NIST TIME AND FREQUENCY BULLETIN 

NIST IR 6653-09 

No. 609 September 2008

 

1. GENERAL BACKGROUND INFORMATION ........................................................................................... 2 

 

2. TIME SCALE INFORMATION ............................................................................................................... 2 

 

3. PHASE DEVIATIONS FOR WWVB AND LORAN-C ............................................................................... 4 

 

4. BROADCAST OUTAGES OVER FIVE MINUTES AND WWVB 

 PHASE PERTURBATIONS ..................................................................................................................... 5 

5. NOTES ON NIST TIME SCALES AND PRIMARY STANDARDS ........................................................... 5 

6. BIBLIOGRAPHY ....................................................................................................................................... 5 

 

This bulletin is published monthly. Address correspondence to: 

 

Eyvon M. Petty, Editor 

Time and Frequency Division 

National Institute of Standards and Technology 

325 Broadway 

Boulder, CO 8O3O5-3328 

(3O3) 497-3295 

Email: pettye@boulder.nist.gov

U.S. DEPARTMENT OF COMMERCE, CARLOS M. GUTIERREZ, Secretary

NATIONAL INSTITUTE OF STANDARDS AND TECHNOLOGY, James M. Turner, Deputy Director 

 

 1 
2 

1. GENERAL BACKGROUND INFORMATION 

_______________________________________________________________________________________________________ 

ACRONYMS AND ABBREVIATIONS USED IN THIS BULLETIN

ACTS - Automated Computer Time Service 

BIPM - Bureau International des Poids et Mesures 

CS - Cesium Standard 

GPS - Global Positioning System 

IERS - International Earth Rotation Service 

LORAN - Long Range Navigation 

MC - Master Clock 

MJD - Modified Julian Date 

NIST - National Institute of Standards and Technology 

NOAA - National Oceanic and Atmospheric Administration 

NVLAP - National Voluntary Laboratory Accreditation Program ns - nanosecond 

SI - International System of Units μs - microsecond 

TA - Atomic Time ms - millisecond 

TAI - International Atomic Time s - second 

USNO - United States Naval Observatory min - minute 

UTC - Coordinated Universal Time 

_______________________________________________________________________________________________________ 

2. TIME SCALE INFORMATION 

 The values listed below are based on data from the IERS, the USNO, and NIST. The UTC(USNO,MC) - UTC(NIST) values are 

averaged measurements from all available common-view GPS satellites (see bibliography on page 5). UTC - UTC(NIST) data are 

on page 3. 

 0000 HOURS COORDINATED UNIVERSAL TIME 

AUG 2008 MJD UT1-UTC(NIST) 

 (±5 ms) 

UTC(USNO,MC) - UTC(NIST) 

 (±20 ns) 

7 54685 -457 ms 5 ns 

14 54692 -457 ms 5 ns 

21 54699 -460 ms 3 ns 

28 54706 -462 ms -2 ns 

 The master clock pulses used by the WWV, WWVH, and WWVB time-code transmissions are referenced to the UTC(NIST) time 

scale. Occasionally, 1 s is added to the UTC time scale. This second is called a leap second. Its purpose is to keep the UTC time 

scale within ±0.9 s of the UT1 astronomical time scale, which changes slightly due to variations in the Earth’s rotation. 

 NOTE: A positive leap second will be added at the end of December 2008. 

 Positive leap seconds, beginning at 23 h 59 min 60 s UTC and ending at 0 h 0 min 0 s UTC, were inserted in the UTC 

timescale on 30 June 1972, 1981-1983, 1985, 1992, 1993, 1994, and 1997, and on 31 December 1972-1979, 1987, 1989, 

1990,1995, 1998, and 2005. 

 The use of leap seconds ensures that UT1 - UTC will always be held within ±0.9 s. The current value of UT1 - UTC is called the 

DUT1 correction. DUT1 corrections are broadcast by WWV, WWVH, WWVB, and ACTS and are printed below. These corrections 

may be added to received UTC time signals in order to obtain UT1. 

 -0.5 s beginning 0000 UTC 07 August 2008 

 -0.4 s beginning 0000 UTC 13 March 2008 

DUT1 = UT1 - UTC = -0.3 s beginning 0000 UTC 29 November 2007 

 -0.2 s beginning 0000 UTC 14 June 2007 
3 

 

The difference between UTC(NIST) and UTC has been within ±100 ns since July 6, 1994. The table below shows values 

of UTC - UTC(NIST) as supplied by the BIPM in their Circular T publication for the most recent 310 day period in which 

data are available. Data are given at 10-day intervals. Five day interval data are available in Circular T. 

