# Official Journal L 63/2020

| | ISSN 1977-0677 |
| --- | --- |
| Official Journal
  of the European Union | L 63 |
| | |
| English edition | Legislation | Volume 633 March 2020 |

ISSN 1977-0677

Official Journal

of the European Union

L 63

English edition

Legislation

Volume 633 March 2020

---

| Contents | | II   Non-legislative acts | page |
| --- | --- | --- | --- |
| | | REGULATIONS |
| | * | Commission Implementing Regulation (EU) 2020/349 of 2 March 2020 amending Implementing Regulation (EU) 2019/1916 in respect of operational conditions in certain urban or interurban areas | 1 |
| | | DECISIONS |
| | * | Commission Implementing Decision (EU) 2020/350 of 28 February 2020 amending Decision 2002/364/EC as regards definitions of first–line assays and confirmatory assays, requirements for devices for self-testing and requirements for HIV and HCV rapid tests, confirmatory and supplementary assays
  (notified under document C(2020) 1086)

   (
  1
  ) | 3 |

Contents

II   Non-legislative acts

page

REGULATIONS

*

Commission Implementing Regulation (EU) 2020/349 of 2 March 2020 amending Implementing Regulation (EU) 2019/1916 in respect of operational conditions in certain urban or interurban areas

1

DECISIONS

*

Commission Implementing Decision (EU) 2020/350 of 28 February 2020 amending Decision 2002/364/EC as regards definitions of first–line assays and confirmatory assays, requirements for devices for self-testing and requirements for HIV and HCV rapid tests, confirmatory and supplementary assays
  (notified under document C(2020) 1086)

   (
  1
  )

(notified under document C(2020) 1086)

1

3

| | |
| --- | --- |
| | (1)   Text with EEA relevance. |

---

(1)   Text with EEA relevance.

1

| EN | Acts whose titles are printed in light type are those relating to day-to-day management of agricultural matters, and are generally valid for a limited period.
  The titles of all other Acts are printed in bold type and preceded by an asterisk. |
| --- | --- |

EN

Acts whose titles are printed in light type are those relating to day-to-day management of agricultural matters, and are generally valid for a limited period.

The titles of all other Acts are printed in bold type and preceded by an asterisk.

---

II Non-legislative acts

REGULATIONS

| 3.3.2020 | EN | Official Journal of the European Union | L 63/1 |
| --- | --- | --- | --- |

3.3.2020

EN

Official Journal of the European Union

L 63/1

---

COMMISSION IMPLEMENTING REGULATION (EU) 2020/349

of 2 March 2020

amending Implementing Regulation (EU) 2019/1916 in respect of operational conditions in certain urban or interurban areas

THE EUROPEAN COMMISSION,

Having regard to the Treaty on the Functioning of the European Union,

Having regard to Council Directive 96/53/EC of 25 July 1996 laying down for certain road vehicles circulating within the Community the maximum authorised dimensions in national and international traffic and the maximum authorised weights in international traffic (1), and in particular Article 8b(4) thereof,

Whereas:

| (1) | Following the adoption of Commission Implementing Regulation (EU) 2019/1916 (2), it has appeared that, in respect of operational conditions in certain urban or interurban areas, its terms were not consistent with Article 8b(3) of Directive 96/53/EC. |
| --- | --- |

(1)

Following the adoption of Commission Implementing Regulation (EU) 2019/1916 (2), it has appeared that, in respect of operational conditions in certain urban or interurban areas, its terms were not consistent with Article 8b(3) of Directive 96/53/EC.

| (2) | Therefore Implementing Regulation (EU) 2019/1916 should be amended accordingly. |
| --- | --- |

(2)

Therefore Implementing Regulation (EU) 2019/1916 should be amended accordingly.

| (3) | This Regulation should enter into force as a matter of urgency on the day following that of its publication in the Official Journal of the European Union. |
| --- | --- |

(3)

This Regulation should enter into force as a matter of urgency on the day following that of its publication in the Official Journal of the European Union.

| (4) | The measures provided for in this Regulation are in accordance with the opinion of the Committee referred to in Article 10i(2) of Directive 96/53/EC, |
| --- | --- |

