- About NRC           - The Commission
      - Statutory Authority
      - Strategic Plan
      - Budget & Performance
      - Organization & Functions
      - Locations
      - History
      - Values
      - Policymaking
      - Radiation Protection
      - Fire Protection
      - Regulation
      - Emergency Preparedness   and Response
      - Public Affairs
      - Congressional Affairs
      - International Programs
      - State and Tribal Programs
      - Alternative Dispute   Resolution Programs
      - Contact Us
      - Employment
      - Contracting
      - Grants
      - Civil Rights
      -     
    - NuclearReactors           - Power Reactors
      - Research and Test Reactors
      - Operating Reactors
      - Operator Licensing
      - New Reactors
      - Advanced Reactors
      - Operator Licensing for New Reactors
      -     
    - NuclearMaterials           - Special Nuclear Material
      - Source Material
      - Byproduct Material
      - Med, Academic & Ind Uses
      - Source Materials Facilities
      - Uranium Recovery Facilites
      - Fuel Cycle Facilities
      - Materials Transportation
      -     
    - RadioactiveWaste           - Low-Level Waste
      - Waste Incidental to Reprocessing
      - High-Level Waste
      - Uranium Mill Tailings
      - LLW Disposal
      - HLW Disposal
      - Storage of Spent Nuclear Fuel
      - Transporation of Spent Nuclear Fuel
      -     
    - NuclearSecurity           - Domestic Safeguards
      - Information Security
      - Radioactive Material
      - &;   
    - Public Meetings& Involvement           - Open Government
      - Public Meeting Schedule
      - Commission Schedule
      - Conferences & Symposia
      - Documents for Comment
      - Involving Stakeholders
      - Info Quality Guidelines
      - Subscribe to E-mail Notices 
      -     

UNITED STATESNUCLEAR REGULATORY COMMISSIONOFFICE OF NUCLEAR MATERIAL SAFETY AND SAFEGUARDSWASHINGTON, D.C. 20555

December 13, 1996

**Contents**

- Addressees
- Purpose
- Description of Circumstances
- Discussion
- Attachment 1: Fraction of Original Sr-90 Activity Remaining after (t) Years (table)

## Addressees

All U.S. Nuclear Regulatory Commission Medical Use Licensees authorized to use strontium-90 (Sr-90) eye applicators.

## Purpose

The U.S. Nuclear Regulatory Commission (NRC) is issuing this information notice to alert addressees to recent misadministrations causedby incorrect source strength determinations of Sr-90 eye applicators.  It is expected that recipients will review the information forapplicability to their facilities and consider actions, as appropriate, to avoid similar problems.  However, suggestions contained in thisinformation notice are not NRC requirements; therefore, no specific action nor written response is required.

## Description of Circumstances

The primary causes of two recent events have been problems with calibration and calculation of the dose rates from Sr-90 applicators.  Asa result, the administered doses were not within 20 percent of the prescribed dose.

Case 1.  In October 1995, during an inspection of a licensee authorized to use a Sr-90 eye applicator in Mayag�ez, Puerto Rico, it wasdetermined that the calibration record for the licensee's Sr-90 eye applicator was missing.  A previous owner of the source lost the originalcalibration certificate, and his medical physics consultant performed a check of the source strength.  This check was performed with aninappropriate measurement instrument and resulted in an incorrect determination of the source strength as 0.24 Gray (Gy)/sec (24 rad/sec). The licensee obtained the source in 1994 and relied on the erroneous source strength determination during treatments.  Also, the sourcehad not been decay-corrected by either owner since 1990.  NRC urged the licensee to have the eye applicator recalibrated and to beinstructed in the proper method to calculate the decay of the source.  

The licensee subsequently had the eye applicator recalibrated by the National Institute of Standards and Technology.  The resultantrecalibration revealed a dose rate of 0.53 Gray  (53 rad)/sec rather than the 0.24 Gy (24 rad)/sec used by the licensee.  The licensee hadprescribed a total dose of between 10 to 25 Gy (1000 to 2500 rad) for approximately 70 patients, but actually administered about twice thisamount.  An NRC medical consultant has reviewed the cases and has recommended that the patients be evaluated for any adverse effects. 

