NIST HANDBOOK 150-11 

2007 Edition 

National 

Voluntary

Laboratory 

Accreditation 

Program 

ELECTROMAGNETIC

COMPATIBILITY AND

TELECOMMUNICATIONS

PROGRAM

Kurt B. Fischer 

Carroll S. Brickenkamp 

National Voluntary Laboratory Accreditation Program 

Standards Services Division 

Technology Services 

September 2007 

U.S. Department of Commerce 

Carlos M. Gutierrez, Secretary 

National Institute of Standards and Technology 

James M. Turner, Acting Director 
NVLAP AND THE NVLAP LOGO 

The term NVLAP and the NVLAP logo are registered 

marks of the Federal Government, which retains 

exclusive rights to control the use thereof. Permission 

to use the term and symbol (NVLAP logo with 

approved caption) is granted to NVLAP-accredited 

laboratories for the limited purpose of announcing 

their accredited status, and for use on reports that 

describe only testing and calibration within the scope 

of accreditation. NVLAP reserves the right to control 

the quality of the use of the NVLAP term, logo, and 

symbol.
Contents 

Foreword.......................................................................................................................................................v

Introduction..................................................................................................................................................vi

1 General information .............................................................................................................................1

1.1 Scope of handbook ......................................................................................................................1

1.2 Organization of handbook ...........................................................................................................1

1.3 Program description.....................................................................................................................1

1.4 References ...................................................................................................................................2

1.5 Terms and definitions ..................................................................................................................2

1.6 Program documentation...............................................................................................................3

2 LAP establishment, development and implementation........................................................................4

3 Accreditation process ...........................................................................................................................4

3.1 General ........................................................................................................................................4

3.2 Management system review ........................................................................................................4

3.3 On-site assessment.......................................................................................................................5

4 Management requirements for accreditation........................................................................................9

4.1 Organization ................................................................................................................................9

4.2 Management system ....................................................................................................................9

4.3 Document control ......................................................................................................................10

4.4 Review of requests, tenders and contracts.................................................................................10

4.5 Subcontracting of tests and calibrations ....................................................................................11

4.6 Purchasing services and supplies...............................................................................................11

4.7 Service to the customer..............................................................................................................11

4.8 Complaints.................................................................................................................................11

4.9 Control of nonconforming testing and/or calibration work.......................................................11

4.10 Improvement..............................................................................................................................11

4.11Corrective action .......................................................................................................................12

4.12 Preventive action .......................................................................................................................12

4.13Control of records......................................................................................................................12

4.14 Internal audits ............................................................................................................................12

4.15 Management reviews.................................................................................................................12

5 Technical requirements for accreditation...........................................................................................13

5.1 General ......................................................................................................................................13

5.2 Personnel ...................................................................................................................................13

5.3 Accommodation and environmental conditions ........................................................................13

5.4 Test and calibration methods and method validation ................................................................13

5.5 Equipment..................................................................................................................................14

5.6 Measurement traceability ..........................................................................................................15

5.7 Sampling....................................................................................................................................15

5.8 Handling of test and calibration items.......................................................................................15

5.9 Assuring the quality of test and calibration results....................................................................15

5.10Reporting the results..................................................................................................................16

NIST Handbook 150-11:2007 iii 
6 Additional requirements.....................................................................................................................16

Annex A (informative) 

 Information about selected ECT standards ........................................................................................17

iv NIST Handbook 150-11:2007 
Foreword 

The NIST Handbook 150 publication series sets forth the procedures, requirements, and guidance for the 

accreditation of testing and calibration laboratories by the National Voluntary Laboratory Accreditation 

Program (NVLAP). The series is comprised of the following publications: 

• NIST Handbook 150, NVLAP Procedures and General Requirements, which contains the general 

procedures and requirements under which NVLAP operates as an unbiased third-party accreditation 

body; 

• NIST Handbook 150-xx program-specific handbooks, which supplement NIST Handbook 150 by 

providing additional requirements, guidance, and interpretive information applicable to specific 

NVLAP laboratory accreditation programs (LAPs). 

The program-specific handbooks are not stand-alone documents, but rather are companion documents to 

NIST Handbook 150. They tailor the general criteria found in NIST Handbook 150 to the specific tests, 

calibrations, or types of tests or calibrations covered by a LAP. 

NIST Handbook 150-11, NVLAP Electromagnetic Compatibility and Telecommunications, presents the 

technical requirements and guidance for the accreditation of laboratories under the NVLAP 

Electromagnetic Compatibility and Telecommunications (ECT) LAP. The 2007 edition incorporates 

changes resulting from the release of the newest editions of ISO/IEC 17025, General requirements for the 

competence of testing and calibration laboratories, and NIST Handbook 150, as well as editorial 

improvements. The 2007 edition of NIST Handbook 150-11 supersedes and replaces the 1995 edition, 

Electromagnetic Compatibility and Telecommunications FCC Methods, and the 1994 draft of the program 

handbook, Electromagnetic Compatibility MIL-STD-462. 

The handbook was revised with the participation of technical experts in the field of electromagnetic 

compatibility and telecommunications testing and was approved by NVLAP. The following main 

changes have been made to this handbook with respect to the previous edition: 

• all references to applicable international guides and standards have been updated; 

• on-site assessment checklists and the test method selection list are no longer included in order that 

they may be provided as separate documents, which may be updated at different intervals than the 

handbook; 

• the body of the handbook has been restructured to conform with internationally accepted rules for the 

structure and drafting of standards, where appropriate, to promote ease of use and understanding. 

Annex A (informative) provides a list of major ECT standards-issuing bodies, the acronyms most 

commonly cited, and the national economies for which the standards have been issued. 

This handbook is also available on the NVLAP web site (http://www.nist.gov/nvlap). 

Questions or comments concerning this handbook should be submitted to NVLAP, National Institute of 

Standards and Technology, 100 Bureau Drive, Stop 2140, Gaithersburg, MD, 20899-2140; phone: 

301-975-4016; fax: 301-926-2884; e-mail: nvlap@nist.gov. 

NIST Handbook 150-11:2007 v 
Introduction 

General 

All electrical devices are potential emitters of electromagnetic radiation and are potentially affected by 

electromagnetic radiation emitted from other electrical devices in their vicinity. These emissions may 

interfere with the performance and safe operation of a device. In fact, an electrical device is classified as 

either a non-intentional radiator, generating emissions as a by-product of normal operation (for example, a 

television or personal computer), or as an intentional radiator (for example, a citizens band radio or cell 

phone). 

Electromagnetic Compatibility (EMC) is the ability of a device, product, or system to operate properly in 

its intended electromagnetic environment without degradation and without being a source of 

electromagnetic interference (EMI). As more sophisticated and sensitive electronic devices enter the 

marketplace, electromagnetic compatibility becomes more and more important. 

