Index | Site       Map | FAQ | Facility       Info | Reading       Rm | New | Help | Glossary | Contact UsProtecting People and the EnvironmentUNITED STATES NUCLEAR REGULATORY COMMISSION -   About NRC    - The     Commission 
  - Statutory Authority 
  - Strategic Plan 
  - Budget &      Performance 
  - Organization & Functions 
  - Locations 
  - History 
  - Values 
  - Policymaking 
  - Radiation Protection 
  - Fire Protection 
  - How We Regulate 
  - Grants 
  - Civil Rights 
  - 
- NuclearReactors    - Operating Reactor Locations 
  - Reactor Quick Links 
  - Power Reactors 
  - Research and Test     Reactors 
  - Operating Reactors 
  - Operator Licensing 
  - New Reactors 
  - Operator     Licensing for New Reactors 
  - Reactor Safety Research 
  - Advisory Comm (ACRS) 
  - Decommissioning     of Nuclear Facilities 
  - 
- NuclearMaterials    - Fuel Cycle Facility Locations 
  - Uranium     Recovery Locations 
  - Materials Quick Links 
  - Special Nuclear Material 
  - Source Material 
  - Byproduct Material 
  - Med, Academic & Ind Uses 
  - Source Materials Facilities 
  - Uranium Recovery 
  - Fuel Cycle Facilities 
  - Materials     Transportation 
  - Materials Research 
  - Advisory Comm (ACMUI) 
  - Decommissioning     of Nuclear Facilities 
  - Materials     Environmental Reviews 
  - 
- RadioactiveWaste    - Waste Quick Links 
  - Low-Level Waste 
  - Waste Incidental to     Reprocessing 
  - Incidental Waste Criteria 
  - High-Level Waste 
  - Uranium Mill Tailings 
  - LLW Disposal 
  - HLW Disposal 
  - Spent Fuel Storage 
  - Spent Fuel Transport 
  - Waste Research 
  - 
- NuclearSecurity    - Security     Spotlight 
  - NRC     Responsibilities 
  - Domestic Safeguards 
  - Radioactive Material Security 
  - Facility Protection 
  - Nuclear Material Control 
  - International Safeguards 
  - Information Security 
  - Emergency     Preparedness 
  - &; 
- Public Meetings&   Involvement    - Public Meeting Schedule 
  - Commission     Schedule 
  - Conferences & Symposia 
  - Documents for Comment 
  - Brochures & Fact Sheets 
  - Involving Stakeholders 
  - Info Quality Guidelines 
  - Rulemaking Process 
  - Licensing 
  - Enforcement 
  - Hearing Process 
  - Hearing Applications 
  - 

Home       > Electronic Reading       Room       > Document       Collections       > General       Communications       > Information       Notices       > 1999       > IN 99-34

UNITED STATESNUCLEAR REGULATORY COMMISSIONOFFICE OF NUCLEAR REACTOR REGULATIONWASHINGTON, D.C. 20555-0001

December 28, 1999
NRC INFORMATION NOTICE 99-34:POTENTIAL FIRE HAZARD IN THE USE OF POLYALPHAOLEFIN IN       TESTING OF AIR FILTERS- Addressees 
- Purpose 
- Description   of Circumstances 
Addressees
All holders of licenses for nuclear reactors and fuel cycle facilities.
Purpose
The U.S. Nuclear Regulatory Commission (NRC) is issuing this information notice to alert addressees to a recent event in which a flame was emitted from a thermal aerosol generator being used for in-place testing of a high-efficiency particulate air (HEPA) filter. The aerosol was generated from a synthetic aliphatic hydrocarbon, polyalphaolefin (PAO). The flame did not result in personnel injury, but it had the potential to create serious consequences. It is expected that recipients will review the information for applicability to their facilities and consider actions, as appropriate. However, suggestions contained in this information notice are not NRC requirements; therefore, no specific action or written response to this notice is required.
Description of Circumstances
On April 28, 1999, at the Department of Energy's West Valley Demonstration Project (WVDP), operations personnel were conducting in-place testing of a high-efficiency particulate air (HEPA) filter on a main filter bank using a thermal aerosol generator (NUCON F-1000-DG Model F) to produce a test aerosol of polyalphaolefin (PAO).

In accordance with West Valley's established procedure, the generator heater temperature had stabilized at 720 °F. The liquid flow valve was placed in the "on" position and adjusted to produce a predetermined flow rate. Then the vapor adjust control valve (carrier air) was opened slowly until a steady supply of aerosol was observed. As the vapor adjust control valve was slowly opened, a 2-to 3-foot-long flame was emitted from the generator discharge port. The attending operator was able to extinguish the flame by immediately closing both the vapor control valve and the liquid flow valve.

The thermal aerosol generator had earlier been used to generate dioctyl phthalate (DOP) test aerosol. Such an aerosol is produced in the generator at a heater block temperature of approximately 720°F; the auto-ignition temperature of DOP is 735°F. The heater block temperature is not adjustable, but the temperature regulator can be replaced with one that will maintain the heater block temperature at about 625°F, which is adequate for producing the PAO aerosol while maintaining a margin below the PAO auto-ignition temperature of 650°F. WVDP had changed the test aerosol to PAO but had not modified the heater controls to produce the lower temperature applicable to PAO.

The operators actuated the two control valves in an order opposite to the order presented in the vendor's operating manual. The vendor's manual recommends that air flow ("carrier air") be initiated before establishing liquid flow.

Following the event, West Valley suspended all HEPA filter testing on site until corrective actions were completed. The vendor of the thermal aerosol generator simulated the event under the same flow conditions and valve manipulations. The vendor observed flames at the aerosol discharge port in approximately 75 percent of the tests.

West Valley has modified its generator heater block controls and its valve operating procedures appropriately and, in conjunction with the vendor, has modified the carrier air valve so that it remains open a small amount even when in the closed position.

This information notice requires no specific action or written response. If you have any questions about the information in this notice, please contact one of the technical contacts listed below, the appropriate regional office, or the appropriate Project Manager of the Office of Nuclear Reactor Regulation (NRR) or of the Office of Nuclear Material Safety and Safeguards (NMSS).
Original signed byOriginal signed byMichael F. Weber, Director Division of Fuel Cycle Safety and       SafeguardsOffice of Nuclear Material Safety and SafeguardsLedyard B. Marsh, Chief Events Assessment, Generic Communications             and Safeguards and Non-Power       Reactors BranchDivision of Regulatory Improvement ProgramsOffice       of Nuclear Reactor RegulationTechnical Contacts:Todd J. Jackson, Region I 610-337-5308 E-mail:       tjj@nrc.govJohn P. Segala, NRR 301-415-1858 E-mail:     jps1@nrc.govCandice M. Drummond, NMSS 301-415-6433 E-mail: cmd@nrc.govAttachments:1. List of Recently Issued NMSS Information Notices 2.       List of Recently Issued NRC Information Notices(ADAMS Accession Number ML993550113)