13339 Hagers Ferry Road

Huntersville, NC 28078-7929

McGuire Nuclear Complex - MG03A2

Phone: 980-875-5245 Fax: 980-875-4349

Order Summary Report

Analytical Laboratory

Order Number: J11080089

Project Name: WWTS - Biweekly

Lab Contact: Jason C Perkins

Date: 8/24/2011

Customer Address: 3195 Pine Hall Rd

Customer Name(s): Bill Kennedy, Melonie Martin, Wayne Chapman, Tom Johnson

Mailcode: Belews Steam Station

Belews Creek, NC 28012

Phone: 980-875-5348

Report Authorized By:

(Signature)

Program Comments:

FGD BiMonthly - LL Hg

Data Flags & Calculations:

Any analytical tests or individual analytes within a test flagged with a Qualifier indicate a deviation from the method quality

system or quality control requirement. The qualifier description is found at the end of the Certificate of Analysis (sample results) 

under the qualifiers heading. All results are reported on a dry weight basis unless otherwise noted.

Data Package:

This data package includes analytical results that are applicable only to the samples described in this narrative. An estimation of 

the uncertainty of measurement for the results in the report is available upon request. This report shall not be reproduced, except 

in full, without the written consent of the Analytical Laboratory. Please contact the Analytical laboratory with any questions. The 

order of individual sections within this report is as follows:

Job Summary Report, Sample Identification, Technical Validation of Data Package, Analytical Laboratory Certificate of Analysis,

Analytical Laboratory QC Reports, Sub-contracted Laboratory Results, Customer Specific Data Sheets, Reports & 

Documentation, Customer Database Entries, Test Case Narratives, Chain of Custody (COC)

Certification:

The Analytical Laboratory holds the following State Certifications : North Carolina (DENR) Certificate #248, South Carolina 

(DHEC) Laboratory ID # 99005. Contact the Analytical Laboratory for definitive information about the certification status of 

specific methods. 

Analytical Lab 

Page 1 of 29
Sample ID's & Descriptions:

Sample ID Plant/Station

Collection 

Date and Time Collected By Sample Description

2011016915 BELEWS 10-Aug-11 8:00 AM T. Johnson FGD Purge Eff 

2011016916 BELEWS 10-Aug-11 8:00 AM T. Johnson EQ TANK EFF. 

2011016917 BELEWS 10-Aug-11 8:00 AM T. Johnson BIOREACTOR 1 INF. 

2011016918 BELEWS 10-Aug-11 8:00 AM T. Johnson BIOREACTOR 2 INF. 

2011016919 BELEWS 10-Aug-11 8:00 AM T. Johnson BIOREACTOR 2 EFF. 

2011016920 BELEWS 02-Aug-11 2:15 PM L.DAVIS FILTER BLANK 

2011016921 BELEWS 02-Aug-11 2:15 PM L.DAVIS Trip Blank 

2011016934 BELEWS 10-Aug-11 12:20 PM David Morris (Prism) BIOREACTOR 1 INF. 

2011016935 BELEWS 02-Aug-11 2:15 PM L.DAVIS HG BLANK BIOREACTOR 1 INF. 

2011016936 BELEWS 10-Aug-11 12:30 PM David Morris (Prism) BIOREACTOR 2 INF. 

2011016937 BELEWS 02-Aug-11 2:15 PM L.DAVIS Hg Blk BioReactor 2 Inf 

2011016938 BELEWS 10-Aug-11 12:25 PM David Morris (Prism) BIOREACTOR 2 EFF. 

2011016939 BELEWS 02-Aug-11 2:15 PM L.DAVIS Hg Blk BioReactor 2 Eff 

13 Total Samples

Analytical Lab 

Page 2 of 29
COC and .pdf report are in agreement with sample totals 

and analyses (compliance programs and procedures).

All Results are less than the laboratory reporting limits.

All laboratory QA/QC requirements are acceptable.

