NASA/TM--1999-209732

Construction of a Chemical Sensor/

Instrumentation Package Using Fiber Optic

and Miniaturization Technology

(MSFC Center Director's Discretionary Fund Final Report,

Project No. 97-12)

R.L. Newton

Marshall Space Flight Center, Marshall Space Flight Center, Alabama

October 1999

https://ntrs.nasa.gov/search.jsp?R=19990111739 2018-05-14T13:09:51+00:00Z
The NASA STI Program Office...in Profile

Since its founding, NASA has been dedicated to

the advancement of aeronautics and space

science. The NASA Scientific and Technical

Information (STI) Program Office plays a key

part in helping NASA maintain this important

role.

The NASA STI Program Office is operated by

Langley Research Center, the lead center for

NASA's scientific and technical information. The

NASA STI Program Office provides access to the

NASA STI Database, the largest collection of

aeronautical and space science STI in the world. The

Program Office is also NASA's institutional

mechanism for disseminating the results of its

research and development activities. These results

are published by NASA in the NASA STI Report

Series, which includes the following report types:

TECHNICAL PUBLICATION. Reports of

completed research or a major significant phase

of research that present the results of NASA

programs and include extensive data or

theoretical analysis. Includes compilations of

significant scientific and technical data and

information deemed to be of continuing reference

value. NASA's counterpart of peer-reviewed

formal professional papers but has less stringent

limitations on manuscript length and extent of

graphic presentations.

TECHNICAL MEMORANDUM. Scientific and

technical findings that are preliminary or of

specialized interest, e.g., quick release reports,

working papers, and bibliographies that contain

minimal annotation. Does not contain extensive

analysis.

CONTRACTOR REPORT. Scientific and

technical findings by NASA-sponsored

contractors and grantees.

CONFERENCE PUBLICATION. Collected

papers from scientific and technical conferences,

symposia, seminars, or other meetings sponsored

or cosponsored by NASA.

SPECIAL PUBLICATION. Scientific, technical,

or historical information from NASA programs,

projects, and mission, often concerned with

subjects having substantial public interest.

TECHNICAL TRANSLATION.

English-language translations of foreign scientific

and technical material pertinent to NASA's

mission.

Specialized services that complement the STI

Program Office's diverse offerings include creating

custom thesauri, building customized databases,

organizing and publishing research results...even

providing videos.

For more information about the NASA STI Program

Office, see the following:

• Access the NASA STI Program Home Page at

http://www.sti.nasa.gov

• E-mail your question via the Internet to

help@ sti.nasa.gov

• Fax your question to the NASA Access Help

Desk at (301) 621-0134

• Telephone the NASA Access Help Desk at (301)

621-0390

Write to:

NASA Access Help Desk

NASA Center for AeroSpace Information

7121 Standard Drive

Hanover, MD 21076-1320
NASA/TM--1999-209732

Construction of a Chemical Sensor/

Instrumentation Package Using Fiber Optic

and Miniaturization Technology

(MSFC Center Director's Discretionary Fund Final Report,

Project No. 97-12)

R.L. Newton

Marshall Space Flight Center, Marshall Space Flight Center, Alabama

National Aeronautics and

Space Administration

Marshall Space Flight Center

October 1999
NASA Center _r AeroSpacelntbnnation

7121 Standard Drive

Hanove_MD 21076-1320

(301)621-0390

Available from:

ii

National Technical tnfornlation Service

5285 Port Royal Road

Springfield, VA 2216 I

(703) 487-4650
TABLE OF CONTENTS

1. INTRODUCTION ........................................................................................................................

1.1 Research Motivation .............................................................................................................

1.2 Fiber Optic Technology ........................................................................................................

1.3 Micromanufacturing/Microsystems Technology ..................................................................

2. EXPERIMENTAL APPROACH ..................................................................................................

2.1 Spectrum Selection ...............................................................................................................

2.2 Spectrometer Selection .........................................................................................................

