1

NIST TIME AND FREQUENCY BULLETIN 

NISTIR 6629-09

 NO. 549 SEPTEMBER 2003 

1. GENERAL BACKGROUND INFORMATION ............................................................................................... 2

2. TIME-SCALE INFORMATION ..................................................................................................................... 2

3. PHASE DEVIATIONS FOR WWVB AND LORAN-C ................................................................................... 4

4. BROADCAST OUTAGES OVER FIVE MINUTES AND WWVB 

 PHASE PERTURBATIONS........................................................................................................................... 5

5. NOTES ON NIST TIME SCALES AND PRIMARY STANDARDS............................................................... 5

6. BIBLIOGRAPHY.............................................................................................................................................. 5

7. SPECIAL ANNOUNCEMENTS..................................................................................................................... 7

This bulletin is published monthly. Address correspondence to: 

Eyvon M. Petty, Editor 

Time and Frequency Division 

National Institute of Standards and Technology

325 Broadway 

Boulder, CO 8O3O5-3328 

(3O3) 497-3295

Email: pettye@boulder.nist.gov

NOTE TO SUBSCRIBERS: Please include your address label (or a copy) 

with any correspondence regarding this bulletin. 

U.S. DEPARTMENT OF COMMERCE, DONALD L. EVANS, Secretary

TECHNOLOGY ADMINISTRATION, Dr. Cheryl L. Shavers, Under Secretary of Commerce for Technology

NATIONAL INSTITUTE OF STANDARDS AND TECHNOLOGY, Dr. Arden L. Bement, Jr., Director
2

1. GENERAL BACKGROUND INFORMATION

_______________________________________________________________________________________________________

ACRONYMS AND ABBREVIATIONS USED IN THIS BULLETIN

BIPM - Bureau International des Poids et Mesures

Cs - Cesium standard

GPS - Global Positioning System

IERS - International Earth Rotation Service

LORAN - Long Range Navigation

MC - Master Clock

MJD - Modified Julian Date

NVLAP - National Voluntary Laboratory Accreditation Program

NIST - National Institute of Standards and Technology

NOAA - National Oceanic and Atmospheric Administration ns - nanosecond

SI - International System of Units μs - microsecond

TA - Atomic Time ms - millisecond

TAI - International Atomic Time s - second

USNO - United States Naval Observatory min - minute

UTC - Coordinated Universal Time

_______________________________________________________________________________________________________

2. TIME-SCALE INFORMATION

 The values listed below are based on data from the IERS, the USNO, and NIST. The UTC(USNO,MC) - UTC(NIST) values are 

averaged measurements from all available common-view GPS satellites (see bibliography on page 5). UTC - UTC(NIST) data are on 

page 3.

 0000 HOURS COORDINATED UNIVERSAL TIME

AUG 2003 MJD UT1-UTC(NIST)

 (±5 ms)

UTC(USNO,MC) - UTC(NIST)

 (±20 ns)

7 52858 -354 ms 4 ns

14 52865 -352 ms 4 ns

21 52872 -352 ms 2 ns

28 52879 -351 ms 0 ns

 The master clock pulses used by the WWV, WWVH, and WWVB time-code transmissions are referenced to the UTC(NIST) time 

scale. Occasionally, 1 s is added to the UTC time scale. This second is called a leap second. Its purpose is to keep the UTC time 

scale within ±0.9 s of the UT1 astronomical time scale, which changes slightly due to variations in the rate of rotation of the Earth. 

NOTE: NO leap second will be inserted at the end of December 2003.

 Positive leap seconds, beginning at 23 h 59 min 60 s UTC and ending at 0 h 0 min 0 s UTC, were inserted in the UTC timescale 

on 30 June 1972, 1981-1983, 1985, 1992, 1993, 1994, and 1997, and on 31 December 1972-1979, 1987, 1989, 1990,1995, and 

1998. There have been 22 leap seconds inserted in total. 

 The use of leap seconds ensures that UT1 - UTC will always be held within ±0.9 s. The current value of UT1 - UTC is called the 

DUT1 correction. DUT1 corrections are broadcast by WWV, WWVH, and WWVB and are printed below. These corrections may be 

added to received UTC time signals in order to obtain UT1. 

