### 13339 Hagers Ferry Road Huntersville, NC 28078-7929 McGuire Nuclear Complex - MG03A2 Phone: 980-875-5245 Fax: 980-875-4349 

# Order Summary Report 

# Analytical Laboratory

Order Number: J12110390 Project Name: WWTS - Biweekly (1)

L ab Contact: Jason C Perkins


### Date: 12/17/2012 Customer Address: 253 Plant Allen Road Customer Name(s): BillK-RonL--RobnJ-DonS

Belmont, NC 28012

Phone: 980-875-5348
Report Authorized By: (Signature)



### Program Comments: Please contact the Program Manager (Jason C Perkins) with any questions regarding this report.
Data Flags & Calculations: Any analytical tests or individual analytes within a test flagged with a Qualifier indicate a deviation from the method quality system or quality control requirement. The qualifier description is found at the end of the Certificate of Analysis (sample results) under the qualifiers heading. All results are reported on a dry weight basis unless otherwise noted. Subcontracted data included on the Duke Certificate of Analysis is to be used as information only. Certified vendor results can be found in the subcontracted lab final report. Duke Energy Analytical Laboratory subcontracts analyses to other vendor laboratories that have been qualified by Duke Energy to perform these analyses except where noted. Data Package: This data package includes analytical results that are applicable only to the samples described in this narrative. An estimation of the uncertainty of measurement for the results in the report is available upon request. This report shall not be reproduced, except in full, without the written consent of the Analytical Laboratory. Please contact the Analytical laboratory with any questions. The order of individual sections within this report is as follows: Job Summary Report, Sample Identification, Technical Validation of Data Package, Analytical Laboratory Certificate of Analysis, Analytical Laboratory QC Reports, Sub-contracted Laboratory Results, Customer Specific Data Sheets, Reports & Documentation, Customer Database Entries, Test Case Narratives, Chain of Custody (COC)
Certification: The Analytical Laboratory holds the following State Certifications : North Carolina (DENR) Certificate #248, South Carolina (DHEC) Laboratory ID # 99005. Contact the Analytical Laboratory for definitive information about the certification status of specific methods.

## Sample ID's & Descriptions:

Sample ID Plant/Station Collection Date and Time Collected By Sample Description 2012025277 ALLEN 26-Nov-12 9:30 AM Chris Williams FGD Purge Eff 2012025278 ALLEN 26-Nov-12 9:41 AM Chris Williams EQ Tank Eff 2012025279 ALLEN 26-Nov-12 9:56 AM Chris Williams BioReactor 1 Inf 2012025280 ALLEN 26-Nov-12 10:14 AM Chris Williams BioReactor 2 Inf 2012025281 ALLEN 26-Nov-12 10:31 AM Chris Williams BioReactor 2 Eff 2012025282 ALLEN 26-Nov-12 10:50 AM Chris Williams Filter Blk 2012025284 ALLEN 26-Nov-12 9:56 AM Chris Williams BioReactor 1 Inf 2012025285 ALLEN 26-Nov-12 9:56 AM Chris Williams Hg Blk BioReactor 1 Inf 2012025286 ALLEN 26-Nov-12 10:14 AM Chris Williams BioReactor 2 Inf 2012025287 ALLEN 26-Nov-12 10:14 AM Chris Williams Hg Blk BioReactor 2 Inf 2012025288 ALLEN 26-Nov-12 10:31 AM Chris Williams BioReactor 2 Eff 2012025289 ALLEN 26-Nov-12 10:31 AM Chris Williams Hg Blk BioReactor 2 Eff 12 Total Samples

### COC and .pdf report are in agreement with sample totals

and analyses (compliance programs and procedures). All Results are less than the laboratory reporting limits. All laboratory QA/QC requirements are acceptable. Yes No



# Technical Validation Review Checklist:

Yes No

Yes No



## Report Sections Included:

Job Summary Report Sub-contracted Laboratory Results Sample Identification Customer Specific Data Sheets, Reports, & Documentation Technical Validation of Data Package Customer Database Entries Analytical Laboratory Certificate of Analysis Analytical Laboratory QC Report Chain of Custody

Reviewed By: 