3. DISCUSSION OF RESULTS ......................................................................................................

3.1 Experimental Data ................................................................................................................

3.2 Data Analysis and Discussion ..............................................................................................

4. CONCLUSIONS AND RECOMMENDATIONS .......................................................................

APPENDIX--Microspectrometer Technical Specifications ...............................................................

REFERENCES ....................................................................................................................................

6

6

6

9

9

ll

12

13

14

°,.

III

LIST OF FIGURES

I.

2.

3.

4.

5.

6.

7.

8.

9.

10.

II.

12.

13.

14.

Fiber optic gyroscope ..........................................................................................................

Schematic of cladding applied to optical fiber ....................................................................

A comparison of the nano. micro, and macro "world". ......................................................

SEM of a top view of the multiturned micromachined inductor ........................................

Electromagnetic spectrum investigated in this experiment .................................................

Exploded view of spectrometer diffraction grating .............................................................

Photodiode cross section .....................................................................................................

Test apparatus for the sampling of iodine ...........................................................................

Closeup view of test configuration for iodine analysis .......................................................

Raw data of spectrum distilled water "'blank" used in iodine investigation ........................

Three parts per million iodine in distilled water. ................................................................

Thirty parts per million iodine in distilled water. ...............................................................

Three hundred parts per million iodine in distilled water ...................................................

"'Raw" data calibration curve for iodine ..............................................................................

,-)

,-)

4

10

10

11

V

LIST OF ACRONYMS/ABBREVIATIONS

CDDF

FOG

LIGA

MEMS

SiO_

Center Director's Discretionary Fund

fiberoptic gyroscope

Lithographie, Galvanoformung, Abformung

microelectromechanical systems

silica

vii

TECHNICAL MEMORANDUM

CONSTRUCTION OF A CHEMICAL SENSOR/INSTRUMENTATION PACKAGE

USING FIBER OPTIC AND MINIATURIZATION TECHNOLOGY

(MSFC Center Director's Discretionary Fund Final Report, Project No. 97-12)

1. INTRODUCTION

1.1 Research Motivation

Since the beginning of space flight, lowering the cost of access to space has been a priority. Over

the past several years, NASA has refocused its efforts on providing low-cost access to space. The stated

goal of reducing the cost of a launch by a factor of 100 by Administrator Golden I necessitates the need

to employ advanced materials and technologies into launch vehicles. The use of composite materials and

more efficient engines allows significant reductions in weight while retaining required strength and

thrust. Whether used for instrumentation related to payloads or for vehicle health monitoring, fiber optic

and microsystem technology may provide additional savings in weight and volume. The purpose of this

research is to construct and evaluate a sensor instrument package using the technologies mentioned

above.

1.2 Fiber Optic Technology

The fiber optic industry has experienced tremendous growth in the last several years. This growth

has been driven in large part by the communications industry. In 1997 over $9 billion were spent within

the United States on fiber optic equipment. 2 To date, 12 billion km of fiber optic cable have been

installed in America) This has resulted in making optical tibet a low-cost, easily obtained product.

Optical fibers for light transmission serve as waveguides for the light signals. The retention

of light within the optical fiber is made possible by having the light pass through the central core glass

which has a higher refractive index than the outer clad glass. It is by monitoring this refractive index

change that the fiber can itself be used as a "'sensing" device and not only to carry voice or data transmissions. The vast majority of optical fiber is made of silica (SiO2); however, fibers made of plastic

are now commercially available. Optical sensors are capable of measuring a variety of properties, 4 many

of which are listed below:

• Temperature

• Pressure

• Flow

• Liquid level

• Electric fields

• Magnetic fields

• Vibration

• Acceleration

• Radiation

• Strain

• Displacement (position)

• Force

• Rotation

• Chemical species

• pH

• Velocity

• Humidity

• Acoustic fields
Manycompanieshaveshownparticularinterestin fiberoptic gyroscopes(FOG's).Theall solid

statenatureof theFOG givesit severaladvantagesovertheir mechanicalcounterparts.Figure 1shows

thebasiccomponentsof a FOG.Theseincludelow cost,long shelflife, rapidstartup,smallsize,low

weight,andruggedconstruction.FOG'sexploit theSagnaceffect,in whichtwo beamstravelingin

oppositedirectionsaroundacircle will arrivebackat amovingreferencepoint with aphasedifference

dueto rotation.5

Detector

: Modulator ......

