# 

GLP10

- 01

- 02; I nterim Report 27

- Analysis of PFBS, PFHS, and PFOS in

Groundwater Samples Collected from Decatur, AL, 1 st 

## Quarter 2012

Study Title

Analysis of Perfluorooctane Sulfonate (PFOS), Perfluorohexane Sulfonate (PFHS) and Perfluorobutane S ulfonate

(PFBS) in Groundwater, Soil and Sediment for the 3M Decatur Phase 3 Site

- Related Monitoring Program



## Data Requirement

EPA TSCA Good Laboratory Practice Standards 40 CFR Part 792



## Study Director

Jaisimha Kesari P.E., DEE

Weston Solutions, Inc.

1400 Weston Way

West Chester, PA 19380

Phone: 610

- 701

- 3761



## Author

Susan Wolf

3M Environmental Laboratory



## Interim Report Completion Date

Date of signing



## Performing Laboratory

3M Environmental Health and Safety Operations

Environmental Laboratory

3M Center, Bldg 260

- 05 -N- 17

St. Paul , MN 55144



## Project Identification

GLP10

- 01

- 02

- 27



## Total Number of Pages

114



### Testing Cert #2052.01 The testing reported herein meet the requirements of ANSI/ ISO/IEC 17025:2005 “General

Requirements for the Competence of Testing and Calibration Laboratories”, in

accordance with the A2LA Testing Certificate # 2052.01. Testing that complies with this

International Standard also meets principles of ISO 9001:2000.

### This page has been reserved for specific country requirements.GLP10-01-02 ; Interi m Repor t 2 7

Analysi s o f PFBS

, PFHS

, an d PFO S i n Groundwate r

Decatur

, A L

- 1s

t Quarte r 201 2 Samplin g



## GL P COMPLIANC E STATEMEN T

Repor t Title : GLP10-01-02 ; Interi m Repor t 2 7

- Analysi s o f PFBS

, PFHS

, an d PFO S i n Groundwate r Sample s Collecte d fro m Decatur

, AL

, 1 Quarte r 201 2

Study : Analysi s o f Perfluorooctan e Sulfonat e (PFOS)

, Perfluorohexan e Sulfonat e (PFHS ) an d

Perfiuorobutan e Sulfonat e (PFBS ) i n Groundwater

, Soi l an d Sedimen t fo r th e 3 M Decatu r Phas e 3

Site-Relate d Monitorin g Program

.

Thi s analytica l phas e wa s conducte d i n complianc e wit h Toxi c Substance s Contro l Ac t (TSCA ) Goo d

Laborator y Practic e (GLP ) Standards

, 4 0 CF R 792

, wit h th e exception s liste d below :

• Thes e ar e environmenta l sample s wher e ther e i s n o specifi c tes t substance

, n o specifi c tes t

syste m an d n o dosin g o f a tes t system

.

• Th e referenc e substance s hav e no t bee n characterize d unde r th e GLP s an d th e stabilit y unde r

storag e condition s a

t th e tes t sit e hav e no

t bee n determine d unde r GLPs

.



# 2 2 /

Gar y Ho|^enstein f Sponso r Representativ e Dat e

Jaisimh a Kesari

, P.E.

, DEE

, Stud y Directo r Dat e

Pag e 3 o f 11 4GLP10-01-02 ; Interi m Repor t 2 7

Analysi s o f PFBS

, PFHS

, an d PFO S i n Groundwate r

Decatur

, A L

- 1s

t Quarte r 201 2 Samplin g



## QUALIT Y ASSURANC E STATEMEN T

Repor t Title : GLP10-01-02 ; Interi m Repor t 2 7

- Analysi s o f PFBS

, PFHS

, an d PFO S i n Groundwate r Sample s Collecte d fro m Decatur

, AL

, 1 Quarte r 201 2

Study : Analysi s o f Perfluorooctan e Sulfonat e (PFOS)

, Perfluorohexan e Sulfonat e (PFHS ) an d

Perfluorobutan e Sulfonat e (PFBS ) i n Groundwater

, Soi l an d Sedimen t fo r th e 3 M Decatu r Phas e 3

Site-Relate d Monitorin g Program

.

Thi s analytica l phas e wa s audite d b y th e 3 M Environmenta l Laborator y Qualit y Assuranc e Uni t (QAU)

,

a s indicate d i n th e followin g table

. Th e finding s wer e reporte d t o th e principa l investigato r (P.I.)

, laborator y managemen t an d stud y director

.

Inspectio n Date s Phas e Dat e R e porte d t o

Inspectio n Date s Phas e Testin g Facilit y IVIanagemen t Stud y Directo r

5/7/1 2 Dat a / Interi m Repor

t 5/9/1 2 5/9/1 2



## QA U Representativ e / 2

_

Pag e 4 o f 11 4

# T ABLE OF 

# C ONTENTS

GLP Compliance Statement

...................................................................................................................... 3

Quality Assurance Statement

.................................................................................................................... 4

Table of Contents

................................................................................................ ....................................... 5

List of Tables

................................................................................................ .............................................. 6

1 Study Information

................................................................................................ ................................ 8

2 Summary

................................................................................................ ............................................. 9

3 Introduction

................................................................................................ ........................................ 11

4 Test & Control Substances

.............................................................................................................. 11

5 Reference Substances

..................................................................................................................... 12

6 Test System

................................................................................................ ...................................... 13

7 Method Summary

............................................................................................................................. 14

7.1 Methods

............................................................................................................................ 14

7.2 Sample Collection

............................................................................................................. 14

7.3 Sample Preparation

.......................................................................................................... 14

7.4 Analys is

............................................................................................................................. 14

8 Analytical Results

.............................................................................................................................. 16

8.1 Calibration

......................................................................................................................... 16

8.2 System Suitabilit y

............................................................................................................. 16

8.3 Limit of Quantitation (LOQ)

.............................................................................................. 16

8.4 Continuing Calibration

...................................................................................................... 17

8.5 Blanks

................................................................................................ ................................ 17

8.6 Lab Control Spikes (LCSs)

.............................................................................................. 17

8.7 Analytical Method Uncertainty

......................................................................................... 19

8.8 Field Matrix Spikes (FMS)

................................................................................................ 19

### 9 Data Summary and Discussion

....................................................................................................... 20

10 Conclusion

................................................................................................ ......................................... 32

11 Data/Sample Retention

.................................................................................................................... 32

12 Attachments

................................................................................................ ...................................... 32

13 Signatures

................................................................................................ ......................................... 33



# L IST OF 

# T ABLES

Table 1. Summarized PFBS, PFHS, and PFOS Results (Decatur Groundwater

– Q1 2012).

. 10

Table 2. Sample Description Key Code.

....................................................................................... 13

Table 3. Instrument Parameters.

................................................................................................... 15

Table 4. Liquid Chromatography Conditions.

............................................................................... 15

Table 5. Mass Transitions.

............................................................................................................. 15

Table 6. Limit of Quantitation (LOQ).

............................................................................................. 17

Table 7. Laboratory Control Spike Recovery.

............................................................................... 18

Table 8. Analytical Uncertainty.

..................................................................................................... 19

Table 9. Field Matrix Spikes.

.......................................................................................................... 20

Table 10. DAL GW 203L 120327

.................................................................................................. 22

Table 11. DAL GW 220R 120323

................................................................................................ . 22

Table 12. DAL GW 220L 120323

.................................................................................................. 23

Table 13. DAL GW 222R 120327

................................................................................................ . 23

Table 14. DAL GW 227R 120316

................................................................................................ . 24

Tabl e 15. DAL GW 227L 120316

.................................................................................................. 24

Table 16. DAL GW 310R 120320

................................................................................................ . 25

Table 17. DAL GW RW312R 120328

........................................................................................... 25

Table 18. DAL GW 317L 120320

.................................................................................................. 26

Table 19. DAL GW 324L 120314

.................................................................................................. 26

Table 20. DAL GW RW327R 1203 23

........................................................................................... 27

### Table 21. DAL GW 328R 120314

................................................................................................ . 27

Table 22. DAL GW 328L 120314

.................................................................................................. 28

Table 23. DAL GW 330R 120314

................................................................................................ . 28

Table 24. DAL GW 330L 120314

.................................................................................................. 29

Table 25. DAL GW RW331S 120328

........................................................................................... 29

Table 26. DAL GW 335R 120322

................................................................................................ . 30

Table 27. DAL GW GRS04 120328

.............................................................................................. 30

Table 28. Trip Blank 1

.................................................................................................................... 31

Table 29. Equipment Rinseate Blank

............................................................................................ 31

## 1 **Study Information**

Sponsor

3M Company

Sponsor Representative

Gary Hohenstein

3M EHS Operations

3M Building 22 4- 5W

- 03

Saint Paul, MN 55144

- 1000

Phone: (651) 737

- 3570

Study Director

Jaisimha Kesari, P.E., DEE

Weston Solutions, Inc.

West Chester, PA 19380

Phone: (610) 701

- 3761

Fax: (610) 701

- 7401

j.kesari@westonsolutions.com

Study Location

Testing Facility

3M EHS Operations

3M Environmental Laboratory

Building 260

- 5N

- 17

St. Paul, MN 55144

Study Personnel

William K. Reagen, Ph.D., 3M Laboratory Manager

Cleston Lange, Ph.D., Principal Analytical Investigator, ( clange@mmm.com ); phone (651)

- 733

- 9860

Susan Wolf, 3M Analyst

Chelsie Grochow; analyst

Kelly Ukes; analyst

Study Dates

Study Initiation: March 8, 2010

Interim 2 7 Experimental Termination: May 2 , 2012

Interim Report Completion: Date of Interim Report Signing

Location of Archiv es

All original raw data and the analytical report have been archived at the 3M Environmental

Laboratory according to 40 CFR Part 792. The test substance and analytical reference standard

reserve samples are archived at the 3M Environmental Laboratory acc ording to 40 CFR Part 792

## 2 **Summary**

The 3M Environmental Laboratory received groundwater samples from wells located in Decatur, AL,

representing eighteen (18) different sampling locations collected March 14

- 28, 2012. A total of

seventy

- seven sample bottles were received at the 3M Environmental Laboratory for perfluorooctane

sulfonate (PFOS), perfluorohexane sulfonate (PFHS) and perfluorobutane sulfonate (PFBS)

.

For each sampling location, four sample bottles were collected; sample, sample duplicate, low field

matrix spike, and high field matrix spike. One trip blank containing Milli

- QTM water and approp riate trip

blank spikes were analyzed with the groundwater samples. In addition, one equipment rinseate blank

sample was collected . The equipment rinseate blank did not have FMS samples prepared for

determination of P FBS, PFHS, or PFOS recovery. The reference standard used to prepare the FMS

samples consisted of both linear and branched isomers of PFOS. All groundwater samples for this

project were logged under GLP10

- 01

- 02

- 27

.

The groundwater samples, trip and equipment rinseate blanks for GLP10

- 01

- 02

- 27 were received from

Weston personnel on April 4, 2012

. All of the samples were prepared and analyzed for PFBS, PFHS,

and PFOS following 3M Environmental Laboratory Method ETS -8- 044. 1. Many of the groundwater

sample s required dilution to ob tain PFBS, PFHS, and /or PFOS concentrations within the range of the

curve, in some instances up to a 500

- fold dilution were required. The average measured PFBS, PFHS,

and PFOS concentrations are summarized in Table 1 . Equipment rinseate and the trip blanks were

below the lower limit of quantitation (LLOQ), indicating adequate control of sample contamination

during shipping and sample collections.

The PFBS concentration results for all g roundwater locations ranged from <0.0 25 0 ng/mL to 2300

ng/mL. The PFHS concentration results for all groundwater locations ranged from 0.0877 ng/mL to

8280 ng /mL. The PFOS concentration results for the reported groundwater locations ranged from

0.230 ng/ mL to 3 60 0 ng/mL. T he analytical uncertainties associated with the reported results using

external standard calibration are as follows: PFBS ± 21 %, PFH S ± 1 7 %, and PFOS ± 34 %. For

several sampling locations, improved FMS recoveries were seen for P FO S usi ng internal standard

calibration. The internal standard was added to the sample or sample dilution prior to analysis.

S amples analyzed using internal calibration for PFOS have an analytical method uncertainty of ± 17 %.

### Table 1

. Summarized PFBS, PFHS, and PFOS Results (Decatur Groundwater

– Q1 2012 ).

PFBS (1) 

### PFHS (1) 

### PFOS (1)



### Sampl ing Location Avg. Conc

. (ng/mL)

± %RPD Avg. Conc

. (ng/mL)

± %RPD Avg. Conc

. (ng/mL)

± %RPD

DAL GW 203L 120327 30.8 ± 1.9 % 202 ± 3.0 % 666 ± 11 % (2)



### DAL GW 220R 120323 6.90 ± 6.8 % 29.8 ± 1.3 % 33.0 ± 26 % (3)



### DAL GW 220L 120323 8.52 ± 4.9 % 41.4 ± 0.24% 49.5 ± 12 %

DAL GW 222R 120327 64.5 ± 5.1 % 407 ± 2.2 % 351 ± 25 % (3)



### DAL GW 227R 120316 14.8 ± 6.8 % 86.3 ± 1.4 % 616 ± 8.4 % (2)



### DAL GW 227L 120316 2 20 ± 3 .6% 40.1 ± 0.25 % 220 ± 2.3 % (2)



### DAL GW 310R 120320 986 ± 0.71 % 328 ± 4.9 % 671 ± 7.9 % (2 ,4 )



### DAL GW RW312R 120328 740 ± 0.14 % 494 ± 0.61 % 992 ± 7.7 % (2)



### DAL GW 317L 120320 <0.0 250 0.0 877 ± 2.7 % 0.230 ± 4.3 % (2)



### DAL GW 324L 120314 82.1 ± 2.3 % 88.4 ± 1.0 % 194 ± 0.52 % (2)



### DAL GW RW327R 120323 62.1 ± 3.5 % 146 ± 4.1 % 345 ± 4.6%

DAL GW 328R 120314 31.0 ± 1.3 % 57. 0 ± 5.1 % 144 ± 6.9 % (2)



### DAL GW 328L 120314 47.9 ± 3.3 % 18.7 ± 4.3 % 1.14 ± 2.6 % (2)



### DAL GW 330R 120314 2120 ± 0.47 % 185 ± 2.2 % 462 ± 1.5 %

DAL GW 330L 120314 397 ± 4.3 % 359 ± 2.5 % 146 ± 1 4%

DAL GW RW331S 120328 997 ± 2.6 % 473 ± 1.9 % 850 ± 4.2 % (2 ,4 )



### DAL GW 335R 120322 864 ± 1.0 % 1600 ± 2.5 % 3600 ± 6.7 %

DAL GW GRS04 120328 2300 ± 2.2 % 8280 ± 1.7 % NR (5)



### Trip Blank (Milli

- QTM Water) <0.0250 <0.0250 <0.0232

Rinseate Blank <0.0250 <0.0250 <0. 0 232

(1) All samples reported using external standard calibration unless noted otherwise. The analytical method

uncertainties associated with the reported results u sing external calibration are: PFBS ± 21%, PFHS ± 17%,

and PFOS ± 34%.

(2) Sample results reported using internal standard calibration. The analytical method uncertainty associated with

the reported results using internal calibration is ± 1 7% for PFOS.

(3) The sample/sample duplicate RPD did not meet method acceptance criteria of ≤ 20%.

(4) The FMS recovery did not meet acceptance criteria. The analytical uncertainty has been adjusted for PFOS

to ± 32 % for DAL GW 310R and to ± 36% for DAL GW RW331S

. (5) NR = Not Reportable; Sampling location not reportable due to the FMS recovery being bel ow 50%.

## 3 **Introduction**

This analytical study was conducted as part of the Phase 3 En vironmental M onitoring and A ssessment

P rogram for the 3M facility located in Decatur, Alabama. The objective of the overall program is to gain

information regarding concentrations of perfluorooctane sulfonate (PFOS), perfluorohexane sulfonate

(PFHS) and perfluorobutane sulfonate (PFBS)

, in various environmental media such as groundwater,

soils and sediments that are associated with and near the Decatur facility. This analytical study was

conducted to analyze ground water samples collected from various wells located in Decatur, AL for

PFBS, PFHS, and PFOS in an effort to characterize on

- site groundwater conditions.

The 3M Environmental Laboratory prepared sample containers (250 mL high

- density polyethylene

bottles) which were shipped to Decatur, AL Weston personnel prior to field sampling. Sample

containers for each sampling location included a field sample, field sample duplicate, and field spike

samples. Each empty container was marked with a “fill to here” line to produce a final sample volume

of 200 mL. Containers designated for field matrix samples were fortified with an appropriate matrix

spike solution containing PFBS (linear), PFHS (linear), and PFOS (linear a nd branched) prior to being

sent to the field for sample collection. See section 8.8 of the report for field matrix spike levels.

Samples were prepared and analyzed according to the procedure defined in 3M Environmental

Laboratory method ETS -8- 044. 1 “M ethod of Analysis for the Determination of Perfluorinated

Compounds in Water by LC/MS/MS; Direct Injection Analysis ”. For several sampling locations,

improved FMS recoveries were seen for POFS using internal standard calibration. The internal

standard was added to the sample or sample dilution prior to analysis. Samples analyzed using

internal standard calibration for PFOS included: DAL GW 203L, DAL GW 227R, DAL GW 227L, DAL

GW 310R, DAL GW RW312R, DAL GW 317L, DAL GW 324L, DAL GW 328R, DAL GW 328L, DAL

GW RW331S, DAL GW GRS04, and the Trip Blank. Results for PFOS for sampling location DAL GW

GRS04 are not reported due to the FMS recovery being below <50%.

Table 1 summarizes the average PFBS, PFHS, and PFOS concentrations for the duplicate samples

col lected, trip blanks and equipment rinseate samples. Tables 10

- 29 summarize the individual sample

results and the associated FMS recoveries. All results for the quality control samples prepared and

analyzed with the samples are reported and discussed elsewhere in this report



## 4 **Test & Control Substances**

There was not a test substance or control substances in the classic sense of a GLP study. This study

was purely analytical in nature.

## 5 **Reference Substances**

Reference Substance PFBS

(predominantly linear) PFHS

(linear)



### Chemical Name Perfluorobutane sulfonate Perfluorohexane sulfonate

Chemical Formula 

### C 4 

### F 9 

### SO 3 - 

### K + 

### C 6 

### F 13 

### SO 3 - 

### Na +



### Identifier NA L- PFHXS

Use Target Analyte Reference

Standard Target Analyte Reference

Standard

Source 3M Wellington

Expiration Da te 1/10/2017 3/25/2018

Storage Conditions Frozen Frozen

Chemical Lot Number 41

- 2600

- 8442 -5 LPFHxSAM08

TCR Number TCR

- 121 TCR08

- 0018

Physical Description White Powder Crystalline

Purity 96.7% 100 %

Reference Substance PFOS

(linear + branched) PFOS

(li near + branched)



### Chemical Name Perfluorooctane sulfonate Perfluorooctane sulfonate

Chemical Formula 

### C 8 

### F 17 

### SO 3 - 

### K + 

### C 8 

### F 17 

### SO 3 - 

### K +



### Identifier CAS # 2795

- 39 -3 Br- PFOSK

Use FMS Reference Standard Target Analyte Reference

Standard

Source Sigma Aldrich Wellington

Expiration Date 2/4/2014 3/17/2014; 12/01/2014

Storage Conditions Room Temperature Frozen

Chemical Lot Number 1424328V brPFOSK0708; brPFOSK1111

TCR Number TCR11

- 0028 TCR11

- 0010; TCR11

- 0041

Physical Description White Powder Liquid

Purity 99.7% 99. 9%Reference Substance 13 C

8 - PFOS

(linear)



### Chemical Name Sodium

Perfluorooctanesulfonate

Chemical Formula 13 

### C 8 

### F 17 

### SO 3 - 

### Na +



### Identifier MPFC -C-0 112

Use Internal Standard

Source Wellington

Expiration Date 01/24/2015

Storage Conditions Frozen

Ch emical Lot Number 012312

TCR Number TCR12

- 0004

Physical Description Liquid

Purity 5 μ g/mL (1)

(1) Custom mixture of seven mass

- labeled ( 13 C) perfluoroalkylcarboxylic acids, two

mass

- labeled ( 13 C ) perfluoroalkylsulfonates and one mass

- labeled ( 13 C )

perfluoro -1- octanesulfonamide.



## 6 **Test System**

The test system for this study are ground water samples collected from wells located in Decatur, AL by

Weston Solutions, Inc. personnel. Samples for this study are “real world” samples, not dosed with a

specific lot o f test substance.

Table 2

. Sample Description Key Code.

**Example DAL GW 203L DB**  **120327**

String Number String Descriptor Example

**1 Sampling Location**  **DAL = Decatur, Alabama**

**2 Sample Type**  **GW= Ground w**  **ater**

**3 Well Identifier**  **Ex ample:**  **203 L** 

**4 Well Level**  **R = Residu u**  **m shallow water**

**- bearing zone** 

**L = Bedrock water**

**- bearing zo**  **ne**

**S = Epikarst middle water**

**- bearing zone** 

**5 Sample Type**  **0=primary sample**

**DB =duplicate sample** 

**LS = low spike**

**HS = high spike**

**6 Sampling Date**  **120327 = March 27, 2012** 

## 7 **Method Summary**

7.1 Methods

Analysis for all analytes was completed following 3M Environmental Laboratory method ETS -8- 044. 1

“ Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS;

Direct Injection Analysis ”.



## 7.2 Sample Collection

Samples were collected in 250 mL Nalgene TM 

### (high

- density polyethylene) bottles prepared at the 3M

Environmental Laboratory. Sample bottles were returned to the laboratory at ambient conditions on April

4, 2012. Samples were stored refrigerated at the laboratory after receipt. A set of laboratory prepared

Trip Blank and Trip Blank field matrix spikes were sent with the sample collection bottles.



## 7.3 Sample Preparation

Samples were prepared by removing an aliquot of the well mixed sample and placing it in an autovial for

analysis.

Samples that required dilution were prepared as follows:

1:10 dilutions were prepared by diluting 1mL sample with 9 mL of Milli Q water, and

1:100 dilutions were prepared by diluting 0.1mL sample with 9.9mL of Mi lli Q water.

Samples may also have been diluted further by varying the injection volume during analysis.

The following sampling locations were reanalyzed for PFOS using internal calibration by preparing as

follows:

A 5 mL aliquot of either a sample di lution or undiluted sample was fortified with an aliquot of a separate

internal standard spiking solution containing 13 

### C 8 

### - PFOS at a nomi nal concentration of 1 ng/mL : DAL GW

203L, DAL GW 227R, DAL GW 227L, DAL GW 310R, DAL GW RW312R, DAL GW 317L, DAL GW

324 L, DAL GW 328R, DAL GW 328L, DAL GW RW331S, DAL GW GRS04, and Trip Blank.



## 7.4 Analysis

All study samples and quality control samples were analyzed for PFBS, PFHS, and PFOS using high

performance liquid chromatography/ tandem mass spectrometry (HPLC/MS/MS). Detailed instrument

parameters, the liquid chromatography gradient program, and the specific mass transitions analyzed are

described in the raw data hard copies placed in the final data packet, and are briefly described below in

Table 3

, Table 4 and Table 5.

### Table 3 . Instrument Parameters

.

Instrument Name ETS Buster ETS Buster

Analytical Method Followed ETS

- 8

- 044.1 ETS

- 8

- 044.1

Analysis Date 4/11/12 4/30/12

Liquid Chromatograph Agilent 1100 Agilent 1100

Guard column Betasil C18 (4.6 mm X 100 mm), 5 μ Betasil C18 (4.6 mm X 100 mm), 5 μ

Analytical column Betasil C18 (4.6 mm X 100 mm), 5 μ Betasil C18 (4.6 mm X 100 mm), 5 μ

Injection Volume 2 or 10 μ L 25 μ L

Mass Spectrometer Applied Biosystems API 4 000 Applied Biosystems API 4 000

Ion Source Turbo Spray Turbo Spray

Electrode Turbo ion electrode Turbo ion electrode

Polarity Negative Negative

Software Analyst 1.4.2 Analyst 1.4.2



### Table 4. Liquid Chromatography Conditions

.

**Step**

**Number Total Time** 

**(min) Flow Rate** 

**(** μ

**L/min) Percent**  **A**

**(2 mM ammonium**

**acetate) Percent B** 

**(Methanol)**

**ETS**

**- 8** 

**- 044.1**  **Analysis**

0 0 750 97.0 3.0

1 2.0 750 97.0 3.0

2 14.5 750 5.0 95.0

3 15.5 750 5.0 95.0

4 16.5 750 97.0 3.0

5 20.0 750 97.0 3.0



### Table 5. Mass Transitions.

**Analyte Re**  **ference**

**Material**

**Structure Mass Transition** 

**Q1/Q3 Internal Standard**  **Mass Transition**

**Q1/Q3**

**PFBS Linear**  **299/80**

**NA NA**  **299/99**

**PFHS Linear**  **399/80**

**NA NA**  **399/99**

**PFOS Linear +** 

**Branched 499/80** 

**[ 13**  

### C **8** 

### ]PFOS **(1)** 

### 507/80 **499/99**

**499/130**

Dwell time was 75 msec ( 4/11 /12 analysis) or 5 0 msec ( 4/30/12 analysis) f or each transition. The individual transitions

were summed to produce a “total ion chromatogram” (TIC), which was used for quantitation.