### DBA Account Date: 

### 12/17/2012 Electronic Data Deliverable (EDD) Sent Separately

# Certificate of Laboratory Analysis This report shall not be reproduced, except in full.



### Order # J12110390

2012025277 Collection Date: 26-Nov-12 9:30 AM 

### Site: FGD Purge Eff Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sample #:

Analyst DF INORGANIC IONS BY IC Bromide 700 mg/L EPA 300.0 12/05/2012 09:05 JAHERMA 10 100 MERCURY (COLD VAPOR) IN WATER Mercury (Hg) 49.9 ug/L EPA 245.1 11/29/2012 15:22 AGIBBS 2.5 50
TOTAL RECOVERABLE METALS BY ICP Boron (B) 72.4 mg/L EPA 200.7 12/05/2012 12:59 DJSULL1 0.5 10
DISSOLVED METALS BY ICP-MS Selenium (Se) 1830 ug/L EPA 200.8 12/03/2012 15:03 KRICHAR 10 10
TOTAL RECOVERABLE METALS BY ICP-MS Arsenic (As) 401 ug/L EPA 200.8 12/04/2012 11:05 KRICHAR 10 10 Chromium (Cr) 249 ug/L EPA 200.8 12/04/2012 11:05 KRICHAR 10 10 Copper (Cu) 319 ug/L EPA 200.8 12/04/2012 11:05 KRICHAR 10 10 Nickel (Ni) 270 ug/L EPA 200.8 12/04/2012 11:05 KRICHAR 10 10 Selenium (Se) 7510 ug/L EPA 200.8 12/04/2012 11:05 KRICHAR 10 10 Silver (Ag) < 10 ug/L EPA 200.8 12/04/2012 11:05 KRICHAR 10 10 Zinc (Zn) 445 ug/L EPA 200.8 12/04/2012 11:05 KRICHAR 10 10 SELENIUM SPECIATION - (Analysis Performed by Applied Speciation and Consulting, LLC) Vendor Parameter Complete Vendor Method V_AS&C
TOTAL DISSOLVED SOLIDS TDS 8600 mg/L SM2540C 11/28/2012 15:53 SWILLI3 200 1



### 2012025278 Collection Date: 26-Nov-12 9:41 AM 

### Site: EQ Tank Eff Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sample #:

Analyst DF MERCURY (COLD VAPOR) IN WATER Mercury (Hg) 19.3 ug/L EPA 245.1 11/29/2012 15:25 AGIBBS 2.5 50
TOTAL RECOVERABLE METALS BY ICP Boron (B) 48.0 mg/L EPA 200.7 12/05/2012 13:03 DJSULL1 0.5 10
DISSOLVED METALS BY ICP-MS Selenium (Se) 951 ug/L EPA 200.8 12/03/2012 15:06 KRICHAR 10 10

# Certificate of Laboratory Analysis This report shall not be reproduced, except in full.



### Order # J12110390

2012025278 Collection Date: 26-Nov-12 9:41 AM 

### Site: EQ Tank Eff Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sample #:

Analyst DF TOTAL RECOVERABLE METALS BY ICP-MS Arsenic (As) 123 ug/L EPA 200.8 12/04/2012 11:22 KRICHAR 10 10 Chromium (Cr) 98.2 ug/L EPA 200.8 12/04/2012 11:22 KRICHAR 10 10 Copper (Cu) 134 ug/L EPA 200.8 12/04/2012 11:22 KRICHAR 10 10 Nickel (Ni) 130 ug/L EPA 200.8 12/04/2012 11:22 KRICHAR 10 10 Selenium (Se) 3140 ug/L EPA 200.8 12/04/2012 11:22 KRICHAR 10 10 Silver (Ag) < 10 ug/L EPA 200.8 12/04/2012 11:22 KRICHAR 10 10 Zinc (Zn) 193 ug/L EPA 200.8 12/04/2012 11:22 KRICHAR 10 10



### 2012025279 Collection Date: 26-Nov-12 9:56 AM 

### Site: BioReactor 1 Inf Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sample #:

Analyst DF MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC) Vendor Parameter Complete Vendor Method V_BRAND
TOTAL RECOVERABLE METALS BY ICP Boron (B) 28.9 mg/L EPA 200.7 12/05/2012 13:07 DJSULL1 0.5 10
DISSOLVED METALS BY ICP-MS Selenium (Se) 354 ug/L EPA 200.8 12/03/2012 15:10 KRICHAR 10 10
TOTAL RECOVERABLE METALS BY ICP-MS Arsenic (As) < 10 ug/L EPA 200.8 12/04/2012 11:12 KRICHAR 10 10 Chromium (Cr) < 10 ug/L EPA 200.8 12/04/2012 11:12 KRICHAR 10 10 Copper (Cu) 17.8 ug/L EPA 200.8 12/04/2012 11:12 KRICHAR 10 10 Nickel (Ni) 13.2 ug/L EPA 200.8 12/04/2012 11:12 KRICHAR 10 10 Selenium (Se) 368 ug/L EPA 200.8 12/04/2012 11:12 KRICHAR 10 10 Silver (Ag) < 10 ug/L EPA 200.8 12/04/2012 11:12 KRICHAR 10 10 Zinc (Zn) < 10 ug/L EPA 200.8 12/04/2012 11:12 KRICHAR 10 10 SELENIUM SPECIATION - (Analysis Performed by Applied Speciation and Consulting, LLC) Vendor Parameter Complete Vendor Method V_AS&C



