SW-846 Update V 6020B - 1 Revision 2 

October 2012 

METHOD 6020B 

INDUCTIVELY COUPLED PLASMA—MASS SPECTROMETRY

SW-846 is not intended to be an analytical training manual. Therefore, method procedures 

are written based on the assumption that they will be performed by analysts who are formally 

trained in at least the basic principles of chemical analysis and in the use of the subject 

technology. 

In addition, SW-846 methods, with the exception of required method use for the analysis of 

method-defined parameters, are intended to be guidance methods which contain general 

information on how to perform an analytical procedure or technique which a laboratory can use 

as a basic starting point for generating its own detailed standard operating procedure (SOP), 

either for its own general use or for a specific project application. The performance data 

included in this method are for guidance purposes only, and are not intended to be and must not 

be used as absolute quality control (QC) acceptance criteria for purposes of laboratory 

accreditation. 

1.0 SCOPE AND APPLICATION 

 1.1 Inductively coupled plasma-mass spectrometry (ICP-MS) is applicable to the 

determination of sub-μg/L concentrations of a large number of elements in water samples and in 

waste extracts or digests (Refs. 1 and 2). When dissolved constituents are required, samples 

must be filtered and acid-preserved prior to analysis. No digestion is required prior to analysis 

for dissolved elements in water samples. Acid digestion prior to filtration and analysis is required 

for groundwater, aqueous samples, industrial wastes, soils, sludges, sediments, and other solid 

wastes for which total (acid-leachable) elements are required. The analyst should insure that a 

sample digestion method is chosen that is appropriate for each analyte and the intended use of 

the data. Refer to Chapter Three for the appropriate digestion procedures. 

 1.2 ICP-MS has been applied to the determination of over 60 elements in various 

matrices. Analytes for which the acceptability of Method 6020 has been demonstrated through 

multi-laboratory testing on solid and aqueous wastes are listed below. 

Element Symbol CASRNa Element Symbol CASRNa 

Aluminum Al 7429-90-5 Magnesium Mg 7439-95-4 

Antimony Sb 7440-36-0 Manganese Mn 7439-96-5 

Arsenic As 7440-38-2 Mercury Hg 7439-97-6 

Barium Ba 7440-39-3 Nickel Ni 7440-02-0 

Beryllium Be 7440-41-7 Potassium K 7440-09-7 
SW-846 Update V 6020B - 2 Revision 2 

October 2012 

Element Symbol CASRNa Element Symbol CASRNa 

Cadmium Cd 7440-43-9 Selenium Se 7782-49-2 

Calcium Ca 7440-70-2 Silver Ag 7440-22-4 

Chromium Cr 7440-47-3 Sodium Na 7440-23-5 

Cobalt Co 7440-48-4 Thallium TI 7440-28-0 

Copper Cu 7440-50-8 Vanadium V 7440-62-2 

Iron Fe 7439-89-6 Zinc Zn 7440-66-6 

Lead Pb 7439-92-1 

 

aChemical Abstract Service Registry Number 

 The performance acceptability of ICP-MS for the determination of the listed elements was 

based upon comparison of the multi-laboratory testing results with those obtained from either 

furnace atomic absorption spectrophotometry or inductively coupled plasma—atomic emission 

spectrometry. It should be noted that one multi-laboratory study was conducted in 1988. As 

advances in ICP-MS instrumentation and software have been made since that time, other 

elements have been added through validation and with additional improvements in performance 

of the method. Performance, in general, presently exceeds the original multi-laboratory 

performance data for the listed elements (and others) that are provided in Sec. 13.0. Instrument 

detection limits (IDLs), lower limits of quantitation (LLOQs) and linear ranges will vary with the 

matrices, instrumentation, and operating conditions. In relatively simple matrices, IDLs will 

generally be < 0.1 μg/L. For less sensitive elements (e.g., Se and As) and desensitized major 

elements, IDLs may be ≥ 1.0 μg/L. 

 1.3 If Method 6020 is used to determine any analyte not listed in Sec. 1.2, it is the 

responsibility of the analyst to demonstrate the precision and bias of the method for the waste to 

be analyzed. The analyst must always monitor potential sources of interferences and take 

appropriate action to ensure data of known quality (see Sec. 9.0). Other elements and matrices 

may be analyzed by this method if performance is demonstrated for the analyte of interest, in the 

matrices of interest, at the concentration levels of interest in the same manner as the listed 

elements and matrices (see Sec. 9.0). 