The Vendor Laboratories have been qualified by the 

Analytical Laboratory

Yes No

Technical Validation Review

Checklist:

Yes No

Yes No

Report Sections Included:

Job Summary Report Sub-contracted Laboratory Results

Sample Identification Customer Specific Data Sheets, Reports, & Documentation

Technical Validation of Data Package Customer Database Entries

Analytical Laboratory Certificate of Analysis

Analytical Laboratory QC Report

Chain of Custody

Reviewed By: Mary Ann Ogle Date: 8/24/2011

Yes

Electronic Data Deliverable (EDD) Sent Separately

Analytical Lab 

Page 3 of 29
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J11080089

2011016915

Collection Date: 10-Aug-11 8:00 AM

Site: FGD Purge Eff 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

INORGANIC IONS BY IC

Bromide 97 mg/L EPA 300.0 11-Aug-11 20:40 CLEEMAN 10

MERCURY (COLD VAPOR) IN WATER

Mercury (Hg) 244 ug/L EPA 245.1 12-Aug-11 09:37 AGIBBS 5

TOTAL RECOVERABLE METALS BY ICP

Boron (B) 168 mg/L EPA 200.7 15-Aug-11 12:28 DJSULL1 0.5

DISSOLVED METALS BY ICP-MS

Selenium (Se) 220 ug/L EPA 200.8 15-Aug-11 12:11 KRICHAR 10

TOTAL RECOVERABLE METALS BY ICP-MS

Arsenic (As) 303 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

Chromium (Cr) 329 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

Copper (Cu) 259 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

Nickel (Ni) 260 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

Selenium (Se) 7470 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 20

Silver (Ag) < 10 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

Zinc (Zn) 466 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 20

SELENIUM SPECIATION

Vendor Parameter Complete V_AS&C

TOTAL DISSOLVED SOLIDS

TDS 14000 mg/L SM2540C 12-Aug-11 14:30 TJA7067 10

2011016916

Collection Date: 10-Aug-11 8:00 AM

Site: EQ TANK EFF. 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

MERCURY (COLD VAPOR) IN WATER

Mercury (Hg) 317 ug/L EPA 245.1 12-Aug-11 09:45 AGIBBS 5

TOTAL RECOVERABLE METALS BY ICP

Boron (B) 179 mg/L EPA 200.7 15-Aug-11 12:31 DJSULL1 0.5

DISSOLVED METALS BY ICP-MS

Selenium (Se) 326 ug/L EPA 200.8 15-Aug-11 12:14 KRICHAR 10

TOTAL RECOVERABLE METALS BY ICP-MS

Arsenic (As) 345 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

Analytical Lab 

Page 4 of 29
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J11080089

2011016916

Collection Date: 10-Aug-11 8:00 AM

Site: EQ TANK EFF. 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

TOTAL RECOVERABLE METALS BY ICP-MS

Chromium (Cr) 412 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

Copper (Cu) 315 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

Nickel (Ni) 295 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

Selenium (Se) 9530 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 20

Silver (Ag) < 10 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

Zinc (Zn) 541 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 20

2011016917

Collection Date: 10-Aug-11 8:00 AM

Site: BIOREACTOR 1 INF. 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

TOTAL RECOVERABLE METALS BY ICP

Boron (B) 167 mg/L EPA 200.7 15-Aug-11 12:35 DJSULL1 0.5

DISSOLVED METALS BY ICP-MS

Selenium (Se) 210 ug/L EPA 200.8 15-Aug-11 12:17 KRICHAR 5

TOTAL RECOVERABLE METALS BY ICP-MS

Arsenic (As) < 10 ug/L EPA 200.8 17-Aug-11 10:43 MHH7131 10

Chromium (Cr) < 10 ug/L EPA 200.8 17-Aug-11 10:43 MHH7131 10

Copper (Cu) < 10 ug/L EPA 200.8 17-Aug-11 10:43 MHH7131 10

Nickel (Ni) < 10 ug/L EPA 200.8 17-Aug-11 10:43 MHH7131 10

Selenium (Se) 242 ug/L EPA 200.8 17-Aug-11 10:43 MHH7131 10

Silver (Ag) < 10 ug/L EPA 200.8 17-Aug-11 10:43 MHH7131 10

Zinc (Zn) < 20 ug/L EPA 200.8 17-Aug-11 10:43 MHH7131 20

SELENIUM SPECIATION

Vendor Parameter Complete V_AS&C

2011016918

Collection Date: 10-Aug-11 8:00 AM

Site: BIOREACTOR 2 INF. 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

TOTAL RECOVERABLE METALS BY ICP

Boron (B) 168 mg/L EPA 200.7 15-Aug-11 12:39 DJSULL1 0.5

TOTAL RECOVERABLE METALS BY ICP-MS

Arsenic (As) < 10 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