In order to achieve EMC, governments set limits, and device manufacturers, including device purchasers, 

set requirements for the design, production and operation of electronic systems that are 

electromagnetically compatible with their environments. 

Types of requirements 

In the United States, most electrical devices are required to meet specific limits for electromagnetic 

emissions, but not immunity to emissions. The U.S. Federal Communications Commission (FCC) sets 

requirements for commercial and consumer electronic devices and systems. Many nations set 

requirements for a device’s immunity to electromagnetic emissions as well as limits for device emissions. 

The most cited standards outside the United States include International Electrotechnical Commission 

(IEC) and Comite International Special des Perturbations Radioelectriques (International Special 

Committee on Radio Interference) (CISPR) standards. 

With the global nature of trade, devices manufactured in one place must meet the EMC requirements of 

multiple locations in which they are sold. Governments, buyers and manufacturers often cite international 

EMC voluntary standards. Even so, NVLAP currently tracks and accredits to more than 900 separate test 

method requirements in the ECT LAP. Part of the reason for the large number of standards in this field is 

because of the many ways to categorize electrical and electronic devices for EMC. 

Some categories have evolved from older application categories, including telephony and radio 

communications. As early as 1977, the International Organization for Standardization (ISO) began to 

develop its Open Systems Interconnection Basic Reference Model as an abstract description for 

communications and computer network protocol design, including physical standards, protocol standards, 

and interoperability standards. Also included are “harms to the network” test methods for wired 

telecommunications. Network Equipment Building System (NEBS) standards apply to the broad array of 

devices intended for the central office (CO) environment and procurements by Local Exchange Carriers 

(LECs), Competitive Access Providers (CAPs), Competitive Local Exchange Carriers (CLECs), Internet 

Service Providers (ISPs), and Access Service Providers (ASPs). 

Some EMC requirements are based on the intended usage environment of the product, often more 

extreme than normal business and other public applications. For example, U.S. Military Standards (MILvi NIST Handbook 150-11:2007 
STD) 461/462 impose requirements for devices used in various ground, flight, and naval environments. 

Radio Technical Commission for Aeronautics, now RTCA, Inc., (RTCA) DO-160 test methods apply to 

devices dedicated to aeronautical environments. 

Some standards that are not directly EMC standards are closely related, such as environmental and 

product safety, including human exposure limits (maximum permissible exposure [MPE] and specific 

absorption rate [SAR]), are also included in the ECT LAP. 

History of the NVLAP ECT Program 

The NVLAP ECT Program for FCC test methods was established in October 1985 in response to a 

request from five private-sector testing laboratories. The purpose of the program was to formally 

recognize laboratories found competent to perform testing in accordance with Title 47 of the U.S. Code of 

Federal Regulations (CFR) Part 15-Radio Frequency Devices and 47 CFR Part 68-Connection of 

Terminal Equipment to the Telephone Network. The program was expanded in 1988 in response to a 

request from the Naval Air Systems Command (NAVAIR) for the establishment and maintenance of 

adequate technical resources for MIL-STD-462 Acceptance Testing. The purpose of that part of the 

program is to assess and accredit laboratories that produce reliable test data for the U.S. military. 

Present status of the NVLAP ECT Program 

At the time of publication, this handbook covers test methods used to demonstrate compliance with FCC 

requirements given in 47 CFR, Telecommunication, Parts 0 through 101, the test methods in MIL-STD 

461/462, the ANSI C63 standards, and the international standards, IEC 61000 and CISPR 22. 

Other test methods used to demonstrate compliance with specific national standards for electromagnetic 

compatibility are also covered. These standards include Australia and New Zealand Standard (AS/NZS) 

3548, Australian Communications Authority (ACA) Technical Specifications (TSs), Chinese National 

Standard (CNS) 13438, and Canadian Compliance Specification (CS) 03. (CS 03 is the Canadian 

regulatory requirements to FCC part 68.) In addition, by virtue of a memorandum of understanding 

between NVLAP and the Voluntary Control Council for Interference (VCCI) of Japan, NVLAP provides 

ISO/IEC 17025 accreditation of any electromagnetic compatibility-testing laboratory to the Normative 

Annex 1 Technical Requirements of Regulations (VCCI V-3) for voluntary control measures of VCCI. 

Because of the constant updates and modifications to international, national, and regional requirements, 

users of this handbook should check frequently with the issuing bodies for changes and additions to 

requirements and regulations. Please check the NVLAP web site or contact the ECT Program Manager on 

questions about specific standards for which accreditation is available or may be made available. 

Informative Annex A provides the names and acronyms of the most common EMC standards 

organizations and, where applicable, the national economies for which the standards have been issued. 

NIST Handbook 150-11:2007 vii 
This page intentionally left blank
1 General information 

1.1 Scope of handbook 

1.1.1 NIST Handbook 150-11 specifies the technical requirements and provides guidance for the 

accreditation of laboratories under the NVLAP Electromagnetic Compatibility and Telecommunications 

Laboratory Accreditation Program (ECT Program). It supplements the NVLAP procedures and general 

requirements found in NIST Handbook 150, by tailoring the general criteria found in NIST Handbook 

150 to the specific tests and types of tests covered by the ECT Program. 

1.1.2 NIST Handbook 150, this handbook, and their respective checklists (see 1.6) constitute the 

collective body of requirements that must be met by a laboratory seeking NVLAP accreditation for the 

ECT Program. 

1.1.3 This handbook is intended for information and use by accredited ECT laboratories, assessors 

conducting on-site assessments, laboratories seeking accreditation, other laboratory accreditation systems, 

users of laboratory services, and others needing information on the requirements for accreditation under 

the ECT Program. 

1.2 Organization of handbook 

The numbering and titles of the first five clauses of this handbook match those of NIST Handbook 150. 

The primary subclauses in clauses 4 and 5 (e.g., 4.1, 4.2, etc.) are also numbered and titled to correspond 

with those of NIST Handbook 150, even when there are no requirements additional to those in NIST 

Handbook 150. 

1.3 Program description 

The purpose of the ECT LAP is to accredit testing laboratories found capable and competent to perform 

EMC conformance testing to FCC, MIL-STD, IEC, CISPR and other test method standards that have 

been and may be added to the program. 

The program includes both intentional radiators (i.e., radio transmitters) and unintentional radiators (i.e., 

digital devices), as well as wireless and wired telecommunications products. The program also 

encompasses various test methods that are part of the FCC regulatory requirements associated with radio 

frequency (RF) safety including specific absorption rate (SAR) and maximum permissible exposure 

(MPE). 

