science for a changing world 

Borehole Dilatometer Installation, Operation, and 

Maintenance at sites in Hawaii 

By G.D.Myren, M.J.S. Johnston and R.J. Mueller 

Open File Report 2006-1 103 

U.S. Department of the Interior 

U.S. Geological Survey 
U.S. Department of the Interior 

Dirk Kempthorne, Secretary of Interior 

U.S. Geological Survey 

P. Patrick Leahy, Acting Director 

U.S. Geological Survey, Reston, Virginia 2006 

Revised and reprinted: 2006 

For product and ordering information: 

World Wide Web: http://www.usgs.gov/pubprod 

Telephone: 1-888-ASK-USGS 

For more information on the USGS-the Federal source for science about the Earth, 

its natural and living resources, natural hazards, and the environment: 

World Wide Web: http://www.usgs.gov 

Telephone: 1-888-ASK-USGS 

Any use of trade, product, or firm names is for descriptive purposes only and does not imply 

endorsement by the U.S. Government. 

Although this report is in the public domain, permission must be secured from the individual 

copyright owners to reproduce any copyrighted material contained within this report. 
Contents 

1. Introduction 

2. Siting 

3. Drilling 

4. Dilatometer Instrumentation 

5. Dilatometer Installation 

6. Additional Work and Installation 

7. Electronics Enclosures 

8. Surface Electronics 

9. Basic Principle of Operation (SOC Box) 

10. Valve opening algorithm 

11. Cable & Connector Summary 

12. Barometer 

13. Telemetry 

14. Dilatometer Maintenance 

15. Latitude &Longitude of Sites 

16. Satellite Telemetry Configuration 

Strip Rd 

Hokukano 

Mauna Loa 

Keller Well 

ZENOSOFT 

Strip Rd 

Hokukano 

Mauna Loa 

Keller Well 

17. Installation Field Notes 

18. Drawings 
1 Introduction: 

In response to concerns about the potential hazard of Mauna Loa volcano in Hawaii, the USGS began efforts in 

1998 to add four high-resolution borehole sites. Located at these sites are; strainmeters, tiltmeters, seismometers, 

accelerometers and other instrumentation. These instruments are capable of providing continuous monitoring of 

the magma movement under Mauna Loa. Each site was planned to provide multi-parameter monitoring of 

volcanic activity. 

In June of 2000, a contract was let for the core drilling of three of these four sites. They are located at Hokukano 

(west side of Mauna Loa) above Captain Cook, HI; at Mauna Loa Observatory (1 1,737' near the summit), and at 

Mauna Loa Strip Road (east side of Mauna Loa). Another site was chosen near Halema'uma u' and Kilauea's 

summit, in the Keller deep well. (See maps). The locations of these instruments are shown in Figure 1 with their 

latitude and longitude in Table 1. 

The purpose of this network is to monitor crustal deformation associated with volcanic intrusions and 

earthquakes on Mauna Loa and Kilauea volcanoes. This report describes the methods used to locate sites, install 

dilatometers, other instrumentation, and telemetry. We also provide a detailed description of the electronics used 

for signal amplification and telemetry, plus techniques used for instrument maintenance. Instrument sites were 

selected in regions of hard volcanic rock where the expected signals from magmatic activity were calculated to be 

a maximum and the probability of earthquakes with magnitude 4 or greater is large. At each location, an attempt 

was made to separate tectonic and volcanic signals from known noise sources for each instrument type. 

2 Siting: 

Using seismicity, geologic and topographic maps together with geophysical knowledge and geologists 

recommendations, a list of preliminary sites were selected. Available access, and telemetry issues were checked 

out in detail during field visits. When the final site choice was made, permits were obtained from landowners and 

a drilling contract was drawn up to begin exploratory drilling. 

3 Drilling: 

The primary drilling method used, involved core drilling with a PQ (4.80"/12.192 cm old.) drill to deal with the 

typical volcano geology in Hawaii. This geology consists of layered basaltic flows separated by ash deposits. Each 

hole was core drilled to about 350 to 400 feet, allowing retrieval of 3.265" 0.d. core. This continuous core 

provided scientist with an unprecedented look into the eruption history of Mauna Loa over the past 10,000 years. 

When 3 m of competent unfractured rock was identified in the recovered core below 350 feet, that location was 

recorded as a possible installation site. After the best possibilities for installation sites were selected, the hole was 

cased with drill steel. Cementing of the casing was not successful due to the voids between flows. All holes were 

located well above the water table so all holes had to be drilled with air. Some holes were logged with a televiewer 

borehole inspection camera to help in selection of the best installation sites. Overall, this drilling method was the 

least expensive for drilling in volcano geology. An inclinometer (measured the verticality of the holes) and a 

cement bailer were run in each hole to check for clearance, and depth. When these parameters were all 

determined, the hole was deemed ready for installation. 

4 DILATOMETER INSTRUMENTATION : 

The Sacks-Evertson dilational strainmeter used in this experiment (Sacks met alfR., 1971) are installed at 
depths between 367' and 1192' below the surface on MaunaLoa andKilauea. The sensors, 

installed as part of a cooperative program between the U.S. Geological Survey and the Carnegie Institution of 

Washington, are cemented in the borehole with expansive grout having density characteristics approximating 

those of the host material. The borehole is then filled to the surface with cement to avoid long-term strains from 

hole relaxation effects. (Re-equilibration of the aquifer system was not an issue at these dry volcanic boreholes.) 

The sensor consists of a 3 m long stainless steel oil filled reservoir that is filled with 100 cs silicon oil. Small 

compressions on the side of this reservoir force oil into a small bellows. Displacement of the end of the bellows is 

monitored by an LVDT (linear voltage displacement transducer) which produces an output that is proportional to 

the imposed dilational strain. There are two LVDT's in the strainmeters installed in Hawaii. The first transducer 

measures the rock strain, and the second transducer acts as reservoir volume monitor 1 lo gain transducer I 

thermistor. The mechanical gain of the first LVDT is about 67,000. The frequency response is flat from about 20 

Hz to less than 0.000000001 Hz. The 20 Hz high frequency cut-off is caused by the hydraulic filter effect as the 

oil flows from the large reservoir into the bellows chamber through a small orifice. 

5 DILATOMETER INSTALLATION 

Introduction 

Installation was accomplished with the use of a truck mounted hydraulic winch and derrick. Before installation a 

28 foot (8.534 meters) long grout dump was lowered to the bottom of the hole to check that the instrument would 

not get hung up during installation. Electronics for signal conditioning, amplification, data collection and 

transmission are temperature tested in the lab and installed in closed bottom concrete block surface enclosures. 

The electronics are powered by solar charged batteries. They are located in similar concrete block surface 

enclosures within 15 feet (4.57 meters) of the electronics. 

At these sites additional effort has been made to record the seismic portion of the signal generated by the 

dilatometer. This has been done using 24 bit recorders with Spread Spectrum radio telemetry to HVO. They are 

installed with solar charged batteries in a separate enclosure. 

Installation Procedure 

After a instrumentation site has been selected, the core hole has been drilled and competent rock has been 

located below the casing, installation can begin. 

A day before installation, the instrument is checked for correct operation and correct resistances between pin 

outs. Voltage is applied and readings of the signal out and signal change after the valve is closed and opened are 

taken and recorded. (see Installation Notes in Figures, Tables and Schematics). The dilatometer is manufactured 

with additional ballast weight in the center section to help sink the instrument in the installation grout. 

Prior to installation, a small hydraulic crane and winch is setup, its wire cable reeled out, measured and color 

coded every 50 feet(15m.) for 1200 feet(365 m.). The grout dump, which is transported in three sections, is 

assembled. Its bottom opening trip mechanism is put together, greased, threaded to the bottom of the grout dump, 

and checked for operation.. The hydraulic derrick and winch are moved over the hole and the 27 foot (8.23 m.) 

grout dump is lowered to the bottom. When this clears, the bottom trip opens, and the hole depth agrees with the 

depth determined by the drilling, the drill rig is allowed to leave the site. 

The instrument cable (mounted on a cable reel stand in the back of a pick-up) is unreeled, measured and marked 

with colored tape every 50 feet (15 m.). Twenty feet from the determined instrument depth a warning mark is 

attached. At the bottom depth mark for the instrument a bright colored tape is attached over a 2 foot (600 mm.) 
section of the cable with the beginning of the tape nearest the instrument. The cable is reeled back in and 

positioned next to the hole. The instrument is removed from its crate, a wire rope is attached for lining, and the 

instrument is secured vertically next to the hydraulic derrick, it is tested again for proper operation and this 

information is recorded (see Installation Notes). 

The grout dump is raised over the hole with bottom trip attached, and run in the hole twice to recheck the depth. 

If a site in the hole has been found above the bottom, the bottom is raised to reach the site. This is done by mixing 

in a mortar mixer the right amount of grout to reach that depth and then lowering it to the bottom of the hole in the 

grout dump. Approximately 6 hours is allowed between each dump for proper hardening of the grout. (see photo 

18). The grout dump is then run down the hole empty to check the hardness and the depth. (See attached copy on 

'SET-GROUT" in ATTACHMENTS). The cement dump has a capacity of 0.925 cu.ft. which fills 

approximately 10.59 linear feet of 3.76" hole. Each bag of non-shrink grout (Corps of Engineers Spec. for nonshrink grout CRD-C 621), 50-lb., is mixed with 9.8 lb. water (1.225 gal.) to get about .46 cu.ft. It takes 2.01 bags 

to fill the cement dump for one trip in the hole to cover the instrument. Three bags of grout and 3.675 gallons of 

water are usually mixed in a mortar mixer for each instrument emplacement. This leaves approximately 5.5 feet of 

grout above the instrument if the instrument settles to the bottom. 

Now that the instrument is ready, the expansive non-shrink grout can be mixed for instrument installation. 

Mixing takes at least 15 minutes. The grout should have a slump of 12, or zero cone, rather like a thick malt. The 

cable test, and recheck of the instrument are recorded with the date and time. (see Installation Field Notes, 

attached in Figures, Tables and Schematics, under Resistance Check). A cup of water is poured in the grout dump 

(to act as a cushion when the grout is poured in), then the grout is poured in using 5 gal. plastic pails and a funnel. 

Because the grout begins to harden in 1 hour, time should be noted. Using the hydraulic derrick, the grout dump is 

lowered to 15 feet (4.572 meters) of the bottom, the tension of the cable is checked, and the dump is allowed to 

free fall. This trips the bottom device, the dump is slowly raised and the tension is checked for the weight of the 

grout. As the dump is raised the strain on the hydraulic motor is noticed to help determine if the grout has been 

released. This slow raise allows time for the grout to flow out evenly (with no turbulence which can potentially 

cause uneven mixing). When the dump is about 30 feet (9.15 m.) offthe bottom it is raised quickly to the surface. 

The dump is set aside and the instrument is positioned over the hole using the derrick and winch. As the 

instrument is lowered to the bottom of the hole, the instrument cable is taped to the weight bearing wire rope. The 

instrument cable depth marks are called off as they go in the hole. When the target depth, as marked on the cable, 

approaches the top of the casing, it's descent is slowed by the contact with the grout. The instrument will begin to 

enter the grout and sink to within 1-2 feet of the bottom. 

At this time instrument and cable resistances are read, and power is applied to read its strain response as the 

cement sets. These values are recorded along with the date and time. (see installation notes in appendix). 

If, before one hour has elapsed, any resistances are bad or the instrumentis 

working improperly, than it should be pulled out of the grout slowly. Once the instrument is on the surface, 

the top of the cement becomes the hole bottom. If there is another place above the original site, than another 

attempt can be made to install, if not the hole is made available for other instrumentation. 

If the instrument passes operation checks, the installation can proceed.(the strainmeter should have approx 5.0 

ft of grout over the top of it). The instrument cable is tied off using 112 inch rope with 3 or 4 half hitches tied to the 

surface casing. One to two days are allowed before the tension is relieved and the wire rope and cable are cable 

clamped to the casing. 

6 Additional Instrumentation 
In order to optimize the science from these boreholes, additional cement is placed over the instrument and other 

instruments are installed. At Strip Rd, Hokukano, and Mauna Loa Observatory sites, borehole 

seismometer/acceleration packages and tiltmeter instruments were installed. 

At these sites the hole bottom was brought up to a point 100' above the top of the strainmeter by placing neat 

cement in the borehole. One inch flush joint tremmie pipe is lowered into the borehole in 20' sections by using the 

hydraulic derrick. Once this tremmie gently contacts the top of the strainmeter cement and the depth and volume 

to a seismic package site is determined, a new bottom can be set. The tremmie is pulled back 20' and a neat 

cement mix is prepared and delivered by gravity through the tremmie to the new target depth. After the tremmie is 

pulled from the hole and washed, this mix is allowed to set overnight. The seismic/acceleration package is placed 

in the hole the next day using the same tremmie pipe It is cemented in place, and the hole bottom is brought up 

again (using the same method of volume calculation & tremmie and cement placement) to 50 feet from the 

surface. The tiltmeter can now be installed. It is attached to 1" flush joint PVC tremmie pipe by a pvc press fit 

joint with wire rope bearing the load. 

7 ELECTRONICS ENCLOSURES 

After the instruments are all installed and the hole is cemented to the surface, electronic surface 

enclosures are installed. These enclosures consist of a 3 ft. (0.9144 m.) wide, 4 ft. ( 1.2192 m.) long, by 1 Vi ft 

(0.4572 m.) high closed bottom concrete block structures with a steel lid for the electronics, and the same for the 

batteries. The battery enclosure is a short distance from the electronics with solar panels for battery charging near 

by. All enclosures are connected by pvc conduit for power and signal routing. 