### 2012025280 Collection Date: 26-Nov-12 10:14 AM 

### Site: BioReactor 2 Inf Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sample #:

Analyst DF MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC) Vendor Parameter Complete Vendor Method V_BRAND

# Certificate of Laboratory Analysis This report shall not be reproduced, except in full.



### Order # J12110390

2012025280 Collection Date: 26-Nov-12 10:14 AM 

### Site: BioReactor 2 Inf Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sample #:

Analyst DF TOTAL RECOVERABLE METALS BY ICP Boron (B) 23.3 mg/L EPA 200.7 12/05/2012 13:11 DJSULL1 0.5 10
TOTAL RECOVERABLE METALS BY ICP-MS Arsenic (As) < 10 ug/L EPA 200.8 12/04/2012 11:15 KRICHAR 10 10 Chromium (Cr) < 10 ug/L EPA 200.8 12/04/2012 11:15 KRICHAR 10 10 Copper (Cu) < 10 ug/L EPA 200.8 12/04/2012 11:15 KRICHAR 10 10 Nickel (Ni) < 10 ug/L EPA 200.8 12/04/2012 11:15 KRICHAR 10 10 Selenium (Se) 47.5 ug/L EPA 200.8 12/04/2012 11:15 KRICHAR 10 10 Silver (Ag) < 10 ug/L EPA 200.8 12/04/2012 11:15 KRICHAR 10 10 Zinc (Zn) < 10 ug/L EPA 200.8 12/04/2012 11:15 KRICHAR 10 10



### 2012025281 Collection Date: 26-Nov-12 10:31 AM 

### Site: BioReactor 2 Eff Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sample #:

Analyst DF INORGANIC IONS BY IC Bromide 890 mg/L EPA 300.0 12/05/2012 09:24 JAHERMA 10 100 MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC) Vendor Parameter Complete Vendor Method V_BRAND MERCURY (COLD VAPOR) IN WATER Mercury (Hg) < 1 ug/L EPA 245.1 11/29/2012 15:27 AGIBBS 1 20
TOTAL RECOVERABLE METALS BY ICP Boron (B) 23.1 mg/L EPA 200.7 12/05/2012 13:14 DJSULL1 0.5 10
TOTAL RECOVERABLE METALS BY ICP-MS Arsenic (As) < 5 ug/L EPA 200.8 12/04/2012 11:46 KRICHAR 5 5 Chromium (Cr) < 5 ug/L EPA 200.8 12/04/2012 11:46 KRICHAR 5 5 Copper (Cu) < 5 ug/L EPA 200.8 12/04/2012 11:46 KRICHAR 5 5 Nickel (Ni) < 5 ug/L EPA 200.8 12/04/2012 11:46 KRICHAR 5 5 Selenium (Se) 17.2 ug/L EPA 200.8 12/04/2012 11:46 KRICHAR 5 5 Silver (Ag) < 5 ug/L EPA 200.8 12/04/2012 11:46 KRICHAR 5 5 Zinc (Zn) < 5 ug/L EPA 200.8 12/04/2012 11:46 KRICHAR 5 5 SELENIUM SPECIATION - (Analysis Performed by Applied Speciation and Consulting, LLC) Vendor Parameter Complete Vendor Method V_AS&C

# Certificate of Laboratory Analysis This report shall not be reproduced, except in full.



### Order # J12110390

2012025282 Collection Date: 26-Nov-12 10:50 AM 

### Site: Filter Blk Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sample #:

Analyst DF DISSOLVED METALS BY ICP-MS Selenium (Se) 1.26 ug/L EPA 200.8 12/03/2012 14:41 KRICHAR 1 1



### 2012025284 Collection Date: 26-Nov-12 9:56 AM 

### Site: BioReactor 1 Inf Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sample #:

Analyst DF MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC) Vendor Parameter Complete Vendor Method V_BRAND



### 2012025285 Collection Date: 26-Nov-12 9:56 AM 

### Site: Hg Blk BioReactor 1 Inf Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sample #:

Analyst DF MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC) Vendor Parameter Complete Vendor Method V_BRAND



### 2012025286 Collection Date: 26-Nov-12 10:14 AM 

### Site: BioReactor 2 Inf Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sample #:

Analyst DF MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC) Vendor Parameter Complete Vendor Method V_BRAND



### 2012025287 Collection Date: 26-Nov-12 10:14 AM 

### Site: Hg Blk BioReactor 2 Inf Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sample #:

Analyst DF MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC) Vendor Parameter Complete Vendor Method V_BRAND

# Certificate of Laboratory Analysis This report shall not be reproduced, except in full.



### Order # J12110390

2012025288 Collection Date: 26-Nov-12 10:31 AM 

### Site: BioReactor 2 Eff Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sample #:

Analyst DF MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC) Vendor Parameter Complete Vendor Method V_BRAND



