Name: Jon A Finlinson

Title: President & Chief Operations Officer

Company: Intermountain Power Service Corp

Sign Date: 8/31/2010

Name:

Title:

Company:

Sign Date:

Permitted Facility Representative / Date Testing Company Representative / Date

6481_INTERMOUNTAIN

Final Test Report 

for

/ /

Based on information and belief formed 

after reasonable inquiry, I certify that the 

statements and information in this test 

report are true, accurate, and complete.

I have reviewed all testing details and 

results in this test report and hereby 

certify that the test report is authentic 

and accurate.

Monday, February 28, 2011 Page 1 of 33
6481_INTERMOUNTAIN

8/31/2010

Facility Information:

6481_INTERMOUNTAIN

850 West Brush Wellman Road

Delta UT 84624

Contact: Jon Christensen

Phone: (435) 864-4414

Fax: (435) 864-6670

Email: jon-c@ipsc.com

Testing Company:

Applied Environmental Consultants, a JBR company

1553 West Elna Rae

Tempe AZ 85281

Contact:Jason Nockleby

Phone: (480) 829-0457

Fax: (480) 829-8985

Email: Jnockleby@jbrenv.com

Air Permit Number:

2700010003

Permitted Source ID/Name:

6481 INTERMOUNTAIN

Permitted Maximum Process Rate: Max. Normal Operation Process Rate: Target Process Test Rate

Description of the source (including control equipment). Please see the attachments for source or

 process flow diagram:

The Intermountain Power Project (IPP) consists of two 950 MWhe (gross) coal-fired thermal cycle steam-electric 

generating units. Each identical unit has a dry bottom boiler utilizing opposed wall low-NOX control burners with 

associated over-fire air for NOX reduction during combustion. Other pollution control devices (PCD) include fabric 

filter baghouses with reverse air cleaning and multi-module counterflow wet spray limestone flue gas desulfurization 

(FGD) scrubbers. 

Each IPP unit uses a continuous emissions monitoring system (CEMS) certified under both Part 75 for the Acid Rain 

Markets Program, and Part 60 New Source Performance Standard (NSPS) to monitor CO2, NOX, and SO2. Each 

unit also utilizes an uncertified mercury CEMS that measures flue gas concentrations only, since a data acquisition 

and handling system (DAHS) has not yet been integrated due to the remand of the Clean Air Mercury Rule (CAMR).

Industry/SCC/NAIS 221112

Sampling Location Information:

Test Parameter Information:

AFS #: 4902700010 FRS #: 110000470053

SCC / Description 10100202 External Combustion Boilers - Electric Generation - Bituminous/Subbituminous 

Coal - Pulverized Coal: Dry Bottom (Bituminous Coal) 

The following state and federal regulations that apply to the proposed testing:

Location

Round

Duct

Diam.

Distance from 

upstream dist.

Distance from

downstream dist.

Number of

Traverse Ports

Equiv. 

Diam

Min.Travers 

Points

Rect. Duct 

Length /Width

2SGA 336 0 0 336 2880 4200 4 12

Monday, February 28, 2011 Page 2 of 33
6481_INTERMOUNTAIN

8/31/2010

The following describes any modifications and/or deviations to the applicable test 

methods. If alternative methods were requested, see the attachments for documentation 

of request AND approval, including dates.

Monday, February 28, 2011 Page 3 of 33
6481_INTERMOUNTAIN

8/31/2010

Sampling / Stack Data Results Summary

Location 2SGA - EPA Method 316

Average

Run Number M316-1 M316-2 M316-3

Test Date 6/16/2010 6/16/2010 6/16/2010

Run Start Time 8:37:00 AM 11:14:00 

AM

1:43:00 PM

Run Finish Time 10:27:00 AM 1:04:00 PM 3:33:00 PM

Net Run Time, minutes 120 120 120

Dry Gas Meter Volume Sampled, dscf 67.104 69.337 69.616 68.686

Moisture Content of Stack Gas, % 12.88 13.35 13.59 13.273

Moisture Saturation at Stack Gas Temperature, % 13.84 13.93 14.22 13.997

Carbon Dioxide, % 13.89 13.89 13.83 13.870

Oxygen, % 5.59 5.6 5.65 5.613

Average Stack Gas Temperature, °F 119.92 120.17 120.92 120.337

Dry Volumetric Flow Rate, dry scfm 1,970,550.5 2,010,070.0 2,015,875.4 1,998,831.96

7

Actual Wet Volumetric Flue Gas Flow Rate, acfm 2,994,771.4 3,072,725.6 3,094,153.8 3,053,883.60