 1.4 Use of this method should be restricted to spectroscopists who are knowledgeable 

in the recognition and correction of spectral, chemical, and physical interferences in ICP-MS 

analysis. 

 1.5 An appropriate internal standard is necessary for each analyte determined by 

ICP-MS. Recommended internal standards are 6

Li, 45Sc, 89Y, 103Rh, 115In, 159Tb, 165Ho, and 209Bi. 

The lithium internal standard should have an enriched abundance of 6

Li, so that interference 
SW-846 Update V 6020B - 3 Revision 2 

October 2012 

from lithium native to the sample is minimized. Other elements may need to be used as internal 

standards when samples contain significant native amounts of the recommended internal 

standards as indicated by high bias of internal standard recoveries. 

Note: Other potential causes of a high bias should also be considered before a final decision is 

made that the internal standard high bias is caused by an excessive concentration of the internal 

standard isotope in the sample. 

1.6 Prior to employing this method, analysts are advised to consult the preparatory 

method for each type of procedure that may be employed in the overall analysis (e.g., Methods 

3005, 3010, 3015, 3031, 3040, 3050, 3051, 3052, 7000, and 6800) for additional information on 

QC procedures, development of QC acceptance criteria, calculations, and general guidance. 

Analysts also should consult the disclaimer statement at the front of the manual and the 

information in Chapter Two for guidance on the intended flexibility in the choice of methods, 

apparatus, materials, reagents, and supplies, and on the responsibilities of the analyst for 

demonstrating that the techniques employed are appropriate for the analytes of interest, in the 

matrix of interest, and at the levels of concern. 

In addition, analysts and data users are advised that, except where explicitly specified in a 

regulation, the use of SW-846 methods is not mandatory in response to Federal testing 

requirements. The information contained in this method is provided by EPA as guidance to be 

used by the analyst and the regulated community in making judgments necessary to generate 

results that meet the data quality objectives (DQOs) for the intended application. 

 1.7 This method is restricted to use by, or under supervision of, properly experienced 

and trained personnel. Each analyst must demonstrate the ability to generate acceptable results 

with this method. 

2.0 SUMMARY OF METHOD 

 2.1 Prior to analysis, aqueous and solid samples are solubilized or digested using the 

appropriate sample preparation methods (see Chapter Three). When analyzing groundwater 

samples for dissolved constituents, acid digestion is not necessary, if the samples are filtered 

and acid-preserved prior to analysis (e.g., Methods 3005, 3010, 3015, 3031, 3050, 3051 and 

3052). For oils, greases, or waxes, use the solvent dissolution procedure in method 3040 to 

prepare the samples. 

 2.2 This method describes multi-element determinations using ICP-MS in 

environmental samples. The method measures ions produced by a radio-frequency inductively 

coupled plasma. Analyte species in liquid are nebulized and the resulting aerosol is transported 

by argon gas into the plasma torch. The ions produced by high temperatures are entrained in 

the plasma gas and introduced, by means of an interface, into a mass spectrometer. The ions 

produced in the plasma are sorted according to their mass-to-charge (m/z) ratios and quantified 

with a channel electron multiplier. Interferences must be assessed and valid corrections applied 

or the data flagged to indicate problems. Interference correction must include compensation for 

background ions contributed by the plasma gas, reagents, and constituents of the sample 

matrix. 
SW-846 Update V 6020B - 4 Revision 2 

October 2012 

3.0 DEFINITIONS 

 Refer to Chapter One, Chapter Three, and the manufacturer's instructions for definitions 

that may be relevant to this procedure. 

4.0 INTERFERENCES 

 4.1 Isobaric elemental interferences in ICP-MS are caused by isotopes of different 

elements forming atomic ions with the same nominal m/z ratio. A data system must be used to 

correct for these interferences. This involves determining the signal for another isotope of the 

interfering element and subtracting the appropriate signal from the analyte isotope signal. 

 4.2 Isobaric molecular and doubly charged ion interferences in ICP-MS are caused by 

ions consisting of more than one atom or charge, respectively. Most isobaric interferences that 

could affect ICP-MS determinations have been identified in the literature (Refs. 3 and 4). 