8 SURFACE ELECTRONICS 

The electronics consists of signal amplifiers, a barometric pressure transducer, and two data collection systems 

housed in the electronics enclosure. Electronics for signal conditioning, amplification, data collection and 

transmission are temperature tested in the lab and installed in closed bottom concrete block surface enclosures. 

The electronics are powered by solar charged batteries. They are located in a closed bottom concrete block surface 

enclosures within 15 feet (4.57 meters) of the electronics. A water resistant box houses the electronics for the 

strainmeter. 

The strainmeter electronics consists of a dcldc converter powering 2 op amps for 2 different DC (strain) signals, 

and a automatic valve opener driven by microprocessor control. The operation of the strainmeter electronics is a 

follows. As the strainmeter in the borehole is squeezed by the surrounding rock, silicon oil in the instrument is 

forced through an orifice, displaces a bellows, which moves the attached core of a transducer. The movement of 

this core is approximately .3 18Vl.01 in.(.3 18V/.254mm), as powered by a 6.8V voltage regulator. The movement 

of this transducer is measured as a voltage at the surface in the SOC Box (Strainmeter Operation & Control Box). 

This voltage is monitored by a micro processor to control pressure relief of the transducer in the strainmeter. As 

stated in the Dilatometer Instrumentation section, there are two LVDT's, one measures the strain on the rock, the 

second acts as a reservoir monitor/thermistor/lo gain transducer. The operation of the SOC Box is designed so that 

if LVDT #1 exceeds a predetermined threshold voltage of 0.4 volts it's valve will openlclose and pass this 

pressure to LVDT #2. After 2 hours LVDT #2's valve will open and pass this pressure in the form of fluid to the 

reservoir volume space of the strainmeter. (see appendix). If during the daily cycle of the instrument operation, the 

battery powering the strainmeter should drop below 10 volts, the electronics will automatically shut down. It will 

open both valves at this time preventing pressure from exceeding the physical limits of the LVDT's in the 

instrument. 
This electronics package draws approximately 380 milliamps. The electronics is powered by a 12 volt deepcycle maintenance free gelled electrolyte trickle charged battery. This battery is kept charged 

by two 50 watt solar panels using a automatic sequencing charger. This charger stops charging at 14.3 volts +I-.2 

volts and resumes at 13.2 +I-0.3 volts. During the night a blocking diode acts to prevent discharge of the battery 

through the panel. 

High frequency data from the dilatometer in the 0.005 Hz to 100 Hz can be recorded on 24 bit telemetry 

systems with a least count noise of less than10 -1 1. Low frequency data, from 0 Hz to 0.002 Hz are transmitted 

via a 17-bit digital telemetry system through the GOES satellite system to Men10 Park, Cal. A separate polled 

spread spectrum telemetry to USGS Hawaii Volcano Observatory provides 1 minute data. The least count noise 

on the high gain satellite telemetry system for the dilatometer is about 2 * 10 -1 1. For the low gain channel the 

least count noise is about 1.2 * 10 -8. These instruments all record earth strain tides, strain transients related to 

volcanic deformation and numerous strain seismograms from local and tele-seismic earthquakes with magnitudes 

between 1 and 9. These strain seismograms are used to calculate the dynamic earthquake moments. 

Static moments and total earthquake moments are determined from the co-seismic strains and total strain 

changes observed with larger events. Should pre-seismic strains occur before an expected volcanic eruption, they 

can be resolved at about the 10 to -1 1 level if they occur quickly, and about 10 to -8 level if they occur days to 

weeks before the event. 

9. Basic Principle of Operation: 

Summary: 

SOC Box Operation (as described by Carnegie Institute of Washington 1 DTM) is as follows: 

The strainmeter control box contains the electronics, which control and monitor the strainmeter. It also filters and 

passes the analog signals from the strainmeter to an external device. The external device is usually an Analog-toDigital converter passing digital data to either local storage or a telemetry system or both. 

Detailed Description: 

Supply power to the strainmeter control box is monitored internally by the electronics to determine if sufficient 

voltage is present at the input to the strainmeter control box. If the voltage dips below the preset trip voltage ( 

should be about lOVdc ), the controller will disconnect power until the voltage rises to an acceptable level. There 

is some hysteresis designed into the power monitoring circuitry to avoid the unwanted condition of power-off, 

power-on, power-off, etc ... This on-off-on-off-on is something that could be quite common with a solar-cell 

charged battery system if the hysteresis was not implemented. To avoid any problems that could arise from 

sustained power cycling, hysteresis is used in the power monitoring circuit. Additionally supply power is 

monitored by the microcontroller by way of analog input to the 16-bit multiplexing ND. If the voltage measured 

by the ND drops to an unacceptable level the microcontroller will immediately open the strainmeter valves in an 

effort to protect the strainmeter. If this precaution was not taken when power is sagging, the strainmeter control 

box might shut down with valves closed during a seismic event. This could easily rupture the bellows in the 

strainmeter, leaving it unusable. 

The strainmeter control box supplies power to the Differential Transformers contained in the strainmeter. The 

control box will supply a regulated 6.8VDC potential to the DT power input 

The Valves in the strainmeters (designed and manufactured at the Carnegie Institution of Washington) open or 

close depending on the polarity of the potential voltage applied to the valves. Most of the older land-based 

strainmeters operate with a valve operating potential of 24VDC. The newer land- and water-based strainmeters 
have valves that operate at 48VDC'. The serial connection is optically isolated using the RS-232 standard. It uses 

the following parameters. 

BAUD: 9600, DATA BITS: 8, STOP BITS: 1, PARITY: None 

As of the writing of this manual the Dept. of Terrestrial Magnetism has adopted a policy to use only 48VDC 

valves on all future water- and land-based strainmeters. 

10. VALVE OPENING ALGORITHM 

June 7.2000 

Low Threshold Voltage: 

0.4Volts 

Low Threshold Period: 

Valve 1 = 900 seconds (15 minutes) 

Valve 2 = 660 seconds (1 1 minutes) 

High Threshold Voltage: 

3.0Volts (60% of AID'S maximum voltage level) 

Valve behavior if DTl voltage exceeds low threshold for 15 consecutive minutes: 

Valve 1 opens 

Valve 2 is scheduled to open 2 hours later. 

Valve 1 closes 15 seconds after opening. 

Valve 2 closes 15 seconds after opening. 

Valve2 behavior if DT2 voltage exceeds low threshold for 11 consecutive minutes: 

Valve 2 opens. 

Valve 2 closes 15 seconds after opening. 

Valve1 behavior if DTl voltage exceeds high threshold: 

Valve 1 opens. 

Valve 1 closes 1 minute afer opening 

Valve 2 is scheduled to open 11 minutes after Valve 1 closes 

Valve 2 remains open for 5 seconds. 

If Valve 1 opens again in less than 11 minutes, Valve2 opening is 

rescheduled for 1 1 minutes following next Valve1 closing. 

Valve2 behavior if DT2 voltage exceeds high threshold: 

Valve 2 opens. 

Valve 2 closes 5 SEC after opening. 

Interlock: 

If Valve 1 needs to open we check if Valve2 is open. If Valv e2 is open, we close Valve 2 before opening Valve 1. 

If Valve 2 needs to open we check if Valve 1 is open. If Valve 1 is open, we close Valve 1 before opening Valv e2 
Here is a chart describing the new firmware algorithm. 

You should note that the High Level threshold does not change, it is static 

at 3.0VDC. The interlock disables the operation that opens valve 2, 2 

hours after valve 1 has been opened over the low threshold. This is intended to 

keep DT2 close to zero so that the spring associated with DT2 does not 

deform such that its zero point would move slightly due to long term extension. 

Again, this interlock disable is only to be used to troubleshoot 

very specific problems. Most probably you won't have any need of this 

feature. So, make sure you do not select the interlock disable feature. 

The 12VDC power occasionally displaying 12VDC is most probably a Windows 

'feature'. Don't worry about it too much this is Windows interrupting its 

own RS232 stream. However, if you see that the SOC box is opening the 

valves as this 1 VDC power display occurs, you should be concerned. If this 

is happening you need to let DTMICIW* know because that would be very serious, 

Valve Close after 

low threshold 

opening 

15 seconds since 

valve closed? 

Average next 128 

samples. Represents 

zero offset level. 

Difference = 

LVT -ABS(Average) 

Difference 

LT. 

10% of LVT? 

LVT = 

LVT + 0.1 *LVT 

ABS(Difference) 

GT. 

(LVT+O. 1 *LVT) ? 

LVT = LVT +LVTO 

ABS(Zero offset) 

LT. 

LVT? 

LVT!=LVTO? 

LVT-LVT-LVTO 

LVT+LVTO 

LT. 

HVT? 

LVTO-Default Low VoltageThreshold = 0.4V 

LVT=Low Voltage Threshold 

HVT-High VoltageThreshold=3V 

Algorithm for Low Voltage Threshold Adjustment 

Applies to DT1 and DT2 
DT1 and DT2 can have different Low Voltage Thresholds 

NO 

NO 

Return leaving LVT 

unchanged 

NO 

Return 

NO 

LVT = LVTO? 

Return 

NO 

NO 

*DTM/CIW = Department of Terrrestrial Magnetism / Carnegie Institute of Technology 

5241 Broadbranch Road, N.W. 

Washington, DC 200 15 

Call 202-478-8843, 8829, or 8835 
11. Cable & Connector Summary (as described by Carnegie Institute of Washington 1 DTM) 

Power 

The Five (5) pin power connector provides input power to the 'Strainmeter Control Box'. The input power is a 

nominal +12Vdc (Strainmeter Operation and Control Box (SOC Box)) 

( 1 l-24Vdc allowable range). 

Strainmeter 

The eighteen (1 8) pin connector interfaces the control box to the strainmeter. This connector provides power to the 

Differential Transformers, DT's, and valves in the strainmeter. It also feeds the DT outputs from the strainmeter 

into the control box. 

Ground Post Post 

This post is used to make a connection with the common ground. In an effort to reduce ground loops, which can 

induce noise into the system, this is by design the sole connection to common ground. 

RS232 

This connector provides serial communication to a PC for the purpose of monitoring the status of the strainmeter 

and manual control of the valves. 

Output A 

Output B 

These two connectors are analog outputs from the strainmeter control box. These are the buffered DT signals from 

the strainmeter. OUTPUT A and B have identical pinouts. 

* Note: 

There is a hole in one of the faceplates. This hole feeds to a water tight tube used to feed outside air pressure to the 

pressure sensor inside the box 

** Note: Refer (to connector pin-out map for a description of connector pin assignments. 
12. Barometer: 

There is a barometric pressure transducer operating over a 300 millibar range. This on site transducer aids in 

the reduction of the strain data as it is effected by barometric pressure. (see barometric pressure transducer in 

Additional Electronics Section.) Setra Model #270 

The barometric pressure transducer is powered by 1 12 volt deep-cycle maintenance free gelled electrolyte trickle 

charged batteries. This is kept charged by a similar automatic sequencing charger hooked to a 20 watt solar panel. 

13 Telemetry: 

Coastal Environmental Systems ZENO Model 3200 was selected as a Data Collection Platform. This system 

draws 84ma at 12 volts DC during collection and 3 amps at transmission of data to the GOES satellite. Data is 

collected once every 10 minutes to 17 bit accuracy and transmitted at 10 minute intervals. A second ZENO is also 

installed for transmission of 1 minute data via Spread Spectrum radio to the Hawaii Volcano Observatory. It also 

uses a auto sequencing charger and a 50 watt solar panel to keep the battery charged. 

14. PROCEDURE FOR DILATOMETER MAINTENANCE: 

In general the maintenance of a dilatometer installation is fairly straightforward. There may be the specific 

instance when a visit may be made for unexpected problems, but for the most part it is a routine procedure. In 

summary, the data from each instrument is looked at daily for proper operation. It is inspected for tidal response 

(data quality), data dropouts (satellite problems, computer problems, missing transmisions), time of transmission, 

transmission power levels, and battery voltages. Information obtained from this helps in the proper field 

maintenance. 

bk05 = Setra #270 600-900mbar at 0 - 5vdc +I-5.12 volt range 3.91~10-5 

bk06 = tiltmeter x axis Pinnacle 5000 +I-5.12 volt range 3.91~10-5 

at 14meter depth gain 3 in tiltmeter = 85.26mvlmicroradian 

or 0.000458 1574 rnicroradianslct 

bk07 =tiltmeter y axis Pinnacle 5000 +I-5.12 volt range 3.91~10-5 

bk08 = YSI4401 thermister C x ,001 (ie, 15666 x ,001 = +15.666) 

bk09 = +I- 12 bit range = ,007816291 voltslct ,007816291 by 1720 = 13.44 volts dc 

bklO =tipping bucket raingage, 1 count = lmillimeter 

Number of assigned DCP bits 01 - 08 are +/-I7 bit; 09 is +/-I2 bit, 10 = rain gage 

Mama Loa Strainmeter 

lat 19 32.227 long 155 35.347 

Data Logger = Coastal Environmental Systems ZENO 3200 sn #? 