NA = Not Applicable

(1) Internal standard was only used for the analysis of locations DAL GW 203L, DAL GW 227R, DAL GW 227L, DAL

GW 310R, DAL GW RW312R, DAL GW 317L, DAL GW 324L, DAL GW 328R, DAL GW 328L, DAL GW

RW331S, DAL GW GRS04, and Trip Blank for PFOS.

## 8 **Analytical Results**

8.1 Calibration

8.1.1 External Standard Cali bration

– 4/11/12 analysis:

Samples were analyzed against an external standard calibration curve. Calibration standards were

prepared by spiking known amounts of the stock solution containing the target analytes into Milli Q

water. A total of fifteen spiked standards ranging from 0.02 5 ng/mL to 100 ng/mL (nominal) were

prepared

. A quadratic, 1/x weighted, calibration curve of the peak area counts was used to fit the data

for each analyte. The data were not forced through zero during the fitting process. Calculating the

standard concentrations using the peak area confirmed accuracy of each curve point.

Each curve point was quantitated using the overall calibration curve and reviewed for accuracy. Method

calibration accuracy requirements of 100±25% (100±30% for the lowest curve point) were met for all

analytes. The correlation coefficient (r) was greater than 0.995 for PFBS, PFHS, and PFOS.

8.1.2 Internal Standard Calibration

– 4/30 /1 2 analysis:

Several sampling locations were re

- analyzed for PFOS against a matrix

- matched stable isotope internal

standard calibration curve. Calibration standards were prepared by spiking known amounts of the stock

solution containing the target analytes into a laboratory

- prepared synthetic groundwater containing

calcium and magnesium. The calibration standards contained an internal standard mix containing 13 

### C 8 

### - PFOS at a nominal concentration of 1 ng/mL. A total of sixteen spiked standards ranging from 0.025

ng/mL to 100 ng/mL were analyzed for PFOS. The PFOS reference standard used for the preparation

of the calibration standards contained both linear and branched isomers. A quadratic, 1/x weighted,

calibration curve of the ratio of the standard peak area counts over the internal standard peak area

counts was used to fit the data for each analyte. The data were not forced through zero during the fitting

process.

Each curve point was quantitated using the overall calibration curve and reviewed for accuracy. Method

calibration accuracy requirements of 100±25% (100±30% for the lowest curve point) were met for all

analytes. The correlation coefficient (r) was greater than 0.995 for PFOS.



## 8.2 System Suitability

A calibration standard was analyzed four times at the beginning of the analytical sequence to

demonstrate overall system suitability. The acceptance criteria of less than or equal to 5% relative

standard deviation (RSD) for peak area and retention time criteria of less than or equal to 2% RSD was

met for PFBS, PFHS, and PFOS for each analytical batch

.



## 8.3 Limit of Quant itation (LOQ)

The LOQ for this analysis is the lowest non

- zero calibration standard in the curve that meets linearity

and accuracy requirements and for which the area counts are at least twice those of the appropriate

blanks. The LOQs associated with the sample analysis are listed in the table below in Table 6

.

### Table 6 . Limit of Quantitation (LOQ).

**Analysis**

**Date Dilution**  **PFBS**

**LOQ, ng/mL PFHS** 

**LOQ, ng/mL PFOS** 

**LOQ, ng/mL**

4/11/12 1 0.0250 0.0250 0.0232

10 0.250 0.250 0.232

100 2.50 2.50 2.32

500 12.5 12.5 11.6

4/30 /1 2 1 NA NA 0.0278

10 NA NA 0.278

100 NA NA 2.78

NA = Not Applicable; analyte was not included in the analytical run

.



## 8.4 Continuing Calibration

During the course of each analytical sequence, continuing calibration verification samples (CCVs) were

analyzed to confirm that the instrument response and the initial calibration curve were still in control. All

CCVs met method criteria of 100% ± 25% for PFBS, PFHS, and PFOS

.



## 8.5 Blanks

Three types of blanks were prepared and analyzed with the samples: solvent blanks, trip blanks, and

equipment rinseate blanks. Solvent blank results were reviewed and used to evaluate method

performance to determine the LOQ for PFBS, PFHS, and PFOS. Trip blanks reflect the shipping and

sa mple collection conditons the sample bottles and samples experience. Equipment rinseate blanks

are aqueous samples that reflect the efficiency of equipment cleaning in the field between different

sample collections and are proof of no cross contamination of samples from the equipment.



## 8.6 Lab Control Spikes (LCSs)

Three lab control spike levels were prepared and analyzed in triplicate with each preparation set. LCSs

were prepared by spiking known amounts of the analyte into 10 mL of Milli Q water or synthe tic

groundwater to produce the desired concentration. The spiked water samples were then analyzed in

the same manner as the samples. The method acceptance criteria, average of LCS at each level

should be within 100% ± 20% with an RSD ≤20% , were met for all LCS samples

.

The following calculations were used to generate data in Table 7 for laboratory control spikes.

100% *

ion Concentrat Spike ion Concentrat Calculated Recovery Percent LC S

=

100% *

recovery LCS average replicates LCS deviation standard RSD LCS%

=

### Table 7. Laboratory Control Spike Recovery

.

ETS

- 8

- 044.1

Analyzed 4/11/12

External Standard

Calibration 

### PFBS **PFHS**

**Lab ID Spiked** 

**Concentration**

**(ng/mL) Calculated** 

**Concentration**

**(ng/mL) %Recovery**  **Spiked**

**Concentration**

**(ng/mL) Calculated** 

**Concentration**

**(ng/mL) %Recovery** 

LCS

- 120405

- 1 0.498 0.494 99.2 0.497 0.476 95.7

LCS

- 120405

- 2 0.498 0.471 94.6 0.497 0.478 96.2

LCS

- 120405

- 3 0.498 0.475 95.3 0.497 0.498 100

Average ± %RSD 96.4 % ± 2.6 % 97.3 % ± 2.4 %

LCS

- 120405

- 4 4.98 5.00 100 4.97 5.00 101

LCS

- 120405

- 5 4.98 4.99 100 4.97 4.92 98.9

LCS

- 120405

- 6 4.98 4.98 100 4.97 4.88 98.2

Average ± %RSD 100 % ± 0.0 % 99.4 % ± 1.5 %

LCS

- 120405

- 7 29.9 29.7 99.4 29.9 28.4 94.8

LCS

- 12040 5

- 8 29.9 29.2 97.6 29.9 27.6 92.2

LCS

- 120405

- 9 29.9 29.7 99.2 29.9 28.9 96.7

Average ± %RSD 98.7 % ± 1.0 % 94.6 % ± 2.4 %

ETS

- 8

- 044.1

Analyzed 4/11/12

External Standard

Calibration 

### PFOS (Linear + Branched)

**Lab ID Spiked** 

**Concentration**

**(ng/mL) Calculated** 

**Concentration**

**(ng/mL) %Recovery** 

LCS

- 120405

- 1 0.462 0.386 83.6

LCS

- 120405

- 2 0.462 0.390 84.3

LCS

- 120405

- 3 0.462 0.367 79.5

Average ± %RSD 82.5 % ± 3.1 %

LCS

- 120405

- 4 4.62 4.55 98.4

LCS

- 120405

- 5 4.62 4.48 97.0

LCS

- 120405

- 6 4.62 4.49 97.1

Average ± %R SD 97.5 % ± 0.80 %

LCS

- 120405

- 7 27.8 26.0 93.6

LCS

- 120405

- 8 27.8 26.4 95.0

LCS

- 120405

- 9 27.8 27.0 97.0

Average ± %RSD 95.2 % ± 1.8 %

### Table 7 continued. Laboratory Control Spike Recovery.

ETS

- 8

- 044.1

Analyzed 4/30 /12

Internal Standard

Calibration 

### PFOS **(Linear + Branched)**

**Lab ID Spiked** 

**Concentration**

**(ng/mL) Calculated** 

**Concentration**

**(ng/mL) %Recovery** 

LCS

- 120425

- 1 0.184 0.194 105

LCS

- 120425

- 2 0.184 0.187 102

LCS

- 120425

- 3 0.184 0.179 97.4

Average ± %RSD 101 % ± 3.8 %

LCS

- 120425

- 4 1.84 1.85 100

LCS

- 120425

- 5 1.84 1.86 101

LCS

- 120425

- 6 1.84 1.82 98.8

Average ± %RSD 99.9 % ± 1.1 %

LCS

- 120425

- 7 18.4 18.2 99.0

LCS

- 120425

- 8 18.4 17.2 93.5

LCS

- 120425

- 9 18.4 17.8 96.5

Average ± %RSD 96.3 % ± 2.9 %



## 8.7 Analytical Method Uncertainty

Analytical uncertainty is based on historical LCS data that is control charted and used to evaluate

method accuracy and precision. The method uncertainty is calculated following ETS

- 12

- 012.2. The

standard deviation is calculated for the set of accuracy results (in %) obtained for the LCS samples.

The expanded uncertainty is calculated by multiplying the standard deviation by a factor of 2, which

corresponds to a confidence level of 95%. The most recent 50 data points were used to generate the

method uncertainty values.

Table 8 . Analytical Uncertainty.

Analyte Calibration Standard Deviation Method Uncertainty

PFBS External 10.7 ± 21 %

PFHS External 8.69 ± 1 7 %

PFOS External 17.2 ± 34 %

PFOS Internal 8.33 ± 17 %



## 8.8 Field Matrix Spikes (FMS)

Field matrix s pikes were collected at each sampling point to verify that the analytical method is

applicable to the collected matrix. Field matrix spikes were generated by adding a measured volume of

field sample to a container spiked by the laboratory with PFBS (linear), PFHS (linear), and PFOS (linear

and branched) prior to shipping sample containers for sample collection. Field matrix spike recoveries

within method acceptance criteria of 100±30% confirm that “unknown” components in the sample matrix

do not significantly interfere with the extraction and analysis of the analytes of interest. Field matrix spike

concentrations must be 50% of the sample concentration to be considered an appropriate field spike.

Field matrix spikes are presented in section 9 of this rep ort.

### Table 9 . Field Matrix Spikes.

Sampling Location Spike Level PFBS

(ng/mL) PFHS

(ng/mL) PFOS

(ng/mL)

317L Low FMS 1.00 0.998 0.998

High FMS 10.0 9.98 9.98

220R, 220L, 328L Low FMS 10.0 9.98 9.98

High FMS 100 99.8 99.8

203L, 222R, 227L, 227R, 324L,

330L, RW327R, 328R Low FMS 100 99.8 99.8

High FMS 5 00 499 499

310R, 330R, RW331S Low FMS 5 00 499 499

High FMS 1 000 998 998

GRS04, RW312R, 335R Low FMS 1 0 00 998 998

High FMS 5 0 00 4990 4990

Trip Blank 1 Low FMS 10.0 9 .98 9.98

Mid FMS 100 99.8 99.8

High FMS 1000 998 998

100%*

on Concentrat Spike Dup.) Sample Field & Sample Field:ion Concentrat Average FMS ofion Concentrat (Sample Recovery FMS −

=



## 9 **Data Summary and Discussion**

The tables below summarize the sample results and field matrix spike recoveries for the sampling

locations as well as the Trip Blanks and rinseate blank. Results and average values are rounded to

three significant figures according to EPA rounding rules. Because of rounding, values may vary slightly

from those listed in the raw data. Field matrix spike recoveries meeting the method acceptance crite ria

of ± 30%, demonstrate that the method was appropriate for the given matrix and their respective

quantitative ranges

.

DAL GW 220R

– The sample/sample duplicate RPD was 26% for PFOS.

DAL GW 222R

– The sample/sample duplicate RPD was 25% for PFOS.

DA L GW 3 10 R

– The recovery of the LS for PFOS was 68.0% and 72.1% for the HS

. Since both FMS

spike levels were ap propriate as compared to the sample concentration, the analytical uncertainty has

been adjusted for PFOS to ± 32%.

DAL GW 330 R

– The recovery o f th e LS for PFOS was 78.3 % and 66.0% for the HS

. Both FMS spike

levels were ap propriate as compared to the sample concentration; however, no adjustment was made

to the analytical uncertainty since the recovery was within the analytical method uncertainty of ± 34%

for PFOS by external standard calibration. Sampling location did not contain and appropriate FMS

level for PFBS to access recovery.

DAL GW 330 L – The recovery of the LS for PFOS was 85.7 % and 66.6% for the HS

. Both FMS spike

levels were ap propriate as compared to the sample concentration; however, no adjustment was made

to the analytical uncertainty since the recovery was within the analytical method uncertainty of ± 34%

for PFOS by external standard calibration.

### DAL GW RW331S

– The recovery of the LS and HS for PFOS was 64.1 % and 69.1%, respectively

. Since both FMS samples were ap propriate as compared to the sample concentration, the analytical

uncertainty has been adjusted for PFOS ± 36 %.

DAL GW GRS04

– The recovery of the HS for PF O S was 4 5.9 %. Since this was the only appropriate

spike level, the sampling location was not reported due to the FMS recovery being below 50%.

### Table 10 . DAL GW 203L 1 20327

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/m L)**  **%Recovery Concentration** 

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 001 DAL

- GW

- 203L

- 0

- 120327 31.1 NA 199 NA 702 NA

GLP10

- 01

- 02

- 27

- 002 DAL

- GW

- 203L

- DB

- 120327 30.5 NA 205 NA 630 NA

GLP10

- 01

- 02

- 27

- 003 DAL

- GW

- 203L

- LS

- 120327 124 93.2 291 89.2 NA (1 ) 

### NA (1 )



### GLP10

- 01

- 02

- 27

- 004 DAL

- GW

- 203L

- HS

- 120327 507 95.2 656 91.0 1050 77.0

**Average Concentration (ng/mL) ±**  **%RPD 30.8**  **ng/mL ±**  **1 .9%**  **202 ng/mL**  **± 3.0**  **% 666**  **ng/mL ±**  **11 %** 

NA = Not Applicable

All s amples were diluted 10 0x.

PFBS and PFHS were analyzed on 4/11/12 by external standard calibration and PFOS was analyzed on 4/30/12 by internal standard calibration.

(1) Sample not analyzed due to inappropriate field matrix spike level.



### Table 11 . DAL GW 220R 1 20323

**PF BS**  **P F** 

**HS PF**  **OS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 005 DAL

- GW

- 220R

- 0

- 120323 6.66 NA 30.0 NA 37.3 NA

GLP10

- 01

- 02

- 27

- 006 DAL

- GW

- 220R

- DB

- 120323 7.13 NA 29.6 NA 28.6 NA

GLP10

- 01

- 02

- 27

- 007 DAL

- GW

- 220R

- LS

- 120323 15.6 87.1 41.0 NC 52.5 NC

GLP10

- 01

- 02

- 27

- 008 DAL

- GW

- 220R

- HS

- 120323 96.2 89.3 119 89.4 122 89.2

**Average Concentration (ng/mL) ±**  **%RPD 6.90**  **ng/mL ±**  **6.8 %**  **29.8 ng/mL**  **± 1.3**  **% 33.0**  **ng/mL ±**  **26 %**  **(1 )** 

NA = Not App licable

NC = Not Calculated; Endogenous sample concentration greater than 2x spike level.

All s amples were diluted 10x and analyzed on 4/11/12 by external standard calibration.

(1 ) Sample/sample duplicate RPD did not meet acceptance criteria of ≤ 20%.

### Ta ble 12 . DAL GW 220L 1 20323

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 009 DAL

- GW

- 220L

- 0

- 120323 8.73 NA 41.4 NA 52.5 NA

GLP10

- 01

- 02

- 27

- 010 DAL

- GW

- 220L

- DB

- 120323 8.31 NA 41.3 NA 46.4 NA

GLP10

- 01

- 02

- 27

- 011 DAL

- GW

- 220L

- LS

- 120323 17.6 90.8 51.0 NC 56.7 NC

GLP10

- 01

- 02

- 27

- 012 DAL

- GW

- 220L

- HS

- 120323 103 94.5 134 92.8 119 69.7

**Averag e Concentration (ng/mL)**  **± %RPD**  **8.52 ng/mL**  **± 4.9**  **% 41.4**  **ng/mL ±**  **0.24 %**  **49.5 ng/mL**  **± 12**  **%**

NA = Not Applicable

NC = Not Calculated; Endogenous sample concentration greater than 2x spike level.

All samples were diluted 10x and analyzed on 4/11/12 by external standard calibration.



### Table 13 . DAL GW 222R 1 20327

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 013 DAL

- GW

- 222R

- 0

- 120327 62.8 NA 402 NA 395 NA

GLP10

- 01

- 02

- 27

- 0 14 DAL

- GW

- 222R

- DB

- 120327 66.1 NA 411 NA 306 NA

GLP10

- 01

- 02

- 27

- 015 DAL

- GW

- 222R

- LS

- 120327 152 87.6 517 NC 444 NC

GLP10

- 01

- 02

- 27

- 016 DAL

- GW

- 222R

- HS

- 120327 526 92.3 862 91.3 701 70.2

**Average Concentration (ng/mL) ±**  **%RPD 64.5**  **ng/mL ±**  **5.1 %**  **407 ng/mL**  **± 2.2**  **% 35**  **1 ng/mL**  **± 25**  **% (1**  **)**

NA = Not Applicable

NC = Not Calculated; Endogenous sample concentration greater than 2x spike level.

All samples were diluted 100x and analyzed on 4/11/12 by external standard calibration.

(1) Sample/sample duplicate RPD did not meet accep tance criteria of ≤ 20%.

### Table 14

. DAL GW 227R 1 20316

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 017 DAL

- GW

- 227R

- 0

- 120316 14.3 NA 85.7 NA 642 NA

GLP10

- 01

- 02

- 27

- 018 DAL

- GW

- 227R

- DB

- 120316 15.3 NA 86.9 NA 590 NA

GLP10

- 01

- 02

- 27

- 019 DAL

- GW

- 227R

- LS

- 120316 111 96.2 179 92.9 NA (1 ) 

### NA (1 )



### GLP10

- 01

- 02

- 27

- 020 DAL

- GW

- 227R

- HS

- 120316 486 94.2 546 92.1 991 75.2

**Average Concent ration (ng/mL)**  **± %RPD**  **14.8 ng/mL**  **± 6.8**  **% 86.3**  **ng/mL ±**  **1.4 %**  **616 ng/mL**  **± 8.4**  **%**

NA = Not Applicable

All samples were diluted 100x.

PFBS and PFHS were analyzed on 4/11/12 by external standard calibration and PFOS was analyzed on 4/30/12 by internal standard ca libration.

(1) Sample not analyzed due to inappropriate field matrix spike level.



### Table 15

. DAL GW 227L 1 20316

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 021 DAL

- GW

- 227L

- 0

- 120316 224 NA 40.0 NA 222 NA

GLP10

- 01

- 02

- 27

- 022 DAL

- GW

- 227L

- DB

- 120316 216 NA 40.1 NA 217 NA

GLP10

- 01

- 02

- 27

- 023 DAL

- GW

- 227L

- LS

- 120316 310 NC 134 94.1 301 NC

GLP10

- 01

- 02

- 27

- 024 DAL

- GW

- 227L

- HS

- 120316 65 8 87.6 487 89.6 595 75.3

**Average Concentration (ng/mL) ±**  **%RPD 220**  **ng/mL ±**  **3 .6%**  **40.1 ng/mL**  **± 0.25**  **% 220**  **ng/mL ±**  **2.3 %** 

NA = Not Applicable

NC = Not Calculated; Endogenous sample concentration greater than 2x spike level.

All samples were diluted 100x.

PFBS and PFHS were analyzed on 4/11/12 by external standard calibration and PFOS was analyzed on 4/30/12 by internal standard cali bration.

### Table 16 . DAL GW 310R 120320

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL )**  **%Recovery Concentration** 

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 025 DAL

- GW

- 310R

- 0

- 120320 989 NA 336 NA 697 NA

GLP10

- 01

- 02

- 27

- 026 DAL

- GW

- 310R

- DB

- 120320 982 NA 320 NA 644 NA

GLP10

- 01

- 02

- 27

- 027 DAL

- GW

- 310R

- LS

- 120320 1360 74.9 7 60 86.6 1010 68.0 (1 )



### GLP10

- 01

- 02

- 27

- 028 DAL

- GW

- 310R

- HS

- 120320 1840 85.5 1180 85.4 1390 72.1

**Average Concentration (ng/mL) ±**  **%RPD 986**  **ng/mL ±**  **0.71 %**  **328 ng/mL**  **± 4.9**  **% 671**  **ng/mL ±**  **7.9 %**  **(2)**

NA = Not Applicable

All samples were diluted 100x.

PFBS and PFHS were analyzed on 4/11/12 by external standard calibration and PFOS was analyzed on 4/30/12 by internal standard calibration.

(1 ) FMS did not meet acceptance criteria of 100 ± 30%.

(2 ) PFOS method uncertainty is adjusted to ± 32%.



### Table 17 . DAL GW RW 312R 1 20328

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 029 DAL

- GW

- RW312R

- 0

- 120328 739 NA 495 NA 1030 NA

GLP10

- 01

- 02

- 27

- 0 30 DAL

- GW

- RW312R

- DB

- 120328 740 NA 492 NA 954 NA

GLP10

- 01

- 02

- 27

- 031 DAL

- GW

- RW312R

- LS

- 120328 1620 88.1 1370 87.8 1740 74.9

GLP10

- 01

- 02

- 27

- 032 DAL

- GW

- RW312R

- HS

- 120328 5260 90.4 5100 92.3 4930 78.9

**Average Concentration (ng/mL) ±**  **%RPD 740**  **ng/mL ±**  **0.14 %**  **494 ng/mL**  **± 0.61**  **% 992**  **ng/mL ±**  **7.7 %** 

NA = Not Applicable

The s ample, sample duplicate, and LS were diluted 100 x. The HS was diluted 500x

PFBS and PFHS were analyzed on 4/11/12 by external standard calibration and PFOS was analyzed on 4/30/12 by internal stand ard calibration.

### Table 18 . DAL GW 317L 1 20320

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 033 DAL

- GW

- 317L

- 0

- 120320 <0.0250 NA 0.0865 NA 0.235 NA

GLP10

- 01

- 02

- 27

- 034 DAL

- GW

- 317L

- DB

- 120320 <0.0250 NA 0.0889 NA 0.225 NA

GLP10

- 01

- 02

- 27

- 035 DAL

- GW

- 317L

- LS

- 120320 0.936 93.6 0.957 87.1 1.06 83.2

GLP10

- 01

- 02

- 27

- 036 DAL

- GW

- 317L

- HS

- 120320 9.67 96.7 8.79 87.2 NR (1 ) 

### NR (1 )



### Av **erage Concentration (ng/mL) ±**  **%RPD <0.025**  **0 ng/mL 0.0877**  **ng/mL ±**  **2.7 %**  **0.230 ng/mL**  **± 4.3**  **%**

NA = Not Applicable

PFBS and PFHS were analyzed on 4/11/12 by external standard calibration and PFOS was analyzed on 4/30/12 by internal standard calibration.

(1) NR = No t Reportable; Incorrect sample was analyzed.



### Table 19 . DAL GW 324L 1 20314

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 037 DAL

- GW

- 324L

- 0

- 120314 81.1 NA 87.9 NA 193 NA

GLP10

- 01

- 02

- 27

- 038 DAL

- GW

- 324L

- DB

- 120314 83.0 NA 88.8 NA 194 NA

GLP10

- 01

- 02

- 27

- 039 DAL

- GW

- 324L

- LS

- 120314 176 94.0 181 92.8 303 110

GLP10

- 01

- 02

- 27

- 040 DAL

- GW

- 324L

- HS

- 120314 512 86.0 497 81.9 613 84.1

**Av erage Concentration (ng/mL)**  **± %RPD**  **82.1 ng/mL**  **± 2.3**  **% 88.4**  **ng/mL ±**  **1.0 %**  **194 ng/mL**  **± 0.**  **52 %** 

NA = Not Applicable

All samples were diluted 100x.

PFBS and PFHS were analyzed on 4/11/12 by external standard calibration and PFOS was analyzed on 4/30/12 by intern al standard calibration.

### Table 20 . DAL GW RW 327R 1 20323

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 041 DAL

- G W

- RW327R

- 0

- 120323 63.2 NA 149 NA 337 NA

GLP10

- 01

- 02

- 27

- 042 DAL

- GW

- RW327R

- DB

- 120323 61.0 NA 143 NA 353 NA

GLP10

- 01

- 02

- 27

- 043 DAL

- GW

- RW327R

- LS

- 120323 157 94.9 234 88.2 407 NC

GLP10

- 01

- 02

- 27

- 044 DAL

- GW

- RW327R

- HS

- 120323 517 91.0 605 92.0 721 75.4

**Average C oncentration (ng/mL)**  **± %RPD**  **62.1 ng/mL**  **± 3.5**  **% 146**  **ng/mL ±**  **4.1 %**  **345 ng/mL**  **± 4.6**  **%**

NA = Not Applicable

NC = Not Calculated; Endogenous sample concentration greater than 2x spike level.

