San Joaquin Valley Air Pollution Control DistrictJune 30, 2011____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-12011 Air Monitoring Network PlanAppendix B: Detailed Air Monitoring Site InformationSites organized by County, alphabetical therein:CountyMSASite NamePage NumberClovis–Villa B-2Fresno–Drummond B-6Fresno–First B-9Fresno–Pacific B-13Fresno–Sky Park B-15Huron B-17Parlier B-19Fresno FresnoTranquillity B-22Arvin–Di Giorgio B-24Bakersfield–Planz B-26Bakersfield–California B-28Edison B-31Lebec B-33Maricopa B-35Oildale B-37Kern BakersfieldShafter B-39Corcoran–Patterson B-42Hanford–Irwin B-45Kings Hanford – CorcoranSanta Rosa Rancheria B-48Madera--City B-50Madera MaderaMadera--Pump Yard B-52Merced–Coffee Road B-55Merced MercedMerced M Street B-57Manteca B-59Stockton–Hazelton B-61Stockton–Wagner/Holt B-64San Joaquin StocktonTracy–Airport B-66Modesto–14thStreet B-69Stanislaus ModestoTurlock B-72Porterville B-75Sequoia–Ash Mountain B-77Sequoia–Lower Kaweah B-79Visalia–Airport B-81Tulare Visalia – PortervilleVisalia–Church B-83San Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-22011 Air Monitoring Network PlanSite name Clovis–VillaAIRS #060195001CountyFresnoCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyData reported by SJVAPCD: PM2.5 FEM, CO,NO2,NMHC (PAMS), TOTAL NMOC, MeteorologyData reported by CARB: PM10 FRMSite Start Date9/1/90Pollutant ParametersOzone, PM10 FRM, PM2.5 FEM, CO,NO2, NMHC (PAMS), TOTAL NMOCMeteorological ParametersWind speed, wind direction, outdoor temperature, relative humidity, barometric pressure, solar radiationAddress908 N. Villa Ave., Clovis CA 93612Latitude36.81944Longitude-119.716Elevation (m)86LocationPortable building in lotDistance to road500 m + (east)Traffic Count4876Ground CoverPavedSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-32011 Air Monitoring Network PlanClovis–Villa (1 of 3)PollutantOzone PM10 FRM PM2.5 FEMParameter Code 44201 81102 88101Spatial scaleNeighborhood Neighborhood NeighborhoodSite typePopulation Population PopulationMonitor objective Timely/public,standards/strategy,research supportStandards/strategy,research supportTimely/publicMonitor typeSLAMS SLAMS SLAMSPOC (or primary monitor for PM2.5 and PM10) 1 1 3Method code087 063 170Sampling method (List Instrument) 400 E Sierra Andersen Met One 1020Analysis method UV Gravimetric Beta attenuationStart date1/1/1990 1/1/1990 4/26/2005Operation schedule (e.g. 1:1, 1:3) 1:1 1:6 1:1Sampling season ALL YEAR ALL YEAR ALL YEARProbe height (meters) 7.5 m 7.0 m 7.0 mDistance from supporting structure (meters) 4.5 m 0.25 m 4.0 mDistance from obstructions on roof ____ _____ _____Distance from obstructions not on roof (meters) 32.0 m 31.5 m 31.0 mDistance from trees (meters) 24.5 m 27.5 m 25.0 mDistance to furnace or incinerator flue (meters) 16.0 m 15.5 m 17.0 mDistance between collocated monitors (meters) _____3.7 m 2.5 mUnrestricted airflow (degrees) 355 355 355Probe material (Teflon, etc.) TEFLON _____ ALUMINUMResidence time (seconds) 12.6 _____ _____Frequency of flow rate verification for manual PMsamplers audit_____ Quarterly _____Frequency of flow rate verification for automatedPM analyzers audit_____ _____ Bi-weeklyFrequency of one-point QC check (gaseous) 1:1 __________Last Annual Performance Evaluation (gaseous) 10/20/2009, 11/2/2010 _____ _____Last two semi-annual flow rate audits for PMmonitors_____ 5/27/2010, 11/2/2010 5/27/2010, 11/2/2010Changes planned within the next 18 months(Y/N))N NNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-42011 Air Monitoring Network PlanClovis–Villa (2 of 3)PollutantCONO2Total NMOC NMHC (PAMS)Parameter code 42101 42602 43102 ManySpatial scaleNeighborhood Neighborhood Neighborhood NeighborhoodSite typePopulation High concentration Population PopulationMonitor objective Standards/strategy Standards/strategy,researchResearch ResearchMonitor typeSLAMS PAMS PAMS PAMSPOC1 1 1 1Sampling method (List Instrument) 48i-TLE 42C 910A,925 55Method code054 074 164 177Analysis method IR CL GC GCStart date1/1/1990 1/1/1990 1/1/1990 1/1/1990Operation schedule (e.g. 1:1, 1:3) 1:1 1:1 1:3 1:1Sampling season ALL YEAR ALL YEAR JUN-JUL-AUG ALL YEARProbe height (meters) 7.5 m 7.5 m 6.5 m 7.5 mDistance from supporting structure (meters) 4.5 m 4.5 m 0.25 m 4.5 mDistance from obstructions on roof ____ ____ ____ ____Distance from obstructions not on roof (meters) 32.0 m 32.0 m 33.5 m 32.0 mDistance from trees (meters) 24.5 m 24.5 m 28.0 m 24.5 mDistance to furnace or incinerator flue (meters) 16.0 m 16.0 m 13.5 m 16.0 mDistance between collocated monitors (meters) __________ ____ ____Unrestricted airflow (degrees) 355 355 350 355Probe material (Teflon, etc.) TEFLON TEFLON S. STEEL TEFLONResidence time (seconds) 11.6 11.6 _____ _____Frequency of flow rate verification for manualPM samplers audit_____ _____ _____ _____Frequency of flow rate verification for automatedPM analyzers audit_____ _____ _____ _____Frequency of one-point QC check (gaseous) 1:1 1:1 _____ 1:1Last Annual Performance Evaluation (gaseous) 11/2/2010 11/2/2010 5/10/2010 _____Last two semi-annual flow rate audits for PMmonitors_____ _____ _____ _____Changes planned within the next 18 months(Y/N))NNNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-52011 Air Monitoring Network PlanClovis–Villa (3 of 3)PollutantMet ParametersParameter codeManySpatial scaleRegionalSite typeGeneralMonitor objectiveResearch, timely/publicMonitor typePAMSPOC1Method codeManySampling method (List Instrument) ITP-125-125 HV,OT-06A-2, BP-092, RH-HMP45D, SRD-Mod.8-48, WD020C, WS-010CAnalysis method_____Start date1/1/1990Operation schedule (e.g. 1:1, 1:3) 1:1Sampling seasonALL YEARProbe height (meters)9.6 mDistance from supporting structure (meters) 2.7 mDistance from obstructions on roof _____Distance from obstructions not on roof (meters) 29.5 mDistance from trees (meters) 25.5 mDistance to furnace or incinerator flue (meters) _____Distance between collocated monitors (meters) _____Unrestricted airflow (degrees) 360Probe material (Teflon, etc.) _____Residence time (seconds) _____Frequency of flow rate verification for manual PMsamplers audit_____Frequency of flow rate verification for automated PManalyzers audit_____Frequency of one-point QC check (gaseous) _____Last Annual Performance Evaluation (gaseous) _____Last two semi-annual flow rate audits for PM monitors _____Changes planned within the next 18 months (Y/N)) NSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-62011 Air Monitoring Network PlanSite name Fresno–DrummondAIRS #060190007CountyFresnoCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyData reported by SJVAPCD: Ozone, CO,NO2Data reported by CARB: PM10 FRMSite Start Date7/1/84Pollutant ParametersOzone, PM10 FRM, CO,NO2Meteorological ParametersWind speed, wind direction, outdoor temperature, barometric pressureAddress4706 E. Drummond Street, Fresno CA 93725Latitude36.70556Longitude-119.741Elevation (m)89LocationPortable building in parking lotDistance to road42.5 m (north), 121 m (east)Traffic Count600Ground CoverPavedSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-72011 Air Monitoring Network PlanFresno–Drummond (1 of 2)PollutantOzone PM10 FRM CONO2Parameter code 44201 81102 42101 42602Spatial scaleNeighborhood Neighborhood Neighborhood NeighborhoodSite typePopulation,regional transportPopulation Population High concentrationMonitor objective Timely/public,standards/strategy,research supportStandards/strategy,research supportStandards/strategy Standards/strategyMonitor typeSLAMS SLAMS SLAMSSLAMSPOC1 111Method code087 063 054 074Sampling method (List Instrument) 400 E Sierra Andersen 48 42CAnalysis method UV Gravimetric IR CLStart date7/1/1984 7/1/1984 7/1/1984Operation schedule (e.g. 1:1, 1:3) 1:1 1:6 1:1 1:1Sampling season ALL YEAR ALL YEAR ALL YEAR ALL YEARProbe height (meters) 8.5 m 6 m 8.5 m 8.5 mDistance from supporting structure (meters) _____ 10.5 m _____ _____Distance from obstructions on roof _____ 0.5 m _____ _____Distance from obstructions not on roof (meters) _____ 5 m _____ _____Distance from trees (meters) 25 m 24 m 25 m 25 mDistance to furnace or incinerator flue (meters) 23.5 m 23 m 23.5 m 23.5 mDistance between collocated monitors (meters) __________ _____ _____Unrestricted airflow (degrees) 360 260 360 360Probe material (Teflon, etc.) TEFLON _____ TEFLON TEFLONResidence time (seconds) 12.8 _____ 12.6 12.9Frequency of flow rate verification for manual PMsamplers audit_____ Quarterly _____ _____Frequency of flow rate verification for automatedPM analyzers audit_____ _____ _____ _____Frequency of one-point QC check (gaseous) 1:1 _____1:1 1:1Last Annual Performance Evaluation (gaseous) 3/1/2011 _____ 3/1/2011 3/1/2011Last two semi-annual flow rate audits for PMmonitors_____ 2/10/2010,8/1/2010, 3/1/2011_____ _____Changes planned within the next 18 months (Y/N) NNNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-82011 Air Monitoring Network PlanFresno–Drummond (2 of 2)PollutantMet ParametersParameter codeManySpatial scaleRegionalSite typeGeneralMonitor objectiveResearch, timely/publicMonitor typeSLAMSPOC1Method codeManySampling method (List Instrument) ITP-125-125 HV,OT-060A-2, BP-092, WD-020C, WS-010CAnalysis method_____Start date10/7/2004Operation schedule (e.g. 1:1, 1:3) 1:1Sampling seasonALL YEARProbe height (meters)10 mDistance from supporting structure (meters) _____Distance from obstructions on roof _____Distance from obstructions not on roof (meters) _____Distance from trees (meters) 25 mDistance to furnace or incinerator flue (meters) 23mDistance between collocated monitors (meters) _____Unrestricted airflow (degrees) 360Probe material (Teflon, etc.) _____Residence time (seconds) _____Frequency of flow rate verification for manual PMsamplers audit_____Frequency of flow rate verification for automated PManalyzers audit_____Frequency of one-point QC check (gaseous) _____Last Annual Performance Evaluation (gaseous) _____Last two semi-annual flow rate audits for PM monitors _____Changes planned within the next 18 months (Y/N) NSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-92011 Air Monitoring Network PlanSite name Fresno–FirstAIRS #060190008CountyFresnoCollecting (Operating)AgencyAll equipment operated by CARBReporting AgencyData reported by CARB: PM10 FRM, PM10 FEM,PM2.5 FRM, PM2.5 Non-FEM, CO,NO2, ToxicsData reported by NPS:SO2Site Start Date1/1/90Pollutant ParametersOzone, PM10 FRM, PM10 FEM, PM2.5 FRM, PM2.5 Non-FEM, CO,NO2,SO2, ToxicsMeteorological ParametersWind speed, wind direction, outdoor temperature, barometric pressureAddress3425 N. First St., Fresno CA 93726LatitudeN 37o46’ 55”LongitudeN 119o46’ 23”Elevation (m)96LocationDistance to road75 mTraffic Count3000Ground CoverRoofSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-102011 Air Monitoring Network PlanFresno–First (1 of 3)Pollutant Ozone PM10 FRM PM10 FEM PM2.5 FRMParameter code 44201 81102 85101 88101Spatial scale Neighborhood Neighborhood Neighborhood NeighborhoodSite type Population High concentration High concentration HighconcentrationMonitor objective Max PrecursorEmissions ImpactMax PrecursorEmissions ImpactMax PrecursorEmissions ImpactPopulationExposureMonitor type SLAMS SLAMS SPM SLAMSPOC1 1 1 1Method code 087 063 063 118Sampling method (List Instrument) API/Teledyne 400Andersen SA1200 Met One 1020 R&P 2025Analysis method UV Gravimetric Beta Attenuation GravimetricStart dateOperation schedule (e.g. 1:1, 1:3) 1:1 1:6 1-Hour DailySampling season ALL YEAR ALL YEAR ALL YEAR ALL YEARProbe height (meters)Distance from supporting structure(meters)Distance from obstructions on roofDistance from obstructions not on roof(meters)None None None NoneDistance from trees (meters) None None None NoneDistance to furnace or incinerator flue(meters)None None None NoneDistance between collocated monitors(meters)-- -- -- --Unrestricted airflow (degrees) 360 360 360 360Probe material (Teflon, etc.) Teflon Teflon TeflonTeflonResidence time (seconds) 4.2 -- -- --Frequency of flow rate verification formanual PM samplers audit-- Once a Month -- Once a MonthFrequency of flow rate verification forautomated PM analyzers audit-- -- Twice a Month --Frequency of one-point QC check(gaseous)Twice a month -- -- --San Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-112011 Air Monitoring Network PlanFresno–First (1 of 3) continuedPollutant Ozone PM10 FRM PM10 FEM PM2.5Last Annual Performance Evaluation(gaseous)09/22/2010 -- -- --Last two semi-annual flow rate audits forPM monitors-- 09/20/2010 09/20/2010 06/08/2010Changes planned within the next 18months (Y/N))N NNNFresno–First (2 of 3)Pollutant PM2.5 PM2.5 Non-FEM CONO2Parameter