Case 2.  In November 1995, while reviewing treatment records during a routine NRC inspection of a licensee authorized to use a Sr-90 eyeapplicator in Honolulu, Hawaii, an NRC inspector determined that the licensee had incorrectly calculated the decay of the strength of a Sr-90 eye applicator and subsequently delivered doses 20 percent over the prescribed doses to 16 patients.  The root cause of this event wasthe licensee improperly calculating the decay of the Sr-90 source.  Specifically, the licensee linearly decayed the strength of the sourceinstead of logarithmically decaying the source strength of the eye applicator.  Before May 1995, a technologist had correctly calculated thedecay of the source; however, when this technologist left, the physician incorrectly calculated the source decay based on a linear decayfrom the previous calculations.  In addition, the incorrect half-life of Sr-90 was used in the calculation.

Following a correct determination of the source strength, the licensee determined that  17 misadministrations involving 16 patientsoccurred, from May 6, 1995 to November 16, 1995, including one patient who was treated on both eyes.  The delivered doses range from21.1 to 22.7 percent greater than the prescribed dose of 40 Gy (4000 rad). 

## Discussion

10 CFR 35.32, "Quality Management Program," requires licensees to develop written procedures to meet five objectives associated withany brachytherapy dose, including assurance that the prescribed dose is the administered dose.  A necessary part of this is to ensure thatthe dose rate emitted from an applicator is correct.  If the manufacturer's certificate of calibration or original activity/dose rate name plateis missing, the licensee should arrange with a qualified expert to determine the dose rate from the Sr-90 source.  (For additional details, seeNRC IN 94-17, "Sr-90 Eye Applicators:  Submission of Quality Management Plan, Calibration and Use," dated March 11, 1994.)  Failureto implement the Quality Management Rule and make required reports may subject a licensee to orders, civil penalties, and notices ofviolation.  

In view of the nature of the cited events, it appears worthwhile to review some of the properties of Sr-90 eye applicators and the relatedprocesses for decay correction over time.  New Sr-90 eye applicators typically contain a 2 gigabecquerel (GBq) [54 millicurie(mCi)]source, exhibiting a surface dose rate of about 0.50 Gy (50 rad)/sec.  The half-life of the parent Sr-90 is 28.5 yrs [maximum beta energyequal to 0.54 mega-electron volts (MeV)], and the yttrium-90 (Y-90) daughter half-life is 64.2 hrs (beta-max, 2.27 MeV); therefore, bothisotopes are in equilibrium on the eye applicator.  Since Sr-90 and Y-90 are in equilibrium, emissions from both isotopes must beaccounted for in dosimetry calculations.

The dose rate (D. t) at a time (t) can be calculated from the initial dose rate (D. o) at the time of the most recent calibration with thefollowing formula:

where e, the base of the natural logarithm (2.718), is raised to the power and is referred to as the "decay factor" (df),where is the half-life of the isotope, and t is the elapsed time.  For Sr-90,  years; therefore,.  The values for df can be used to determine the fraction of original activity remaining after t years. These values can then be used to calculate dose rates by using the following formula: 

The fraction of activity remaining after a given number of years from the original measurement date is given in Table 1 (Attachment 1).

For example

An eye applicator calibrated by the manufacturer on January 1, 1978, delivered an initial dose rate () of 0.75 Gy/sec (75 rad/sec) oncontact.  If, on January 1, 1996, the eye applicator needs to be used, the elapsed time (in years) since calibration is 18, and from Table 1, df= 0.646 for Sr-90.

Using formula F2:

Table 1 may be extended using formula F1 and the value for Sr-90.  Other values of df that need to be evaluated bylicensees will need to be re-calculated by using the appropriate half-life for that isotope.  Alternatively, a semi-logarithmic plot of theabove data will yield a straight line, which may be extended beyond 25 years.

This information notice requires no specific action nor written response.  If you have any questions about the information in this notice,please contact one of the technical contacts listed below or the appropriate regional office.

**Donald A. Cool, DirectorDivision of Industrial and   Medical Nuclear SafetyOffice of Nuclear Material Safety  and Safeguards**

## Attachment 1: Fraction of Original Sr-90 Activity Remaining after (t) Years

 TABLE 1. FRACTION (EXPRESSED AS DECIMAL) OF ORIGINAL SR-90 ACTIVITY REMAINING AFTER (t) YEARS