All s amples were diluted 100x and analyzed on 4/11/12 by external standard calibration.



### Table 21 . DAL GW 328R 1 20314

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 045 DAL

- GW

- 328R

- 0

- 120314 3 0.8 NA 55.5 NA 139 NA

GLP10

- 01

- 02

- 27

- 046 DAL

- GW

- 328R

- DB

- 120314 31.2 NA 58.4 NA 149 NA

GLP10

- 01

- 02

- 27

- 047 DAL

- GW

- 328R

- LS

- 120314 123 92.0 154 97.2 242 98.2

GLP10

- 01

- 02

- 27

- 048 DAL

- GW

- 328R

- HS

- 120314 475 88.8 497 88.2 529 77.2

**Average Concentration (ng/mL) ±**  **%RPD 31.0**  **ng/mL ±**  **1.3 %**  **57.0 ng/mL**  **± 5.1**  **% 144**  **ng/mL ±**  **6.9 %** 

NA = Not Applicable

All samples were diluted 100x.

PFBS and PFHS were analyzed on 4/11/12 by external standard calibration and PFOS was analyzed on 4/30/12 by internal standard calibration.

### Table 22 . DAL GW 328L 1 20314

**PF BS**  **(1)** 

### PF **HS (1)**  

### PF **OS (2)** 



### 3M LIMS ID **Description Concentration** 

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 049 DAL

- GW

- 328L

- 0

- 120314 47.1 NA 1 9.1 NA 1.12 NA

GLP10

- 01

- 02

- 27

- 050 DAL

- GW

- 328L

- DB

- 120314 48.7 NA 18.3 NA 1.15 NA

GLP10

- 01

- 02

- 27

- 051 DAL

- GW

- 328L

- LS

- 120314 59.5 NC 27.7 90.2 9.16 80.4

GLP10

- 01

- 02

- 27

- 052 DAL

- GW

- 328L

- HS

- 120314 140 92.1 112 93.5 77.5 76.5

**Average Concentration (ng/mL) ±**  **%R PD**  **47.9 ng/mL**  **± 3.3**  **% 18.7**  **ng/mL ±**  **4.3 %**  **1.14 ng/mL**  **± 2.6**  **%**

NA = Not Applicable

NC = Not Calculated; Endogenous sample concentration greater than 2x spike level.

(1) All s amples were diluted 10x and analyzed on 4/11/12 by external standard calibration.

(2) The s ample, sample duplicate, and LS were not diluted while the HS was diluted 10x. Samples were analyzed on 4/30/12 by internal standard calibration.



### Table 23 . DAL GW 330R 1 20314

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Conce ntration** 

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 053 DAL

- GW

- 330R

- 0

- 120314 2120 NA 187 NA 458 NA

GLP10

- 01

- 02

- 27

- 054 DAL

- GW

- 330R

- DB

- 120314 2110 NA 183 NA 465 NA

GLP10

- 01

- 02

- 27

- 055 DAL

- GW

- 330R

- LS

- 120314 2430 NC 606 84.4 852 78.3

GLP10

- 01

- 02

- 27

- 056 DAL

- GW

- 330R

- HS

- 120314 2940 NC 1070 88.7 1120 66.0 (2 )



### Average Concentration (ng/mL) **± %RPD**  **2120 ng/mL**  **± 0.47**  **% (1**  **)** 

### 185 **ng/mL ±**  **2.2 %**  **462 ng/mL**  **± 1.5**  **%**

NA = Not Applicable

NC = Not Calculated; Endogenous sample concentration greater than 2x spike level.

All s amples were diluted 100x and analyzed on 4/11/12 by external standard calibration.

(1) Sampling location did not have an appropriate FMS spike level for PFBS to access accuracy

. (2) FMS did not meet acceptance criteria of 100 ± 30%. No adjustment was made to the analytical uncertainty since the recovery was within the analytical method uncertainty of ± 34% for

PFOS by external standard calibration.

### Table 24 . DAL GW 330L 1 20314

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 057 DAL

- GW

- 330L

- 0

- 120314 388 NA 354 NA 135 NA

GLP10

- 01

- 02

- 27

- 058 DAL

- GW

- 330L

- DB

- 120314 405 NA 363 NA 156 NA

GLP10

- 01

- 02

- 27

- 059 DAL

- GW

- 330L

- LS

- 120314 484 NC 438 NC 231 85.7

GLP10

- 01

- 02

- 27

- 060 DAL

- GW

- 330L

- HS

- 120314 851 90.9 810 90.5 478 66.6 (1 )



### Average Concentration (ng/mL) **± %RPD**  **397 ng/mL**  **± 4.3**  **% 359**  **ng/mL ±**  **2.5 %**  **146 ng/mL**  **± 1**  **4 %** 

NA = Not Appl icable

NC = Not Calculated; Endogenous sample concentration greater than 2x spike level.

All s amples were diluted 100x and analyzed on 4/11/12 by external standard calibration.

(1) FMS did not meet acceptance criteria of 100 ± 30%. No adjustment was made to t he analytical uncertainty since the recovery was within the analytical method uncertainty of ± 34% for

PFOS by external standard calibration.



### Table 25 . DAL GW RW 331S 1 20328

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 061 DAL

- GW

- RW331S

- 0

- 120328 1010 NA 477 NA 832 NA

GLP10

- 01

- 02

- 27

- 062 DAL

- GW

- RW331S

- DB

- 120328 984 NA 468 NA 868 NA

GLP10

- 01

- 02

- 27

- 063 DAL

- GW

- RW331S

- LS

- 120 328 1440 88.6 939 93.5 1170 64.1 (1 )



### GLP10

- 01

- 02

- 27

- 064 DAL

- GW

- RW331S

- HS

- 120328 1790 79.3 1260 78.9 1540 69.1 (1 )



### Average Concentration (ng/mL) **± %RPD**  **997 ng/mL**  **± 2.6**  **% 473**  **ng/mL ±**  **1.9 %**  **850 ng/mL**  **± 4.2**  **% (2)** 

NA = Not Applicable

All samples were diluted 1 00x.

PFBS and PFHS were analyzed on 4/11/12 by external standard calibration and PFOS was analyzed on 4/30/12 by internal standard calibration.

(1) FMS did not meet acceptance criteria of 100 ± 30%.

(2) PFOS method uncertainty is adjusted to ± 36%.

### Table 26 . DAL GW 335R 1 20322

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 065 DAL

- GW

- 335R

- 0

- 120322 859 NA 1620 NA 3480 NA

GLP10

- 01

- 02

- 27

- 066 DAL

- GW

- 335R

- DB

- 120322 868 NA 1580 NA 3720 NA

GLP10

- 01

- 02

- 27

- 067 DAL

- GW

- 335R

- LS

- 120322 1770 90.7 2610 NC 4520 NC

GLP10

- 01

- 02

- 27

- 068 DAL

- GW

- 335R

- HS

- 120322 5410 90.9 6000 88.2 7750 83.2

**Average Concentration (ng/mL) ±**  **%RPD 864**  **ng/mL ±**  **1 .0**  **% 1600**  **ng/mL ±**  **2.5 %**  **3600 ng/mL**  **± 6.7**  **%**

NA = Not Applicable

NC = Not Calculated; Endogenous sample concentration greater than 2x spike level.

The sample, sample duplicate, and LS were diluted 100x. The HS was diluted 500x

All samples were analyzed on 4/11/12 by external standard calibration.



### Table 27 . DAL GW GRS04 1 20328

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 0 69 DAL

- GW

- GRS04

- 0

- 120328 2320 NA 8350 NA

Not Reportable GLP10

- 01

- 02

- 27

- 070 DAL

- GW

- GRS04

- DB

- 120328 2270 NA 8210 NA

GLP10

- 01

- 02

- 27

- 071 DAL

- GW

- GRS04

- LS

- 120328 3180 88.5 9220 NC

GLP10

- 01

- 02

- 27

- 072 DAL

- GW

- GRS04

- HS

- 120328 6730 88.7 12000 74.5

**Average Conc entration (ng/mL)**  **± %RPD**  **2300 ng/mL**  **± 2.2**  **% 8280**  **ng/mL ±**  **1.7 %**  **NR (1**  **)**

NA = Not Applicable

NC = Not Calculated; Endogenous sample concentration greater than 2x spike level.

The sample, sample duplicate, and LS were diluted 100x. The HS was diluted 500x

PF BS and PFHS were analyzed on 4/11/12 by external standard calibration and PFOS was analyzed on 4/30/12 by internal standard c alibration.

(1) NR = Not Reportable; Sampling location not reportable due to the FMS recovery being below 50 %.

### Table 28 . Trip Blank 1

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 27

- 074 DAL

- GW

- TRIP01

- 0

- (1) 

### <0.0250 NA <0.0250 NA <0.0232 NA

GLP10

- 01

- 02

- 2 7

- 075 DAL

- GW

- TRIP01

- LS

- (1) 

### 9.82 98.2 9.46 94.8 8.49 85.1

GLP10

- 01

- 02

- 27

- 076 DAL

- GW

- TRIP01

- MS

- (1) 

### 92.7 92.7 91.6 91.8 78.9 79.1

GLP10

- 01

- 02

- 27

- 077 DAL

- GW

- TRIP01

- HS

- (1) 

### 982 98.2 952 95.4 826 82.8

GLP10

- 01

- 02

- 27

- 074 DAL

- GW

- TRIP01

- 0

- (2) 

### NA NA NA NA <0.0 278 NA

GLP10

- 01

- 02

- 27

- 075 DAL

- GW

- TRIP01

- LS

- (2) 

### NA NA NA NA 8.38 84.0

GLP10

- 01

- 02

- 27

- 076 DAL

- GW

- TRIP01

- MS

- (2) 

### NA NA NA NA 71.8 71.9

GLP10

- 01

- 02

- 27

- 077 DAL

- GW

- TRIP01

- HS

- (2) 

### NA NA NA NA 804 80.6

NA = Not Applicable

(1) Sample and LS undiluted, MS diluted 10x, HS diluted 100x and analyzed on 4/11/12 by external standard calibration

. (2) Sample and LS undiluted, MS diluted 10x, HS diluted 100x and analyzed on 4/30/12 by internal standard calibration.



### Table 29

. Equipment Rinseate Blank

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 27

- 073 DAL

- GW

- 203L

- RB

- 120327 <0.0250 <0.0250 <0.0232

The s ample was analyzed undiluted on 4/11/12 by external standard ca libration.

## 10 **Conclusion**

Laboratory control spikes and field matrix spikes were used to determine the analytical method

accuracy and precision for PFBS, PFHS, and PFOS. A nalysis was successfully completed following

3M Environmental Laboratory method ETS -8-0 44. 1 described herein.



## 11 **Data/Sample Retention**

All remaining samples and associated project data (hardcopy and electronic) will be archived according

to 3M Environmental Laboratory standard operating procedures.



## 12 **Attachments**

Attachment A: Protocol Amendment 2 7 (General Project Outline)

Attachment B: Representative Chromatograms and Calibration Curves

Attachment C: Analytical Method

Attachment D: Method DeviationGLP10-01-02 ; Interi m Repor t 2 7

Analysi s o f PFBS

, PFHS

, an d PFO S i n Groundwate r

Decatur

, A L

- 1s

t Quarte r 201 2 Samplin g



## 13 Signatures

Clesto n Lange

, Ph.D.

, 3 M Principa l Analytica l Investigato r ' Dat e

Willia m K

. Reagen

, Ph.D.

, 3 M Environmenta l Laborator y Departmen t Manage r Dat e

Gar y Hohensiein^ponso r Representativ e Dat e

Jaisimh a Kfeiari

, Stud y Directo r ' / Dat e

Pag e 3 3 o f 11 4

# A TTACHMENT 

# A: P ROTOCOL 

# A MENDMENT

# A TTACHMENT 

# B: R EPRESENTATIVE 

# S AMPLE 

# C HROMATOGRAMS AND



# C ALIBRATION 

# C URVE 

# ( S 

# )b120411a.rdb (PFOS): "Quadratic" Regression ("1 / x" weighting): y = -1.91e+003 x^2 + 5.73e+005 x + 1.68e+004 (r = 0.9996)0 5 10 15 20 25 30 35 40 45 50 55 60 65 70 75 80 85 90Concentration, ng/mL0.02.0e64.0e66.0e68.0e61.0e71.2e71.4e71.6e71.8e72.0e72.2e72.4e72.6e72.8e73.0e73.2e73.4e73.6e73.8e7

Area, countsWorkstation: ETSBUSTERBatch Name: b120411a.dab*Data worked up by KJUPage 1 of 1Printing Time: 10:01:44 AMPrinting Date: Tuesday, April 17, 2012b120411a.rdb (PFHS): "Quadratic" Regression ("1 / x" weighting): y = -3.87e+003 x^2 + 5.26e+005 x + 5.52e+003 (r = 0.9993)0 5 10 15 20 25 30 35 40 45 50 55 60 65 70 75 80 85 90 95 100Concentration, ng/mL0.01.0e62.0e63.0e64.0e65.0e66.0e67.0e68.0e69.0e61.0e71.1e71.2e71.3e71.4e71.5e71.6e71.7e71.8e71.9e72.0e72.1e72.2e72.3e72.4e72.5e72.6e7

Area, countsWorkstation: ETSBUSTERBatch Name: b120411a.dab*Data worked up by KJUPage 1 of 1Printing Time: 9:59:28 AMPrinting Date: Tuesday, April 17, 2012b120411a.rdb (PFBS): "Quadratic" Regression ("1 / x" weighting): y = -3.15e+003 x^2 + 4.91e+005 x + 8.42e+003 (r = 0.9992)0 5 10 15 20 25 30 35 40 45 50 55 60 65 70 75 80 85 90 95 100Concentration, ng/mL0.01.0e62.0e63.0e64.0e65.0e66.0e67.0e68.0e69.0e61.0e71.1e71.2e71.3e71.4e71.5e71.6e71.7e71.8e71.9e72.0e72.1e72.2e72.3e72.4e72.5e72.6e72.7e7

Area, countsWorkstation: ETSBUSTERBatch Name: b120411a.dab*Data worked up by KJUPage 1 of 1Printing Time: 10:01:26 AMPrinting Date: Tuesday, April 17, 2012Sample Name: "b120411a020" Sample ID: "11012-10-11" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "10 ng/mL FC std in Milli Q Water" Annotation: ""Sample Index: 20Sample Type: StandardConcentration: 10.0 ng/mLCalculated Conc: 10.2 ng/mLAcq. Date: 4/11/2012Acq. Time: 7:29:04 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 4697398 countsHeight: 1.28e+006 cpsStart Time: 13.5 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e61.25e6

Intensity, cps13.5713.43Sample Name: "b120411a028" Sample ID: "Solvent Blank" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "Milli Q Water TN12-0132" Annotation: ""Sample Index: 28Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/11/2012Acq. Time: 10:17:34 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 1483 countsHeight: 5.08e+002 cpsStart Time: 13.5 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min01002003004005006007008009001000110012001300140015001600170018001900

Intensity, cpsSample Name: "b120411a029" Sample ID: "LCS-120405-1" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "0.5ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 0.498 ng/mLCalculated Conc: 0.494 ng/mLAcq. Date: 4/11/2012Acq. Time: 10:38:41 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 250124 countsHeight: 6.93e+004 cpsStart Time: 13.5 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e4

Intensity, cps13.5714.2415.4715.8711.7514.62Sample Name: "b120411a032" Sample ID: "LCS-120405-4" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "5ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 4.98 ng/mLCalculated Conc: 5.00 ng/mLAcq. Date: 4/11/2012Acq. Time: 11:41:54 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 2384031 countsHeight: 6.59e+005 cpsStart Time: 13.5 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps13.58*** Buster J2930203Printing Time: 2:28:03 PMPrinting Date: Friday, May 11, 2012Page 1 of 23Sample Name: "b120411a035" Sample ID: "LCS-120405-7" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "30ppb LCS" Annotation: ""Sample Index: 35Sample Type: QCConcentration: 29.9 ng/mLCalculated Conc: 29.7 ng/mLAcq. Date: 4/12/2012Acq. Time: 12:45:10 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 11807788 countsHeight: 3.02e+006 cpsStart Time: 13.5 minEnd Time: 14.0 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e6

Intensity, cps13.57Sample Name: "b120411a138" Sample ID: "GLP10-01-02-27-001" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-203L-0-" Annotation: ""Sample Index: 138Sample Type: UnknownConcentration: N/ACalculated Conc: 31.1 ng/mLAcq. Date: 4/13/2012Acq. Time: 12:58:18 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 160914 countsHeight: 4.32e+004 cpsStart Time: 13.3 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e44.2e4

Intensity, cpsSample Name: "b120411a141" Sample ID: "GLP10-01-02-27-004" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-203L-HS-" Annotation: ""Sample Index: 141Sample Type: UnknownConcentration: N/ACalculated Conc: 507. ng/mLAcq. Date: 4/13/2012Acq. Time: 2:01:29 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 2417512 countsHeight: 6.78e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps13.52Sample Name: "b120411a142" Sample ID: "GLP10-01-02-27-005" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-220R-0-" Annotation: ""Sample Index: 142Sample Type: UnknownConcentration: N/ACalculated Conc: 6.66 ng/mLAcq. Date: 4/13/2012Acq. Time: 2:22:32 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 333863 countsHeight: 9.23e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e4

Intensity, cps13.52*** Buster J2930203Printing Time: 2:28:03 PMPrinting Date: Friday, May 11, 2012Page 2 of 23Sample Name: "b120411a145" Sample ID: "GLP10-01-02-27-008" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-220R-HS-" Annotation: ""Sample Index: 145Sample Type: UnknownConcentration: N/ACalculated Conc: 96.2 ng/mLAcq. Date: 4/13/2012Acq. Time: 3:25:46 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 4439899 countsHeight: 1.23e+006 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e6

Intensity, cps13.52Sample Name: "b120411a149" Sample ID: "GLP10-01-02-27-009" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-220L-0-" Annotation: ""Sample Index: 149Sample Type: UnknownConcentration: N/ACalculated Conc: 8.73 ng/mLAcq. Date: 4/13/2012Acq. Time: 4:50:17 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 434541 countsHeight: 1.17e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e51.15e5

Intensity, cps13.52Sample Name: "b120411a152" Sample ID: "GLP10-01-02-27-012" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-220L-HS-" Annotation: ""Sample Index: 152Sample Type: UnknownConcentration: N/ACalculated Conc: 103. ng/mLAcq. Date: 4/13/2012Acq. Time: 5:53:34 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 4728707 countsHeight: 1.31e+006 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e61.25e6

Intensity, cps13.52Sample Name: "b120411a153" Sample ID: "GLP10-01-02-27-013" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-222R-0-" Annotation: ""Sample Index: 153Sample Type: UnknownConcentration: N/ACalculated Conc: 62.8 ng/mLAcq. Date: 4/13/2012Acq. Time: 6:14:42 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 315333 countsHeight: 8.55e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e4

Intensity, cps13.52*** Buster J2930203Printing Time: 2:28:03 PMPrinting Date: Friday, May 11, 2012Page 3 of 23Sample Name: "b120411a156" Sample ID: "GLP10-01-02-27-016" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-222R-HS-" Annotation: ""Sample Index: 156Sample Type: UnknownConcentration: N/ACalculated Conc: 526. ng/mLAcq. Date: 4/13/2012Acq. Time: 7:17:53 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 2503463 countsHeight: 6.98e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps13.52Sample Name: "b120411a160" Sample ID: "GLP10-01-02-27-017" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-227R-0-" Annotation: ""Sample Index: 160Sample Type: UnknownConcentration: N/ACalculated Conc: 14.3 ng/mLAcq. Date: 4/13/2012Acq. Time: 8:42:27 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 78585 countsHeight: 2.13e+004 cpsStart Time: 13.4 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e42.0e42.1e4

Intensity, cpsSample Name: "b120411a163" Sample ID: "GLP10-01-02-27-020" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-227R-HS-" Annotation: ""Sample Index: 163Sample Type: UnknownConcentration: N/ACalculated Conc: 486. ng/mLAcq. Date: 4/13/2012Acq. Time: 9:45:41 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 2320431 countsHeight: 6.38e+005 cpsStart Time: 13.4 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e5

Intensity, cps13.52Sample Name: "b120411a164" Sample ID: "GLP10-01-02-27-021" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-227L-0-" Annotation: ""Sample Index: 164Sample Type: UnknownConcentration: N/ACalculated Conc: 224. ng/mLAcq. Date: 4/13/2012Acq. Time: 10:06:53 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 1093556 countsHeight: 2.93e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e5

Intensity, cps13.52*** Buster J2930203Printing Time: 2:28:03 PMPrinting Date: Friday, May 11, 2012Page 4 of 23Sample Name: "b120411a167" Sample ID: "GLP10-01-02-27-024" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-227L-HS-" Annotation: ""Sample Index: 167Sample Type: UnknownConcentration: N/ACalculated Conc: 658. ng/mLAcq. Date: 4/13/2012Acq. Time: 11:10:08 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 3099860 countsHeight: 8.57e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e5

Intensity, cps13.52Sample Name: "b120411a171" Sample ID: "GLP10-01-02-27-025" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-310R-0-" Annotation: ""Sample Index: 171Sample Type: UnknownConcentration: N/ACalculated Conc: 989. ng/mLAcq. Date: 4/14/2012Acq. Time: 12:34:33 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 4554659 countsHeight: 1.23e+006 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e6

Intensity, cps13.52Sample Name: "b120411a174" Sample ID: "GLP10-01-02-27-028" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-310R-HS-" Annotation: ""Sample Index: 174Sample Type: UnknownConcentration: N/ACalculated Conc: 1840. ng/mLAcq. Date: 4/14/2012Acq. Time: 1:37:56 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 7973133 countsHeight: 2.11e+006 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e62.0e6

Intensity, cps13.52Sample Name: "b120411a175" Sample ID: "GLP10-01-02-27-029" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-RW312R-0-" Annotation: ""Sample Index: 175Sample Type: UnknownConcentration: N/ACalculated Conc: 739. ng/mLAcq. Date: 4/14/2012Acq. Time: 1:59:00 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 3464830 countsHeight: 9.41e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e5

Intensity, cps13.52*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 5 of 23Sample Name: "b120411a178" Sample ID: "GLP10-01-02-27-032" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-RW312R-HS-" Annotation: ""Sample Index: 178Sample Type: UnknownConcentration: N/ACalculated Conc: 5260. ng/mLAcq. Date: 4/14/2012Acq. Time: 3:02:18 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 4824687 countsHeight: 1.29e+006 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e61.25e6

Intensity, cps13.51Sample Name: "b120411a182" Sample ID: "GLP10-01-02-27-033" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-317L-0-" Annotation: ""Sample Index: 182Sample Type: UnknownConcentration: N/ACalculated Conc: 0.00209 ng/mLAcq. Date: 4/14/2012Acq. Time: 4:26:41 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 9447 countsHeight: 2.35e+003 cpsStart Time: 13.4 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0100200300400500600700800900100011001200130014001500160017001800190020002100220023002400

Intensity, cpsSample Name: "b120411a185" Sample ID: "GLP10-01-02-27-036" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-317L-HS-" Annotation: ""Sample Index: 185Sample Type: UnknownConcentration: N/ACalculated Conc: 9.67 ng/mLAcq. Date: 4/14/2012Acq. Time: 5:30:02 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 4460815 countsHeight: 1.21e+006 cpsStart Time: 13.4 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e6

Intensity, cps13.5213.38Sample Name: "b120411a186" Sample ID: "GLP10-01-02-27-037" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-324L-0-" Annotation: ""Sample Index: 186Sample Type: UnknownConcentration: N/ACalculated Conc: 81.1 ng/mLAcq. Date: 4/14/2012Acq. Time: 5:51:13 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 404183 countsHeight: 1.02e+005 cpsStart Time: 13.3 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e5

Intensity, cps13.52*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 6 of 23Sample Name: "b120411a189" Sample ID: "GLP10-01-02-27-040" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-324L-HS-" Annotation: ""Sample Index: 189Sample Type: UnknownConcentration: N/ACalculated Conc: 512. ng/mLAcq. Date: 4/14/2012Acq. Time: 6:54:28 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 2438207 countsHeight: 6.79e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps13.5213.38Sample Name: "b120411a193" Sample ID: "GLP10-01-02-27-041" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-RW327R-0-" Annotation: ""Sample Index: 193Sample Type: UnknownConcentration: N/ACalculated Conc: 63.2 ng/mLAcq. Date: 4/14/2012Acq. Time: 8:19:04 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 317186 countsHeight: 8.59e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e4

Intensity, cps13.52Sample Name: "b120411a196" Sample ID: "GLP10-01-02-27-044" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-RW327R-HS-" Annotation: ""Sample Index: 196Sample Type: UnknownConcentration: N/ACalculated Conc: 517. ng/mLAcq. Date: 4/14/2012Acq. Time: 9:22:24 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 2463440 countsHeight: 6.97e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps13.51Sample Name: "b120411a197" Sample ID: "GLP10-01-02-27-045" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-328R-0-" Annotation: ""Sample Index: 197Sample Type: UnknownConcentration: N/ACalculated Conc: 30.8 ng/mLAcq. Date: 4/14/2012Acq. Time: 9:43:31 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 159292 countsHeight: 4.21e+004 cpsStart Time: 13.3 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e4