_______________________________________________________________________________________________________

+0.1 s beginning 0000 UTC 19 October 2000

+0.0 s beginning 0000 UTC 01 March 2001

DUT1 = UT1 - UTC = -0.1 s beginning 0000 UTC 04 October 2001

-0.2 s beginning 0000 UTC 14 February 2002

-0.3 s beginning 0000 UTC 24 October 2002

-0.4 s beginning 0000 UTC 03 April 2003
3

 

The deviation of UTC(NIST) from UTC has been within +/–100 ns since July 6, 1994. The table below shows values of UTC 

- UTC(NIST) as supplied by the BIPM in their Circular T publication for the most recent 310-day period in which data are 

available. Data are given at ten day intervals. Five day interval data are available in Circular T. 

0000 Hours Coordinated Universal Time

DATE MJD UTC-UTC(NIST) ns

Jul. 29, 2003 52849 10.7

Jul. 19, 2003 52839 12.7

Jul. 9, 2003 52829 12.6

Jun. 29, 2003 52819 9.2

Jun. 19, 2003 52809 7.5

Jun. 9, 2003 52799 3.8

May 30, 2003 52789 0.6

May 20, 2003 52779 4.0

May 10, 2003 52769 10.4

Apr. 30, 2003 52759 8.9

Apr. 20, 2003 52749 10.7

Apr. 10, 2003 52739 10.9

Mar. 31, 2003 52729 11

Mar. 21, 2003 52719 12

Mar. 11, 2003 52709 11

Mar. 1, 2003 52699 7

Feb. 19, 2003 52689 8

Feb. 9, 2003 52679 7

Jan. 30, 2003 52669 2

Jan. 20, 2003 52659 -1

Jan. 10, 2003 52649 -5

Dec. 31, 2002 52639 -2

Dec. 21, 2002 52629 -3

Dec. 11, 2002 52619 -4

Dec. 1, 2002 52609 -3

Nov. 22, 2002 52599 2

Nov. 12, 2002 52589 3

Nov. 2, 2002 52579 4

Oct. 22, 2002 52569 5

Oct. 12, 2002 52559 4

Oct. 2, 2002 52549 4
4

3. PHASE DEVIATIONS FOR WWVB AND LORAN-C

WWVB - The values shown for WWVB are the time differences between the time markers of the UTC(NIST) time scale 

and the first positive -going zero voltage crossover measured at the transmitting antenna. The unce rtainty of 

the individual measurements is ±0.5 μs. The values listed are for 1300 UTC.

LORAN-C - The values shown for Loran-C represent the daily accumulated phase shift (in ns). The phase shift is

measured by comparing the output of a Loran receiver to the UTC(NIST) time scale for a period of 24 h. If data 

were not recorded on a particular day, the symbol ( -) is printed. The stations monitored are Baudette, ND

(8970-Y) and Fallon, NV (9940). The monitoring is done from the NIST laboratories in Boulder, Colorado.

Note: The values shown for Loran-C are in nanoseconds.

UTC(NIST)-WWVB (60 kHz) UTC(NIST) - LORAN PHASE (ns)

ANTENNA PHASE LORAN-C (BAUDETTE) LORAN-C (FALLON)

DATE MJD (μs) (8970) (9940)