Chromium (Cr) < 10 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

Copper (Cu) < 10 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

Nickel (Ni) < 10 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

Analytical Lab 

Page 5 of 29
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J11080089

2011016918

Collection Date: 10-Aug-11 8:00 AM

Site: BIOREACTOR 2 INF. 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

TOTAL RECOVERABLE METALS BY ICP-MS

Selenium (Se) 52.1 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

Silver (Ag) < 10 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

Zinc (Zn) < 20 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 20

2011016919

Collection Date: 10-Aug-11 8:00 AM

Site: BIOREACTOR 2 EFF. 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

INORGANIC IONS BY IC

Bromide 95 mg/L EPA 300.0 11-Aug-11 20:55 CLEEMAN 10

MERCURY (COLD VAPOR) IN WATER

Mercury (Hg) < 1 ug/L EPA 245.1 12-Aug-11 09:47 AGIBBS 1

TOTAL RECOVERABLE METALS BY ICP

Boron (B) 173 mg/L EPA 200.7 15-Aug-11 12:43 DJSULL1 0.5

TOTAL RECOVERABLE METALS BY ICP-MS

Arsenic (As) < 5 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 5

Chromium (Cr) < 5 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 5

Copper (Cu) < 5 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 5

Nickel (Ni) < 5 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 5

Selenium (Se) 5.01 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 5

Silver (Ag) < 5 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 5

Zinc (Zn) < 10 ug/L EPA 200.8 17-Aug-11 10:46 MHH7131 10

SELENIUM SPECIATION

Vendor Parameter Complete V_AS&C

2011016920

Collection Date: 02-Aug-11 2:15 PM

Site: FILTER BLANK 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

DISSOLVED METALS BY ICP-MS

Selenium (Se) < 1 ug/L EPA 200.8 15-Aug-11 11:51 KRICHAR 1

2011016921

Collection Date: 02-Aug-11 2:15 PM

Site: Trip Blank 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

Analytical Lab 

Page 6 of 29
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J11080089

2011016921

Collection Date: 02-Aug-11 2:15 PM

Site: Trip Blank 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

TOTAL RECOVERABLE METALS BY ICP

Boron (B) < 0.05 mg/L EPA 200.7 15-Aug-11 12:24 DJSULL1 0.05

TOTAL RECOVERABLE METALS BY ICP-MS

Arsenic (As) < 1 ug/L EPA 200.8 17-Aug-11 10:27 MHH7131 1

Chromium (Cr) < 1 ug/L EPA 200.8 17-Aug-11 10:27 MHH7131 1

Copper (Cu) < 1 ug/L EPA 200.8 17-Aug-11 10:27 MHH7131 1

Nickel (Ni) < 1 ug/L EPA 200.8 17-Aug-11 10:27 MHH7131 1

Selenium (Se) < 1 ug/L EPA 200.8 17-Aug-11 10:27 MHH7131 1

Silver (Ag) < 1 ug/L EPA 200.8 17-Aug-11 10:27 MHH7131 1

Zinc (Zn) < 2 ug/L EPA 200.8 17-Aug-11 10:27 MHH7131 2

SELENIUM SPECIATION

Vendor Parameter Complete V_AS&C

2011016934

Collection Date: 10-Aug-11 12:20 PM

Site: BIOREACTOR 1 INF. 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

MERCURY 1631

Vendor Parameter Complete V_BRAND

2011016935

Collection Date: 02-Aug-11 2:15 PM

Site: HG BLANK BIOREACTOR 1 INF. 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

MERCURY 1631

Vendor Parameter Complete V_BRAND

2011016936

Collection Date: 10-Aug-11 12:30 PM

Site: BIOREACTOR 2 INF. 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

MERCURY 1631

Vendor Parameter Complete V_BRAND

2011016937

Collection Date: 02-Aug-11 2:15 PM

Site: Hg Blk BioReactor 2 Inf 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

Analytical Lab 

Page 7 of 29
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J11080089

2011016937

Collection Date: 02-Aug-11 2:15 PM

Site: Hg Blk BioReactor 2 Inf 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

MERCURY 1631

Vendor Parameter Complete V_BRAND

2011016938

Collection Date: 10-Aug-11 12:25 PM

Site: BIOREACTOR 2 EFF. 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

MERCURY 1631

Vendor Parameter Complete V_BRAND

2011016939

Collection Date: 02-Aug-11 2:15 PM

Site: Hg Blk BioReactor 2 Eff 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