Examples include 75ArCl+

 ion on the 75As signal and MoO+ ions on the cadmium isotopes. While 

the approach used to correct for molecular isobaric interferences is demonstrated below using 

the natural isotope abundances from the literature (Ref. 5), the most precise coefficients for an 

instrument can be determined from the ratio of the net isotope signals observed for a standard 

solution of the interfering element at a concentration which produces sufficient interference at 

the isotopes of interest that a reliable measurement can be made. Because the 35Cl natural 

abundance of 75.77% is 3.13 times the 37Cl abundance of 24.23%, the chloride correction for

arsenic can be calculated (approximately) as follows (where the 38Ar37Cl+ contribution at m/z 75 

is a negligible 0.06% of the 40Ar35Cl+ signal): 

Corrected arsenic signal (using the abundances of natural isotopes 

for coefficient approximations) = 

(m/z 75 signal) - (3.13) [(m/z 77 signal) - (0.87) (m/z 82 signal)] 

 where, the final term adjusts for any selenium contribution at 77 m/z, 

NOTE: Arsenic values can be biased high by this type of equation when the net signal at m/z 82 

is caused by ions other than 82Se+

, (e.g., 81BrH+ from bromine wastes [Ref. 6]). 

NOTE: The coefficients should be verified experimentally using the procedures or coefficients 

provided by the instrument manufacturer. 

Similarly, 

Corrected cadmium signal (using the abundances of natural isotopes 

for coefficient approximations) = 

(m/z 114 signal) - (0.027)(m/z 118 signal) - (1.63)(m/z 108 signal) 
SW-846 Update V 6020B - 5 Revision 2 

October 2012 

 where, the last 2 terms adjust for any 114Sn+

 or 114MoO+ contributions at m/z 114. 

NOTE: Cadmium values will be biased low by this type of equation when 92ZrO+ ions contribute 

at m/z 108, but use of m/z 111 for Cd is even subject to direct (94ZrOH+) and indirect 

(

90ZrO+

) additive interferences when Zr is present. 

NOTE: With respect to the arsenic equation above, the coefficients could be improved. For 

example, the coefficient to modify “3.13” (in the equation above) for a particular 

instrument can be determined from the observed ratio of the m/z 75 to the m/z 77 net 

isotope signals for a solution of hydrochloric acid. The concentration of HCl used 

should provide enough signal at the measured isotopes to ensure that a reliable 

measurement can be made, while not exceeding the linear range of the detector. 

 The accuracy of these types of equations is based upon the constancy of the observed

isotopic ratios for the interfering species. Corrections that presume a constant fraction of a 

molecular ion relative to the "parent" ion have not been found (Ref. 7) to be reliable, e.g., oxide 

levels can vary with operating conditions. If a correction for an oxide ion is based upon the ratio 

of parent-to-oxide ion intensities, the correction must be adjusted for the degree of oxide 

formation by the use of an appropriate oxide internal standard previously demonstrated to form a 

similar level of oxide as the interferent. For example, this type of correction has been reported 

(Ref. 7) for oxide-ion corrections using ThO+/Th+ for the determination of rare earth elements. 

The use of aerosol desolvation and/or mixed gas plasmas have been shown to greatly reduce 

molecular interferences (Ref. 8). These techniques can be used, provided that IDL, bias, and 

precision specifications for analysis of the samples can be met. 

4.3 As technology continues to develop, modifications to existing ICP-MS 

instrumentation can reduce or completely remove common interferences thus eliminating the 

need for reliance on correction equations. Instruments must be able to demonstrate successful 

freedom from interferences. Examples of such modifications are discussed in more detail below: 

4.3.1 Recent ICP-MS instruments may include collision or reaction cells for 

removal of molecular isobaric interferences. This type of interference removal is effective, 

and highly recommended for complex and/or varying matrices. The systems work either by 

collision of molecular species with an inert gas (usually helium) or by reaction of molecular 

species or the target analyte with reactive gases (e.g., ammonia or methane). 

Manufacturer recommendations should be followed for the configuration of the 

collision/reaction cell. This technique may eliminate the need for most correction 

equations, but freedom from interference still needs to be demonstrated using the spectral 

interference check (SIC) solutions described in sections 7.23 and 9.9. 