(4)

The measures provided for in this Regulation are in accordance with the opinion of the Committee referred to in Article 10i(2) of Directive 96/53/EC,

HAS ADOPTED THIS REGULATION:

Article 1

Amendment to Implementing Regulation (EU) 2019/1916

In Article 3, paragraph 1 is replaced by the following:

‘1.   Member States may prohibit the circulation of vehicles or vehicle combinations equipped with devices in the in-use position in view of the special characteristics of urban or interurban areas where speed limits are 50 km/h or lessand where vulnerable road users are likely to be present.’.

Article 2

Entry into force

This Regulation shall enter into force on the day following that of its publication in the Official Journal of the European Union.

Official Journal of the European Union

This Regulation shall be binding in its entirety and directly applicable in all Member States.

Done at Brussels, 2 March 2020.

For the Commission

The President

Ursula VON DER LEYEN

---

(1)  
  OJ L 235, 17.9.1996, p. 59.

1

(2)  Commission Implementing Regulation (EU) 2019/1916 of 15 November 2019 laying down detailed provisions as regards the use of rear aerodynamic devices pursuant to Council Directive 96/53/EC (OJ L 297, 18.11.2019, p. 3).

2

---

DECISIONS

| 3.3.2020 | EN | Official Journal of the European Union | L 63/3 |
| --- | --- | --- | --- |

3.3.2020

EN

Official Journal of the European Union

L 63/3

---

COMMISSION IMPLEMENTING DECISION (EU) 2020/350

of 28 February 2020

amending Decision 2002/364/EC as regards definitions of first–line assays and confirmatory assays, requirements for devices for self-testing and requirements for HIV and HCV rapid tests, confirmatory and supplementary assays

(notified under document C(2020) 1086)

(Text with EEA relevance)

THE EUROPEAN COMMISSION,

Having regard to the Treaty on the Functioning of the European Union,

Having regard to Directive 98/79/EC of the European Parliament and of the Council of 27 October 1998 on in vitro diagnostic medical devices (1), and in particular the second subparagraph of Article 5(3) thereof,

Whereas:

| (1) | Pursuant to the first subparagraph of Article 5(3) of Directive 98/79/EC, Member States are to presume compliance with the essential requirements referred to in Article 3 of that Directive in respect of devices designed and manufactured in conformity with common technical specifications. The common technical specifications for in vitro diagnostic medical devices are laid down in Commission Decision 2002/364/EC (2). |
| --- | --- |

(1)

Pursuant to the first subparagraph of Article 5(3) of Directive 98/79/EC, Member States are to presume compliance with the essential requirements referred to in Article 3 of that Directive in respect of devices designed and manufactured in conformity with common technical specifications. The common technical specifications for in vitro diagnostic medical devices are laid down in Commission Decision 2002/364/EC (2).

| (2) | In the interest of public health and patient safety and in order to reflect scientific and technological progress, including the evolution in the intended use, performance, and analytical sensitivity of certain devices, it is appropriate to update the common technical specifications laid down in Decision 2002/364/EC. |
| --- | --- |

(2)

In the interest of public health and patient safety and in order to reflect scientific and technological progress, including the evolution in the intended use, performance, and analytical sensitivity of certain devices, it is appropriate to update the common technical specifications laid down in Decision 2002/364/EC.

| (3) | The definitions of first-line assays and confirmatory assays, requirements for devices for self-testing and requirements for HIV and HCV rapid tests, confirmatory and supplementary assays should be amended in order to take into account the evolved state of the art, the changes in clinical needs, new scientific knowledge available and the new types of devices present on the market. |
| --- | --- |

(3)

The definitions of first-line assays and confirmatory assays, requirements for devices for self-testing and requirements for HIV and HCV rapid tests, confirmatory and supplementary assays should be amended in order to take into account the evolved state of the art, the changes in clinical needs, new scientific knowledge available and the new types of devices present on the market.