01 - 08 are +/-I7 bit; 09 +/-I2 bit 

mlO 1 = transducer # 1 hi gain +I-,512 volt range 3.91~10-6 

m102 = " lo " +I-5.12 volt range 3.91~10-5 

m103 = #2 10 " +I-5.12 volt range 3.91~10-5 

m104 = inputto ND as a short to ground +I-5.12 volt range 3.91~10-5 

m105 = Setra #270 600-900mbar at 0-5vdc +I-5.12 volt range 3.91~10-5 

m106 = tiltmeter x axis Pinnacle 5000 +I-5.12 volt range 3.9 1x10-5 

at 14meter depth gain 3 in tiltmeter = 85.26mvlmicroradian 

or 0.000458 1574 rnicroradianslct 

m107 = tiltmeter y axis Pinnacle 5000 +I-5.12 volt range 3.91~10-5 

at 14meter depth gain 3 in tiltmeter = 119.38mv/microradian 

or 0.00032721 143 rnicroradianslct 

ml08 = YSI4401 thermister C x ,001 (ie, 15666 x ,001 = +15.666) 

m109 = +I- 12 bit range = ,007816291 voltslct ,007816291 by 1720 = 13.44 volts dc 

ml10 ==tipping bucket raingage, 1 count = lmillimeter 

Number of assigned DCP bits 01 - 08 are +/-I7 bit; 09 is +/-I2 bit, 10 = rain gage 

Keller Strainmeter 

lat 19 23.557long 155 17.154 

Data Logger = Coastal Environmental Systems ZENO 3200 sn #? 

01 - 08 are +/-I7 bit; 09 is +/-I2 bit, 10 & 11 are +I-17bit 

kwO 1 = transducer # 1 hi gain +I-,512 volt range 3.91~10-6 

kw02 = lo +I-5.12 volt range 3.91~10-5 

kw03 = #2 10 " +I-.5 12 volt range 3.91~10-6 

kw04 = inputto ND as a short to ground +I-5.12 volt range 3.9 1x10-5 

kw05 = Setra #270 800-1 lmbar at 0-5vdc +I-5.12 volt range 3.91~10-5 

kw06 = tiltmeter x axis Pinnacle 5000 +I-5.12 volt range 3.91~10-5 

at 14meter depth gain 3 in tiltmeter = 85.26mvlmicroradian 

or 0.000458 1574 rnicroradianslct 

kw07 = tiltmeter y axisPinnacle 5000 +I-5.12 volt range 3.91~10-5 

kw08 = YSI4401 thermister C x ,001 (ie, 15666 x ,001 = +15.666) 

kw09 = +I- 12 bit range = ,007816291 voltslct 

,007816291 by 1720 = 13.44 volts dc 

kwlO =tipping bucket raingage, 1 count = lmillimeter 

kwl 1 = grounded input +I-5.12 volt range 3.91~10-5 
Number of assigned DCP bits 01 - 08 are +/-I7 bit; 09 is +/-I2 bit, 

10 = rain gage, 1 1 is +/- 17 bit spare 

Programming ZENO 

Accessing the Coastal Environmental ZENO 3200 for programming is done through a manufacturer supplied 

rs232 cable and customer supplied PC or MAC with a terminal emulation program. (Crosstalk, Mirror, Windows 

Terminal, Hyperterminal, or ProComm). 

Access the ZENO via the rs232 cable with the Hyperterminal program on. Then power the ZENO. 

The program will display the following: 

Strip Road Program: 

"apply power to ZENO 

Watchdog Reset 

ZENO-3200 using ZENOSOFT V1.813 Dec 1 1998 14:56:07 CS EEBE 

(C)opyright 1995-1998, Coastal Environmental Systems, Seattle, WA, USA 

System Time = 05/12/02 23:47:05 

Initializing Zeno 3200 ... / 

Verifying GOES Transmitter Initialization ... 

WARNING: GOES Transmitter not initialized since system restart. 

Use the 'Initialize GOES' command inside the GOES menu. 

Zeno 3200 is Data Sampling. Type 'U<enter> to access the User Interface 

('V <enter> entering this command will not appear on screen. However the response follows) 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

COMMTJNTCATIONS MRNU 

(Cnlm) Change Item n To Value m (Tn) Terminal Mode On COM Port n 

(M) Modem Menu (E) Save Parameters To EEPROM 

(P) Power Control Menu (U) User Menu 

(G) GOES Menu (Q) Quit 

(D) Digital Control Menu (H) Help 

Item 1: 9600 

Item 2: 9600 

Item 3: 9600 

Item 4: RS232 

Item 5: GOES 

Item 6: RS232 

Item 7: NO 

Item 8: NO 

(COM1 Baud Rate) 

(COM2 Baud Rate) 

(COM3 Baud Rate) 

(COM1 Port Type) 

(COM2 Port Type) 

(COM3 Port Type) 

(COM3 User Interface Exclusive) 

(Enable Exclusive CCSAIL Access) 
GOES MRNU 

(Cdm) Change Item n To Value m (E) Save Parameters To EEPROM 

(D) Run GOES Diagnostics (U) User Menu 

(R) Reset GOES Errors (Q) Quit 

(I) Initialize GOES (H) Help 

Item 1: 2637e3e0 

Item 2: 104 

Item 3: 00:00:10:00 

Item 4: 00:06:00 

Item 5: 1 

Item 6: SHORT 

Item 7: 151 

Item 8: 00:OO:OO 

Item 9: 00:05:00 

(Data Collection Platform Address) 

(Self-Timed Transmit Channel Number) 

(Self-Timed Transmission Interval) 

(Self-Timed Transmission Offset) 

(Transmission Window Length) 

(Satellite Link Parameter: Preamble) 

(Random Transmit Channel Number) 

(Random Transmission Interval) 

(Random Disable Time) 

Current Date and Time: 05/12/01 22:51:46 

Enter new Date and Time: 05/12/02 22:53:00 

GOES Transmitter Initialization ... successful 

Note: Next sample interval begins in 6 minutes and 59 seconds 

GOES MRNU 

(Cdm) Change Item n To Value m (E) Save Parameters To EEPROM 

(D) Run GOES Diagnostics (U) User Menu 

(R) Reset GOES Errors (Q) Quit 

(I) Initialize GOES (H) Help 

Item 1 : 2637e3e0 (Data Collection Platform Address) 

Item 2: 104 (Self-Timed Transmit Channel Number) 

Item 3: 00:OO: 10:OO (Self-Timed Transmission Interval) 

Item 4: 00:06:00 (Self-Timed Transmission Offset) 

Item 5: 1 (Transmission Window Length) 

Item 6: SHORT (Satellite Link Parameter: Preamble) 

Item 7: 151 (Random Transmit Channel Number) 

Item 8: 00:00:00 (Random Transmission Interval) 

Item 9: 00:05:00 (Random Disable Time) 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

SAMPLE PERIOD MRNU 

(Cdm) Change Item n To Value m (Q) Quit 

(E) Save Parameters To EEPROM (H) Help 

(U) User Menu 

Item 1 : 600 (Sample Interval Time) 

Item 2: 14 (Sample Duration Time) 

Item 3: 0 (Sample Time Offset) 
> u 

Checking Scan List records 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

Enter Administrator Password:   

Waiting for all data acquisition tasks to finish . . . 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu 

(P) Process Menu 

(D) Data Output Menu 

(T) Sensor Timing Loop Menu 

(0) Output Message Timing Menu 

(I,) System Load Menu 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 1 of 9: 

Item 1 : Sensor Tvne Code 

(W) Password Menu 

(R) Reset System 

(E) Save Parameters To EEPROM 

(U) User Menu 

(Q) Quit 

0 Help 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

2 (1 8-bit Single-Ended A to D) 

dt0l 

1 - 

, 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation . 

Item 6: Switched Power Code 0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannup Time 0 

2 (1.0 seconds) 

0 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 
Sensor Items for Record 2 of 9: 

Item 1 : Sensor Type Code 2 (1 8-bit Single-Ended A to D) 

Item 2: Sensor Name dtOl 

Item 3: Sensor Input Channel 1 - 

Item 4: Analog Channel Gain 1 

Item 5: Analog Channel Attenuation 10 

Item 6: Switched Power Code 0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannuu Time 0 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 3 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: Switched Power Code 

8 

0 

2 (1.0 seconds) 

0 

0.5 

n 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

2 (1 8-bit Single-Ended A to D) 

dt02 

I+ 

1 

10 

0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannuv Time 0 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 4 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: Switched Power Code 

8 

0 

2 (1.0 seconds) 

0 

0.5 

0 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

2 (1 8-bit Single-Ended A to D) 

short 

2- 

1 

1 

0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 
Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannup Time 0 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 5 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: switched Power Code 

Item 7: Sensor Excitation Voltage Code 

Item 8: Switched Excitation Return 

Item 9: Switched Power Wannup Time 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 6 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: switched Power Code 

Item 7: Sensor Excitation Voltage Code 

Item 8: Switched Excitation Return 

Item 9: Switched Power Wannup Time 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

2 (1.0 seconds) 

0 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

2 (1 8-bit Single-Ended A to D) 

bar0 

2+ 

1 

10 

0 (NO SWITCHED POWER) 

0 (NO EXCITATION VOLTAGE) 

0 

0 

8 

0 

2 (1.0 seconds) 

0 

0.5 

0 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

2 (1 8-bit Single-Ended A to D) 

x-axis 

3- 

1 

10 

0 (NO SWITCHED POWER) 

0 (NO EXCITATION VOLTAGE) 

0 

0 

8 

0 

2 (1.0 seconds) 

0 

0.5 

0 
N 

SENSOR MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Sensor Items for Record 7 of 9: 

Item 1 : Sensor Type Code 2 (1 8-bit Single-Ended A to D) 

Item 2: Sensor Name y-axis 

Item 3: Sensor Input Channel 3+ 

Item 4: Analog Channel Gain 1 

Item 5: Analog Channel Attenuation 10 

Item 6: Switched Power Code 0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannup Time 0 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 8 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

2 (1.0 seconds) 

0 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

Item 6: Switched Power Code 

Item 7: Sensor Excitation Voltage Code 

Item 8: Switched Excitation Return 

Item 9: Switched Power Wannup Time 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

2 (1 8-bit Single-Ended A to D) 

temp 

4- 

1 

10 

0 (NO SWITCHED POWER) 

2 (EXC = 2.50 VDC) 

A 

0 

8 

0 

2 (1.0 seconds) 

0 

0.5 

0 

SENSOR MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 
(Jn) Jump To Record n (H) Help 

Sensor Items for Record 9 of 9: 

Item 1 : Sensor Type Code 1 (1 2-bit Analog to Digital) 

Item 2: Sensor Name Battery 

Item 3: Sensor Input Channel BATTERY VOLTAGE 

Item 6: Switched Power Code 0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannuu Time 0 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu 

(P) Process Menu 

(D) Data Output Menu 

(T) Sensor Timing Loop Menu 

(0) Output Message Timing Menu 

(I,) System Load Menu 

PROCESS MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

(W) Password Menu 

(R) Reset System 

(E) Save Parameters To EEPROM 

(U) User Menu 

(Q) Quit 

0 Help 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

Process Items for Record 1 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data Sl : dtOl 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 2 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S2 : dtOl 

PROCESS MRNU 
(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 3 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S3 : dt02 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 4 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S4 : short 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 5 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S5 : bare 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 6 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S6 : x-axis 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 
(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 7 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S7 : y-axis 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 8 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S8 : temp 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 9 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S9 : Battery 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu 

(P) Process Menu 

(D) Data Output Menu 

(T) Sensor Timing Loop Menu 

(0) Output Message Timing Menu 

(I,) System Load Menu 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 1 of 9: 

Item 1: Field Type code 

(W) Password Menu 

(R) Reset System 

(E) Save Parameters To EEPROM 

(U) User Menu 

(Q) Quit 

0 Help 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 
Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 2 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 3 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 4 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

dt0l 

P2.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

short 

P4.1 

0 

3 
DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 5 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 6 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 7 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

bar0 

P5.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

x-axis 

P6.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

y-axis 

P7.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 
Data Items for Record 8 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 9 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu 

(P) Process Menu 

(D) Data Output Menu 

(T) Sensor Timing Loop Menu 

(0) Output Message Timing Menu 

(I,) System Load Menu 

SENSOR TTMTNG LOOP MRNU 

(Cdm) Change Item n To Value m 

(Z) Zeno Program Menu 

12 : GOES Binary Format Field 

1 

temp 

P8.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

BATTERY 

P9.1 

0 

3 

(W) Password Menu 

(R) Reset System 

(E) Save Parameters To EEPROM 

(U) User Menu 

(Q) Quit 

0 Help 

Item 1 : 0.5 (Timing Loop #I Period) 

Item 2: 1.0 (Timing Loop #2 Period) 

Item 3: 120.0 (Timing Loop #3 Period) 

Item 4: 10.0 (Timing Loop #4 Period) 

OUTPUT MRSSAGE TTMTNG MRNU 

(Cdm) Change Item n To Value m (H) Help 

(Z) Zeno Program Menu 

Item 1 : COM2 (Data Output Message #I COM Port) 

Item 2: COM3 (Data Output Message #2 COM Port) 

Item 3: COM3 (Data Output Message #3 COM Port) 

Item 4: COM3 (Data Output Message #4 COM Port) 

Item 5: 0.0 (Data Output Message #I Period) 
Item 6: 0.0 (Data Output Message #2 Period) 

Item 7: 0.0 (Data Output Message #3 Period) 

Item 8: 0.0 (Data Output Message #4 Period) 

> e 

Verifying parameters can be stored in EEPROM 

Saving parameters to EEPROM . . . 

Saving Scan List parameters to EEPROM . . . 

475 out of 2048 bytes used in EEPROM. 