The NVLAP ECT test method standards are in the process of being grouped into one or more of the 

following categories: 

• Electromagnetic emissions; 

• Electromagnetic immunity; 

• Electromagnetic compatibility (both emissions and immunity); 

NIST Handbook 150-11:2007 1 
• Environmental simulation; 

• Harms to the public telecommunications network; 

• Radio conformance; 

• Performance; 

• Interoperability; 

• RF safety; 

• Product safety under normal and/or faulty use. 

Laboratories may seek accreditation in ECT test methods in any of the areas listed above. 

1.4 References 

The following references are important for the application of this handbook. For dated references, only 

the edition cited applies. For undated references, the latest edition of the referenced document applies. 

a) 47 U.S. Code of Federal Regulations (CFR) Telecommunication, Parts 0 through 101 

b) MIL-STD 461, Department of Defense Interface Standard, Requirements for the Control of 

Electromagnetic Interference Characteristics of Subsystems and Equipment 

c) MIL-STD 462, Department of Defense, Measurement of Electromagnetic Interference 

Characteristics of Subsystems and Equipment 

d) ANSI C63 Electromagnetic Compatibility Standards; for OATS site attenuation measurements, 

ANSI C63.4:2003 applies. 

e) IEC 61000 Electromagnetic compatibility (EMC) 

f) CISPR 22, Information technology equipment - Radio disturbance characteristics - Limits and 

methods of measurement 

g) For terms and definitions 1.5.1 through 1.5.4 below: 

BIPM/IEC/IFCC/ISO/IUPAC/IUPAP/OIML, International Vocabulary of Basic and General 

Terms in Metrology (VIM), Draft 3rd Edition, Final 2007-05-18, BIPM Joint Committee for Guides 

in Metrology (JCGM)/WG 2 Document N341. 

1.5 Terms and definitions 

For the purposes of this handbook, the terms and definitions given in NIST Handbook 150 and the 

following apply. 

2 NIST Handbook 150-11:2007 
1.5.1 

calibration 

Operation that, under specified conditions, in a first step establishes a relation between the quantity values 

with measurement uncertainties provided by measurement standards and corresponding indications with 

associated measurement uncertainties and, in a second step, uses this information to establish a relation 

for obtaining a measurement result from an indication. A calibration may be expressed by a statement, 

calibration function, calibration diagram, calibration curve, or calibration table. In some cases it may 

consist of an additive or multiplicative correction of the indication with associated measurement 

uncertainty. 

1.5.2 

uncertainty budget 

Statement of a measurement uncertainty, of the components of that measurement uncertainty, and of their 

calculation and combination. 

1.5.3 

validation 

See verification. 

1.5.4 

verification 

Provision of objective evidence that a given item fulfills specified requirements. When applicable, 

measurement uncertainty should be taken into consideration. 

1.6 Program documentation 

1.6.1 General 

Assessors use NVLAP test method review summary sheets and checklists to ensure that each laboratory 

receives an assessment comparable to that received by others. Checklists assist assessors in documenting 

the assessment to certain specific test methods and to the NVLAP requirements found in NIST Handbook 

150 and this handbook. Checklists and test method review summary sheets form part of the On-Site 

Assessment Report (see NIST Handbook 150). The current versions of these forms are available on the 

NVLAP web site <http://www.nist.gov/nvlap>. 

1.6.2 NIST Handbook 150 Checklist 

All NVLAP programs use the NIST Handbook 150 Checklist (formerly called the General Operations 

Checklist), which contains the requirements published in NIST Handbook 150. The checklist items are 

numbered to correspond to clauses 4 and 5 and annexes A and B of NIST Handbook 150. 

1.6.3 NIST Handbook 150-11 Checklist 

The NIST Handbook 150-11 Checklist (also referred to as the ECT Program-Specific Checklist) 

addresses the requirements specific to electromagnetic compatibility and telecommunications testing 

given in NIST Handbook 150-11. The checklist contains guidance expressed at a more detailed level than 

found in this handbook. Other checklists (such as the FCC Parts 2, 15, and 18 Checklist) may apply and 

are available on the NVLAP web site. 

NIST Handbook 150-11:2007 3 
1.6.4 Test Method Review Summary {if applicable} 

Because of the very large number of relevant standards and test methods in the ECT LAP, the assessor 

uses Test Method Review Summary sheets, along with applicable checklists, to review the laboratory’s 

ability to perform the ECT test methods. The review of the test methods by the assessor ranges from 

observing tests to having laboratory staff describe the test procedures. The assessor notes on the Test 

Method Review Summary the depth into which each part of the test method was reviewed (Observed 

Test, Examined Apparatus, Walked/Talked Through Test, Listened to Description of Procedures). 

1.6.5 NVLAP Lab Bulletins 

NVLAP Lab Bulletins are issued to laboratories and assessors, when needed, to clarify program-specific 

requirements and to provide information about program additions and changes. Current Lab Bulletins are 

posted on the program-specific handbooks page of the NVLAP web site. 

2 LAP establishment, development and implementation 

This clause contains no information additional to that provided in NIST Handbook 150, clause 2. 

3 Accreditation process 

3.1 General 

3.1.1 This clause discusses the assessment and accreditation process for laboratories in the ECT LAP. 

3.1.2 An overview of the laboratory accreditation process is provided in NIST Handbook 150, clause 3, 

and includes information pertaining to application for accreditation; on-site assessment; proficiency 

testing; accreditation decision; granting accreditation; renewal of accreditation; changes to scope of 

accreditation; monitoring visits; and suspension, denial, revocation, and voluntary termination of 

accreditation. 

3.1.3 The assessment process consists of a NVLAP review of the application and laboratory 

management system documentation and an on-site assessment visit. 

3.1.4 For an initial accreditation, NVLAP management may consider a pre-assessment on-site visit to 

better define the laboratory’s requested scope of accreditation. In such cases, the pre-assessment costs will 

be charged to the laboratory in addition to the On-Site Assessment Fee. 

3.2 Management system review 

3.2.1 Prior to applying to NVLAP for accreditation, a laboratory shall have a fully implemented 

management system. A copy of the latest quality manual and relevant associated documents shall be sent 

to NVLAP with the application forms. 

3.2.2 The laboratory shall create a cross-reference document allowing the laboratory and a NVLAP 

assessor to verify that all requirements of clauses 4 and 5 and annexes A and B of NIST Handbook 150 

4 NIST Handbook 150-11:2007 
and the corresponding NIST Handbook 150-11 are addressed in the management system documentation. 

The cross-reference document shall verify that all requirements of this handbook and clauses 4 and 5 and 

annexes A and B of NIST Handbook 150 are addressed and their locations clearly identified in the 

management system documentation. 