Polarizer

Lioht

Source

Figure I. Fiber optic gyroscope. 6

Fiber optic sensors are currently available for environmental and biological monitoring. Over the

past several years fiber optic sensors have been fabricated to measure a range of compounds such as

ammonia, oxygen, methane, lead, and many hydrocarbons. 7 Many of these sensors operate by the

application of a coating or cladding material to the outside of a bare optical fiber. This cladding material

contains molecules that wilt chemically react with specific target compounds. The interaction of the light

with the cladding material when target compounds are present will be of a different wavelength than

light interacting with unreacted cladding material. Figure 2 illustrates this specifically for a biosensor, 8

but the principle is essentially the same as that used in a variety of environmental and biosensor applications. This appears to be a very promising technology for the detection of agents that might be used in

chemical and biological warfare. One day soldiers will perhaps wear clothing that have these fiber optic

sensors woven directly into them and will know instantaneously whether or not they are being exposed

to harmful agents.

Figure 2. Schematic of cladding applied to optical fiber.
A realstrengthof fiberoptic sensorsisthattheycanactasdistributedsensorsthatcandetect

changesovertheentirelengthof thefiber,whichmaybemetersor kilometersin length. This technique

is beingdemonstratedby thedevelopmentof adistributeddosimeternetworkfor detectinghydrazine

vaporatrocketlaunchsites.9Theeffectivenessof thisconfigurationhasbeendemonstratedby using

fibersaslongasI km for sensorapplications.

Anotherpromisingapplicationof fiberoptic sensorsis in theareaof strainsensors.Operatingin

thismode,the fibers themselves are monitored for changes in length or optical path of the light beam.

This change can be calibrated to movement. This technology is already being applied to the monitoring

of civil engineering construction such as bridges. 10 Langley Research Center designed fiber optic sensors to measure strain, temperature, and hydrogen on the X-33 and shuttle fuel tanks. These sensors

must operate within an environmental temperature range of-252 to 121 °C and withstand launch and

reentry. 11

1.3 Micromanufaeturing]Microsystems Technology

Worldwide, micromanufacturing technology is being employed in automotive, communications,

medical, and environmental markets. The global market is estimated at between $6 and $14 billion for

1998 and, according to one estimate, is projected upward of $38 billion by 2002. Much of the cun'ent

market is focused on accelerometers used in automobile air bag systems and inkjet printer head technology. There are more than 600 companies, universities, and research organizations worldwide currently

engaged in this technology. 12 This technology shows particular promise in space-related applications

where size, cost, and weight are critical issues in hardware design.

Micromanufacturing, in the narrow sense, comprises the use of a set of manufacturing tools

based on batch thin-fihn fabrication techniques commonly used in the electronics industry. In the

broader sense, micromanufacturing describes one of many precision engineering disciplines which take

advantage of serial direct write technologies, as well as of more traditional precision machining methods. These tools are used for creating small three-dimensional structures with dimensions ranging from

subcentimeters to submicrometers, involving sensors, actuators, or other microcomponents and

microsystems. 13 Another term that is often used with respect to micromanufacturing and related fields is

microelectromechanical systems (MEMS).

Figure 3 illustrates the scaling of microsystems to other technologies and compares this with the

size of items familiar to us. Many of the features and components of MEMS devices are much smaller

than a grain of sand. For example, in figure 4 the components of this device are on the order of hundreds

of microns. Overall, MEMS is considered an enabling technology, one that will allow for new applications for existing instrumentation and sensing techniques.

A very interesting aspect of MEMS technology is being applied to the area of micropropulsion.