code 88502 88501 42101 42602Spatial scale Neighborhood Neighborhood Neighborhood NeighborhoodSite type High concentration High concentration Population PopulationMonitor objective Population Exposure PopulationExposureMax PrecursorEmissions ImpactMax Precursor EmissionsImpactMonitor type Trend Improve SPM SLAMSSLAMSPOC5 1 3 1 1Method code 810 707 731 067 074Sampling method (List Instrument) R&P 2025 MetOne 1020 Dasibi 3008 API 200EAnalysis method Gravimetric Beta AttenuationStart dateOperation schedule (e.g. 1:3, 1-Hour,etc.)Daily 1-Hour 1:1 1:1Sampling season ALL YEAR All year All year All yearProbe height (meters)Distance from supporting structure(meters)Distance from obstructions on roofDistance from obstructions not on roof(meters)None None None NoneDistance from trees (meters) None None None NoneDistance to furnace or incinerator flue(meters)None None None NoneSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-122011 Air Monitoring Network PlanFresno–First (2 of 3) continuedPollutant PM2.5 PM2.5 Non-FEM CONO2Distance between collocated monitors(meters)-- 1.5 -- --Unrestricted airflow (degrees) 360 360 360 360Probe material (Teflon, etc.) Teflon Teflon TeflonTeflonResidence time (seconds) -- -- 6.2Frequency of flow rate verification formanual PM samplers auditOnce a Month -- -- --Frequency of flow rate verification forautomated PM analyzers audit-- Twice a month -- --Frequency of one-point QC check(gaseous)-- -- Twice a month Twice a monthLast Annual Performance Evaluation(gaseous)-- -- 09/22/2010 09/22/2010Last two semi-annual flow rate audits forPM monitors09/22/2010 09/22/2010 -- --Changes planned within the next 18months (Y/N)NNNNFresno–First (3 of 3)PollutantSO2Toxics Met ParametersParameter code 42401 Many ManySpatial scale Neighborhood Neighborhood RegionalSite typePopulation Population GeneralMonitor objective Other Unknown Research, timely/publicMonitor type SLAMS Many ManyPOC1 ManyManyMethod code 009 Many ManySampling method (List Instrument) - Xontech 924Analysis method TECO 43 --Start dateOperation schedule (e.g. 1:1, 1:3) 1:1 1:1 1:1Sampling season All year All year All yearProbe height (meters)San Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-132011 Air Monitoring Network PlanFresno–First (3 of 3) continuedPollutantSO2Toxics Met ParametersDistance from supporting structure (meters)Distance from obstructions on roofDistance from obstructions not on roof(meters)None NoneDistance from trees (meters) None None NoneDistance to furnace or incinerator flue(meters)None None NoneDistance between collocated monitors(meters)None -- --Unrestricted airflow (degrees) -- 360 360Probe material (Teflon, etc.) 360 Teflon TeflonResidence time (seconds) TeflonFrequency of flow rate verification for manualPM samplers audit5.9 -- --Frequency of flow rate verification forautomated PM analyzers audit-- -- --Frequency of one-point QC check (gaseous) -- Twicea month --Last Annual Performance Evaluation(gaseous)Twice a month 09/23/2010 --Last two semi-annual flow rate audits for PMmonitors09/22/2010 -- --Changes planned within the next 18 months(Y/N)NNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-142011 Air Monitoring Network PlanSite name Fresno–PacificAIRS #060195025CountyFresnoCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyAll data reported by Ventura County APCDSite Start Date1/1/00Pollutant ParametersPM2.5 FRMMeteorological ParametersNoneAddress1716 Winery, Fresno CA 93726Latitude36.72639Longitude-119.733Elevation (m)100LocationOn school roofDistance to road62.0 m (north), 52.0 m (east)Traffic Count2539Ground CoverRoof materialSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-152011 Air Monitoring Network PlanFresno–PacificPollutantPM2.5 FRMParameter code88101Spatial scaleNeighborhoodSite typePopulationMonitor objectiveStandards/strategy, research supportMonitor typeSLAMSPOC1Method code120Sampling method (List Instrument) Partisol 2025 installed on 9/15/2010Analysis methodGRAVI-METRICStart date1/1/2000Operation schedule (e.g. 1:1, 1:3) 1:3, 1:6Sampling seasonALL YEARProbe height (meters)8.0 mDistance from supporting structure (meters) 6.0 mDistance from obstructions on roof 54.5 mDistance from obstructions not on roof (meters) _____Distance from trees (meters) 76.0 mDistance to furnace or incinerator flue (meters) _____Distance between collocated monitors (meters) _____Unrestricted airflow (degrees) 360Probe material (Teflon, etc.) _____Residence time (seconds) _____Frequency of flow rate verification for manual PMsamplers auditMONTHLYFrequency of flow rate verification for automated PManalyzers audit_____Frequency of one-point QC check (gaseous) _____Last Annual Performance Evaluation (gaseous) _____Last two semi-annual flow rate audits for PM monitors 9/15/2010, 3/3/2011Changes planned within the next 18 months (Y/N) NSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-162011 Air Monitoring Network PlanSite name Fresno–Sky ParkAIRS #060190242CountyFresnoCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyAll data reported by SJVAPCDSite Start Date7/1/86Pollutant ParametersOzone, CO,NO2Meteorological ParametersWind speed, wind direction, outdoor temperatureAddress4508 Chennault Ave, Fresno CA 93722Latitude36.84056Longitude-119.874Elevation (m)65LocationPortable buildingDistance to road11.5 m (west)Traffic Count100Ground CoverGravelSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-172011 Air Monitoring Network PlanFresno–Sky ParkPollutantOzone CONO2Met ParametersParameter code 44201 42101 42602 ManySpatial scale Neighborhood NeighborhoodNeighborhood RegionalSite typePopulation, regionaltransportPopulation Population GeneralMonitor objective Timely/public,standards/strategy,research supportStandards/strategyStandards/strategyResearch,timely/publicMonitor typeSLAMS SLAMSSLAMSSLAMSPOC1 1 1 1Method code 087 054 074 ManySampling method (List Instrument) 400E 48 42C ITP-125-125 HV, OT06A-2, WD-020C, WS010BAnalysis method UV IR CL ____Start date7/1/1986 7/1/1986 7/1/1986 7/1/1986Operation schedule (e.g. 1:1, 1:3) 1:1 1:1 1:1 1:1Sampling season ALL YEAR ALL YEAR ALL YEAR ALL YEARProbe height (meters) 4 m 4 m 4 m 5 mDistance from supporting structure (meters) ____ ____ ____ ____Distance from obstructions on roof ____ ____ ____ ____Distance from obstructions not on roof (meters) 5 m/ 16 m 5 m / 16 m 5 m / 16 m 5 m / 16 mDistance from trees (meters) 27 m / 20 m 27 m / 20m 27 m / 20 m 27 m / 20 mDistance to furnace or incinerator flue (meters) ____ ____ ____ ____Distance between collocated monitors (meters) ________ ____ ____Unrestricted airflow (degrees) 280 280 280 280Probe material (Teflon, etc.) TEFLON TEFLON TEFLON____Residence time (seconds) 10.0 9.4 10.1 ____Frequency of flow rate verification for manualPM samplers audit____ ____ ____ ____Frequency of flow rate verification forautomated PM analyzers audit____ ____ ____ ____Frequency of one-point QC check (gaseous) 1:1 1:1 1:1 ____Last Annual Performance Evaluation (gaseous) 3/2/2011 3/2/2011 3/2/2011 3/2/2011San Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-182011 Air Monitoring Network PlanFresno–Sky Park (continued)PollutantOzone CONO2Met ParametersLast two semi-annual flow rate audits for PMmonitors____ ____ ____ ____Changes planned within the next 18 months(Y/N)NNNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-192011 Air Monitoring Network PlanSite name HuronAIRS #060192008CountyFresnoCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyAll data reported by SJVAPCDSite Start Date10/12/09Pollutant ParametersPM2.5 Non-FEMMeteorological ParametersBarometric pressureAddress16875 4thSt., Huron, CA 93234Latitude36.583Longitude-119.5Elevation (m)112LocationIn school roomDistance to road202 m (west), 99.5 m (north)Traffic Count1205Ground CoverPaved/vegetatedSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-202011 Air Monitoring Network PlanHuronPollutantPM2.5 Non-FEM Met ParametersParameter code64101Spatial scaleNeighborhood NeighborhoodSite typePopulation PopulationMonitor objectiveTimely/public Timely/publicMonitor typeSPM -POC3Method code731Sampling method (List Instrument) Anderson ITP-125-50-HV, BP-092Analysis methodBETA-ATTENUATIONStart dateQ3-2009 2/1/2010Operation schedule (e.g. 1:1, 1:3, 1-Hour) 1-Hour 1:1Sampling seasonALL YEAR ALL YEARProbe height (meters) 4.5 mDistance from supporting structure (meters) 1.5 mDistance from obstructions on roof _____Distance from obstructions not on roof (meters) _____Distance from trees (meters) 41.5 mDistance to furnace or incinerator flue (meters) _____Distance between collocated monitors (meters) _____Unrestricted airflow (degrees) 270Probe material (Teflon, etc.) ALUMINUMResidence time (seconds) _____Frequency of flow rate verification for manual PMsamplers audit_____Frequency of flow rate verification for automated PManalyzers auditBI-WEEKLYFrequency of one-point QC check (gaseous) _____Last Annual Performance Evaluation (gaseous) _____Last two semi-annual flow rate audits for PMmonitors7/21/2010, 11/2/2010Changes planned within the next 18 months (Y/N) NNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-212011 Air Monitoring Network PlanSite name ParlierAIRS #060194001CountyFresnoCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyAll data reported by SJVAPCDSite Start Date1/1/06Pollutant ParametersOzone,NO2, TOTAL NMOC, NMHC (PAMS)Meteorological ParametersWind speed, wind direction, outdoor temperature, relative humidity, barometric pressure, solar radiationAddress9240 S. Riverbend Ave., Parlier CA 93648Latitude36.59722Longitude-119.504Elevation (m)78LocationPortable building in university fieldDistance to road500 m+ (north)Traffic Count8700Ground CoverDirt/vegetatedSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-222011 Air Monitoring Network PlanParlier (1 of 2)PollutantOzoneNO2Total NMOC NMHC (PAMS)Parameter code 44201 42602 43102 ManySpatial scaleNeighborhood Neighborhood Neighborhood NeighborhoodSite typeHigh concentration,regional transportPopulation Population PopulationMonitor objective Timely/public,standards/strategy,research supportStandards/strategy,researchResearch ResearchMonitor typePAMS PAMSPAMSPAMSPOC1 1 1 1Method code087 074 164 177Sampling method (List Instrument) 400 E 42C / 200Einstalledon 1/10/2011910A 55CAnalysis method UV CL GC GCStart date3/1/1983 3/1/1983 3/1/1983 3/1/83Operation schedule (e.g. 1:1, 1:3) 1:1 1:1 1:3 1:1Sampling season ALL YEAR ALL YEAR JUN-JUL-AUG ALL YEARProbe height (meters) 9.0 m 9.0 m 7.0 m 9.0 mDistance from supporting structure (meters) ____________ ______ ______Distance from obstructions on roof ______ ______ ______ ______Distance from obstructions not on roof (meters) ______ ______ ______ ______Distance from trees (meters) ______ ______ 12.5 m ______Distance to furnace or incinerator flue (meters) ______ ______ ______ ______Distance between collocated monitors (meters) ______ ______ ______ ______Unrestricted airflow (degrees) 360 360 270 360Probe material (Teflon, etc.) TEFLON TEFLON S. STEEL TEFLONResidence time (seconds) 13.6 13.3 ______ 12.9Frequency of flow rate verification for manual PMsamplers audit______ ______ ______ ______Frequency of flow rate verification for automatedPM analyzers audit______ ______ ______ ______Frequency of one-point QC check (gaseous) 1:1 1:1 ______ 1:1Last Annual Performance Evaluation (gaseous) 1/27/2011 1/27/2011 5/10/2010 ______San Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-232011 Air Monitoring Network PlanParlier (1 of 2) continuedPollutantOzoneNO2TOTAL NMOC NMHC (PAMS)Last 2 semi-annual flow rate audits, PM monitors ______ ______ ______ ______Changes planned within the next 18 months (Y/N) NNNYParlier (2 of 2)PollutantMet ParametersParameter codeManySpatial scaleRegionalSite typeGeneralMonitor objectiveResearch, timely/publicMonitor typePAMSPOC1Method codeManySampling method (List Instrument) ITP-125-125 HV,OT-06A-2, BP-092, RH-HMP45D, SRD-Mod.8-48, WD020C, WS-010CAnalysis method______Start date3/1/83Operation schedule (e.g. 1:1, 1:3) 1:1Sampling seasonALL YEARProbe height (meters)9.5 mDistance from supporting structure (meters) ______Distance from obstructions on roof ______Distance from obstructions not on roof (meters) ______Distance from trees (meters) ______Distance to furnace or incinerator flue (meters) ______Distance between collocated monitors (meters) ______Unrestricted airflow (degrees) 360Probe material (Teflon, etc.) ______Residence time (seconds) ______Frequency of flow rate verification for manual PMsamplers audit______Frequency of flow rate verification for automated PManalyzers audit______San Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-242011 