Total EEPROM Writes: 43 Checksum: 238 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu (W) Password Menu 

(P) Process Menu (R) Reset System 

(D) Data Output Menu (E) Save Parameters To EEPROM 

(T) Sensor Timing Loop Menu (U) User Menu 

(0) Output Message Timing Menu (Q) Quit 

(I,) System Load Menu (H) Help 

> b 

Waiting for all data acquisition tasks to finish 

BACK DOOR MRNU 

(Cdm) Change Item n To Value m (X) Display Stack Usage 

(F) Calculate Free Heap Memory (E) Save Parameters To EEPROM 

(A) Auto-Calibrate Compass (U) User Menu 

(I) Initialize Compass (HI Help 

(R) Reset Parameters To Defaults 

Item 1: 16777 

Item 2: 1 

Item 3: 60 

Item 4: 18 

Item 5: COUNTS 

Item 6: YES 

Item 7: 32768.00 

Item 8: 0 

Item 9: 1.0000 

(Processor Clock Speed) 

(RAMROM Wait States) 

(50160 Hz Rejection For 18-bit ADC) 

(1311 8 Bit Operation Of 18-bit ADC) 

(A To D Conversion Results) 

(Expert Menu Mode) 

(Real-time Clock Crystal Frequency At 25 Degrees C) 

(Speed vs. Noise Tradeoffs For 18-bit ADC, Factory Only) 

(12-bit ADC Correction Factor) 

SAMPLE PERIOD MRNU 

(Cdm) Change Item n To Value m (Q) Quit 

(E) Save Parameters To EEPROM (H) Help 

(U) User Menu 

Item 1 : 600 (Sample Interval Time) 

Item 2: 14 (Sample Duration Time) 

Item 3: 0 (Sample Time Offset) 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 
SYSTEM FUNCTIONS MRNU 

(Cdm) Change Item n to Value m (E) Save Parameters To EEPROM 

(S) System Date And Time (U) User Menu 

(T) Calibrate Internal Temperature (Q) Quit 

(V) Program Version (H) Help 

Item 1 : 1223 (Primary UnitExperiment ID) 

Item 2: 2 (Secondary UnitExperiment ID) 

Item 3: 1 (Data Dump Format) 

Item 4: 1 (Real Time Output Format) 

Item 5: 0 (Add Compass To Vane) 

Item 6: 0 (Compass Offset) 

Item 7: 0 (Barometer Elevation) 

> v 

ZENO-3200 using ZENOSOFT V1.813 Dec 1 1998 14:56:07 CS EEBE 

(C)opyright 1995-1998, Coastal Environmental Systems, Seattle, WA, USA 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

>cg1 

Current Date and Time: 05/12/02 23:39:26 

Enter new Date and Time: 05/12/02 23:45:00 

GOES Transmitter Initialization ... successful 

Note: Next sample interval begins in 4 minutes and 59 seconds. 

GOES MRNU 

(Cdm) Change Item n To Value m (E) Save Parameters To EEPROM 

(D) Run GOES Diagnostics (U) User Menu 

(R) Reset GOES Errors 

(I) Initialize GOES 

Item 1: 2637e3e0 

Item 2: 104 

Item 3: 00:00:10:00 

Item 4: 00:05:15 

Item 5: 1 

Item 6: SHORT 

Item 7: 151 

Item 8: 00:OO:OO 

Item 9: 00:05:00 

(Data Collection Platform Address) 

(Self-Timed Transmit Channel Number) 

(Self-Timed Transmission Interval) 

(Self-Timed Transmission Offset) 

(Transmission Window Length) 

(Satellite Link Parameter: Preamble) 

(Random Transmit Channel Number) 

(Random Transmission Interval) 

(Random Disable Time) 

Verifying GOES Transmitter Initialization ... successful 

Exiting user interface. 
Hokukano Program 

"apply power to ZEN0 

Watchdog Reset 

ZENO-3200 using ZENOSOFT V1.813 Dec 1 1998 14:56:07 CS EEBE 

(C)opyright 1995-1998, Coastal Environmental Systems, Seattle, WA, USA. 

System Time = 05/12/02 23:47:05 

Initializing Zeno 3200 ... / 

Verifying GOES Transmitter Initialization ... 

WARNING: GOES Transmitter not initialized since system restart. 

Use the 'Initialize GOES' command inside the GOES menu. 

Zeno 3200 is Data Sampling. Type 'U<enter> to access the User Interface 

('V <enter> entering this command will not appear on screen. However the response follows) 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

COMMTJNTCATIONS MRNU 

(Cnlm) Change Item n To Value m (Tn) Terminal Mode On COM Port n 

(M) Modem Menu (E) Save Parameters To EEPROM 

(P) Power Control Menu (U) User Menu 

(G) GOES Menu (Q) Quit 

(D) Digital Control Menu (H) Help 

Item 1: 9600 

Item 2: 9600 

Item 3: 9600 

Item 4: RS232 

Item 5: GOES 

Item 6: RS232 

Item 7: NO 

Item 8: NO 

(COM1 Baud Rate) 

(COM2 Baud Rate) 

(COM3 Baud Rate) 

(COM1 Port Type) 

(COM2 Port Type) 

(COM3 Port Type) 

(COM3 User Interface Exclusive) 

(Enable Exclusive CCSAIL Access) 
GOES MRNU 

(Cdm) Change Item n To Value m (E) Save Parameters To EEPROM 

(D) Run GOES Diagnostics (U) User Menu 

(R) Reset GOES Errors (Q) Quit 

(I) Initialize GOES (H) Help 

Item 1 : 263880d4 (Data Collection Platform Address) 

Item 2: 104 (Self-Timed Transmit Channel Number) 

Item 3: 00:OO: 10:OO (Self-Timed Transmission Interval) 

Item 4: 00:05: 15 (Self-Timed Transmission Offset) 

Item 5: 1 (Transmission Window Length) 

Item 6: SHORT (Satellite Link Parameter: Preamble) 

Item 7: 151 (Random Transmit Channel Number) 

Item 8: 00:00:00 (Random Transmission Interval) 

Item 9: 00:05:00 (Random Disable Time) 

Current Date and Time: 05/12/01 22:51:46 

Enter new Date and Time: 05/12/02 22:53:00 

GOES Transmitter Initialization ... successful 

Note: Next sample interval begins in 6 minutes and 59 seconds 

GOES MRNU 

(Cdm) Change Item n To Value m (E) Save Parameters To EEPROM 

(D) Run GOES Diagnostics (U) User Menu 

(R) Reset GOES Errors (Q) Quit 

(I) Initialize GOES (H) Help 

Item 1 : 263880d4 (Data Collection Platform Address) 

Item 2: 104 (Self-Timed Transmit Channel Number) 

Item 3: 00:OO: 10:OO (Self-Timed Transmission Interval) 

Item 4: 00:05: 15 (Self-Timed Transmission Offset) 

Item 5: 1 (Transmission Window Length) 

Item 6: SHORT (Satellite Link Parameter: Preamble) 

Item 7: 151 (Random Transmit Channel Number) 

Item 8: 00:00:00 (Random Transmission Interval) 

Item 9: 00:05:00 (Random Disable Time) 

USER MRNU 

(C) Communications Menu ( T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

SAMPLE PERIOD MRNU 

(Cdm) Change Item n To Value m (Q) Quit 

(E) Save Parameters To EEPROM (H) Help 

(U) User Menu 

Item 1 : 600 (Sample Interval Time) 
Item 2: 14 (Sample Duration Time) 

Item 3: 0 (Sample Time Offset) 

> u 

Checking Scan List records 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

> zUSER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

Enter Administrator Password: ** ** 

Waiting for all data acquisition tasks to finish 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu (W) Password Menu 

(P) Process Menu (R) Reset System 

(D) Data Output Menu (E) Save Parameters To EEPROM 

(T) Sensor Timing Loop Menu (U) User Menu 

(0) Output Message Timing Menu (Q) Quit 

(I,) System Load Menu (H) Help 

SENSOR MRNU 

(Cnlm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Sensor Items for Record 1 of 9: 

Item 1 : Sensor Type Code 2 (1 8-bit Single-Ended A to D) 

Item 2: Sensor Name dtOl 

Item 3: Sensor Input Channel 1 - 

Item 4: Analog Channel Gain 1 

Item 5: Analog Channel Attenuation 1 

Item 6: Switched Power Code 0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannup Time 0 

Item 10: Sensor Sample Count 8 

Item 11 : Maximum Sensor Readings 0 

Item 12: Sensor Timing Loop 2 (1.0 seconds) 

Item 13: Conversion Coefficient A 0 

Item 14: Conversion Coefficient B 0.5 

Item 15: Conversion Coefficient C 0 
N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

Sensor Items for Record 2 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analoa - Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: Switched Power Code 

Item 7: Sensor Excitation Voltage Code 

Item 8: Switched Excitation Return 

Item 9: Switched Power Wannup Time 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

2 (1 8-bit Single-Ended A to D) 

dt0l 

1 - 

1 

10 - ~ 

0 (NO SWITCHED POWER) 

0 (NO EXCITATION VOLTAGE) 

0 

0 

8 

0 

2 (1.0 seconds) 

0 

0.5 

0 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Sensor Items for Record 3 of 9: 

Item 1 : Sensor Type Code 2 (1 8-bit Single-Ended A to D) 

Item 2: Sensor Name dt02 

Item 3: Sensor Input Channel I+ 

Item 4: Analog Channel Gain 1 

Item 5: Analog Channel Attenuation 10 

Item 6: Switched Power Code 0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannup Time 0 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

2 (1.0 seconds) 

0 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 
Sensor Items for Record 4 of 9: 

Item 1 : Sensor Type Code 2 (1 8-bit Single-Ended A to D) 

Item 2: Sensor Name short 

Item 3: Sensor Input Channel 2- 

Item 4: Analog Channel Gain 1 

Item 5: Analog Channel Attenuation 1 

Item 6: Switched Power Code 0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannuu Time 0 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 5 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: Switched Power Code 

8 

0 

2 (1.0 seconds) 

0 

0.5 

n 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

2 (1 8-bit Single-Ended A to D) 

bar0 

2+ 

1 

10 

0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannuv Time 0 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 6 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: Switched Power Code 

8 

0 

2 (1.0 seconds) 

0 

0.5 

0 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

2 (1 8-bit Single-Ended A to D) 

x-axis 

3- 

1 

10 

0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 
Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannup Time 0 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 7 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: switched Power Code 

Item 7: Sensor Excitation Voltage Code 

Item 8: Switched Excitation Return 

Item 9: Switched Power Wannup Time 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 8 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: switched Power Code 

Item 7: Sensor Excitation Voltage Code 

Item 8: Switched Excitation Return 

Item 9: Switched Power Wannup Time 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

2 (1.0 seconds) 

0 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

2 (1 8-bit Single-Ended A to D) 

y-axis 

3+ 

1 

10 

0 (NO SWITCHED POWER) 

0 (NO EXCITATION VOLTAGE) 

0 

0 

8 

0 

2 (1.0 seconds) 

0 

0.5 

0 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

2 (1 8-bit Single-Ended A to D) 

temp 

4- 

1 

10 

0 (NO SWITCHED POWER) 

2 (EXC = 2.50 VDC) 

A 

0 

8 

0 

2 (1.0 seconds) 

0 

0.5 

0 
SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 9 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

1 (12-bit Analog to Digital) 

Batterv 

Item 3: Sensor Input Channel BATTERY VOLTAGE 

Item 6: Switched Power Code 0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannup Time 0 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu 

(P) Process Menu 

(D) Data Output Menu 

(T) Sensor Timing Loop Menu 

(0) Output Message Timing Menu 

(I,) System Load Menu 

PROCESS MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Process Items for Record 1 of 9: 

Item 1 : Process Category 

0 

1 (0.5 seconds) 

0 

(W) Password Menu 

(R) Reset System 

(E) Save Parameters To EEPROM 

(U) User Menu 

(Q) Quit 

0 Help 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

1 : General -. 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data Sl : dtOl 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 
Process Items for Record 2 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S2 : dtOl 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 3 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S3 : dt02 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 4 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S4 : short 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 5 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S5 : bare 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 6 of 9: 

Item 1 : Process Category 1 : General 
Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S6 : x-axis 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 7 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S7 : y-axis 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 8 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S8 : temp 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 9 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S9 : Battery 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu (W) Password Menu 

(P) Process Menu (R) Reset System 

(D) Data Output Menu (E) Save Parameters To EEPROM 

(T) Sensor Timing Loop Menu (U) User Menu 

(0) Output Message Timing Menu (Q) Quit 

(I,) System Load Menu (H) Help 

DATA OUTPUT MRNU 
(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 1 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 2 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 3 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

dt0l 

P2.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

dt02 

P3.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

Data Items for Record 4 of 9: 
Item 1: Field Type code 12 : GOES Binary Format Field 

Item 2: Output Message($ 1 

Item 3: Field Name short 

Item 4: Input Record and Element P4.1 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 5 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 6 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 7 of 9: 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

bar0 

P5.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

x-axis 

P6.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

Item 1: Field Type code 12 : GOES Binary Format Field 

Item 2: Output Message($ 1 

Item 3: Field Name y-axis 

Item 4: Input Record and Element P7.1 

Item 5: Field Decimal Places 0 

Item 6: Field Width 3 
DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 8 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 9 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu 

(P) Process Menu 

(D) Data Output Menu 

(T) Sensor Timing Loop Menu 

(0) Output Message Timing Menu 

(I,) System Load Menu 

SENSOR TTMTNG LOOP MRNU 

(Cdm) Change Item n To Value m 

(Z) Zeno Program Menu 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

temp 

P8.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

BATTERS 

P9.1 

0 

3 

(W) Password Menu 

(R) Reset System 

(E) Save Parameters To EEPROM 

(U) User Menu 

(Q) Quit 

0 Help 

Item 1 : 0.5 (Timing Loop #I Period) 

Item 2: 1.0 (Timing Loop #2 Period) 

Item 3: 120.0 (Timing Loop #3 Period) 

Item 4: 10.0 (Timing Loop #4 Period) 

> 0 

OUTPUT MRSSAGE TTMTNG MRNU 
(Cdm) Change Item n To Value m (H) Help 

(Z) Zeno Program Menu 

Item 1: COM2 

Item 2: COM3 

Item 3: COM3 

Item 4: COM3 

Item 5: 0.0 

Item 6: 0.0 

Item 7: 0.0 

Item 8: 0.0 

(Data Output Message #I COM Port) 

(Data Output Message #2 COM Port) 

(Data Output Message #3 COM Port) 

(Data Output Message #4 COM Port) 

(Data Output Message #I Period) 

(Data Output Message #2 Period) 

(Data Output Message #3 Period) 

(Data Output Message #4 Period) 

> e 

Verifying parameters can be stored in EEPROM . . . 