One particular resealvh in this area has focused on developing thrusters with impulses of 104 to 10-('

N-sec for high-accuracy station-keeping and attitude control of microspacecraft (e.g. <1 kg). 14This

technology is still in its infancy. Many issues related to the basic properties of materials must be examined on the microscale before MEMS propulsion, or for that matter, many other MEMS devices find

widespread use other than the specific applications already mentioned.
Micromachined

Gear _+100pm

-- 1 m I Elephant_+2m high I

-- 0.1 m=10 cm

'0,

0.01 m=l I ICChip +1 cm Wide [I cm

I Grain of Sand -+1mm I

,.,,. lmm

I Dust Particle_+1-5Nm ]

I Quantum Structures_+200 Electron],c A i

-...,,. 0.1 mm=100pm [Biological Cell-+18pm ]

z

.,....-- 0.01 mm=10 pm(,..... O _

..m lpm

Micromachining __

"_. i i I Smallest IC Features_+0.35_m i

I 0.1 _tm=100am

NanomachiningI _ %e

r" ] O.Ol pm=lo nm :u@

I Atomic Lettering Using i

Scanning Tunneling

Microscope

Atoms 1-4 ,&,]

1 nm=10 ,&,

0.1 nm=IA

I DNA_+2nm wide I

Figure 3. A comparison of the nano, micro, and macro "world".]5
ElectroplatedMagneticCore

Air Gap(50pm)

BondingPad

ElectroplatedConductorLines

Figure 4. SEM of a top view of the multiturned micromachined inductor. 16

Spectrometers such as the one used in this investigation have been one of the first analytical tools

miniaturized using MEMS technology, due in part to the ability to fabricate microdiffraction gratings.

Spectrometers measure the transmission of light through a sample, and according to Beer's law, the

absorbance of a solute in a solution is a function of its concentration at a particular wavelength. Thus,

absorbance measurements can be used to determine the concentration of solutions. 17

5
2. EXPERIMENTAL APPROACH

2.1 Spectrum Selection

The goal of this research was to combine fiber optic sensor technology with miniaturization

technology to yield a lightweight, low-cost sensor package. An additional goal of the research was to

incorporate wireless data transmission from the sensor package to a computer located some distance

away (e.g., 100 ft). It was also decided that the research would initially focus on collecting measurements of species located in the visible region (see fig. 5) of the electromagnetic spectrum due to the

availability of miniaturized spectrophotometers manufactured for sensing in the ultraviolet-visible range.

Iodine is used in the Space Station water recovery system for disinfectant purposes and has a maximum

absorption at 462 nm. Therefore, initial tests were focused on the remote, aqueous detection of this

chemical.

Wavelength(cm)

10-10 10 -8 10-6 10-4 10-2 1 10 2 104

Figure 5.

1019 1017 1015 10 _3 10 _ 10 9 107

Frequency (Hz)

Electromagnetic spectrum investigated in this experiment. 18

2.2 Spectrometer Selection

The spectrometer selected for the iodine studies is manufactured by microParts j9 and is supplied

with a 1-m fiber optic pigtail permanently attached to the spectrometer. Specifications of the spectrometer are given in the appendix. Microsystem technology is employed in this spectrometer via the selffocusing reflection grating, as indicated in figure 6. This particular grating was manufactured using the

Lithographie, Galvanoformung, Ablormung (LIGA) method.

The light entering the spectrometer via the optical fiber is separated on the curved reflection

grating located at the far end of the enclosure. The individual wavelengths of light are then directed back

to a diode array which converts the light intensity into an electrical response via a photodiode that is then

transmitted to the data system. A diagram of one photodiode that would be coupled with many others to

form an array is shown is figure 7. Photodiodes consist of a layer of silicon doped with atoms carrying

extra valence electrons (p-type semiconductors) on top of a layer doped with atoms carrying one valence

electron less than silicon (n-type) semiconductors. 20

6
Input ]

;_1 -_'x //

/

Optical Fiber

Figure 6.