Air Monitoring Network PlanParlier (2 of 2)PollutantMet ParametersFrequency of one-point QC check (gaseous) ______Last Annual Performance Evaluation (gaseous) ______Last two semi-annual flow rate audits for PM monitors ______Changes planned within the next 18 months (Y/N) NSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-252011 Air Monitoring Network PlanSite name TranquillityAIRS #060192009CountyFresnoCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyAll data reported by SJVAPCDSite Start Date11/9/2009Pollutant ParametersOzone, PM2.5 FEMMeteorological ParametersWind speed, wind direction, outdoor temperature, barometric pressureAddress32650 W. Adams, Tranquillity CA 93668Latitude36.600833Longitude-120.382222Elevation (m)59LocationPortable shedDistance to road186 m (south)Traffic CountGround CoverGravel/vegetationSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Fresno County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-262011 Air Monitoring Network PlanTranquillityPollutantOzone PM2.5 FEM Met ParametersParameter code44201 ManySpatial scaleUrban Urban UrbanSite typePopulation Population PopulationMonitor objectiveTimely/public Timely/public Timely/publicMonitor typeSPM SPMPOC1 3Method code087 170 ManySampling method (List Instrument) 400 E 1020 ITP-020B, OT-060,BP-092C, WD-020C,WS-010CAnalysis methodUV BETA-ATTENUATIONStart date10/30/2009 10/30/2009 10/30/2009Operation schedule (e.g. 1:1, 1:3) 1:1 1-Hour 1:1Sampling seasonALL YEAR ALL YEAR ALL YEARProbe height (meters)4 m 4 m 10 m / 20 mDistance from supporting structure (meters)Distance from obstructions on roofDistance from obstructions not on roof (meters)Distance from trees (meters) 102 m 102 m 102 mDistance to furnace or incinerator flue (meters) 97.5 m 97.5 m 97.5 mDistance between collocated monitors (meters)Unrestricted airflow (degrees) 360 360 360Probe material (Teflon, etc.) TEFLON ALUMINUMResidence time (seconds) 6.0Frequency of flow rate verification for manual PM samplersauditFrequency of flow rate verification for automated PManalyzers auditBI-WEEKLYFrequency of one-point QC check (gaseous) 1:1Last Annual Performance Evaluation (gaseous) 7/21/2010Last 2 semi-annual flow rate audits, PM monitors 10/21/2010, 4/12/2011Changes planned within the next 18 months (Y/N) NSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-272011 Air Monitoring Network PlanSite name Arvin–Di GiorgioAIRS #060295002CountyKernCollecting (Operating)AgencyAll equipment operated by CARBReporting AgencyAll data reported by CARBSite Start DatePollutant ParametersOzoneMeteorological ParametersOutdoor temperatureAddress19405 Buena Vista Blvd, Arvin CA 93203LatitudeN 35o14’ 21”LongitudeN 118o47’ 18.6”Elevation (m)107LocationDistance to road10 mTraffic Count500Ground CoverDirtSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-282011 Air Monitoring Network PlanArvin–Di GiorgioPollutantOzone Met ParametersParameter code 44201 62101Spatial scaleNeighborhood RegionalSite typeMonitor objective Population Exposure Research, timely/publicMonitor typePAMS PAMSPOC1 1Method code 087 040Sampling method (List Instrument) 400 EAnalysis method UVStart date6/1/1989Operation schedule (e.g. 1:1, 1:3) 1:1 1:1Sampling season ALL YEAR ALL YEARProbe height (meters)Distance from supporting structure (meters) ____ ____Distance from obstructions on roof ____ ____Distance from obstructions not on roof (meters) ____ ____Distance from trees (meters)Distance to furnace or incinerator flue (meters) ____ ____Distance between collocated monitors (meters) ________Unrestricted airflow (degrees) 350Probe material (Teflon, etc.) TEFLON ____Residence time (seconds) 10.7 ____Frequency of flow rate verification for manualPM samplers audit____ ____Frequency of flow rate verification for automatedPM analyzers audit____ ____Frequency of one-point QC check (gaseous) 1:1 ____Last Annual Performance Evaluation (gaseous) 07/27/2010 ____Last two semi-annual flow rate audits for PMmonitors____Changes planned within the next 18 months(Y/N)Y YSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-292011 Air Monitoring Network PlanSite name Bakersfield–PlanzAIRS #060290016CountyKernCollecting (Operating)AgencyAll equipment operated by CARBReporting AgencyAll data reported by Ventura County APCDSite Start Date9/19/00Pollutant ParametersPM2.5 FRMMeteorological ParametersNoneAddress401 E. Planz Rd., Bakersfield CA 93307LatitudeN 36o19’ 52”LongitudeW 118o59’ 59”Elevation (m)145LocationDistance to road500 mTraffic Count1000Ground CoverAsphaltSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-302011 Air Monitoring Network PlanBakersfield–PlanzPollutantPM2.5 FRMParameter code88101Spatial scaleNeighborhoodSite typePopulationMonitor objectivePopulation ExposureMonitor typeSLAMSPOC1Method code120Sampling method (List Instrument) R&P 2025Analysis methodGravimetricStart dateOperation schedule (e.g. 1:1, 1:3) 1:3Sampling seasonAll yearProbe height (meters)Distance from supporting structure (meters)Distance from obstructions on roofDistance from obstructions not on roof (meters) NoneDistance from trees (meters) NoneDistance to furnace or incinerator flue (meters) NoneDistance between collocated monitors (meters) NoneUnrestricted airflow (degrees) 360Probe material (Teflon, etc.) TeflonResidence time (seconds) NAFrequency of flow rate verification for manual PMsamplers auditOnce a monthFrequency of flow rate verification for automated PManalyzers audit--Frequency of one-point QC check (gaseous) --Last Annual Performance Evaluation (gaseous) --Last two semi-annual flow rate audits for PM monitors 01/29/2010Changes planned within the next 18 months (Y/N) YSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-312011 Air Monitoring Network PlanSite name Bakersfield–CaliforniaAIRS #060290014CountyKernCollecting (Operating)AgencyEquipment operated by CARB: PM10 FRM,PM2.5 FRM, PM2.5 FEM,NO2, Toxics,MeteorologyEquipment operated by SJVAPCD: TemporaryPM10 FEMReporting AgencyData reported by CARB: PM10FRM, PM2.5 FRM,NO2, Toxics,MeteorologyData reported by Ventura CountyAPCD: PM2.5 FEMData reported by SJVAPCD:Temporary PM10 FEMSite Start Date3/1/94Pollutant ParametersOzone, PM10 FRM, PM10 FEM (temporarily operated bythe SJVAPCD), PM2.5 FRM, PM2.5 FEM,NO2,ToxicsMeteorological ParametersWind speed, wind direction, outdoor temperature, barometric pressureAddress5558 California Ave., Bakersfield CA 93309LatitudeN 35o21’ 24”LongitudeN 119o3’ 46”Elevation (m)117LocationDistance to road300 mTraffic Count10000Ground CoverRoofSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-322011 Air Monitoring Network PlanBakersfield–California (1 of 2)PollutantOzone PM10 FRM PM10 FEM+PM2.5SN20021399PM2.5SN20021078Parameter code 44201 81102 85101Spatial scale Neighborhood Neighborhood Neighborhood NeighborhoodSite typePopulation Population Population PopulationMonitor objective Unknown Unknown PopulationExposureOtherMonitor type SLAMS SLAMS SLAMS Non-regulatory Non-regulatoryPOC1 1 1 3 4Method code 087 063 063 731 731Sampling method (List Instrument) API/Teledyne400Sierra Anderson1200R&P 2025 R&P 2025Analysis method UV Gravimetric Gravimetric GravimetricStart dateOperation schedule (e.g. 1:6, Daily, etc.) 1:1 1:61-Hour 1-Hour 1-HourSampling seasonProbe height (meters)Distance from supporting structure (meters)Distance from obstructions on roofDistance from obstructions not on roof(meters)None None None NoneDistance from trees (meters) None None None NoneDistance to furnace or incinerator flue(meters)None None None NoneDistance between collocated monitors(meters)-- 3.0 3.0 3.0Unrestricted airflow (degrees) 360 360 360 360Probe material (Teflon, etc.) Teflon Teflon TeflonTeflonResidence time (seconds) 10.0 -- -- --Frequency of flow rate verification for manualPM samplers audit-- Once per month Once permonthOnce per monthFrequency of flow rate verification forautomated PM analyzers audit-- -- -- --Frequency of one-point QC check (gaseous) Twice permonth -- -- --Last Annual Performance Evaluation(gaseous)10/28/2009 -- -- --San Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-332011 Air Monitoring Network PlanBakersfield–California (1 of 2) continuedPollutantOzone PM10 FRM PM10 FEM+PM2.5SN20021399PM2.5SN20021078Last two semi-annual flow rate audits for PMmonitors-- 01/29/2010 01/29/2010 01/29/2010Changes planned within the next 18 months(Y/N)NNYNN+-- Temporary monitorBakersfield–California (2 of 2)Pollutant PM2.5 FEMNO2Toxics Met ParametersParameter code 88101 42602 Many ManySpatial scale Neighborhood Neighborhood Neighborhood RegionalSite type Population Population Population GeneralMonitor objective Timely/public Unknown Unknown Research, Timely/publicMonitor type SLAMS SLAMS ManyManyPOC1,2 1 ManyManyMethod code 120 074 ManyManySampling method (List Instrument) Met One 1020 API200A Xontech 924Analysis method Beta Attenuation CLStart dateOperation schedule (e.g. 1:1, 1-Hour) 1-Hour 1:1 1:1 1:1Sampling seasonProbe height (meters)Distance from supporting structure(meters)Distance from obstructions on roofDistance from obstructions not on roof(meters)None None None NoneDistance from trees (meters) None None None NoneDistance to furnace or incinerator flue(meters)None None None NoneDistance between collocated monitors(meters)3.0 -- --Unrestricted airflow (degrees) 360 360 360 360San Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-342011 Air Monitoring Network PlanBakersfield–California (2 of 2) continuedPollutant PM2.5 FEMNO2Toxics Met ParametersProbe material (Teflon, etc.) Teflon Teflon TeflonTeflonResidence time (seconds) -- 15.2 -- --Frequency of flow rate verification formanual PM samplers audit-- -- -- --Frequency of flow rate verification forautomated PM analyzers auditTwice per month -- -- --Frequency of one-point QC check(gaseous)-- Twice per month Twice per month --Last Annual Performance Evaluation(gaseous)-- 11/04/2009 12/13/2009 --Last two semi-annual flow rate audits forPM monitors01/29/2010 -- -- --Changes planned within the next 18months (Y/N)NNNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-352011 Air Monitoring Network PlanSite name EdisonAIRS #060290007CountyKernCollecting (Operating)AgencyAll equipment operated by CARBReporting AgencyAll data reported by CARBSite Start Date1/1/80Pollutant ParametersOzone,NO2Meteorological ParametersWind speed, wind direction, outdoor temperatureAddressJohnson Farm-Shed Rd, Edison CA 93320LatitudeN 35o20’ 45”LongitudeN 119o51’ 6”Elevation (m)172LocationDistance to road450Traffic Count50000Ground CoverDirtSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-362011 Air Monitoring Network PlanEdisonPollutantOzoneNO2Met ParametersParameter code44201 42602 ManySpatial scaleNeighborhood Neighborhood RegionalSite typeHigh concentration,regional transportPopulation GeneralMonitor objectiveUnknown Unknown Research, timely/publicMonitor typeSLAMS SLAMS OtherPOC1 1 1Method code087 074 ManySampling method (List Instrument) API/Teledyne 400API 200 AAnalysis methodUV CLStart dateOperation schedule (e.g. 1:1, 1:3) 1:1 1:1 1:1Sampling seasonAll year All year All yearProbe height (meters)Distance from supporting structure (meters)Distance from obstructions on roofDistance from obstructions not on roof (meters) None None NoneDistance from trees (meters) 18.5 18.5 18.5Distance to furnace or incinerator flue (meters) None None NoneDistance between collocated monitors (meters) -- ----Unrestricted airflow (degrees) 360 360 360Probe material (Teflon, etc.) Teflon Teflon TeflonResidence time (seconds) 15 13.6 --Frequency of flow rate verification for manual PMsamplers audit-- -- --Frequency of flow rate verification for automated PManalyzers audit-- -- --Frequency of one-point QC check (gaseous) Twice a month Twice a month --Last Annual Performance Evaluation (gaseous) 02/02/2010 02/08/2010 --Last two semi-annual flow rate audits for PM monitors -- -- --Changes planned within the next 18 months (Y/N) NNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-372011 Air Monitoring Network PlanSite name LebecAIRS #060292009CountyKernCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyAll data reported by SJVAPCDSite Start Date1/20/2009Pollutant ParametersPM2.5 Non-FEMMeteorological ParametersWind speed, wind direction, outdoor temperature, barometric pressureAddressBeartrap Road (no #), Lebec, CA 91350Latitude34.8415Longitude-118.861Elevation (m)1063LocationDistance to roadTraffic Count67000Ground CoverDirt, vegetatedSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-382011 Air Monitoring Network PlanLebecPollutantPM2.5 Non-FEM Met ParametersParameter code88502 ManySpatial scaleNeighborhood