4.3.2 High resolution ICP-MS instruments are available based on several mass 

analyzer designs with much higher mass resolution within the mass range of traditional 

ICP-MS instruments. These mass analyzers are not based on quadrupole mass analyzers 

and have orders of magnitude resolution above quadrupoles, which helps reduce or 

eliminate interference from polyatomic ions with the same nominal mass. These mass 

analyzers reduce or eliminate the need for most correction equations, but the instrument 

needs to be operated at sufficient resolution to remove the expected interference. For 
SW-846 Update V 6020B - 6 Revision 2 

October 2012 

example, resolving 52Cr from 40Ar12C requires a resolution of around 4000, while resolving 

75As from 40Ar35Cl requires a resolution of around 8000. Freedom from interferences needs 

to be demonstrated for the particular higher resolution mass analyzers ICP-MS. 

 4.4 Additionally, solid-phase chelation may be used to eliminate isobaric interferences 

from both element and molecular sources. An on-line method has been demonstrated for 

environmental waters such as sea water, drinking water and acid decomposed samples. Acid 

decomposed samples refer to samples decomposed by methods similar to methods 3052, 3051, 

3050 or 3015. Samples with% levels of iron and aluminum should be avoided. The method also 

provides a method for preconcentration to enhance detection limits simultaneously with 

elimination of isobaric interferences. The method relies on chelating resins such as 

imminodiacetate or other appropriate resins and selectively concentrates the elements of 

interest while eliminating interfering elements from the sample matrix. By eliminating the 

elements that are direct isobaric interferences or those that form isobaric interfering molecular 

masses, the mass region is simplified and these interferences cannot occur. The method has 

been proven effective for the certification of reference materials and validated using reference 

materials (Refs. 13-15). The method has the potential to be used on-line or off-line as an 

effective sample preparation method specifically designed to address interference problems. 

4.5 Since commercial quadrupole ICP-MS instruments nominally provide unit resolution 

at 10% of the peak height, very high ion currents at adjacent masses can also contribute to ion 

signals at the mass of interest. Although this type of interference is uncommon, it is not easily 

corrected, and samples exhibiting a significant problem of this type could need resolution 

improvement, matrix separation, or analysis using another verified and documented isotope, or 

otherwise the use of another method. 

 4.6 Physical interferences are associated with the sample nebulization and transport 

processes as well as with ion-transmission efficiencies. Nebulization and transport processes 

can be affected if a matrix component causes a change in surface tension or viscosity. Changes 

in matrix composition can cause significant signal suppression or enhancement (Ref. 9). 

Dissolved solids can deposit on the nebulizer tip of a pneumatic nebulizer and on the interface 

skimmers (reducing the orifice size and the instrument performance). Dissolved solid levels 

below 0.2% (2,000 mg/L) have been currently recommended (Ref. 10) to minimize solid 

deposition, although currently-available ICP-MS systems may be able to tolerate much higher 

levels. An internal standard can be used to correct for physical interferences, if it is carefully 

matched to the analyte so that the two elements are similarly affected by matrix changes (Ref. 

11). When intolerable physical interferences are present in a sample, a significant suppression 

of the internal standard signals (to less than 30% of the signals in the calibrations standard) will 

be observed. Dilution of the sample five-fold (i.e., dilute one part sample with four parts diluent 

[1:5 = 1+4]) will usually eliminate the problem. 

 4.7 Memory interferences or carry-over can occur when there are large concentration 

differences between samples or standards which are analyzed sequentially. Sample deposition 

on the sampler and skimmer cones, spray chamber design, and the type of nebulizer affect the 

extent of the memory interferences which are observed. The rinse period between samples 

must be long enough to eliminate significant memory interference. 

 4.8 Reagents and sample processing hardware may yield artifacts and/or interferences 

to sample analysis. All of these materials must be demonstrated to be free from interferences 
SW-846 Update V 6020B - 7 Revision 2 

October 2012 

under the conditions of the analysis by analyzing method blanks. Specific selection of reagents 

may be necessary. Refer to each method to be used for specific guidance on QC procedures. 

5.0 SAFETY 

 5.1 This method does not address all safety issues associated with its use. The 

laboratory is responsible for maintaining a safe work environment and a current awareness file 

of OSHA regulations regarding the safe handling of the chemicals specified in this method. A 

reference file of material safety data sheets (MSDSs) should be available to all personnel 

involved in these analyses. 

 5.2 Concentrated nitric and hydrochloric acids are moderately toxic and extremely 

irritating to skin and mucus membranes. Use these reagents in a hood and if eye or skin contact 

occurs, flush with large volumes of water. Always wear safety glasses or a shield for eye 

protection when working with these reagents. 