3.2.3 Prior to the on-site assessment, the assigned assessor will review all relevant management system 

documentation for conformity with NVLAP requirements, including the requirements of this handbook 

and NIST Handbook 150. During this review, the assessor may request additional management system 

documents and/or records, which will be returned upon request. Because of the very large number of 

relevant standards in the ECT LAP, relevant test method(s), operator instructions and/or test procedures 

may be requested by the assessor for review in advance of the on-site assessment. The laboratory shall 

provide requested test method(s), operator instructions and/or test procedures to the assessor. 

3.3 On-site assessment 

3.3.1 General information 

3.3.1.1 The purpose of the on-site assessment is to determine whether the laboratory is following its 

documented management system and to assess the capability and competence of the laboratory’s delivery 

of its testing services. 

3.3.1.2 The on-site assessment will take place at the laboratory site. Prior to the visit, the NVLAP 

assessor will provide a preliminary agenda, which may change due to findings observed during the on-site 

assessment. Efforts will be made to minimize disruption to the normal working routines during the 

assessment. The assessor will need time and workspace to complete assessment documentation during 

his/her time at the laboratory site. 

3.3.1.3 The laboratory shall make available all supporting technical information in a format that is 

conducive to a detailed review. If relevant documentation is in a language other than English, all relevant 

documentation shall be provided to NVLAP and its assessors in English, prior to the assessment. The 

assessor may request additional information to clarify issues regarding nonconformities or to delve more 

deeply into any technical issues. 

3.3.2 Typical on-site assessment 

3.3.2.1 Opening meeting 

The NVLAP assessor will meet with laboratory management, supervisory personnel, and other 

appropriate staff members to explain the purpose of the on-site assessment and to discuss the schedule for 

the assessment activities. Information provided by the laboratory on its application form may be 

discussed during this meeting. At the discretion of the laboratory manager, other staff may attend the 

briefing. 

3.3.2.2 Staff interviews 

The NVLAP assessor will ask the laboratory manager to assist in arranging times for individual 

interviews with laboratory staff members. The assessor will interview staff members filling key positions 

(e.g., Laboratory Manager, Technical Director, Quality Manager, Authorized Representative, Approved 

Signatories) and staff members who have an effect on the outcome of the testing, including staff who 

conduct the testing. The assessor does not need to talk to all staff members; however, the assessor will 

NIST Handbook 150-11:2007 5 
select staff members representing all aspects of the laboratory. These interviews are conducted to 

determine if the staff members are properly trained, assigned, and supervised, and are technically 

competent for the tasks assigned to them. 

3.3.2.3 Records review 

The NVLAP assessor will review laboratory documentation, including the management system, quality 

manual, equipment and maintenance records, record-keeping procedures, testing procedures, laboratory 

test records and reports, personnel competency records, personnel training plans and records, and 

safeguards for the protection of sensitive and proprietary information. The assessor will review the 

laboratory’s own detailed instructions to perform ECT testing according to the standard test procedures 

for which it seeks accreditation, the range of test items and conditions it can test, and the descriptions of 

the maintenance and calibration of its specific equipment. Also subject to review are: 

 — item identification and tracking procedures and copies of completed test reports; 

 — records of internal audits and use of quality control procedures and participation in 

interlaboratory comparisons or other similar programs; 

 — personnel records, including résumés and job descriptions of key personnel and competency 

evaluations for all staff members who routinely perform the test method for which accreditation 

is sought. 

Laboratory staff shall be available to answer questions; however, the assessor may wish to review the 

documents and records alone. The assessor usually does not ask to remove any laboratory documents or 

records from the laboratory premises. 

NVLAP assessors do not need access to employee information that may be considered sensitive or private 

such as salary, medical information, or performance reviews for work done outside the scope of the 

laboratory’s accreditation. However, this information is often stored together with technical information 

that an assessor will need to check (e.g., job descriptions, résumés, and technical performance reviews). 

In these cases, the assessor will work with the laboratory to ensure the review is performed without 

violating individual privacy. At the discretion of the laboratory, a member of its human resources 

department may be present during the review of personnel information. 

3.3.2.4 Internal audit and management review 

The NVLAP assessor will review and discuss with the laboratory staff the laboratory’s internal audit and 

management review activities, which are separate and distinct activities. The discussion will include all 

aspects of those activities including the management system procedures, the audit findings, the root cause 

determination, the actions taken to resolve problems identified, the actions taken to prevent recurrence, 

and the results of the management review. 

3.3.2.5 Equipment and software 

The NVLAP assessor will examine and determine the suitability of all equipment and facilities required 

to perform the standard test methods for which the laboratory is accredited (or is seeking accreditation). 

The appropriate environmental conditions required for testing will be assessed. The assessor will review 

test data, as well as examine hardware and software for function and appropriateness. 

6 NIST Handbook 150-11:2007 
3.3.2.6 Demonstrations 

The demonstrations requested may be selective or all-inclusive. The NVLAP assessor will observe the 

demonstration of testing procedures by technical personnel assigned to conduct the tests, and will discuss 

the tests with the technical personnel to assure their understanding of the procedures. The demonstrations 

will include preparation of test items, establishment of test conditions, and the setup/use of major 

equipment. The assessor uses the Test Method Review Summary and the ECT Program-Specific 

Checklist in reviewing and summarizing the laboratory’s ability to conduct the test methods. For 

program-specific information about demonstrations, see 3.3.3 and 3.3.4. 

3.3.2.7 Proficiency testing 

The NVLAP assessor will discuss all aspects of proficiency testing results with appropriate staff. Test 

methodology and records documenting the laboratory’s execution of the testing will be reviewed and 

discussed. Any unusual trends or outlying results will be discussed. 

NVLAP reserves the right to provide a test item as a proficiency test sample. The assessor may request 

testing or a demonstration using this or another test item. 

3.3.2.8 On-site assessment report 

The assessor will complete an On-Site Assessment Report, which summarizes the findings and clearly 

lists all nonconformities and comments (positive or negative). This report normally consists of the OnSite Assessment Summary, the On-Site Assessment Narrative Summary, the NIST Handbook 150 

Checklist, the ECT Program-Specific Checklist, and the Test Method Review Summary. The first page of 

the report shall be signed by the assessor and the laboratory’s Authorized Representative or designee to 

acknowledge the discussion, but this does not necessarily indicate agreement by the laboratory. A copy of 

the report is given to the laboratory representative for retention and the assessor sends the original to 

NVLAP. All observations made by the NVLAP assessor are held in the strictest confidence. 