### 2012025289 Collection Date: 26-Nov-12 10:31 AM 

### Site: Hg Blk BioReactor 2 Eff Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sample #:

Analyst DF MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC) Vendor Parameter Complete Vendor Method V_BRAND3958 6 th Ave NW • Seattle, WA 98107 • T: 206

- 632

- 6206 • F: 206

- 632

- 6017 • www.brooksrand.com • brl@brooksrand.com 

### December 14, 2012 Duke Energy ATTN : Jay Perkins Scientific Support

- Laboratory 13339 Hagers Ferry Road Huntersville NC 28078 jcperkins@duke

- energy.com labcustomer@duke

- energy.com

RE: Project DUK

- HV1 201 Client P roject : J12110390 Dear Mr. Perkins, O n November 30 , 2012 , Brooks Rand Labs ( BRL ) received three (3 ) wastewater samples and three (3 ) corresponding field blanks . The sample s were logged

- in for total mercury ( Hg ) analysis. All samples were received, prepared, analyzed, and stored according to BRL SOPs and EPA methodology. The results were blank

- corrected as described in the calculations section of the relevant SOP and may have been evaluated using reporting limits that have been adjusted to account for sample aliquot size. Please refer to the Sample Results page for sample

- specific MDLs, MRLs, and other details. The total Hg result of all three field blanks were detectable at 0.20, 0.16, and 0.35 ng/L and were qualified B . These concentrations were less than the method defined control limit of 0.50 ng/L however; and all the associated field sample results were greater than 10x the concentration of the blank. Contamination was considered insignificant. All data was reported without additional qualification, aside from concentration qualifiers, and a ll associa ted q uality control sample results met the acceptance criteria. Aside from concentration qualifiers, all data was reported without further qualification and all associated quality control sample results met the acceptance criteria. BRL, an accredited laboratory, certifies the reported results of all analyses for which BRL is NELAP accredited meet all NELAP requirements. For more details , see the Report Information page of the report. Please feel free to contact us if you have any questions regarding this report. Sincerely,

Tiffany Stilwater Mi Sun Um Project Manager Data Manager tiffany@brooksrand.co m misun@brooksrand.comProject ID: DUK-HV1201

PM: Tiffany Stilwater Client PM: Jay Perkins Client PO: 141391



### Definition of Data Qualifiers

(Effective 9/23/09) 

### Laboratory Accreditation BRL is accredited by the National Environmental Laboratory Accreditation Program (NELAP) through the State of Florida Department of Health, Bureau of Laboratories (E87982) and is certified to perform many environmental analyses. BRL is also certified by many other states to perform environmental analyses. For a current list of our accreditations /certifications, please visit our website at <http://www.brooksrand.com/default.asp?contentID=586>. Results reported relate only to the samples listed in the report. 

# Report Information

BLK BRL BS CAL CCV
CRM D DUP ICV MSD ND NR PS REC RPD RSD SCV SOP method blank Brooks Rand Labs laboratory fortified blank calibration standard continuing calibration verification
certified reference material dissolved fraction duplicate initial calibration verification matrix spike duplicate non-detect non-reportable post preparation spike percent recovery relative percent difference relative standard deviation secondary calibration verification standard operating procedure MDL MRL MS

method detection limit method reporting limit matrix spike

SRM T COC

standard reference material total recoverable fraction chain of custody record 

### Common Abbreviations

These qualifiers are based on those previously utilized by Brooks Rand Labs, those found in the EPA SOW ILM03.0, Exhibit B, Section III, pg. B-18, and the USEPA Contract Laboratory Program National Functional Guidelines for Inorganic Superfund Data Review; USEPA; January 2010. These supersede all previous qualifiers ever employed by BRL. Detected by the instrument, the result is > the MDL but ≤ the MRL. Result is reported and considered an estimate. B E An estimated value due to the presence of interferences. A full explanation is presented in the narrative. H Holding time and/or preservation requirements not met. Result is estimated. J Estimated value. A full explanation is presented in the narrative. J-M Duplicate precision (RPD) for associated QC sample was not within acceptance criteria. Result is estimated. J-N Spike recovery for associated QC sample was not within acceptance criteria. Result is estimated. M Duplicate precision (RPD) was not within acceptance criteria. Result is estimated. N Spike recovery was not within acceptance criteria. Result is estimated. R Rejected, unusable value. A full explanation is presented in the narrative. U Result is ≤ the MDL or client requested reporting limit (CRRL). Result reported as the MDL or CRRL. X Result is not BLK-corrected and is within 10x the absolute value of the highest detectable BLK in the batch. Result is estimated. 

### Field Quality Control Samples Please be notified that certain EPA methods require the collection of field quality control samples of an appropriate type and frequency; failure to do so is considered a deviation from some methods and for compliance purposes should only be done with the approval of regulatory authorities. Please see the specific EPA methods for details regarding required field quality control samples.