0

Percent Isokinetic of Sampling Rate, % 98.9 100.2 100.3 99.800

F-Factor, dscfm/MMBtu @ %O2 9647 9652 9695 9,664.667

Fw 10566 10532 10559 10,552.333

Fc 0 0 0 0.000

Monday, February 28, 2011 Page 4 of 33
6481_INTERMOUNTAIN

8/31/2010

Location 2SGA - Method 1 - 4

Average

Run Number 1

Test Date 6/16/2010

Run Start Time 4:03:00 PM

Run Finish Time 5:53:00 PM

Net Run Time, minutes 120

Dry Gas Meter Volume Sampled, dscf 62.870 62.870

Moisture Content of Stack Gas, % 12.57 12.570

Moisture Saturation at Stack Gas Temperature, % 14.78 14.780

Carbon Dioxide, % 13.22 13.220

Oxygen, % 6.11 6.110

Average Stack Gas Temperature, °F 122.00 122.000

Dry Volumetric Flow Rate, dry scfm 2,201,846.7 2,201,846.70

0

Actual Wet Volumetric Flue Gas Flow Rate, acfm 3,374,936.7 3,374,936.70

0

Percent Isokinetic of Sampling Rate, %

F-Factor, dscfm/MMBtu @ %O2 9763 9,763.000

Fw 10654 10,654.000

Fc 0 0.000

Location 2SGA - Method 10

Average

Run Number 1 2 3

Test Date 6/16/2010 6/16/2010 6/16/2010

Run Start Time 8:32:00 AM 11:13:00 

AM

1:40:00 PM

Run Finish Time 10:31:00 AM 1:12:00 PM 3:39:00 PM

Carbon Dioxide, % 13.42 13.22 13.11 13.250

Oxygen, % 5.91 6.07 6.2 6.060

Dry Volumetric Flow Rate, dry scfm 2263815 231951 2331351 1,609,039.00

0

F-Factor, dscfm/MMBtu @ %O2 9647 9652 9695 9,664.667

Moisture, % 12.96 13.43 13.68 13.357

Fuel Type Override Override Override

Fw 10566 10532 10559 10,552.333

Fc 0 0 0 0.000

Monday, February 28, 2011 Page 5 of 33
6481_INTERMOUNTAIN

8/31/2010

Location 2SGA - Method 10 3

Average

Run Number 4

Test Date 6/16/2010

Run Start Time 4:00:00 PM

Run Finish Time 5:59:00 PM

Carbon Dioxide, % 13.22 13.220

Oxygen, % 6.11 6.110

Dry Volumetric Flow Rate, dry scfm 2517279 2,517,279.00

0

F-Factor, dscfm/MMBtu @ %O2 9763 9,763.000

Moisture, % 12.66 12.660

Fuel Type Override

Fw 10654 10,654.000

Fc 0 0.000

Location 2SGA - Method 25A

Average

Run Number 1 2 3

Test Date 6/16/2010 6/16/2010 6/16/2010

Run Start Time 8:32:00 AM 11:13:00 

AM

1:40:00 PM

Run Finish Time 10:31:00 AM 1:12:00 PM 3:39:00 PM

Carbon Dioxide, % 13.42 13.22 13.11 13.250

Oxygen, % 5.91 6.07 6.2 6.060

Dry Volumetric Flow Rate, dry scfm 2263815 2319571 2331351 2,304,912.33

3

F-Factor, dscfm/MMBtu @ %O2 9647 9652 9695 9,664.667

Moisture, % 12.96 13.43 13.68 13.357

Fuel Type Override Override Override

Fw 10566 10532 10559 10,552.333

Fc 0 0 0 0.000

Monday, February 28, 2011 Page 6 of 33
6481_INTERMOUNTAIN

8/31/2010

Location 2SGA - Method 3A CO2

Average

Run Number 1 2 3

Test Date 6/16/2010 6/16/2010 6/16/2010

Run Start Time 8:32:00 AM 11:13:00 

AM

1:40:00 PM

Run Finish Time 10:31:00 AM 1:12:00 PM 3:39:00 PM

Carbon Dioxide, % 13.42 13.22 13.11 13.250

Oxygen, % 5.91 6.07 6.2 6.060

Dry Volumetric Flow Rate, dry scfm 2263815 2319571 2331351 2,304,912.33

3

F-Factor, dscfm/MMBtu @ %O2 9647 9652 9695 9,664.667

Moisture, % 12.96 13.43 13.68 13.357

Fuel Type Override Override Override

Fw 10566 10532 10559 10,552.333

Fc 0 0 0 0.000

Location 2SGA - Method 3A CO2 3

Average

Run Number 4

Test Date 6/16/2010

Run Start Time 4:00:00 PM

Run Finish Time 5:59:00 PM

Carbon Dioxide, % 13.22 13.220

Oxygen, % 6.11 6.110

Dry Volumetric Flow Rate, dry scfm 2517279 2,517,279.00

0

F-Factor, dscfm/MMBtu @ %O2 9763 9,763.000

Moisture, % 12.66 12.660

Fuel Type Override

Fw 10654 10,654.000

Fc 0 0.000

Monday, February 28, 2011 Page 7 of 33
6481_INTERMOUNTAIN

8/31/2010

Location 2SGA - Method 3A O2

Average

Run Number 1 2 3

Test Date 6/15/2010 6/16/2010 6/16/2010

Run Start Time 8:32:00 AM 11:13:00 

AM

1:40:00 PM

Run Finish Time 10:31:00 AM 1:12:00 PM 3:39:00 PM

Carbon Dioxide, % 13.42 13.22 13.11 13.250

Oxygen, % 5.91 6.07 6.2 6.060

Dry Volumetric Flow Rate, dry scfm 2263815 2319571 2331351 2,304,912.33

3

F-Factor, dscfm/MMBtu @ %O2 9647 9652 9695 9,664.667

Moisture, % 12.96 13.43 13.68 13.357

Fuel Type Override Override Override

Fw 10566 10532 10559 10,552.333

Fc 0 0 0 0.000

Location 2SGA - Method 3A O2 3

Average

Run Number 4

Test Date 6/16/2010

Run Start Time 4:00:00 PM

Run Finish Time 5:59:00 PM

Carbon Dioxide, % 13.22 13.220

Oxygen, % 6.11 6.110

Dry Volumetric Flow Rate, dry scfm 2517279 2,517,279.00

0

F-Factor, dscfm/MMBtu @ %O2 9763 9,763.000

Moisture, % 12.66 12.660

Fuel Type Override

Fw 10654 10,654.000

Fc 0 0.000

Monday, February 28, 2011 Page 8 of 33
6481_INTERMOUNTAIN

8/31/2010

Emissions Summary

Location: Stack Platform - Method 0011

Compound: Formaldehyde

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.0153 0.0109 0.0076 0.011

lb/mmBtuO2 6.42E-06 4.59E-06 3.08E-06 0.000

ppm 5.36E-03 3.85E-03 2.59E-03 0.004

ppm@7%O2 5.96E-03 4.26E-03 2.86E-03 0.004

Location: Stack Platform - Method 10

Compound: Carbon Monoxide

Average

Run PJ4 R1 PJ4 R2 PJ4 R3

Elb/hr 2.56E+00 4.43E+00 3.86E+00 3.617

lb/mmBtuO2 4.29E-04 7.55E-04 6.60E-04 0.001

mg/dscm 4.69E-01 8.16E-01 7.03E-01 0.663

Location: Stack Platform - Method 23

Compound: 1,2,3,4,6,7,8-HpCDF

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.00000000501 0.00000000531 0.00000000503 0.000

lb/mmBtuO2 5.21E-13 5.57E-13 5.38E-13 0.000

ng/dscm 5.70E-04 6.02E-04 5.73E-04 0.001

ng/dscm@7%O2 6.05E-04 6.46E-04 6.24E-04 0.001

Compound: 1,2,3,7,8-PeCDD

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.00000000189 0.00000000179 0.00000000176 0.000

lb/mmBtuO2 1.97E-13 1.88E-13 1.88E-13 0.000

ng/dscm 2.15E-04 2.03E-04 2.00E-04 0.000

ng/dscm@7%O2 2.28E-04 2.18E-04 2.18E-04 0.000

Compound: 1,2,3,7,8-PeCDF

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.00000000167 0.00000000177 0.00000000146 0.000

lb/mmBtuO2 1.74E-13 1.86E-13 1.56E-13 0.000

ng/dscm 1.90E-04 2.01E-04 1.66E-04 0.000

ng/dscm@7%O2 2.02E-04 2.16E-04 1.81E-04 0.000

Compound: 2,3,3',4,4',5,5'-HpCB (PCB189)

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.00000000364 0.00000000619 0.000000006 0.000

lb/mmBtuO2 3.79E-13 6.49E-13 6.41E-13 0.000

ng/dscm 4.14E-04 7.02E-04 6.83E-04 0.001

ng/dscm@7%O2 4.39E-04 7.53E-04 7.43E-04 0.001

Monday, February 28, 2011 Page 9 of 33
6481_INTERMOUNTAIN

8/31/2010

Compound: 2,3,3',4,4',5/2,3,3',4,4',5'-HxCB (PC

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.0000001145 0.000000192 0.000000242 0.000

lb/mmBtuO2 1.19E-11 2.01E-11 2.59E-11 0.000

ng/dscm 1.30E-02 2.18E-02 2.76E-02 0.021

ng/dscm@7%O2 1.38E-02 2.34E-02 3.00E-02 0.022

Compound: 2,3,3',4,4'-PeCB (PCB105)

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.000000447 0.000000728 0.000000895 0.000

lb/mmBtuO2 4.65E-11 7.63E-11 9.57E-11 0.000

ng/dscm 5.08E-02 8.25E-02 1.02E-01 0.078

ng/dscm@7%O2 5.39E-02 8.85E-02 1.11E-01 0.084

Compound: 2,3',4,4',5,5'-HxCB (PCB167)

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.0000000421 0.0000000787 0.0000000943 0.000

lb/mmBtuO2 4.38E-12 8.25E-12 1.01E-11 0.000

ng/dscm 4.79E-03 8.92E-03 1.07E-02 0.008

ng/dscm@7%O2 5.08E-03 9.57E-03 1.16E-02 0.009

Compound: 2,3,4,4',5-PeCB (PCB114)

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.0000000282 0.0000000526 0.0000000695 0.000

lb/mmBtuO2 2.93E-12 5.52E-12 7.43E-12 0.000

ng/dscm 3.21E-03 5.96E-03 7.91E-03 0.006

ng/dscm@7%O2 3.41E-03 6.39E-03 8.61E-03 0.006

Compound: 2,3',4,4',5-PeCB (PCB118)

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.0000014 0.00000235 0.00000278 0.000

lb/mmBtuO2 1.46E-10 2.46E-10 2.97E-10 0.000

ng/dscm 1.59E-01 2.66E-01 3.17E-01 0.247

ng/dscm@7%O2 1.69E-01 2.85E-01 3.45E-01 0.266

Compound: 2',3,4,4',5-PeCB (PCB123)

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.0000000277 0.0000000543 0.0000000652 0.000

lb/mmBtuO2 2.88E-12 5.69E-12 6.97E-12 0.000

ng/dscm 3.15E-03 6.16E-03 7.42E-03 0.006

ng/dscm@7%O2 3.34E-03 6.61E-03 8.08E-03 0.006

Compound: 2,3,4,7,8-PeCDF

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.00000000161 0.00000000171 0.00000000143 0.000

lb/mmBtuO2 1.68E-13 1.79E-13 1.53E-13 0.000

Monday, February 28, 2011 Page 10 of 33
6481_INTERMOUNTAIN

8/31/2010

ng/dscm 1.83E-04 1.94E-04 1.63E-04 0.000

ng/dscm@7%O2 1.94E-04 2.08E-04 1.77E-04 0.000

Compound: 2,3,7,8-TCDD

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.00000000125 0.00000000121 0.000000000944 0.000

lb/mmBtuO2 1.30E-13 1.27E-13 1.01E-13 0.000

ng/dscm 1.42E-04 1.37E-04 1.07E-04 0.000

ng/dscm@7%O2 1.51E-04 1.47E-04 1.16E-04 0.000

Compound: 2,3,7,8-TCDF

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.00000000105 0.00000000106 0.000000000977 0.000

lb/mmBtuO2 1.09E-13 1.11E-13 1.04E-13 0.000

ng/dscm 1.19E-04 1.20E-04 1.11E-04 0.000

ng/dscm@7%O2 1.26E-04 1.29E-04 1.21E-04 0.000

Compound: 3,3',4,4',5,5'-HxCB (PCB169)

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.00000000723 0.00000000486 0.00000000626 0.000

lb/mmBtuO2 7.52E-13 5.10E-13 6.69E-13 0.000

ng/dscm 8.22E-04 5.51E-04 7.13E-04 0.001

ng/dscm@7%O2 8.72E-04 5.91E-04 7.76E-04 0.001

Compound: 3,3',4,4',5-PeCB (PCB126)

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.00000000578 0.000000011 0.0000000106 0.000

lb/mmBtuO2 6.02E-13 1.15E-12 1.13E-12 0.000

ng/dscm 6.57E-04 1.25E-03 1.21E-03 0.001

ng/dscm@7%O2 6.97E-04 1.34E-03 1.32E-03 0.001

Compound: 3,3',4,4'-TCB (PCB77)

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.0000000981 0.000000163 0.000000161 0.000

lb/mmBtuO2 1.02E-11 1.71E-11 1.72E-11 0.000

ng/dscm 1.12E-02 1.85E-02 1.83E-02 0.016

ng/dscm@7%O2 1.19E-02 1.98E-02 1.99E-02 0.017

Compound: 3,4,4',5-TCB (PCB81)

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.00000000688 0.0000000145 0.0000000119 0.000

lb/mmBtuO2 7.16E-13 1.52E-12 1.27E-12 0.000

ng/dscm 7.82E-04 1.64E-03 1.36E-03 0.001

ng/dscm@7%O2 8.30E-04 1.76E-03 1.48E-03 0.001

Compound: OCDD

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Monday, February 28, 2011 Page 11 of 33
6481_INTERMOUNTAIN