| (4) | The manufacturers should be allowed time to adapt to the changes in common technical specifications. The date of application of this Decision should therefore be deferred. However, in the interest of public health and patient safety, manufacturers should be allowed to comply with the common technical specifications as amended by this Decision before its date of application on a voluntary basis. |
| --- | --- |

(4)

The manufacturers should be allowed time to adapt to the changes in common technical specifications. The date of application of this Decision should therefore be deferred. However, in the interest of public health and patient safety, manufacturers should be allowed to comply with the common technical specifications as amended by this Decision before its date of application on a voluntary basis.

| (5) | The measures provided for in this Decision are in accordance with the opinion of the Committee established by Article 6(2) of Council Directive 90/385/EEC (3), |
| --- | --- |

(5)

The measures provided for in this Decision are in accordance with the opinion of the Committee established by Article 6(2) of Council Directive 90/385/EEC (3),

HAS ADOPTED THIS DECISION:

Article 1

The Annex to Decision 2002/364/EC is amended in accordance with the Annex to this Decision.

Article 2

1.   This Decision shall apply from 2 March 2021.

2.   Notwithstanding paragraph 1, from 2 March 2020 until 1 July 2020 Member States shall apply the presumption of compliance laid down in Article 5(3) of Directive 98/79/EC for all in vitro diagnostic medical devices that comply with any of the following:

| (a) | the common technical specifications laid down in Decision 2002/364/EC as amended by Commission Decision 2011/869/EU (4); |
| --- | --- |

(a)

the common technical specifications laid down in Decision 2002/364/EC as amended by Commission Decision 2011/869/EU (4);

| (b) | the common technical specifications laid down in Decision 2002/364/EC as amended by Commission Implementing Decision (EU) 2019/1244 (5); |
| --- | --- |

(b)

the common technical specifications laid down in Decision 2002/364/EC as amended by Commission Implementing Decision (EU) 2019/1244 (5);

| (c) | the common technical specifications laid down in Decision 2002/364/EC as amended by this Decision. |
| --- | --- |

(c)

the common technical specifications laid down in Decision 2002/364/EC as amended by this Decision.

3.   Notwithstanding paragraph 1, from 2 July 2020 until 1 March 2021 Member States shall apply the presumption of compliance laid down in Article 5(3) of Directive 98/79/EC for all in vitro diagnostic medical devices that comply with either of the following:

| (a) | the common technical specifications laid down in Decision 2002/364/EC as amended by Implementing Decision (EU) 2019/1244; |
| --- | --- |

(a)

the common technical specifications laid down in Decision 2002/364/EC as amended by Implementing Decision (EU) 2019/1244;

| (b) | the common technical specifications laid down in Decision 2002/364/EC as amended by this Decision. |
| --- | --- |

(b)

the common technical specifications laid down in Decision 2002/364/EC as amended by this Decision.

Article 3

This Decision is addressed to the Member States.

Done at Brussels, 28 February 2020.

For the Commission

Stella KYRIAKIDES

Member of the Commission

---

(1)  
  OJ L 331, 7.12.1998, p. 1.

1

(2)  Commission Decision 2002/364/EC of 7 May 2002 on common technical specifications for in vitro-diagnostic medical devices (OJ L 131, 16.5.2002, p. 17).

2

(3)  Council Directive 90/385/EEC of 20 June 1990 on the approximation of the laws of the Member States relating to active implantable medical devices (OJ L 189, 20.7.1990, p. 17).

3

(4)  Commission Decision 2011/869/EU of 20 December 2011 amending Decision 2002/364/EC on common technical specifications for in vitro diagnostic medical devices (OJ L 341, 22.12.2011, p. 63).

4

(5)  Commission Implementing Decision (EU) 2019/1244 of 1 July 2019 amending Decision 2002/364/EC as regards requirements for HIV and HCV antigen and antibody combined tests and as regards requirements for nucleic acid amplification techniques with respect to reference materials and qualitative HIV assays (OJ L 193, 19.07.2019, p.1).