RegionalSite typePopulation GeneralMonitor objectiveTimely/public Research, timely/publicMonitor typeSPM SPMPOC3 ManyMethod code731 ManySampling method (List Instrument) BAM 1020Analysis methodBETAATTENUATIONITP-125-5O HV, OT-060A-2, BP-092, WD-020C, WS-010CStart dateOperation schedule (e.g. 1:1, 1-Hour) 1-Hour 1:1Sampling seasonALL YEAR ALL YEARProbe height (meters) 5.5 m 9.6 mDistance from supporting structure (meters)Distance from obstructions on roofDistance from obstructions not on roof (meters)Distance from trees (meters)Distance to furnace or incinerator flue (meters)Distance between collocated monitors (meters)Unrestricted airflow (degrees) 360 360Probe material (Teflon, etc.) ALUMINUMResidence time (seconds)Frequency of flow rate verification for manual PMsamplers auditFrequency of flow rate verification for automated PManalyzers auditBI-WEEKLYFrequency of one-point QC check (gaseous)Last Annual Performance Evaluation (gaseous)Last two semi-annual flow rate audits for PM monitors7/20/2010, 12/14/2010Changes planned within the next 18 months (Y/N)) NSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-392011 Air Monitoring Network PlanSite name MaricopaAIRS #060290008CountyKernCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyAll data reported by SJVAPCDSite Start Date7/1/87Pollutant ParametersOzoneMeteorological ParametersWind speed, wind direction, outdoor temperature, barometric pressureAddress755 Stanislaus St., Maricopa CA 93352Latitude35.05139Longitude-119.403Elevation (m)289LocationIn old school buildingDistance to road500 m + (north)Traffic Count0Ground CoverGravelSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-402011 Air Monitoring Network PlanMaricopaPollutantOzone Met ParametersParameter code 44201 ManySpatial scaleNeighborhood RegionalSite typeRegional transport GeneralMonitor objective Timely/public,standards/strategy,research supportResearch, timely/publicMonitor typeSLAMS SLAMSPOC1 1Method code087 ManySampling method (List Instrument) 400 E ITP-125-50HV, OT-06A-2, BP-092, WD-020C, WS-010CAnalysis method UV _____Start date7/1/1987 7/1/1987Operation schedule (e.g. 1:1, 1:3) 1:1 1:1Sampling season ALL YEAR ALL YEARProbe height (meters) 5 m 5 mDistance from supporting structure (meters) _____ 2.7 m (OT)Distance from obstructions on roof _____ 5 m (BP)1.5 m (OT)Distance from obstructions not on roof (meters) _____ _____Distance from trees (meters) _____ _____Distance to furnace or incinerator flue (meters) _____ _____Distance between collocated monitors (meters) __________Unrestricted airflow (degrees) 360 360 (WD,WS, BP),270 (OT)Probe material (Teflon, etc.) TEFLON _____Residence time (seconds) 5.9 _____Frequency of flow rate verification for manual PMsamplers audit_____ _____Frequency of flow rate verification for automatedPM analyzers audit_____ _____Frequency of one-point QC check (gaseous) 1:1 _____Last Annual Performance Evaluation (gaseous) 10/11/2010 _____Last two semi-annual flow rate audits for PMmonitors_____ _____Changes planned within the next 18 months (Y/N) NNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-412011 Air Monitoring Network PlanSite name OildaleAIRS #060290232CountyKernCollecting (Operating)AgencyAll equipment operated by CARBReporting AgencyAll data reported by CARBSite Start Date1/1/80Pollutant ParametersOzone, PM10 FRMMeteorological ParametersNoneAddress3311 Manor St, Oildale CA 93308LatitudeN 35o28’17”LongitudeN 119o1’ 0”Elevation (m)183LocationDistance to road150 mTraffic Count10000Ground CoverDirtSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-422011 Air Monitoring Network PlanOildalePollutantOzone PM10 FRMParameter code44201 81102Spatial scaleNeighborhood NeighborhoodSite typeRegional transport PopulationMonitor objectiveHighest Concentration UnknownMonitor typeSLAMS SLAMSPOC1 2Method code087 063Sampling method (List Instrument)API/Teledyne 400 Sierra Anderson 1200Analysis methodUV GravimetricStart dateOperation schedule (e.g. 1:1, 1:3)1:1 1:6Sampling seasonAll year All yearProbe height (meters)Distance from supporting structure (meters)Distance from obstructions on roofDistance from obstructions not on roof (meters) None NoneDistance from trees (meters)None NoneDistance to furnace or incinerator flue (meters) None NoneDistance between collocated monitors (meters) -- --Unrestricted airflow (degrees)360 360Probe material (Teflon, etc.)Teflon TeflonResidence time (seconds)9.0 --Frequency of flow rate verification for manual PM samplers audit -- Once a monthFrequency of flow rate verification for automated PM analyzers audit -- --Frequency of one-point QC check (gaseous) Twice a month --Last Annual Performance Evaluation (gaseous) 01/06/2009Last two semi-annual flow rate audits for PM monitors -- 10/19/2009Changes planned within the next 18 months (Y/N) NNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-432011 Air Monitoring Network PlanSite name ShafterAIRS #060296001CountyKernCollecting (Operating)AgencyEquipment operated by CARB: Ozone,NO2Equipment operated by SJVAPCD:Meteorology, NMOC, NMHC (PAMS)Reporting AgencyData reported by CARB: Ozone,NO2Data reported by SJVAPCD: NMOC, NMHC(PAMS), MeteorologySite Start Date1/1/89Pollutant ParametersOzone,NO2, TOTAL NMOC, NMHC (PAMS)Meteorological ParametersWind speed, wind direction, outdoor temperature, solar radiationAddress578 Walker St, Shafter CA 93263LatitudeN 35o30’13”LongitudeN 119o16’ 21”Elevation (m)126LocationDMV buildingDistance to road10 mTraffic CountGround CoverAsphaltSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-442011 Air Monitoring Network PlanShafter (1 of 2)PollutantOzoneNO2Total NMOC NMHC (PAMS)Parameter code 44201 42602 43102 ManySpatial scaleNeighborhood Neighborhood Neighborhood NeighborhoodSite typeGeneral/background Population Population PopulationMonitor objective Population Exposure Population Exposure Research ResearchMonitor typePAMSPAMSPAMS PAMSPOC1 1 1 1Method code087 074 164 177Sampling method (List Instrument) 400E (ARB) TECO 42, 42C, 42i 910A 55 samplerAnalysis method UV CL GC GCStart date1/1/1989 1/1/1989 7/1/1994 7/1/1994Operation schedule (e.g. 1:1, 1:3) 1:1 1:1 1:3 1:1Sampling season ALL YEAR ALL YEAR JUN-JUL-AUG ALL YEARProbe height (meters) 10.0 m 10.0 m 7.0 m 7.0 mDistance from supporting structure (meters) ____ ____ ____ ____Distance from obstructions on roof ____ ____ ____ ____Distance from obstructions not on roof (meters) ____ ____ ____ ____Distance from trees (meters) ____ ____ ____ ____Distance to furnace or incinerator flue (meters) ____ ____ 10.5 m 11.0 mDistance between collocated monitors (meters) ________ ____ ____Unrestricted airflow (degrees) 360 360 360 360Probe material (Teflon, etc.) TEFLON TEFLON S. STEEL TEFLONResidence time (seconds) 10.1 8.9 ____ 9.6Frequency of flow rate verification for manual PMsamplers audit____ ____ ____ ____Frequency of flow rate verification for automatedPM analyzers audit____ ____ ____ ____Frequency of one-point QC check (gaseous) 1:1 1:1 ____ 1:1Last Annual Performance Evaluation (gaseous) 05/26/201010/13/201005/26/201010/13/20105/10/2010 ____Last two semi-annual flow rate audits for PMmonitors____ ____ ____ ____Changes planned within the next 18 months (Y/N) NNNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kern County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-452011 Air Monitoring Network PlanShafter (2 of 2)PollutantMet ParametersParameter codeManySpatial scaleRegionalSite typeGeneralMonitor objectiveResearch, timely/publicMonitor typeOtherPOC1Method codeManySampling method (List Instrument) ITP-BA512AABB, OT-06A-2, SRD-Mod. 8-48, WD-020B, WS-010C, BP-092Analysis method____Start date1/1/1989Operation schedule (e.g. 1:1, 1:3) 1:1Sampling seasonALL YEARProbe height (meters)10.0 mDistance from supporting structure (meters) ____Distance from obstructions on roof ____Distance from obstructions not on roof (meters) ____Distance from trees (meters) ____Distance to furnace or incinerator flue (meters) ____Distance between collocated monitors (meters) ____Unrestricted airflow (degrees) 360Probe material (Teflon, etc.) ____Residence time (seconds) ____Frequency of flow rate verification for manual PMsamplers audit____Frequency of flow rate verification for automated PManalyzers audit____Frequency of one-point QC check (gaseous) ____Last Annual Performance Evaluation (gaseous) ____Last two semi-annual flow rate audits for PM monitors ____Changes planned within the next 18 months (Y/N) NSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kings County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-462011 Air Monitoring Network PlanSite name Corcoran–PattersonAIRS #060310004CountyKingsCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyData reported by SJVAPCD:PM2.5 FEM, PM10 TEOM,MeteorologyData reported by CARB: PM10FRMData reported by Ventura CountyAPCD: PM2.5 FRMSite Start Date10/1/96Pollutant ParametersPM10 FRM, PM10 TEOM, PM2.5 FRM, PM2.5 FEMMeteorological ParametersWind speed, wind direction, outdoor temperature, barometric pressureAddress1520 Patterson Ave, Corcoran CA 93212Latitude36.10222Longitude-119.566Elevation (m)62LocationPortable buildingDistance to road35.0 (east), 38.5 (south)Traffic Count1035Ground CoverGravelSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kings County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-472011 Air Monitoring Network PlanCorcoran–Patterson (1 of 2)PollutantPM10 FRM PM10 TEOM PM2.5 FRMParameter code 81102 81102 88101Spatial scale Neighborhood Neighborhood NeighborhoodSite typeHigh concentration High concentration High concentrationMonitor objective Standards/strategy, research support Timely/public Standards/strategy,research supportMonitor type SLAMS Other SLAMS SLAMS SLAMSPOC1(Primary)3(Collocated)4(Alternate)7 1Method code 063079 120Sampling method (List Instrument) Sierra Andersen 1020 Andersen 300 - unit out ofservice December 2010Analysis method Gravimetric Tapered Element GravimetricStart date10/1/1996 8/8/2005 10/1/1996Operation schedule (e.g. 1:1, 1:3, 1-Hour) 1:31-Hour 1:3, 1:6Sampling season ALL YEAR ALL YEAR ALL YEARProbe height (meters) 6 m6 m 6 mDistance from supporting structure (meters) ____________ ______Distance from obstructions on roof ______ ______ ______Distance from obstructions not on roof(meters)______ ______ ______Distance from trees (meters) 59.5 m (1), 65.0 m (2)50.0 m (3) 48.0 m 50.0 mDistance to furnace or incinerator flue (meters) ______ ______ ______Distance between collocated monitors(meters)2.0 m (1 to 2) 2 m (1 to 3) 1.3 m 1.2 mUnrestricted airflow (degrees) 360360 360Probe material (Teflon, etc.) ______ TEFLON ______Residence time (seconds) ______ ______ ______Frequency of flow rate verification for manualPM samplers auditQUARTERLY ______ MONTHLYFrequency of flow rate verification forautomated PM analyzers audit______ BI-WEEKLY ______Frequency of one-point QC check (gaseous) ______ ______ ______Last Annual Performance Evaluation(gaseous)______ ______ ______San Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kings County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-482011 Air Monitoring Network PlanCorcoran–Patterson (1 of 2) continuedPollutantPM10 FRM PM10 TEOM PM2.5 FRMLast two semi-annual flow rate audits for PMmonitors9/8/2010, 1/26/11 8/4/10, 1/26/2011 3/23/2010, 9/30/2010Changes planned within the next 18 months(Y/N)NNN* AIRS codes for collocated pair: 06-031-0004-88102-Primary and 06-031-0003-88102-Collocated.Corcoran–Patterson (2 of 2)PollutantPM2.5 FEM Met ParametersParameter code88101 ManySpatial scaleNeighborhood RegionalSite typeHigh concentration GeneralMonitor objectiveTimely/public Research, timely/publicMonitor typeSPM ManyPOC3 ManyMethod code170 ManySampling method (List Instrument) 1020 ITP - 110-50HV, OT-06A-2, BP-090D,WD-020C, WS-010BAnalysis methodBETA-ATTENUATION ______Start date4/13/2002 10/1/1996Operation schedule (e.g. 1:1, 1-Hour) 1-Hour 1:1Sampling seasonALL YEAR ALL YEARProbe height (meters) 6 m 9.6 mDistance from supporting structure (meters) ____________Distance from obstructions on roof ______ ______Distance from obstructions not on roof (meters) ______ ______Distance from trees (meters) 50.0 m 51.5 mDistance to furnace or incinerator flue (meters) ______ ______Distance between collocated monitors (meters) 1.2 m______Unrestricted airflow (degrees) 360 360Probe material (Teflon, etc.) ALUMINUM ______Residence time (seconds) ______ ______Frequency of flow rate verification for manual PMsamplers audit______ ______San Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kings County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-492011 Air Monitoring