8/31/2010

Mass_mg 0.00000847 0.00000000577 0.00000000556 0.000

lb/mmBtuO2 8.81E-10 6.05E-13 5.94E-13 0.000

ng/dscm 9.63E-01 6.54E-04 6.33E-04 0.321

ng/dscm@7%O2 1.02E+00 7.01E-04 6.89E-04 0.340

Compound: OCDF

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.00000522 0.00000000697 0.00000000573 0.000

lb/mmBtuO2 5.43E-10 7.31E-13 6.13E-13 0.000

ng/dscm 5.94E-01 7.90E-04 6.52E-04 0.198

ng/dscm@7%O2 6.30E-01 8.47E-04 7.10E-04 0.211

Compound: Other HpCDD

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0 0 0 0.000

lb/mmBtuO2 0.00E+00 0.00E+00 0.00E+00 0.000

ng/dscm 0.00E+00 0.00E+00 0.00E+00 0.000

ng/dscm@7%O2 0.00E+00 0.00E+00 0.00E+00 0.000

Compound: Other HpCDF

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0 0 0 0.000

lb/mmBtuO2 0.00E+00 0.00E+00 0.00E+00 0.000

ng/dscm 0.00E+00 0.00E+00 0.00E+00 0.000

ng/dscm@7%O2 0.00E+00 0.00E+00 0.00E+00 0.000

Compound: Other HxCDD

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0 0.00000000778 0 0.000

lb/mmBtuO2 0.00E+00 8.16E-13 0.00E+00 0.000

ng/dscm 0.00E+00 8.82E-04 0.00E+00 0.000

ng/dscm@7%O2 0.00E+00 9.46E-04 0.00E+00 0.000

Compound: Other HxCDF

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0 0.00000000237 0.00000000151 0.000

lb/mmBtuO2 0.00E+00 2.49E-13 1.61E-13 0.000

ng/dscm 0.00E+00 2.69E-04 1.72E-04 0.000

ng/dscm@7%O2 0.00E+00 2.89E-04 1.87E-04 0.000

Compound: Other PeCDD

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0 0.00000000545 0 0.000

lb/mmBtuO2 0.00E+00 5.72E-13 0.00E+00 0.000

ng/dscm 0.00E+00 6.18E-04 0.00E+00 0.000

ng/dscm@7%O2 0.00E+00 6.63E-04 0.00E+00 0.000

Monday, February 28, 2011 Page 12 of 33
6481_INTERMOUNTAIN

8/31/2010

Compound: Other PeCDF

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0 0 0 0.000

lb/mmBtuO2 0.00E+00 0.00E+00 0.00E+00 0.000

ng/dscm 0.00E+00 0.00E+00 0.00E+00 0.000

ng/dscm@7%O2 0.00E+00 0.00E+00 0.00E+00 0.000

Compound: Other TCDD

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0 0.00000000462 0.00000000231 0.000

lb/mmBtuO2 0.00E+00 4.84E-13 2.47E-13 0.000

ng/dscm 0.00E+00 5.24E-04 2.63E-04 0.000

ng/dscm@7%O2 0.00E+00 5.62E-04 2.86E-04 0.000

Compound: Other TCDF

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.00000000284 0.00000000357 0.00000000227 0.000

lb/mmBtuO2 2.96E-13 3.74E-13 2.43E-13 0.000

ng/dscm 3.23E-04 4.05E-04 2.58E-04 0.000

ng/dscm@7%O2 3.43E-04 4.34E-04 2.81E-04 0.000

Location: Stack Platform - Method 25A

Compound: Total organic compounds (TOC) as 

Average

Run PJ4 R1 PJ4 R2 PJ4 R3

Elb/hr 3.02E+00 7.08E+00 2.03E+00 4.043

gr/dscf 2.40E-04 5.61E-04 1.60E-04 0.000

lb/mmBtuO2 5.28E-04 1.23E-03 3.49E-04 0.001

mg/dscm 5.50E-01 1.28E+00 3.67E-01 0.732

ppm 3.00E-01 7.00E-01 2.00E-01 0.400

ppm@7%O2 3.34E-01 7.75E-01 2.21E-01 0.443

Location: Stack Platform - Method 26A

Compound: Hydrogen Chloride

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.97 0.99 0.9 0.953

Elb/hr 1.55E+00 1.58E+00 1.44E+00 1.523

lb/mmBtuO2 2.50E-04 2.55E-04 2.34E-04 0.000

mg/dscm 2.71E-01 2.80E-01 2.57E-01 0.269

mg/dscm@7%O2 2.90E-01 2.96E-01 2.71E-01 0.286

Compound: Hydrogen Fluoride

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.4 0.4 0.42 0.407

Elb/hr 6.39E-01 6.40E-01 6.72E-01 0.650

lb/mmBtuO2 1.03E-04 1.03E-04 1.09E-04 0.000

mg/dscm 1.12E-01 1.13E-01 1.20E-01 0.115

Monday, February 28, 2011 Page 13 of 33
6481_INTERMOUNTAIN

8/31/2010

mg/dscm@7%O2 1.20E-01 1.19E-01 1.27E-01 0.122

Location: Stack Platform - Method 29 

Compound: Antimony

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 1.184247 1.229485 1.20238 1.205

Elb/hr 9.43E-01 9.73E-01 9.47E-01 0.954

lb/mmBtuO2 1.60E-04 1.68E-04 1.62E-04 0.000

mg/dscm 1.73E-01 1.79E-01 1.72E-01 0.175

mg/dscm@7%O2 1.86E-01 1.95E-01 1.88E-01 0.190

Compound: Arsenic

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.02022 0.01888 0.01881 0.019

Elb/hr 1.61E-02 1.49E-02 1.48E-02 0.015

lb/mmBtuO2 2.74E-06 2.58E-06 2.53E-06 0.000

mg/dscm 2.95E-03 2.76E-03 2.69E-03 0.003

mg/dscm@7%O2 3.17E-03 3.00E-03 2.93E-03 0.003

Compound: Beryllium

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.00172 0.00119 0.00151 0.001

Elb/hr 1.37E-03 9.42E-04 1.19E-03 0.001

lb/mmBtuO2 2.33E-07 1.63E-07 2.03E-07 0.000

mg/dscm 2.51E-04 1.74E-04 2.16E-04 0.000

mg/dscm@7%O2 2.70E-04 1.89E-04 2.35E-04 0.000

Compound: Cadmium

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.001529 0.00125 0.001524 0.001

Elb/hr 1.22E-03 9.89E-04 1.20E-03 0.001

lb/mmBtuO2 2.07E-07 1.71E-07 2.05E-07 0.000

mg/dscm 2.23E-04 1.82E-04 2.18E-04 0.000

mg/dscm@7%O2 2.40E-04 1.98E-04 2.38E-04 0.000

Compound: Chromium

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.06523 0.05429 0.038052 0.053

Elb/hr 5.19E-02 4.30E-02 3.00E-02 0.042

lb/mmBtuO2 8.83E-06 7.43E-06 5.12E-06 0.000

mg/dscm 9.53E-03 7.92E-03 5.44E-03 0.008

mg/dscm@7%O2 1.02E-02 8.61E-03 5.93E-03 0.008

Compound: Filterable Particulate

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 241.5 241.9 226.3 236.567

Elb/hr 1.92E+02 1.91E+02 1.78E+02 187.000

Monday, February 28, 2011 Page 14 of 33
6481_INTERMOUNTAIN

8/31/2010

lb/mmBtuO2 3.27E-02 3.31E-02 3.04E-02 0.032

mg/dscm 3.53E+01 3.53E+01 3.24E+01 34.333

mg/dscm@7%O2 3.79E+01 3.84E+01 3.53E+01 37.200

Compound: Lead

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.029889 0.02882 0.029033 0.029

Elb/hr 2.38E-02 2.28E-02 2.29E-02 0.023

lb/mmBtuO2 4.05E-06 3.94E-06 3.90E-06 0.000

mg/dscm 4.37E-03 4.21E-03 4.15E-03 0.004

mg/dscm@7%O2 4.69E-03 4.58E-03 4.52E-03 0.005

Compound: Manganese

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.319968 0.29902 0.293857 0.304

Elb/hr 2.55E-01 2.37E-01 2.32E-01 0.241

lb/mmBtuO2 4.33E-05 4.09E-05 3.95E-05 0.000

mg/dscm 4.68E-02 4.36E-02 4.20E-02 0.044

mg/dscm@7%O2 5.03E-02 4.74E-02 4.58E-02 0.048

Compound: Nickel

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 4.30507 3.96703 4.16365 4.145

Elb/hr 3.43E+00 3.14E+00 3.28E+00 3.283

lb/mmBtuO2 5.83E-04 5.43E-04 5.60E-04 0.001

mg/dscm 6.29E-01 5.79E-01 5.95E-01 0.601

mg/dscm@7%O2 6.76E-01 6.29E-01 6.49E-01 0.651

Compound: Selenium

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 0.03101 0.03352 0.0344 0.033

Elb/hr 2.47E-02 2.65E-02 2.71E-02 0.026

lb/mmBtuO2 4.20E-06 4.59E-06 4.63E-06 0.000

mg/dscm 4.53E-03 4.89E-03 4.92E-03 0.005

mg/dscm@7%O2 4.87E-03 5.31E-03 5.36E-03 0.005

Location: Stack Platform - Method 6C

Compound: Sulfur Dioxide

Average

Run PJ4 R1 PJ4 R2 PJ4 R3

Elb/hr 3.93E+03 4.08E+03 4.09E+03 4,033.333

lb/mmBtuO2 6.34E-01 6.60E-01 6.66E-01 0.653

mg/dscm 6.87E+02 7.24E+02 7.31E+02 714.000

mg/dscm@7%O2 7.35E+02 7.65E+02 7.72E+02 757.333

ppm 2.58E+02 2.72E+02 2.75E+02 268.333

ppm@7%O2 2.76E+02 2.88E+02 2.90E+02 284.667

Location: Stack Platform - Method 7E

Monday, February 28, 2011 Page 15 of 33
6481_INTERMOUNTAIN

8/31/2010

Process Run Data

Compound: Nitrogen oxides (NOx)