Saving parameters to EEPROM . . . 

Saving Scan List parameters to EEPROM . . . 

475 out of 2048 bytes used in EEPROM. 

Total EEPROM Writes: 43 Checksum: 238 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu (W) Password Menu 

(P) Process Menu (R) Reset System 

(D) Data Output Menu (E) Save Parameters To EEPROM 

(T) Sensor Timing Loop Menu (U) User Menu 

(0) Output Message Timing Menu (Q) Quit 

(I,) System Load Menu (H) Help 

> b 

Waiting for all data acquisition tasks to finish 

BACK DOOR MRNU 

(Cdm) Change Item n To Value m (X) Display Stack Usage 

(F) Calculate Free Heap Memory (E) Save Parameters To EEPROM 

(A) Auto-Calibrate Compass (U) User Menu 

(I) Initialize Compass (H) Help 

(R) Reset Parameters To Defaults 

Item 1: 16777 (Processor Clock Speed) 

Item 2: 1 (RAMROM Wait States) 

Item 3: 60 (50160 Hz Rejection For 18-bit ADC) 

Item 4: 18 (1311 8 Bit Operation Of 18-bit ADC) 

Item 5: COUNTS (A To D Conversion Results) 

Item 6: YES (Expert Menu Mode) 

Item 7: 32768.00 (Real-time Clock Crystal Frequency At 25 Degrees C) 

Item 8: 0 (Speed vs. Noise Tradeoffs For 18-bit ADC, Factory Only) 

Item 9: 1.0000 (12-bit ADC Correction Factor) 

SAMPLE PERIOD MRNU 

(Cdm) Change Item n To Value m (Q) Quit 

(E) Save Parameters To EEPROM (H) Help 

(U) User Menu 

Item 1 : 600 (Sample Interval Time) 

Item 2: 14 (Sample Duration Time) 

Item 3: 0 (Sample Time Offset) 
USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

SYSTEM FUNCTIONS MRNU 

(Cdm) Change Item n to Value m (E) Save Parameters To EEPROM 

(S) System Date And Time (U) User Menu 

(T) Calibrate Internal Temperature (Q) Quit 

(V) Program Version (HI Help 

Item 1 : 1223 (Primary UnitExperiment ID) 

Item 2: 2 (Secondary UnitExperiment ID) 

Item 3: 1 (Data Dump Format) 

Item 4: 1 (Real Time Output Format) 

Item 5: 0 (Add Compass To Vane) 

Item 6: 0 (Compass Offset) 

Item 7: 0 (Barometer Elevation) 

> v 

ZENO-3200 using ZENOSOFT V1.813 Dec 1 1998 14:56:07 CS EEBE 

(C)opyright 1995-1998, Coastal Environmental Systems, Seattle, WA, USA 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

>cg1 

Current Date and Time: 05/12/02 23:39:26 

Enter new Date and Time: 05/12/02 23:45:00 

GOES Transmitter Initialization ... successful 

Note: Next sample interval begins in 4 minutes and 59 seconds 

GOES MRNU 

(Cdm) Change Item n To Value m (E) Save Parameters To EEPROM 

(D) Run GOES Diagnostics (U) User Menu 

(R) Reset GOES Errors (Q) Quit 

(I) Initialize GOES (H) Help 

Item 1 : 263880d4 (Data Collection Platform Address) 

Item 2: 104 (Self-Timed Transmit Channel Number) 

Item 3: 00:OO: 10:OO (Self-Timed Transmission Interval) 

Item 4: 00:05: 15 (Self-Timed Transmission Offset) 

Item 5: 1 (Transmission Window Length) 

Item 6: SHORT (Satellite Link Parameter: Preamble) 

Item 7: 151 (Random Transmit Channel Number) 
Item 8: 00:00:00 (Random Transmission Interval) 

Item 9: 00:05:00 (Random Disable Time) 

>'I 

Verifying GOES Transmitter Initialization ... successful 

Exiting user interface 

Mauna Loa Program 

"apply power to ZEN0 

Watchdog Reset 

ZENO-3200 using ZENOSOFT V1.813 Dec 1 1998 14:56:07 CS EEBE 

(C)opyright 1995-1998, Coastal Environmental Systems, Seattle, WA, USA 

System Time = 05/12/02 23:47:05 

Initializing Zeno 3200 ... / 

Verifying GOES Transmitter Initialization ... 

WARNING: GOES Transmitter not initialized since system restart. 

Use the 'Initialize GOES' command inside the GOES menu. 

Zeno 3200 is Data Sampling. Type 'U<enter> to access the User Interface 

('V <enter> entering this command will not appear on screen. However the response follows) 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

COMMTJNTCATIONS MRNU 
(Cdm) Change Item n To Value m (Tn) Terminal Mode On COM Port n 

(M) Modem Menu (E) Save Parameters To EEPROM 

(P) Power Control Menu (U) User Menu 

(G) GOES Menu (Q) Quit 

(D) Digital Control Menu (H) Help 

Item 1: 9600 

Item 2: 9600 

Item 3: 9600 

Item 4: RS232 

Item 5: GOES 

Item 6: RS232 

Item 7: NO 

Item 8: NO 

(COM1 Baud Rate) 

(COM2 Baud Rate) 

(COM3 Baud Rate) 

(COM1 Port Type) 

(COM2 Port Type) 

(COM3 Port Type) 

(COM3 User Interface Exclusive) 

(Enable Exclusive CCSAIL Access) 

GOES MRNU 

(Cdm) Change Item n To Value m (E) Save Parameters To EEPROM 

(D) Run GOES Diagnostics (U) User Menu 

(R) Reset GOES Errors (Q) Quit 

(I) Initialize GOES (H) Help 

Item 1: 2638202~ 

Item 2: 104 

Item 3: 00:00:10:00 

Item 4: 00:05:40 

Item 5: 1 

Item 6: SHORT 

Item 7: 151 

Item 8: 00:OO:OO 

Item 9: 00:05:00 

(Data Collection Platform Address) 

(Self-Timed Transmit Channel Number) 

(Self-Timed Transmission Interval) 

(Self-Timed Transmission Offset) 

(Transmission Window Length) 

(Satellite Link Parameter: Preamble) 

(Random Transmit Channel Number) 

(Random Transmission Interval) 

(Random Disable Time) 

Current Date and Time: 05/12/01 22:51:46 

Enter new Date and Time: 05/12/02 22:53:00 

GOES Transmitter Initialization ... successful 

Note: Next sample interval begins in 6 minutes and 59 seconds 

GOES MRNU 

(Cdm) Change Item n To Value m (E) Save Parameters To EEPROM 

(D) Run GOES Diagnostics (U) User Menu 

(R) Reset GOES Errors (Q) Quit 

(I) Initialize GOES (H) Help 

Item 1: 2638202~ 

Item 2: 104 

Item 3: 00:00:10:00 

Item 4: 00:05:40 

Item 5: 1 

Item 6: SHORT 

Item 7: 151 

Item 8: 00:OO:OO 

Item 9: 00:05:00 

(Data Collection Platform Address) 

(Self-Timed Transmit Channel Number) 

(Self-Timed Transmission Interval) 

(Self-Timed Transmission Offset) 

(Transmission Window Length) 

(Satellite Link Parameter: Preamble) 

(Random Transmit Channel Number) 

(Random Transmission Interval) 

(Random Disable Time) 
USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

SAMPLE PERIOD MRNU 

(Cdm) Change Item n To Value m (Q) Quit 

(E) Save Parameters To EEPROM (H) Help 

(U) User Menu 

Item 1 : 600 (Sample Interval Time) 

Item 2: 14 (Sample Duration Time) 

Item 3: 0 (Sample Time Offset) 

> u 

Checking Scan List records 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

Enter Administrator Password:   

Waiting for all data acquisition tasks to finish 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu (W) Password Menu 

(P) Process Menu (R) Reset System 

(D) Data Output Menu (E) Save Parameters To EEPROM 

(T) Sensor Timing Loop Menu (U) User Menu 

(0) Output Message Timing Menu (Q) Quit 

(I,) System Load Menu (H) Help 

SENSOR MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Sensor Items for Record 1 of 9: 

Item 1 : Sensor Type Code 2 (1 8-bit Single-Ended A to D) 

Item 2: Sensor Name dtOl 

Item 3: Sensor Input Channel 1 - 

Item 4: Analog Channel Gain 1 

Item 5: Analog Channel Attenuation 1 

Item 6: Switched Power Code 0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 
Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannup Time 0 

Item 10: Sensor Sample Count 8 

Item 11 : Maximum Sensor Readings 0 

Item 12: Sensor Timing Loop 2 (1.0 seconds) 

Item 13: Conversion Coefficient A 0 

Item 14: Conversion Coefficient B 0.5 

Item 15: Conversion Coefficient C 0 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Sensor Items for Record 2 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: Switched Power Code 

Item 7: Sensor Excitation Voltage Code 

Item 8: Switched Excitation Return 

Item 9: Switched Power Wannup Time 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 3 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: switched Power Code 

Item 7: Sensor Excitation Voltage Code 

Item 8: Switched Excitation Return 

Item 9: Switched Power Wannup Time 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

2 (1 8-bit Single-Ended A to D) 

dt01 

1 - 

1 

10 

0 (NO SWITCHED POWER) 

0 (NO EXCITATION VOLTAGE) 

0 

0 

2 (1.0 seconds) 

0 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

2 (1 8-bit Single-Ended A to D) 

dt02 

I+ 

1 

10 

0 (NO SWITCHED POWER) 

0 (NO EXCITATION VOLTAGE) 

0 

0 

8 

0 

2 (1.0 seconds) 

0 

0.5 

0 
N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

2 (1 8-bit Single-Ended A to D) 

short 

2- 

1 

1 

Sensor Items for Record 4 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Invut Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: Switched Power Code 0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannup Time 0 

Item 10: Sensor Sample Count 8 

Item 11 : Maximum Sensor Readings 0 

Item 12: Sensor Timing Loop 2 (1.0 seconds) 

Item 13: Conversion Coefficient A 0 

Item 14: Conversion Coefficient B 0.5 

Item 15: Conversion Coefficient C 0 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Sensor Items for Record 5 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: Switched Power Code 

Item 7: Sensor Excitation Voltage Code 

Item 8: Switched Excitation Return 

Item 9: Switched Power Wannup Time 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

2 (1 8-bit Single-Ended A to D) 

bar0 

2+ 

1 

10 

0 (NO SWITCHED POWER) 

0 (NO EXCITATION VOLTAGE) 

0 

0 

8 

0 

2 (1.0 seconds) 

0 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 
Sensor Items for Record 6 of 9: 

Item 1 : Sensor Type Code 2 (1 8-bit Single-Ended A to D) 

Item 2: Sensor Name x-axis 

Item 3: Sensor Input Channel 3- 

Item 4: Analog Channel Gain 1 

Item 5: Analog Channel Attenuation 10 

Item 6: Switched Power Code 0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannuv Time 0 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 7 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: Switched Power Code 

8 

0 

2 (1.0 seconds) 

0 

0.5 

0 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

2 (1 8-bit Single-Ended A to D) 

y-axis 

3+ 

1 

10 

0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannup Time 0 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

8 

0 

2 (1.0 seconds) 

0 

0.5 

0 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 8 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: Switched Power Code 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

2 (1 8-bit Single-Ended A to D) 

temp 

4- 

1 

10 

0 (NO SWITCHED POWER) 
Item 7: Sensor Excitation Voltage Code 2 (EXC = 2.50 VDC) 

Item 8: Switched Excitation Return A 

Item 9: Switched Power Warmup Time 0 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 9 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 6: Switched Power Code 

2 (1.0 seconds) 

0 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

1 (12-bit Analog to Digital) 

Battery 

BATTERY VOLTAGE 

0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Warmup Time 0 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu 

(P) Process Menu 

(D) Data Output Menu 

(T) Sensor Timing Loop Menu 

(0) Output Message Timing Menu 

(I,) System Load Menu 

PROCESS MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Process Items for Record 1 of 9: 

1 (0.5 seconds) 

0 

(W) Password Menu 

(R) Reset System 

(E) Save Parameters To EEPROM 

(U) User Menu 

(Q) Quit 

0 Help 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data Sl : dtOl 
PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 2 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S2 : dtOl 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 3 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S3 : dt02 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 4 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S4 : short 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 5 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S5 : bare 
PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 6 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S6 : x-axis 

PROCESS Mwm 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 7 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S7 : y-axis 

PROCESS MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 8 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S8 : temp 

PROCESS MRNU 

(Cnlm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Process Items for Record 9 of 9: 