 5.3 Hydrofluoric acid is a very toxic acid and penetrates the skin and tissues 

deeply if not treated immediately. Injury occurs in two stages: firstly, by hydration that 

induces tissue necrosis; and secondly, by penetration of fluoride ions deep into the tissue and 

thereby reacting with calcium. Boric acid and/or other complexing reagents and appropriate 

treatment agents should be administered immediately. 

WARNING: Consult appropriate safety literature for determining the proper protective eyewear, 

clothing and gloves to use when handling hydrofluoric acid. Always have 

appropriate treatment materials readily available prior to working with this 

acid. See Method 3052 for additional recommendations for handling hydrofluoric 

acid from a safety and an instrument standpoint. 

 5.4 Many metal salts, are extremely toxic if inhaled or swallowed. 

WARNING: Exercise extreme care to ensure that samples and standards are handled safely 

and properly and that all exhaust gases are properly vented. Wash hands 

thoroughly after handling. 

6.0 EQUIPMENT AND SUPPLIES 

 6.1 Inductively coupled plasma-mass spectrometer: 

 6.1.1 The system must be capable of providing resolution, better than or equal 

to 1.0 u (unified atomic mass unit) at 10% peak height. The system must have a mass 

range from at least 6 to 240 u and a data system that allows corrections for isobaric 

interferences and the application of the internal standard technique. Use of a mass-flow 

controller for the nebulizer argon and a peristaltic pump for the sample solution are 

recommended. 

 6.1.2 Argon gas, high-purity grade (99.99%). 
SW-846 Update V 6020B - 8 Revision 2 

October 2012 

 6.2 Volumetric flasks of suitable material composition, precision and accuracy 

 6.3 Volumetric pipets of suitable material composition, precision and accuracy 

 This section does not list all common laboratory ware (e.g., beakers) that might be used. 

7.0 REAGENTS AND STANDARDS 

 7.1 Reagent-grade, and whenever necessary, ultra-high purity-grade chemicals, must 

be used in all tests. Unless otherwise indicated, it is intended that all reagents conform to the 

specifications of the Committee on Analytical Reagents of the American Chemical Society, 

where such specifications are available. Other grades may be used, provided it is first 

ascertained that the reagent is of sufficiently high purity to permit its use without lessening the 

accuracy of the determination. 

 7.2 Reagent water - Reagent water must be interference free. All references to water 

in this method refer to reagent water unless otherwise specified. 

 7.3 Ultra high-purity or equivalent acids must be used in the preparation of standards 

and for sample processing. Redistilled acids are recommended because of the high sensitivity 

of ICP-MS. Nitric acid at less than 2% (v/v) is necessary for ICP-MS to minimize damage to the 

interface and to minimize isobaric molecular-ion interferences with the analytes. Many more 

molecular-ion interferences are observed when hydrochloric and sulfuric acids are used (Refs. 3 

and 4). The use of 1% (v/v) HCl is necessary for the stability of antimony and silver 

concentrations in the range of 50 - 500 μg/L. For concentrations greater than 500 μg/L silver, 

additional HCl will be needed. As a consequence, the accuracy of analytes that need significant 

chloride molecular-ion corrections (e.g., As and V) will degrade. 

 7.3.1 Nitric acid (concentrated), HNO3

 7.3.2 Nitric acid (50% [v/v]), HNO3 - Prepare by adding 500 mL concentrated 

HNO3 to 400 mL water and diluting to 1 L. 

 7.3.3 Nitric acid (1% [v/v]), HNO3 - Prepare by adding 10 mL concentrated 

HNO3 to 400 mL water and diluting to 1 L. 

 7.3.4 Hydrochloric acid (concentrated), HCl 

 7.3.5 Hydrochloric acid (37%), HCl - Prepare by adding 370 mL concentrated 

HCl to 400 mL water and diluting to 1L. 