Intensity, cps13.52*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 7 of 23Sample Name: "b120411a200" Sample ID: "GLP10-01-02-27-048" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-328R-HS-" Annotation: ""Sample Index: 200Sample Type: UnknownConcentration: N/ACalculated Conc: 475. ng/mLAcq. Date: 4/14/2012Acq. Time: 10:46:59 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 2271052 countsHeight: 6.52e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps13.51Sample Name: "b120411a204" Sample ID: "GLP10-01-02-27-049" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-328L-0-" Annotation: ""Sample Index: 204Sample Type: UnknownConcentration: N/ACalculated Conc: 47.1 ng/mLAcq. Date: 4/14/2012Acq. Time: 12:11:25 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 2251791 countsHeight: 5.79e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e5

Intensity, cps13.51Sample Name: "b120411a207" Sample ID: "GLP10-01-02-27-052" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-328L-HS-" Annotation: ""Sample Index: 207Sample Type: UnknownConcentration: N/ACalculated Conc: 140. ng/mLAcq. Date: 4/14/2012Acq. Time: 1:14:51 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 6278231 countsHeight: 1.68e+006 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e6

Intensity, cps13.51Sample Name: "b120411a208" Sample ID: "GLP10-01-02-27-053" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-330R-0-" Annotation: ""Sample Index: 208Sample Type: UnknownConcentration: N/ACalculated Conc: 2120. ng/mLAcq. Date: 4/14/2012Acq. Time: 1:35:58 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 8997990 countsHeight: 2.31e+006 cpsStart Time: 13.3 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e62.0e62.1e62.2e6

Intensity, cps13.51*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 8 of 23Sample Name: "b120411a211" Sample ID: "GLP10-01-02-27-056" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-330R-HS-" Annotation: ""Sample Index: 211Sample Type: UnknownConcentration: N/ACalculated Conc: 2940. ng/mLAcq. Date: 4/14/2012Acq. Time: 2:39:22 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 11722827 countsHeight: 2.98e+006 cpsStart Time: 13.3 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e6

Intensity, cps13.51Sample Name: "b120411a215" Sample ID: "GLP10-01-02-27-057" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-330L-0-" Annotation: ""Sample Index: 215Sample Type: UnknownConcentration: N/ACalculated Conc: 388. ng/mLAcq. Date: 4/14/2012Acq. Time: 4:04:00 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 1865394 countsHeight: 5.02e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e54.6e54.8e55.0e5

Intensity, cps13.52Sample Name: "b120411a218" Sample ID: "GLP10-01-02-27-060" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-330L-HS-" Annotation: ""Sample Index: 218Sample Type: UnknownConcentration: N/ACalculated Conc: 851. ng/mLAcq. Date: 4/14/2012Acq. Time: 5:07:22 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 3956624 countsHeight: 1.09e+006 cpsStart Time: 13.3 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e6

Intensity, cps13.52Sample Name: "b120411a219" Sample ID: "GLP10-01-02-27-061" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-RW331S-0-" Annotation: ""Sample Index: 219Sample Type: UnknownConcentration: N/ACalculated Conc: 1010. ng/mLAcq. Date: 4/14/2012Acq. Time: 5:28:32 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 4646545 countsHeight: 1.27e+006 cpsStart Time: 13.3 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e61.25e6

Intensity, cps13.52*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 9 of 23Sample Name: "b120411a222" Sample ID: "GLP10-01-02-27-064" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-RW331S-HS-" Annotation: ""Sample Index: 222Sample Type: UnknownConcentration: N/ACalculated Conc: 1790. ng/mLAcq. Date: 4/14/2012Acq. Time: 6:31:47 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 7779398 countsHeight: 2.03e+006 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e62.0e6

Intensity, cps13.52Sample Name: "b120411a226" Sample ID: "GLP10-01-02-27-065" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-335R-0-" Annotation: ""Sample Index: 226Sample Type: UnknownConcentration: N/ACalculated Conc: 859. ng/mLAcq. Date: 4/14/2012Acq. Time: 7:56:20 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 3990967 countsHeight: 9.57e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e59.5e5

Intensity, cps13.5213.38Sample Name: "b120411a229" Sample ID: "GLP10-01-02-27-068" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-335R-HS-" Annotation: ""Sample Index: 229Sample Type: UnknownConcentration: N/ACalculated Conc: 5410. ng/mLAcq. Date: 4/14/2012Acq. Time: 8:59:33 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 4954060 countsHeight: 1.34e+006 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e59.5e51.0e61.1e61.1e61.2e61.2e61.3e61.3e6

Intensity, cps13.51Sample Name: "b120411a230" Sample ID: "GLP10-01-02-27-069" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-GRS04-0-" Annotation: ""Sample Index: 230Sample Type: UnknownConcentration: N/ACalculated Conc: 2320. ng/mLAcq. Date: 4/14/2012Acq. Time: 9:20:38 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 9697711 countsHeight: 2.45e+006 cpsStart Time: 13.3 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e62.0e62.1e62.2e62.3e62.4e6

Intensity, cps13.52*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 10 of 23Sample Name: "b120411a243" Sample ID: "GLP10-01-02-27-069" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-GRS04-0-" Annotation: ""Sample Index: 243Sample Type: UnknownConcentration: N/ACalculated Conc: 2700. ng/mLAcq. Date: 4/16/2012Acq. Time: 12:02:18 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.6 minArea: 2567928 countsHeight: 7.01e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps13.55Sample Name: "b120411a053" Sample ID: "GLP10-01-02-27-074" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TRIP01-0-" Annotation: ""Sample Index: 53Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/12/2012Acq. Time: 7:04:44 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.7 minArea: 1164 countsHeight: 2.55e+002 cpsStart Time: 13.6 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0501001502002503003504004505005506006507007508008509009501000105011001150120012501300

Intensity, cpsSample Name: "b120411a054" Sample ID: "GLP10-01-02-27-075" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TRIP01-LS-" Annotation: ""Sample Index: 54Sample Type: UnknownConcentration: N/ACalculated Conc: 9.82 ng/mLAcq. Date: 4/12/2012Acq. Time: 7:25:55 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 4523843 countsHeight: 1.22e+006 cpsStart Time: 13.4 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e6

Intensity, cps13.5513.39Sample Name: "b120411a055" Sample ID: "GLP10-01-02-27-076" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TRIP01-MS-" Annotation: ""Sample Index: 55Sample Type: UnknownConcentration: N/ACalculated Conc: 92.7 ng/mLAcq. Date: 4/12/2012Acq. Time: 7:46:56 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 4289698 countsHeight: 1.14e+006 cpsStart Time: 13.4 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e6

Intensity, cps13.54*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 11 of 23Sample Name: "b120411a056" Sample ID: "GLP10-01-02-27-077" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TRIP01-HS-" Annotation: ""Sample Index: 56Sample Type: UnknownConcentration: N/ACalculated Conc: 982. ng/mLAcq. Date: 4/12/2012Acq. Time: 8:08:05 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 4524684 countsHeight: 1.23e+006 cpsStart Time: 13.4 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e6

Intensity, cps13.54Sample Name: "b120411a044" Sample ID: "GLP10-01-02-27-073" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-203L-RB-" Annotation: ""Sample Index: 44Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/12/2012Acq. Time: 3:55:03 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 1395 countsHeight: 3.95e+002 cpsStart Time: 13.5 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min050100150200250300350400450500550600650700750800850900950100010501100115012001250

Intensity, cpsSample Name: "b120411a020" Sample ID: "11012-10-11" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "10 ng/mL FC std in Milli Q Water" Annotation: ""Sample Index: 20Sample Type: StandardConcentration: 9.98 ng/mLCalculated Conc: 10.1 ng/mLAcq. Date: 4/11/2012Acq. Time: 7:29:04 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 4916953 countsHeight: 1.39e+006 cpsStart Time: 15.0 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e59.5e51.0e61.1e61.1e61.2e61.2e61.3e61.3e61.4e6

Intensity, cps15.0917.57Sample Name: "b120411a028" Sample ID: "Solvent Blank" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "Milli Q Water TN12-0132" Annotation: ""Sample Index: 28Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/11/2012Acq. Time: 10:17:34 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.3 minArea: 1354 countsHeight: 2.98e+002 cpsStart Time: 15.2 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min05001000150020002500300035004000450050005500600065007000750080008500

Intensity, cps*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 12 of 23Sample Name: "b120411a029" Sample ID: "LCS-120405-1" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "0.5ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 0.497 ng/mLCalculated Conc: 0.476 ng/mLAcq. Date: 4/11/2012Acq. Time: 10:38:41 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 254814 countsHeight: 7.55e+004 cpsStart Time: 15.0 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e4

Intensity, cps15.1017.5516.5315.5817.1916.39Sample Name: "b120411a032" Sample ID: "LCS-120405-4" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "5ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 4.97 ng/mLCalculated Conc: 5.00 ng/mLAcq. Date: 4/11/2012Acq. Time: 11:41:54 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 2537144 countsHeight: 7.35e+005 cpsStart Time: 15.0 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e5

Intensity, cps15.0917.55Sample Name: "b120411a035" Sample ID: "LCS-120405-7" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "30ppb LCS" Annotation: ""Sample Index: 35Sample Type: QCConcentration: 29.9 ng/mLCalculated Conc: 28.4 ng/mLAcq. Date: 4/12/2012Acq. Time: 12:45:10 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 11814663 countsHeight: 3.19e+006 cpsStart Time: 14.9 minEnd Time: 15.4 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e6

Intensity, cps15.08Sample Name: "b120411a138" Sample ID: "GLP10-01-02-27-001" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-203L-0-" Annotation: ""Sample Index: 138Sample Type: UnknownConcentration: N/ACalculated Conc: 199. ng/mLAcq. Date: 4/13/2012Acq. Time: 12:58:18 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 1037815 countsHeight: 2.61e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e52.4e52.5e52.6e5

Intensity, cps15.03*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 13 of 23Sample Name: "b120411a141" Sample ID: "GLP10-01-02-27-004" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-203L-HS-" Annotation: ""Sample Index: 141Sample Type: UnknownConcentration: N/ACalculated Conc: 656. ng/mLAcq. Date: 4/13/2012Acq. Time: 2:01:29 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 3290555 countsHeight: 9.14e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e5

Intensity, cps15.0414.90Sample Name: "b120411a142" Sample ID: "GLP10-01-02-27-005" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-220R-0-" Annotation: ""Sample Index: 142Sample Type: UnknownConcentration: N/ACalculated Conc: 30.0 ng/mLAcq. Date: 4/13/2012Acq. Time: 2:22:32 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 1551193 countsHeight: 3.88e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e5

Intensity, cps15.0414.90Sample Name: "b120411a145" Sample ID: "GLP10-01-02-27-008" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-220R-HS-" Annotation: ""Sample Index: 145Sample Type: UnknownConcentration: N/ACalculated Conc: 119. ng/mLAcq. Date: 4/13/2012Acq. Time: 3:25:46 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 5728677 countsHeight: 1.56e+006 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e6

Intensity, cps15.0314.90Sample Name: "b120411a149" Sample ID: "GLP10-01-02-27-009" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-220L-0-" Annotation: ""Sample Index: 149Sample Type: UnknownConcentration: N/ACalculated Conc: 41.4 ng/mLAcq. Date: 4/13/2012Acq. Time: 4:50:17 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 2116410 countsHeight: 5.30e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e54.6e54.8e55.0e55.2e5

Intensity, cps15.0314.89*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 14 of 23Sample Name: "b120411a152" Sample ID: "GLP10-01-02-27-012" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-220L-HS-" Annotation: ""Sample Index: 152Sample Type: UnknownConcentration: N/ACalculated Conc: 134. ng/mLAcq. Date: 4/13/2012Acq. Time: 5:53:34 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 6340843 countsHeight: 1.77e+006 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e6

Intensity, cps15.0314.89Sample Name: "b120411a153" Sample ID: "GLP10-01-02-27-013" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-222R-0-" Annotation: ""Sample Index: 153Sample Type: UnknownConcentration: N/ACalculated Conc: 402. ng/mLAcq. Date: 4/13/2012Acq. Time: 6:14:42 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 2057826 countsHeight: 4.95e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e54.6e54.8e5

Intensity, cps15.0314.89Sample Name: "b120411a156" Sample ID: "GLP10-01-02-27-016" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-222R-HS-" Annotation: ""Sample Index: 156Sample Type: UnknownConcentration: N/ACalculated Conc: 862. ng/mLAcq. Date: 4/13/2012Acq. Time: 7:17:53 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 4251483 countsHeight: 1.16e+006 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e6

Intensity, cps15.0314.89Sample Name: "b120411a160" Sample ID: "GLP10-01-02-27-017" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-227R-0-" Annotation: ""Sample Index: 160Sample Type: UnknownConcentration: N/ACalculated Conc: 85.7 ng/mLAcq. Date: 4/13/2012Acq. Time: 8:42:27 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 453688 countsHeight: 1.20e+005 cpsStart Time: 14.8 minEnd Time: 15.1 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e51.15e5

Intensity, cps15.04*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 15 of 23Sample Name: "b120411a163" Sample ID: "GLP10-01-02-27-020" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-227R-HS-" Annotation: ""Sample Index: 163Sample Type: UnknownConcentration: N/ACalculated Conc: 546. ng/mLAcq. Date: 4/13/2012Acq. Time: 9:45:41 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 2763505 countsHeight: 8.16e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e5

Intensity, cps15.0414.9017.47Sample Name: "b120411a164" Sample ID: "GLP10-01-02-27-021" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-227L-0-" Annotation: ""Sample Index: 164Sample Type: UnknownConcentration: N/ACalculated Conc: 40.0 ng/mLAcq. Date: 4/13/2012Acq. Time: 10:06:53 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 215580 countsHeight: 5.29e+004 cpsStart Time: 14.8 minEnd Time: 15.1 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e44.2e44.4e44.6e44.8e45.0e45.2e4

Intensity, cps15.04Sample Name: "b120411a167" Sample ID: "GLP10-01-02-27-024" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-227L-HS-" Annotation: ""Sample Index: 167Sample Type: UnknownConcentration: N/ACalculated Conc: 487. ng/mLAcq. Date: 4/13/2012Acq. Time: 11:10:08 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 2476096 countsHeight: 7.35e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e5

Intensity, cps15.0314.9017.46Sample Name: "b120411a171" Sample ID: "GLP10-01-02-27-025" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-310R-0-" Annotation: ""Sample Index: 171Sample Type: UnknownConcentration: N/ACalculated Conc: 336. ng/mLAcq. Date: 4/14/2012Acq. Time: 12:34:33 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 1729101 countsHeight: 4.15e+005 cpsStart Time: 14.7 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e5

Intensity, cps15.0314.89*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 16 of 23Sample Name: "b120411a174" Sample ID: "GLP10-01-02-27-028" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-310R-HS-" Annotation: ""Sample Index: 174Sample Type: UnknownConcentration: N/ACalculated Conc: 1180. ng/mLAcq. Date: 4/14/2012Acq. Time: 1:37:56 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 5676830 countsHeight: 1.57e+006 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e6

Intensity, cps15.0314.89Sample Name: "b120411a175" Sample ID: "GLP10-01-02-27-029" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-RW312R-0-" Annotation: ""Sample Index: 175Sample Type: UnknownConcentration: N/ACalculated Conc: 495. ng/mLAcq. Date: 4/14/2012Acq. Time: 1:59:00 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 2512884 countsHeight: 5.97e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e5

Intensity, cps15.0314.90Sample Name: "b120411a178" Sample ID: "GLP10-01-02-27-032" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-RW312R-HS-" Annotation: ""Sample Index: 178Sample Type: UnknownConcentration: N/ACalculated Conc: 5100. ng/mLAcq. Date: 4/14/2012Acq. Time: 3:02:18 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 4968008 countsHeight: 1.46e+006 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e6

Intensity, cps15.02Sample Name: "b120411a182" Sample ID: "GLP10-01-02-27-033" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-317L-0-" Annotation: ""Sample Index: 182Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0865 ng/mLAcq. Date: 4/14/2012Acq. Time: 4:26:41 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 51001 countsHeight: 1.32e+004 cpsStart Time: 14.8 minEnd Time: 15.1 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.0500.01000.01500.02000.02500.03000.03500.04000.04500.05000.05500.06000.06500.07000.07500.08000.08500.09000.09500.01.0e41.1e41.1e41.2e41.2e41.3e41.3e4

Intensity, cps*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 17 of 23Sample Name: "b120411a185" Sample ID: "GLP10-01-02-27-036" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-317L-HS-" Annotation: ""Sample Index: 185Sample Type: UnknownConcentration: N/ACalculated Conc: 8.79 ng/mLAcq. Date: 4/14/2012Acq. Time: 5:30:02 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 4331555 countsHeight: 1.26e+006 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e6

Intensity, cps15.0317.47Sample Name: "b120411a186" Sample ID: "GLP10-01-02-27-037" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-324L-0-" Annotation: ""Sample Index: 186Sample Type: UnknownConcentration: N/ACalculated Conc: 87.9 ng/mLAcq. Date: 4/14/2012Acq. Time: 5:51:13 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 465228 countsHeight: 1.10e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e5

Intensity, cps15.03Sample Name: "b120411a189" Sample ID: "GLP10-01-02-27-040" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-324L-HS-" Annotation: ""Sample Index: 189Sample Type: UnknownConcentration: N/ACalculated Conc: 497. ng/mLAcq. Date: 4/14/2012Acq. Time: 6:54:28 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 2526063 countsHeight: 7.22e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e5

Intensity, cps15.0314.89Sample Name: "b120411a193" Sample ID: "GLP10-01-02-27-041" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-RW327R-0-" Annotation: ""Sample Index: 193Sample Type: UnknownConcentration: N/ACalculated Conc: 149. ng/mLAcq. Date: 4/14/2012Acq. Time: 8:19:04 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 782327 countsHeight: 1.81e+005 cpsStart Time: 14.8 minEnd Time: 15.1 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e5

Intensity, cps15.0414.90*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 18 of 23Sample Name: "b120411a196" Sample ID: "GLP10-01-02-27-044" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-RW327R-HS-" Annotation: ""Sample Index: 196Sample Type: UnknownConcentration: N/ACalculated Conc: 605. ng/mLAcq. Date: 4/14/2012Acq. Time: 9:22:24 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 3048259 countsHeight: 8.59e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e5

Intensity, cps15.0314.89Sample Name: "b120411a197" Sample ID: "GLP10-01-02-27-045" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-328R-0-" Annotation: ""Sample Index: 197Sample Type: UnknownConcentration: N/ACalculated Conc: 55.5 ng/mLAcq. Date: 4/14/2012Acq. Time: 9:43:31 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 296272 countsHeight: 7.13e+004 cpsStart Time: 14.8 minEnd Time: 15.1 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e4

Intensity, cps15.03Sample Name: "b120411a200" Sample ID: "GLP10-01-02-27-048" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-328R-HS-" Annotation: ""Sample Index: 200Sample Type: UnknownConcentration: N/ACalculated Conc: 497. ng/mLAcq. Date: 4/14/2012Acq. Time: 10:46:59 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 2526993 countsHeight: 7.23e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e5

Intensity, cps15.0314.8917.47Sample Name: "b120411a204" Sample ID: "GLP10-01-02-27-049" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-328L-0-" Annotation: ""Sample Index: 204Sample Type: UnknownConcentration: N/ACalculated Conc: 19.1 ng/mLAcq. Date: 4/14/2012Acq. Time: 12:11:25 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 995980 countsHeight: 1.84e+005 cpsStart Time: 14.7 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e5

Intensity, cps15.0214.88*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 19 of 23Sample Name: "b120411a207" Sample ID: "GLP10-01-02-27-052" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-328L-HS-" Annotation: ""Sample Index: 207Sample Type: UnknownConcentration: N/ACalculated Conc: 112. ng/mLAcq. Date: 4/14/2012Acq. Time: 1:14:51 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 5430838 countsHeight: 1.50e+006 cpsStart Time: 14.7 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e6

Intensity, cps15.0314.89Sample Name: "b120411a208" Sample ID: "GLP10-01-02-27-053" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-330R-0-" Annotation: ""Sample Index: 208Sample Type: UnknownConcentration: N/ACalculated Conc: 187. ng/mLAcq. Date: 4/14/2012Acq. Time: 1:35:58 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 976145 countsHeight: 2.52e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e52.4e52.5e5

Intensity, cps15.03Sample Name: "b120411a211" Sample ID: "GLP10-01-02-27-056" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-330R-HS-" Annotation: ""Sample Index: 211Sample Type: UnknownConcentration: N/ACalculated Conc: 1070. ng/mLAcq. Date: 4/14/2012Acq. Time: 2:39:22 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 5209638 countsHeight: 1.47e+006 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e6

Intensity, cps15.0314.89Sample Name: "b120411a215" Sample ID: "GLP10-01-02-27-057" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-330L-0-" Annotation: ""Sample Index: 215Sample Type: UnknownConcentration: N/ACalculated Conc: 354. ng/mLAcq. Date: 4/14/2012Acq. Time: 4:04:00 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 1817466 countsHeight: 4.53e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e5

Intensity, cps15.0314.89*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 20 of 23Sample Name: "b120411a218" Sample ID: "GLP10-01-02-27-060" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-330L-HS-" Annotation: ""Sample Index: 218Sample Type: UnknownConcentration: N/ACalculated Conc: 810. ng/mLAcq. Date: 4/14/2012Acq. Time: 5:07:22 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 4015200 countsHeight: 1.11e+006 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e6

Intensity, cps15.0314.90Sample Name: "b120411a219" Sample ID: "GLP10-01-02-27-061" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-RW331S-0-" Annotation: ""Sample Index: 219Sample Type: UnknownConcentration: N/ACalculated Conc: 477. ng/mLAcq. Date: 4/14/2012Acq. Time: 5:28:32 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 2426130 countsHeight: 5.89e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e5

Intensity, cps15.0314.89Sample Name: "b120411a222" Sample ID: "GLP10-01-02-27-064" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-RW331S-HS-" Annotation: ""Sample Index: 222Sample Type: UnknownConcentration: N/ACalculated Conc: 1260. ng/mLAcq. Date: 4/14/2012Acq. Time: 6:31:47 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 6022474 countsHeight: 1.63e+006 cpsStart Time: 14.7 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e6

Intensity, cps15.0314.89Sample Name: "b120411a226" Sample ID: "GLP10-01-02-27-065" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-335R-0-" Annotation: ""Sample Index: 226Sample Type: UnknownConcentration: N/ACalculated Conc: 1620. ng/mLAcq. Date: 4/14/2012Acq. Time: 7:56:20 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 7532029 countsHeight: 1.57e+006 cpsStart Time: 14.7 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e6

Intensity, cps15.0314.89*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 21 of 23Sample Name: "b120411a229" Sample ID: "GLP10-01-02-27-068" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-335R-HS-" Annotation: ""Sample Index: 229Sample Type: UnknownConcentration: N/ACalculated Conc: 6000. ng/mLAcq. Date: 4/14/2012Acq. Time: 8:59:33 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 5760780 countsHeight: 1.57e+006 cpsStart Time: 14.7 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e6

Intensity, cps15.02Sample Name: "b120411a230" Sample ID: "GLP10-01-02-27-069" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-GRS04-0-" Annotation: ""Sample Index: 230Sample Type: UnknownConcentration: N/ACalculated Conc: No InterceptAcq. Date: 4/14/2012Acq. Time: 9:20:38 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 26412325 countsHeight: 4.60e+006 cpsStart Time: 14.7 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e63.2e63.4e63.6e63.8e64.0e64.2e64.4e6

Intensity, cps15.0314.89Sample Name: "b120411a243" Sample ID: "GLP10-01-02-27-069" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-GRS04-0-" Annotation: ""Sample Index: 243Sample Type: UnknownConcentration: N/ACalculated Conc: 8350. ng/mLAcq. Date: 4/16/2012Acq. Time: 12:02:18 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.1 minArea: 7715860 countsHeight: 1.62e+006 cpsStart Time: 14.7 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e6

Intensity, cps15.0814.93Sample Name: "b120411a053" Sample ID: "GLP10-01-02-27-074" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TRIP01-0-" Annotation: ""Sample Index: 53Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/12/2012Acq. Time: 7:04:44 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.0 minArea: 648 countsHeight: 2.42e+002 cpsStart Time: 15.0 minEnd Time: 15.1 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0500100015002000250030003500400045005000550060006500700075008000

Intensity, cps*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 22 of 23Sample Name: "b120411a054" Sample ID: "GLP10-01-02-27-075" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TRIP01-LS-" Annotation: ""Sample Index: 54Sample Type: UnknownConcentration: N/ACalculated Conc: 9.46 ng/mLAcq. Date: 4/12/2012Acq. Time: 7:25:55 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 4636005 countsHeight: 1.40e+006 cpsStart Time: 14.9 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e59.5e51.0e61.1e61.1e61.2e61.2e61.3e61.3e61.4e6