Analyst

MERCURY 1631

Vendor Parameter Complete V_BRAND

Analytical Lab 

Page 8 of 29
 

August 22, 2011 

Jay Perkins 

Duke Energy Analytical Laboratory 

Mail Code MGO3A2 (Building 7405) 

13339 Hagers Ferry Rd. 

Huntersville, NC 28078 

(704) 875-5245 

Project: Belews – FGD WWTS (Bi-Monthly Sampling) (LIMS #J11080089) 

Dear Mr. Perkins, 

Attached is the report associated with four (4) aqueous samples submitted for selenium 

speciation analysis on August 15, 2011. The samples were received in a sealed cooler at -0.4

oC 

on August 16, 2011. Selenium speciation analysis was performed via ion chromatography 

inductively coupled plasma dynamic reaction cell mass spectrometry (IC-ICP-DRC-MS). Any 

issues associated with the analysis are addressed in the following report. 

If you have any questions, please feel free to contact me at your convenience. 

 Sincerely, 

 

 Ben Wozniak

 Project Manager 

 Applied Speciation and Consulting, LLC 

 

Analytical Lab 

Page 9 of 29
Applied Speciation and Consulting, LLC 

Report prepared for: 

Jay Perkins 

Duke Energy Analytical Laboratory 

Mail Code MGO3A2 (Building 7405) 

13339 Hagers Ferry Rd. 

Huntersville, NC 28078 

Project: Belews – FGD WWTS (Bi-Monthly Sampling) (LIMS #J11080089) 

August 22, 2011 

1. Sample Reception 

Four (4) aqueous samples in 125mL HDPE bottles (provided by Applied Speciation and 

Consulting) were submitted for selenium speciation analysis on August 15, 2011. The 

samples were received on August 16, 2011 in a sealed container at -0.4

oC. 

The samples were received in a laminar flow clean hood void of trace metals contamination 

and ultra-violet radiation. Upon reception, the samples were designated discrete sample 

identifiers. An aliquot of each sample was filtered (0.45μm) and these filtrates were stored in 

a secure, monitored cryofreezer (maintained at a temperature of -80oC) until selenium 

speciation analysis could be performed via ion chromatography inductively coupled plasma 

dynamic reaction cell mass spectrometry (IC-ICP-DRC-MS). 

2. Sample Preparation 

All sample preparation is performed in laminar flow clean hoods known to be free from trace 

metals contamination. All applied water for dilutions and sample preservatives are 

monitored for contamination to account for any biases associated with the sample results. 

Selenium Speciation Analysis by IC-ICP-DRC-MS Prior to analysis, an aliquot of each 

sample was filtered with a syringe filter (0.45μm) and injected directly into a sealed 

autosampler vial. No further sample preparation was performed as any chemical alteration of 

the samples may shift the equilibrium of the system resulting in changes in speciation ratios. 

3. Sample Analysis 

All sample analysis is preceded by a minimum of a five-point calibration curve spanning the 

entire concentration range of interest. Calibration curves are performed at the beginning of 

Analytical Lab 

Page 10 of 29
each analytical day. All calibration curves, associated with each species of interest, are 

standardized by linear regression resulting in a response factor. All sample results are 

instrument blank corrected to account for any operational biases associated with the 

analytical platform. 

Prior to sample analysis, all calibration curves are verified using second source standards 

which are identified as initial calibration verification standards (ICV). 

Ongoing instrument performance is identified by the analysis of continuing calibration 

verification standards (CCV) and continuing calibration blanks (CCB) at a minimal interval 

of every ten analytical runs. 

Selenium Speciation Analysis by IC-ICP-DRC-MS All samples for selenium speciation 

analysis were analyzed by ion chromatography inductively coupled plasma dynamic reaction 

cell mass spectrometry (IC-ICP-DRC-MS) on August 17, 2011. An aliquot of each sample is

injected onto an anion exchange column and mobilized by a basic (pH > 7) gradient. The 

eluting selenium species are then introduced into a radio frequency (RF) plasma where 

energy-transfer processes cause desolvation, atomization, and ionization. The ions are 

extracted from the plasma through a differentially-pumped vacuum interface and travel 

through a pressurized chamber (DRC) containing a specific reactive gas which preferentially 

reacts with interfering ions of the same target mass to charge ratios (m/z). A solid-state 

detector detects ions transmitted through the mass analyzer and the resulting current is 

processed by a data handling system. 