3.3.2.9 Closing meeting 

The NVLAP assessor will conduct a closing meeting with the laboratory manager, supervisory personnel, 

and other appropriate staff members to discuss the findings. During the visit the assessor will have 

categorized all problems identified as nonconformities and comments. They will be discussed at the 

closing meeting and resolutions may be mutually agreed upon. The assessor will specifically note items 

that have been corrected during the on-site assessment along with any recommendations for other 

action(s). The process for resolving nonconformities identified during the on-site is documented in NIST 

Handbook 150. Disagreements between the laboratory and an assessor may be referred to NVLAP for 

resolution. 

3.3.3 Specific requirements for ECT on-site assessments 

3.3.3.1 All laboratory equipment required to perform accredited testing shall be available for assessment 

and in good working order. The assessor will physically examine equipment and facilities. This includes 

storage areas, shielded enclosures, open area test sites (OATS), anechoic chambers, pre-scan areas, test 

benches, electronics, test jigs, and antennas, as appropriate. 

3.3.3.2 The laboratory shall have appropriate OATS measurement description reports and OATS 

attenuation data available. The assessor will review OATS measurement description reports and OATS 

NIST Handbook 150-11:2007 7 
attenuation data. Unless the OATS meets the description of a standard site, the site attenuation 

measurements shall be performed using the volumetric techniques described in ANSI C63.4:2003. 

3.3.3.3 For FCC CFR Part 68-Connection of Terminal Equipment to the Telephone Network, and other 

similar standards and regulations, an appropriate test artifact shall be used to demonstrate the test 

equipment. 

3.3.3.4 Both automatic and manual systems will be evaluated in detail by the assessor as applicable. 

3.3.4 Demonstrations 

3.3.4.1 Assessor safety 

Some test methods may be harmful to staff and/or assessors if appropriate safety measures are not 

employed. The purpose of the assessment is not to do an Occupational Safety and Health Administration 

(OSHA) workplace safety assessment; however, the assessors or observers will not knowingly expose 

themselves or others to a hazardous or inappropriate test method environment. All appropriate personnel 

safety measures shall be taken. These measures include: 

 — observing any warnings and precautions contained in user manuals; 

 — following all appropriate OSHA workplace statutory mandates. 

Examples of potentially hazardous test methods or situations include: 

 — inappropriate implementation of techniques required by the technical standard; 

 — applying surges to products without associated personal protective equipment such as eye 

protection or physical barriers; 

 — exposing observers to the radiated electromagnetic fields associated with radio frequency 

immunity test methods (if demonstrating in an OATS environment); 

 — employing field strengths that expose observers to magnitudes that exceed the MPE 

requirements specified by the FCC in CFR 47, Part 2, 1091; 

 — faulty implementation of “ground isolation” requirements. 

The assessor will not observe potentially hazardous test methods without appropriate measures taken to 

protect all observers and staff from harm. Standards associated with these test methods will be removed 

from the scope of accreditation if appropriate protective measures have not been taken. 

3.3.4.2 Conducted and radiated emissions measurements 

Demonstrations shall include the use of receivers and/or spectrum analyzers in shielded enclosures, prescan areas, OATS, and/or fully or semi-anechoic chambers. 

3.3.4.3 OATS validation as part of demonstration 

As part of the demonstration of measurement, an OATS or an alternative site, shall be validated at three 

frequencies of measure in both horizontal and vertical polarization. 

8 NIST Handbook 150-11:2007 
3.3.4.4 Requested demonstrations to be minimally disruptive 

The assessor will attempt to minimize disruption to the normal working routine; however, the laboratory 

shall demonstrate the appropriate test methods in the scope of accreditation. NVLAP may not accredit to 

test methods that are not demonstrated unless other suitable means of demonstrating competency in this 

area are found acceptable. 

3.3.4.5 Assessor transportation 

The laboratory shall ensure transportation to OATS or alternative site for the assessor. 

3.3.4.6 Demonstrations for multiple OATS 

If the laboratory maintains more than one OATS and/or alternative site, the assessor will ask questions to 

determine whether all sites are operated and equipped in the same manner. Usually one site will be 

examined; however, at the discretion of the assessor, more than one site may be examined. 

3.3.4.7 OATS validation reports 

Validation reports for all OATS and/or alternative sites shall be available that include all the requirements 

stated in 47 CFR Part 2.948. 

3.3.5 Nonconformity resolution 

3.3.5.1 The laboratory shall resolve or formulate a plan to resolve all nonconformities and provide a 

response to NVLAP within 30 days from the date of the on-site assessment. For the case of an initial 

application, all non-conformities must be resolved before accreditation is granted (i.e., plans are not 

acceptable). 

3.3.5.2 The laboratory shall review all comments for potential improvements in their electromagnetic 

compatibility testing program. 

4 Management requirements for accreditation 

4.1 Organization 

There are no requirements additional to those set forth in NIST Handbook 150. 

4.2 Management system 

4.2.1 The laboratory shall ensure that the requirements of NIST Handbook 150 are met so that staff are 

knowledgeable of the electronic- or paper-based documentation system and can demonstrate, if 

authorized, the retrieval of needed documents and/or records. 

4.2.2 The laboratory shall create a cross-reference document allowing the laboratory and a NVLAP 

assessor to verify that all requirements of clauses 4 and 5 and annexes A and B of NIST Handbook 150 

and the corresponding NIST Handbook 150-11 are addressed in the management system documentation. 

NIST Handbook 150-11:2007 9 
The cross-reference document requirement can be satisfied in a number of ways. One way is to number 

and organize the management system documentation to be the same as the NIST Handbook 150 

Checklist. 

4.2.3 The laboratory shall have readily available the regulation(s) and the applicable version of the 

standard(s) for the test methods for which accreditation has been requested. 

4.2.4 If a customer, for whatever reason (e.g., regulatory requirement), requires accreditation to a 

version of a test method that is not the latest published version, then the laboratory shall document that 

requirement and shall have readily available the required version of the test method. 

4.2.5 When a test method references another test method, guide, practice, or specification, the 

laboratory shall have readily available the referenced documents, where relevant. 

4.2.6 In addition to the information specified in NIST Handbook 150, the management system 

documentation shall include: 

a) testing facilities and scope of relevant services offered; 

b) policy and procedures for use of subcontractors, if applicable; 

c) procedures for receipt, identification, and tracking of test items; 

d) procedures by which the laboratory describes the test items and the criteria, if any, for determining 

if the item is acceptable for testing; 

e) procedures and actions concerning damaged or altered test items; 

f) procedures for maintenance and calibration of the equipment used in conducting the tests; 

g) descriptions of the procedures, practices, and equipment that the laboratory uses in conducting tests; 

h) if the laboratory participates in any laboratory comparisons, procedures for interlaboratory 

comparison and the laboratory’s participation in proficiency testing, a summary of the results, and a 

description of any corrective actions taken because of the results; 

i) the personnel training and competency evaluations, which demonstrate that the test procedures are 

being followed correctly. 