5

---

ANNEX

The Annex to Decision 2002/364/EC is amended as follows:

| 1. | Section 2 is amended as follows:






  (a) | the following definition of ‘First line assay’ is inserted between the definition of ‘Whole system failure rate’ and the definition of ‘Confirmation assay’:

  ‘
  First-line assay

  First-line assay means an assay used to detect a marker or analyte, and which may be followed by a confirmatory assay. Devices intended solely to be used to monitor a previously determined marker or analyte are not considered first-line assays.’; |
| --- | --- | --- |

1.

Section 2 is amended as follows:

(a)

the following definition of ‘First line assay’ is inserted between the definition of ‘Whole system failure rate’ and the definition of ‘Confirmation assay’:

‘
  First-line assay

First-line assay means an assay used to detect a marker or analyte, and which may be followed by a confirmatory assay. Devices intended solely to be used to monitor a previously determined marker or analyte are not considered first-line assays.’;

| (b) | the definition of ‘Confirmation assay’ is replaced by the following:

  ‘
  Confirmatory assay

  Confirmatory assay means an assay used for the confirmation of a reactive result from a first-line assay.’; |
| --- | --- |

(b)

the definition of ‘Confirmation assay’ is replaced by the following:

‘
  Confirmatory assay

Confirmatory assay means an assay used for the confirmation of a reactive result from a first-line assay.’;

| 2. | Section 3 is amended as follows:






  (a) | Sub-section 3.1.1 is replaced by the following:






  ‘3.1.1. | Devices which detect virus infections shall meet the requirements for sensitivity and specificity set out in Table 1, Table 3, Table 4 and Table 5, which apply to them taking account of the intended purpose of the devices concerned, virus type and entities to be detected (antigen and/or antibody). See also principle 3.1.11 for first-line assays.’; |
| --- | --- | --- | --- |

2.

Section 3 is amended as follows:

(a)

Sub-section 3.1.1 is replaced by the following:

‘3.1.1.

Devices which detect virus infections shall meet the requirements for sensitivity and specificity set out in Table 1, Table 3, Table 4 and Table 5, which apply to them taking account of the intended purpose of the devices concerned, virus type and entities to be detected (antigen and/or antibody). See also principle 3.1.11 for first-line assays.’;

| (b) | Sub-section 3.1.3 is replaced by the following:






  ‘3.1.3. | Devices for self-testing shall meet the same CTS requirements for sensitivity and specificity as respective devices for professional use. Relevant parts of the performance evaluation shall be carried out (or repeated) by appropriate lay users to validate the operation of the device and the instructions for use. The lay users selected for the performance evaluation shall be representative of the intended users groups.
  Performance evaluation of a device for self-testing shall include, for each body fluid claimed for use with the device, e.g. whole blood, urine, saliva, etc., at least 200 lay users that are known positive for the infection and at least 400 lay users that do not know their status, of which at least 200 are at high risk of acquiring the infection. The sensitivity and specificity of the device for self-testing in the hands of lay users shall be defined against the confirmed patient infectious status.’; |
| --- | --- | --- |

(b)

Sub-section 3.1.3 is replaced by the following:

‘3.1.3.

Devices for self-testing shall meet the same CTS requirements for sensitivity and specificity as respective devices for professional use. Relevant parts of the performance evaluation shall be carried out (or repeated) by appropriate lay users to validate the operation of the device and the instructions for use. The lay users selected for the performance evaluation shall be representative of the intended users groups.

Performance evaluation of a device for self-testing shall include, for each body fluid claimed for use with the device, e.g. whole blood, urine, saliva, etc., at least 200 lay users that are known positive for the infection and at least 400 lay users that do not know their status, of which at least 200 are at high risk of acquiring the infection. The sensitivity and specificity of the device for self-testing in the hands of lay users shall be defined against the confirmed patient infectious status.’;

| (c) | Sub-section 3.1.9 is replaced by the following:






  ‘3.1.9. | Performance evaluation of first line assays shall include 25 positive (if available in the case of rare infections) ‘same day’ fresh serum samples (≤ 1 day after sampling).’; |
| --- | --- | --- |

(c)

Sub-section 3.1.9 is replaced by the following:

‘3.1.9.