Retention times for each eluting species are compared to known standards for species 

identification. 

4. Analytical Issues 

The overall analyses went well and no significant analytical issues were encountered. All 

quality control parameters associated with these samples were within acceptance limits.

The estimated method detection limits (eMDLs) for selenite, selenate, and selenocyanate are 

generated from replicate analyses of the lowest standard in the calibration curve. Not all 

selenium species are present in preparation blanks; therefore, eMDL calculations based on 

preparation blanks are artificially biased low. 

The eMDL for methylseleninic acid and selenomethionine is calculated from the average 

eMDL of selenite, selenate, and selenocyanate. The calibration does not contain 

methylseleninic acid or selenomethionine due to impurities in these standards which would 

bias the results for other selenium species. 

Analytical Lab 

Page 11 of 29
If you have any questions or concerns regarding this report, please feel free to contact me. 

Sincerely, 

Ben Wozniak 

Project Manager 

Applied Speciation and Consulting, LLC 

Analytical Lab 

Page 12 of 29
Sample Results

Sample ID Se(IV) Se(VI) SeCN MeSe(IV) SeMe

Unknown Se 

Species (n)

FGD Purge Eff 203 138 ND (<3.9) ND (<4.6) ND (<4.6) 0 (0)

BioReactor 1 Inf 99.6 122 3.76 2.7 ND (<1.1) 5.4 (2)

BioReactor 2 Eff ND (<1.6) ND (<0.90) ND (<0.98) ND (<1.1) ND (<1.1) 0 (0)

Metals Trip Blk ND (<0.31) ND (<0.18) ND (<0.20) ND (<0.23) ND (<0.23) 0 (0)

All results reflect the applied dilution and are reported in μg/L

ND = Not detected at the applied dilution

SeCN = Selenocyanate

MeSe(IV) = Methylseleninic acid

SeMe = Selenomethionine

Unknown Se Species = Total concentration of all unknown Se species observed by IC-ICP-MS

n = number of unknown Se species observed

Report Generated by: Ben Wozniak

Applied Speciation and Consulting, LLC

Selenium Speciation Results for Duke Energy

Project Name: Belews – FGD WWTS (Bi-Monthly Sampling) 

Contact: Jay Perkins

Date: August 22, 2011

LIMS #J11080089

Analytical Lab 

Page 13 of 29
Quality Control Summary - Preparation Blank Summary

Analyte (μg/L) PBW1 PBW2 PBW3 PBW4 Mean StdDev eMDL* eMDL 10x eMDL 50x eMDL 200x

Se(IV) 0.000 0.000 0.000 0.000 0.000 0.000 0.031 0.31 1.6 6.2

Se(VI) 0.000 0.000 0.000 0.000 0.000 0.000 0.018 0.18 0.90 3.6

SeCN 0.000 0.000 0.000 0.000 0.000 0.000 0.020 0.20 0.98 3.9

MeSe(IV) 0.000 0.000 0.000 0.000 0.000 0.000 0.023 0.23 1.1 4.6

SeMe 0.000 0.000 0.000 0.000 0.000 0.000 0.023 0.23 1.1 4.6

eMDL = Estimated Method Detection Limit

*Please see narrative regarding eMDL calculations

Quality Control Summary - Certified Reference Materials

Analyte (μg/L) CRM True Value Result Recovery

Se(IV) LCS 9.57 9.759 102.0

Se(VI) LCS 9.48 9.480 100.0

SeCN LCS 8.92 8.929 100.1

MeSe(IV) LCS 6.47 6.625 102.4

SeMe LCS 9.32 9.408 100.9

Selenium Speciation Results for Duke Energy

Project Name: Belews – FGD WWTS (Bi-Monthly Sampling) 

Contact: Jay Perkins

LIMS #J11080089

Date: August 22, 2011

Report Generated by: Ben Wozniak

Applied Speciation and Consulting, LLC

Analytical Lab 

Page 14 of 29
Quality Control Summary - Matrix Duplicates

Analyte (μg/L) Sample ID Rep 1 Rep 2 Mean RPD

Se(IV) BioReactor 2 Eff ND (<1.6) ND (<1.6) NC NC

Se(VI) BioReactor 2 Eff ND (<0.90) ND (<0.90) NC NC

SeCN BioReactor 2 Eff ND (<0.98) ND (<0.98) NC NC

MeSe(IV) BioReactor 2 Eff ND (<1.1) ND (<1.1) NC NC

SeMe BioReactor 2 Eff ND (<1.1) ND (<1.1) NC NC

ND = Not detected at the applied dilution

NC = Value was not calculated due to one or more concentrations below the eMDL

Quality Control Summary - Matrix Spike/ Matrix Spike Duplicate

Analyte (μg/L) Sample ID Spike Conc MS Result Recovery Spike Conc MSD Result Recovery RPD