Intensity, cps15.0617.5215.43Sample Name: "b120411a055" Sample ID: "GLP10-01-02-27-076" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TRIP01-MS-" Annotation: ""Sample Index: 55Sample Type: UnknownConcentration: N/ACalculated Conc: 91.6 ng/mLAcq. Date: 4/12/2012Acq. Time: 7:46:56 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 4498203 countsHeight: 1.31e+006 cpsStart Time: 14.9 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e59.5e51.0e61.1e61.1e61.2e61.2e61.3e61.3e6

Intensity, cps15.06Sample Name: "b120411a056" Sample ID: "GLP10-01-02-27-077" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TRIP01-HS-" Annotation: ""Sample Index: 56Sample Type: UnknownConcentration: N/ACalculated Conc: 952. ng/mLAcq. Date: 4/12/2012Acq. Time: 8:08:05 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 4664092 countsHeight: 1.36e+006 cpsStart Time: 14.9 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e59.5e51.0e61.1e61.1e61.2e61.2e61.3e61.3e61.4e6

Intensity, cps15.06Sample Name: "b120411a044" Sample ID: "GLP10-01-02-27-073" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-203L-RB-" Annotation: ""Sample Index: 44Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/12/2012Acq. Time: 3:55:03 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.2 minArea: 2034 countsHeight: 2.51e+002 cpsStart Time: 15.0 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0500100015002000250030003500400045005000550060006500700075008000

Intensity, cps*** Buster J2930203Printing Time: 2:28:04 PMPrinting Date: Friday, May 11, 2012Page 23 of 23Sample Name: "b120411a017" Sample ID: "11012-10-8" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "1.0 ng/mL FC std in Milli Q Water" Annotation: ""Sample Index: 17Sample Type: StandardConcentration: 0.928 ng/mLCalculated Conc: 0.899 ng/mLAcq. Date: 4/11/2012Acq. Time: 6:25:52 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 95Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.1 minArea: 530667 countsHeight: 1.19e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e51.15e5

Intensity, cps16.0615.9116.4717.8215.7414.33Sample Name: "b120411a028" Sample ID: "Solvent Blank" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "Milli Q Water TN12-0132" Annotation: ""Sample Index: 28Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/11/2012Acq. Time: 10:17:34 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 95Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 16.1 minArea: 1955 countsHeight: 7.33e+002 cpsStart Time: 16.0 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0500100015002000250030003500400045005000550060006500

Intensity, cps16.46Sample Name: "b120411a029" Sample ID: "LCS-120405-1" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "0.5ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 0.462 ng/mLCalculated Conc: 0.386 ng/mLAcq. Date: 4/11/2012Acq. Time: 10:38:41 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.1 minArea: 237973 countsHeight: 5.57e+004 cpsStart Time: 15.7 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e4

Intensity, cps16.0515.9116.4417.7815.7517.6417.2716.7814.3818.0818.28Sample Name: "b120411a032" Sample ID: "LCS-120405-4" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "5ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 4.62 ng/mLCalculated Conc: 4.55 ng/mLAcq. Date: 4/11/2012Acq. Time: 11:41:54 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 2584026 countsHeight: 5.90e+005 cpsStart Time: 15.7 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e5

Intensity, cps16.0515.9115.7316.47*** Buster J2930203Printing Time: 2:31:41 PMPrinting Date: Friday, May 11, 2012Page 1 of 6Sample Name: "b120411a035" Sample ID: "LCS-120405-7" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "30ppb LCS" Annotation: ""Sample Index: 35Sample Type: QCConcentration: 27.8 ng/mLCalculated Conc: 26.0 ng/mLAcq. Date: 4/12/2012Acq. Time: 12:45:10 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 13637286 countsHeight: 2.73e+006 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e62.0e62.1e62.2e62.3e62.4e62.5e62.6e62.7e6

Intensity, cps16.0415.9015.72Sample Name: "b120411a142" Sample ID: "GLP10-01-02-27-005" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-220R-0-" Annotation: ""Sample Index: 142Sample Type: UnknownConcentration: N/ACalculated Conc: 37.3 ng/mLAcq. Date: 4/13/2012Acq. Time: 2:22:32 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 2129400 countsHeight: 2.84e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e5

Intensity, cps15.9815.82Sample Name: "b120411a145" Sample ID: "GLP10-01-02-27-008" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-220R-HS-" Annotation: ""Sample Index: 145Sample Type: UnknownConcentration: N/ACalculated Conc: 122. ng/mLAcq. Date: 4/13/2012Acq. Time: 3:25:46 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 6735111 countsHeight: 1.17e+006 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e6

Intensity, cps15.9815.8315.67Sample Name: "b120411a149" Sample ID: "GLP10-01-02-27-009" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-220L-0-" Annotation: ""Sample Index: 149Sample Type: UnknownConcentration: N/ACalculated Conc: 52.5 ng/mLAcq. Date: 4/13/2012Acq. Time: 4:50:17 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 2974264 countsHeight: 3.99e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e5

Intensity, cps15.9815.82*** Buster J2930203Printing Time: 2:31:41 PMPrinting Date: Friday, May 11, 2012Page 2 of 6Sample Name: "b120411a152" Sample ID: "GLP10-01-02-27-012" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-220L-HS-" Annotation: ""Sample Index: 152Sample Type: UnknownConcentration: N/ACalculated Conc: 119. ng/mLAcq. Date: 4/13/2012Acq. Time: 5:53:34 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 6574899 countsHeight: 1.05e+006 cpsStart Time: 15.5 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e6

Intensity, cps15.9815.8315.67Sample Name: "b120411a153" Sample ID: "GLP10-01-02-27-013" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-222R-0-" Annotation: ""Sample Index: 153Sample Type: UnknownConcentration: N/ACalculated Conc: 395. ng/mLAcq. Date: 4/13/2012Acq. Time: 6:14:42 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 2251458 countsHeight: 4.04e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e5

Intensity, cps15.9815.83Sample Name: "b120411a156" Sample ID: "GLP10-01-02-27-016" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-222R-HS-" Annotation: ""Sample Index: 156Sample Type: UnknownConcentration: N/ACalculated Conc: 701. ng/mLAcq. Date: 4/13/2012Acq. Time: 7:17:53 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 3940590 countsHeight: 7.19e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e5

Intensity, cps15.9815.8315.67Sample Name: "b120411a193" Sample ID: "GLP10-01-02-27-041" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-RW327R-0-" Annotation: ""Sample Index: 193Sample Type: UnknownConcentration: N/ACalculated Conc: 337. ng/mLAcq. Date: 4/14/2012Acq. Time: 8:19:04 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 1926840 countsHeight: 2.21e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e5

Intensity, cps15.9815.83*** Buster J2930203Printing Time: 2:31:41 PMPrinting Date: Friday, May 11, 2012Page 3 of 6Sample Name: "b120411a196" Sample ID: "GLP10-01-02-27-044" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-RW327R-HS-" Annotation: ""Sample Index: 196Sample Type: UnknownConcentration: N/ACalculated Conc: 721. ng/mLAcq. Date: 4/14/2012Acq. Time: 9:22:24 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 4053890 countsHeight: 6.03e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e5

Intensity, cps15.9815.8215.67Sample Name: "b120411a208" Sample ID: "GLP10-01-02-27-053" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-330R-0-" Annotation: ""Sample Index: 208Sample Type: UnknownConcentration: N/ACalculated Conc: 458. ng/mLAcq. Date: 4/14/2012Acq. Time: 1:35:58 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 2601486 countsHeight: 4.31e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e5

Intensity, cps15.9815.82Sample Name: "b120411a211" Sample ID: "GLP10-01-02-27-056" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-330R-HS-" Annotation: ""Sample Index: 211Sample Type: UnknownConcentration: N/ACalculated Conc: 1120. ng/mLAcq. Date: 4/14/2012Acq. Time: 2:39:22 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 6195197 countsHeight: 1.12e+006 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e6

Intensity, cps15.9715.8215.67Sample Name: "b120411a215" Sample ID: "GLP10-01-02-27-057" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-330L-0-" Annotation: ""Sample Index: 215Sample Type: UnknownConcentration: N/ACalculated Conc: 135. ng/mLAcq. Date: 4/14/2012Acq. Time: 4:04:00 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 785700 countsHeight: 9.51e+004 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e4

Intensity, cps15.9815.82*** Buster J2930203Printing Time: 2:31:41 PMPrinting Date: Friday, May 11, 2012Page 4 of 6Sample Name: "b120411a218" Sample ID: "GLP10-01-02-27-060" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-330L-HS-" Annotation: ""Sample Index: 218Sample Type: UnknownConcentration: N/ACalculated Conc: 478. ng/mLAcq. Date: 4/14/2012Acq. Time: 5:07:22 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 2711366 countsHeight: 4.61e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e54.6e5

Intensity, cps15.9815.8315.67Sample Name: "b120411a226" Sample ID: "GLP10-01-02-27-065" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-335R-0-" Annotation: ""Sample Index: 226Sample Type: UnknownConcentration: N/ACalculated Conc: 3480. ng/mLAcq. Date: 4/14/2012Acq. Time: 7:56:20 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.8 minArea: 17642133 countsHeight: 1.74e+006 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e6

Intensity, cps15.8115.9715.67Sample Name: "b120411a229" Sample ID: "GLP10-01-02-27-068" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-335R-HS-" Annotation: ""Sample Index: 229Sample Type: UnknownConcentration: N/ACalculated Conc: 7750. ng/mLAcq. Date: 4/14/2012Acq. Time: 8:59:33 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 8444258 countsHeight: 1.14e+006 cpsStart Time: 15.5 minEnd Time: 16.3 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e6

Intensity, cps15.9615.80Sample Name: "b120411a044" Sample ID: "GLP10-01-02-27-073" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-203L-RB-" Annotation: ""Sample Index: 44Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/12/2012Acq. Time: 3:55:03 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 95Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 16.0 minArea: 1090 countsHeight: 4.35e+002 cpsStart Time: 16.0 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min050010001500200025003000350040004500500055006000

Intensity, cps*** Buster J2930203Printing Time: 2:31:41 PMPrinting Date: Friday, May 11, 2012Page 5 of 6Sample Name: "b120411a053" Sample ID: "GLP10-01-02-27-074" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TRIP01-0-" Annotation: ""Sample Index: 53Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/12/2012Acq. Time: 7:04:44 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 95Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 16.0 minArea: 2270 countsHeight: 5.03e+002 cpsStart Time: 16.0 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min020040060080010001200140016001800200022002400260028003000320034003600380040004200440046004800500052005400

Intensity, cps16.4317.7416.2816.0415.6515.5317.5817.3116.7817.9518.40Sample Name: "b120411a054" Sample ID: "GLP10-01-02-27-075" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TRIP01-LS-" Annotation: ""Sample Index: 54Sample Type: UnknownConcentration: N/ACalculated Conc: 8.49 ng/mLAcq. Date: 4/12/2012Acq. Time: 7:25:55 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 4746206 countsHeight: 9.12e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e5

Intensity, cps16.0115.8615.7016.41Sample Name: "b120411a055" Sample ID: "GLP10-01-02-27-076" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TRIP01-MS-" Annotation: ""Sample Index: 55Sample Type: UnknownConcentration: N/ACalculated Conc: 78.9 ng/mLAcq. Date: 4/12/2012Acq. Time: 7:46:56 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 4422989 countsHeight: 8.54e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e5

Intensity, cps16.0215.86Sample Name: "b120411a056" Sample ID: "GLP10-01-02-27-077" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TRIP01-HS-" Annotation: ""Sample Index: 56Sample Type: UnknownConcentration: N/ACalculated Conc: 826. ng/mLAcq. Date: 4/12/2012Acq. Time: 8:08:05 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 4622536 countsHeight: 8.93e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e5

Intensity, cps16.0115.86*** Buster J2930203Printing Time: 2:31:41 PMPrinting Date: Friday, May 11, 2012Page 6 of 6b120430a.rdb (PFOS): "Quadratic" Regression ("1 / x" weighting): y = -0.00331 x^2 + 1.47 x + 0.0199 (r = 0.9997)0 5 10 15 20 25 30 35 40 45 50 55 60 65 70 75 80 85 90 95Analyte Conc. / IS Conc.05101520253035404550556065707580859095100105110

Analyte Area / IS AreaWorkstation: ETSBUSTERBatch Name: b120430a.dab*Data worked up by KJUPage 1 of 1Printing Time: 1:09:05 PMPrinting Date: Wednesday, May 02, 2012Sample Name: "b120430a018" Sample ID: "11012-62-9" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "2.5 ng/mL FC std in Synth. Water" Annotation: ""Sample Index: 18Sample Type: StandardConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/30/2012Acq. Time: 9:39:17 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 478724 countsHeight: 1.14e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e5

Intensity, cps16.0317.1715.7917.4217.0817.9416.1817.5215.5515.2516.5018.1115.1514.7713.87Sample Name: "b120430a028" Sample ID: "12001-160" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "Method Blank" Annotation: ""Sample Index: 28Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 1:11:42 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 441020 countsHeight: 1.12e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e5

Intensity, cps16.02Sample Name: "b120430a029" Sample ID: "LCS-120425-1" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "0.2ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 0.953 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 1:32:54 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 386683 countsHeight: 9.77e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e4

Intensity, cps16.03Sample Name: "b120430a032" Sample ID: "LCS-120425-4" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "2ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 0.953 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 2:36:42 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 473092 countsHeight: 1.22e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e51.15e51.20e5

Intensity, cps16.04*** Buster J2930203Printing Time: 2:34:56 PMPrinting Date: Friday, May 11, 2012Page 1 of 15Sample Name: "b120430a035" Sample ID: "LCS-120425-7" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "20ppb LCS" Annotation: ""Sample Index: 35Sample Type: QCConcentration: 0.953 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 3:40:24 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 550212 countsHeight: 1.38e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e41.0e51.1e51.1e51.2e51.2e51.3e51.3e51.4e5

Intensity, cps16.02Sample Name: "b120430a046" Sample ID: "GLP10-01-02-27-001" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-203L-0-" Annotation: ""Sample Index: 46Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 7:33:57 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 629623 countsHeight: 1.67e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e5

Intensity, cps16.01Sample Name: "b120430a048" Sample ID: "GLP10-01-02-27-004" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-203L-HS-" Annotation: ""Sample Index: 48Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 8:16:26 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 652377 countsHeight: 1.69e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e5

Intensity, cps16.01Sample Name: "b120430a049" Sample ID: "GLP10-01-02-27-017" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-227R-0-" Annotation: ""Sample Index: 49Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 8:37:45 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 634945 countsHeight: 1.61e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e5

Intensity, cps16.01*** Buster J2930203Printing Time: 2:34:56 PMPrinting Date: Friday, May 11, 2012Page 2 of 15Sample Name: "b120430a051" Sample ID: "GLP10-01-02-27-020" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-227R-HS-" Annotation: ""Sample Index: 51Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 9:20:08 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 619179 countsHeight: 1.57e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e5

Intensity, cps16.00Sample Name: "b120430a055" Sample ID: "GLP10-01-02-27-021" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-227L-0-" Annotation: ""Sample Index: 55Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 10:45:11 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 627094 countsHeight: 1.66e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e5

Intensity, cps16.00Sample Name: "b120430a058" Sample ID: "GLP10-01-02-27-024" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-227L-HS-" Annotation: ""Sample Index: 58Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 11:48:58 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 614996 countsHeight: 1.64e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e5

Intensity, cps15.99Sample Name: "b120430a059" Sample ID: "GLP10-01-02-27-025" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-310R-0-" Annotation: ""Sample Index: 59Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 12:10:16 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 590671 countsHeight: 1.48e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e5

Intensity, cps15.99*** Buster J2930203Printing Time: 2:34:56 PMPrinting Date: Friday, May 11, 2012Page 3 of 15Sample Name: "b120430a062" Sample ID: "GLP10-01-02-27-028" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-310R-HS-" Annotation: ""Sample Index: 62Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 1:13:54 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 627917 countsHeight: 1.58e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e5

Intensity, cps16.00Sample Name: "b120430a066" Sample ID: "GLP10-01-02-27-033" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-317L-0-" Annotation: ""Sample Index: 66Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 2:38:58 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 420518 countsHeight: 1.02e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e5

Intensity, cps16.00Sample Name: "b120430a068" Sample ID: "GLP10-01-02-27-035" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-317L-LS-" Annotation: ""Sample Index: 68Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 3:21:31 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 358297 countsHeight: 8.62e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e4

Intensity, cps15.99Sample Name: "b120430a070" Sample ID: "GLP10-01-02-27-037" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-324L-0-" Annotation: ""Sample Index: 70Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 4:04:05 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 632110 countsHeight: 1.57e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e5

Intensity, cps15.99*** Buster J2930203Printing Time: 2:34:57 PMPrinting Date: Friday, May 11, 2012Page 4 of 15Sample Name: "b120430a073" Sample ID: "GLP10-01-02-27-040" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-324L-HS-" Annotation: ""Sample Index: 73Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 5:07:52 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 564774 countsHeight: 1.41e+005 cpsStart Time: 15.8 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e5

Intensity, cps16.00Sample Name: "b120430a077" Sample ID: "GLP10-01-02-27-045" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-328R-0-" Annotation: ""Sample Index: 77Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 6:33:00 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 576218 countsHeight: 1.46e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e5

Intensity, cps16.00Sample Name: "b120430a080" Sample ID: "GLP10-01-02-27-048" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-328R-HS-" Annotation: ""Sample Index: 80Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 7:36:52 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 608812 countsHeight: 1.56e+005 cpsStart Time: 15.8 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e5

Intensity, cps16.00Sample Name: "b120430a081" Sample ID: "GLP10-01-02-27-049" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-328L-0-" Annotation: ""Sample Index: 81Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 7:58:10 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 351762 countsHeight: 9.56e+004 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e4

Intensity, cps15.99*** Buster J2930203Printing Time: 2:34:57 PMPrinting Date: Friday, May 11, 2012Page 5 of 15Sample Name: "b120430a083" Sample ID: "GLP10-01-02-27-051" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-328L-LS-" Annotation: ""Sample Index: 83Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 8:40:46 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 347312 countsHeight: 9.03e+004 cpsStart Time: 15.8 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e4

Intensity, cps15.99Sample Name: "b120430a088" Sample ID: "GLP10-01-02-27-061" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-RW331S-0-" Annotation: ""Sample Index: 88Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 10:27:11 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 632443 countsHeight: 1.60e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e5

Intensity, cps15.99Sample Name: "b120430a091" Sample ID: "GLP10-01-02-27-064" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-RW331S-HS-" Annotation: ""Sample Index: 91Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 11:30:57 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 576912 countsHeight: 1.49e+005 cpsStart Time: 15.9 minEnd Time: 16.3 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e5

Intensity, cps15.99Sample Name: "b120430a100" Sample ID: "GLP10-01-02-27-029" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-RW312R-0-" Annotation: ""Sample Index: 100Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/2/2012Acq. Time: 2:42:41 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 542377 countsHeight: 1.40e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e41.0e51.1e51.1e51.2e51.2e51.3e51.3e51.4e5

Intensity, cps15.99*** Buster J2930203Printing Time: 2:34:57 PMPrinting Date: Friday, May 11, 2012Page 6 of 15Sample Name: "b120430a103" Sample ID: "GLP10-01-02-27-032" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-RW312R-HS-" Annotation: ""Sample Index: 103Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/2/2012Acq. Time: 3:46:47 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 526105 countsHeight: 1.24e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e51.15e51.20e5

Intensity, cps15.99Sample Name: "b120430a042" Sample ID: "GLP10-01-02-27-074" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-TRIP01-0-" Annotation: ""Sample Index: 42Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 6:09:03 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 678955 countsHeight: 1.70e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e5

Intensity, cps16.01Sample Name: "b120430a043" Sample ID: "GLP10-01-02-27-075" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-TRIP01-LS-" Annotation: ""Sample Index: 43Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 6:30:19 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 674914 countsHeight: 1.78e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e5

Intensity, cps16.02Sample Name: "b120430a044" Sample ID: "GLP10-01-02-27-076" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-TRIP01-MS-" Annotation: ""Sample Index: 44Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 6:51:30 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 444349 countsHeight: 1.21e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e51.15e51.20e5

Intensity, cps16.02*** Buster J2930203Printing Time: 2:34:57 PMPrinting Date: Friday, May 11, 2012Page 7 of 15Sample Name: "b120430a045" Sample ID: "GLP10-01-02-27-077" File: "b120430a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-TRIP01-HS-" Annotation: ""Sample Index: 45Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 5/1/2012Acq. Time: 7:12:46 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 607100 countsHeight: 1.60e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e5

Intensity, cps16.02Sample Name: "b120430a018" Sample ID: "11012-62-9" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "2.5 ng/mL FC std in Synth. Water" Annotation: ""Sample Index: 18Sample Type: StandardConcentration: 2.32 ng/mLCalculated Conc: 2.43 ng/mLAcq. Date: 4/30/2012Acq. Time: 9:39:17 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 1781697 countsHeight: 3.07e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e5

Intensity, cps16.0315.8816.41Sample Name: "b120430a028" Sample ID: "12001-160" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "Method Blank" Annotation: ""Sample Index: 28Sample Type: UnknownConcentration: N/ACalculated Conc: 0.00238 ng/mLAcq. Date: 5/1/2012Acq. Time: 1:11:42 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 10403 countsHeight: 1.66e+003 cpsStart Time: 15.8 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min05001000150020002500300035004000450050005500600065007000

Intensity, cpsSample Name: "b120430a029" Sample ID: "LCS-120425-1" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "0.2ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 0.184 ng/mLCalculated Conc: 0.194 ng/mLAcq. Date: 5/1/2012Acq. Time: 1:32:54 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 123356 countsHeight: 2.16e+004 cpsStart Time: 15.7 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e42.0e42.1e4

Intensity, cps16.0216.4015.8816.5017.7217.7917.5716.5717.1515.7218.38*** Buster J2930203Printing Time: 2:34:57 PMPrinting Date: Friday, May 11, 2012Page 8 of 15Sample Name: "b120430a032" Sample ID: "LCS-120425-4" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "2ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 1.84 ng/mLCalculated Conc: 1.85 ng/mLAcq. Date: 5/1/2012Acq. Time: 2:36:42 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 1353938 countsHeight: 2.50e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e52.4e5

Intensity, cps16.0415.8916.4015.71Sample Name: "b120430a035" Sample ID: "LCS-120425-7" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "20ppb LCS" Annotation: ""Sample Index: 35Sample Type: QCConcentration: 18.4 ng/mLCalculated Conc: 18.2 ng/mLAcq. Date: 5/1/2012Acq. Time: 3:40:24 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 14821227 countsHeight: 2.50e+006 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e62.0e62.1e62.2e62.3e62.4e6

Intensity, cps16.0215.8715.70Sample Name: "b120430a046" Sample ID: "GLP10-01-02-27-001" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-203L-0-" Annotation: ""Sample Index: 46Sample Type: UnknownConcentration: N/ACalculated Conc: 702. ng/mLAcq. Date: 5/1/2012Acq. Time: 7:33:57 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 6708243 countsHeight: 9.99e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e59.5e5

Intensity, cps16.0115.85Sample Name: "b120430a048" Sample ID: "GLP10-01-02-27-004" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-203L-HS-" Annotation: ""Sample Index: 48Sample Type: UnknownConcentration: N/ACalculated Conc: 1050. ng/mLAcq. Date: 5/1/2012Acq. Time: 8:16:26 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 10294934 countsHeight: 1.53e+006 cpsStart Time: 15.5 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e6

Intensity, cps16.0115.8615.70*** Buster J2930203Printing Time: 2:34:57 PMPrinting Date: Friday, May 11, 2012Page 9 of 15Sample Name: "b120430a049" Sample ID: "GLP10-01-02-27-017" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-227R-0-" Annotation: ""Sample Index: 49Sample Type: UnknownConcentration: N/ACalculated Conc: 642. ng/mLAcq. Date: 5/1/2012Acq. Time: 8:37:45 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 6202720 countsHeight: 8.17e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e5

Intensity, cps16.0015.85Sample Name: "b120430a051" Sample ID: "GLP10-01-02-27-020" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-227R-HS-" Annotation: ""Sample Index: 51Sample Type: UnknownConcentration: N/ACalculated Conc: 991. ng/mLAcq. Date: 5/1/2012Acq. Time: 9:20:08 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 9245086 countsHeight: 1.32e+006 cpsStart Time: 15.5 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e59.5e51.0e61.1e61.1e61.2e61.2e61.3e61.3e6

Intensity, cps16.0015.8515.69Sample Name: "b120430a055" Sample ID: "GLP10-01-02-27-021" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-227L-0-" Annotation: ""Sample Index: 55Sample Type: UnknownConcentration: N/ACalculated Conc: 222. ng/mLAcq. Date: 5/1/2012Acq. Time: 10:45:11 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 2151023 countsHeight: 3.15e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e5

Intensity, cps16.0015.84Sample Name: "b120430a058" Sample ID: "GLP10-01-02-27-024" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-227L-HS-" Annotation: ""Sample Index: 58Sample Type: UnknownConcentration: N/ACalculated Conc: 595. ng/mLAcq. Date: 5/1/2012Acq. Time: 11:48:58 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 5576961 countsHeight: 8.74e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e5

