Index | Site       Map | FAQ | Facility       Info | Reading       Rm | New | Help | Glossary | Contact UsProtecting People and the EnvironmentUNITED STATES NUCLEAR REGULATORY COMMISSION -   About NRC    - The     Commission 
  - Statutory Authority 
  - Strategic Plan 
  - Budget &      Performance 
  - Organization & Functions 
  - Locations 
  - History 
  - Values 
  - Policymaking 
  - Radiation Protection 
  - Fire Protection 
  - How We Regulate 
  - Grants 
  - Civil Rights 
  - 
- NuclearReactors    - Operating Reactor Locations 
  - Reactor Quick Links 
  - Power Reactors 
  - Research and Test     Reactors 
  - Operating Reactors 
  - Operator Licensing 
  - New Reactors 
  - Operator     Licensing for New Reactors 
  - Reactor Safety Research 
  - Advisory Comm (ACRS) 
  - Decommissioning     of Nuclear Facilities 
  - 
- NuclearMaterials    - Fuel Cycle Facility Locations 
  - Uranium     Recovery Locations 
  - Materials Quick Links 
  - Special Nuclear Material 
  - Source Material 
  - Byproduct Material 
  - Med, Academic & Ind Uses 
  - Source Materials Facilities 
  - Uranium Recovery 
  - Fuel Cycle Facilities 
  - Materials     Transportation 
  - Materials Research 
  - Advisory Comm (ACMUI) 
  - Decommissioning     of Nuclear Facilities 
  - Materials     Environmental Reviews 
  - 
- RadioactiveWaste    - Waste Quick Links 
  - Low-Level Waste 
  - Waste Incidental to     Reprocessing 
  - Incidental Waste Criteria 
  - High-Level Waste 
  - Uranium Mill Tailings 
  - LLW Disposal 
  - HLW Disposal 
  - Spent Fuel Storage 
  - Spent Fuel Transport 
  - Waste Research 
  - 
- NuclearSecurity    - Security     Spotlight 
  - NRC     Responsibilities 
  - Domestic Safeguards 
  - Radioactive Material Security 
  - Facility Protection 
  - Nuclear Material Control 
  - International Safeguards 
  - Information Security 
  - Emergency     Preparedness 
  - &; 
- Public Meetings&   Involvement    - Public Meeting Schedule 
  - Commission     Schedule 
  - Conferences & Symposia 
  - Documents for Comment 
  - Brochures & Fact Sheets 
  - Involving Stakeholders 
  - Info Quality Guidelines 
  - Rulemaking Process 
  - Licensing 
  - Enforcement 
  - Hearing Process 
  - Hearing Applications 
  - 

Home       > Electronic Reading       Room       > Document       Collections       > General       Communications       > Information       Notices       > 1999       > IN 99-23

UNITED STATESNUCLEAR REGULATORY COMMISSIONOFFICE OF NUCLEAR MATERIAL SAFETY AND SAFEGUARDSWASHINGTON, D.C. 20555-0001

July 6, 1999
NRC INFORMATION NOTICE 99-23:      SAFETY CONCERNS RELATED TO REPEATED CONTROL UNIT FAILURES OF THE       NUCLETRON CLASSIC MODEL HIGH-DOSE-RATE REMOTE AFTERLOADING BRACHYTHERAPY       DEVICES- Addressees: 
- Purpose: 
- Description   of Circumstances: 
- Discussion: 
Addressees:
All U.S. Nuclear Regulatory Commission (NRC) medical licensees authorized to use brachytherapy sources in Nucletron Classic Model high-dose-rate (HDR) remote afterloaders.
Purpose:
NRC is issuing this information notice to alert you to ongoing control unit failures with Nucletron Classic Model HDR devices. You should review this information for applicability to your facilities and consider actions, as appropriate, to minimize the impact of such failures on patient treatments and personnel exposures. However, suggestions contained in this information notice are not NRC requirements; therefore, no specific action nor written response is required.
Description of Circumstances:
By January 1996, NRC was aware of three incidents where the Nucletron Classic Model HDR control units failed (locked up) for unexplained reasons. These control unit failures exposed a deficiency in the design of the door interlock circuitry for the Nucletron Classic Model HDR. Nucletron issued a Safety Alert on March 4, 1996, describing this problem and providing information on the proper operator response to such failures. This safety alert was also issued as an attachment to NRC Information Notice 96-21, issued on April 10, 1996, that discussed both the door interlock problem and the control unit failures. Following these three reported failures of the control unit, Nucletron Corporation made field modifications to these devices to correct the door interlock failure mechanism and to eliminate the unexplained control unit failures.

Following these modifications, two similar failures were reported in 1997, four in 1998, and three to date in 1999. These additional nine control unit failures, reported since the 1996 corrective action, indicate a ongoing problem with unexplained control unit failures. However, no further problems have been reported with the operation of the door interlocks. Nucletron has continued to investigate the ongoing control unit failures and now believes it has found the root cause of the failures and subsequently developed appropriate corrective measures. We understand that the newly developed corrective measures are presently undergoing testing before implementation.
Discussion:
In all 12 reported control unit failures no patient or personnel overexposures were reported. In each case the treatment unit continued to operate, as designed, when communications with the control unit was lost. In this event, the microprocessor in the treatment unit continues the preprogrammed treatment to the catheter being treated when the failure occurs, halting upon completion. If the failure occurs during a single catheter treatment or during the last catheter in a multi-catheter operation, then the treatment will proceed to completion. Otherwise, it will terminate with completion of the catheter being treated, leaving any remaining catheters untreated. Most users, however, when faced with a blank or frozen display on an inoperative control unit, immediately abort the treatment, using the "Emergency Stop" button on the control unit. The "Treatment Interrupt" button (called for in the 1st step on the Safety Alert procedures) on the control unit is rendered inoperative by the control unit failure. Irrespective of the method of terminating the patient's treatment, the actual treatment data can subsequently be read from the treatment unit, using a handheld terminal device.

Most patients receive less than the prescribed dose in the event of a control unit failure and, at the discretion of the Authorized User, either the Written Directive must be revised to reflect the actual dose given or the remaining portion of the treatment given after repair or replacement of the defective control unit. To date, none of the reported control unit failures has resulted in overexposures or misadministrations. However, the failure of the control unit removes the first layer of radiation safety protection. At this point, the avoidance of excessive radiation exposures depends upon either the continued proper operation of the treatment unit or operator intervention.

NRC previously issued Information Notice 96-21 on April 10, 1996, and included a copy of the Nucletron issued Safety Alert, addressing the control unit failures, as an attachment. This Safety Alert sets forth a series of four steps to be taken if, for any reason, the Control Unit stops updating the status of the treatment in progress. These steps should ensure that the treatment is interrupted and the source retracted before licensee personnel enter the treatment room. You are encouraged to follow these recommendations from the device vendor, as contained in its Safety Alert of March 4, 1996. Any questions you may have related to these control unit failures and related corrective actions, or to get a copy of the Safety Alert, should be addressed to your Nucletron Corporation representative.

This information notice requires no specific action nor written response. If you have any questions about the information in this notice, please contact the technical contact listed below or the appropriate regional office.
/s/'d by       
Donald A. Cool, DirectorDivision of Industrial and Medical Nuclear       SafetyOffice of Nuclear Material Safety and   Safeguards
Contact:Robert L. Ayres, NMSS (301) 415-5746 E-mail: rxa1@nrc.gov      Attachments:1. List of Recently Issued NMSS Information Notices 2. List of       Recently Issued NRC Information Notices(NUDOCS Accession Number 9907010001)