Se(IV) BioReactor 2 Eff 278.0 275.8 99.2 278.0 274.2 98.6 0.6

Se(VI) BioReactor 2 Eff 252.3 264.5 104.9 252.3 265.8 105.4 0.5

SeCN BioReactor 2 Eff 228.8 223.7 97.8 228.8 225.1 98.4 0.6

Applied Speciation and Consulting, LLC

Selenium Speciation Results for Duke Energy

Project Name: Belews – FGD WWTS (Bi-Monthly Sampling) 

Contact: Jay Perkins

LIMS #J11080089

Date: August 22, 2011

Report Generated by: Ben Wozniak

Analytical Lab 

Page 15 of 29
Analytical Lab 

Page 16 of 29
 

3958 6th Ave NW • Seattle, WA 98107 • T: 206-632-6206 • F: 206-632-6017 • www.brooksrand.com • brl@brooksrand.com 

August 23, 2011 

Duke Energy 

ATTN: Jay Perkins 

Scientific Support-Laboratory

13339 Hagers Ferry Road 

Huntersville NC 28078 

jcperkins@duke-energy.com 

labcustomer@duke-energy.com 

RE: Project DUK-HV1101 Client Project: J11080089 

Dear Mr. Perkins, 

On August 11, 2011, Brooks Rand Labs (BRL) received three (3) flue gas desulfurization (FGD) 

waste water samples and three (3) corresponding field blanks. Samples were logged-in for total 

mercury (Hg) analysis. All samples were received, prepared, analyzed, and stored according to 

BRL SOPs and EPA methodology. 

The results were blank-corrected as described in the calculations section of the applicable 

SOP(s) and may be evaluated using adjusted reporting limits to account for sample aliquot size. 

Please refer to the Sample Results page for sample-specific detection limits and other details. 

All other quality assurance criteria were satisfied and, aside from concentration qualifiers, all 

data was reported without additional qualification. 

BRL, an accredited laboratory, certifies the reported results of all analyses for which BRL is 

NELAP accredited meet all NELAP requirements. For more details, see the Report Information

page of the report. 

Please feel free to contact me if you have any questions regarding this report. 

Sincerely, 

 

Tiffany Stilwater 

Project Manager 

tiffany@brooksran.com 

Analytical Lab 

Page 17 of 29
Project ID: DUK-HV1101

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Definition of Data Qualifiers

(Effective 9/23/09)

Laboratory Accreditation

BRL is accredited by the National Environmental Laboratory Accreditation Program (NELAP) through the State of Florida 

Department of Health, Bureau of Laboratories (E87982) and is certified to perform many environmental analyses. BRL is 

also certified by many other states to perform environmental analyses. For a current list of our accreditations /certifications, 

please visit our website at <http://www.brooksrand.com/default.asp?contentID=586>. Results reported relate only to the 

samples listed in the report.

Report Information

BLK

BRL

BS

CAL

CCV

CRM

D

DUP

ICV

MSD

ND

NR

PS

REC

RPD

RSD

SCV

SOP

method blank 

Brooks Rand Labs

laboratory fortified blank

calibration standard

continuing calibration verification

certified reference material

dissolved fraction

duplicate

initial calibration verification

matrix spike duplicate

non-detect

non-reportable

post preparation spike

percent recovery

relative percent difference

relative standard deviation

secondary calibration verification

standard operating procedure

MDL

MRL

MS

method detection limit

method reporting limit

matrix spike

SRM

T

COC

standard reference material

total recoverable fraction

chain of custody record 

Common Abbreviations

These qualifiers are based on those previously utilized by Brooks Rand, Ltd., those found in the EPA SOW ILM03.0, 

Exhibit B, Section III, pg. B-18, and the USEPA Laboratory Data Validation Functional Guidelines for Evaluating Inorganic 

Analyses; USEPA; July 2002. These supersede all previous qualifiers ever employed by BRL.