Item 1 : Process Category 1 : General 

Item 2: Process Number 2 : Averaging Process 

Item 3: Input for Average Data S9 : Battery 

ZEN0 PROGRAM MENU 

(S) Sensor Menu (W) Password Menu 
(P) Process Menu 

(D) Data Output Menu 

(T) Sensor Timing Loop Menu 

(0) Output Message Timing Menu 

(I,) System Load Menu 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 1 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

(R) Reset System 

(E) Save Parameters To EEPROM 

(U) User Menu 

(Q) Quit 

0 Help 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

dt0l 

PI1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

Data Items for Record 2 of 9: 

Item 1: Field Type code 12 : GOES Binary Format Field 

Item 2: Output Message($ 1 

Item 3: Field Name dtOl 

Item 4: Input Record and Element P2.1 

Item 5: Field Decimal Places 0 

Item 6: Field Width 3 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Data Items for Record 3 of 9: 

Item 1: Field Type code 12 : GOES Binary Format Field 

Item 2: Output Message($ 1 

Item 3: Field Name dt02 

Item 4: Input Record and Element P3.1 

Item 5: Field Decimal Places 0 

Item 6: Field Width 3 
DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Data Items for Record 4 of 9: 

Item 1: Field Type code 12 : GOES Binary Format Field 

Item 2: Output Message($ 1 

Item 3: Field Name short 

Item 4: Input Record and Element P4.1 

Item 5: Field Decimal Places 0 

Item 6: Field Width 3 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Data Items for Record 5 of 9: 

Item 1: Field Type code 12 : GOES Binary Format Field 

Item 2: Output Message($ 1 

Item 3: Field Name bar0 

Item 4: Input Record and Element P5.1 

Item 5: Field Decimal Places 0 

Item 6: Field Width 3 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Data Items for Record 6 of 9: 

Item 1: Field Type code 12 : GOES Binary Format Field 

Item 2: Output Message($ 1 

Item 3: Field Name x-axis 

Item 4: Input Record and Element P6.1 

Item 5: Field Decimal Places 0 

Item 6: Field Width 3 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 
Data Items for Record 7 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 8 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 9 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu 

(P) Process Menu 

(D) Data Output Menu 

(T) Sensor Timing Loop Menu 

(0) Output Message Timing Menu 

(I,) System Load Menu 

SENSOR TTMTNG LOOP MRNU 

(Cdm) Change Item n To Value m 

(Z) Zeno Program Menu 

12 : GOES Binary Format Field 

1 

y-axis 

P7.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

temp 

P8.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

BATTERY 

P9.1 

0 

3 

(W) Password Menu 

(R) Reset System 

(E) Save Parameters To EEPROM 

(U) User Menu 

(Q) Quit 

0 Help 
Item 1 : 0.5 (Timing Loop #I Period) 

Item 2: 1.0 (Timing Loop #2 Period) 

Item 3: 120.0 (Timing Loop #3 Period) 

Item 4: 10.0 (Timing Loop #4 Period) 

OUTPUT MRSSAGE TTMTNG MRNU 

(Cdm) Change Item n To Value m (H) Help 

(Z) Zeno Program Menu 

Item 1: COM2 

Item 2: COM3 

Item 3: COM3 

Item 4: COM3 

Item 5: 0.0 

Item 6: 0.0 

Item 7: 0.0 

Item 8: 0.0 

(Data Output Message #I COM Port) 

(Data Output Message #2 COM Port) 

(Data Output Message #3 COM Port) 

(Data Output Message #4 COM Port) 

(Data Output Message #I Period) 

(Data Output Message #2 Period) 

(Data Output Message #3 Period) 

(Data Output Message #4 Period) 

> e 

Verifying parameters can be stored in EEPROM 

Saving parameters to EEPROM . . . 

Saving Scan List parameters to EEPROM . . . 

475 out of 2048 bytes used in EEPROM. 

Total EEPROM Writes: 43 Checksum: 238 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu (W) Password Menu 

(P) Process Menu (R) Reset System 

(D) Data Output Menu (E) Save Parameters To EEPROM 

(T) Sensor Timing Loop Menu (U) User Menu 

(0) Output Message Timing Menu (Q) Quit 

(I,) System Load Menu (H) Help 

> b 

Waiting for all data acquisition tasks to finish . . . 

BACK DOOR MRNU 

(Cdm) Change Item n To Value m (X) Display Stack Usage 

(F) Calculate Free Heap Memory (E) Save Parameters To EEPROM 

(A) Auto-Calibrate Compass (U) User Menu 

(I) Initialize Compass (H) Help 

(R) Reset Parameters To Defaults 

Item 1: 16777 

Item 2: 1 

Item 3: 60 

Item 4: 18 

Item 5: COUNTS 

Item 6: YES 

Item 7: 32768.00 

Item 8: 0 

Item 9: 1.0000 

(Processor Clock Speed) 

(RAMROM Wait States) 

(50160 Hz Rejection For 18-bit ADC) 

(1311 8 Bit Operation Of 18-bit ADC) 

(A To D Conversion Results) 

(Expert Menu Mode) 

(Real-time Clock Crystal Frequency At 25 Degrees C) 

(Speed vs. Noise Tradeoffs For 18-bit ADC, Factory Only) 

(12-bit ADC Correction Factor) 
SAMPLE PERIOD MRNU 

(Cdm) Change Item n To Value m (Q) Quit 

(E) Save Parameters To EEPROM (H) Help 

(U) User Menu 

Item 1 : 600 (Sample Interval Time) 

Item 2: 14 (Sample Duration Time) 

Item 3: 0 (Sample Time Offset) 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

SYSTEM FUNCTIONS MRNU 

(Cdm) Change Item n to Value m (E) Save Parameters To EEPROM 

(S) System Date And Time (U) User Menu 

(T) Calibrate Internal Temperature (Q) Quit 

(V) Program Version (H) Help 

Item 1 : 1223 (Primary UnitExperiment ID) 

Item 2: 2 (Secondary UnitExperiment ID) 

Item 3: 1 (Data Dump Format) 

Item 4: 1 (Real Time Output Format) 

Item 5: 0 (Add Compass To Vane) 

Item 6: 0 (Compass Offset) 

Item 7: 0 (Barometer Elevation) 

> v 

ZENO-3200 using ZENOSOFT V1.813 Dec 1 1998 14:56:07 CS EEBE 

(C)opyright 1995-1998, Coastal Environmental Systems, Seattle, WA, USA. 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

>cg1 

Current Date and Time: 05/12/02 23:39:26 

Enter new Date and Time: 05/12/02 23:45:00 

GOES Transmitter Initialization ... successful 

Note: Next sample interval begins in 4 minutes and 59 seconds 

GOES MRNU 

(Cdm) Change Item n To Value m (E) Save Parameters To EEPROM 

(D) Run GOES Diagnostics (U) User Menu 

(R) Reset GOES Errors (Q) Quit 
(I) Initialize GOES (H) Help 

Item 1: 2638202~ 

Item 2: 104 

Item 3: 00:00:10:00 

Item 4: 00:05:40 

Item 5: 1 

Item 6: SHORT 

Item 7: 151 

Item 8: 00:OO:OO 

Item 9: 00:05:00 

(Data Collection Platform Address) 

(Self-Timed Transmit Channel Number) 

(Self-Timed Transmission Interval) 

(Self-Timed Transmission Offset) 

(Transmission Window Length) 

(Satellite Link Parameter: Preamble) 

(Random Transmit Channel Number) 

(Random Transmission Interval) 

(Random Disable Time) 

Verifying GOES Transmitter Initialization ... successful 

Exiting user interface 

KELLER WELL PROGRAM 

"apply power to ZEN0 

Watchdog Reset 

ZENO-3200 using ZENOSOFT V1.813 Dec 1 1998 14:56:07 CS EEBE 

(C)opyright 1995-1998, Coastal Environmental Systems, Seattle, WA, USA. 

System Time = 05/12/02 23:47:05 

Initializing Zeno 3200 ... / 

Verifying GOES Transmitter Initialization ... 

WARNING: GOES Transmitter not initialized since system restart. 

Use the 'Initialize GOES' command inside the GOES menu. 
Zeno 3200 is Data Sampling. Type 'U<enter> to access the User Interface 

('V <enter> entering this command will not appear on screen. However the response follows) 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

COMMTJNTCATIONS MRNU 

(Cdm) Change Item n To Value m (Tn) Terminal Mode On COM Port n 

(M) Modem Menu (E) Save Parameters To EEPROM 

(P) Power Control Menu (U) User Menu 

(G) GOES Menu (Q) Quit 

(D) Digital Control Menu (H) Help 

Item 1: 9600 

Item 2: 9600 

Item 3: 9600 

Item 4: RS232 

Item 5: GOES 

Item 6: RS232 

Item 7: NO 

Item 8: NO 

(COM1 Baud Rate) 

(COM2 Baud Rate) 

(COM3 Baud Rate) 

(COM1 Port Type) 

(COM2 Port Type) 

(COM3 Port Type) 

(COM3 User Interface Exclusive) 

(Enable Exclusive CCSAIL Access) 

>g 

GOES MRNU 

(Cdm) Change Item n To Value m (E) Save Parameters To EEPROM 

(D) Run GOES Diagnostics (U) User Menu 

(R) Reset GOES Errors (Q) Quit 

(I) Initialize GOES (H) Help 

Item 1 : 2637fOd6 (Data Collection Platform Address) 

Item 2: 104 (Self-Timed Transmit Channel Number) 

Item 3: 00:OO: 10:OO (Self-Timed Transmission Interval) 

Item 4: 00:06:20 (Self-Timed Transmission Offset) 

Item 5: 1 (Transmission Window Length) 

Item 6: SHORT (Satellite Link Parameter: Preamble) 

Item 7: 151 (Random Transmit Channel Number) 

Item 8: 00:00:00 (Random Transmission Interval) 

Item 9: 00:05:00 (Random Disable Time) 

Current Date and Time: 05/12/01 22:51:46 

Enter new Date and Time: 05/12/02 22:53:00 

GOES Transmitter Initialization ... successful 

Note: Next sample interval begins in 6 minutes and 59 seconds 

GOES MRNU 

(Cdm) Change Item n To Value m (E) Save Parameters To EEPROM 

(D) Run GOES Diagnostics (U) User Menu 

(R) Reset GOES Errors (Q) Quit 
(I) Initialize GOES (H) Help 

Item 1: 2637fOd6 

Item 2: 104 

Item 3: 00:00:10:00 

Item 4: 00:06:20 

Item 5: 1 

Item 6: SHORT 

Item 7: 151 

Item 8: 00:OO:OO 

Item 9: 00:05:00 

(Data Collection Platform Address) 

(Self-Timed Transmit Channel Number) 

(Self-Timed Transmission Interval) 

(Self-Timed Transmission Offset) 

(Transmission Window Length) 

(Satellite Link Parameter: Preamble) 

(Random Transmit Channel Number) 

(Random Transmission Interval) 

(Random Disable Time) 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

SAMPLE PERIOD MRNU 

(Cnlm) Change Item n To Value m (Q) Quit 

(E) Save Parameters To EEPROM (H) Help 

(U) User Menu 

Item 1 : 600 (Sample Interval Time) 

Item 2: 14 (Sample Duration Time) 

Item 3: 0 (Sample Time Offset) 

> u 

Checking Scan List records 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

> zUSER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

Enter Administrator Password: ** ** 

Waiting for all data acquisition tasks to finish 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu (W) Password Menu 

(P) Process Menu (R) Reset System 

(D) Data Output Menu (E) Save Parameters To EEPROM 

(T) Sensor Timing Loop Menu (U) User Menu 

(0) Output Message Timing Menu (Q) Quit 
(I,) System Load Menu 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

Sensor Items for Record 1 of 9: 

Item 1 : Sensor Tvne , Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: Switched Power Code 0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannup Time 0 

2 (1 8-bit Single-Ended A to D) 

dt0l 

1 - 

1 

1 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 2 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: Switched Power Code 

Item 7: Sensor Excitation Voltage Code 

Item 8: Switched Excitation Return 

Item 9: Switched Power Wannup Time 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

2 (1.0 seconds) 

0 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

2 (1 8-bit Single-Ended A to D) 

dt0l 

1 - 

1 

10 

0 (NO SWITCHED POWER) 

0 (NO EXCITATION VOLTAGE) 

0 

0 

8 

0 

2 (1.0 seconds) 

0 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 
(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Sensor Items for Record 3 of 9: 

Item 1 : Sensor Type Code 2 (1 8-bit Single-Ended A to D) 

Item 2: Sensor Name dt02 

Item 3: Sensor Input Channel I+ 

Item 4: Analog Channel Gain 1 

Item 5: Analog Channel Attenuation 10 

Item 6: Switched Power Code 0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannuv Time 0 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 4 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: Switched Power Code 

8 

0 

2 (1.0 seconds) 

0 

0.5 

0 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

2 (1 8-bit Single-Ended A to D) 

short 

2- 

1 

1 

0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannup Time 0 

Item 10: Sensor Sample Count 8 

Item 11 : Maximum Sensor Readings 0 

Item 12: Sensor Timing Loop 2 (1.0 seconds) 

Item 13: Conversion Coefficient A 0 

Item 14: Conversion Coefficient B 0.5 

Item 15: Conversion Coefficient C 0 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Sensor Items for Record 5 of 9: 

Item 1 : Sensor Type Code 2 (1 8-bit Single-Ended A to D) 

Item 2: Sensor Name bar0 

Item 3: Sensor Input Channel 2+ 

Item 4: Analog Channel Gain 1 
Item 5: Analog Channel Attenuation 

Item 6: Switched Power Code 

Item 7: Sensor Excitation Voltage Code 

Item 8: Switched Excitation Return 

Item 9: Switched Power Wannup Time 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