Average

Run PJ4 R1 PJ4 R2 PJ4 R3

Elb/hr 1.45E+03 1.57E+03 1.42E+03 1,480.000

lb/mmBtuO2 2.34E-01 2.53E-01 2.32E-01 0.240

mg/dscm 2.54E+02 2.78E+02 2.55E+02 262.333

mg/dscm@7%O2 2.72E+02 2.94E+02 2.69E+02 278.333

ppm 1.33E+02 1.45E+02 1.33E+02 137.000

ppm@7%O2 1.42E+02 1.53E+02 1.40E+02 145.000

Location: Stack Platform - OTM - 27/28

Compound: Filterable PM2.5

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 29.6 35.4 39.9 34.967

lb/mmBtuO2 6.73E-03 8.04E-03 9.11E-03 0.008

Compound: Inorganic (Aqueous) Condensible Pa

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 110.9 66 104.6 93.833

lb/mmBtuO2 2.52E-02 1.50E-02 2.39E-02 0.021

Compound: Organic Condensible Particulate

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 2 1.8 0.7 1.500

lb/mmBtuO2 4.55E-04 4.09E-04 1.60E-04 0.000

Compound: Total PM2.5

Average

RunNumber PJ4 R1 PJ4 R2 PJ4 R3

Mass_mg 142.5 103.2 145.2 130.300

lb/mmBtuO2 3.24E-02 2.35E-02 3.32E-02 0.030

Monday, February 28, 2011 Page 16 of 33
6481_INTERMOUNTAIN

8/31/2010

Process Run Data

Run Name Value UOM Comments

Target

Value

1 Bituminous Coal Burned 8552.5 1000 Tons/hr 0

2 Bituminous Coal Burned 8552.5 1000 Tons/hr 0

3 Bituminous Coal Burned 8514.8 1000 Tons/hr 0

4 Bituminous Coal Burned 8471.5 1000 Tons/hr 0

2 Coal Feed Rate - 1SGA 381.54 ton/hr 0

3 Coal Feed Rate - 1SGA 383.69 ton/hr 0

4 Coal Feed Rate - 1SGA 381.27 ton/hr 0

2 Steam Pressure - 1SGA 2384.1 psi 0

3 Steam Pressure - 1SGA 2383.7 psi 0

4 Steam Pressure - 1SGA 2383.3 psi 0

2 Gross Electric Output - 1SGA 950.01 MWh 0

3 Gross Electric Output - 1SGA 949.99 MWh 0

4 Gross Electric Output - 1SGA 949.9 MWh 0

2 Oxygen (O2) - 1SGA 3.3 % 0

3 Oxygen (O2) - 1SGA 3.45 % 0

4 Oxygen (O2) - 1SGA 3.16 % 0

2 Fuel Oil Feed Rate - 1SGA 0.03 klbh 0

3 Fuel Oil Feed Rate - 1SGA 0.03 klbh 0

4 Fuel Oil Feed Rate - 1SGA 0.03 klbh 0

2 Steam Flow Rate - 1SGA 6684.5 klbh 0

3 Steam Flow Rate - 1SGA 6701.9 klbh 0

4 Steam Flow Rate - 1SGA 6706 klbh 0

2 Steam Temperature - 1SGA 1002.5 F 0

3 Steam Temperature - 1SGA 1000.3 F 0

4 Steam Temperature - 1SGA 1001.6 F 0

1 Total Heat Input - 2SGA 8479.5 MMBtu/hr 0

2 Total Heat Input - 2SGA 8428.7 MMBtu/hr 0

3 Total Heat Input - 2SGA 8467.0 MMBtu/hr 0

4 Total Heat Input - 2SGA 8474.5 MMBtu/hr 0

1 Steam Temperature - 2SGA 998.33 F 0

2 Steam Temperature - 2SGA 998.33 F 0

3 Steam Temperature - 2SGA 999.13 F 0

4 Steam Temperature - 2SGA 998.64 F 0

Monday, February 28, 2011 Page 17 of 33
6481_INTERMOUNTAIN

8/31/2010

APCD Run Data

1 Steam Flow Rate - 2SGA 6485.1 klbh 0

2 Steam Flow Rate - 2SGA 6503.1 klbh 0

3 Steam Flow Rate - 2SGA 6524.6 klbh 0

4 Steam Flow Rate - 2SGA 6513.5 klbh 0

1 Fuel Oil Feed Rate - 2SGA 0 klbh 0

2 Fuel Oil Feed Rate - 2SGA 0 klbh 0

3 Fuel Oil Feed Rate - 2SGA 0 klbh 0

4 Fuel Oil Feed Rate - 2SGA 0 klbh 0

1 Oxygen (O2) - 2SGA 3.24 % 0

2 Oxygen (O2) - 2SGA 3.29 % 0

3 Oxygen (O2) - 2SGA 3.5 % 0

4 Oxygen (O2) - 2SGA 3.45 % 0

1 Gross Electric Output - 2SGA 949.91 MWh 0

2 Gross Electric Output - 2SGA 949.97 MWh 0

3 Gross Electric Output - 2SGA 950.08 MWh 0

4 Gross Electric Output - 2SGA 949.95 MWh 0

1 Steam Pressure - 2SGA 2358.7 psi 0

2 Steam Pressure - 2SGA 2357.7 psi 0

3 Steam Pressure - 2SGA 2356.5 psi 0

4 Steam Pressure - 2SGA 2358.4 psi 0

1 Coal Feed Rate -2SGA 384.29 ton/hr 0

2 Coal Feed Rate -2SGA 381.96 ton/hr 0

3 Coal Feed Rate -2SGA 383.69 ton/hr 0

4 Coal Feed Rate -2SGA 384.04 ton/hr 0

2 Net Electric Output - 1SGA 895.38 MWh 0

3 Net Electric Output - 1SGA 896.29 MWh 0

4 Net Electric Output - 1SGA 895.86 MWh 0

1 Net Electric Output - 2SGA 890.36 MWh 0

2 Net Electric Output - 2SGA 888.77 MWh 0

3 Net Electric Output - 2SGA 888.59 MWh 0

4 Net Electric Output - 2SGA 889.5 MWh 0

Run Name Value UOM Comments

Target

Value

1 Scrubber Differential Pressure - Module F - 2SGA 1.72 inWC 0

2 Scrubber Differential Pressure - Module F - 2SGA 1.76 inWC 0

Monday, February 28, 2011 Page 18 of 33
6481_INTERMOUNTAIN

8/31/2010

3 Scrubber Differential Pressure - Module F - 2SGA 1.77 inWC 0

4 Scrubber Differential Pressure - Module F - 2SGA 1.76 inWC 0

1 Scrubber Differential Pressure - Module C - 2SGA 1.68 inWC 0

2 Scrubber Differential Pressure - Module C - 2SGA 1.7 inWC 0

3 Scrubber Differential Pressure - Module C - 2SGA 1.72 inWC 0

4 Scrubber Differential Pressure - Module C - 2SGA 1.72 inWC 0

1 Scrubber Differential Pressure - Module D - 2SGA 1.92 inWC 0

2 Scrubber Differential Pressure - Module D - 2SGA 1.94 inWC 0

3 Scrubber Differential Pressure - Module D - 2SGA 1.97 inWC 0

4 Scrubber Differential Pressure - Module D - 2SGA 1.97 inWC 0

1 Scrubber Differential Pressure - Module E - 2SGA 1.79 inWC 0

2 Scrubber Differential Pressure - Module E - 2SGA 1.81 inWC 0

3 Scrubber Differential Pressure - Module E - 2SGA 1.83 inWC 0

4 Scrubber Differential Pressure - Module E - 2SGA 1.82 inWC 0

1 Fabric Filter Differential Pressure - A Casing - 

2SGA

9.35 inWC 0

2 Fabric Filter Differential Pressure - A Casing - 

2SGA

9.45 inWC 0

3 Fabric Filter Differential Pressure - A Casing - 

2SGA

9.6 inWC 0

4 Fabric Filter Differential Pressure - A Casing - 

2SGA

9.33 inWC 0

1 Fabric Filter Differential Pressure - B Casing - 

2SGA

8.95 inWC 0

2 Fabric Filter Differential Pressure - B Casing - 

2SGA

9.04 inWC 0

3 Fabric Filter Differential Pressure - B Casing - 

2SGA

9.2 inWC 0

4 Fabric Filter Differential Pressure - B Casing - 

2SGA

8.91 inWC 0

1 Fabric Filter Differential Pressure - C Casing - 