Network PlanCorcoran–Patterson (2 of 2) continuedPollutantPM2.5 FEM Met ParametersFrequency of flow rate verification for automated PManalyzers auditBI-WEEKLY ______Frequency of one-point QC check (gaseous) ______ ______Last Annual Performance Evaluation (gaseous) ____________Last two semi-annual flow rate audits for PM monitors 6/10/2010, 1/26/2011 ______Changes planned within the next 18 months (Y/N) NNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kings County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-502011 Air Monitoring Network PlanSite name Hanford–IrwinAIRS #060311004CountyKingsCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyData reported by SJVAPCD: Ozone, PM10 TEOM,PM2.5 FEM,NO2, MeteorologyData reported by CARB: PM10 FRMSite Start Date10/11/93Pollutant ParametersOzone, PM10 FRM, PM10 TEOM, PM2.5 FEM,NO2Meteorological ParametersWind speed, wind direction, outdoor temperature, barometric pressureAddress807 S Irwin St, Hanford CA 93230Latitude36.31472Longitude-119.644Elevation (m)82LocationSchool roofDistance to road158 m (south)Traffic Count3383Ground CoverVegetation/roof materialSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kings County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-512011 Air Monitoring Network PlanHanford–Irwin (1 of 2)PollutantOzone PM10 FRM PM10 TEOMParameter code 44201 81102 85101Spatial scaleNeighborhood Neighborhood NeighborhoodSite typePopulation Population PopulationMonitor objective Timely/public,standards/strategy,research supportStandards/strategy,research supportTimely/publicMonitor typeSLAMS SLAMS SPMPOC1 1 3Method code087 063 079Sampling method (List Instrument) 400 E Sierra AndersenAnalysis method UV Gravimetric TEOMStart date2/25/2010 10/11/1993 7/14/2010Operation schedule (e.g. 1:1, 1:3, 1-Hour) 1:1 1:61-HourSampling season ALL YEAR ALL YEAR ALL YEARProbe height (meters) 5.5 m 5.5 m 5.5 mDistance from supporting structure (meters)Distance from obstructions on roofDistance from obstructions not on roof (meters)Distance from trees (meters)Distance to furnace or incinerator flue (meters)Distance between collocated monitors (meters)Unrestricted airflow (degrees) 360 360Probe material (Teflon, etc.) TEFLONResidence time (seconds) 12.7Frequency of flow rate verification for manual PMsamplers auditQUARTERLYFrequency of flow rate verification for automatedPM analyzers auditFrequency of one-point QC check (gaseous) 1:1Last Annual Performance Evaluation (gaseous) 1/24/2011Last two semi-annual flow rate audits for PMmonitors9/21/2010, 1/24/2011Changes planned within the next 18 months(Y/N)NNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kings County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-522011 Air Monitoring Network PlanHanford–Irwin (2 of 2)PollutantPM2.5 FEMNO2Met ParametersParameter code88101 42602 ManySpatial scaleNeighborhood Neighborhood NeighborhoodSite typePopulation Population PopulationMonitor objective Timely/public Timely/public,standards/strategy,research supportTimely/public,standards/strategy,research supportMonitor typeSPM SLAMS ManyPOC3 1 ManyMethod code170 074 ManySampling method (List Instrument) BAM 1020 API 200E ITP-110-50HV, OT-06A-2,BP-092, WD-020C, WS010CAnalysis method BETA CLStart date2/25/2010 2/25/2010 2/25/2010Operation schedule (e.g. 1:1, 1-Hour) 1-Hour 1:1 1:1Sampling season ALL YEAR ALL YEAR ALL YEARProbe height (meters) 5.5 m 5.5 m 9.6 mDistance from supporting structure (meters)Distance from obstructions on roofDistance from obstructions not on roof (meters)Distance from trees (meters)Distance to furnace or incinerator flue (meters)Distance between collocated monitors (meters)Unrestricted airflow (degrees) 360 360 360Probe material (Teflon, etc.) ALUMINUM TEFLONResidence time (seconds) 14.8Frequency of flow rate verification for manual PMsamplers auditFrequency of flow rate verification for automatedPM analyzers auditBI-WEEKLYFrequency of one-point QC check (gaseous) 1:1Last Annual Performance Evaluation (gaseous) 1/24/2011Last two semi-annual flow rate audits for PMmonitors6/9/2010, 1/24/2011Changes planned within the next 18 months (Y/N) N NNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kings County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-532011 Air Monitoring Network PlanSite name Santa Rosa Rancheria▲AIRS #060310500CountyKingsCollecting (Operating)AgencyAll equipment operated by Tachi-Yokut TribeReporting AgencyAll data reported by Tachi-Yokut TribeSite Start DatePollutant ParametersOzone, PM10 FRMMeteorological ParametersUnknownAddress17225 Jersey Ave., Lemoore, CA 93245Latitude36.233318Longitude-119.765251Elevation (m)68LocationDistance to roadTraffic CountGround Cover▲Tribal MonitorSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Kings County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-542011 Air Monitoring Network PlanSanta Rosa Rancheria▲PollutantOzone PM10 FRM Met ParametersParameter code44201 81102 UnknownSpatial scaleSite typeMonitor objectiveMonitor typeTribal MonitorPOC1 1 1Method code087 063 ManySampling method (List Instrument)Analysis methodStart dateOperation schedule (e.g. 1:1, 1:3) 1-Hour 1:6Sampling seasonProbe height (meters)Distance from supporting structure (meters)Distance from obstructions on roofDistance from obstructions not on roof (meters)Distance from trees (meters)Distance to furnace or incinerator flue (meters)Distance between collocated monitors (meters)Unrestricted airflow (degrees)Probe material (Teflon, etc.)Residence time (seconds)Frequency of flow rate verification for manual PMsamplers auditFrequency of flow rate verification for automated PManalyzers auditFrequency of one-point QC check (gaseous)Last Annual Performance Evaluation (gaseous)Last two semi-annual flow rate audits for PM monitorsChanges planned within the next 18 months (Y/N) Unknown UnknownUnknown▲Tribal MonitorSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Madera County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-552011 Air Monitoring Network PlanSite name Madera–CityAIRS #060392010CountyMaderaCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyAll data reported by SJVAPCDSite Start Date6/1/2010Pollutant ParametersOzone, PM10 TEOM, PM2.5 FEMMeteorological ParametersWind speed, wind direction, outdoor temperature, relative humidity, barometric pressure, solar radiationAddress28261 Avenue 14, Madera CA 93638Latitude36.953282Longitude-120.03421Elevation (m)84LocationPortable buildingDistance to road686 mTraffic CountGround CoverAsphaltSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Madera County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-562011 Air Monitoring Network PlanMadera—CityPollutantOzone PM10 TEOM PM2.5 FEM Met ParametersParameter code 44201 85101 88101 ManySpatial scaleNeighborhood Neighborhood Neighborhood NeighborhoodSite typeGeneral/backgroundPopulation Population General/backgroundMonitor objective Timely/public,standards/strategy,research supportTimely/public Timely/public Timely/public,standards/strategy,research supportMonitor typeSLAMS SLAMS SPM SLAMSPOC1 3 3 1Method code087 079 170 ManySampling method (List Instrument) 400 E TEOM BAM ITP-110-50HV, OT06A-2, BP-092, WD020C, WS-010CAnalysis method UV TE BETAStart date6/1/2010 6/1/2010 6/1/2010Operation schedule (e.g. 1:1, 1-Hour) 1:1 1-Hour 1-HourSampling season ALL YEAR ALL YEARProbe height (meters) 5.5 m 5.5 m 5.5 mDistance from supporting structure (meters) 0.1 m 0.5 m 0.5 mDistance from obstructions on roofDistance from obstructions not on roof (meters) 39m 35 m 37.5 mDistance from trees (meters) 13 m 15.5 m 14.5 mDistance to furnace or incinerator flue (meters) 48m 43.5 m 45 mDistance between collocated monitors (meters)Unrestricted airflow (degrees) 360 360 360Probe material (Teflon, etc.) TEFLON STAINLESS STEEL ALUMINUMResidence time (seconds) 13.5Frequency of flow rate verification for manual PMsamplers auditFrequency of flow rate verification for automatedPM analyzers auditBI-WEEKLY BI-WEEKLYFrequency of one-point QC check (gaseous) 1:1San Joaquin Valley Air Pollution Control DistrictJune 30, 2011Madera County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-572011 Air Monitoring Network PlanMadera—City (continued)PollutantOzone PM10 TEOM PM2.5 FEM Met ParametersLast Annual Performance Evaluation (gaseous) 7/22/2010Last two semi-annual flow rate audits for PMmonitors7/22/2010,12/7/2010Changes planned within the next 18 months (Y/N) NNNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Madera County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-582011 Air Monitoring Network PlanSite name Madera–Pump YardAIRS #060390004CountyMaderaCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyAll data reported by SJVAPCDSite Start Date10/1/99Pollutant ParametersOzone,NO2, TOTAL NMOC, NMHC (PAMS)Meteorological ParametersWind speed, wind direction, outdoor temperature, relative humidity, barometric pressure, solar radiationAddressAv 8 and Road 29 1/2, Madera CA 93637Latitude36.86722Longitude-120.01Elevation (m)85LocationPortable building, outside schoolDistance to road16.0 m (west)Traffic Count0Ground CoverDirt, pavedSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Madera County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-592011 Air Monitoring Network PlanMadera–Pump Yard (1 of 2)PollutantOzoneNO2TOTAL NMOC NMHC (PAMS)Parameter code 44201 42602 43102 ManySpatial scaleNeighborhood Neighborhood Neighborhood NeighborhoodSite typeGeneral/background Population Population PopulationMonitor objective Timely/public,standards/strategy,research supportStandards/strategy,researchResearch ResearchMonitor typePAMS PAMSPAMSPAMSPOC1 1 1 1Method code087 074 164 177Sampling method (List Instrument) 400E 42 910A 55CAnalysis method UV CL GC GCStart date10/1/1999 10/1/1999 10/1/1999 10/1/1999Operation schedule (e.g. 1:1, 1:3) 1:1 1:1 1:3 1:1Sampling season ALL YEAR ALL YEAR JUN-JUL-AUG ALL YEARProbe height (meters) 9.0 m 9.0 m 6.0 m 6.0 mDistance from supporting structure (meters) _____ _____ _____ _____Distance from obstructions on roof _____ _____ _____ _____Distance from obstructions not on roof (meters) _____ _____ _____ _____Distance from trees (meters) 41.0 m 41.0 m 41.5 m 41.5 mDistance to furnace or incinerator flue (meters) _____ _____ _____ _____Distance between collocated monitors (meters) __________ _____ _____Unrestricted airflow (degrees) 360 360 360 360Probe material (Teflon, etc.) TEFLON TEFLON S. STEEL TEFLONResidence time (seconds) 16.9 15.0 16.9Frequency of flow rate verification for manual PMsamplers audit_____ _____ _____ _____Frequency of flow rate verification for automatedPM analyzers audit_____ _____ _____ _____Frequency of one-point QC check (gaseous) 1:1 1:1 _____ 1:1Last Annual Performance Evaluation (gaseous) 11/4/2010 11/4/2010 5/10/2010 _____Last two semi-annual flow rate audits for PMmonitors_____ _____ _____ _____Changes planned within the next 18 months (Y/N) N NNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Madera County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-602011 Air Monitoring Network PlanMadera–Pump Yard (2 of 2)PollutantMet ParametersParameter codeManySpatial scaleRegionalSite typeGeneralMonitor objectiveResearch, timely/publicMonitor typeManyPOCManyMethod codeManySampling method (List Instrument) ITP-125-125-HV, OT-060A-2, BP-092, RH-HMP45D, SRD-Mod. 8-48, WD020C, WS-010CAnalysis method_____Start date10/1/1999Operation schedule (e.g. 1:1, 1:3) 1:1Sampling seasonALL YEARProbe height (meters)9.0 mDistance from supporting structure (meters) _____Distance from obstructions on roof _____Distance from obstructions not on roof (meters) _____Distance from trees (meters) 41.0 mDistance to furnace or incinerator flue (meters) _____Distance between collocated monitors (meters) _____Unrestricted airflow (degrees) 360Probe material (Teflon, etc.) _____Residence time (seconds) _____Frequency of flow rate verification for manual PMsamplers audit_____Frequency of flow rate verification for automated PManalyzers audit_____Frequency of one-point QC check (gaseous) _____Last Annual Performance Evaluation (gaseous) _____Last two semi-annual flow rate audits for PM monitors _____Changes planned within the next 18 months (Y/N) NSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Merced County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-612011 Air Monitoring Network PlanSite name Merced–CoffeeAIRS #060470003CountyMercedCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyAll data reported by SJVAPCDSite Start Date10/1/91Pollutant ParametersOzone, PM2.5 FEM,NO2Meteorological ParametersWind speed, wind