4.3 Document control 

The master list that identifies the current revision status and distribution of documents shall include all 

national and/or international standards on the scope of accreditation (see NIST Handbook 150, 4.3.2.1). 

4.4 Review of requests, tenders and contracts 

All requests, tenders and contracts shall be available for selection and examination by the assessor. At 

least one project record will be examined by the assessor. The record will include the request, the 

10 NIST Handbook 150-11:2007 
quotation, the proposal, and the review of the contract along with receipt of the equipment under test 

(EUT). 

4.5 Subcontracting of tests and calibrations 

There are no requirements additional to those set forth in NIST Handbook 150. 

NOTE Subcontracting applies to any of the test methods on the scope of accreditation. 

4.6 Purchasing services and supplies 

There are no requirements additional to those set forth in NIST Handbook 150. 

NOTE Laboratories should pay special attention to the purchasing of calibration services from calibration service 

providers (CSP). The technical requirements of the calibration need to be specified by the laboratory as well as 

conformance to the appropriate traceability requirements in Annex B of NIST Handbook 150. Assessors will seek to 

determine that laboratory calibration records identify the measurement parameters, as well as the traceability chain 

for each parameter. 

For example, for Line Impedance Stabilization Network used for FCC-conducted emissions measurements, the 

technical requirements of the calibrations are contained in ANSI C63.4:2003, Annex F; and, traceability needs to be 

established for attenuation, impedance, and frequency. 

4.7 Service to the customer 

There are no requirements additional to those set forth in NIST Handbook 150. 

4.8 Complaints 

There are no requirements additional to those set forth in NIST Handbook 150. 

4.9 Control of nonconforming testing and/or calibration work 

There are no requirements additional to those set forth in NIST Handbook 150. 

NOTE Laboratories are expected to provide data within the expected band of measurement uncertainty established 

by the laboratory. If data falls outside the expected measurement uncertainty, then this data is “nonconforming 

work.” Written notification should be provided to each client who has been affected. Even if the outcome of the test 

remains the same (i.e., the conducted emissions data still meets the requirements of the FCC Part 15.207 limit for 

Class B equipment), the client should still be informed if the data are changed significantly (positively or 

negatively). 

4.10 Improvement 

There are no requirements additional to those set forth in NIST Handbook 150. 

NIST Handbook 150-11:2007 11 
4.11 Corrective action 

There are no requirements additional to those set forth in NIST Handbook 150. 

4.12 Preventive action 

There are no requirements additional to those set forth in NIST Handbook 150. 

4.13 Control of records 

There are no requirements additional to those set forth in NIST Handbook 150. 

4.14 Internal audits 

4.14.1 The internal audit shall cover compliance with NVLAP, the laboratory management system, as 

well as regulatory, contractual, and testing requirements. 

4.14.2 An internal audit of test method competency shall be included as part of the internal audit plan 

and its execution. 

4.14.3 An applicant laboratory shall conduct at least one complete internal audit prior to the first on-site 

assessment. The records will be reviewed by the NVLAP assessor before or during the on-site assessment 

visit. 

4.14.4 For accredited laboratories, internal audit reports conducted since the previous on-site assessment 

shall be made available for review. 

4.14.5 Internal audits are separate and distinct from management reviews (see 4.15). 

4.15 Management reviews 

4.15.1 Periodic reviews of the management system shall reflect adherence to NVLAP requirements and 

the laboratory’s quality objectives. 

4.15.2 Management reviews shall review all nonconformities and may reflect positive aspects of the 

management system. 

4.15.3 An applicant laboratory shall perform at least one complete management review prior to the first 

on-site assessment. The records will be reviewed by the NVLAP assessor before or during the on-site 

assessment visit. 

4.15.4 The report of the management review shall be available during the NVLAP on-site assessment. 

12 NIST Handbook 150-11:2007 
5 Technical requirements for accreditation 

5.1 General 

There are no requirements additional to those set forth in NIST Handbook 150. 

5.2 Personnel 

5.2.1 An observation and an evaluation of performance shall be conducted at least annually by the 

immediate supervisor or a designee appointed by the laboratory director. A record of the annual 

evaluation of each staff member shall be dated and signed by the supervisor and the employee. 

5.2.2 Accredited test method competency shall be included as part of the personnel competency 

evaluations. 

5.3 Accommodation and environmental conditions 

5.3.1 All appropriate personnel safety precautions and warnings shall be taken. 

5.3.2 FCC Part 15-Radio Frequency Devices: A facility layout plan of the laboratory, including a 

complete description of the laboratory’s OATS, and drawings and descriptions of the surrounding area 

and adjacent structures shall be available, if applicable. If a facility other than an OATS is used, a 

complete description shall be available along with documentation of equivalence. The laboratory’s 

submission to the FCC Description of Measurement Facilities Program shall be documented. 

5.3.3 All parts of OATS shall be operational and available for inspection during the on-site visit. The 

site attenuation shall be checked per ANSI C63.4 at least once per year and complete written records shall 

be maintained. The site attenuation shall also be checked if significant changes are made in or near the 

OATS. This information will be reviewed during the on-site assessment visit. 

5.3.4 FCC Part 68-Connection of Terminal Equipment to the Telephone Network: The laboratory shall 

have a procedure for checking the testing system before each use. This is especially important for 

automated systems. The laboratory shall have at least one telephone device reserved for use in periodic 

checks of the test system. 

5.4 Test and calibration methods and method validation 

5.4.1 All appropriate safety precautions and warnings shall be incorporated in test and calibration 

methods. 

5.4.2 Measurement uncertainty (MU) shall be estimated for all test methods within the laboratory’s 

scope of accreditation. 

NIST Handbook 150-11:2007 13 
5.4.3 MU is handled in different ways depending on the test method standards that are employed. 

a) If MU can be calculated from Type A and B uncertainties, then the procedure shall follow GUM or 

NIST Technical Note 1297 (see references in NIST Handbook 150, 1.4). Unless stated by the 

standard, the coverage factor (k) shall be equal to 2 (two) such that the confidence interval is 95 %. 

b) In some instances, the standard provides a MU budget as part of the test method. Examples include 

European Telecommunications Standards Institute (ETSI) standards concerning radio 

measurements. Each MU budget shall be supported with calibration and computational data 

applicable to the test method as performed by that laboratory. 

c) In some instances, the standard provides a tolerance for the test method (and does not refer to 

“measurement uncertainty”). Examples include MIL-STD 461E:1999, 4.2.1 (d), which defines a 

+/- 3 dB tolerance for the measurement system (and an antenna to receiver tolerance of +/- 3 dB). 