2SGA

8.29 inWC 0

2 Fabric Filter Differential Pressure - C Casing - 

2SGA

8.36 inWC 0

3 Fabric Filter Differential Pressure - C Casing - 

2SGA

8.54 inWC 0

4 Fabric Filter Differential Pressure - C Casing - 

2SGA

8.3 inWC 0

1 Average Scrubber Differential Pressure - 2SGA 1.78 inWC 0

2 Average Scrubber Differential Pressure - 2SGA 1.8 inWC 0

3 Average Scrubber Differential Pressure - 2SGA 1.82 inWC 0

Monday, February 28, 2011 Page 19 of 33
6481_INTERMOUNTAIN

8/31/2010

4 Average Scrubber Differential Pressure - 2SGA 1.82 inWC 0

1 Average Fabric Filter Differential Pressure - 2SGA 8.86 inWC 0

2 Average Fabric Filter Differential Pressure - 2SGA 8.95 inWC 0

3 Average Fabric Filter Differential Pressure - 2SGA 9.11 inWC 0

4 Average Fabric Filter Differential Pressure - 2SGA 8.84 inWC 0

1 Scrubber pH Module C - 2SGA 5.9 pH 0

2 Scrubber pH Module C - 2SGA 5.9 pH 0

3 Scrubber pH Module C - 2SGA 5.9 pH 0

4 Scrubber pH Module C - 2SGA 5.9 pH 0

1 Scrubber pH Module E - 2SGA 5.88 pH 0

2 Scrubber pH Module E - 2SGA 5.88 pH 0

3 Scrubber pH Module E - 2SGA 5.89 pH 0

4 Scrubber pH Module E - 2SGA 5.86 pH 0

1 Scrubber pH Module F - 2SGA 5.92 pH 0

2 Scrubber pH Module F - 2SGA 5.92 pH 0

3 Scrubber pH Module F - 2SGA 5.89 pH 0

4 Scrubber pH Module F - 2SGA 5.86 pH 0

1 Average Scrubber pH - 2SGA 5.9 pH 0

2 Average Scrubber pH - 2SGA 5.9 pH 0

3 Average Scrubber pH - 2SGA 5.9 pH 0

4 Average Scrubber pH - 2SGA 5.88 pH 0

1 Scrubber Limestone Feed Rate - 2SGA 20986 lb/hr 0

2 Scrubber Limestone Feed Rate - 2SGA 20695 lb/hr 0

3 Scrubber Limestone Feed Rate - 2SGA 18916 lb/hr 0

4 Scrubber Limestone Feed Rate - 2SGA 16067 lb/hr 0

1 Scrubber Slurry Density - 2SGA 33.32 % 0

2 Scrubber Slurry Density - 2SGA 33.11 % 0

3 Scrubber Slurry Density - 2SGA 32.99 % 0

4 Scrubber Slurry Density - 2SGA 32.94 % 0

1 Total Spray Flow Rate - 2SGA 181142 gpm 0

2 Total Spray Flow Rate - 2SGA 181299 gpm 0

3 Total Spray Flow Rate - 2SGA 181469 gpm 0

4 Total Spray Flow Rate - 2SGA 181910 gpm 0

1 Scrubber Slurry Feed Rate Module C - 2SGA 28.54 gpm 0

2 Scrubber Slurry Feed Rate Module C - 2SGA 32.56 gpm 0

3 Scrubber Slurry Feed Rate Module C - 2SGA 26.34 gpm 0

Monday, February 28, 2011 Page 20 of 33
6481_INTERMOUNTAIN

8/31/2010

4 Scrubber Slurry Feed Rate Module C - 2SGA 24.07 gpm 0

1 Scrubber Slurry Feed Rate Module D - 2SGA 32.75 gpm 0

2 Scrubber Slurry Feed Rate Module D - 2SGA 29.17 gpm 0

3 Scrubber Slurry Feed Rate Module D - 2SGA 33.29 gpm 0

4 Scrubber Slurry Feed Rate Module D - 2SGA 28.94 gpm 0

1 Scrubber Slurry Feed Rate Module E - 2SGA 32.52 gpm 0

2 Scrubber Slurry Feed Rate Module E - 2SGA 33.44 gpm 0

3 Scrubber Slurry Feed Rate Module E - 2SGA 27.04 gpm 0

4 Scrubber Slurry Feed Rate Module E - 2SGA 22.18 gpm 0

1 Scrubber Slurry Feed Rate Module F - 2SGA 11.16 gpm 0

2 Scrubber Slurry Feed Rate Module F - 2SGA 9 gpm 0

3 Scrubber Slurry Feed Rate Module F - 2SGA 8.9 gpm 0

4 Scrubber Slurry Feed Rate Module F - 2SGA 6.09 gpm 0

1 C Module HP Spray Amps - 2SGA 25.37 ohms 0

2 C Module HP Spray Amps - 2SGA 25.49 ohms 0

3 C Module HP Spray Amps - 2SGA 25.66 ohms 0

4 C Module HP Spray Amps - 2SGA 25.88 ohms 0

1 C Module IP Spray Amps - 2SGA 26.38 ohms 0

2 C Module IP Spray Amps - 2SGA 26.5 ohms 0

3 C Module IP Spray Amps - 2SGA 26.52 ohms 0

4 C Module IP Spray Amps - 2SGA 26.55 ohms 0

1 C Module LP Spray Amps - 2SGA 22.26 ohms 0

2 C Module LP Spray Amps - 2SGA 22.27 ohms 0

3 C Module LP Spray Amps - 2SGA 22.38 ohms 0

4 C Module LP Spray Amps - 2SGA 22.52 ohms 0

1 D Module HP Spray Amps - 2SGA 26.09 ohms 0

2 D Module HP Spray Amps - 2SGA 25.99 ohms 0

3 D Module HP Spray Amps - 2SGA 25.93 ohms 0

4 D Module HP Spray Amps - 2SGA 26.02 ohms 0

1 D Module IP Spray Amps - 2SGA 24.62 ohms 0

2 D Module IP Spray Amps - 2SGA 24.82 ohms 0

3 D Module IP Spray Amps - 2SGA 24.88 ohms 0

4 D Module IP Spray Amps - 2SGA 24.81 ohms 0

1 D Module LP Spray Amps - 2SGA 23.16 ohms 0

2 D Module LP Spray Amps - 2SGA 23.15 ohms 0

3 D Module LP Spray Amps - 2SGA 23.16 ohms 0

Monday, February 28, 2011 Page 21 of 33
6481_INTERMOUNTAIN

8/31/2010

Process Lab Run Data

4 D Module LP Spray Amps - 2SGA 23.21 ohms 0

1 E Module HP Spray Amps - 2SGA 26.18 ohms 0

2 E Module HP Spray Amps - 2SGA 26.21 ohms 0

3 E Module HP Spray Amps - 2SGA 26.23 ohms 0

4 E Module HP Spray Amps - 2SGA 26.28 ohms 0

1 E Module IP Spray Amps - 2SGA 25.26 ohms 0

2 E Module IP Spray Amps - 2SGA 25.31 ohms 0

3 E Module IP Spray Amps - 2SGA 25.36 ohms 0

4 E Module IP Spray Amps - 2SGA 25.51 ohms 0

1 E Module LP Spray Amps - 2SGA 24.1 ohms 0

2 E Module LP Spray Amps - 2SGA 23.92 ohms 0

3 E Module LP Spray Amps - 2SGA 23.79 ohms 0

4 E Module LP Spray Amps - 2SGA 23.81 ohms 0

1 F Module HP Spray Amps - 2SGA 27.72 ohms 0

2 F Module HP Spray Amps - 2SGA 27.57 ohms 0

3 F Module HP Spray Amps - 2SGA 27.41 ohms 0

4 F Module HP Spray Amps - 2SGA 27.21 ohms 0

1 F Module IP Spray Amps - 2SGA 24.12 ohms 0

2 F Module IP Spray Amps - 2SGA 24.29 ohms 0

3 F Module IP Spray Amps - 2SGA 24.44 ohms 0

4 F Module IP Spray Amps - 2SGA 24.61 ohms 0

1 F Module LP Spray Amps - 2SGA 25.09 ohms 0

2 F Module LP Spray Amps - 2SGA 25.11 ohms 0

3 F Module LP Spray Amps - 2SGA 25.13 ohms 0

4 F Module LP Spray Amps - 2SGA 25.21 ohms 0

1 Scrubber pH Module D - 2SGA 5.9 pH 0

2 Scrubber pH Module D - 2SGA 5.91 pH 0

3 Scrubber pH Module D - 2SGA 5.91 pH 0

4 Scrubber pH Module D - 2SGA 5.88 pH 0

Run Name Value UOM Comments

2 Formaldehyde (Gas Stream) - 1SGA 9.39 ug