 7.3.6 Hydrofluoric acid (concentrated), HF 

 7.3.7 Phosphoric acid (concentrated), H3PO4

 7.3.8 Phosphoric acid (85% [v/v]), H3PO4 - Prepare by adding 850 mL 

concentrated H3PO4 to 100 mL water and diluting to 1 L. 
SW-846 Update V 6020B - 9 Revision 2 

October 2012 

 7.3.9 Sulfuric acid (concentrated), H2SO4

 7.3.10 Sulfuric acid (96% [v/v]) H2SO4, - Prepare by adding 40 mL water to a 2 L 

glass beaker. While gently stirring, carefully add 960 mL concentrated H2SO4 to the 

beaker. Mix until combined. Allow to cool. Carefully, quantitatively transfer solution to a 1-

L volumetric flask. Bring to volume with additional water if necessary. Mix thoroughly 

through inversion to combine. 

WARNING: Considerable heat is generated upon combining sulfuric acid and water. The 

use of appropriate personal protection (e.g. proper gloves, safety glasses and protective 

clothing) is necessary to avoid personal injury such as thermal burns or acid burns due to 

solution splatter. Also, always add acid to water (rather than water to acid) to reduce 

splatter. 

 7.3.11 Citric acid, HO2CCH2C(OH)(CO2H)CH2CO2H 

 7.4 Bismuth(III) oxide, Bi2O3

 7.5 Holmium(III) carbonate pentahydrate, Ho2(CO3)3•5H2O 

 7.6 Indium (powder), In 

7.7 Lithium[6Li] carbonate (95 atom % 6Li), 6Li2CO3 

 7.8 Ammonium hexachlororhodate(III), (NH4)3RhCl6 

 7.9 Scandium(III) oxide, Sc2O3

 7.10 Terbium(III) carbonate pentahydrate, Tb2(CO3)3•5H2O 

 7.11 Yttrium(III) carbonate, Y2(CO3)3•3H2O 

 7.12 Ammonium hexafluorotitanate(IV), (NH4)2TiF6 

 7.13 Ammonium molybdate(VI) (NH4)2MoO4 

 7.14 Aluminum(III) nitrate nonahydrate, Al(NO3)3•9H2O 

 7.15 Calcium carbonate, CaCO3

 7.16 Iron powder, Fe 

 7.17 Magnesium oxide, MgO 

 7.18 Sodium carbonate, Na2CO3

 7.19 Potassium carbonate, K2CO3
SW-846 Update V 6020B - 10 Revision 2 

October 2012 

 7.20 Standard stock solutions - Purchase standard stock solutions from an appropriate 

commercial source. Otherwise, prepare them manually in the laboratory using only ultra, highpurity grade chemicals or metals (≥ 99.99% purity). See Method 6010 for instructions on 

preparing standard solutions from solids. Replace stock standards when succeeding dilutions 

for the preparation of calibration standards cannot be verified. 

 7.20.1 Bismuth internal standard stock solution (100 μg/mL Bi) - Dissolve 

0.1115 g Bi2O3 in a minimum amount of dilute HNO3. Add 10 mL concentrated HNO3 and 

dilute to 1 L with reagent water. 

 7.20.2 Holmium internal standard stock solution (100 μg/mL Ho) - Dissolve 

0.1757 g Ho2(CO3)3•5H2O in 10 mL reagent water and 10 mL concentrated HNO3. After 

dissolution is complete, warm the solution to degas. Add 10 mL concentrated HNO3 and 

dilute to 1 L with reagent water. 

 7.20.3 Indium internal standard stock solution (100 μg/mL In) - Dissolve 0.1000 g 

indium in 10 mL concentrated HNO3. Dilute to 1 L with reagent water. 

 7.20.4 Lithium internal standard stock solution (100 μg/mL 6

Li) - Dissolve 

0.6312 g 6Li2CO3 (95% atomic abundance) in 10 mL of reagent water and 10 mL 

concentrated HNO3. After dissolution is complete, warm the solution to degas. Add 10 mL 

concentrated HNO3 and dilute to 1 L with reagent water. 

 7.20.5 Rhodium internal standard stock solution (100 μg/mL Rh) - Dissolve 

0.3593 g (NH4)3RhCl6 in 10 mL reagent water. Add 100 mL concentrated HCl and dilute to 

1 L with reagent water. 

 7.20.6 Scandium internal standard stock solution (100 μg/mL Sc) - Dissolve 

0.15343 g Sc2O3 in 10 mL 50% hot HNO3. Add 5 mL concentrated HNO3 and dilute to 1 L 

with reagent water. 

 7.20.7 Terbium internal standard stock solution (100 μg/mL Tb) - Dissolve 

0.1828 g Tb2(CO3)3•5H2O in 10 mL 50% HNO3. After dissolution is complete, warm the 

solution to degas. Add 5 mL concentrated HNO3 and dilute to 1 L with reagent water. 