direction, outdoor temperatureAddress385 S. Coffee St., Merced CA 95340Latitude37.28167Longitude-120.434Elevation (m)86LocationPortable building, residential areaDistance to road20 m (east)Traffic Count0Ground CoverDirt, vegetatedSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Merced County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-622011 Air Monitoring Network PlanMerced–CoffeePollutantOzone PM2.5 FEMNO2Met ParametersParameter code 44201 88101 42602 ManySpatial scale Neighborhood Neighborhood Neighborhood RegionalSite typePopulation Population Population GeneralMonitor objective Timely/public,standards/strategy,research supportTimely/public Standards/strategy Research, timely/publicMonitor typeSLAMS SPM SLAMS OtherPOC1 3 1 ManyMethod code 087 170 074 ManySampling method (List Instrument) 400E BAM 1020 42C ITP - 110-50HV, OT06A-2, WD-020C,WS-010CAnalysis method UV BETA CL ____Start date10/1/1991 10/1/1991 10/1/1991Operation schedule (e.g. 1:1, 1-Hour) 1:1 1-Hour 1:1 1:1Sampling season ALL YEAR ALL YEAR ALL YEAR ALL YEARProbe height (meters) 5.0 m 5.5 m 5.0 m 8.0 mDistance from supporting structure (meters) ____ ____ ____Distance from obstructions on roof ____ ____ ____Distance from obstructions not on roof (meters) ____ ____ ____Distance from trees (meters) 13.5 m 13.5 m 13.5 m 13.5 mDistance to furnace or incinerator flue (meters) ____ ____ ____Distance between collocated monitors (meters) ________ ____Unrestricted airflow (degrees) 345 345 345 345Probe material (Teflon, etc.) TEFLON ALUMINUM TEFLON ____Residence time (seconds) 11.9 13.7 ____Frequency of flow rate verification for manualPM samplers audit____ ____ ____Frequency of flow rate verification forautomated PM analyzers audit____ BI-WEEKLY ____ ____Frequency of one-point QC check (gaseous) 1:1 1:1____Last Annual Performance Evaluation (gaseous) 10/5/2010 10/5/2011 ____San Joaquin Valley Air Pollution Control DistrictJune 30, 2011Merced County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-632011 Air Monitoring Network PlanMerced–Coffee (continued)PollutantOzone PM2.5 FEMNO2Met ParametersLast two semi-annual flow rate audits for PMmonitors____ 10/5/2010,4/25/2011____ ____Changes planned within the next 18 months(Y/N)NNNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Merced County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-642011 Air Monitoring Network PlanSite name Merced—M StreetAIRS #060472510CountyMercedCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyData reported by CARB: PM10 FRM Data reported by Ventura County APCD: PM2.5FRMSite Start Date4/1/99Pollutant ParametersPM10 FRM, PM2.5 FRMMeteorological ParametersNoneAddress2334 M Street, Merced CA 95340Latitude37.30861Longitude-120.48Elevation (m)35LocationRoof, post officeDistance to road100 m (railroad, east); PM10: 66 m (north) & 72.5 m(south); PM2.5: 52.5 m (north), 87 m (south)Traffic Count22400Ground CoverGravelSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Merced County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-652011 Air Monitoring Network PlanMerced—M StreetPollutantPM10 FRM PM2.5 FRMParameter code81102 88101Spatial scaleNeighborhood NeighborhoodSite typeRepresentative concentration Representative concentrationMonitor objectiveStandards/strategy, research support Standards/strategy, research supportMonitor typeSLAMS SLAMSPOC11Method code063 120Sampling method (List Instrument) Sierra Andersen Partisol in service on 9/14/2010Analysis methodGRAVI-METRIC GRAVI-METRICStart date4/1/1999 4/1/1999Operation schedule (e.g. 1:1, 1:3) 1:6 1:3, 1:6Sampling seasonALL YEAR ALL YEARProbe height (meters)8.7 m 8.7 mDistance from supporting structure (meters) _____ _____Distance from obstructions on roof _____ _____Distance from obstructions not on roof (meters) _____ _____Distance from trees (meters) _____ _____Distance to furnace or incinerator flue (meters) 38.5 m 45.0 mDistance between collocated monitors (meters) __________Unrestricted airflow (degrees) 360 360Probe material (Teflon, etc.) _____ _____Residence time (seconds) _____ _____Frequency of flow rate verification for manual PMsamplers auditQUARTERLY MONTHLYFrequency of flow rate verification for automated PManalyzers audit_____ _____Frequency of one-point QC check (gaseous) _____ _____Last Annual Performance Evaluation (gaseous) __________Last two semi-annual flow rate audits for PM monitors 10/4/2010, 3/1/2011 10/4/2010, 4/5/2011Changes planned within the next 18 months (Y/N) NNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011San Joaquin County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-662011 Air Monitoring Network PlanSite name MantecaAIRS #060772010CountySan JoaquinCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyAll data reported by SJVAPCDSite Start Date11/16/10Pollutant ParametersPM2.5 FEM; PM10 TEOMMeteorological ParametersWind speed, wind direction, outdoor temperature, barometric pressureAddress530 Fishback Rd., Manteca CA 95337Latitude37.7933804512Longitude-121.24778867Elevation (m)11LocationPortable building, cement pad, dirt, corner near schoolDistance to road12 M to Fishback RdTraffic CountGround CoverSidewalk, dirt, grassSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011San Joaquin County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-672011 Air Monitoring Network PlanMantecaPollutantPM2.5 FEM PM10 TEOM Met ParametersParameter code 88101 85101 ManySpatial scaleNeighborhood Neighborhood NeighborhoodSite typePopulation Population PopulationMonitor objectiveStandards/StrategyResearch SupportStandards/StrategyResearch SupportStandards/StrategyResearch SupportMonitor typeSLAMS SLAMS Non-regulatoryPOC3 3 1Method code170 079 ManySampling method (List Instrument) BAM 1020 TEOM 1405 ITP-125-125,OT-06A2;BP-092; WD-020C;WS010CAnalysis methodStart date11/16/10 5/2/11 11/16/10Operation schedule (e.g. 1:1, 1-Hour) 1-Hour 1-Hour1:1Sampling season All Year All Year All YearProbe height (meters) 6M 6M 10MDistance from supporting structure (meters) 1.5 M 1.5 MDistance from obstructions on roof 0 0Distance from obstructions not on roof (meters) 87.5 M 87.5 M 87.5 MDistance from trees (meters) 53.5 M 53.5 M 53.5 MDistance to furnace or incinerator flue (meters) n/a n/a n/aDistance between collocated monitors (meters) n/a n/a n/aUnrestricted airflow (degrees) 360 360 360Probe material (Teflon, etc.) Aluminum TeflonResidence time (seconds)Frequency of flow rate verification for manual PMsamplers auditFrequency of flow rate verification for automatedPM analyzers auditBi-weekly Bi-WeeklyFrequency of one-point QC check (gaseous) n/a n/a n/aSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011San Joaquin County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-682011 Air Monitoring Network PlanManteca (continued)PollutantPM2.5 FEM PM10 TEOM Met ParametersLast Annual Performance Evaluation (gaseous) n/a n/a n/aLast two semi-annual flow rate audits for PM monitors 11/10/10 4/15/11Changes planned within the next 18 months (Y/N) N NNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011San Joaquin County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-692011 Air Monitoring Network PlanSite name Stockton–HazeltonAIRS #060771002CountySan JoaquinCollecting (Operating)AgencyAll equipment operated by CARBReporting AgencyAll data reported by CARBSite Start DatePollutant ParametersOzone, PM10 FRM, PM2.5FRM, PM2.5 FEM, CO,NO2, ToxicsMeteorological ParametersOutdoor temperatureAddress1593 E. Hazelton St., Stockton CA 95205LatitudeN 37o57’ 6”LongitudeN 121o16’ 8”Elevation (m)4LocationDistance to road62 mTraffic Count1000Ground CoverRoofSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011San Joaquin County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-702011 Air Monitoring Network PlanStockton–Hazelton (1 of 2)PollutantOzone PM10 FRM PM2.5 FRM PM2.5 FEMParameter code 44201 81102 88101 88101Spatial scale Neighborhood Neighborhood Neighborhood NeighborhoodSite typePopulation Population Population PopulationMonitor objective Unknown Highest Concentration Population Exposure PopulationExposureMonitor type SLAMS SLAMS SLAMS SLAMSPOC1 2 1 3Method code 087 063 118 170Sampling method (List Instrument) API/Teledyne 400Sierra Anderson 1200 R&P 2025 Met One 1020Analysis method UV Gravimetric Gravimetric BetaAttenuationStart dateOperation schedule (e.g. 1:1, 1:3, 1-Hour) 1:1 1:61:3 1-HourSampling season All year All year All year All yearProbe height (meters)Distance from supporting structure (meters)Distance from obstructions on roofDistance from obstructions not on roof (meters) None None None NoneDistance from trees (meters) 0.0 0.0 0.0 0.0Distance to furnace or incinerator flue (meters) None None None NoneDistance between collocated monitors (meters) -- ---- --Unrestricted airflow (degrees) 360 360 360 360Probe material (Teflon, etc.) Teflon Teflon TeflonTeflonResidence time (seconds) 8.5 -- -- --Frequency of flow rate verification for manualPM samplers audit-- Once a month Once a month --Frequency of flow rate verification forautomated PM analyzers audit-- -- -- Twice a monthFrequency of one-point QC check (gaseous) Twice a month -- -- --Last Annual Performance Evaluation (gaseous) 10/25/2010 -- -- --Last two semi-annual flow rate audits for PMmonitors-- 08/23/2010 10/26/2010 10/26/2010Changes planned within the next 18 months(Y/N)N NNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011San Joaquin County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-712011 Air Monitoring Network PlanStockton–Hazelton (2 of 2)Pollutant CONO2ToxicsSN20021014ToxicsSN20021016Met ParametersParameter code 42101 42602 Many Many ManySpatial scale NeighborhoodNeighborhood Neighborhood Neighborhood RegionalSite typePopulation Population Population Population GeneralMonitor objective PopulationExposureUnknown Unknown Unknown Research,timely/publicMonitor type SLAMS SLAMS ManyManyManyPOC1 2 ManyManyManyMethod code 054 074 ManyManyManySampling method (List Instrument) Dasibi 3008 Teco42, 42C, 42i Xontech 924 Xontech 924Analysis method IR CLStart dateOperation schedule (e.g. 1:1, 1:3) 1:1 1:1 1:1 1:11:1Sampling season All year All year All year All yearAll yearProbe height (meters)Distance from supporting structure (meters)Distance from obstructions on roofDistance from obstructions not on roof(meters)None None None None NoneDistance from trees (meters) 0.0 0.0 0.0. 0.0. 0.0.Distance to furnace or incinerator flue(meters)None None None None NoneDistance between collocated monitors(meters)-- -- 2 2 --Unrestricted airflow (degrees) 360 360 360 360 360Probe material (Teflon, etc.) Teflon Teflon TeflonTeflon TeflonResidence time (seconds) 7.9 8.7 -- -- --Frequency of flow rate verification formanual PM samplers auditFrequency of flow rate verification forautomated PM analyzers auditFrequency of one-point QC check (gaseous)Last Annual Performance Evaluation(gaseous)10/25/2010 10/25/2010 10/26/2010 10/26/2010San Joaquin Valley Air Pollution Control DistrictJune 30, 2011San Joaquin County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-722011 Air Monitoring Network PlanStockton–Hazelton (2 of 2) continuedPollutant CONO2ToxicsSN20021014ToxicsSN20021016Met ParametersLast two semi-annual flow rate audits for PMmonitorsChanges planned within the next 18 months(Y/N)N NNNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011San Joaquin County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-732011 Air Monitoring Network PlanSite name Stockton–Wagner/HoltAIRS #060773010CountySan JoaquinCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyAll data reported by CARBSite Start Date10/1/96Pollutant ParametersPM10 FRMMeteorological ParametersNoneAddress8778 Brattle Pl., Stockton CA 95209Latitude38.02972Longitude-121.353Elevation (m)7LocationOn school roofDistance to road30 m (north), 60 m (west)Traffic Count0Ground CoverFelt/rubberSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011San Joaquin County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-742011 Air Monitoring Network PlanStockton–Wagner/HoltPollutantPM10 FRMParameter code81102Spatial scaleNeighborhoodSite typePopulationMonitor objectiveStandards/strategy, research supportMonitor typeSLAMSPOC1Method code063Sampling method (List Instrument) AndersonAnalysis methodGRAVI-METRICStart date10/1/1996Operation schedule (e.g. 1:1, 1:3) 1:6Sampling seasonALL YEARProbe height (meters)10 mDistance from supporting structure (meters) 1.5 mDistance from obstructions on roof 11.8 mDistance from obstructions not on roof (meters) ____Distance from trees (meters) 12.5 mDistance to furnace or incinerator flue (meters) ____Distance between