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.comProject ID: DUK-HV1201

PM: Tiffany Stilwater Client PM: Jay Perkins Client PO: 141391



# Sample Information

Report Matrix Type Received Sampled Sample Lab ID 1248033-01 BioReactor 1 Inf 11/26/2012 11/30/2012 Sample Influent 1248033-02 Hg Blk BioReactor 1 Inf 11/26/2012 11/30/2012 Field Blank DIW 1248033-03 BioReactor 2 Inf 11/26/2012 11/30/2012 Sample Influent 1248033-04 Hg Blk BioReactor 2 Inf 11/26/2012 11/30/2012 Field Blank DIW 1248033-05 BioReactor 2 Eff 11/26/2012 11/30/2012 Sample Effluent 1248033-06 Hg Blk BioReactor 2 Eff 11/26/2012 11/30/2012 Field Blank DIW 

# Batch Summary

Analyte Prepared Analyzed Sequence Batch Lab Matrix Method B122262 12/05/2012 12/10/2012 1200914 Water EPA 1631 Hg

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.comProject ID: DUK-HV1201

PM: Tiffany Stilwater Client PM: Jay Perkins Client PO: 141391



# Sample Results

Sample Sequence Result MDL MRL Unit Batch

Qualifier Analyte Basis

Report Matrix BioReactor 1 Inf 136 T 1200914 B122262 Influent 10.1 3.79 1248033-01 Hg ng/L

BioReactor 2 Eff 35.0 T 1200914 B122262 Effluent 0.39 0.15 1248033-05 Hg ng/L

BioReactor 2 Inf 92.9 T 1200914 B122262 Influent 1.02 0.38 1248033-03 Hg ng/L

Hg Blk BioReactor 1 Inf 0.20 T B 1200914 B122262 DIW 0.41 0.15 1248033-02 Hg ng/L

Hg Blk BioReactor 2 Eff 0.16 T B 1200914 B122262 DIW 0.39 0.15 1248033-06 Hg ng/L

Hg Blk BioReactor 2 Inf 0.35 T B 1200914 B122262 DIW 0.39 0.15 1248033-04 Hg ng/L

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.comProject ID: DUK-HV1201

PM: Tiffany Stilwater Client PM: Jay Perkins Client PO: 141391



# Accuracy & Precision Summary

Batch: B122262

Analyte Result Units Native Spike REC & Limits RPD & Limits Method: EPA 1631 Lab Matrix: Water

Sample Certified Reference Material (1249026, NIST 1641d 1000x dilution) B122262-SRM1 15.68 102% ng/L 85-115 Hg 15.95

Matrix Spike (1248033-01) B122262-MS2 505.1 102% ng/L 71-125 Hg 650.0

135.7

Matrix Spike Duplicate (1248033-01) B122262-MSD2 505.1 104% ng/L 71-125 242% Hg 659.9

135.7



# Method Blanks & Reporting Limits

Batch: B122262
Method: EPA 1631 Matrix: Water Analyte: Hg Result Units Sample B122262-BLK1 ng/L 0.12 B122262-BLK2 ng/L 0.14 B122262-BLK3 ng/L 0.15 B122262-BLK4 ng/L 0.17 MDL: 0.15 Average: 0.15 Standard Deviation: 0.02 Limit: 0.10 Limit: 0.50 MRL: 0.40

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.comProject ID: DUK-HV1201

PM: Tiffany Stilwater Client PM: Jay Perkins Client PO: 141391



# Instrument Calibration

Sequence: 1200914 Method: EPA 1631 Instrument: THG-05 Date: 12/10/2012 Total Mercury and Mercury Speciation by CVAFS