08/01/03 52852 5.72 -23 +119

08/02/03 52853 5.72 +117 -351

08/03/03 52854 5.72 +9 -231

08/04/03 52855 5.72 -35 -316

08/05/03 52856 5.72 -11 -368

08/06/03 52857 5.72 +14 -54

08/07/03 52858 5.73 -116 +73

08/08/03 52859 5.72 +10 -59

08/09/03 52860 5.72 +90 +140

08/10/03 52861 5.72 -51 -443

08/11/03 52862 5.72 -62 -37

08/12/03 52863 5.73 -40 +346

08/13/03 52864 5.73 -21 -143

08/14/03 52865 5.73 -10 -249

08/15/03 52866 5.71 -41 +123

08/16/03 52867 5.70 -19 +148

08/17/03 52868 5.70 -117 +36

08/18/03 52869 5.69 +14 -329

08/19/03 52870 5.70 +75 -227

08/20/03 52871 5.71 +12 -205

08/21/03 52872 5.70 -101 +171

08/22/03 52873 5.70 -2 -38

08/23/03 52874 5.70 +44 -102

08/24/03 52875 5.70 -35 -15

08/25/03 52876 5.70 -47 -269

08/26/03 52877 5.71 +5 +171

08/27/03 52878 5.70 -51 -27

08/28/03 52879 5.70 +1 +268

08/29/03 52880 5.72 +13 -362

08/30/03 52881 5.71 -36 -78

08/31/03 52882 5.71 +5 +81
5

4. BROADCAST OUTAGES OVER FIVE MINUTES AND WWVB PHASE PERTURBATIONS

OUTAGES OF 5 MINUTES OR MORE

WWVB 60 kHz

PHASE PERTURBATIONS

2 ms

Station AUG

2003

 MJD Began 

UTC

Ended UTC Freq. AUG

2003

MJD Began

 UTC

End

UTC

WWVB

WWV

WWVH

5. NOTES ON NIST TIME SCALES AND PRIMARY STANDARDS

Primary frequency standards developed and maintained by NIST are used to provide accuracy (rate) input to the BIPM. NIST-7, which 

had served as the U.S. primary standard since 1994, has been replaced by NIST-F1, a cesium fountain frequency standard. The 

uncertainty of the new standard is currently 1.7 parts in 1015.

 The AT1 scale is run in real-time using data from an ensemble of cesium standards and hydrogen masers. It is a free-running scale 

whose frequency is maintained as nearly constant as possible by choosing the optimum weight for each clock that contributes to the 

computation.

 UTC(NIST) is generated as an offset from our real-time scale AT1. It is steered in frequency towards UTC using data published by the 

BIPM in its Circular T. Changes in the steering frequency will be made, if necessary, at 0000 UTC on the first day of the month, and 

very occasionally at mid-month. A change in frequency is limited to no more than ± 2 ns/day. The frequency of UTC(NIST) is kept as 

stable as possible at other times. 

 UTC is generated at the BIPM using a post-processed time-scale algorithm and is not available in real-time. The parameters that we 

use to generate UTC(NIST) in real-time are therefore based on an extrapolation of UTC from the most recent data available. 

6. BIBLIOGRAPHY

Allan, D.W.; Hellwig, H.; and Glaze, D.J., "An accuracy algorithm for an atomic time scale," Metrologia, Vol.11, No.3,

 pp.133-138 (September 1975). 

Allan, D.W. and Weiss, M.A., "Accurate time and frequency transfer during common view of a GPS satellite," Proc. 34th

 Annual Symposium on Frequency Control, p.334 (1980). 

Allan, D.W. and Barnes, J.A., "Optimal time and frequency using GPS signals," Proc. 36th Annual Symposium on

 Frequency Control, p.378 (1982). 

Drullinger, R.E.; Glaze, D.J.; Lowe, J.P.; and Shirley, J.H., "The NIST optically pumped cesium frequency standard," IEEE

 Trans. Instrum. Meas., IM-40, 162-164 (1991).

Glaze, D.J.; Hellwig, H.; Allan, D.W.; and Jarvis, S., "NBS-4 and NBS-6: The NIST primary frequency standards,"

 Metrologia, Vol.13, pp.17-28 (1977). 

Wineland, D.J.; Allan, D.W.; Glaze, D.J.; Hellwig, H.; and Jarvis, S., "Results on limitations in primary cesium standard

 operation," IEEE Trans. Instrum. Meas., IM-25, pp.453-458 (December 1976). 
6

Table 7.1 lists parameters that are used to define UTC(NIST) with respect to our real-time scale AT1. To find the value of 

UTC(NIST) - AT1 at any time T (expressed as a Modified Julian Day, including a fraction if needed), the appropriate 

equation to use is the one for which the desired T is greater than or equal to the entry in the T0 column and less than the 

entry in the last column. The values of xls, x, and y for that month are then used in the equation below to find the desired 

value. The parameters x and y represent the offset in time and in frequency, respectively, between UTC(NIST) and AT1; 

the parameter xls is the number of leap seconds applied to both UTC(NIST) and UTC as specified by the IERS. Leap 

seconds are not applied to AT1.

