# FLIR QUARK 2

Longwave Infrared Thermal Camera

Quark 2 provides leading-edge imaging performance and reliability in an affordable, compact, and lightweight package that offers the best in “SWaP-C” value. Quark 2 is available in resolutions of 336 and 640, both with 17-micron pixels. With high shock and vibration tolerance, Quark 2 is designed for years of maintenance-free operation. And now Quark 2 offers a host of new features, including support options for 60Hz frame rates, accurate radiometric capabilities, and powerful image processing modes that can be set manually.



## IMPROVED IMAGE PROCESSING For clearer imagery, edge sharpening, and contrast •	Second generation Digital Detail EnhancementTM (DDE)

•	Active Contrast EnhancementTM (ACE)

•	Smart Scene OptimizationTM (SSO)

•	Information Based HEQTM (IBHEQ) automatically adjusts AGC

•	Silent Shutterless NUCTM for continuous image improvement



## ACCURATE TEMPERATURE MEASUREMENT Supports radiometry, analytics and telemetry •	TLinear output places temperature data in each pixel •	Adjustable isotherm thresholds colorize temperatures of interest •	Rugged and reliable in all terrain



## COMMON FEATURES ACROSS MODELS Fosters improved OEM integration •	22mm x 22mm x 12mm (w/o lens)

•	640 and 336 resolutions •	Weight: 18.3g

- 23.0g (depending on lens) •	Multiple lens and FOV options
•	9Hz, 30Hz and 60Hz frame rates available

•	Mechanical / electrical compatibility across all models •	Rugged and Reliable New Isotherm capabilities

Edge sharpening with 2nd Gen. DDE



### www.flir.com/Quark2

## Imaging Specifications

www.flir.com NASDAQ: FLIR SANTA BARBARA FLIR Systems, Inc. 70 Castilian Drive Goleta, CA 93117 USA PH: +1 805.690.5097 PORTLAND Corporate Headquarters FLIR Systems, Inc. 27700 SW Parkway Ave. Wilsonville, OR 97070 USA PH: +1 866.477.3687

EUROPE FLIR Systems Luxemburgstraat 2 2321 Meer Belgium PH : +32 (0) 3665 5100 CHINA-SHANGHAI FLIR Systems Co.,Ltd. K301-302, No 26 Lane 168, Daduhe Road, Putuo District, Shanghai 200062, P.R.China PH: +86-21-5169 7628

Equipment described herein may require US Government authorization for export purposes. Diversion contrary to US law is prohibited. Imagery for illustration purposes only. Specifications are subject to change without notice. ©2014 FLIR Systems, Inc. All rights reserved. (Updated 11/21/14) * When registered within 60 days of purchase.



### www.flir.com/Quark2

System Overview System Type Uncooled LWIR Thermal Imager Quark 640: 640 × 512 VOx Microbolometer Quark 336: 336 × 256 VOx Microbolometer Pixel Size 17 μm Spectral Band 7.5 - 13.5 μm Performance <50 mK @ f/1.0 Outputs Analog Video Field-switchable between NTSC and PAL Quark 640: 30 Hz (NTSC); 25 Hz (PAL); <9Hz option for export Quark 336: 30/60 Hz (NTSC); 25/50 Hz (PAL) ; <9Hz export option Digital Video 8- or 14-bit serial LVDS; 8- or 14-bit parallel CMOS; 8-bit BT.656 Operation & Control

Image Control Invert, revert, 2× & 4× digital zoom, polarity, false color or monochrome, AGC, digital detail enhancement (DDE) Camera Control Autonomous; Manual via GUI or serial command Signal Interface 60-pin SAMTEC connector: power, comm., video, digital data, external sync, discrete commands Accessories Video, Power & Communication (VPC) expansion board Physical Attributes Size / Weight 22 × 22 x 12 mm (less lens) / 8 g (camera body only) Mounting Interface 4 M1.6 x 0.35 on rear of camera frame Power Input Voltage 3.3 +/- 0.1 VDC Power Dissipation <1.0 W (Quark 336); <1.2 W (Quark 640) Time to Image <4 seconds (Quark 336); <5 seconds (Quark 640) Environmental Operating Temperature Range -40° C to +80° C external temp Storage Temperature Range -55° C to +105° C external temp Scene Temp Range To 150° C standard Shock / Temperature Shock 500 g; 0.8 msec shock pulse (all axes)/5/min Vibration 4.3 g 3 axes, 8 hours each Humidity 5 - 95% non-condensing Operational Altitude +40,000 feet ROHS, REACH, and WEEE Compliant 

## Applications:

Unmanned Vehicles Handheld Imagers Security Cameras Maritime Cameras Military-grade Goggles