Performance evaluation of first line assays shall include 25 positive (if available in the case of rare infections) ‘same day’ fresh serum samples (≤ 1 day after sampling).’;

| (d) | Sub-section 3.1.11 is replaced by the following:






  ‘3.1.11. | For performance evaluations for first line assays (Table 1 and Table 3) blood donor populations shall be investigated from at least two blood donation centers and consist of consecutive blood donations, which have not been selected to exclude first time donors.’; |
| --- | --- | --- |

(d)

Sub-section 3.1.11 is replaced by the following:

‘3.1.11.

For performance evaluations for first line assays (Table 1 and Table 3) blood donor populations shall be investigated from at least two blood donation centers and consist of consecutive blood donations, which have not been selected to exclude first time donors.’;

| (e) | Sub-section 3.4.2 is replaced by the following:






  ‘3.4.2. | The manufacturer’s batch release testing for first line assays shall include at least 100 specimens negative for the relevant analyte.’. |
| --- | --- | --- |

(e)

Sub-section 3.4.2 is replaced by the following:

‘3.4.2.

The manufacturer’s batch release testing for first line assays shall include at least 100 specimens negative for the relevant analyte.’.

| 3. | Table 1 is replaced by the following:

  ‘Table 1


  First-line assays, excluding rapid tests: anti-HIV 1/2, HIV 1/2 Ag/Ab, anti-HTLV I/II, anti-HCV, HCV Ag/Ab, HBsAg, anti-HBc | | anti-HIV 1/2, HIV 1/2 Ag/Ab | Anti-HTLV-I/II | anti-HCV, HCV Ag/Ab | HBsAg | Anti-HBc |
| --- | --- | --- | --- | --- | --- | --- | --- |
| Diagnostic sensitivity | Positive specimens | 400 HIV-1
  100 HIV-2
  including 40 non-B-subtypes, all available HIV/1 subtypes shall be represented by at least 3 samples per subtype | 300 HTLV-I
  100 HTLV-II | 400 (positive samples)
  Including samples from different stages of infection and reflecting different antibody patterns.
  Genotype 1-4: > 20 samples per genotype (including non-a subtypes of genotype 4);
  5: > 5 samples;
  6: if available | 400
  including subtype-consideration | 400
  including evaluation of other HBV-markers |
| Sero-conversion panels | 20 panels
  10 further panels (at notified body or manufacturer) | To be defined when available | 20 panels
  10 further panels (at notified body or manufacturer) | 20 panels
  10 further panels (at notified body or manufacturer) | To be defined when available |
| Analytical sensitivity | Standards | | | | 0,130 IU/ml ( WHO International Standard: Third International Standard for HBsAg, subtypes ayw1/adw2, HBV genotype B4, NIBSC code: 12/226) | |
| Specificity | Unselected donors (including first-time donors) | 5 000 | 5 000 | 5 000 | 5 000 | 5 000 |
| Hospitalized patients | 200 | 200 | 200 | 200 | 200 |
| Potentially cross-reacting blood-specimens (RF+, related viruses, pregnant women, etc) | 100 | 100 | 100 | 100 | 100’ |

3.

Table 1 is replaced by the following:

‘Table 1

First-line assays, excluding rapid tests: anti-HIV 1/2, HIV 1/2 Ag/Ab, anti-HTLV I/II, anti-HCV, HCV Ag/Ab, HBsAg, anti-HBc

anti-HIV 1/2, HIV 1/2 Ag/Ab

Anti-HTLV-I/II

anti-HCV, HCV Ag/Ab

HBsAg

Anti-HBc

Diagnostic sensitivity

Positive specimens

400 HIV-1

100 HIV-2

including 40 non-B-subtypes, all available HIV/1 subtypes shall be represented by at least 3 samples per subtype

300 HTLV-I

100 HTLV-II

400 (positive samples)

Including samples from different stages of infection and reflecting different antibody patterns.