Self-Focusing

Reflection Grating

(D=0,2 pro; G=2pm) )

Light Output

With 45° Sidewall

DiodeArray

Exploded view of spectrometer diffraction grating.

p-n Photodiode

Si02

Depletion Metal

Region Contacts

Figure 7. Photodiode cross section.

Since a fiber optic probe was already attached to the spectrometer, the experiments were performed using the spectrometer as a "passive" device. That is, the spectrometer only responded to the

amount and wavelength of light that was introduced into the fiber tip. No coating or cladding was

applied to the fiber. The test setup for the aqueous iodine solution experiment is given in figures 8 and 9.

The input signal from the fiber optic cable is processed by the microspectrometer and converted

to an electrical signal that is relayed via the wireless modem to another wireless modem connected to a

PC in a laboratory =30 m away. Care must be taken to ensure that no dirt or liquid enters the fiber tip.

Also, the fiber will break if bent in too tight of a radius. On this particular model, the fiber is permanently mated to the spectrometer so a broken fiber renders the complete spectrometer useless. The

spectrorneter measures 13 mm x 35 mm x 2.5 mm.
Wireless

Modem and

Controller

i tilll_tltJi!!tfttI

I

Microspectrometer

Figure 8. Test apparatus for the sampling of iodine.

Light Enters

at End of

Fiber

_,, Containing

Iodine

Solution

!

Figure 9. Closeup view of test configuration for iodine analysis.

8
3. DISCUSSION OF RESULTS

3.1 Experimental Data

In order to test the sensitivity and performance of the spectrometer a set of iodine solutions were

prepared at various concentration levels. Additionally, distilled water was used as a "blank" in which no

iodine was present. As previously mentioned, iodine has a maximum absorption at 462 nm. Figure 10

demonstrates the spectrum for the blank and figures 11-13 show the spectrum of three concentrations

of iodine.

g

0

-0.02

-0.04

-0.06

-0.08

300

I I I I I I

400 462 500 600 700 800

Wavelength(nm)

Figure 10. Raw data of spectrum distilled water "blank" used in iodine investigation.

=

e0.06

0.04

0.02

-0.02 I I I J

300 400 462 500 600 700 800

Wavelength(nm)

Figure 11. Three parts per million iodine in distilled water.

9
0.05

0

-0.05

x

_J

-0.10

-0.15 I I i I I

300 400 462 500 600 700 800

Wavelength(nm)

Figure 12. Thirty parts per million iodine in distilled water.

0.15

0.10

0.05

x

'1' 0

-0.05

I I I I I I I I

400 450 462 500 550 600 650 700 750

Wavelenolh (nm)

Figure 13. Three hundred parts per million iodine in distilled water.

10
3.2 Data Analysis and Discussion

The spectrometer exhibited somewhat good sensitivity all the way down to 3 ppm. However, as

figure 14 demonstrates, the response does not scale with linear changes in iodine concentration, possibly

because of the need for more input signal into the spectrometer at higher dilutions of the iodine mixtures. The use of distilled water should have eliminated any potential for interference from other species.

The wireless data transfer via the radio modem worked well. The range on these particular units

is on the order of a few hundred meters, but with modification, wireless data can be transmitted several

kilometers, even with low-cost units. Also, in many terrestrial applications of this sensor technology,

data could easily be transmitted using the internet.

0.15

0.1

0.05

0

-0.05

- y=O.O42.x-O.045_b,

I I I I

Figure 14.

3 30

Iodine(ppm)

"Raw" data calibration curve for iodine.

300

11
4. CONCLUSIONS AND RECOMMENDATIONS

This Center Director's Discretionary Fund (CDDF) was designed as a 2-yr investigation into the

construction and testing of an instrument assembly based on fiber optic and miniaturization technology.

However, during the first year of this CDDF project, the principle investigator was granted a full-time

study award from Marshall Space Flight Center management. In order for this effort to be completed

in a timely manner and since the basic objectives of the research were demonstrated, the investigation

was only carried through year one.