Intensity, cps15.9915.8315.67*** Buster J2930203Printing Time: 2:34:57 PMPrinting Date: Friday, May 11, 2012Page 10 of 15Sample Name: "b120430a059" Sample ID: "GLP10-01-02-27-025" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-310R-0-" Annotation: ""Sample Index: 59Sample Type: UnknownConcentration: N/ACalculated Conc: 697. ng/mLAcq. Date: 5/1/2012Acq. Time: 12:10:16 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 6250621 countsHeight: 6.75e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps16.0015.8315.70Sample Name: "b120430a062" Sample ID: "GLP10-01-02-27-028" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-310R-HS-" Annotation: ""Sample Index: 62Sample Type: UnknownConcentration: N/ACalculated Conc: 1390. ng/mLAcq. Date: 5/1/2012Acq. Time: 1:13:54 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 13040022 countsHeight: 1.76e+006 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e6

Intensity, cps16.0015.8515.69Sample Name: "b120430a066" Sample ID: "GLP10-01-02-27-033" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-317L-0-" Annotation: ""Sample Index: 66Sample Type: UnknownConcentration: N/ACalculated Conc: 0.235 ng/mLAcq. Date: 5/1/2012Acq. Time: 2:38:58 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 160895 countsHeight: 2.17e+004 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e42.0e42.1e4

Intensity, cpsSample Name: "b120430a068" Sample ID: "GLP10-01-02-27-035" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-317L-LS-" Annotation: ""Sample Index: 68Sample Type: UnknownConcentration: N/ACalculated Conc: 1.06 ng/mLAcq. Date: 5/1/2012Acq. Time: 3:21:31 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 591220 countsHeight: 8.23e+004 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e4

Intensity, cps15.9915.8316.3615.6816.4617.67*** Buster J2930203Printing Time: 2:34:57 PMPrinting Date: Friday, May 11, 2012Page 11 of 15Sample Name: "b120430a070" Sample ID: "GLP10-01-02-27-037" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-324L-0-" Annotation: ""Sample Index: 70Sample Type: UnknownConcentration: N/ACalculated Conc: 193. ng/mLAcq. Date: 5/1/2012Acq. Time: 4:04:05 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 1886898 countsHeight: 2.33e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e5

Intensity, cps15.9915.8415.6816.36Sample Name: "b120430a073" Sample ID: "GLP10-01-02-27-040" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-324L-HS-" Annotation: ""Sample Index: 73Sample Type: UnknownConcentration: N/ACalculated Conc: 613. ng/mLAcq. Date: 5/1/2012Acq. Time: 5:07:52 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 5270664 countsHeight: 7.44e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e5

Intensity, cps15.9915.8415.70Sample Name: "b120430a077" Sample ID: "GLP10-01-02-27-045" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-328R-0-" Annotation: ""Sample Index: 77Sample Type: UnknownConcentration: N/ACalculated Conc: 139. ng/mLAcq. Date: 5/1/2012Acq. Time: 6:33:00 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 1245100 countsHeight: 1.39e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e41.0e51.1e51.1e51.2e51.2e51.3e51.3e51.4e5

Intensity, cps15.9915.83Sample Name: "b120430a080" Sample ID: "GLP10-01-02-27-048" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-328R-HS-" Annotation: ""Sample Index: 80Sample Type: UnknownConcentration: N/ACalculated Conc: 529. ng/mLAcq. Date: 5/1/2012Acq. Time: 7:36:52 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 4909784 countsHeight: 7.32e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e5

Intensity, cps16.0015.8315.69*** Buster J2930203Printing Time: 2:34:57 PMPrinting Date: Friday, May 11, 2012Page 12 of 15Sample Name: "b120430a081" Sample ID: "GLP10-01-02-27-049" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-328L-0-" Annotation: ""Sample Index: 81Sample Type: UnknownConcentration: N/ACalculated Conc: 1.12 ng/mLAcq. Date: 5/1/2012Acq. Time: 7:58:10 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.8 minArea: 614713 countsHeight: 5.74e+004 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e4

Intensity, cps15.8316.00Sample Name: "b120430a083" Sample ID: "GLP10-01-02-27-051" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-328L-LS-" Annotation: ""Sample Index: 83Sample Type: UnknownConcentration: N/ACalculated Conc: 9.16 ng/mLAcq. Date: 5/1/2012Acq. Time: 8:40:46 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 4801748 countsHeight: 6.83e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps15.9915.8315.6816.36Sample Name: "b120430a088" Sample ID: "GLP10-01-02-27-061" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-RW331S-0-" Annotation: ""Sample Index: 88Sample Type: UnknownConcentration: N/ACalculated Conc: 832. ng/mLAcq. Date: 5/1/2012Acq. Time: 10:27:11 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 7964776 countsHeight: 8.10e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e5

Intensity, cps15.9915.8415.70Sample Name: "b120430a091" Sample ID: "GLP10-01-02-27-064" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-RW331S-HS-" Annotation: ""Sample Index: 91Sample Type: UnknownConcentration: N/ACalculated Conc: 1540. ng/mLAcq. Date: 5/1/2012Acq. Time: 11:30:57 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 13226491 countsHeight: 1.65e+006 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e6

Intensity, cps15.9915.8315.68*** Buster J2930203Printing Time: 2:34:57 PMPrinting Date: Friday, May 11, 2012Page 13 of 15Sample Name: "b120430a100" Sample ID: "GLP10-01-02-27-029" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-RW312R-0-" Annotation: ""Sample Index: 100Sample Type: UnknownConcentration: N/ACalculated Conc: 1030. ng/mLAcq. Date: 5/2/2012Acq. Time: 2:42:41 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 8446999 countsHeight: 9.10e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e5

Intensity, cps15.9915.8315.69Sample Name: "b120430a103" Sample ID: "GLP10-01-02-27-032" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-RW312R-HS-" Annotation: ""Sample Index: 103Sample Type: UnknownConcentration: N/ACalculated Conc: 4930. ng/mLAcq. Date: 5/2/2012Acq. Time: 3:46:47 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 35326023 countsHeight: 4.26e+006 cpsStart Time: 15.5 minEnd Time: 16.3 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e63.2e63.4e63.6e63.8e64.0e64.2e6

Intensity, cps15.9915.8415.68Sample Name: "b120430a042" Sample ID: "GLP10-01-02-27-074" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TRIP01-0-" Annotation: ""Sample Index: 42Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 5/1/2012Acq. Time: 6:09:03 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 9686 countsHeight: 1.59e+003 cpsStart Time: 15.8 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min05001000150020002500300035004000450050005500600065007000

Intensity, cpsSample Name: "b120430a043" Sample ID: "GLP10-01-02-27-075" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TRIP01-LS-" Annotation: ""Sample Index: 43Sample Type: UnknownConcentration: N/ACalculated Conc: 8.38 ng/mLAcq. Date: 5/1/2012Acq. Time: 6:30:19 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 8559297 countsHeight: 1.35e+006 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e59.5e51.0e61.1e61.1e61.2e61.2e61.3e61.3e61.4e6

Intensity, cps16.0215.8715.7116.40*** Buster J2930203Printing Time: 2:34:57 PMPrinting Date: Friday, May 11, 2012Page 14 of 15Sample Name: "b120430a044" Sample ID: "GLP10-01-02-27-076" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TRIP01-MS-" Annotation: ""Sample Index: 44Sample Type: UnknownConcentration: N/ACalculated Conc: 71.8 ng/mLAcq. Date: 5/1/2012Acq. Time: 6:51:30 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 4844068 countsHeight: 7.43e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e5

Intensity, cps16.0215.86Sample Name: "b120430a045" Sample ID: "GLP10-01-02-27-077" File: "b120430a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TRIP01-HS-" Annotation: ""Sample Index: 45Sample Type: UnknownConcentration: N/ACalculated Conc: 804. ng/mLAcq. Date: 5/1/2012Acq. Time: 7:12:46 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 7387728 countsHeight: 1.15e+006 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e6

Intensity, cps16.0215.8715.70*** Buster J2930203Printing Time: 2:34:57 PMPrinting Date: Friday, May 11, 2012Page 15 of 15

# A TTACHMENT 

# C: A NALYTICAL 

# M ETHOD 

# ( S 

# )

### ETS -8- 044. 1 Page 2 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis 

## 1 Scope and Application

This method describes the direct injection analysis of perfluorinated compounds (PFCs) from

water matrices using high

- performance liquid chromatography tandem mass spectrometry

(HPLC/MS/MS). The method is generally applicable but not limited to the measurement of

perfluoroalkyl sulfonamides and perfluorinated alkyl acids (PFAAs) such as perfluorosulfonic

acids (PFSAs) and perfluorocarboxylic acids (PFCAs) (Table 1). Water samples containing

heavy particulate may require preparation by an alternate method such as ETS -8- 154

“ Determination of Perfluorinated Acids, Alcohols, Amides, and Sulfonates In Water By Solid

Phase Extraction and High Performance Liquid Chromatography/Mass Spectrometry ”. The

method is applicable to both external standard and internal standard calibration 1



### .

Table 1. Representative Tar get Analytes

**Acronym Analyte**  **Chemical Abstract Services**

**Registry Number (CASRN)**

PFBA (C4 Acid) Perfluorobutanoic acid 375

- 22

- 4

PFPeA (C5 Acid) Perfluoropentanoic acid 2706

- 90

- 3

PFHxA (C6 Acid) Perfluorohexanoic acid 307

- 24

- 4

PFHpA (C7 Acid) Perfluoroheptanoic acid 375

- 85

- 9

PFOA (C8 Acid) Perfluorooctanoic acid 335

- 67

- 1

PFNA (C9 Acid) Perfluorononanoic acid 375

- 95

- 1

PFDA (C10 Acid) Perfluorodecanoic acid 335

- 76

- 2

PFUnA (C11 Acid) Perfluoroundecanoic acid 2058

- 94

- 8

PFDoA (C12 Acid) Perfluorododecanoic acid 307

- 55

- 1

PF TrD A (C1 3 Acid) Perfluorotridecanoic acid 72629

- 94

- 8

PFBS (C4 Sulfonate) Perfluorobutanesulfonic acid 375

- 73

- 5

PFHS (C6 Sulfonate) Perfluorohexanesulfonic acid 355

- 46

- 4

PFOS (C8 Sulfonate) Perfluorooctanesulfonic acid 1763

- 23

- 1

FBSA (C4 Sulfonamide Perfluorobutanesulfonamide 30334

- 69

- 1

FOSA (C8 Sulfonamide) Pefluorooctanesulfonamide 754

- 91

- 6



### The Minimum Reporting Level (MRL) is the Limit of Quantitation (LOQ) that meets Data

Quality Objectives (DQOs) that are developed based on the intended use of this method.

Method Flexibility

– This is a performance

- based method and may be generally applied to the

determination of perfluorinated compounds in water matrices when analysis batch quality

c ontrol ( QC ) criteria are met 2



### . Each set of samples are prepared in an analysis batch with

calibration standards, LCS s, blanks, and continuing calibration check standards analyzed on

the same instrument during a time period that begins and ends with the analysis of the

appropriate continuing calibration check standards. The laboratory is permitted to modify the

LC column, mobile phase composition, LC conditions, and MS/MS conditions. Method

modifications should be considered to improve method performance or to meet data quality

objectives for the study. In all cases where method modifications are implemented, the batch

1 

### The method is supported by validation with internal standard calibration for C4

- C13 PFCAs, C4, C6, and C8 PFSAs,

and C8 perfluoroalkane sulfonamide in laboratory control samples under 3M method validation E11

- 0667.

2 

### Guidance for establishing method QC Criteria based on a.) FDA May 2001, “Guidance for Industry, Bioanalytical

Method Vali dation”, b.) EPA Method 537, and c.) European Commission: Guidance for Generating and Reporting

Methods of Analysis in Support of Pre

- registration Data Requirements for Annex II (Part A, section 4) and Annex III

(Part A,section 5) of Directive 91/414, SANCO/3029/99 rev. 4 (11/07/00).

### ETS -8- 044. 1 Page 3 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis analytical QCs ( section 9) must be completed and pass QC acceptance criteria ( section 13 ) if

the data from the analytical batch are to be reported

.



## 2 Method Summary

Water samples are analyzed as neat aqueous sample or as solvent diluted aqueous samples

by direct injection using LC/ MS/MS. Samples containing heavy particulate may not be

suitable for analysis by this method. Samples containing suspended particulate should be

centrifuge d or filtered prior to removing a sample aliquot or diluting with solvent

. The water

sample is mixe d well prior to removing an aliquot or diluting, if necessary, with ASTM Type I

water, HPLC water, other suitable water, or solvent (methanol).

Quantitation is by stable isotope internal standard calibration in laboratory reagent water. All

perfluorinat ed compounds ( PFC s) target analyte concentrations of perfluorosulfonic acids

(PFSAs) and perfluorocarboxylic acids (PFCAs) are reported as anions and corrected for

their salt or free acid forms. Alternatively, quantitation may be performed by external

st andard calibration.

This is a performance

- based method. Method uncertainty for each target analyte is

determined for each analytical batch using multiple laboratory control spikes at multiple

concentrations. This method also requires that the precision and accuracy for each sample

be determined using field matrix spikes to verify that the method is applicable to each sample

matrix.

Calibration standards for PFUnA, PFDoA, PFTrDA, and FOSA have been found to be

un stable for more than 2 days in 100% water. Samples requiring analysis for these

compounds by this method should be diluted 1:1 with methanol and analyzed against a

calibration curve prepared in 1:1 synthetic groundwater:MeOH.



## 3 Definitions

3.1 Analysis Batch

A set of study samples that are prepared with calibration standards, laboratory control

samples, and procedural blanks , and analyzed on the same instrument during a time period

that begins and ends with the analysis of the appropriate continuing calibration check

standards.



## 3.2 Analytical Sample

A portion of a laboratory sample prepared for analysis.



## 3.3 Calibration Standard

A solution prepared by spiking a known volume of the Working Standard (WS) into a

predete rmined amount of ASTM Type I, HPLC grade water, or other suitable water (i.e.

matrix water), and analyzed according to this method. Calibration standards are used to

calibrate the instrument response with respect to analyte concentration.



## 3.4 Laboratory Duplicate Sample (LDS, or Lab Dup)

A laboratory duplicate sample is a separate aliquot of a sample taken in the analytical

laboratory that is analyzed separately with identical procedures. Analysis of LDSs compared

to that of the first aliquot give a measure of the precision associated with laboratory

procedures, but not with sample collection, preservation, or storage procedures.

### ETS -8- 044. 1 Page 4 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis 

## 3.5 Field Blank (FB)/Trip Blank (TB)

ASTM Type I, HPLC grade water, or other suitable water, placed in a sample container in the

laboratory and treated as a sample in all respects, including exposure to samp ling site

conditions, storage, preservation and all analytical procedures. The purpose of the TB is to

determine if test substances or other interferences are present in the field environment. This

sample is also referred to as a Trip Blank.



## 3.6 Field Duplicate Sample (FDS, Field Dup)

A sample collected in duplicate at the same time from the same location as the sample. The

FDS is handled under identical circumstances and treated exactly the same throughout field

and laboratory procedures. Analysis of the FDS compared to that of the first sample gives a

measure of the precision associated with sample collection, preservation and storage, as well

as with laboratory procedures.



## 3.7 Field Matrix Spike (FMS)

A sample to which known quantities of the target analyte s , ISs and SRSs are added to the

sample bottle in the laboratory before the bottles are sent to the field for collection of aqueous

samples. A known, specific volume of sample must be added to the sample container

without rinsing. This may be accomplished by making a “fill to this level” line on the outside of

the sample container. The FMS is analyzed to ascertain if any matrix effects, interferences,

or stability issues may complicate the interpretation of the sample analysis.



## 3.8 Trip Blank Matrix Spike ( T BMS )

An aliquot of ASTM Type I, HPLC grade water, or other suitable water, to which known

quantities of the target analytes , ISs and SRSs are added in the laboratory prior to the

shipment of the collection bottles. The TBMS is analyzed exactly like a study sample to help

determine if the method is in control and whether a loss of analyte or analytical bias could be

attributed to sample holding time, sample storage and/or shipment issues. A low and high

TBMS are appropriate when expected sample concentrations are not known or may vary.



## 3.9 Internal Standard (IS)

A compound added to each study sample, calibration standard, laboratory control samples,

and procedural blanks at a consistent level (typically around 1 ng/mL). T he internal

standard (s) are stable isotope labeled versions of the target analyte s . The area count ratio of

the target analyte to the internal standard is used for calibration. Surrogate ISs are applied

when stable isotope ISs of target analytes are unavailable. A surrogate IS is not necessar ily

a stable isotope labeled version of the target analyte, but is treated as an internal standard for

quantitation.



## 3.10 Laboratory Control Sample (LCS)

An aliquot of control matrix to which known quantities of the target analytes , ISs and SRSs

(when applicabl e) are added in the laboratory at the time when samples are aliquotted . At

least three levels (two levels for SRSs) in triplicate are included, one generally at the low end

of the calibration curve and one near the mid range and the upper end of the curve. The

LCSs are analyzed exactly like a laboratory sample to determine whether the stability of the

standards . LCSs should be prepared each day samples are aliquoted

.



## 3.11 Laboratory Matrix Spike (LMS)

A laboratory matrix spike is an aliquot of a sample to which known quantities of target

analytes , ISs and SRSs (when applicable) are added in the laboratory. The LMS is analyzed

exactly like a laboratory sample to determine whether the sample matrix contributes bias to

the analytical results. The endogenous concentrations of the analytes in the sample matrix

must be determined in a separate aliquot and the measured values in the LMS corrected for

these concentrations. LMSs are optional for analysis of aqueous samples.

### ETS -8- 044. 1 Page 5 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis 

## 3.12 Laboratory Sample

A portion or aliquot of a sample received from the field for testing.



## 3.13 Limit of Quantitation (LOQ)

The lower limit of quantitation (LLOQ) for an analytical batch is the lowest concentration that

can be reliably quantitated within the specified limits of precision and accuracy. The LLOQ is

generally selected as the lowest non

- zero standard in the calibration curve that meets method

acceptance criteria. The LLOQ for each target analyte is established for each analysis batch

as the lowest calibration standard with area counts at least twice that of the average area

counts of the procedural blanks.

The upper limit of quantitation (ULOQ) for an analytical batch is the highest concentration that

can be reliably quantitated within the specified limits of precision and accuracy. The highest

standard in the calibration curve that meets method acceptance criteria is defined as the

ULOQ.



## 3.14 Method /Procedural Blank

An aliquot of control matrix that is treated exactly like a laboratory sample including exposure

to all glassware, equipment, solvents, and reagents that are used with other laboratory

samples. The method blank is used to determine if test substances or other interferences are

present in the laboratory environment, the reagents, or the apparatus.



## 3.15 Sample

A sample is an aliquot removed from a larger quantity of material intended to represent the

original source material.



## 3.16 Stock Standard Solution (SSS)

A concentrated solution of a single

- analyte prepared in the laboratory with an assayed

reference compound.



## 3.17 Surrogate Internal Standard

An IS that is not necessarily a stable isotopically labeled target analyte, but is treated as an

internal standard for quantitation. Surrogate ISs are used when isotopically labeled

counter parts of the target analyte are not commercially or re adily available.



## 3.18 Surrogate Recovery Standard (SRS)

An isotopically labeled standard , not used as an internal standard, that is added to each

sample and appropriate QC sample as a means to evaluate the method performance for a

chemical class of compounds (e .g., PFSAs, PFCAs)

.



## 3.19 Working Standard (WS)

A solution of several analytes prepared in the laboratory from SSSs and diluted as needed to

prepare calibration standards and other required analyte solutions.



## 4 Warnings and Cautions

4.1 Health and Safety

The acute and chronic toxicity of the standards for this method have not been precisely

determined; however, each should be treated as a potential health hazard. The analyst

should wear gloves, a lab coat, and safety glasses to prevent exposure to chemicals that

might be present.

### ETS -8- 044. 1 Page 6 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis The laboratory is responsible for maintaining a safe work environment and a current

awareness of local regulations regarding the handling of the chemicals used in this method.

A reference file of material safety data sheets (MSDS) should be available to all personnel

involved in these analyses.



## 4.2 Cautions

The analyst must be familiar with the laboratory equipment and potential hazards including,

but not limited to, the use of solvents, pressurized gas and solvent lines, high voltage, and

vacuum systems. Refer to the appropriate equipment procedure or operator manual for

additional information and cautions.



## 5 Interferences

During sample preparation and analysis, major potential contaminant sources are reagents

and glassware. All materials used in the analyses shall be demonstrated to be free from

interferences under conditions of analysis by running method blanks.

Parts and supplies that contain Teflon® should be avoided or minimized due to the possibility

of interference and/or contamination. These may include, but are not limited to: wash bottles,

Teflon ® 

### lined caps, autovial caps, HPLC parts, etc.

The use of disposable micropipettes or pipettes to aliquot standard solutions is recommended

to make calibration standards and matrix spikes.



## 6 Instrumen tation, Supplies, and Materials

6.1 Instrumentation

Analytical balance capable of reading to 0.0001g

HPLC/MS/MS or HPLC/MS system, as described in Section 10

.



## 6.2 Supplies and Materials

Sample collection bottles

— HDPE (e.g., NalgeneTM) wide

- mouth bottles with screw cap.

Note: Do not use fluorinated or Teflon® bottles or lined caps.

Coolers or boxes for sample shipment.

15

- mL and 50

- mL disposable polypropylene centrifuge tubes.

Class A pipettes and volumetric flasks, various.

2 mL HPLC auto vial s

Disposable pipettes, polypropylene or glass as appropriate

Centrifuge capable of spinning 15

- mL and 50

- mL polypropylene tubes at 3000 rpm.



## 7 Reagents and Standards

Note: Suppliers and catalog numbers are for illustrative purposes only. Equivalent

performance may be ach ieved using chemicals obtained from other suppliers. Do not use a

lesser grade of chemical than those listed.



## 7.1 Chemicals

Water

– Milli

- Q, HPLC grade, or other suitably appropriate sources

Calcium Acetate

- A.C.S. Reagent Grade

### ETS -8- 044. 1 Page 7 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis Magnesium Acetate

- A.C.S. Re agent Grade

Methanol

– HPLC grade

Ammonium Acetate

– A.C.S. Reagent Grade



## 7.2 Representative Target Analytes, ISs, and SRSs

PFBA, Heptafluorobutyric Acid, (C 4 

### Perfluorinated Acid)

PFPeA , Nonafluoropentanoic Acid (C 5 

### Perfluorinated Acid)

PFHxA, Perfluorohexanoi c Acid (C 6 

### Perfluorinated Acid)

PFHpA, Tridecafluoroheptanoic Acid, (C 7 

### Perfluorinated Acid)

PFOA, Ammonium perfluorooctanoate, (C 8 

### Perfluorinated Acid)

PFNA, Heptadecafluorononanoic Acid, (C 9 

### Perfluorinated Acid)

PFDA, Nonadecafluorodecanoic Acid (C 10 

### Per fluorinated Acid)

PFUnA, Perfluoroundecanoic Acid, (C 11 

### Perfluorinated Acid)

PFDoA, Perfluorododecanoic Acid, (C 12 

### Perfluorinated Acid)

PFTrDA, Perfluorotridecanoic Acid, (C 13 

### Perfluorinated Acid)

FBSA, Perfluorobutanesulfonamide

FOSA, Perfluorooctanesulfo nylamide

PFBS, Potassium Perfluorobutane sulfonate

PFHS, Perfluorohexanesulfonate

PFOS, Potassium perfluorooctanesulfonate

PFOA [1,2, 3, 4

- 13 

### C], 13 

### C 4 

### - isotopically labeled perfluorooctanoic acid (SRS)

PFOS [1,2, 3, 4

- 13 

### C], 13 

### C 4 

### - isotopically labeled Perfluorooctanesulfonate (SRS)

PFUnA [1,2

- 13 

### C], 13 

### C 2 

### - isotopically labeled Perfluoroundecanoic acid (SRS)

A custom mix of ISs in a methanolic solution containing ([1,2,3,4

- 13 

### C 4 

### ]PFBA, [1,2

-

13 

### C 2 

### ]PFHxA, [1,2,3,4 ,5,6,7,8

- 13 

### C 8 

### ]PFOA, [1,2,3,4,5 ,6,7,8,9

- 13 

### C 9 

### ]PFN A, [1,2

- 13 

### C 2 

### ]PFDA,

[1,2 ,3,4,5,6,7

- 13 

### C 7 

### ]PFUnA, [1,2

- 13 

### C 2 

### ]PFDoA, [1,2 ,3

- 13 

### C 3 

### ]PFHS, [1,2,3,4 ,5,6,7,8

- 13 

### C 8 

### ]PFOS

, and [1,2,3,4,5,6,7,8

- 13 

### C 8 

### ]PFOSA (Wellington Laboratories, Guelph, ON) in combination with

added ([1,2,3,4 ,5

- 13 

### C 5 

### ]PFPe A

, ([1,2,3,4

- 13 

### C 4 

### ]PFHpA, and [ 18 

### O 2 

### ]PFBS can be used to prepare

a stock IS solution. Alternatively, individual stable isotope ISs can be used to prepare a stock

IS mixture.

Other ISs can be applied.



## 7.3 Reagent Preparation

2 mM A mmonium acetate solution (Analysis)

— Weigh 0.3 g of A mmonium acetate and

dissolve in 2.0 L of reagent water.