3 Formaldehyde (Gas Stream) - 1SGA 9.1 ug

4 Formaldehyde (Gas Stream) - 1SGA 9.36 ug

2 Higher Heating Value (Fuel) - 1SGA 11216 Btu/lb

Monday, February 28, 2011 Page 22 of 33
6481_INTERMOUNTAIN

8/31/2010

3 Higher Heating Value (Fuel) - 1SGA 11085 Btu/lb

4 Higher Heating Value (Fuel) - 1SGA 11231 Btu/lb

2 Moisture (Fuel) - 1SGA 8.72 %

3 Moisture (Fuel) - 1SGA 8.34 %

4 Moisture (Fuel) - 1SGA 7.44 %

2 Residule Moisture (Fuel) - 1SGA

Moisture (Fuel) - 1SGA

2.02 %

3 Residule Moisture (Fuel) - 1SGA

Moisture (Fuel) - 1SGA

2 %

4 Residule Moisture (Fuel) - 1SGA

Moisture (Fuel) - 1SGA

2.24 %

2 Ash Proximate (Fuel) - 1SGA 12.82 %

3 Ash Proximate (Fuel) - 1SGA 13.84 %

4 Ash Proximate (Fuel) - 1SGA 13.84 %

2 Volatiles Proximate (Fuel) - 1SGA 37.52 %

3 Volatiles Proximate (Fuel) - 1SGA 36.76 %

4 Volatiles Proximate (Fuel) - 1SGA 37.65 %

2 Fixed Carbon (Fuel) - 1SGA 44.85 %

3 Fixed Carbon (Fuel) - 1SGA 44.8 %

4 Fixed Carbon (Fuel) - 1SGA 44.37 %

2 Carbon Ultimate (Fuel) - 1SGA 68.81 %

3 Carbon Ultimate (Fuel) - 1SGA 67.91 %

4 Carbon Ultimate (Fuel) - 1SGA 67.77 %

2 Hydrogen Ultimate (Fuel) - 1SGA 4.93 %

3 Hydrogen Ultimate (Fuel) - 1SGA 4.85 %

4 Hydrogen Ultimate (Fuel) - 1SGA 4.83 %

2 Nitrogent Ultimate (Fuel) - 1SGA 1.48 %

3 Nitrogent Ultimate (Fuel) - 1SGA 1.44 %

4 Nitrogent Ultimate (Fuel) - 1SGA 1.44 %

2 Methane Content (Gas Stream) - 1SGA 2.3 ppm

3 Methane Content (Gas Stream) - 1SGA 2.2 ppm

4 Methane Content (Gas Stream) - 1SGA 2.2 ppm

1 Formaldehyde (Gas Stream) - 2SGA 9.8 ug

2 Formaldehyde (Gas Stream) - 2SGA 9.44 ug

3 Formaldehyde (Gas Stream) - 2SGA 23.1 ug

4 Formaldehyde (Gas Stream) - 2SGA 0 ug

1 Higher Heating Value (Fuel) - 2SGA 11196 Btu/lb

2 Higher Heating Value (Fuel) - 2SGA 11397 Btu/lb

3 Higher Heating Value (Fuel) - 2SGA 11514 Btu/lb

4 Higher Heating Value (Fuel) - 2SGA 11592 Btu/lb

1 Moisture (Fuel) - 2SGA 8 %

2 Moisture (Fuel) - 2SGA 6.62 %

3 Moisture (Fuel) - 2SGA 5.82 %

4 Moisture (Fuel) - 2SGA 6.19 %

Monday, February 28, 2011 Page 23 of 33
6481_INTERMOUNTAIN

8/31/2010

1 Residule Moisture (Fuel) - 2SGA 1.88 %

2 Residule Moisture (Fuel) - 2SGA 1.52 %

3 Residule Moisture (Fuel) - 2SGA 1.28 %

4 Residule Moisture (Fuel) - 2SGA 1.4 %

1 Ash Proximate (Fuel) - 2SGA 13.62 %

2 Ash Proximate (Fuel) - 2SGA 14.73 %

3 Ash Proximate (Fuel) - 2SGA 15.24 %

4 Ash Proximate (Fuel) - 2SGA 14.49 %

1 Volatiles Proximate (Fuel) - 2SGA 37.46 %

2 Volatiles Proximate (Fuel) - 2SGA 36.64 %

3 Volatiles Proximate (Fuel) - 2SGA 36.35 %

4 Volatiles Proximate (Fuel) - 2SGA 36.52 %

1 Fixed Carbon (Fuel) - 2SGA 44.48 %

2 Fixed Carbon (Fuel) - 2SGA 44.99 %

3 Fixed Carbon (Fuel) - 2SGA 45.22 %

4 Fixed Carbon (Fuel) - 2SGA 45.62 %

1 Carbon Ultimate (Fuel) - 2SGA 67.87 %

2 Carbon Ultimate (Fuel) - 2SGA 67.88 %

3 Carbon Ultimate (Fuel) - 2SGA 68.09 %

4 Carbon Ultimate (Fuel) - 2SGA 68.95 %

1 Hydrogen Ultimate (Fuel) - 2SGA 4.85 %

2 Hydrogen Ultimate (Fuel) - 2SGA 4.8 %

3 Hydrogen Ultimate (Fuel) - 2SGA 4.8 %

4 Hydrogen Ultimate (Fuel) - 2SGA 4.99 %

1 Nitrogent Ultimate (Fuel) - 2SGA 1.45 %

2 Nitrogent Ultimate (Fuel) - 2SGA 1.52 %

3 Nitrogent Ultimate (Fuel) - 2SGA 1.57 %

4 Nitrogent Ultimate (Fuel) - 2SGA 1.58 %

1 Methane Content (Gas Stream) - 2SGA 2.5 ppm

2 Methane Content (Gas Stream) - 2SGA 2.5 ppm

3 Methane Content (Gas Stream) - 2SGA 2.5 ppm

4 Methane Content (Gas Stream) - 2SGA 0 ppm

Monday, February 28, 2011 Page 24 of 33
6481_INTERMOUNTAIN

8/31/2010

Sampling / Stack Data Results Detail

Location 2SGA - EPA Method 316

Average

Run Number M316-1 M316-2 M316-3

Test Date 6/16/2010 6/16/2010 6/16/2010

Run Start Time 8:37:00 AM 11:14:00 AM 1:43:00 PM

Run Finish Time 10:27:00 AM 1:04:00 PM 3:33:00 PM

Net Traversing Points 12 12 12

Net Run Time, minutes 120 120 120

Nozzle Diameter, inches 0.18 0.18 0.18 0.180

Pitot Tube Coefficient 0.84 0.84 0.84 0.840

Dry Gas Meter Calibration Factor 1.0033 1.0033 1.0033 1.003

Barometric Pressure, inches of Mercury 24.86 24.86 24.86 24.860

Average Orifice Meter Differential, inches H2O 1.41 1.48 1.49 1.460

Dry Gas Meter Volume Sampled, cubic feet 83.053 86.623 87.211 85.629

Average Dry Gas Meter Temperature, °F 87.04 92.29 93.83 91.053

Dry Gas Meter Volume Sampled, dscf 67.104 69.337 69.616 68.686

Total Moisture Liquid collected, g 210.7 226.9 232.6 223.400

Volume of Water Vapor, standard cubic feet 9.92 10.68 10.95 10.517

Moisture Content of Stack Gas, % 12.88 13.35 13.59 13.273

Moisture Saturation at Stack Gas Temperature, % 13.84 13.93 14.22 13.997

Dry Mole Fraction 0.8712 0.8665 0.8641 0.867

Carbon Dioxide, % 13.89 13.89 13.83 13.870

Oxygen, % 5.59 5.6 5.65 5.613

Carbon Monoxide & Nitrogen, % 80.52 80.51 80.52 80.517

Fuel Factor 1.10 1.10 1.10

Dry Molecular Weight, lb/lb-Mole 30.45 30.45 30.44 30.447

Wet Molecular weight, lb/lb-Mole 28.85 28.79 28.75 28.797

Flue Gas Static Pressure, inches of H2O -0.6 -0.6 -0.6 -0.600

Absolute Flue Gas Pressure, inches of Mercury 24.82 24.82 24.82 24.820

Average Stack Gas Temperature, °F 119.92 120.17 120.92 120.337

Average Velocity Head, inches of H2O 1.5733360916

8427

1.652152669

3252

1.670552727

40467

1.632

Average Stack Gas Velocity, feet/second 81.06 83.17 83.75 82.660

Stack Cross-Sectional Area, square feet 615.752 615.752 615.752 615.752

Dry Volumetric Flow Rate, dry scfm 1,970,550.5 2,010,070.0 2,015,875.4 1,998,831.9

67

Actual Wet Volumetric Flue Gas Flow Rate, acfm 2,994,771.4 3,072,725.6 3,094,153.8 3,053,883.6