Analyte: Hg REC & Limits Result True Value Lab ID Units 1200914-IBL1 1.82 pg of Hg 1200914-IBL2 3.61 pg of Hg 1200914-IBL3 3.80 pg of Hg 1200914-IBL4 3.96 pg of Hg 1200914-CAL1 10.00 10.15 pg of Hg 101% 1200914-CAL2 25.00 25.89 pg of Hg 104% 1200914-CAL3 100.0 97.30 pg of Hg 97% 1200914-CAL4 500.0 490.6 pg of Hg 98% 1200914-CAL5 2500 2540 pg of Hg 102% 1200914-CAL6 10000 9830 pg of Hg 98% 1200914-ICV1 1568 1595 pg of Hg 102% 85-115 1200914-CCB1 9.35 pg of Hg 1200914-CCV1 500.0 510.7 pg of Hg 102% 77-123 1200914-CCB2 6.14 pg of Hg 1200914-CCB3 6.21 pg of Hg 1200914-CCB4 5.47 pg of Hg 1200914-CCV2 500.0 515.7 pg of Hg 103% 77-123 1200914-CCB5 8.05 pg of Hg 1200914-CCV3 500.0 522.1 pg of Hg 104% 77-123 1200914-CCB6 7.24 pg of Hg 1200914-CCV4 500.0 527.7 pg of Hg 106% 77-123 1200914-CCB7 7.46 pg of Hg 1200914-CCV5 500.0 519.9 pg of Hg 104% 77-123 1200914-CCB8 6.08 pg of Hg 1200914-CCV6 500.0 512.8 pg of Hg 103% 77-123 1200914-CCB9 6.65 pg of Hg 1200914-CCV7 500.0 513.0 pg of Hg 103% 77-123 1200914-CCBA 5.48 pg of Hg 1200914-CCV8 500.0 513.4 pg of Hg 103% 77-123 1200914-CCBB 5.91 pg of Hg 1200914-CCV9 500.0 508.0 pg of Hg 102% 77-123 1200914-CCBC 5.61 pg of Hg 1200914-CCVA 500.0 511.3 pg of Hg 102% 77-123 1200914-CCBD 5.10 pg of Hg 1200914-CCVB 500.0 504.0 pg of Hg 101% 77-123 1200914-CCBE 4.77 pg of Hg 1200914-CCVC 500.0 502.3 pg of Hg 100% 77-123 1200914-CCBF 5.93 pg of Hg 1200914-CCVD 500.0 501.9 pg of Hg 100% 77-123 1200914-CCBG 5.30 pg of Hg

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.comProject ID: DUK-HV1201

PM: Tiffany Stilwater Client PM: Jay Perkins Client PO: 141391



# Instrument Calibration

Sequence: 1200914 Method: EPA 1631 Instrument: THG-05 Date: 12/10/2012 Total Mercury and Mercury Speciation by CVAFS

Analyte: Hg REC & Limits Result True Value Lab ID Units 1200914-CCVE 500.0 502.4 pg of Hg 100% 77-123 1200914-CCBH 7.85 pg of Hg 1200914-CCVF 500.0 509.5 pg of Hg 102% 77-123 1200914-CCBI 6.96 pg of Hg 1200914-CCVG 500.0 500.3 pg of Hg 100% 77-123 1200914-CCBJ 9.76 pg of Hg 1200914-CCVH 500.0 502.7 pg of Hg 101% 77-123 1200914-CCBK 5.67 pg of Hg 1200914-CCVI 500.0 510.6 pg of Hg 102% 77-123 1200914-CCBL 5.36 pg of Hg 1200914-CCVJ 500.0 519.9 pg of Hg 104% 77-123 1200914-CCBM 6.16 pg of Hg 1200914-CCVK 500.0 514.7 pg of Hg 103% 77-123 1200914-CCBN 5.66 pg of Hg 1200914-ICV2 1568 1685 pg of Hg 107% 85-115 1200914-CCVL 500.0 522.4 pg of Hg 104% 77-123 1200914-CCBO 4.57 pg of Hg 1200914-CCVM 500.0 525.9 pg of Hg 105% 77-123 1200914-CCBP 3.60 pg of Hg 1200914-CCVN 500.0 523.0 pg of Hg 105% 77-123 1200914-CCBQ 5.87 pg of Hg

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.comProject ID: DUK-HV1201

PM: Tiffany Stilwater Client PM: Jay Perkins Client PO: 141391



# Sample Containers

Lab ID: 1248033-01 Report Matrix: Influent Sample Type: Sample Received: 11/30/2012 Sample: BioReactor 1 Inf Collected: 11/26/2012 Ship. Cont. pH P-Lot Preservation Lot Size Container Des A 71666330 10 Cooler

none n/a 500 mL Bottle FLPE Hg-T

Lab ID: 1248033-02 Report Matrix: DIW Sample Type: Field Blank Received: 11/30/2012 Sample: Hg Blk BioReactor 1 Inf Collected: 11/26/2012 Ship. Cont. pH P-Lot Preservation Lot Size Container Des A 71666330 10 Cooler

none n/a 500 mL Bottle FLPE Hg-T

Lab ID: 1248033-03 Report Matrix: Influent Sample Type: Sample Received: 11/30/2012 Sample: BioReactor 2 Inf Collected: 11/26/2012 Ship. Cont. pH P-Lot Preservation Lot Size Container Des A 71666330 10 Cooler

none n/a 500 mL Bottle FLPE Hg-T

Lab ID: 1248033-04 Report Matrix: DIW Sample Type: Field Blank Received: 11/30/2012 Sample: Hg Blk BioReactor 2 Inf Collected: 11/26/2012 Ship. Cont. pH P-Lot Preservation Lot Size Container Des A 71666330 10 Cooler