The tolerance stated in the standard shall be supported by calibration data, MU budgets and/or other 

appropriate calculations. 

d) In some instances, the standard provides a tolerance for the test components and/or instrumentation, 

but not the test method. The tolerance of the instrument should be supported by calibration and 

computational data or comparison to some other appropriate measurement standard (see NIST 

Handbook 150, Annex B, B.3.5). At this time, MU calculation or budgets are not required but 

should be estimated in so far as possible and reasonable. 

e) When the normative standard references neither a MU nor tolerance, the laboratory’s own 

uncertainties associated with the method, its instrumentation, and comparisons between operators, 

locations, etc. should be estimated in so far as possible and reasonable. 

5.5 Equipment 

5.5.1 Shielded enclosure 

The laboratory shall specify how it will monitor and record the performance of its shielded enclosure, 

how often, and what data shall be recorded. For example, any changes made in or near the shielded 

enclosure should warrant that the enclosure’s performance be verified. Requirements for checking 

associated critical equipment, such as power line filters, and grounding systems shall also be specified 

and the results documented. 

5.5.2 Line impedance stabilization networks 

Line impedance stabilization networks (LISN) shall be calibrated for insertion loss and the impedance 

verified at least once per year. 

5.5.3 Surge generator 

The waveform of the surge generator shall be verified with an oscilloscope at least once per year and 

photographs of the waveform shall be kept on file. 

14 NIST Handbook 150-11:2007 
5.5.4 Software and firmware 

Software and/or firmware associated with automated test equipment (either stand-alone or computercontrolled) shall be validated before use. This includes validation of any software updates from the 

original equipment manufacturer (OEM) or other source. 

5.6 Measurement traceability 

5.6.1 Antennas, spectrum analyzers, measurement receivers 

If a laboratory calibrates its own antennas, spectrum analyzers, and/or measurement receivers, explicit 

procedures and instructions for those calibrations shall be maintained. Measurement uncertainties 

associated with these calibrations shall be estimated and reported in the calibration report. Antennas shall 

be calibrated to a standard (e.g., ANSI C63.5:2005 or SAE ARP-958). 

5.6.2 Automated test systems including spectrum analyzer measurements 

When automated test equipment or automated test systems are used, appropriate documentation, 

instructions for use and training are required. 

5.6.3 Computer software and firmware 

5.6.3.1 Computer software shall be included in the laboratory inventory system. 

5.6.3.2 When software or firmware in an automated system is changed by the equipment manufacturer or 

supplier, whether for maintenance, repair, or upgrade, the laboratory shall verify or validate that the 

changed software is correct before using it in testing. 

5.6.3.3 A staff member shall be responsible for implementing, documenting and verifying changes to 

software and associated hardware. 

5.6.3.4 Documentation shall include initial testing, error detection and resolution, updates, and upgrades. 

5.7 Sampling 

There are no requirements additional to those set forth in NIST Handbook 150. 

NOTE The requirements in NIST Handbook 150 for sampling pertain to a laboratory’s selecting the sample to be 

tested. For most ECT test methods, the sample(s) is(are) selected by the laboratory’s customer. 

5.8 Handling of test and calibration items 

There are no requirements additional to those set forth in NIST Handbook 150. 

5.9 Assuring the quality of test and calibration results 

There are no requirements additional to those set forth in NIST Handbook 150. 

NIST Handbook 150-11:2007 15 
5.10 Reporting the results 

There are no requirements additional to those set forth in NIST Handbook 150. 

NOTE Certain standards have special reporting requirements (e.g., ANSI C63.4, Section 10 and/or 

MIL-STD-462). 

6 Additional requirements 

The Declaration of Conformity (DoC) for the FCC Equipment Authorization (EA) program requires 

laboratory accreditation and foreign economy/country participation in a Mutual Recognition Agreement 

(MRA) with the United States. A NVLAP-accredited laboratory shall not provide a test report that uses 

subcontractor data from a non-MRA member foreign economy as part of the DoC equipment approval 

process. The most current list of MRAs with the United States is available at 

<http://ts.nist.gov:80/Standards/Global/mra.cfm>. 

NOTE Annex A of NIST Handbook 150 [A.1. i)] requires that any data provided by a subcontractor be clearly 

identified, and that the report prominently display the statement at the beginning of the report: “This report contains 

data that were produced under subcontract by Laboratory X.” If the subcontracted laboratory is accredited by 

NVLAP, then its Lab Code should also be stated. If the subcontracted laboratory is accredited by a body other than 

NVLAP, then the name of the accreditation body and the laboratory’s number or other unique identifier should also 

be stated. Under the FCC EA program’s DoC, the subcontracted laboratory shall be accredited. 

16 NIST Handbook 150-11:2007 
Annex A 

(informative) 

Information about selected ECT standards 

This annex provides information about common ECT standards. The information is sorted by the name 

of the standards-issuing body (Table 1), the acronym for the standards-issuing body (Table 2), and, if 

available, the national economy for which the standard has been issued (Table 3). 

Table 1. Names of standards-issuing bodies for common ECT standards 

Name of standard or standards body Acronym National economy 

SEMI SEMI 

American National Standards Institute ANSI United States 

Association of Radio Industries and 

Businesses ARIB Japan 

ASTM International (formerly American 

Standards and Testing Materials) ASTM United States 

Australian Communications Authority ACA Australia 

Australian Communications Industry Forum ACIF Australia 

Australian Standard/New Zealand Standard AS/NZS Australia and New Zealand 

British National standard BN United Kingdom 

British Standard BS United Kingdom 

Broadcasting Equipment Technical Standards BETS Canada 

Bureau of Standards, Metrology and 

Inspection BSMI Taiwan 

Canadian Standards Association CSA Canada 

Chinese National Standards CNS Taiwan 

Comité European de Normalisation 

Electrotechnique (European Committee for 

Electrotechnical Standardization) 

CENELEC European Union 

Directorate General of Telecommunications DGT Taiwan 

Electronic Industries Alliance EIA United States 

European Norms (European Standards) EN European Union 

European Telecommunications Standards 

Institute ETS or ETSI European Union 

Federal Communications Commission FCC United States 

Federal Transit Administration (formerly the 

Urban Mass Transportation Administration) FTA United States 

General Requirement (see Network 

Equipment Building System) GR 

Hong Kong Telecommunications Authority HKTA Hong Kong 

NIST Handbook 150-11:2007 17 
Name of standard or standards body Acronym National economy 