B Detected by the instrument, the result is > the MDL but ≤ the MRL. Result is reported and considered an estimate.

E An estimated value due to the presence of interferences. A full explanation is presented in the narrative.

H Holding time and/or preservation requirements not met. Result is estimated.

J Estimated value. A full explanation is presented in the narrative.

J-M Duplicate precision (RPD) for associated QC sample was not within acceptance criteria. Result is estimated.

J-N Spike recovery for associated QC sample was not within acceptance criteria. Result is estimated.

M Duplicate precision (RPD) was not within acceptance criteria. Result is estimated.

N Spike recovery was not within acceptance criteria. Result is estimated.

R Rejected, unusable value. A full explanation is presented in the narrative.

U Result is ≤ the MDL or client requested reporting limit (CRRL). Result reported as the MDL or CRRL.

X Result is not BLK-corrected and is within 10x the absolute value of the highest detectable BLK in the batch. 

Result is estimated.

Field Quality Control Samples

Please be notified that certain EPA methods require the collection of field quality control samples of an appropriate type 

and frequency; failure to do so is considered a deviation from some methods and for compliance purposes should only be 

done with the approval of regulatory authorities. Please see the specific EPA methods for details regarding required field 

quality control samples.

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Analytical Lab 

Page 18 of 29
Project ID: DUK-HV1101

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Sample Information

Sample Lab ID Report Matrix Type Sampled Received

BioReactor 1 Inf 1134009-01 FGD Wastewater Sample 08/10/2011 08/16/2011

Hg Blk BioReactor 1 Inf 1134009-02 DIW Field Blank 08/02/2011 08/16/2011

BioReactor 2 Inf 1134009-03 FGD Wastewater Sample 08/10/2011 08/16/2011

Hg Blk BioReactor 2 Inf 1134009-04 DIW Field Blank 08/02/2011 08/16/2011

BioReactor 2 Eff 1134009-05 FGD Wastewater QC Sample 08/10/2011 08/16/2011

Hg Blk BioReactor 2 Eff 1134009-06 DIW Field Blank 08/02/2011 08/16/2011

Batch Summary

Analyte Lab Matrix Method Prepared Analyzed Batch Sequence

Hg Water EPA 1631 08/18/2011 08/22/2011 B111247 1100569

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Analytical Lab 

Page 19 of 29
Project ID: DUK-HV1101

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Sample Results

Sample Analyte Report Matrix Fraction Result Qualifier MDL MRL Unit Batch Sequence

BioReactor 1 Inf

1134009-01 Hg FGD Wastewater T 60.2 3.03 8.08 ng/L B111247 1100569

BioReactor 2 Eff

1134009-05 Hg FGD Wastewater T 31.1 0.78 2.07 ng/L B111247 1100569

BioReactor 2 Inf

1134009-03 Hg FGD Wastewater T 49.9 3.03 8.08 ng/L B111247 1100569

Hg Blk BioReactor 1 Inf

1134009-02 Hg DIW T 0.15 U 0.15 0.40 ng/L B111247 1100569

Hg Blk BioReactor 2 Eff

1134009-06 Hg DIW T 0.15 U 0.15 0.41 ng/L B111247 1100569

Hg Blk BioReactor 2 Inf

1134009-04 Hg DIW T 0.15 U 0.15 0.41 ng/L B111247 1100569

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Analytical Lab 

Page 20 of 29
Project ID: DUK-HV1101

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Accuracy & Precision Summary

Analyte Native Spike Result Units REC & Limits RPD & Limits

Batch: B111247

Method: EPA 1631

Lab Matrix: Water

Sample

B111247-SRM1 Certified Reference Material (1133002, NIST 1641d 1000x dilution)

Hg 15.68 14.68 ng/L 94% 85-115

B111247-MS1 Matrix Spike (1134009-05)

Hg 31.14 157.1 214.1 ng/L 116% 71-125

B111247-MSD1 Matrix Spike Duplicate (1134009-05)