none n/a 500 mL Bottle FLPE Hg-T

Lab ID: 1248033-05 Report Matrix: Effluent Sample Type: Sample Received: 11/30/2012 Sample: BioReactor 2 Eff Collected: 11/26/2012 Ship. Cont. pH P-Lot Preservation Lot Size Container Des A 71666330 10 Cooler

none n/a 500 mL Bottle FLPE Hg-T

Lab ID: 1248033-06 Report Matrix: DIW Sample Type: Field Blank Received: 11/30/2012 Sample: Hg Blk BioReactor 2 Eff Collected: 11/26/2012 Ship. Cont. pH P-Lot Preservation Lot Size Container Des A 71666330 10 Cooler

none n/a 500 mL Bottle FLPE Hg-T

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.comProject ID: DUK-HV1201

PM: Tiffany Stilwater Client PM: Jay Perkins Client PO: 141391



# Shipping Containers

Cooler
Tracking No: 535305196306 via FedEx Temperature: -0.1 °C Coolant Type: Ice Description: Cooler Damaged in transit? No Returned to client? No Custody seals present? No Custody seals intact? No COC present? Yes Received: November 30, 2012 9:00

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

## December 1 3 , 201 2

Jay Perkins Duke Energy Analytical Laboratory Mail Code MGO3A2 (Building 7405) 13339 Hagers Ferry Rd. Huntersville, NC 28078 (704) 875-5 245 Project: Bi -Monthly Allen Shay/MillerCreek (LIMS #J12110390)

Dear Mr. Perkins, Attached is the report associated with three (3 ) aqueous samples submitted for selenium speciation on November 29, 2012. The samples were received in a sealed cooler at -0. 5 ° C on November 30 , 2012

. Selenium speciation analysis was performed via ion chromatography inductively coupled plasma collision reaction cell mass spectrometry (IC-ICP- CRC -MS). Any issues associated with the analysis are addressed in the following report. If you have any questions, please feel free to contact me at your convenience.
Sincerely,

Russell Gerads Vice President Applied Speciation and Consulting, LLC

## Applied Speciation and Consulting, LLC Report prepared for: Jay Perkins Duke Energy Analytical Laboratory Mail Code MGO3A2 (Building 7405) 13339 Hagers Ferry Rd. Huntersville, NC 28078 Project: Bi -Monthly Allen Shay/MillerCreek (LIMS #J12110390)

December 1 3 , 2012

1. Sample Reception Three (3 ) aqueous samples in 125mL HDPE bottles (provided by Applied Speciation and Consulting) were su bmitted for selenium speciation analysis on November 29 , 2012. The samples were received on November 30 , 201 2 in a sealed container at -0. 5°C. The samples were received in a laminar flow clean hood, void of trace metals contamination and ultra-violet radiation, and were designated discrete sample identifiers. An aliquot of each sample was filtered (0.45μm) and each filtrate w as stored in a secure, monitored cryofreezer (maintained at a temperature of -80 o 

## C) until selenium speciation analysis could be performed via ion chromatography inductively coupled plasma collision reaction cell mass spectrometry (IC-ICP- CRC -MS).

2. Sample Preparation All sample preparation is performed in laminar flow clean hoods known to be free from trace metals contamination. All applied water for dilutions and sample preservatives are monitored for contamination to account for any biases associated with the sample results. Selenium Speciation Analysis by IC -ICP-CRC- MS Prior to analysis, an aliquot of each sample was filtered with a syringe filter (0.45μm) and injected directly into an autosampler vial. No further sample preparation was performed as any chemical alteration of a sample may shift the equilibrium of the system, resulting in changes in speciation ratios.

3. Sample Analysis All sample analysis is preceded by a minimum of a five-point calibration curve spanning the entire concentration range of interest. Calibration curves are performed at the beginning of

## each analytical day. All calibration curves, associated with each species of interest, are

standardized by linear regression resulting in a response factor. All sample results are instrument blank corrected to account for any operational biases associated with the analytical platform. Prior to sample analysis, all calibration curves are verified using second source standard s which are identified as initial calibration verification standards (ICV). Ongoing instrument performance is identified by the analysis of continuing calibration verification standards (CCV) and continuing calibration blanks (CCB) at a minimum interval of every ten analytical runs. Selenium Speciation Analysis by IC -ICP-CRC- MS Each sample for selenium speciation analysis w as analyzed by ion chromatography inductively coupled plasma collision reaction cell mass spectrometry (IC-ICP- CRC -MS) on December 1, 2012. An aliquot of each sample is injected onto an anion exchange column and mobilized by a basic (pH > 7) gradient. The eluting selenium species are then introduced into a radio frequency (RF) plasma where energy-transfer processes cause desolvation, atomization, and ionization. The ions are extracted from the plasma through a differentially-pumped vacuum interface and travel through a pressurized chamber ( CRC ) containing a reaction gas which preferentially reacts with interfering ions of the same target mass to charge ratios (m/z). A solid-state detector detects ions transmitted through the mass analyzer and the resulting current is processed by a data handling system. Retention times for each eluting species are compared to known standards for species identification.