0000 Hours Coordinated Universal Time 

DATE MJD UTC-UTC(NIST) ns 

Jul. 22, 2008 54669 3.5 

Jul. 12, 2008 54659 4.1 

Jul. 02, 2008 54649 3.9 

Jun. 22, 2008 54639 1.1 

Jun. 12, 2008 54629 -2.0 

Jun. 02, 2008 54619 -4.6 

May 23, 2008 54609 -7.8 

May 13, 2008 54599 -8.4 

May 03, 2008 54589 -10.2 

Apr. 23, 2008 54579 -10.1 

Apr. 13, 2008 54569 -12.8 

Apr. 03, 2008 54559 -12.5 

Mar. 24, 2008 54549 -12.6 

Mar. 14, 2008 54539 -13.0 

Mar. 04, 2008 54529 -9.9 

Feb. 23, 2008 54519 -6.7 

Feb. 13, 2008 54509 -5.1 

Feb. 03, 2008 54499 -3.8 

Jan. 24, 2008 54489 -1.2 

Jan. 14, 2008 54479 -0.9 

Jan. 04, 2008 54469 1.3 

Dec. 25, 2007 54459 3.0 

Dec. 15, 2007 54449 5.2 

Dec. 05, 2007 54439 5.1 

Nov. 25, 2007 54429 7.1 

Nov. 15, 2007 54419 5.6 

Nov. 05, 2007 54409 5.8 

 
4 

3. PHASE DEVIATIONS FOR WWVB AND LORAN-C

WWVB - The values shown for WWVB are the time differences between the time markers of the UTC(NIST) time scale 

and the first positive-going zero voltage crossover measured at the transmitting antenna. The uncertainty of 

the individual measurements is ±0.5 μs. The values listed are for 1300 UTC. 

LORAN-C - The values shown for Loran-C represent the daily accumulated phase shift. The phase shift is measured by 

comparing the output of a Loran receiver to the UTC(NIST) time scale for a period of 24 h. If data were not 

recorded on a particular day, the symbol (-) is printed. The stations monitored are Baudette, Minnesota (8970) 

and Boise City, Oklahoma (9610). The monitoring is done from the NIST laboratories in Boulder, Colorado. 

Note: The values shown for Loran-C are in nanoseconds. 

UTC(NIST)-WWVB 

(60 kHz) UTC(NIST) - LORAN PHASE (ns)

ANTENNA PHASE 

LORAN-C 

(BAUDETTE) 

LORAN-C 

(BOISE CITY) 

DATE MJD (μs) (8970) (9610) 

08/01/2008 54679 5. 65 +25 +15 

08/02/2008 54680 5. 65 -15 -18 

08/03/2008 54681 5. 65 -43 -33 

08/04/2008 54682 5. 65 -12 +6 

08/05/2008 54683 5. 65 +88 +6 

08/06/2008 54684 5. 65 -104 +15 

08/07/2008 54685 5. 65 +3 -36 

08/08/2008 54686 5. 65 +30 +65 

08/09/2008 54687 5. 65 +41 -3 

08/10/2008 54688 5. 65 +25 +2 

08/11/2008 54689 5. 65 -10 +17 

08/12/2008 54690 5. 65 +113 -3 

08/13/2008 54691 5. 65 +25 +16 

08/14/2008 54692 5. 65 -67 -10 

08/15/2008 54693 5. 65 +53 +28 

08/16/2008 54694 5. 65 -18 -14 

08/17/2008 54695 5. 65 +21 +26 

08/18/2008 54696 5. 65 +16 +18 

08/19/2008 54697 5. 65 +21 +32 

08/20/2008 54698 5. 65 -103 -33 

08/21/2008 54699 5. 65 +62 +48 

08/22/2008 54700 5. 65 -133 -49 

08/23/2008 54701 5. 65 -76 -9 

08/24/2008 54702 5. 65 -76 +14 

08/25/2008 54703 5. 65 -6 -60 

08/26/2008 54704 5. 65 +12 +31 

08/27/2008 54705 5. 65 -14 -30 

08/28/2008 54706 5. 65 +30 +3 

08/29/2008 54707 5. 65 +13 +13 

08/30/2008 54708 5. 65 +180 -4 

08/31/2008 54709 5. 65 -56 -55 
 5 

 

4. BROADCAST OUTAGES OVER FIVE MINUTES AND WWVB PHASE PERTURBATIONS 

OUTAGES OF 5 MINUTES OR MORE PHASE PERTURBATIONS 

2 ms 

Station Aug 

2008 

 MJD Began 

UTC 

Ended 

UTC 

Freq. Aug 

2008 

MJD Began 

 UTC 

End 

UTC 

WWVB 08-06-08 54684 2304 2312 60 kHz

WWVB 08-25-08 54703 0946 1001 60 kHz

WWV 

WWVH 08-21-08 54699 1935 2003 15 MHz

5. NOTES ON NIST TIME SCALES AND PRIMARY STANDARDS 

 Primary frequency standards developed and operated by NIST are used to provide accuracy (rate) input to the BIPM. NIST-F1, a 

cold-atom cesium fountain frequency standard, has served as the U.S. primary time and frequency standard since 1999. The 

uncertainty of NIST-F1 is currently about 5 parts in 1016. 

 The AT1 scale is run in real-time by use of data from an ensemble of cesium standards and hydrogen masers. It is a free-running 

scale whose frequency is maintained as nearly constant as possible by choosing the optimum weight for each clock that contributes 

to the computation. 