Table 7.1

UTC(NIST) - AT1 = xls + x + y*(T ? T0)

Month xls

(s)

x

(ns)

y

(ns/d)

 T0

(MJD)

Valid until 0000 on:

(MJD)

 Oct 03 -32 -255112.45 -39.5* 52913 52944*

 Sep 03 -32 -253927.745 -39.5 52883 52913

Aug 03 -32 -252702.95 -39.5 52852 52883

 Jul 03 -32 -252228.95 -39.5 52840 52852

 Jul 03 -32 -251473.7 -39.75 52821 52840†

 Jun 03 -32 -251076.2 -39.75 52811 52821

 Jun 03 -32 -250276.2 -40.0 52791 52811†

 May 03 -32 -249652.2 -39.0 52775 52791

 May 03 -32 -249052.2 -40.0 52760 52775†

 Apr 03 -32 -248495.7 -39.75 52746 52760

 Apr 03 -32 -247855.7 -40.0 52730 52746†

 Mar 03 -32 -247415.7 -40.0 52719 52730

 Mar 03 -32 -246607.7 -40.4 52699 52719†

 Feb 03 -32 -246284.9 -40.35 52691 52699

 Feb 03 -32 -245474.9 -40.5 52671 52691†

 Jan 03 -32 -244906.5 -40.6 52657 52671

 Jan 03 -32 -244218.0 -40.5 52640 52657†

 Dec 02 -32 -243813.0 -40.5 52630 52640

 Dec 02 -32 -242964.6 -40.4 52609 52630†

 Nov 02 -32 -242399.0 -40.4 52595 52609

 Nov 02 -32 -241751.0 -40.5 52579 52595†

 Oct 02 -32 -240495.5 -40.5 52548 52579

 Sep 02 -32 -240252.5 -40.5 52542 52548

 Sep 02 -32 -239274.5 -40.75 52518 52542†

 Aug 02 -32 -238577.5 -41.0 52501 52518

† Rate change in mid-month

†† Rate change one day early

*Provisional value
7

IMPORTANT NOTICE!

Effective January 1, 2004, NIST will discontinue sending the bulletin by mail.

The Time and Frequency Bulletin data are now online at

www.boulder.nist.gov/timefreq/pubs/bulletin/timescaleindex.htm

7. SPECIAL ANNOUNCEMENTS 

NOTICE TO DISCONTINUE INVOLVEMENT WITH GOES TIME CODE SERVICE

NIST has announced that it will discontinue its involvement with the time code broadcast from the GOES WEST and 

GOES EAST satellites operated by the National Oceanic and Atmosphere Administration (NOAA) on January 1, 2005. 

This decision has been jointly made by NIST and NOAA in response to the fact that nearly all users requiring time more 

accurate than 1 millisecond now use the Global Positioning System (GPS), and as a result, commercial sources for 

GOES timing receivers no longer exist.

NOAA is expected to continue to provide a GOES time code indefinitely after January 1, 2005, and existing receivers 

should be able to continue to receive and decode the time signal. However, the time code will no longer be controlled 

and checked by NIST, and the received time is expected to be less accurate when NIST discontinues its involvement. 

The GOES satellites currently broadcast continuously updated position information in addition to the time, so that GOES 

receivers can automatically correct for path delay changes caused by satellite motion. This allows the current system to 

have a time uncertainty of less than 100 microseconds. NOAA is expected to continuously broadcast a fixed position 

from the satellites, which could increase the time uncertainty to 1 millisecond or more. 

The GOES time broadcasts began in 1974 and have served many applications and thousands of users. NIST will 

continue to control and monitor the time code through January 1, 2005 to allow users who require a high accuracy 

signal sufficient time to replace their existing receivers. If you have additional questions, please contact Michael 

Lombardi, 303-497-3212, or email lombardi@boulder.nist.gov.