Infocomm Development Authority IDA Singapore 

Institute of Electrical and Electronics 

Engineers, Inc. IEEE 

Interference-Causing Equipment Standard ICES Canada 

International Electrotechnical Commission IEC 

International Organization for Standardization ISO 

International Telecommunications Union – 

Telecommunication Standardization Sector ITU-T 

Korean Norms KN Korea 

Korean Standards KS Korea 

Military Standards MIL-STD United States 

Ministry of Information and Communication MIC Korea 

Network Equipment Building System NEBS 

Public Land Mobile Network PLMN Taiwan 

Public Switched Telephone Network PSTN Taiwan 

Radio Research Laboratory RRL Korea 

Radio Standards Specification RSS Canada 

Radio Telecommunications Terminal 

Equipment RTTE Taiwan 

RTCA (formerly Radio Technical 

Commission for Aeronautics) RTCA United States 

Society of Automotive Engineers, Inc. SAE United States 

Special International Committee on Radio 

Interference (see also IEC) CISPR 

Telecommunications Industry Association TIA United States 

Underwriters Laboratories, Inc. UL United States 

Urban Mass Transportation Administration 

(now the Federal Transit Administration) UMTA United States 

Voluntary Control Council for Interference by 

Information Technology Equipment VCCI Japan 

18 NIST Handbook 150-11:2007 
Table 2. Acronyms of standards-issuing bodies for common ECT standards 

Acronym Name of standard or standards body National economy 

ACA Australian Communications Authority Australia 

ACIF Australian Communications Industry Forum Australia 

ANSI American National Standards Institute United States 

ARIB Association of Radio Industries and Businesses Japan 

AS/NZS Australian Standard/New Zealand Standard Australia and New 

Zealand 

ASTM ASTM International (formerly American Standards and 

Testing Materials) United States 

BETS Broadcasting Equipment Technical Standards Canada 

BN British National standard United Kingdom 

BS British Standard United Kingdom 

BSMI Bureau of Standards, Metrology and Inspection Taiwan 

CENELEC 

Comité European de Normalisation Electrotechnique 

(European Committee for Electrotechnical 

Standardization) 

European Union 

CISPR Special International Committee on Radio Interference 

(see also IEC) 

CNS Chinese National Standards Taiwan 

CSA Canadian Standards Association Canada 

DGT Directorate General of Telecommunications Taiwan 

EIA Electronic Industries Alliance United States 

EN European Norms (European Standards) European Union 

ETS or ETSI European Telecommunications Standards Institute European Union 

FCC Federal Communications Commission United States 

FTA Federal Transit Administration (formerly the Urban Mass 

Transportation Administration) United States 

GR General Requirement (see Network Equipment Building 

System) 

HKTA Hong Kong Telecommunications Authority Hong Kong 

ICES Interference-Causing Equipment Standard Canada 

IDA Infocomm Development Authority Singapore 

IEC International Electrotechnical Commission 

IEEE Institute of Electrical and Electronics Engineers, Inc. 

ISO International Organization for Standardization 

ITU-T International Telecommunications Union – 

Telecommunication Standardization Sector 

KN Korean Norms Korea 

NIST Handbook 150-11:2007 19 
Acronym Name of standard or standards body National economy 

KS Korean Standards Korea 

MIC Ministry of Information and Communication Korea 

MIL-STD Military Standards United States 

NEBS Network Equipment Building System 

PLMN Public Land Mobile Network Taiwan 

PSTN Public Switched Telephone Network Taiwan 

RRL Radio Research Laboratory Korea 

RSS Radio Standards Specification Canada 

RTCA RTCA (formerly Radio Technical Commission for 

Aeronautics) United States 

RTTE Radio Telecommunications Terminal Equipment Taiwan 

SAE Society of Automotive Engineers, Inc. United States 

SEMI SEMI 

TIA Telecommunications Industry Association United States 

UL Underwriters Laboratories, Inc. United States 

UMTA Urban Mass Transportation Administration (now the 

Federal Transit Administration) United States 

VCCI Voluntary Control Council for Interference by Information 

Technology Equipment Japan 

20 NIST Handbook 150-11:2007 
Table 3. National economies of standards-issuing bodies for common ECT standards 

National economy Acronym Name of standard or standards body 

SEMI SEMI 

GR General Requirement (see Network Equipment Building 

System) 

IEEE Institute of Electrical and Electronics Engineers, Inc. 

IEC International Electrotechnical Commission 

ISO International Organization for Standardization 

ITU-T International Telecommunications Union – 

Telecommunication Standardization Sector 

NEBS Network Equipment Building System 

CISPR Special International Committee on Radio Interference (see 

also IEC) 

Australia ACA Australian Communications Authority 

Australia ACIF Australian Communications Industry Forum 

Australia and New 

Zealand AS/NZS Australian Standard/New Zealand Standard 

Canada BETS Broadcasting Equipment Technical Standards 

Canada CSA Canadian Standards Association 

Canada ICES Interference-Causing Equipment Standard 

Canada RSS Radio Standards Specification 

European Union CENELEC Comité European de Normalisation Electrotechnique 

(European Committee for Electrotechnical Standardization) 

European Union EN European Norms (European Standards) 

European Union ETS or 

ETSI European Telecommunications Standards Institute 

Hong Kong HKTA Hong Kong Telecommunications Authority 

Japan ARIB Association of Radio Industries and Businesses 

Japan VCCI Voluntary Control Council for Interference by Information 

Technology Equipment 

Korea KN Korean Norms 

Korea KS Korean Standards 

Korea MIC Ministry of Information and Communication 

Korea RRL Radio Research Laboratory 

Singapore IDA Infocomm Development Authority 

Taiwan BSMI Bureau of Standards, Metrology and Inspection 

Taiwan CNS Chinese National Standards 

Taiwan DGT Directorate General of Telecommunications 

NIST Handbook 150-11:2007 21 
National economy Acronym Name of standard or standards body 

Taiwan PLMN Public Land Mobile Network 

Taiwan PSTN Public Switched Telephone Network 

Taiwan RTTE Radio Telecommunications Terminal Equipment 

United Kingdom BN British National standard 

United Kingdom BS British Standard 

United States ANSI American National Standards Institute 

United States ASTM ASTM International (formerly American Standards and 

Testing Materials) 

United States EIA Electronic Industries Alliance 

United States FCC Federal Communications Commission 

United States FTA Federal Transit Administration (formerly the Urban Mass 

Transportation Administration) 

United States MIL-STD Military Standards 

United States RTCA RTCA (formerly Radio Technical Commission for 

Aeronautics) 

United States SAE Society of Automotive Engineers, Inc. 

United States TIA Telecommunications Industry Association 

United States UL Underwriters Laboratories, Inc. 

United States UMTA Urban Mass Transportation Administration (now the Federal 

Transit Administration) 

22 NIST Handbook 150-11:2007