10 

0 (NO SWITCHED POWER) 

0 (NO EXCITATION VOLTAGE) 

0 

2 (1.0 seconds) 

0 

SENSOR MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Sensor Items for Record 6 of 9: 

Item 1 : Sensor Type Code 2 (1 8-bit Single-Ended A to D) 

Item 2: Sensor Name x-axis 

Item 3: Sensor Input Channel 3- 

Item 4: Analog Channel Gain 1 

Item 5: Analog Channel Attenuation 10 

Item 6: Switched Power Code 0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 0 (NO EXCITATION VOLTAGE) 

Item 8: Switched Excitation Return 0 

Item 9: Switched Power Wannup Time 0 

Item 10: Sensor Sample Count 8 

Item 11 : Maximum Sensor Readings 0 

Item 12: Sensor Timing Loop 2 (1.0 seconds) 

Item 13: Conversion Coefficient A 0 

Item 14: Conversion Coefficient B 0.5 

Item 15: Conversion Coefficient C 0 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m (N) Go To Next Record 

(A) Insert After This Record (P) Go To Previous Record 

(B) Insert Before This Record (X) Delete All Records 

(D) Delete This Record (Z) Zeno Program Menu 

(Jn) Jump To Record n (H) Help 

Sensor Items for Record 7 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

Item 6: Switched Power Code 

Item 7: Sensor Excitation Voltage Code 

Item 8: Switched Excitation Return 

Item 9: Switched Power Wannup Time 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

2 (1 8-bit Single-Ended A to D) 

y-axis 

3+ 

1 

10 

0 (NO SWITCHED POWER) 

0 (NO EXCITATION VOLTAGE) 

0 

0 

8 

0 

2 (1.0 seconds) 
Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

N 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 8 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 4: Analog Channel Gain 

Item 5: Analog Channel Attenuation 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

2 (1 8-bit Single-Ended A to D) 

temp 

4- 

1 

10 

Item 6: Switched Power Code 0 (NO SWITCHED POWER) 

Item 7: Sensor Excitation Voltage Code 2 (EXC = 2.50 VDC) 

Item 8: Switched Excitation Return A 

Item 9: Switched Power Wannup Time 0 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

SENSOR MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Sensor Items for Record 9 of 9: 

Item 1 : Sensor Type Code 

Item 2: Sensor Name 

Item 3: Sensor Input Channel 

Item 6: Switched Power Code 

2 (1.0 seconds) 

0 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

Item 7: Sensor Excitation Voltage Code 

Item 8: Switched Excitation Return 

Item 9: Switched Power Wannup Time 

Item 10: Sensor Sample Count 

Item 11 : Maximum Sensor Readings 

Item 12: Sensor Timing Loop 

Item 13: Conversion Coefficient A 

Item 14: Conversion Coefficient B 

Item 15: Conversion Coefficient C 

1 (12-bit Analog to Digital) 

Battery 

BATTERY VOLTAGE 

0 (NO SWITCHED POWER) 

ZEN0 PROGRAM MRNL 

(S) Sensor Menu 

(P) Process Menu 

0 (NO EXCITATION VOLTAGE) 

0 

0 

1 

0 

1 (0.5 seconds) 

0 

1 

0 

(W) Password Menu 

(R) Reset System 
(D) Data Output Menu 

(T) Sensor Timing Loop Menu 

(0) Output Message Timing Menu 

(I,) System Load Menu 

PROCESS MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Process Items for Record 1 of 9: 

Item 1 : Process Category 

Item 2: Process Number 

Item 3: Input for Average Data 

PROCESS MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Process Items for Record 2 of 9: 

Item 1 : Process Category 

Item 2: Process Number 

Item 3: Input for Average Data 

PROCESS MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Process Items for Record 3 of 9: 

Item 1 : Process Category 

Item 2: Process Number 

Item 3: Input for Average Data 

PROCESS MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Process Items for Record 4 of 9: 

(E) Save Parameters To EEPROM 

(U) User Menu 

(Q) Quit 

0 Help 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

1 : General 

2 : Averaging Process 

Sl : dtOl 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

1 : General 

2 : Averaging Process 

S2 : dt0l 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

1 : General 

2 : Averaging Process 

S3 : dt02 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

Item 1 : Process Category 1 : General 
Item 2: Process Number 

Item 3: Input for Average Data 

PROCESS MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Process Items for Record 5 of 9: 

Item 1 : Process Category 

Item 2: Process Number 

Item 3: Input for Average Data 

PROCESS MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Process Items for Record 6 of 9: 

Item 1 : Process Category 

Item 2: Process Number 

Item 3: Input for Average Data 

PROCESS MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Process Items for Record 7 of 9: 

Item 1 : Process Category 

Item 2: Process Number 

Item 3: Input for Average Data 

PROCESS MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Process Items for Record 8 of 9: 

Item 1 : Process Category 

Item 2: Process Number 

2 : Averaging Process 

S4 : short 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

1 : General 

2 : Averaging Process 

S5 : bar0 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

1 : General 

2 : Averaging Process 

S6 : x-axis 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

1 : General 

2 : Averaging Process 

S7 : y-axis 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

1 : General 

2 : Averaging Process 

Item 3: Input for Average Data S8 : temp 
PROCESS MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Process Items for Record 9 of 9: 

Item 1 : Process Category 

Item 2: Process Number 

Item 3: Input for Average Data 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu 

(P) Process Menu 

(D) Data Output Menu 

(T) Sensor Timing Loop Menu 

(0) Output Message Timing Menu 

(I,) System Load Menu 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 1 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 2 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

1 : General 

2 : Averaging Process 

S9 : Battery 

(W) Password Menu 

(R) Reset System 

(E) Save Parameters To EEPROM 

(U) User Menu 

(Q) Quit 

0 Help 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

dt0l 

P2.1 

0 

3 
DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 3 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 4 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 5 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

short 

P4.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

bar0 

P5.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 
(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 6 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 7 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 8 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

DATA OUTPUT MRNU 

(Cdm) Change Item n To Value m 

(A) Insert After This Record 

(B) Insert Before This Record 

(D) Delete This Record 

(Jn) Jump To Record n 

Data Items for Record 9 of 9: 

Item 1: Field Type code 

Item 2: Output Message($ 

Item 3: Field Name 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

x-axis 

P6.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

y-axis 

P7.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

temp 

P8.1 

0 

3 

(N) Go To Next Record 

(P) Go To Previous Record 

(X) Delete All Records 

(Z) Zeno Program Menu 

0 Help 

12 : GOES Binary Format Field 

1 

BATTERY 
Item 4: Input Record and Element 

Item 5: Field Decimal Places 

Item 6: Field Width 

ZENO PROGRAM MRNU 

(S) Sensor Menu 

(P) Process Menu 

(D) Data Output Menu 

(T) Sensor Timing Loop Menu 

(0) Output Message Timing Menu 

(L) System Load Menu 

SENSOR TTMTNG LOOP MRNU 

(Cdm) Change Item n To Value m 

(Z) Zeno Program Menu 

(W) Password Menu 

(R) Reset System 

(E) Save Parameters To EEPROM 

(U) User Menu 

(Q) Quit 

0 Help 

Item 1: 0.5 (Timing Loop #I Period) 

Item 2: 1.0 (Timing Loop #2 Period) 

Item 3: 120.0 (Timing Loop #3 Period) 

Item 4: 10.0 (Timing Loop #4 Period) 

OUTPUT MRSSAGE TTMTNG MRNU 

(Cdm) Change Item n To Value m (H) Help 

(Z) Zeno Program Menu 

Item 1: COM2 

Item 2: COM3 

Item 3: COM3 

Item 4: COM3 

Item 5: 0.0 

Item 6: 0.0 

Item 7: 0.0 

Item 8: 0.0 

(Data Output Message #I COM Port) 

(Data Output Message #2 COM Port) 

(Data Output Message #3 COM Port) 

(Data Output Message #4 COM Port) 

(Data Output Message #I Period) 

(Data Output Message #2 Period) 

(Data Output Message #3 Period) 

(Data Output Message #4 Period) 

> e 

Verifying parameters can be stored in EEPROM 

Saving parameters to EEPROM . . . 

Saving Scan List parameters to EEPROM . . . 

475 out of 2048 bytes used in EEPROM. 

Total EEPROM Writes: 43 Checksum: 238 

ZENO PROGRAM MRNU 

(S) Sensor Menu (W) Password Menu 

(P) Process Menu (R) Reset System 

(D) Data Output Menu (E) Save Parameters To EEPROM 

(T) Sensor Timing Loop Menu (U) User Menu 

(0) Output Message Timing Menu (Q) Quit 

(L) System Load Menu (H) Help 

> b 

Waiting for all data acquisition tasks to finish 
BACK DOOR MRNU 

(Cdm) Change Item n To Value m (X) Display Stack Usage 

(F) Calculate Free Heap Memory (E) Save Parameters To EEPROM 

(A) Auto-Calibrate Compass (U) User Menu 

(I) Initialize Compass (H) Help 

(R) Reset Parameters To Defaults 

Item 1: 16777 

Item 2: 1 

Item 3: 60 

Item 4: 18 

Item 5: COUNTS 

Item 6: YES 

Item 7: 32768.00 

Item 8: 0 

Item 9: 1.0000 

(Processor Clock Speed) 

(RAMROM Wait States) 

(50160 Hz Rejection For 18-bit ADC) 

(1311 8 Bit Operation Of 18-bit ADC) 

(A To D Conversion Results) 

(Expert Menu Mode) 

(Real-time Clock Crystal Frequency At 25 Degrees C) 

(Speed vs. Noise Tradeoffs For 18-bit ADC, Factory Only) 

(12-bit ADC Correction Factor) 

SAMPLE PERIOD MRNU 

(Cdm) Change Item n To Value m (Q) Quit 

(E) Save Parameters To EEPROM (H) Help 

(U) User Menu 

Item 1 : 600 (Sample Interval Time) 

Item 2: 14 (Sample Duration Time) 

Item 3: 0 (Sample Time Offset) 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

SYSTEM FUNCTIONS MRNU 

(Cdm) Change Item n to Value m (E) Save Parameters To EEPROM 

(S) System Date And Time (U) User Menu 

(T) Calibrate Internal Temperature (Q) Quit 

(V) Program Version (H) Help 

Item 1 : 1223 (Primary UnitExperiment ID) 

Item 2: 2 (Secondary UnitExperiment ID) 

Item 3: 1 (Data Dump Format) 

Item 4: 1 (Real Time Output Format) 

Item 5: 0 (Add Compass To Vane) 

Item 6: 0 (Compass Offset) 

Item 7: 0 (Barometer Elevation) 

> v 

ZENO-3200 using ZENOSOFT V1.813 Dec 1 1998 14:56:07 CS EEBE 

(C)opyright 1995-1998, Coastal Environmental Systems, Seattle, WA, USA 
USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

>cg1 

Current Date and Time: 05/12/02 23:39:26 

Enter new Date and Time: 05/12/02 23:45:00 

GOES Transmitter Initialization ... successful 

Note: Next sample interval begins in 4 minutes and 59 seconds. 

GOES MRNU 

(Cnlm) Change Item n To Value m (E) Save Parameters To EEPROM 

(D) Run GOES Diagnostics (U) User Menu 

(R) Reset GOES Errors (Q) Quit 

(I) Initialize GOES (H) Help 

Item 1: 2637fOd6 

Item 2: 104 

Item 3: 00:00:10:00 

Item 4: 00:06:20 

Item 5: 1 

Item 6: SHORT 

Item 7: 151 

Item 8: 00:OO:OO 

Item 9: 00:05:00 

(Data Collection Platform Address) 

(Self-Timed Transmit Channel Number) 

(Self-Timed Transmission Interval) 

(Self-Timed Transmission Offset) 

(Transmission Window Length) 

(Satellite Link Parameter: Preamble) 

(Random Transmit Channel Number) 

(Random Transmission Interval) 

(Random Disable Time) 

>'I 

Verifying GOES Transmitter Initialization ... successful 

Exiting user interface. 

ZENOSOFT 

An alternate to entering the programming by hand would be to download the configuration: 
To Configure a ZENO using a laptop and program from a Diskette 

1 hook black ZENO db9 cable to ZENO and corn port 1 (db9) on back of laptop. 

2 open ZENO Hyperterm tile (configured to 9600 baud, 8 data bits, 1 start bit, 1 stop bit, no parity, no 

flow control) 

3 type U & enter (the U will not display) 

4 Computer responds with : 

USER MRNU 

(C) Communications Menu (T) Test Menu 

(F) System Functions Menu (Z) Zeno Program Menu 

(S) Sample Period Menu (Q) Quit 

(D) Data Retrieval Menu (H) Help 

5 type Z &enter 

6 Enter Administrator Password: (ZEN0 is password) 

7 Computer responds with : 

ZENO PROGRAM MRNU 

(S) Sensor Menu (W) Password Menu 

(P) Process Menu (R) Reset System 

(D) Data Output Menu (E) Save Parameters To EEPROM 

(T) Sensor Timing Loop Menu (U) User Menu 

(0) Output Message Timing Menu (Q) Quit 

(I,) System Load Menu (H) Help 

8 type L &enter 

9 Computer responds with : 

SYSTEM LOAD MRNU 

(R) Receive Configuration From Host (Z) Zeno Program Menu 

(T) Transmit Configuration From Zeno (H) Help 

type XR & enter 
the computer says: 

The Existing System Setup Will Be Lost. Continue? (Y/N) Y & enter 

Ready To Receive X-Modem System Configuration File 

Enter CONTROL-X To Abort transfer. 