Fiber optic sensing is a maturing technology. This research examined and demonstrated that the

technology is viable for the detection of chemical compounds of interest. Commercially procured sensors are currently available from a wide selection of vendors. The feasibility of using miniaturization

technology was also demonstrated. Even on very simplistic experiments, the instrumentation was able

to resolve low concentrations (few ppm) of iodine in an aqueous solution.

By the integration of fiber optic sensors and miniaturization technology, size and weight reductions were realized in the construction of an instrument package. Due to the development of this technology by industry and academia, NASA should see positive benefits from future utilization of this

technology in its programs.

12
APPENDIXmMicrospectrometer Technical Specifications

Component

Material:

Dimension:

Spectral range:

Blaze wavelength:

Grating constant:

Transmission:

Spectral distance:

Resolution:

Order:

poly(methyimethocrylate)

! 3.5 nrn x 35 mm

370-850 nm

560 nm

1.7 Bm

max. 15 percent

0.2 nm/Bm

7 nm (12 nm) with standard diode array

first

Fiber

Step index:

Numerical aperture:

Pigtail length:

Connector:

50/125 Bm or 105/125 Bm

0.22

lm

Optional

Photodiode Array

Hamamatsu CMOS series or others

Applications

Color measurement

Environmental analysis

Chemical analysis

Medical analysis

13
REFERENCES

.

2.

.

4.

.

.

.

.

.

10.

11.

12.

Wingo, W.: Design News, Vol. 54, No. 12, p. 28, June 21, 1999.

Multimedia Telecommunications Association--1997 Multimedia Telecommunications

MarketReview & Forecast, Multimedia Telecommunications Association: 68, January 1, 1998.

Worldwide Fiber and Cable: Demand, Productions, and Price, Lightwave, Vol. 72, January 1999.

"Optical Sensor Technologies," Optoelectronics in Japan and the United States (online), 1992,

URL: http://itri.loyola.edu/opto/c6 s3.htm (cited June 30, 1999).

_ " 13" _ Hewish, M.: Penetratm= the Fiber-Optic Fog, International Defense Review, Vol. 34, No. 5,

p. 151, April 1, 1999.

Associated Products, The Boeing Company (online), URL:http://www.boeing.com/assocproducts/

mdip/fog.htm (cited June 27, 1999).

Tatterson, K.G.: "Pollutants Can't Hide from Fiber Optic Sensors," Photonics Spectra,

pp. 116-118, April 1998.

Appell, D.: "Clad Fiber Detects Biological Agents Fast," Laser Focus World, pp. 26-28,

April 1998.

Klimcak, C.; Radhakrishnan, G.; and Jaduszliwer, B.: "A Remote Fiber Optic Dosimeter Network

for Detecting Hydrazine Vapor," The Aerospace Corporation, Los Angeles, CA.

Iraqi, A.; Morawski, R.Z.; Barwicz, A.; and Bock, W.J.: Distributed Data Processing in a Telemetric System for Monitoring Civil Engineering Constructions, IEEE Instrumentation and Measurement Technology Conference, St. Paul, MN, May 18-20, 1998.

Fiber Sensors'for Space Vehicle Health Monitoring of RLVs, Ames Research Center (online),

URL:http://ack.arc.nasa.gov/ic/projects/photonics/OS/HealthSensors/health.html (cited June 27,

1999).

Grace, R.H.: "The Commercialization of MES/MST: Where Do We Go From Here'?" Senso_:_,

Vol. 16, No 2, pp. 4-8, 1999.

Madou, M.: Fundamentals ofMicro.J_tbrication, CRC Press, New York, NY, p. 1, 1997.

Lewis, D.H.; Siegfried, W.J.; Cohen, R.B.: and Antonsson, E.K.: Digital Micropropulsion,

MEMS'99, Orlando, FL, 1999.