00

Percent Isokinetic of Sampling Rate, % 98.9 100.2 100.3 99.800

Percent Excess Air, % 35.7 35.8 36.2 35.900

F-Factor, dscfm/MMBtu @ %O2 9647 9652 9695 9,664.667

Round Duct Diameter, inches 336 336 336

Rectangular Duct Width, inches 0 0 0

Rectangular Duct Length, inches 0 0 0

Fw 10566 10532 10559 10,552.333

Fc 0 0 0 0.000

Monday, February 28, 2011 Page 25 of 33
6481_INTERMOUNTAIN

8/31/2010

Location 2SGA - Method 1 - 4

Average

Run Number 1

Test Date 6/16/2010

Run Start Time 4:03:00 PM

Run Finish Time 5:53:00 PM

Net Traversing Points 13

Net Run Time, minutes 120

Nozzle Diameter, inches

Pitot Tube Coefficient 0.84 0.840

Dry Gas Meter Calibration Factor 1.0033 1.003

Barometric Pressure, inches of Mercury 24.72 24.720

Average Orifice Meter Differential, inches H2O 1.30 1.300

Dry Gas Meter Volume Sampled, cubic feet 79.184 79.184

Average Dry Gas Meter Temperature, °F 93.38 93.380

Dry Gas Meter Volume Sampled, dscf 62.870 62.870

Total Moisture Liquid collected, g 192.1 192.100

Volume of Water Vapor, standard cubic feet 9.04 9.040

Moisture Content of Stack Gas, % 12.57 12.570

Moisture Saturation at Stack Gas Temperature, % 14.78 14.780

Dry Mole Fraction 0.8743 0.874

Carbon Dioxide, % 13.22 13.220

Oxygen, % 6.11 6.110

Carbon Monoxide & Nitrogen, % 80.67 80.670

Fuel Factor 1.12

Dry Molecular Weight, lb/lb-Mole 30.36 30.360

Wet Molecular weight, lb/lb-Mole 28.81 28.810

Flue Gas Static Pressure, inches of H2O -1.45 -1.450

Absolute Flue Gas Pressure, inches of Mercury 24.61 24.610

Average Stack Gas Temperature, °F 122.00 122.000

Average Velocity Head, inches of H2O 1.9714878957

044

1.971

Average Stack Gas Velocity, feet/second 91.35 91.350

Stack Cross-Sectional Area, square feet 615.752 615.752

Dry Volumetric Flow Rate, dry scfm 2,201,846.7 2,201,846.7

00

Actual Wet Volumetric Flue Gas Flow Rate, acfm 3,374,936.7 3,374,936.7

00

Percent Isokinetic of Sampling Rate, %

Percent Excess Air, % 40.2 40.200

F-Factor, dscfm/MMBtu @ %O2 9763 9,763.000

Round Duct Diameter, inches 336

Rectangular Duct Width, inches 0

Rectangular Duct Length, inches 0

Fw 10654 10,654.000

Fc 0 0.000

Monday, February 28, 2011 Page 26 of 33
6481_INTERMOUNTAIN

8/31/2010

Location 2SGA - Method 10

Average

Run Number 1 2 3

Test Date 6/16/2010 6/16/2010 6/16/2010

Run Start Time 8:32:00 AM 11:13:00 AM 1:40:00 PM

Run Finish Time 10:31:00 AM 1:12:00 PM 3:39:00 PM

Carbon Dioxide, % 13.42 13.22 13.11 13.250

Oxygen, % 5.91 6.07 6.2 6.060

Fuel Factor 1.12 1.12 1.12

Dry Volumetric Flow Rate, dry scfm 2263815 231951 2331351 1,609,039.0

00

F-Factor, dscfm/MMBtu @ %O2 9647 9652 9695 9,664.667

Moisture, % 12.96 13.43 13.68 13.357

Analyzer Make TECO TECO TECO

Analyzer Model 48C 48C 48C

Analyzer Serial Number

Operating Range 505 505 505 505.000

Operating Units ppm ppm ppm

No. Readings/Avg. 60 60 60 60.000

Calibration Set 4 5 5 4.667

Calibration Pre Zero Cylinder ID Zero Gas Zero Gas Zero Gas

Calibration Pre Zero Cylinder Instrument Response -0.36 -1.31 0.09 -0.527

Calibration Pre Zero Cylinder Bias 0.07 -0.11 0.16 0.040

Calibration Pre Zero Cylinder Drift 0 0 0 0.000

Calibration Pre High Cylinder ID CC249349-CO CC71524-

CO

CC71524-CO

Calibration Pre High Cylinder Instrument Response 99.16 187.65 187.16 157.990

Calibration Pre High Cylinder Bias -0.05 -2.26 -2.35 -1.553

Calibration Pre High Cylinder Drift 0 0 0 0.000

Calibration Post Zero Cylinder ID Zero Gas Zero Gas Zero Gas

Calibration Post Zero Cylinder Instrument Response -1.31 0.09 -1.46 -0.893

Calibration Post Zero Cylinder Bias -0.11 0.16 -0.14 -0.030

Calibration Post Zero Cylinder Drift 0.18 0.27 0.3 0.250

Calibration Post High Cylinder ID CC249349-CO CC71524-

CO

CC71524-CO

Calibration Post High Cylinder Instrument Response 99.73 187.16 186.78 157.890

Calibration Post High Cylinder Bias 0.06 -2.35 -2.43 -1.573

Calibration Post High Cylinder Drift 0.11 9.000000000

00003E-02

8.000000000

00001E-02

0.093

Cavg 572.43 387.27 348.92 436.207

Cavg Units ppmvd ppmvd ppmvd

Cgas 568.806 412.6054 372.604 451.338

Cgas Units ppmvd ppmvd ppmvd

Fuel Type Override Override Override

Fw 10566 10532 10559 10,552.333

Fc 0 0 0 0.000

Cgasw 0 0 0 0.000

Cgasw Units

Monday, February 28, 2011 Page 27 of 33
6481_INTERMOUNTAIN

8/31/2010

Location 2SGA - Method 10 3

Average

Run Number 4

Test Date 6/16/2010

Run Start Time 4:00:00 PM

Run Finish Time 5:59:00 PM

Carbon Dioxide, % 13.22 13.220

Oxygen, % 6.11 6.110

Fuel Factor 1.12

Dry Volumetric Flow Rate, dry scfm 2517279 2,517,279.0

00

F-Factor, dscfm/MMBtu @ %O2 9763 9,763.000

Moisture, % 12.66 12.660

Analyzer Make TECO

Analyzer Model 48C

Analyzer Serial Number

Operating Range 505 505.000

Operating Units ppm

No. Readings/Avg. 60 60.000

Calibration Set 5 5.000

Calibration Pre Zero Cylinder ID Zero Gas

Calibration Pre Zero Cylinder Instrument Response -1.46 -1.460

Calibration Pre Zero Cylinder Bias -0.14 -0.140

Calibration Pre Zero Cylinder Drift 0 0.000

Calibration Pre High Cylinder ID CC71524-CO

Calibration Pre High Cylinder Instrument Response 186.78 186.780

Calibration Pre High Cylinder Bias -2.43 -2.430

Calibration Pre High Cylinder Drift 0 0.000

Calibration Post Zero Cylinder ID Zero Gas

Calibration Post Zero Cylinder Instrument Response -0.63 -0.630

Calibration Post Zero Cylinder Bias 0.02 0.020

Calibration Post Zero Cylinder Drift 0.16 0.160

Calibration Post High Cylinder ID CC71524-CO

Calibration Post High Cylinder Instrument Response 187.8 187.800

Calibration Post High Cylinder Bias -2.23 -2.230

Calibration Post High Cylinder Drift 0.2 0.200

Cavg 315.91 315.910

Cavg Units ppmvd

Cgas 336.5864 336.586

Cgas Units ppmvd

Fuel Type Override

Fw 10654 10,654.000

Fc 0 0.000

Cgasw 0 0.000

Cgasw Units

Monday, February 28, 2011 Page 28 of 33
6481_INTERMOUNTAIN

8/31/2010

Location 2SGA - Method 25A

Average

Run Number 1 2 3

Test Date 6/16/2010 6/16/2010 6/16/2010

Run Start Time 8:32:00 AM 11:13:00 AM 1:40:00 PM

Run Finish Time 10:31:00 AM 1:12:00 PM 3:39:00 PM

Carbon Dioxide, % 13.42 13.22 13.11 13.250

Oxygen, % 5.91 6.07 6.2 6.060

Fuel Factor 1.12 1.12 1.12

Dry Volumetric Flow Rate, dry scfm 2263815 2319571 2331351 2,304,912.3

33

F-Factor, dscfm/MMBtu @ %O2 9647 9652 9695 9,664.667

Moisture, % 12.96 13.43 13.68 13.357

Analyzer Make VIG VIG VIG

Analyzer Model 20/2 20/2 20/2

Analyzer Serial Number

Operating Range 10 10 10 10.000

Operating Units ppm ppm ppm

No. Readings/Avg. 60 60 60 60.000

Calibration Set 6 6 6 6.000

Calibration Pre Zero Cylinder ID Zero Gas Zero Gas Zero Gas

Calibration Pre Zero Cylinder Instrument Response 0.08 0.28 0.25 0.203

Calibration Pre Zero Cylinder Bias 0 2 1.7 1.233

Calibration Pre Zero Cylinder Drift 0 0 0 0.000

Calibration Pre High Cylinder ID CC150368-

THC

CC150368-

THC

CC150368-

THC

Calibration Pre High Cylinder Instrument Response 4.12 4.18 4.06 4.120

Calibration Pre High Cylinder Bias 0 0.6 -0.6 0.000

Calibration Pre High Cylinder Drift 0 0 0 0.000

Calibration Post Zero Cylinder ID Zero Gas Zero Gas Zero Gas

Calibration Post Zero Cylinder Instrument Response 0.28 0.25 0.23 0.253

Calibration Post Zero Cylinder Bias 2 1.7 1.5 1.733

Calibration Post Zero Cylinder Drift 2 0.3 0.2 0.833

Calibration Post High Cylinder ID CC150368-

THC

CC150368-

THC

CC150368-

THC

Calibration Post High Cylinder Instrument Response 4.18 4.06 4.17 4.137

Calibration Post High Cylinder Bias 0.6 -0.6 0.5 0.167