At this point you have to go to the "Transfer" drop down of the menu bar at the top of the Hyperterm 

Window. 

Hit "Send File" 

Go to "Browse" and find the configuration you downloaded and saved in the laptop" in "Select File to 

Send" 

And click "open". 

(ifyou take too long you'll be kicked back to section 9) 

Once you hit "send" in the "Send File" window, the program is downloaded (you'll see a window telling 

the download progress.) 

You'll get a response similar to this: 

ccccccccc 

73 Parameters Loaded 

10 Sensor Command Sets Loaded 

10 Process Command Sets Loaded 

10 Data Command Sets Loaded 

SYSTEM LOAD MRNU 

(R) Receive Configuration From Host (Z) Zeno Program Menu 

(T) Transmit Configuration From Zeno (H) Help 

Precede The R Or T Command With An 'X' For X-Modem Transfer 

(e.g. Enter 'XR' To Receive A Configuration File Via X-Modem) 

14 You can now go back to (Z) 

ZEN0 PROGRAM MRNU 

(S) Sensor Menu (W) Password Menu 

(P) Process Menu (R) Reset System 

(D) Data Output Menu (E) Save Parameters To EEPROM 

(T) Sensor Timing Loop Menu (U) User Menu 

(0) Output Message Timing Menu (Q) Quit 

(I,) System Load Menu (H) Help 

Verifying parameters can be stored in EEPROM 

Saving parameters to EEPROM . . . 

Saving Scan List parameters to EEPROM . . . 

51 8 out of 2048 bytes used in EEPROM. 

Total EEPROM Writes: 135 Checksum: 107 
ZEN0 PROGRAM MENU 

(S) Sensor Menu (W) Password Menu 

(P) Process Menu (R) Reset System 

(D) Data Output Menu (E) Save Parameters To EEPROM 

(T) Sensor Timing Loop Menu (U) User Menu 

(0) Output Message Timing Menu (Q) Quit 

(I,) System Load Menu (H) Help 

Then : 

U C G & enter 

GOES MENU 

(Cnlm) Change Item n To Value m (E) Save Parameters To EEPROM 

(D) Run GOES Diagnostics (U) User Menu 

(R) Reset GOES Errors (Q) Quit 

(I) Initialize GOES (H) Help 

Item 1 : 2637E3AO (Data Collection Platfonn Address) 

Item 2: 104 (Self-Timed Transmit Channel Number) 

Item 3: 00:OO: 10:OO (Self-Timed Transmission Interval) 

Item 4: 00:06:20 (Self-Timed Transmission Offset) 

Item 8: 1 (Transmission Window Length) 

Item 6: SHORT (Satellite Link Parameter: Preamble) 

Item 7: 151 (Random Transmit Channel Number) 

Item 8: 00:OO:OO (Random Transmission Interval) 

Item 9: 00:05:00 (Random Disable Time) 

Ensure the platform address and self-timed transmission offset are correct. 

I & enter 

Current Date and Time: 04/10/26 18:30:50 

Enter new Date and Time: ??/??I?? ??:??:?? & enter 

This should initialize everything and you're done 
The following are configuration for the Hawaii sites: 

Strip Road: 

* Zeno 3200 System Setup File 

*Program Version And Date: ZENO-3200 using ZENOSOFT VI.813 Dec 1 1998 14:56:07 CS EEBE 

* (C)opyright 1995-1998, Coastal Environmental Systems, Seattle, WA, USA. 

* Setup File Date And Time: 00/09/14 22:28:46 

PARAMI 6000142240201223296009600 

PARAM2960004001 1000 

PARAM31677716018000012 

PARAM42201 132768000-150 

PARAM5000030000000 

PARAM60008521632006411969601042560153610 

PARAM715101280010000 

PARAMS "NONR" "NONR" "NONR" "NONR" "NONR" "NONR" ,111 "ZENO" 

SENSOR 2 "dtOl" 8 0 0 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "dtOl" 8 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "dt02" 1 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR2"short"900000080200.500 

SENSOR 2 "baro" 2 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "x-axis" 10 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "y-axis" 3 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR2"temp"1103021080200.500 

SENSOR1"Battery"20000001010100 

PROCESS 1 2 S1.1 

PROCESS 1 2 S2.1 

PROCESS 1 2 S3.1 

PROCESS 1 2 S4.1 

PROCESS 1 2 S5.1 

PROCESS 1 2 S6.1 

PROCESS 1 2 S7.1 

PROCESS 1 2 S8.1 

PROCESS 1 2 S9.1 

DATA12 1 "dtO1"Pl.l 03 1 

DATA 12 1 "dtO1" P2.1 0 3 1 

DATA 12 1 "dt02" P3.1 0 3 1 

DATA 12 1 "short" P4.1 0 3 1 

DATA 12 1 "baro" P5.1 0 3 1 

DATA 12 1 "x-axis" P6.1 0 3 1 

DATA 12 1 "y-axis" P7.1 0 3 1 

DATA 12 1 "temp" P8.1 0 3 1 

DATA 12 1 "BATTERY" P9.1 0 3 1 

EOF 

nnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnn 

nnnnnnnnnnn 

Hokukano 

* Zeno 3200 System Setup File 

*Program Version And Date: ZENO-3200 using ZENOSOFT VI.813 Dec 1 1998 14:56:07 CS EEBE 

* (C)opyright 1995-1998, Coastal Environmental Systems, Seattle, WA, USA. 

* Setup File Date And Time: 00/09/19 22:58:20 

PARAMI 6000142240201223296009600 

PARAM2960004001 1000 

PARAM31677716018000012 

PARAM42201 132768000-150 
PARAM5000030000000 

PARAM600085216320064123720410425601295 10 

PARAM715101280010000 

PARAMS "NONR" "NONR" "NONR" "NONR" "NONR" "NONR" ,111 "ZENO" 

SENSOR 2 "dtOl" 8 0 0 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "dtOl" 8 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "dt02" 1 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR2"short"900000080200.500 

SENSOR 2 "baro" 2 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "x-axis" 10 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "y-axis" 3 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR2"temp"1103021080200.500 

SENSOR1"Batten/"20000001010100 

PROCESS 1 2 S1.1 

PROCESS 1 2 S2.1 

PROCESS 1 2 S3.1 

PROCESS 1 2 S4.1 

PROCESS 1 2 S5.1 

PROCESS 1 2 S6.1 

PROCESS 1 2 S7.1 

PROCESS 1 2 S8.1 

PROCESS 1 2 S9.1 

DATA12 1 "dtO1"Pl.l 03 1 

DATA 12 1 "dtO1" P2.1 0 3 1 

DATA 12 1 "dt02" P3.1 0 3 1 

DATA 12 1 "short" P4.1 0 3 1 

DATA 12 1 "baro" P5.1 0 3 1 

DATA 12 1 "x-axis" P6.1 0 3 1 

DATA 12 1 "y-axis" P7.1 0 3 1 

DATA 12 1 "temp" P8.1 0 3 1 

DATA 12 1 "BATTERY" P9.1 0 3 1 

EOF 

nnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnn 

nnnnnnnnnnn 

Mama Loa 

* Zeno 3200 System Setup File 

*Program Version And Date: ZENO-3200 using ZENOSOFT VI.813 Dec 1 1998 14:56:07 CS EEBE 

* (C)opyright 1995-1998, Coastal Environmental Systems, Seattle, WA, USA. 

* Setup File Date And Time: 00/09/17 03:45:07 

PARAMI 6000142240201223296009600 
PARAM5000030000000 

PARAM60008521632006411969601042560153610 

PARAM715101280010000 

PARAMS "NONR" "NONR" "NONR" "NONR" "NONR" "NONR" ,111 "ZENO" 

SENSOR 2 "dtOl" 8 0 0 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "dtOl" 8 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "dt02" 1 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR2"short"900000080200.500 

SENSOR 2 "baro" 2 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "x-axis" 10 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "y-axis" 3 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR2"temp"1103021080200.500 

SENSOR1"Batten/"20000001010100 

PROCESS 1 2 S1.1 

PROCESS 1 2 S2.1 

PROCESS 1 2 S3.1 

PROCESS 1 2 S4.1 

PROCESS 1 2 S5.1 

PROCESS 1 2 S6.1 

PROCESS 1 2 S7.1 

PROCESS 1 2 S8.1 

PROCESS 1 2 S9.1 

DATA12 1 "dtO1"Pl.l 03 1 

DATA 12 1 "dtO1" P2.1 0 3 1 

DATA 12 1 "dt02" P3.1 0 3 1 

DATA 12 1 "short" P4.1 0 3 1 

DATA 12 1 "baro" P5.1 0 3 1 

DATA 12 1 "x-axis" P6.1 0 3 1 

DATA 12 1 "y-axis" P7.1 0 3 1 

DATA 12 1 "temp" P8.1 0 3 1 

DATA 12 1 "BATTERY" P9.1 0 3 1 

EOF 

nnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnnn 

nnnnnnnnnnn 
Keller Well 

* Zeno 3200 System Setup File 

*Program Version And Date: ZENO-3200 using ZENOSOFT VI.813 Dec 1 1998 14:56:07 CS EEBE 

* (C)opyright 1995-1998, Coastal Environmental Systems, Seattle, WA, USA. 

* Setup File Date And Time: 00/09/15 00:48:24 

PARAMI 6000142240201223296009600 

PARAM2960004001 1000 

PARAM31677716018000012 

PARAM42201 132768000-150 

PARAM5000030000000 

PARAM60008521632006412003421042560132010 

PARAM715101280010000 

PARAMS "NONR" "NONR" "NONR" "NONR" "NONR" "NONR" ,111 "ZENO" 

SENSOR 2 "dtOl" 8 0 0 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "dtOl" 8 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "dt02" 1 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR2"short"900000080200.500 

SENSOR 2 "baro" 2 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "x-axis" 10 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "y-axis" 3 0 3 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR2"temp"1103021080200.500 

SENSOR1"BATTERY"20000001010100 

SENSOR 2 "water" 4 0 0 0 0 0 0 8 0 2 0 0.5 0 0 

SENSOR 2 "water" 12 0 0 0 0 0 0 8 0 2 0 0.5 0 0 

PROCESS 1 2 S1.1 

PROCESS 1 2 S2.1 

PROCESS 1 2 S3.1 

PROCESS 1 2 S4.1 

PROCESS 1 2 S5.1 

PROCESS 1 2 S6.1 

PROCESS 1 2 S7.1 

PROCESS 1 2 S8.1 

PROCESS 1 2 S9.1 

PROCESS 1 2 S10.1 

PROCESS 11 Sll.1 

DATA12 1 "dtO1"Pl.l 03 1 

DATA 12 1 "dtO1" P2.1 0 3 1 

DATA 12 1 "dt02" P3.1 0 3 1 

DATA 12 1 "short" P4.1 0 3 1 

DATA 12 1 "baro" P5.1 0 3 1 

DATA 12 1 "x-axis" P6.1 0 3 1 

DATA 12 1 "y-axis" P7.1 0 3 1 

DATA 12 1 "temp" P8.1 0 3 1 

DATA 12 1 "BATTERY" P9.1 0 3 1 

DATA 12 1 "water" PI01 0 3 1 

DATA 12 1 "water" PI11 0 3 1 

EOF 
17. 

Installation DT112 Dilatometer HAWAIIIManunoth 

Site Name 

Instrument #~ 

Surface test 

Wire Color 

Black 

White 

Green 

Orange 

Gdlk trace 

Blue 

WhtBlk trace 

Red/Blk trace 

Blmt trace 

Org/Blk trace 

BlkRed trace 

Red 

BldBlk trace 

Red/Wilt trace 

G&t trace 

Blk/Wilt trace 

Pin numbel 

1 

2 

4 

5 

9 

6 

7 

8 

15 

10 

16 

3 

11 

I3 

14 

12 

Strainmeter Status 

DT1 status 

Power up 

Valve 1 open--close__ 

PC DVM 

Comp A 

Valve 1 open__close__ 

PC DVM 

Comp A 

Valve 1 open--close__ 

PC DVM 

Comp A 

time date 

Depth of instrument^ 

component 

Voltage's 

ohm's downhole PC 

Valve 1 close + 

Valve 1 & 2 close - 

Valve 1 & 2 close - 

DT1 supply TN+ 

DT2 supply TN+ 

DT1&2 supply corn INDTl sig out Om + 

DTICO~ om - - -- - 

NC NC 

DT1&2 supply corn OUT - 

NC NC 

Valve 2 close + __- -- -- 

DT2 sig out Om + 

NC NC 

NC NC 

DT2 corn Om - 

DT2 status 

Valve 2 open__close- time date 

PC DVM 

Comp A - 

Valve 2 open__closePC DVM 

Comp A 

Valve 2 open--closePC DVM 

Comp A 

time date 

time date 

Shut Down (insure valves are open) Electronics power up(va1ves closed) 

Valve 1 open--close__ Valve 2 open--closePC DVM PC DVM 

time date 

Comp A Comp A 
18. Acknowledgements: 

Michael Acierno CIW 1 DTM (SOC Box Operation) 

Dale Evertsen (instrument design) 

Vincent Keller USGS 

Alan Linde CIW IDTM 

Glen Poe CIW IDTM retired (electronics design) 

Selwyn Sacks CIW IDTM (instrument design) 

Brian Scheilgh CIW IDTM (SOC Box Operation) 

Michael Seeman CIW IDTM retired (instrument manufacture) 

Stanley Silverman USGS 

Please refer to: Open-File 89-340 Borehole Dilatometer Installation, Operation and Maintenance at 

Sites along the San Andreas Fault, California, G.D. Myren and M.J.S. Johnston