Synthetic Groundwater (containing 25 ppm Ca and Mg)

– Weigh 0.61 g of Calcium Acetate

and 0.92 g of Magnesium Acetate and dissolve in 6.0 L of reagent water.

Note: Alternative volumes may be prepared as long as the ratios of the solvent to solute ratios

are maintained.

### ETS -8- 044. 1 Page 8 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis 

## 7.4 Stock Standard Solution (SSS) and Working Standard Solution

Preparation

The following standard preparation procedure serves as an example. Weighed amounts and

final volumes may be changed to suit the needs of a particular study. For example, μL

volumes may be spiked into volumetric flasks when diluting stock solutions to appropriate

levels.

100 μg/mL target analyte SSSs

— Weigh out 10 mg of analytical standard **(corrected for**

**percen t salt**  **, acid [ETS -4-**  **031] and purity**  ) and dilute to 100 mL with methanol or other

suitable solvent, in a 100 mL volumetric flask. Transfer to a 125 mL LDPE bottle or other

suitable container. Prepare a separate solution for each analyte. Expiration dates and

storage conditions of stock solutions should be assigned in accordance with laboratory

standard operating procedure. An example of purity and salt correction is given below for

PFOS.



### salt of weight moclecular anion of weight molecular factor correction salt

=

9275.0

538 499 factor correction )salt (K PFOS

== +



### 10 mg C 8 

### F 17 

### S0 3 - 

### K + 

### with purity 90% = 8.35 mg C 8 

### F 17 

### S0 3 – 

### (10 mg*0.90*0.9275=8.35 mg)

10 μg/mL (10,000 ng/mL) mixed working standard

— Add 5.0 mL each of the 100 μg/mL

SSSs to a 50 mL volumetric flask and bring up to volume with solvent.

1 μg/mL (1,000 ng/mL) mixed working stan dard

— Add 0.5 mL of the 100 μg/mL SSSs to a

50 mL volumetric flask and bring up to volume with solvent.

0.1 μg/mL (100 ng/mL) mixed standard

— Add 0.05 mL of the 100 μg/mL SSSs to a 50 mL

volumetric flask and bring up to volume with solvent.

Storage Conditi ons

— Store all SSSs and working standards in accordance with laboratory

standard operating procedure or in a refrigerator at 4°±2°C for a maximum period of 6 months

from the date of preparation.



## 7.5 Calibration Standards

Calibration can be performed by IS or external calibration. Using the working standards

described above, prepare calibration solutions in ASTM Type I water, HPLC water, other

suitable water, or a mixture of solvent and water using the information in Table 2 as a

guideline. Note: Volumes of wa ter or water/solvent mixtures and working standards may be

adjusted to meet the data quality objectives addressed in the general project outline.

Calibration levels other than those listed below can be prepared as needed.

For the quantitation of PFOA and PFOS, reference materials of certified mixed linear and

branched isomer are preferred. Alternately, reference materials of primarily linear isomers of

PFOA and/or PFOS may be used, however, when quantitating with predominantly linear

reference standards, additional LCS samples containing both linear and branched isomers of

PFOA and PFOS are required 3



### .

7.5.1 Internal Standard (IS) and Surrogate Recovery Standard (SRS)

For IS calibration, stable isotope internal standards of each target analyte or appropriate

surrogate ISs should be spiked at the same level in all calibration standards. Once the

calibration standards have been prepared as stated above in Section 7.5, all calibration

standards are spiked with a separate internal standard spiking solution. Typically the

3 A report summarizing an assessment of the use of reference standards containing certified linear and branched isomers of

PFOA/PFOS can be found in 3M report E11

- 0560.

### ETS -8- 044. 1 Page 9 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis concentration of the internal standard is consistent with the internal standard concentration

expected in the samples being prepared, usually 1 ng/mL. The concentration of the internal

standard spiking solution is typically 2 μg/ mL. A separate zero point or method blank is

typically prepared at the same time as the calibration standards, using the same solution

used to prepare the standards (ASTM Type I water, HPLC water, other suitable water, or a

solvent/water mixture), and is spiked with the internal standard at the same concentration as

the calibration curve, typically at 1 ng/mL.

If the sample s being analzyed w ere pre

- spiked with SRSs , the calibration curve prepared in

Section 7.5 is spiked with a separate SRS spiking solution. Typically, the sample bottles are

spiked with a SRS at 0.1 ng/mL. The final calibration curve must consist of at least six

calibration points after analysis. The following table provides an example of spike

concentrations and volumes used to achieve a multi

- point extracted calibration curve with

internal standard and surrogate standard.

Table 1 lists recommended stable isotope internal standards for several PFSA and PFCA

target compounds. A custom mix of isotopically labeled target analytes in a methanolic

solution containing ([1,2,3,4

- 13 

### C 4 

### ]PFBA, [1,2

- 13 

### C 2 

### ]PFHxA, [1,2,3,4,5,6,7,8

- 13 

### C 8 

### ]PFOA,

[1,2,3,4,5,6,7,8,9

- 13 

### C 9 

### ]PFNA, [1,2,3,4,5,6

- 13 

### C 6 

### ]PFDA, [1,2,3,4,5,6,7

- 13 

### C 7 

### ]PFUnA, [1,2

-

13 

### C 2 

### ]PFDoA, [ 1,2,3

- 13 

### C 3 

### ]PFHS, [1,2,3,4,5,6,7,8

- 13 

### C 8 

### ]PFOS, and [1,2,3,4,5,6,7,8

- 13 

### C 8 

### ]FOSA

(Wellington Laboratories, Guelph, ON) in combination with added ([1,2,3,4 ,5

- 13 

### C 5 

### ]PFPe A

, ([1,2,3,4

- 13 

### C 4 

### ]PFHp A , and [ 18 

### O 2 

### ]PFBS can be used to prepare a stock IS solution.

Alternative sources of certified stable isotope labeled target analytes are applicable.

Alternatively, individual stable isotope ISs can be used to prepare a stock IS mixture. The

table below lists the recommended stable isotope ISs and SRSs applied in the method.

Other stable isotope ISs and SRSs of target analytes not listed in the table may be used if

supported by validation and/or analysis batch QCs meeting method acceptance criteria (e.g.,

[ 13 

### C 2 

### ]- PFOA) . The same internal standard should be used for a given analyte throughout the

entire project/study. Note: some of the compounds listed below are appropriate to use as

surrogate ISs when a stable isotope IS of a target analyte is not available. Generally,

surrogate isotopically labeled PFCAs are used for PFCAs, and surrogate i sotopically labeled

PFSA s are used for PFSAs.

Table 2 provides examples of spike concentrations and volumes used to achieve a multi

- point

calibration curve with ISs and SRSs

.

### ETS -8- 044. 1 Page 10 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis Table 1 . Stable Isotope PFCAs and PFSAs used for ISs and SRSs

Compound Name Synonym or Acronym Analytical Purpose Reference Standard

Source

13 C 4 - Perfluorobutanoic acid [1,2,3,4

- 13 C 4 ]PFBA IS for PFBA Wellington Labs

(Mix or Individual)

13 C 4 - Perfluoro pentanoic acid [1,2,3,4 ,5

- 13 C 5 ]PFPe A IS for PFPeA Wellington Labs

(Mix or Individual)

13 C 2 - Perfluorohexanoic acid [1,2

- 13 C 2 ]PFHxA IS for PFHxA Wellington Labs

(Mix or Individual)

13 C 4 - Perfluoro hept anoic acid [1,2,3,4

- 13 C 4 ]PFHp A IS for PF HpA Wellington Labs

(Mix or Individual)

13 C 8 - Perfluorooctanoic acid [1,2,3,4,5,6,7,8

- 13 C 8 ]PFOA IS for PFOA and [1,2,3,4

13 C 4 ]PFOA Wellington Labs

(Mix or Individual)

13 C 9 - Perfluorononanoic acid [1,2,3,4,5,6,7,8,9

- 13 C 9 ]PFNA IS for PFNA Wellington Labs

(Mix or Individual)

13 C 6 - Perfluorodecanoic acid [1,2,3,4,5,6

- 13 C 6 ]PFDA IS for PFDA Wellington Labs

(Mix or Individual)

13 C 7 - Perfluoroundecanoic acid [1,2,3,4,5,6,7

- 13 C 7 ]PFUnA IS for PFUnA Wellington Labs

(Mix or Individual)

13 C 2 - Perfluorododecanoic acid [1,2

- 13 C 2 ]PFDoA IS for PFDoA, *PFTA Wellington Labs

(Mix or Individual)

18 O 2 - Ammonium Perfluorobutane sulfonate [ 18 O 2 ]PFBS IS for PFBS RTI International

(Individual)

13 C 3 - Ammonium Perfluorohexane sulfonate [ 1,2,3

- 13 C 3 ]PFHS IS for PFHS Wellington Labs

(Mix or Individual)

13 C 8 - Sodium Perfluorooctane sulfonate [1,2,3,4,5,6,7,8

- 13 C 8 ]PFOS IS for PFOS and

PFOS[1,2,3,4 13 C 4 ], Wellington Labs

(Mix or Individual)

13 C 8 - Perfluorooctanesulfonamide [1,2,3,4,5,6,7,8

- 13 C 8 ]FOSA IS for FOSA Wellington Labs (mix)

RTI International

(Individual)

13 C 4 - Perfluorooctanoic acid [1,2,3,4

- 13 C 4 ]PFOA SRS for all PFCAs : C4

- C8 Wellington

13 C 2 - Perfluoroundecanoic acid [1,2

- 13 C 2 ]PFUnA SRS for all PFCAs C9

- C13 Wellington

13 C 8 - Perfluorooctane sulfonate [1,2,3,4

- 13 C 4 ]PFOS SRS for all PFSA s: C4,

C6, and C8 Wellington

### ETS -8- 044. 1 Page 11 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; Direct Injection Analysis Table 2 . Example Preparation of Calibration Curve with ISs and SRSs

**Sample Description Concentration** 

**of WS, μ**  **g/mL Volume of** 

**WS,** μ **L Volume of IS** 

**(2 μ**  **g/mL),** μ **L Concentration of** 

**Surrogate, μ**  **g/mL Volume of** 

**Surrogate,** μ **L Volume of ASTM Type I**  **Water, or**

**other suitable solvent**  **(1) , mL** 

0.025 ng/mL curve point 0.10 25 50 0.20 12.5 100

0.030 ng/mL curve point 0.10 30 50 0.20 15 100

0.04 ng/mL curve point 0.10 40 50 0.20 20 100

0.05 ng/mL curve point 0.10 50 50 0.20 25 100

0.1 ng/mL curve point 0.10 100 50 0.20 50 100

0.25 ng/mL curve point 0.10 250 50 0.20 125 100

0.5 ng/mL curve point 1.0 50 50 0.20 250 100

1 ng/mL curve point 1.0 100 50 0.20 500 100

2.5 ng/mL curve point 10.0 25 50 10.0 25 100

5.0 ng/mL curve point 10.0 50 50 10.0 50 100

10.0 ng/mL curve point 10.0 100 50 10.0 100 100

25.0 ng/mL curve point 10.0 250 50 10.0 NA 100

50.0 ng/mL curve point 10.0 500 50 10.0 NA 100

75.0 ng/mL curve point 10.0 750 50 10.0 NA 100

100 ng/mL curve point 10.0 1000 50 10.0 NA 100

N/A

- Not Applicable

(1) Samples requiring analysis for PFUnA, PFDoA, PFTrDA, and FOSA should be analyzed against a calibration curve prepared in 1:1 synthetic groundwa ter:MeOH.

### ETS -8- 044. 1 Page 12 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis 

## 8 Sample Collection and Bottle Preparation

Sample collection bottles are prepared by 3M Environmental Laboratory (or subcontract supplier) personnel for

shipment at ambient temperature to the collection site. Typically, four separate collection bottles are

associated with a single collection site: sample, field duplicate sample, low field matrix spike, and high field

matrix spike. Alternatively, the sample and field duplicate sample may contain SRSs in lieu of additional target

analyte low field matrix spike and target analyte high field matrix spike samples. Depending on the scope of

the project, additional replicates of the field sample and field matrix spikes may be added. Also, it is not

uncommon for additional mid

- level field matrix spikes to be collected if the expected sample concentrations are

truly unknown or could span a large concentration range.

High

- density polyethylene (HDPE) wide

- mouth Nalgene bottles are used for the sample collection containers.

(Volumes of the bottles may vary depending on how much sample is required to meet data quality objectives.)

Sample collection volumes are project specific and based on data quality objectives. The Nalgene bottles do

not require any pretreatment prior to use. Typically, placement of a sample bottle volumetric “fill to here” line is

done by using a sample bottle marker template. Alternatively, bottles may be weighed prior to bottle

preparation and weighed again after samples have been collected.

All bottles should be clearly labeled to indicate its intended use as a sample, field sample duplicate, low field

matrix spike, high field matrix spike, sample/SRS field matrix spike, field duplicate sample/SRS field matrix

spike, trip blank, or trip blank matrix spike. If each location has different designated spike levels, the label

should also clearly indicate the sample location designation. Generally, a set of bottles for a given collection

site are prepared then grouped together in plastic bags for organizational purposes. For each sample

collection event, at least one set of trip blank and trip blank matrix spikes are prepared.

Bottle preparation should be documented in a Note to File or on a sample preparation worksheet and should

include the following information: date prepared, total number of bottles prepared, number of sample sites, the

standard identification numbers and spike volumes used to prepare spiked bottles, the “fill to here” volume, and

any other pertinent information needed for reconstructibility of the data. The Note to File will be included in the

final data package for the project.

Samples are collected in the field and shipped to the laboratory at ambient temperature.



## 8.1 Field Matrix Spike Sample (FMS)

Field matrix spike samples are a requirement of the method. A FMS sample is defined as a QC sample to

which known quantities of appropriate target analytes are added to the sample bottle in the field or in the

laboratory before the bottles are sent to the field. The sample and field duplicate sample may contain

appropriate SRSs in lieu of target analyte FMS samples. Sample quantities are determined vol umetrically or

gravimetrically. A known, specific volume or weight of sample is added to the sample container without

rinsing. Volumetric sample measurements may be acquired by a laboratory applied “fill to this level” line on the

outside of the sample container. Target analyte FMS samples should be spiked at approximately 0.5

- 10 times

the expected analyte concentration in the sample. If the expected range of analyte concentrations is unknown,

multiple spikes at varying levels may be prepared to increase the likelihood that a spike at an appropriate level

is made. Typically a low and a high target analyte spike are prepared for each sampling location. In those

instances where SRSs are to be used in lieu of target analyte FMS samples, the sample and field duplicate

sample are spiked at approximately 2

- 5 times the target LOQ. The FMS is analyzed to ascertain if matrix

effects or sample holding time contributes bias to the analytical results. For the sample bottles designated for

matrix spikes, an appropriate volume of matrix spiking solution is added to the empty bottle prior to sampling.

The volume of spike solution added should produce the desired final concentration of target analytes once the

bottle is filled with sample to the “fill to here line”. The matrix spiking solution(s) should be prepared in a

suitable solvent and contain all of the appropriate target analytes, ISs, and SRSs. The target analyte matrix

spiking solution is often the same as the working standards used to create the calibration standards. An

example of a bottle spike is given below.

“Fill to here” volume = 200 mL (A 250 mL Nalgene bottle is used)

Desired Field Spike Concentration = 0.25 ng/mL

### ETS -8- 044. 1 Page 13 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis 500 μ L of a 0.1 μ g/mL spiking solution (containing the target analytes) is added to the bottle and the bottle cap

promptly sealed.



## 8.2 Internal Standard and Surrogate Recovery Standard

If analysis of a surrogate recovery standard (SRS) is included in the project objectives, an appropriate volume

of a surrogate standard solution is added to all the bottles prior to sampling and SPE. Typically sample bottles

are spiked with surrogate recovery standards at a final desired spike concentration of 0.1 ng/mL.

If quantitation by internal standard (IS) is included in the project objective, an appropriate volume of internal

standard solution is added to all the bottles prior to sampling and SPE . Typically sample bottles are spiked

with internal standard at a final desired spike concentration of 1 ng/mL.

For the trip blank, the SRS spike and IS spike i s added to the bottle and then ASTM Type I water (HPLC grade

reagent water or other suitable water may used) is added to the “fill to here” line. The bottle is capped and

sealing tape may be placed around the outer edge of the cap. Trip blank matrix spikes are prepared by adding

the appropriate volume of target analyte spiking solution, IS, and SRS spiking solutions and filling the bottle to

the desired volume with the appropriate water and capping and sealing the cap.



## 9 Quality Control and Data Quality Objectives

9.1 Data Quality Objectives

This method and required quality control samples is designed to generate data accurate to ±30% with a

targeted LOQ of 0.025 ng/mL. Any deviations from the quality control measures spelled out below will be

documented in the raw data and footnoted in the final report.



## 9.2 Method/Procedural Blanks

The method/procedural blank is zero point calibration standard (which includes ISs) analyzed in a regular basis

with each analysis batch. At a minimum, method blanks are analyzed prior to instrument calibration, prior to

the analysis of CCV samples, after every 10 sample injections, and at the end of the analytical run.

The mean area count or area ratios when using internal standard calibration, for each analyte in the method

blanks m ust be less than 50% of the area count counts or area ratios when using internal standard calibration,

of the LOQ standard. The standard deviation of the area counts, or area ratios when using internal standard

calibration, of these method blanks should be calculated. A specific %RSD acceptance criteria is not specified

but is assessed on an analytical batch basis. If the mean area counts or area ratios when using internal

standard calibration, of the method blanks exceed 50% of the LOQ standard, then the LOQ must be raised to

the first standard level in the curve that meets criteria. Method blanks may be eliminated if technical

justification can be provided (e.g. the procedural blank was analyzed after an unexpectedly high level sample).

If any procedural blanks are removed from the LOQ determination, document in the raw data and report as

appropriate. Laboratory Sample Replicates / Field Duplicate Sample

Typically, samples are collected in duplicates in the field. The relative percent difference (RPD) of duplicate

samples should be ≤20% for the precision of sample preparation and analysis to be considered in control.

Replicate samples not meeting the ≤20% RPD criteria are flagged and reported as outside of QC acceptance

criteria.



## 9.3 Laboratory Matrix S pikes (LMSs)

LMSs may be performed in lieu of FMSs if FMSs have previously been performed for the sample matrix.

Additionally, LMSs may be performed in lieu of FMSs for a sample matrix if the FMS levels were not

appropriate for determining spike recoveries relative to endogenous levels of target analytes and appropriate

SRSs. Generally, each sample location represents a different sample and sample matrix. LMSs are prepared

for each sample and analyzed to determine the matrix effect on spike recovery efficiency of each target analyte

and appropriate SRSs. LMSs should be prepared at a minimum of one level and in duplicate. LMS

concentrations should be prepared at approximately 0.5

- 10 times the endogenous concentration or

approximately 4

- 10 times the LOQ concentration of each target analyte.

### ETS -8- 044. 1 Page 14 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis Lab matrix spike recoveries should fall within ± 30% of expected values. Sample data with LMS recovery

outside of ±30% but within ±50% of the expected value are flagged and reported as outside of QC acceptance

criter ia. Data with LMS recovery outside of ±50% of the expected value are reported as NR, where NR is

defined as “Not Reportable” data outside of QC acceptance criteria.



## 9.4 Lab Control Sample

Lab control spikes are prepared for each analysis batch to determine method accuracy and precision. LCSs

should be prepared at three levels in triplicate for each target analyte and at a minimum of two levels in

triplicate for appropriate SRSs. Low lab control spikes should be prepared at a concentration in the range of

a pproximately four to ten times higher than the targeted lower LOQ, the mid lab control spikes should be

prepared at a concentration near the mid

- point of the calibration curve and the high lab control spikes at

approximately 80% of the upper LOQ. For each target analyte and SRSs, the percent relative standard

deviation (method precision) for each control spike level must be less than or equal to 20% and the average

recovery (method accuracy) for each control spike level must be 80

- 120%. Sample data for target analytes

outside of the laboratory control spike acceptance criteria will be handled as follows:

If the average recovery of a spiking level falls outside method acceptance, but at least 67% (6 out of 9 ) o f LCS

samples are within 20 % of their respective nominal value (33% of the QC samples, not all replicates at the

same concentration, may be outside 20% of nominal value), the average recovery will be flagged as outside

method acceptance criteria. All LCS samples will be control charted as per ETS -4- 02 6. If the average

recovery of one of the spiking levels exceeded the analytical method uncertainty as determined by ETS

- 12

- 012, that analytical batch uncertainty will be expanded for that particular study.

If more than 67% of the LCS samples fail to meet method acceptance criteria, the data will not be reported.

Calibration standards consisting of mixed branched and linear isomer PFOS/PFOA are preferred. However,

for PFOS/PFOA target analytes, if the calibration standards are comprised of predominantly linear isomers

only, at least one level of triplicate LCSs should be prepared using PFOS/PFOA which contains a mix of linear

and branched isomers. These LCSs will be used to demonstrate quantitative equivalency (or quantitative bias)

of the isomeric mix when using a predominantly linear standard for calibration. The mixed linear and branched

isomer PFOS/PFOA LCSs recoveries should fall within ± 30% of expected values. Alternatively, in lieu of

mixed branched and linear isomer PFOS/PFOA LCSs, mixed branched and linear isomer PFOS/PFOA

TBMSs may be applied to demonstrate method accuracy and precision.



## 9.5 Field Matrix Spikes (FMSs) / Surrogate Recovery Standards (SRSs)

FMSs are prepared for each sampling location and analyzed to determine the matrix effect and sample holding

time on the spike recovery of each target analyte and/or appropriate SRS. Generally, each sample location

represents a different sample and sample matrix.

FMSs are QC samples to which known quantities of appropriate target analytes are added to the sample bottle

in the laboratory before the bottles are sent to the field. Typically a low and a high target analyte FMS are

prepared for each sampling location. The sample and field duplicate sample may contain appropriate SRSs in

lieu of target analyte low field matrix spike and target analyte high field matrix spike samples.

Field matrix spike method acceptance criteria are recoveries within ±30% of the expected value. If FMS

recovery (target analyte or SRS spike) is outside of ±30% of the expected value or could not be assessed

because the FMS (target analyte) was spiked at an inappropriate level, the sample result is reported as follows:

1.) If target analyte FMS recovery could not be assessed because the FMS's were at an inappropriate level,

then Laboratory Matrix Spikes (LMS) may be substituted. If LMS recoveries are within ±30% the data are

reportable and flagged to indicate that the FMS spikes levels were inappropriate.

2.) If multiple target analyte FMS's were prepared on a sample and the closest FMS level to the reported

sample meets the ±30% acceptance criteria but additional FMS's are outside the ±30% acceptance range, the

data are reportable and flagged to indicate that while there were failing FMS’s, the uncertainty will not be

expanded since the most appropriate spike level passed.

### ETS -8- 044. 1 Page 15 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis 3.) If the target analyte FMS recoveries are outside of the ±30% acceptance range but at least 30 acceptable

historical reportable FMS sample results are available, the data may be reported but flagged with an expanded

uncertainty and as not meeting FMS criteria.

4.) Sample data with FMS recovery outside of ±30% but within ±50% of the expected value are flagged and

reported as outside of QC acceptance criteria with an expanded uncertainty.

5.) If FMS recovery is outside of ±50%, the sample result is reported as NR, where NR is defined as “Not

Reportable” due to noncompliant QC results.

The targeted fortification levels should be at least 50% of the endogenous level and less than 10 times the

e ndogenous level to be used without justification to determine the statement of accuracy for analytical results.

Note: It is possible for bottles utilized for Field Matrix Spike samples to be under

- filled or over

- filled during

sample collection. Since this scenario will effect the actual concentration of the FMS sample (surrogate and

internal standard concentrations will also be effected, if used), it is important that any obvious under

- filling or

over

- filling of sample bottles be documented in the data package and taken into account in the FMS , ISs, or

SRSs recovery calculations. Samples over

- filled or under

- filled by more than 10% will be require recalculation

of the FMS, ISs, and SRS true values.

The average of the sample and the field duplicate should be used to calculate the recovery.



## 10 Procedures

10.1 Water Sample Preparation

This method is applicable to water samples. Samples containing heavy particulate may not be suitable for

analysis by this method. Samples containing suspended particulate should be centrifuge prior to removing a

sample aliquot, or filtered.

• Thoroughly mix sample before removing a n aliquot and placing in a labeled autovial.

• Dilute sample, if necessary, with ASTM Type I water , HPLC water, other suitable water , or solvent

(methanol)

.

• L ab control spikes are prepared for each analysis batch to determine method accuracy and precision.

LCSs should be prepared at three levels in triplicate for each target analyte and at a minimum of two

levels in triplicate for appropriate SRSs. Low lab control spikes should be prepared at a concentration

in the range of approximately four to ten times higher than the targeted lower LOQ, the mid lab control

spikes should be prepared at a concentration near the mid

- point of the calibration curve and the high

lab control spikes at approximately 80% of the upper LOQ. For IS quantitation, stable isotope internal

standards of each target analyte or appropriate surrogate ISs should be spiked at the same level as the

samples being analyzed, in all LCSs

.