Genotype 1-4: > 20 samples per genotype (including non-a subtypes of genotype 4);

5: > 5 samples;

6: if available

400

including subtype-consideration

400

including evaluation of other HBV-markers

Sero-conversion panels

20 panels

10 further panels (at notified body or manufacturer)

To be defined when available

20 panels

10 further panels (at notified body or manufacturer)

20 panels

10 further panels (at notified body or manufacturer)

To be defined when available

Analytical sensitivity

Standards

0,130 IU/ml ( WHO International Standard: Third International Standard for HBsAg, subtypes ayw1/adw2, HBV genotype B4, NIBSC code: 12/226)

Specificity

Unselected donors (including first-time donors)

5 000

5 000

5 000

5 000

5 000

Hospitalized patients

200

200

200

200

200

Potentially cross-reacting blood-specimens (RF+, related viruses, pregnant women, etc)

100

100

100

100

100’

| 4. | Table 3 is replaced by the following:

  ‘Table 3


  Rapid tests: anti-HIV 1/2, HIV 1/2 Ag/Ab, anti-HCV, HCV Ag/Ab, HBsAg, anti-HBc, anti-HTLV I and II | | anti-HIV 1/2, HIV 1/2 Ag/Ab | anti-HCV, HCV Ag/Ab | HBsAg | anti-HBc | anti-HTLV I and II | Acceptance criteria |
| --- | --- | --- | --- | --- | --- | --- | --- | --- |
| Diagnostic sensitivity | Positive specimens | Same criteria as in Table 1 | Same criteria as in Table 1 | Same criteria as in Table 1 | Same criteria as in Table 1 | Same criteria as in Table 1 | Same criteria as in Table 1 |
| Sero-conversion panels | Same criteria as in Table 1 | Same criteria as in Table 1 | Same criteria as in Table 1 | Same criteria as in Table 1 | Same criteria as in Table 1 | Same criteria as in Table 1 |
| Diagnostic specificity | Negative specimens | 21 000 blood donations | 1 000 blood donations | 1 000 blood donations | 1 000 blood donations | 1 000 blood donations | ≥ 99 %
  (anti-HBc: ≥ 96 %)’ |
| 200 clinical specimens | 200 clinical specimens | 200 clinical specimens | 200 clinical specimens | 200 clinical specimens |
| 200 samples from pregnant women | 200 samples from pregnant women | 200 samples from pregnant women | | 200 samples from pregnant women |
| 100 potentially interfering samples | 100 potentially interfering samples | 100 potentially interfering samples | 100 potentially interfering samples | 100 potentially interfering samples |

4.

Table 3 is replaced by the following:

‘Table 3

Rapid tests: anti-HIV 1/2, HIV 1/2 Ag/Ab, anti-HCV, HCV Ag/Ab, HBsAg, anti-HBc, anti-HTLV I and II

anti-HIV 1/2, HIV 1/2 Ag/Ab

anti-HCV, HCV Ag/Ab

HBsAg

anti-HBc

anti-HTLV I and II

Acceptance criteria

Diagnostic sensitivity

Positive specimens

Same criteria as in Table 1

Same criteria as in Table 1

Same criteria as in Table 1

Same criteria as in Table 1

Same criteria as in Table 1

Same criteria as in Table 1

Sero-conversion panels

Same criteria as in Table 1

Same criteria as in Table 1

Same criteria as in Table 1

Same criteria as in Table 1

Same criteria as in Table 1

Same criteria as in Table 1

Diagnostic specificity

Negative specimens

21 000 blood donations

1 000 blood donations

1 000 blood donations

1 000 blood donations

1 000 blood donations

≥ 99 %

(anti-HBc: ≥ 96 %)’

200 clinical specimens

200 clinical specimens

200 clinical specimens

200 clinical specimens

200 clinical specimens

200 samples from pregnant women

200 samples from pregnant women

200 samples from pregnant women

200 samples from pregnant women

100 potentially interfering samples

100 potentially interfering samples

100 potentially interfering samples

100 potentially interfering samples

100 potentially interfering samples

| 5. | Table 4 is replaced by the following:

  ‘Table 4


  Confirmatory and supplementary assays for anti-HIV 1/ 2, HIV 1/2 Ag/Ab, anti-HTLV I and II, anti-HCV, HCV Ag/Ab, HBsAg | | anti-HIV 1/2, HIV 1/2 Ag/Ab confirmatory assays | anti-HTLV I and II confirmatory assays | anti-HCV, HCV Ag/Ab supplementary assays | HBsAg confirmatory assays | Acceptance criteria |
| --- | --- | --- | --- | --- | --- | --- | --- |
| Diagnostic sensitivity | Positive specimens | 200 HIV-1 and 100 HIV-2
  Including samples from different stages of infection and reflecting different antibody patterns | 200 HTLV-I and 100 HTLV-II | 300 HCV (positive samples)
  Including samples from different stages of infection and reflecting different antibody patterns.
  Genotypes 1 – 4: > 20 samples (including non-a subtypes of genotype 4;
  Genotype 5: > 5 samples;
  Genotype 6: if available | 300 HBsAg
  Including samples from different stages of infection
  20 ‘high pos’ samples (>26 IU/ml); 20 samples in the cut-off range | Correct identification as positive (or indeterminate), not negative |
| Sero-conversion panels | 15 sero-conversion panels/low titre panels | | 15 sero-conversion panels/low titre panels | 15 sero-conversion panels/low titre panels | |
| Analytical sensitivity | Standards | | | | Third International Standard for HBsAg, subtypes ayw1/adw2, HBV genotype B4, NIBSC code: 12/226 | |
| Diagnostic specificity | Negative specimens | 200 blood donations | 200 blood donations | 200 blood donations | 10 false positives as available from the performance evaluation of the first line assay (1). | No false-positive results/ (1) no neutralisation |
| 200 clinical samples including pregnant women | 200 clinical samples including pregnant women | 200 clinical samples including pregnant women |
| 50 potentially interfering samples, including samples with indeterminate results in other confirmatory assays | 50 potentially interfering samples, including samples with indeterminate results in other confirmatory assays | 50 potentially interfering samples, including samples with indeterminate results in other supplementary assays | 50 potentially interfering samples |

5.

Table 4 is replaced by the following:

‘Table 4

Confirmatory and supplementary assays for anti-HIV 1/ 2, HIV 1/2 Ag/Ab, anti-HTLV I and II, anti-HCV, HCV Ag/Ab, HBsAg

anti-HIV 1/2, HIV 1/2 Ag/Ab confirmatory assays

anti-HTLV I and II confirmatory assays

anti-HCV, HCV Ag/Ab supplementary assays

HBsAg confirmatory assays

Acceptance criteria

Diagnostic sensitivity

Positive specimens

200 HIV-1 and 100 HIV-2

Including samples from different stages of infection and reflecting different antibody patterns

200 HTLV-I and 100 HTLV-II

300 HCV (positive samples)

Including samples from different stages of infection and reflecting different antibody patterns.

Genotypes 1 – 4: > 20 samples (including non-a subtypes of genotype 4;

Genotype 5: > 5 samples;

Genotype 6: if available

300 HBsAg

Including samples from different stages of infection

20 ‘high pos’ samples (>26 IU/ml); 20 samples in the cut-off range

Correct identification as positive (or indeterminate), not negative

Sero-conversion panels

15 sero-conversion panels/low titre panels

15 sero-conversion panels/low titre panels

15 sero-conversion panels/low titre panels

Analytical sensitivity

Standards

Third International Standard for HBsAg, subtypes ayw1/adw2, HBV genotype B4, NIBSC code: 12/226

Diagnostic specificity

Negative specimens

200 blood donations

200 blood donations

200 blood donations

10 false positives as available from the performance evaluation of the first line assay (1).

No false-positive results/ (1) no neutralisation

200 clinical samples including pregnant women

200 clinical samples including pregnant women

200 clinical samples including pregnant women

50 potentially interfering samples, including samples with indeterminate results in other confirmatory assays

50 potentially interfering samples, including samples with indeterminate results in other confirmatory assays

50 potentially interfering samples, including samples with indeterminate results in other supplementary assays

50 potentially interfering samples

---

(1)  Acceptance criteria: no neutralisation for HBsAg confirmatory assay.’

---