collocated monitors (meters) ____Unrestricted airflow (degrees) 280Probe material (Teflon, etc.) _____Residence time (seconds) _____Frequency of flow rate verification for manual PMsamplers auditQUARTERLYFrequency of flow rate verification for automated PManalyzers audit_____Frequency of one-point QC check (gaseous) _____Last Annual Performance Evaluation (gaseous) _____Last two semi-annual flow rate audits for PM monitors 11/10/2010, 3/10/2011Changes planned within the next 18 months (Y/N) YSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011San Joaquin County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-752011 Air Monitoring Network PlanSite name Tracy–AirportAIRS #060773005CountySan JoaquinCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyAll data reported by SJVAPCDSite Start Date1/11/05Pollutant ParametersOzone, PM10 TEOM, PM2.5 Non-FEM,NO2Meteorological ParametersWind speed, wind direction, outdoor temperature, barometric pressure, radio acoustic sounding system(RASS)Address5749 S. Tracy Blvd., Tracy CA 95376Latitude37.682682Longitude-121.442393Elevation (m)30LocationMunicipal airport yardDistance to road685.7 mTraffic Count868Ground CoverGravelSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011San Joaquin County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-762011 Air Monitoring Network PlanTracy–Airport (1 of 2)PollutantOzone PM10 TEOM PM2.5 Non-FEMNO2Parameter code44201 81102 88502 42602Spatial scaleNeighborhood Neighborhood Neighborhood NeighborhoodSite typeRegional transport RegionaltransportRegional transport PopulationMonitor objective Timely/public,standards/strategy,research supportTimely/public Timely/public Standards/strategyMonitor typeSLAMS SLAMS SPM SLAMSPOC1 3 3 1Method code087 079 731 074Sampling method (List Instrument) 400E TEOM BAM 1020 42CAnalysis method UV TAPEREDELEMENTBETAATTENUATIONCLStart date1/11/2005 10/25/2005 1/11/2005 1/11/2005Operation schedule (e.g. 1:1, 1-Hour) 1:1 1-Hour 1-Hour 1:1Sampling season ALL YEAR ALL YEAR ALL YEAR ALL YEARProbe height (meters) 7.0 m 6.5 m 6.5 m 7.0 mDistance from supporting structure (meters) _____ _____ _____Distance from obstructions on roof _____ _____ _____Distance from obstructions not on roof (meters) 42.7 m 42.7 m 42.7 m 42.7 mDistance from trees (meters) 41.5 m 41.5 m 41.5 m 41.5 mDistance to furnace or incinerator flue (meters) _____ _____ _____Distance between collocated monitors (meters) _____3.5m 3.5mUnrestricted airflow (degrees) 360 360 360 360Probe material (Teflon, etc.) TEFLON TEFLON ALUMINUM TEFLONResidence time (seconds) 10.6 _____ _____ 13.8Frequency of flow rate verification for manual PMsamplers audit_____ _____ _____ _____Frequency of flow rate verification for automatedPM analyzers audit_____ BI-WEEKLY BI-WEEKLY _____Frequency of one-point QC check (gaseous) 1:1 __________ 1:1Last Annual Performance Evaluation (gaseous) 4/6/2011 _____ _____ 4/6/2011San Joaquin Valley Air Pollution Control DistrictJune 30, 2011San Joaquin County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-772011 Air Monitoring Network PlanTracy–Airport (1 of 2) continuedPollutantOzone PM10 TEOM PM2.5 Non-FEMNO2Last two semi-annual flow rate audits for PMmonitors_____ 7/1/2010,4/6/20117/26/2010,4/6/2011_____Changes planned within the next 18 months (Y/N) N NN NTracy–Airport (2 of 2)PollutantMet ParametersParameter codeManySpatial scaleRegionalSite typeGeneralMonitor objectiveResearch, timely/publicMonitor typeSLAMSPOCManyMethod codeManySampling method (List Instrument) ITP-125-125 HV,OT-06A-2, BP-092, WD-020C, WS-010CAnalysis method_____Start date1/11/2005Operation schedule (e.g. 1:1, 1:3) 1:1Sampling seasonALL YEARProbe height (meters)10 mDistance from supporting structure (meters) _____Distance from obstructions on roof _____Distance from obstructions not on roof (meters) _____Distance from trees (meters) 48.7mDistance to furnace or incinerator flue (meters) _____Distance between collocated monitors (meters) _____Unrestricted airflow (degrees) 360Probe material (Teflon, etc.) _____Residence time (seconds) _____Frequency of flow rate verification for manual PMsamplers audit_____San Joaquin Valley Air Pollution Control DistrictJune 30, 2011San Joaquin County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-782011 Air Monitoring Network PlanTracy–Airport (2 of 2) continuedPollutantMet ParametersFrequency of flow rate verification for automated PManalyzers audit_____Frequency of one-point QC check (gaseous) _____Last Annual Performance Evaluation (gaseous) _____Last two semi-annual flow rate audits for PM monitors _____Changes planned within the next 18 months (Y/N) NSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Stanislaus County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-792011 Air Monitoring Network PlanSite name Modesto–14thStreetAIRS #060990005CountyStanislausCollecting (Operating)AgencyAll equipment operated by CARBReporting AgencyAll data reported by CARBSite Start Date1/1/81Pollutant ParametersOzone, PM10 FRM, PM2.5 FRM, PM2.5 FEM, COMeteorological ParametersWind speed, wind direction, outdoor temperature, barometric pressureAddress814 14th Street, Modesto CA 95354LatitudeN 37o38’ 31”LongitudeW 120o59’ 39”Elevation (m)27LocationDistance to road13 mTraffic Count10000Ground CoverRoofSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Stanislaus County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-802011 Air Monitoring Network PlanModesto–14thStreet (1 of 2)PollutantOzone PM10 FRM PM2.5 FRM PM2.5 FEMParameter code 44201 81102 88101 88101Spatial scaleNeighborhood Neighborhood Neighborhood NeighborhoodSite typePopulation Population Population PopulationMonitor objective Unknown Unknown Population ExposurePopulationExposureMonitor typeSLAMS SLAMS SLAMS SLAMSPOC1 3 1 3Method code087 063 118 170Sampling method (List Instrument) API/Teledyne 400Sierra Anderson1200R&P 2025 Met One 1020Analysis method UV Gravimetric Gravimetric Beta AttenuationStart dateOperation schedule (e.g. 1:1, 1:3, 1-Hour) 1:1 1:61:3 1-HourSampling season All year All year All year All yearProbe height (meters)Distance from supporting structure (meters)Distance from obstructions on roofDistance from obstructions not on roof (meters) None None None NoneDistance from trees (meters) None None None NoneDistance to furnace or incinerator flue (meters) None None None NoneDistance between collocated monitors (meters) -- ---- --Unrestricted airflow (degrees) 360 360 360 360Probe material (Teflon, etc.) Teflon Teflon TeflonTeflonResidence time (seconds) 5.9 -- -- --Frequency of flow rate verification for manualPM samplers audit-- Once a month Once a month --Frequency of flow rate verification for automatedPM analyzers audit-- -- -- Twice a monthFrequency of one-point QC check (gaseous) Twice a month -- -- --Last Annual Performance Evaluation (gaseous) 09/28/2010 -- -- --Last two semi-annual flow rate audits for PMmonitors-- 10/26/2010 10/12/2010 09/28/2010Changes planned within the next 18 months(Y/N)NNNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Stanislaus County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-812011 Air Monitoring Network PlanModesto–14thStreet (2 of 2)Pollutant CO PM2.5 Speciation Met ParametersParameter code 42101 ManyManySpatial scale NeighborhoodNeighborhood RegionalSite type PopulationGeneralMonitor objective UnknownMonitor type SLAMS Supplemental speciation SLAMSPOC 1 5ManyMethod code 067 811/812 ManySampling method (List Instrument)Dasibi 3008Analysis method IR GravimetricStart dateOperation schedule (e.g. 1:1, 1:3) 1:11:1Sampling season All yearAll yearProbe height (meters)Distance from supporting structure(meters)Distance from obstructions on roofDistance from obstructions not onroof (meters)NoneNoneDistance from trees (meters) NoneNoneDistance to furnace or incineratorflue (meters)NoneNoneDistance between collocatedmonitors (meters)----Unrestricted airflow (degrees) 360360Probe material (Teflon, etc.) TeflonTeflonResidence time (seconds) 5.4--Frequency of flow rate verificationfor manual PM samplers audit----Frequency of flow rate verificationfor automated PM analyzers audit----San Joaquin Valley Air Pollution Control DistrictJune 30, 2011Stanislaus County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-822011 Air Monitoring Network PlanModesto–14thStreet (2 of 2) continuedPollutant CO PM2.5 Speciation Met ParametersFrequency of one-point QC check(gaseous)Twice amonth--Last Annual PerformanceEvaluation (gaseous)09/28/2010--Last two semi-annual flow rateaudits for PM monitors----Changes planned within the next18 months (Y/N)NNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Stanislaus County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-832011 Air Monitoring Network PlanSite name TurlockAIRS #060990006CountyStanislausCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyData reported by SJVAPCD: Ozone, PM2.5 FEM,CO,NO2, MeteorologyData reported by CARB: PM10 FRMSite Start Date1994Pollutant ParametersOzone, PM10 FRM, PM2.5 FEM, CO,NO2Meteorological ParametersWind speed, wind direction, outdoor temperature, barometric pressureAddress1034 S. Minaret St., Turlock CA 95380Latitude37.48806Longitude-120.836Elevation (m)30LocationPortable building – neighborhoodDistance to road32 m (east), 4 m (north)Traffic Count670Ground CoverGravelSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Stanislaus County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-842011 Air Monitoring Network PlanTurlock (1 of 2)PollutantOzone PM10 FRM PM2.5 FEM COParameter code 44201 81102 88101 42101Spatial scale Neighborhood Neighborhood Neighborhood NeighborhoodSite typePopulation Population Population PopulationMonitor objective Timely/public,standards/strategy,research supportStandards/strategy,research supportTimely/public Standards/strategyMonitor type SLAMS SLAMS SLAMS SLAMSPOC1 1 3 1Method code 087 063 170 054Sampling method (List Instrument) 400E Sierra Andersen 1020 48CAnalysis method UV GRAVIMETRIC Beta Attenuation IRStart date1994 1994 9/14/2006 1994Operation schedule (e.g. 1:1, 1:3, 1-Hour ) 1:1 1:61-Hour 1:1Sampling season ALL YEAR ALL YEAR ALL YEAR ALL YEARProbe height (meters) 7 m 6.5 m 5.4 m 7 mDistance from supporting structure (meters) ____ ____ ____ ____Distance from obstructions on roof ____ ____ ____ ____Distance from obstructions not on roof(meters)____ ____ ____ ____Distance from trees (meters) 37.5 m 37.5 m 37.5 m 37.5 mDistance to furnace or incinerator flue (meters) 48.0 m 48.0 m 48.0 m 48.0 mDistance between collocated monitors(meters)____ ____ ____ ____Unrestricted airflow (degrees) 360 360 360 360Probe material (Teflon, etc.) TEFLON ____ ALUMINUMTEFLONResidence time (seconds) 14.8 ____ ____ 14Frequency of flow rate verification for manualPM samplers audit____ QUARTERLY ____ ____Frequency of flow rate verification forautomated PM analyzers audit____ ____ BI-WEEKLY ____Frequency of one-point QC check (gaseous) 1:1 ________ 1:1Last Annual Performance Evaluation(gaseous)10/7/2010 ____ ____ 10/7/2010San Joaquin Valley Air Pollution Control DistrictJune 30, 2011Stanislaus County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-852011 Air Monitoring Network PlanTurlock (1 of 2) continuedPollutantOzone PM10 FRM PM2.5 FEM COLast two semi-annual flow rate audits for PMmonitors____ 10/18/2010,2/4/20114/28/2010,9/27/2010____Changes planned within the next 18 months(Y/N)NNNNTurlock (2 of 2)PollutantNO2Met ParametersParameter code42602 ManySpatial scaleNeighborhood RegionalSite typePopulation GeneralMonitor objectiveStandards/strategy Research, timely/publicMonitor typeSLAMS OtherPOC1 ManyMethod code074 ManySampling method (List Instrument) 42C ITP-125-125 HV, OT-060A-2, BP-090D, WD-020C,WS-010CAnalysis methodCL ____Start date1994 1994Operation schedule (e.g. 1:1, 1:3) 1:1 1:1Sampling seasonALL YEAR ALL YEARProbe height (meters) 7 m 7.7 m7 m (OT)Distance from supporting structure (meters) ____ ____Distance from obstructions on roof ____ ____Distance from obstructions not on roof (meters) ____ ____Distance from trees (meters) 37.5 m 37.5 mDistance to furnace or incinerator flue (meters) 48.0 m 48.0 mDistance between collocated monitors (meters) ________Unrestricted airflow (degrees) 360 360Probe material (Teflon, etc.) TEFLON ____Residence time (seconds) 14.1 ____Frequency of flow rate verification for manual PMsamplers audit____ ____San Joaquin Valley Air Pollution Control DistrictJune 30, 2011Stanislaus County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-862011 Air Monitoring Network PlanTurlock (2 of 2) continuedPollutantNO2Met ParametersFrequency of flow rate verification for