Hg 31.14 153.3 204.7 ng/L 113% 71-125 245%

Method Blanks & Reporting Limits

Batch: B111247

Method: EPA 1631

Matrix: Water

Analyte: Hg

Sample Result Units

B111247-BLK1 0.10 ng/L

B111247-BLK2 0.05 ng/L

B111247-BLK3 0.08 ng/L

B111247-BLK4 0.06 ng/L

Average: 0.07 Standard Deviation: 0.02 MDL: 0.15

Limit: 0.50 Limit: 0.10 MRL: 0.41

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Analytical Lab 

Page 21 of 29
Project ID: DUK-HV1101

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Instrument Calibration

Method: EPA 1631

Sequence: 1100569

Instrument: THG-05

Date: 08/22/2011

Total Mercury and Mercury Speciation by CVAFS

Analyte: Hg

Lab ID True Value Result Units REC & Limits

1100569-IBL1 6.42 pg of Hg

1100569-IBL2 8.76 pg of Hg

1100569-IBL3 10.70 pg of Hg

1100569-IBL4 11.08 pg of Hg

1100569-CAL1 25.00 25.81 pg of Hg 103%

1100569-CAL2 100.0 92.73 pg of Hg 93%

1100569-CAL3 500.0 517.2 pg of Hg 103%

1100569-CAL4 2500 2552 pg of Hg 102%

1100569-CAL5 10000 9933 pg of Hg 99%

1100569-ICV1 1568 1468 pg of Hg 94% 85-115

1100569-CCB1 15.8 pg of Hg

1100569-CCV1 500.0 524.8 pg of Hg 105% 77-123

1100569-CCV2 500.0 514.0 pg of Hg 103% 77-123

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Analytical Lab 

Page 22 of 29
Project ID: DUK-HV1101

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Sample Containers

Lab ID: 1134009-01 Report Matrix: FGD Wastewater

Sample: BioReactor 1 Inf Sample Type: Sample Received: 08/16/2011

Collected: 08/10/2011

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71313080

60

Cardboard 

Box

Bottle FLPE Hg-T 125 mL None N/A

Lab ID: 1134009-02 Report Matrix: DIW

Sample: Hg Blk BioReactor 1 Inf Sample Type: Field Blank Received: 08/16/2011

Collected: 08/02/2011

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71313080

60

Cardboard 

Box

Bottle FLPE Hg-T 125 mL 0.1% HCl (BRL) 1121032 <2

Lab ID: 1134009-03 Report Matrix: FGD Wastewater

Sample: BioReactor 2 Inf Sample Type: Sample Received: 08/16/2011

Collected: 08/10/2011

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71313080

60

Cardboard 

Box

Bottle FLPE Hg-T 250 mL None N/A

Lab ID: 1134009-04 Report Matrix: DIW

Sample: Hg Blk BioReactor 2 Inf Sample Type: Field Blank Received: 08/16/2011

Collected: 08/02/2011

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71313080

60

Cardboard 

Box

Bottle FLPE Hg-T 125 mL 0.1% HCl (BRL) 1121032 <2

Lab ID: 1134009-05 Report Matrix: FGD Wastewater

Sample: BioReactor 2 Eff Sample Type: QC Sample Received: 08/16/2011

Collected: 08/10/2011

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71313080

60

Cardboard 

Box

Bottle FLPE Hg-T 250 mL None N/A

Lab ID: 1134009-06 Report Matrix: DIW

Sample: Hg Blk BioReactor 2 Eff Sample Type: Field Blank Received: 08/16/2011

Collected: 08/02/2011

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71313080

60

Cardboard 

Box

Bottle FLPE Hg-T 125 mL 0.1% HCl (BRL) 1121032 <2

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Analytical Lab 

Page 23 of 29
Project ID: DUK-HV1101

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Shipping Containers

Cardboard Box

Tracking No: 4726 7966 2932 via FedEx

Temperature: ambient

Coolant Type: None

Description: Cardboard Box

Damaged in transit? No

Returned to client? No

Custody seals present? No

Custody seals intact? No

COC present? Yes

Received: August 16, 2011 9:00

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Analytical Lab 

Page 24 of 29
Analytical Lab 

Page 25 of 29
Analytical Lab 

Page 26 of 29
From: Perkins, Jay C

To: Chapman, Wayne C

Subject: Br

Date: Monday, August 15, 2011 1:55:00 PM

Attachments: image001.gif

FGD Purge EFF 97 mg/L

BioReactor 2 Eff 95 mg/L

Sampled: August 10

Wayne, this is preliminary data.

Thanks,

Jay Perkins

Duke Energy Analytical Laboratory

W-980-875-5348

C-704-796-6598

Analytical Lab 

Page 27 of 29
Analytical Lab 

Page 28 of 29
Analytical Lab 

Page 29 of 29