 UTC(NIST) is generated as an offset from our real-time scale AT1. It is steered in frequency towards UTC by use of data published 

by the BIPM in its Circular T. Changes in the steering frequency will be made, if necessary, at 0000 UTC on the first day of the 

month, and occasionally at mid-month. A change in frequency is limited to no more than ±2 ns/day. The frequency of UTC(NIST) is 

kept as stable as possible at other times. 

 UTC is generated at the BIPM using a post-processed time-scale algorithm and is not available in real-time. The parameters that 

we use to generate UTC(NIST) in real-time are therefore based on an extrapolation of UTC from the most recent available data. 

 

6. BIBLIOGRAPHY 

Allan, D.W.; Hellwig, H.; and Glaze, D.J., "An accuracy algorithm for an atomic time scale," Metrologia, Vol.11, No.3, pp.133-138 

(1975). 

Allan, D.W.; Davis, D.D.; Weiss, M.A.; Clements, A.; Guinot, B.; Granveaud, M.; Dorenwendt, K.; Fischer, B.; Hetzel, P.; Aoki, S.; 

Fujimoto, M.; Charron, L.; and Ashby, N., “Accuracy of international time and frequency comparisons via global positioning system 

satellites in common-view,” IEEE Transactions on Instrumentation and Measurement, Vol. IM-34, pp.118-125 (1985). 

Heavner, T.P.; Jefferts, S.R.; Donley; E.A.; Shirley, J.H. and Parker, T.E., “NIST F1; recent improvements and accuracy evaluations,” 

Metrologia, Vol. 42, pp. 411-422 (2005). 

Jefferts, S.R.; Shirley, J.; Parker, T.E.; Heavner, T.P.; Meekhof, D.M.; Nelson, C., Levi, F.; Costanza, G.; De Marchi, A.; Drullinger, 

R.; Hollberg, L.; Lee, W.D.; and Walls, F.L., “Accuracy evaluation of NIST-F1,” Metrologia, Vol. 39, pp. 321-336 (2002). 

Lewandowski, W. and Thomas, C.; “GPS Time transfer,” Proceedings of the IEEE, Vol. 79, pp. 991-1000 (1991). 

Parker, T.E.; Jefferts, S.R.; Heavner, T.P.; and Donley, E.A., “Operation of the NIST-F1 caesium fountain primary frequency standard 

with a maser ensemble, including the impact of frequency transfer noise,” Metrologia, Vol. 42, pp. 423-430 (2005). 

Weiss, M.A.; Allan, D.W.; “An NBS Calibration Procedure for Providing Time and Frequency at a Remote Site by Weighting and 

Smoothing of GPS Common View Data,” IEEE Transactions on Instrumentation and Measurement, Vol. IM-36, pp. 572-578 (1987). 
 6 

Table 7.1 lists parameters that are used to define UTC(NIST) with respect to our real-time scale AT1. To find the value of 

UTC(NIST) - AT1 at any time T (expressed as a Modified Julian Day, including a fraction if needed), the appropriate 

equation to use is the one for which the desired T is greater than or equal to the entry in the T0 column and less than the 

entry in the last column. The values of xls, x, and y for that month are then used in the equation below to find the desired 

value. The parameters x and y represent the offset in time and in frequency, respectively, between UTC(NIST) and AT1; 

the parameter xls is the number of leap seconds applied to both UTC(NIST) and UTC as specified by the IERS. Leap 

seconds are not applied to AT1. 

Table 7.1 

UTC(NIST) - AT1 = xls + x + y*(T −T0) 

Month xls

(s) 

x 

(ns) 

y 

(ns/d) 

 T0

(MJD) 

Valid until 0000 on: 

(MJD) 

 Oct 08 -33 -325894.6 -38.3* 54740 54771

 Sep 08 -33 -324745.6 -38.3 54710 54740*

 Aug 08 -33 -323558.3 -38.3 54679 54710

 Jul 08 -33 -322792.3 -38.3 54659 54679

 Jul 08 -33 -322369.9 -38.4 54648 54659†

 Jun 08 -33 -321211.9 -38.6 54618 54648

 May 08 -33 -320594.3 -38.6 54602 54618

 May 08 -33 -320018.3 -38.4 54587 54602†

 Apr 08 -33 -319288.7 -38.4 54568 54587

 Apr 08 -33 -318867.4 -38.3 54557 54568†

 Mar 08 -33 -318178.0 -38.3 54539 54557

 Mar 08 -33 -317684.0 -38.0 54526 54539†

 Feb 08 -33 -316582.0 -38.0 54497 54526

 Jan 08 -33 -315974.0 -38.0 54481 54497

 Jan 08 -33 -315405.5 -37.9 54466 54481†

 Dec 07 -33 -314230.6 -37.9 54424 54466

 Nov 07 -33 -313813.7 -37.9 54435 54424

 Nov 07 -33 -313091.7 -38.0 54405 54435†

 Oct 07 -33 -312635.7 -38.0 54393 54405

 Oct 07 -33 -311911.8 -38.1 54374 54393†

 Sep 07 -33 -310768.8 -38.1 54344 54374

 Aug 07 -33 -309587.7 -38.1 54313 54344

† Rate change in mid-month 

†† Rate change one day early 

*Provisional value