Calibration Post High Cylinder Drift 0.6 1.2 1.1 0.967

Cavg 0.36 0.44 0.36 0.387

Cavg Units ppmvd ppmvd ppmvd

Cgas 0.1832 0.1834 0.1251 0.164

Cgas Units ppmvd ppmvd ppmvd

Fuel Type Override Override Override

Fw 10566 10532 10559 10,552.333

Fc 0 0 0 0.000

Cgasw 0 0 0 0.000

Cgasw Units

Monday, February 28, 2011 Page 29 of 33
6481_INTERMOUNTAIN

8/31/2010

Location 2SGA - Method 3A CO2

Average

Run Number 1 2 3

Test Date 6/16/2010 6/16/2010 6/16/2010

Run Start Time 8:32:00 AM 11:13:00 AM 1:40:00 PM

Run Finish Time 10:31:00 AM 1:12:00 PM 3:39:00 PM

Carbon Dioxide, % 13.42 13.22 13.11 13.250

Oxygen, % 5.91 6.07 6.2 6.060

Fuel Factor 1.12 1.12 1.12

Dry Volumetric Flow Rate, dry scfm 2263815 2319571 2331351 2,304,912.3

33

F-Factor, dscfm/MMBtu @ %O2 9647 9652 9695 9,664.667

Moisture, % 12.96 13.43 13.68 13.357

Analyzer Make Servomex Servomex Servomex

Analyzer Model 4900 4900 4900 4,900.000

Analyzer Serial Number

Operating Range 25 25 25 25.000

Operating Units % % %

No. Readings/Avg. 60 60 60 60.000

Calibration Set 8 8 8 8.000

Calibration Pre Zero Cylinder ID Zero Gas Zero Gas Zero Gas

Calibration Pre Zero Cylinder Instrument Response -0.03 0.1 0.12 0.063

Calibration Pre Zero Cylinder Bias 0.04 0.56 0.64 0.413

Calibration Pre Zero Cylinder Drift 0 0 0 0.000

Calibration Pre High Cylinder ID CC155303-

CO2

CC155303-

CO2

CC155303-

CO2

Calibration Pre High Cylinder Instrument Response 11.79 11.76 11.83 11.793

Calibration Pre High Cylinder Bias -0.76 -0.88 -0.6 -0.747

Calibration Pre High Cylinder Drift 0 0 0 0.000

Calibration Post Zero Cylinder ID Zero Gas Zero Gas Zero Gas

Calibration Post Zero Cylinder Instrument Response 0.1 0.12 0.06 0.093

Calibration Post Zero Cylinder Bias 0.56 0.64 0.4 0.533

Calibration Post Zero Cylinder Drift 0.52 0.08 0.24 0.280

Calibration Post High Cylinder ID CC155303-

CO2

CC155303-

CO2

CC155303-

CO2

Calibration Post High Cylinder Instrument Response 11.76 11.83 11.74 11.777

Calibration Post High Cylinder Bias -0.88 -0.6 -0.96 -0.813

Calibration Post High Cylinder Drift 0.12 0.28 0.36 0.253

Cavg 12.95 12.78 12.65 12.793

Cavg Units %vd %vd %vd

Cgas 13.421 13.2284 13.1024 13.251

Cgas Units %vd %vd %vd

Fuel Type Override Override Override

Fw 10566 10532 10559 10,552.333

Fc 0 0 0 0.000

Cgasw 0 0 0 0.000

Cgasw Units

Monday, February 28, 2011 Page 30 of 33
6481_INTERMOUNTAIN

8/31/2010

Location 2SGA - Method 3A CO2 3

Average

Run Number 4

Test Date 6/16/2010

Run Start Time 4:00:00 PM

Run Finish Time 5:59:00 PM

Carbon Dioxide, % 13.22 13.220

Oxygen, % 6.11 6.110

Fuel Factor 1.12

Dry Volumetric Flow Rate, dry scfm 2517279 2,517,279.0

00

F-Factor, dscfm/MMBtu @ %O2 9763 9,763.000

Moisture, % 12.66 12.660

Analyzer Make Servomex

Analyzer Model 4900 4,900.000

Analyzer Serial Number

Operating Range 25 25.000

Operating Units %

No. Readings/Avg. 60 60.000

Calibration Set 8 8.000

Calibration Pre Zero Cylinder ID Zero Gas

Calibration Pre Zero Cylinder Instrument Response 0.06 0.060

Calibration Pre Zero Cylinder Bias 0.4 0.400

Calibration Pre Zero Cylinder Drift 0 0.000

Calibration Pre High Cylinder ID CC155303-

CO2

Calibration Pre High Cylinder Instrument Response 11.74 11.740

Calibration Pre High Cylinder Bias -0.96 -0.960

Calibration Pre High Cylinder Drift 0 0.000

Calibration Post Zero Cylinder ID Zero Gas

Calibration Post Zero Cylinder Instrument Response 0.08 0.080

Calibration Post Zero Cylinder Bias 0.48 0.480

Calibration Post Zero Cylinder Drift 0.08 0.080

Calibration Post High Cylinder ID CC155303-

CO2

Calibration Post High Cylinder Instrument Response 11.89 11.890

Calibration Post High Cylinder Bias -0.36 -0.360

Calibration Post High Cylinder Drift 0.6 0.600

Cavg 12.8 12.800

Cavg Units %vd

Cgas 13.2232 13.223

Cgas Units %vd

Fuel Type Override

Fw 10654 10,654.000

Fc 0 0.000

Cgasw 0 0.000

Cgasw Units

Monday, February 28, 2011 Page 31 of 33
6481_INTERMOUNTAIN

8/31/2010

Location 2SGA - Method 3A O2

Average

Run Number 1 2 3

Test Date 6/15/2010 6/16/2010 6/16/2010

Run Start Time 8:32:00 AM 11:13:00 AM 1:40:00 PM

Run Finish Time 10:31:00 AM 1:12:00 PM 3:39:00 PM

Carbon Dioxide, % 13.42 13.22 13.11 13.250

Oxygen, % 5.91 6.07 6.2 6.060

Fuel Factor 1.12 1.12 1.12

Dry Volumetric Flow Rate, dry scfm 2263815 2319571 2331351 2,304,912.3

33

F-Factor, dscfm/MMBtu @ %O2 9647 9652 9695 9,664.667

Moisture, % 12.96 13.43 13.68 13.357

Analyzer Make Servomex Servomex Servomex

Analyzer Model 4900 4900 4900 4,900.000

Analyzer Serial Number

Operating Range 25 25 25 25.000

Operating Units % % %

No. Readings/Avg. 60 60 60 60.000

Calibration Set 9 9 9 9.000

Calibration Pre Zero Cylinder ID Zero Gas Zero Gas Zero Gas

Calibration Pre Zero Cylinder Instrument Response 0.04 0.05 0.17 0.087

Calibration Pre Zero Cylinder Bias 0 0.04 0.52 0.187

Calibration Pre Zero Cylinder Drift 0 0 0 0.000

Calibration Pre High Cylinder ID CC155303-O2 CC155303-

O2

CC155303-

O2

Calibration Pre High Cylinder Instrument Response 11.98 12.13 12.13 12.080

Calibration Pre High Cylinder Bias -1 -0.4 -0.4 -0.600

Calibration Pre High Cylinder Drift 0 0 0 0.000

Calibration Post Zero Cylinder ID Zero Gas Zero Gas Zero Gas

Calibration Post Zero Cylinder Instrument Response 0.05 0.17 0.25 0.157

Calibration Post Zero Cylinder Bias 0.04 0.52 0.84 0.467

Calibration Post Zero Cylinder Drift 0.04 0.48 0.32 0.280

Calibration Post High Cylinder ID CC155303-O2 CC155303-

O2

CC155303-

O2

Calibration Post High Cylinder Instrument Response 12.13 12.13 12.11 12.123

Calibration Post High Cylinder Bias -0.4 -0.4 -0.48 -0.427

Calibration Post High Cylinder Drift 0.6 0 0.08 0.227

Cavg 5.91 6.14 6.32 6.123

Cavg Units %vd %vd %vd

Cgas 5.909 6.0701 6.2075 6.062

Cgas Units %vd %vd %vd

Fuel Type Override Override Override

Fw 10566 10532 10559 10,552.333

Fc 0 0 0 0.000

Cgasw 0 0 0 0.000

Cgasw Units

Monday, February 28, 2011 Page 32 of 33
6481_INTERMOUNTAIN

8/31/2010

Location 2SGA - Method 3A O2 3

Average

Run Number 4

Test Date 6/16/2010

Run Start Time 4:00:00 PM

Run Finish Time 5:59:00 PM

Carbon Dioxide, % 13.22 13.220

Oxygen, % 6.11 6.110

Fuel Factor 1.12

Dry Volumetric Flow Rate, dry scfm 2517279 2,517,279.0

00

F-Factor, dscfm/MMBtu @ %O2 9763 9,763.000

Moisture, % 12.66 12.660

Analyzer Make Servomex

Analyzer Model 4900 4,900.000

Analyzer Serial Number

Operating Range 25 25.000

Operating Units %

No. Readings/Avg. 60 60.000

Calibration Set 9 9.000

Calibration Pre Zero Cylinder ID Zero Gas

Calibration Pre Zero Cylinder Instrument Response 0.25 0.250

Calibration Pre Zero Cylinder Bias 0.84 0.840

Calibration Pre Zero Cylinder Drift 0 0.000

Calibration Pre High Cylinder ID CC155303-O2

Calibration Pre High Cylinder Instrument Response 12.11 12.110

Calibration Pre High Cylinder Bias -0.48 -0.480

Calibration Pre High Cylinder Drift 0 0.000

Calibration Post Zero Cylinder ID Zero Gas

Calibration Post Zero Cylinder Instrument Response 0.05 0.050

Calibration Post Zero Cylinder Bias 0.04 0.040

Calibration Post Zero Cylinder Drift 0.8 0.800

Calibration Post High Cylinder ID CC155303-O2

Calibration Post High Cylinder Instrument Response 12.05 12.050

Calibration Post High Cylinder Bias -0.72 -0.720

Calibration Post High Cylinder Drift 0.24 0.240

Cavg 6.18 6.180

Cavg Units %vd

Cgas 6.1159 6.116

Cgas Units %vd

Fuel Type Override

Fw 10654 10,654.000

Fc 0 0.000

Cgasw 0 0.000

Cgasw Units

Monday, February 28, 2011 Page 33 of 33