14
15. Madou, M.: Fundamentals qfMicrofabrication, CRC Press, New York, NY, Figure 9.1, 1997.

16. Allen, M.G.; and Park, J.: "Bar-Type Microinductors and Microtransformers with Electroplated

Alloy Cores", Georgia Tech Micromachine Group (online), URL:http://mems.mirc.gatech.edu/

research/magnetic.html (cited June 30, 1999)

17. Shugar, I.; and Gershon, J.: Chemical Technicians'Ready Reference Handbook, McGraw-Hill,

New York, NY, 1981.

18. Graphic provided by: Galactic Industries Corporation, 395 Main Street, Salem, NH 03079 USA.

19. microParts, American Laubscher Corporation, ALC Building, 80 Finn Court, Farmingdale, NY

11735-1107.

20. "Laser Focus World," Back to Basics, p. 95, May 1996.

15
REPORT DOCUMENTATION PAGE FormApprove_

OMB No. 0704-0188

Public reporting b_Jrden for this collection of information is estimated to average 1 hour per response, including the time tor reviewing instructions, searching existing data sources,

gathering and maintaining the data needed, and completing and reviewing the collection of information, Send comments regarding this burden estimale Orany other aspect of this

collection o! inlormation, including suggestions for reducing this burden, to Washington Headquarters Services, Directorate tot Information Operation and Reports, 1215 Jefferson

Davis Highway, Suite 1204, Arlington, VA 22202-4302, and to the Office of Management and Budget, Paperwork Reduction Project (0704-0188), Washington, DC 20503

1. AGENCY USE ONLY (Leave Blank) 2. REPORT DATE 3. REPORT TYPE AND DATES COVERED

October 1999 Technical Memorandum

4. TITLE AND SUBTITLE 5. FUNDING NUMBERS

Construction of a Chemical Sensor/Instrumentation Package

Using Fiber Optic and Miniaturization Technology

(MSFC Center Director's Discretionary Fund Final Report, Project No. 97-12)

6. AUTHORS

R.L. Newton

7. PERFORMINGORGANIZATIONNAMES(S)ANDADDRESS(ES)

George C. Marshall Space Flight Center

Marshall Space Flight Center, Alabama 358 i 2

9. SPONSORING/MONITORINGAGENCYNAME(S)ANDADDRESS(ES)

National Aeronautics and Space Administration

Washington, DC 20546-0001

8. PERFORMING ORGANIZATION

REPORT NUMBER

M-943

10. SPONSORING/MONITORING

AGENCY REPORT NUMBER

NASA/TM-- 1999-209732

11. SUPPLEMENTARY NOTES

Prepared by Materials, Processes, and Manufacturing Department, Engineering Directorate

12a. DISTRIBUTION/AVAILABILITY STATEMENT

Unclassified-Unlimited

Subject Category 35

Nonstandard Distribution

12b. DISTRIBUTION CODE

13. ABSTRACT (Maximum 200 words)

The objective of this research was to construct a chemical sensor/instrumentation package that

was smaller in weight and volume than conventional instrumentation. This reduction in weight

and volume is needed to assist in further reducing the cost of launching payloads into space. To

accomplish this, fiber optic sensors, miniaturized spectrometers, and wireless modems were

employed. The system was evaluated using iodine as a calibration analyte.

14, SUBJECT TERMS

fiber optics, MEMS, sensors

17. SECURITY CLASSIFICATION

OF REPORT

Unclassified

NSN 7540-01-280-5500

18. SECURITY CLASSIFICATION

OF THIS PAGE

Unclassified

15. NUMBER OF PAGES

24

16. PRICE CODE

A03

19. SECURITY CLASSIFICATION 20. LIMITATION OF ABSTRACT

OF ABSTRACT

Unclassified Unlimited

Standard Form 298 (Rev. 2-89)

Prescnbed byANSi Std 239-18

298-102

National Aeronautics and

Space Administration

AD33

George C. Marshall Space Flight Center

Marshall Space Flight Center, Alabama

35812