automated PManalyzers audit____ ____Frequency of one-point QC check (gaseous) 1:1 ____Last Annual Performance Evaluation (gaseous) 3/24/2009 ____Last two semi-annual flow rate audits for PM monitors ____ ____Changes planned within the next 18 months (Y/N) NNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Tulare County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-872011 Air Monitoring Network PlanSite name PortervilleAIRS #061072010CountyTulareCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyAll data reported by SJVAPCDSite Start Date3/8/2010Pollutant ParametersOzone, PM2.5 FEMMeteorological ParametersWind speed, wind direction, outdoor temperature, barometric pressureAddress1839 S. Newcomb St., Porterville CA 93257Latitude36.031031Longitude-119.055018Elevation (m)41LocationPortable building on parking lotDistance to road160 m (east)Traffic CountGround CoverPavedSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Tulare County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-882011 Air Monitoring Network PlanPortervillePollutantOzone PM2.5 FEM Met ParametersParameter code44201 88101 ManySpatial scaleNeighborhood Neighborhood NeighborhoodSite typePopulation Population PopulationMonitor objectiveTimely/public,standards/strategy,research supportTimely/public Timely/publicMonitor typeSLAMS SPM SLAMSPOC1 3 1Method code087 731 ManySampling method (List Instrument) 400 E 1020 ITP-125-125 HV, OT060A-2, BP-092, WD020C, WS-010CAnalysis methodUV BETA-ATTENUATIONStart date3/8/2010 3/8/2010 3/8/2010Operation schedule (e.g. 1:1, 1:3, 1-Hour) 1:1 1-Hour 1:1Sampling seasonALL YEAR ALL YEAR ALL YEARProbe height (meters) 5.4 m 5.4 m 9.6 mDistance from supporting structure (meters)Distance from obstructions on roofDistance from obstructions not on roof (meters) 10m 10 mDistance from trees (meters)Distance to furnace or incinerator flue (meters)Distance between collocated monitors (meters)Unrestricted airflow (degrees) 345 345 345Probe material (Teflon, etc.) TEFLON ALUMINUMResidence time (seconds) 6.0Frequency of flow rate verification for manual PMsamplers auditFrequency of flow rate verification for automated PManalyzers auditBI-WEEKLYFrequency of one-point QC check (gaseous) 1:1Last Annual Performance Evaluation (gaseous) 7/20/2010Last two semi-annual flow rate audits for PM monitors 7/20/2010, 9/29/2010Changes planned within the next 18 months (Y/N) N NNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Tulare County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-892011 Air Monitoring Network PlanSite name Sequoia–Ash MountainAIRS #061070009CountyTulareCollecting (Operating)AgencyAll equipment operated by NPSReporting AgencyAll data reported by NPSSite Start Date1/1/00Pollutant ParametersOzone, PM2.5 FRM, PM2.5 FEM, CASTnet (dry deposition)Meteorological ParametersWind speed, wind direction, outdoor temperature, relative humidity, solar radiationAddressAsh Mountain, Sequoia National Park CALatitude36.48944Longitude-118.829Elevation (m)535LocationDistance to road122 mTraffic Count1000Ground CoverDirtSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Tulare County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-902011 Air Monitoring Network PlanSequoia–Ash MountainPollutantOzone PM2.5 FRM PM2.5 FEM Met ParametersParameter code 44201 88501 88502 ManySpatial scaleRegional Regional Regional RegionalSite typeRegional transport Regional transport Regional transport GeneralMonitor objective Timely/public,standards/strategy,research supportResearch support Timely/public Research,timely/publicMonitor typeNon-EPA Federal Non-EPA Federal Non-EPA FederalNon-EPA FederalPOC1 1 1 1Method code047 750 707 ManySampling method (List Instrument) TECO 49, 49CAnalysis method UV Gravimetric Beta AttenuationStart date2000 1992 2007Operation schedule (e.g. Hourly, 1:1, 1:3, 1-Hour)1:1 1:6 1-Hour 1:1Sampling season All year All year All year All yearProbe height (meters) 10 5 4Distance from supporting structure (meters) 3 2 1.5Distance from obstructions on roof 5Distance from obstructions not on roof (meters) --Distance from trees (meters) 15 – 20 10 - 20 15 – 20Distance to furnace or incinerator flue (meters) 305 305 305Distance between collocated monitors (meters) 3 3 3Unrestricted airflow (degrees) 360 360 360Probe material (Teflon, etc.) Teflon Teflon TeflonResidence time (seconds) --Frequency of flow rate verification for manual PMsamplers auditFrequency of flow rate verification for automatedPM analyzers auditFrequency of one-point QC check (gaseous)Last Annual Performance Evaluation (gaseous) March2009 December 2008,August 2007Last two semi-annual flow rate audits for PMmonitorsChanges planned within the next 18 months (Y/N) NNNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Tulare County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-912011 Air Monitoring Network PlanSite name Sequoia–Lower KaweahAIRS #061070006CountyTulareCollecting (Operating)AgencyAll equipment operated by NPSReporting AgencyAll data reported by NPSSite Start Date4/1/1981Pollutant ParametersOzone, NADP (wet deposition)Meteorological ParametersWind speed, wind direction, outdoor temperature, relative humidity, solar radiationAddressLower Kaweah Campground, Sequoia National Park, CALatitude36.56611Longitude-118.7776Elevation (m)1890LocationDistance to road1500 mTraffic Count5000Ground CoverDirtSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Tulare County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-922011 Air Monitoring Network PlanSequoia–Lower KaweahPollutantOzone Met ParametersParameter code44201 ManySpatial scaleRegional RegionalSite typeRegional transport GeneralMonitor objectiveTimely/public,standards/strategy,research supportResearch, timely/publicMonitor typeNon-EPA Federal Non-EPA FederalPOC1 1Method code087 ManySampling method (List Instrument) TECO 49, 49CAnalysis methodStart date1982Operation schedule (e.g. 1:1, 1:3) 1:1 1:1Sampling seasonAll year All yearProbe height (meters) 10Distance from supporting structure (meters)Distance from obstructions on roofDistance from obstructions not on roof (meters)Distance from trees (meters) 15 – 20Distance to furnace or incinerator flue (meters) 750Distance between collocated monitors (meters)Unrestricted airflow (degrees)Probe material (Teflon, etc.) TeflonResidence time (seconds)Frequency of flow rate verification for manual PMsamplers auditFrequency of flow rate verification for automated PManalyzers auditFrequency of one-point QC check (gaseous)Last Annual Performance Evaluation (gaseous) March2009Last two semi-annual flow rate audits for PM monitorsChanges planned within the next 18 months (Y/N) NNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Tulare County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-932011 Air Monitoring Network PlanSite name Visalia–AirportAIRS #061073000CountyTulareCollecting (Operating)AgencyAll equipment operated by SJVAPCDReporting AgencyAll data reported by SJVAPCDSite Start DateSeptember 2000Pollutant ParametersNoneMeteorological ParametersWind speed, wind direction, outdoor temperature, relative humidity, barometric pressure, solar radiation,radio acoustic sounding system (RASS)AddressAirport, Visalia CA 93291Latitude36.31389Longitude-119.392Elevation (m)90LocationMunicipal airport yardDistance to road81 m (west), 29.5 (parking lot)Traffic Count32000Ground CoverVegetatedSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Tulare County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-942011 Air Monitoring Network PlanVisalia–AirportPollutantMet ParametersParameter codeManySpatial scaleRegionalSite typeGeneralMonitor objectiveResearch, timely/publicMonitor typePAMSPOC1Method codeManySampling method (List Instrument) ITP-125-125 HV, OT-06A-2, BP-090D, RH-083-0-6, SRD-Mod. 8-48, WD020C, WS-010BAnalysis method______Start date10/1/1999Operation schedule (e.g. 1:1, 1:3) 1:1Sampling seasonALL YEARProbe height (meters)10 mDistance from supporting structure (meters) ______Distance from obstructions on roof ______Distance from obstructions not on roof (meters) ______Distance from trees (meters) 8 mDistance to furnace or incinerator flue (meters) ______Distance between collocated monitors (meters) ______Unrestricted airflow (degrees) 270Probe material (Teflon, etc.) ______Residence time (seconds) ______Frequency of flow rate verification for manual PMsamplers audit______Frequency of flow rate verification for automated PManalyzers audit______Frequency of one-point QC check (gaseous) ______Last Annual Performance Evaluation (gaseous) ______Last two semi-annual flow rate audits for PM monitors ______Changes planned within the next 18 months (Y/N) NSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Tulare County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-952011 Air Monitoring Network PlanSite name Visalia—ChurchAIRS #061072002CountyTulareCollecting (Operating)AgencyAll equipment operated by CARBReporting AgencyAll data reported by CARBSite Start Date7/1/79Pollutant ParametersOzone, PM10 FRM, PM2.5 FRM, PM2.5 FEMMeteorological ParametersWind speed, wind direction, outdoor temperature, barometric pressureAddress310 N. Church St., Visalia CA 93291LatitudeN 36o19’ 57”LongitudeW 119o17’ 27”Elevation (m)97LocationPortable buildingDistance to road23 mTraffic Count10000Ground CoverRoofSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Tulare County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-962011 Air Monitoring Network PlanVisalia—Church (1 of 2)PollutantOzone PM10 FRM PM2.5 FRM PM2.5 Non-FEMParameter code 44201 81102 88101 88501Spatial scale Neighborhood Neighborhood Neighborhood NeighborhoodSite typePopulation Population Population Regional transportMonitor objective Unknown Unknown Population Exposure Population ExposureMonitor typeSLAMS SLAMS SLAMS Non-regulatoryPOC1 2 1 3Method code 087 063 118 731Sampling method (List Instrument) API/Teledyne 400Sierra Anderson1200R&P 2025 Met One 1020Analysis method UV Gravimetric Gravimetric Beta attenuationStart dateOperation schedule (e.g. 1:1, 1:3) 1:1 1:6 1:3 1-HourSampling season All year All year All year All yearProbe height (meters)Distance from supporting structure (meters)Distance from obstructions on roofDistance from obstructions not on roof (meters) None None None NoneDistance from trees (meters) None None None NoneDistance to furnace or incinerator flue (meters) None None None NoneDistance between collocated monitors (meters) -- ---- --Unrestricted airflow (degrees) 360 360 360 360Probe material (Teflon, etc.) Teflon Teflon TeflonTeflonResidence time (seconds) 14.2 -- -- --Frequency of flow rate verification for manualPM samplers audit-- Once a month Once a month --Frequency of flow rate verification forautomated PM analyzers audit-- -- -- Twice a monthFrequency of one-point QC check (gaseous) Twice a month -- -- --Last Annual Performance Evaluation (gaseous) 01/20/2011 -- -- --Last two semi-annual flow rate audits for PMmonitors-- 04/05/2007 12/02/2010 05/19/2010Changes planned within the next 18 months(Y/N)NNNNSan Joaquin Valley Air Pollution Control DistrictJune 30, 2011Tulare County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-972011 Air Monitoring Network PlanVisalia—Church (2 of 2)Pollutant Met ParametersPM2.5 SpeciationParameter code ManyManySpatial scale RegionalNeighborhoodSite type GeneralMonitor objective Research, timely/publicMonitor type ManySupplemental speciationPOC 15Method code Many811/812Sampling method (ListInstrument)Analysis methodGravimetricStart dateOperation schedule (e.g.1:1, 1:3)1:1Sampling season All yearProbe height (meters)Distance from supportingstructure (meters)Distance from obstructionson roofDistance from obstructionsnot on roof (meters)NoneDistance from trees(meters)NoneDistance to furnace orincinerator flue (meters)NoneDistance betweencollocated monitors(meters)--Unrestricted airflow(degrees)360Probe material (Teflon,etc.)TeflonResidence time (seconds) --San Joaquin Valley Air Pollution Control DistrictJune 30, 2011Tulare County____________________________________________________________________________________________________________________Appendix B: Detailed Air Monitoring Site InformationB-982011 Air Monitoring Network PlanVisalia—Church (2 of 2) continuedPollutant Met ParametersPM2.5 SpeciationFrequency of flow rateverification for manual PMsamplers audit--Frequency of flow rateverification for automatedPM analyzers audit--Frequency of one-point QCcheck (gaseous)--Last Annual PerformanceEvaluation (gaseous)--Last two semi-annual flowrate audits for PM monitors--Changes planned within thenext 18 months (Y/N)NN