4. Analytical Issues The overall analyses went well and no significant analytical issues were encountered. All quality control parameters associated with the samples were within acceptance limits. The estimated method detection limits (eMDLs) for selenite, selenate, and selenocyanate are generated from replicate analyses of the lowest standard in the calibration curve. Not all selenium species are present in preparation blanks; therefore, eMDL calculations based on preparation blanks are artificially biased low. The eMDL for methylseleninic acid and selenomethionine is calculated from the average eMDL of selenite, selenate, and selenocyanate. The calibration does not contain methylseleninic acid or selenomethionine due to impurities in these standards which would bias the results for other selenium species.

## If you have any questions or concerns regarding this report, please feel free to contact me.

Sincerely,

Russell Gerads Vice President Applied Speciation and Consulting, LLC

### Sample Results
Sample ID Se(IV) Se(VI) SeCN MeSe(IV) SeMe Unknown Se Species (n) FGD Purge Eff 1720 98.1 104 8.4 5.7 84.3 (2) BioReactor 1 Inf 172 96.4 2.24 6.76 2.77 8.85 (2) BioReactor 2 Eff 0.40 ND (<0.63) ND (<0.51) ND (<0.45) ND (<0.45) 0.0 (0) All results reflect the applied dilution and are reported in μg/L ND = Not detected at the applied dilution SeCN = Selenocyanate MeSe(IV) = Methylseleninic acid SeMe = Selenomethionine Unknown Se Species = Total concentration of all unknown Se species observed by IC-ICP-MS Report Generated by: Russell Gerads Applied Speciation and Consulting, LLC Selenium Speciation Results for Duke Energy Project Name: Bi-Monthly Allen Shay/MillerCreek

Contact: Jay Perkins

Date: December 13, 2012 LIMS #J12110390

### Quality Control Summary - Preparation Blank Summary

Analyte (μg/L) PBW1 PBW2 PBW3 PBW4 Mean StdDev eMDL* eMDL 250x eMDL 1000x Se(IV) 0.000 0.000 0.000 0.000 0.000 0.000 0.001 0.22 0.86 Se(VI) 0.000 0.000 0.000 0.000 0.000 0.000 0.003 0.63 2.5 SeCN 0.000 0.000 0.000 0.000 0.000 0.000 0.002 0.51 2.0 MeSe(IV) 0.000 0.000 0.000 0.000 0.000 0.000 0.002 0.45 1.8 SeMe 0.000 0.000 0.000 0.000 0.000 0.000 0.002 0.45 1.8 eMDL = Estimated Method Detection Limit *Please see narrative regarding eMDL calculations
Quality Control Summary - Certified Reference Materials Analyte (μg/L) CRM True Value Result Recovery Se(IV) LCS 9.57 9.57 100.0 Se(VI) LCS 9.48 9.23 97.3 SeCN LCS 8.92 8.78 98.4 MeSe(IV) LCS 6.47 6.15 95.1 SeMe LCS 9.32 8.78 94.2 Applied Speciation and Consulting, LLC Selenium Speciation Results for Duke Energy Project Name: Bi-Monthly Allen Shay/MillerCreek

Contact: Jay Perkins LIMS #J12110390 Date: December 13, 2012 Report Generated by: Russell Gerads

### Quality Control Summary - Matrix Duplicates Analyte (μg/L) Sample ID Rep 1 Rep 2 Mean RPD Se(IV) Batch QC 0.33 0.26 0.30 21.8 Se(VI) Batch QC ND (<0.63) ND (<0.63) NC NC SeCN Batch QC ND (<0.51) ND (<0.51) NC NC MeSe(IV) Batch QC ND (<0.45) ND (<0.45) NC NC SeMe Batch QC ND (<0.45) ND (<0.45) NC NC ND = Not detected at the applied dilution NC = Value was not calculated due to one or more concentrations below the eMDL

Quality Control Summary - Matrix Spike/ Matrix Spike Duplicate Analyte (μg/L) Sample ID Spike Conc MS Result Recovery Spike Conc MSD Result Recovery RPD Se(IV) Batch QC 1390 1501 108.0 1390 1499 107.8 0.2 Se(VI) Batch QC 1261 1337 106.0 1261 1326 105.1 0.8 SeCN Batch QC 1144 1192 104.3 1144 1191 104.1 0.1 Applied Speciation and Consulting, LLC Selenium Speciation Results for Duke Energy Project Name: Bi-Monthly Allen Shay/MillerCreek

Contact: Jay Perkins LIMS #J12110390 Date: December 13, 2012 Report Generated by: Russell Gerads