• If LCSs are being prepared us ing synthetic groundwater, allow the LCSs samples to equilibrate for a

minimum of 4 hours before aliquoting for analysis or diluting with solvent (methanol).



## 11 Sample Analysis

- LC/MS/MS

11.1 Instrument Setup

Note: In this example, an Applied Biosystems Sciex API 4000 (API 5000 or API 5500) Tandem Mass

Spectrometer (LC/MS/MS) is used. Other brands/models of LC/MS/MS instruments as well as single

quadrupole mass spectrometers (LC/MS) may be used as long as the method acceptance criteria are met.

Brand names, suppliers, part numbers, and models are for illustrative purposes only. Equivalent performance

may be achieved using apparatus and materials other than those specified here, but demonstration of

equivalent performance that meets the requirements of this method is the responsibility of the laboratory. The

operator must optimize and document the equipment and settings used.

### ETS -8- 044. 1 Page 16 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis Establish the LC/MS/MS system and operating conditions equivalent to the following:

Mass Spec: Applied Biosystems API 4000, API 5000, or API 5500

Ion Source: Turbo Ion Spray (ABS)

Mode: Electrospray Negative

Scan Type: MRM (Multiple Reaction Monitoring)

Computer: Dell DHM

Software: Windows 2000 or Windows XP, Analyst 1.4.2 or higher versi ons

HPLC: Agilent Series 1100,1200, or 1290

Agilent Quaternary Pump

Agilent Vacuum Degasser

Agilent Autosampler

Agilent Column Oven

Note: One or more C18 HPLC analytical columns (2.1 mm x 100 mm, 5 μ m or 2.1 mm x 50 mm, 5 μ m) may be

attached on

- line after the purge valve and before the sample injection port to retard and separate any residue

contaminants that may be in the mobile phase and/or HPLC system.

HPLC Column: Betasil C18, 4.6mm x 100mm, 5 μ m (ThermoElectron Corporation)

Column Temperature: 35 ° C

Inje ction Volume: 5μL

Mobile Phase (A): 2mM Ammonium Acetate in ASTM Type I water (See 7.3)

Mobile Phase (B): Methanol

Table 3

. Liquid Chromatography Gradient Program

.

**Step**

**Number Total Time** 

**(min) Flow Rate** 

**(** μ

**L/min) Percent A** 

**(2 mM ammonium** 

**acetate) Percent B** 

**(Methanol)**

0 0 750 97.0 3.0

1 2.0 750 97.0 3.0

2 14.5 750 5.0 95.0

3 15.5 750 5.0 95.0

4 16.5 750 97.0 3.0

5 20.0 750 97.0 3.0



### Note: Other HPLC gradients may be used as long as the method criteria and project data quality objectives are

met.

It may be necessary to adjust the HPLC gradient in order to optimize instrument performance. Columns with

different dimensions (e.g. 2.1mm x 30mm) and columns from different manufacturers (Keystone Betasil C18

etc.) may be used.

### ETS -8- 044. 1 Page 17 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis Table 4 Suggested MRM Transitions for Target Analytes, Surrogates, and Internal Standards

Analyte Analyte Description Mass Transition

Q1 (amu) Mass Transition

Q3 (amu)

PFBA (C4 Acid) Target 213 169

PFPeA (C5 Acid) Target 263 219

PFHxA (C6 Acid) Target 313 269, 119

PFHpA (C7 Acid) Target 363 319, 169

PFOA (C8 Acid) Target 413 369, 219, 169

PFNA (C9 Acid) Target 463 419, 169, 219

PFDA (C10 Acid) Target 513 469, 269, 219

PFUnA (C11 Acid) Target 563 519, 269, 219

PFDoA (C12 Acid) Target 613 569, 169, 319

PFTA (C13 Acid) Target 663 619, 369, 319

FBSA (C4 Sulfonamide) Target 298 78

FOSA (C8 Sulfonamide) Target 498 78

PFBS (C4 Sulfonate) Target 299 99, 80

PFHS (C6 Sulfonate) Target 399 99, 80

PFOS (C8 Sulfonate) Target 499 80, 99, 130

[1,2,3,4

- 13 C 4 ]PFBA IS for PFBA 217 172

[1,2 ,3,4,5

- 13 C 5 ]PFPe A IS for PFPeA 268 223

[1,2

- 13 C 2 ]PFHxA IS for PFHxA 315 270

[1,2,3,4

- 13 C 4 ]PFHpA IS for PFHpA 367 322

[1,2,3,4,5,6,7,8

- 13 C 8 ]PFOA IS for PFOA 421 376

[1,2,3,4,5,6,7,8,9

- 13 C 9 ]PFNA IS for PFNA 472 427

[1,2,3,4,5,6

- 13 C 6 ]PFDA IS for PFDA 519 474

[1,2,3,4,5,6,7

- 13 C 7 ]PFUnA IS for PFUnA 570 525

[1,2

- 13 C 2 ]PFDoA IS for PFDoA and PFTA 615 570

[ 18 O 2 ]PFBS IS for PFBS 303 84

[ 1,2,3

- 13 C 3 ]PFHS IS for PFHS 402 80

[1,2,3,4

- 13 C 4 ]PFOS IS for PFOS 503 80

[1,2,3,4,5,6,7,8

- 13 C 8 ]FOSA IS for FOSA 507 80

[1,2,3,4

- 13 C 4 ]PFOA Surrogate (C4

- C8 Acids) 417 372

[1,2,3,4

- 13 C 4 ]PFOS Surrogate(Sulfonates, FOSA) 503 80

[1,2

- 13 C 2 ]PFUnA Surrogate (C9

- C13 Acids) 565 520



### Multiple transitions for monitoring the analytes is an option. The use of one daughter ion is acceptable if data

sensitivity and selectivity is achieved and provided that retention time criteria are met to assure adequate

specificity. While the daughter ions may be chosen at the discretion of the analyst, mass transition 99 is

suggested for PFOS. Quantitation may be performed using the total ion chromatogram (TIC, or summed

MRMs) for a given analyte. For example, the PFOA TIC would sum all three of the monitored transitions. Use

of the suggested primary ion is recommended. Retention times may vary slightly, on a day

- to

- day basis,

depending on the batch of mobile phase and the gradient, column, guard column(s) used etc. Drift in retention

times is acce ptable within an analytical run, as long as the drift continues through the entire analysis and the

standards are interspersed throughout the analytical run.



## 11.2 Calibration Curve

Quantitation is by internal standard or external standard calibration. Calibration standards may be prepared in

ASTM Type I, HPLC water, other suitable water, or a solvent/water mixture. If internal standard calibration

does not meet calibration acceptance criteria, external calibration can be applied. See Table 1 for

### ETS -8- 044. 1 Page 18 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis recom mended application of available internal standards. Quantitation of PFOA and PFOS is by summed

analyte

- specific mass transitions.

Analyze the standard curve prior to each set of samples. If internal standards were added to the calibration

standards area ratios are used to generate the calibration curve. The standard curve may be plotted using a

linear regression (y = mx + b), weighted 1/x or unweighted, or by quadratic fi t (y = ax 2 

### + bx + c), weighted 1/x or

unweighted

, using suitable software

. The mathematical method used to calculate the calibration curve should

be applied consistently throughout a study. Any change should be thoroughly documented in the raw data.

High and/or low points may be excluded from the calibration curves to provide a better fit over the range

appropriate to the data or because they did not meet the pre

- determined acceptance criteria. Low

- level curve

points should also be excluded if their area counts (or area ratio if quantitating by IS) are not at least twice that

of the ave rage area counts (or area ratio if quantitating by IS) of method and/or solvent blanks. The coefficient

of determination (r 2 

### ) value for the calibration curve must be greater than or equal to 0.990 (or a correlation

coefficient (r) of 0.995). Each point in the curve must be within ± 25% of the theoretical concentration with the

exception of the LLOQ, which may be within ± 30%. Justification for exclusion of calibration curve points will be

noted in the raw data. A minimum of 6 points will be used to construct the calibration curve.

If the calibration curve does not meet acceptance criteria, perform routine maintenance or prepare a new

standard curve (if necessary) and reanalyze.



## 11.3 Continuing Calibration Verification (CCV)

Continuing calibration verifications (CCV) are analyzed to verify the accuracy of the calibration curve. Analyze

a mid

- range calibration standard, one of the same standards used to construct the calibration curve, at a

minimum after every tenth sample, not including solvent blanks, with a minimum of one per sample set.

Calibration verification injections must be within ± 25% to be considered acceptable. The calibration curve and

the last passing CCV will then bracket acceptable samples. Multiple CCV levels may be used. Samples must

be bracketed by passing CCVs or the calibration curve and a passing CCV to be reportable.



## 11.4 System Suitability

A minimum of three system suitability samples should be injected at the beginning of each analytical run, prior

to th e analysis of the calibration curve. Typically these samples are at a concentration near the mid

- level of the

calibration curve and are repeated injections from one autosampler vial. It is suggested that the system

suitability injections have area counts or area ratios when using internal standard calibration, with a target RSD

of ≤5% and a target retention time RSD of ≤2%. There is no defined acceptability limit on these results as the

%RSD value is dependent on the number of MRM transitions being monitored in the LC/MS/MS run or time

period. Ultimately, any effects on these parameters for the System Suitability samples will also be evident on

all standards and QC samples analyzed as part of the analysis batch. Any effect of system suitability is

incorp orated within QC acceptance criteria. 4



## 11.5 Sample Analysis and QCs

For each analysis batch, the instrument analysis run sequence should include an initial calibration curve,

samples, FDSs, interspersed blanks, interspersed CCVs, appropriate QCs (i.e., LCSs, LMSs, FMSs, TBMSs,

and TBs), and a final CCV or calibration curve bracketing samples and appropriate QCs

Inject the same volume (between 5

- 100μL) of each standard, analytical sample and blank into the instrument

(unless an on

- instrument sample dilution is desired).

Samples containing analytes that are quantitated above the concentration of the highest standard in the curve

should be further diluted and reanalyzed.

4 3M Environmental Laboratory study E08

- 0096 evaluated the effect on these results as a function of the number of MRMs being monitored.

### ETS -8- 044. 1 Page 19 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis 

## 12 Data Analysis and Calculations

The chromatography analysis software will typically calculate the amount of target analyte in the sample

extracts using the established calibration curve.

Calculate the percent recovery of the LCS using the following equation:



### 100%*

)

mL ng (ion Concentrat Spike )

mL ng (ion Concentrat LCS

recovery LCS%

=

Calculate the percent recovery of the LMS using the following equation:

100%*

)

mL ng ( ion Concentrat Spike )

mL ng ( Sample of ion Concentrat -)

mL ng ( ion Concentrat LMS

recovery % LMS

=



### For samples fortified with known amounts of analyte prior to extraction, use the following equation to calculate

the percent recovery.



### 100 (ng/mL) added Analyte (ng/mL) sample in found analyte Average- (ng/mL) found analyte Total Recovery × =



## 13 Analysis Batch Method Performance Criteria

Any method performance parameters that are not achieved must be considered in the evaluation of the data.

Nonconformance to any specified parameters must be described and discussed in the final report if the

Technical Manager (non

- GLP study) or Study Director (GLP study) chooses to report the data.

If criteria listed in this method performance section are not met, maintenance may be performed on the system

and samples reanalyzed, or other actions taken as appropriate. Document all actions in the raw data.

If data are to be reported when performance criteria have not been met, the data must be footnoted on tables

and discussed in the text of the report.



## 13.1 System Suitability

– Analysis Batch

A minimum of three system suitability samples should be injected at the beginning of each analytical run.

These samples are run prior to the calibration curve. It is suggested that the system suitability injections have

area counts with a target RSD of ≤5% and a target retention time RSD of ≤2%. There is no defined

acceptability limit on these results as the %RSDs are dependent on the number of MRM transitions being

monitored in the LC/MS/MS run or time period. Any effect of system suitability is incorporated in the QC

acceptance criteria.



## 13.2 Calibration and Limit of Quantitation (LOQ)

– Analysis Batch

Calibration Curve: The coefficient of determination (r 2 

### ) value for the calibration curve must be greater than or

equal to 0 .990 corresponding to a correlation coefficient (r) = 0.995. Each point in the curve must be within

± 25% of the theoretical concentration with the exception of the LLOQ, which may be within ± 30%.

CCV Performance: The calibration standards that are interspersed throughout the analytical sequence are

evaluated as continuing calibration verifications in addition to being part of the calibration curve. The accuracy

### ETS -8- 044. 1 Page 20 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis of each curve point must be within 25% of the theoretical value ( within 30% for lowest curve point ) . Samples

that are bracketed by CCVs not meeting these criteria must be reanalyzed.

Limits of Quantitation (LOQ): The lower LOQ (LLOQ) is the lowest non

- zero active standard in the

calibration curve; the peak area of the LLOQ must be at least 2X that of the average area counts for all

prepared procedural blank(s). By definition, the measured value of the LLOQ must be within 30% of the

theoretical value.

Demonstration of Specificity: Specificity is demonstrated by chromatographic retention time (within 4% of

standard) and the mass spectral response of unique ions.



## 13.3 Blanks

– Method/Procedural Blanks and Trip

Method/ Procedural Blanks: Multiple procedural blanks should be interspersed throughout the analysis batch

and the analytical sequence. At a minimum, method blanks are analyzed prior to instrument calibration, prior

to the analysis of CCV samples, after every 10 sample injections, and at the end of the analytical run.

The mean area counts (or area ratios when using IS calibration) for each analyte must be less than 50% of the

area count of the LOQ standard. If the area counts of the procedural blanks exceed 50% of the LOQ standard,

then the LOQ must be raised to the first standard level that meets criteria.

Trip Blank: A t rip blank of ASTM Type I water (or lab equivalent) is prepared in a sample container in the

laboratory and treated as a sample, including exposure to shipping, sampling site conditions, storage,

preservation and all analytical procedures. The trip blanks results for each analyte are included with the

repo rted sample results.



## 13.4 Data Accuracy and Precision

– Analysis Batch

Lab Control Spikes: The average recovery at each LCS level for each target analyte and appropriate SRS

should be within 80

- 120% and the percent relative standard deviation of the recoveries must be less than or

equal to 20%. If the average recovery of a spiking level falls outside method acceptance, but at least 67% (6

out of 9 ) o f LCS samples are within 20 % of their respective nominal value (33% of the QC samples, not all

replicates at the same concentration, may be outside 20% of nominal value), the average recovery will be

flagged as outside method acceptance criteria. All LCS samples will be control charted as per ETS

- 12

- 012. If

the average recovery of one of the spiking levels exceeded the analytical method uncertainty as determined by

ETS

- 12

- 012, that analytical batch uncertainty will be expanded for that particular study. The average recovery

at each LCS level for mixed branched/linear isomer PFOA and PFOS should be within 70

- 130% and the

percent relative standard deviation of the recoveries must be less than or equal to 20%.

Field Duplicates: The relative percent difference (RPD) of duplicate samples should be less than 20% for the

precision of sample preparation and analysis to be considered in control. Replicate samples not meeting the

20% RPD criteria are flagged and reported as outside of QC acceptance criteria.

Field Matrix Spikes: FMS acceptance criteria are recoveries within ±30% of the expected value for each

target analyte and appropriate SRS. Sample data with FMS recovery outside of ±30% but within ±50% of the

expected value are flagged and reported as outside of QC acceptance criteria. Data with FMS recovery

outside of ±50% of the expected value are reported as NR, where NR is defined as “Not Reportable” data

outside of QC acceptance criteria. If FMS recovery could not be assessed because FMSs were at an

inappropriate level, then Laboratory Matrix Spikes (LMSs) may be substituted. If LMS recoveries are within

±30% for each target analyte and SRSs the data are reportable but flagged as not meeting the FMS method

acceptance criteria.



## 13.5 Analytical Method Uncertainty

Analytical method uncertainty for each target analyte and SRS is determined with control charted historica l

analysis batch LCS data for the method and reported with each analysis batch. 5 

### Uncertainty determinations

5 

### Method uncertainty based on INTERNATIONAL ANS/ISO/IED STANDARD 17025 reference (GUM, Guide to the Expression of

Uncert ainty in Measurement). Method application demonstrated in ETS

- 12

- 012, citing references: a.) EURACHEM/CITAC Guide,

“Quantifying Uncertainty in Analytical Measurement,” Second Edition; Editors: S.L.R. Ellison, M. Rosslein, and A. Williams

. b.)Georgian, Thomas, “Estimation of Laboratory Analytical Uncertainty Using Laboratory Control Samples,” Environmental Testing &

### ETS -8- 044. 1 Page 21 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis are based on INTERNATIONAL ANS/ISO/IED STANDARD 17025 reference (GUM, Guide to the Expression

of Uncertainty in Measurement) and described in ETS -12- 012 . At least thirty data points are required for

determining analytical method uncertainty. The method uncertainty is defined as 2x the standard deviation of

the percent recoveries of the pooled lab control spikes. While all LCS data points are control charted, only the

most recent fifty data points are used for determining the method uncertainty.

When less than thirty LCS data points have been generated for a given analyte, the analysis batch LCSs are

used to determine the data uncertainty. If FMSs meet the ±30% recovery criteria at a level appropriate to the

endogenous level, and the LCS meet the ±20% recovery criteria, then the uncertainty of the data is determined

as within 100±20%.

Analysis batch sample data with FMS recovery outside of ±30% but within ±50% of the expected value are

flagged and reported as outside of QC acceptance criteria with expanded uncertainties. Data with FMS

recovery outside of ±50% of the expected value are reported as NR, where NR is defined as “Not Reportable”

data outside of QC acceptance criteria. If FMS recovery could not be assessed because FMSs were at an

inappropriate level, then Laboratory Matrix Spikes (LMSs) may be substituted. If LMS recoveries are within

±30% for each target analyte and appropriate SRSs the data are reportable but flagged as not meeting the

FMS method acceptance criteria with uncertainties of ±30%. If FMS do not meet the ±30% recovery criteria,

and historical FMS data does not exist, the analytical uncertainty is evaluated on a sample

- by -s ample basis,

the data may be reported with expanded uncertainty and are flagged.



## 13.6 Quantitation of PFOA/PFOS

– Analysis Batch

Calibration standards consisting of mixed branched and linear isomer PFOS/PFOA are preferred.

Quantitation is performed by integrating the linear and branched isomers together. Alternately, the linear and

branched isomers can be integrated separately, applying the appropriate true value to each calibration curve

po int for each isomer. The LCS and samples are then quantitated by integrating the linear and branched

isomers separately (requires separate analytical results files) and quantitating the resulting peak against the

linear or branched calibration curve

. The results from both integrations are then summed to produce the final

result. Integrating the linear and branched isomers separately may be helpful for those samples where the

linear/branched ratios do not closely match those of the reference standards.

Ho wever, for PFOS/PFOA target analytes, if the calibration standards are comprised of predominantly linear

isomers only t he method requires the addition of LCS s of mixed branched/linear isomer PFOS/PFOA. The

purpose of including these LCSs is to demonstrate quantitative equivalency (or quantitative bias) of the

isomeric mix when using a predominantly linear PFOS or PFOA standard for calibration. Alternatively, in lieu

of mixed branched and linear isomer PFOS/PFOA LCSs, mixed branched and linear isomer PFOS/ PFOA

TBMSs may be applied to demonstrate method accuracy and precision.

An alternate method of quantitation can be performed whereby only the linear isomer of PFOS/PFOA is

integrated and used for generating the calibration curve. The LCS and samples are then quantitated by

integrating the linear and branched isomers separately (requires separate analytical results files) and

quantitating the resulting peak against the linear calibration curve. The results from both integrations are then

summed to produce the final result. Integrating the linear and branched isomers separately reduces the on

column concentration for those samples that contain both linear and branched isomers of PFOA/PFOS. This

ensures that the concentration detected is within the a range of the calibration curve that is comparable

regardless of whether the calibration curve was generated using predominantly linear isomers of PFOS/PFOA

or linear plus branched isomers of PFOS/PFOA.



## 14 Pollution Prevention and Waste Management

Waste generated when performing this method will be disposed of appropriately. The original samples will be

archived at the 3M Environmental Laboratory in accordance with internal procedures.

Analysis, November/December 2000. c.)Taylor, B.N. and CE. Kuyatt, NIST Technical Note 1297, 1994 Edition: “Guidelines for

Evaluating and Expressing the Uncertainty of NIST Measurement Results.”d.)Adams, T.M., “A2LA Guide for the Estimation of

Measurement Uncertainty in Testing”, July 2002.

### ETS -8- 044. 1 Page 22 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis 

## 15 Records

Each data package generated for a study must include all supporting information for reconstruction of the data.

Information for the data package must include, but is not limited to the following items: study or project

number, sample and standard prep sheets/records, instrument run log (instrument batch records, instrument

ac quisition method, summary pages), instrument results files, chromatograms, calibration curves, and data

calculations.



## 16 Affected Documents

None.



## 17 Revisions

Revision

Number Summary of Changes

1 Section 1

. Included the use of internal standard calibration by this method.

Section 2. Included the use of internal standard calibration by this method. Included the

use of a solvent/water mixture when analyzing for PFUnA, PFDoA, PFTrDA, and FOSA.

Section 3

. Added definitions for internal standard, surrogate internal standard, and

surrogate recovery standard.

Section 6.Removed the details regarding the instrument parameters to section 10 of the

method

. Section 7. Updated reference standards to include internal standards and surrogates.

Changed concentration levels for working standards and included the use of internal

standards and surrogates.

Section 8. Inserted a new section on sample bottle preparation.

Se ction 9 Quality Control. This section was previously section 10 in ETS -8- 044.0. Updated

QC criteria to be consistent with method ETS -8- 154.4.

Section 10 Procedures . This section was previously section 8 (Sample Handling) in E TS -8-

044.0. Added detail regarding the preparation of LCSs. Included the use of methanol as a

dilution solvent.

Section 11 Samp le Analysis

. This section was previously section 10 in ETS -8- 044.0.

Included the details regarding the instrument parameters.

Section 12 Data Analysis and Calculations. This section was previously section 11 in ETS

8- 044.0. Removed the equation for calculating the analytes concentration, indicating that

this is done by the instrument software.

Section 13 Method Performance. This section was previously section 12 in ETS -8- 044.0.

Updated QC criteria to be consistent with ETS -8- 154.4. Added information on the

determination of analytical method uncertainty and quantitation of PFOA/PFOS.

Section 14 Pollution Prevention. This section was previously section 13 in ETS -8- 044.0.

Section 15 Records. This section was previously section 14 in ETS -8- 044.0.

Section 16 Affected Documents. This section was previously section 15 in ETS -8- 044.0.

Section 17 Revisions. This section was previously section 16 in ETS

- 8

- 044.0.

# A TTACHMENT 

# D: D EVIATION 

# ( S 

# )

# R ECORD OF 

# D EVIATION 

# /N ONCONFORMANCE

ETS -4- 008.7

Documentation of Deviations and Control of Nonconforming Testing Page 1 of 1 

## I. Identification

Study / Project No.

GLP10

- 01

- 0 2

- 2 7 Date(s) of Occurrence:

4/11/12 , 4/30/12 Document Number:

ETS

- 8

- 044. 1

Deviation type  SOP  Equipment Procedure  Method

(Check one)  Protocol  GPO  Other:



## II. Description (attach extra pages as needed)



## Method Requirements:

1. FMS re covery within ±30% (section 13.4 ).

2. RPD values ≤20% (section 13.4).



## Actual procedure/process:

1. The following samples had recoveries that did not meet acceptance criteria for PFOS: DAL GW 310R LS

(68.0%), DAL GW 330R HS (66.0%), DAL GW 330L HS (66.6%), DAL GW RW331S LS (64.1%) and HS

(69.1%), and DAL GW GRS04 HS (45.9%).

2. The sampling location DAL GW 220R had a sample/sample duplicate RPD of 2 6 % for PFOS and DAL

GW 222R had a RPD of 25 % for PFOS

.



## III. Actions Taken

(such as amendment issued, SOP revision, etc.)



## Corrective Action (  Yes  No) Ref erence :

Acceptability of the nonconforming work:

1. The sampling location s with FMS recoveries outside method acceptance criteria will be flagged in the

report . T he analytical uncertainty for sampling location s DAL GW 310R and DAL GW RW331S will be

expanded for PFOS based on the %bias of the most appropriate spike level

. Since the %bias for

sampling locations DAL GW 330R and DAL GW 330L were within the analytical method uncertainty for

PFOS by external standard calibration (± 34%) , no adjustment will be made to the sampling location

analytical uncertainty. Sampling location DAL GW GRS04 will not be reported for PFOS due to the only

appropriate spike level having a recovery below 50%.

2. The non

- compliant RPDs will be flagged in the final report.



### Actions:  Halting of Work  Client Notification  Work Recall  Withholdin g of Report

 Other: Deviations will be noted in final report.



## Project Lead /PAI Approval: Date:

Study Director (if GLP): Date:

Sponsor Approval (for GLP protocol deviations): NA Date: NA

T echnical Reviewer (optional): NA Date: NA

Laboratory Department Manager Approval: Date:



## IV. Authorization to Resume Work

Where halting of work occurred, resumption of work must first be approved by Laboratory Management



## Laboratory Department Manager Approval: NA Date: NA



## Deviation No. ______________

(assigned by Study Director or Team Leader at the end of study or project)