 7.20.8 Yttrium internal standard stock solution (100 μg/mL Y) - Dissolve 0.2316 g 

Y2(CO3)3•3H2O in 10 mL 50% HNO3. Add 5 mL concentrated HNO3 and dilute to 1 L with 

reagent water. 

 7.20.9 Titanium interference stock solution (100 μg/mL Ti) - Dissolve 0.4133 g 

(NH4)2TiF6 in reagent water. Add 2 drops concentrated HF and dilute to 1 L with reagent 

water. 

 7.20.10 Molybdenum interference stock solution (100 μg/mL Mo) - Dissolve 

0.2043 g (NH4)2MoO4 in reagent water. Dilute to 1 L with reagent water. 

 7.20.11 Gold preservative stock solution for mercury (100 μg/mL Au) - Purchase 

as a commercially prepared, high-purity solution of AuCl3 in dilute HCl matrix. 
SW-846 Update V 6020B - 11 Revision 2 

October 2012 

 7.21 Mixed-calibration standard solutions - Prepare by diluting stock standard solutions 

to levels in the linear range for the instrument, using the same combination and concentrations 

of acids used in the preparation of the sample digestates (approximately 1% HNO3). The 

calibration standard solutions must contain a suitable concentration of an appropriate internal 

standard for each analyte. Internal standards may be added on-line at the time of analysis using 

a second channel of the peristaltic pump and an appropriate mixing manifold. Generally, an 

internal standard should be no more than 50 u removed from the analyte. Recommended 

internal standards include 6Li, 45Sc, 89Y, 103Rh, 115In, 159Tb, 169Ho, and 209Bi. Prior to preparing 

the mixed standards, each stock standard solution must be analyzed separately to determine 

possible spectral interferences or the presence of impurities. 

NOTE: Care should be taken when preparing the calibration standards to ensure that the 

elements are compatible and stable when mixed together. Standards which interfere 

with another analyte, or which are contaminated with another analyte, may not be 

included in the same calibration standard as that analyte. 

Transfer the mixed-standard solutions to an appropriate container for storage. Freshly mixed 

standards must be prepared as needed with the realization that concentrations can change upon 

aging. Calibration standards must be initially verified using a QC standard (see Sec. 7.24). 

 7.22 Blanks - Three types of blanks are necessary for analysis: (1) the calibration blank, 

which is used in establishing the calibration curve; (2) the method blank, which is used to 

monitor for possible contamination resulting from the sample preparation procedure; and (3) the 

rinse blank, which is used to flush the system between all samples and standards. 

7.22.1 Calibration blank - Prepare by acidifying reagent water using the same 

combination and concentrations of acids used in the preparation of the matrix-matched 

calibration standards (Sec. 7.21) along with the selected concentrations of internal 

standards, such that there is an appropriate internal standard element for each of the 

target analytes. The use of HCl for antimony and silver is discussed in Sec. 7.3. The 

calibration blank will also be used for all initial calibration blank (ICB) and continuing 

calibration blank (CCB) determinations. 

 7.22.2 Method blank — Prepare by a processing either a volume of reagent 

water equal to that used for actual aqueous samples, or, otherwise, a clean, empty 

container, equivalent to that used for actual solid samples through all of the preparatory 

and instrument determination steps used for making ICP-MS determinations in samples. 

These steps may include, but are not limited to, pre-filtering, digestion, dilution, filtering, 

and analysis (refer to Sec. 9.5). 

 7.22.3 Rinse blank - Prepare as a 1 - 2% HNO3 solution. Prepare a sufficient 

quantity such that it may be used to flush the system in between standards and samples. 

If mercury is to be analyzed, the rinse blank should also contain 2 μg/mL AuCl3. 

 7.23 Spectral interference check (SIC) solutions - Prepare so as to contain known 

concentrations of interfering elements that will demonstrate the appropriate magnitude of 

interferences and provide an adequate test of any corrections. Chloride in the SIC solution 

provides a means to evaluate software corrections for chloride-related interferences such as 
SW-846 Update V 6020B - 12 Revision 2 

October 2012 

35Cl16O+

 on 51V

+

 and 40Ar35Cl+ on 75As+. Iron is used to demonstrate adequate resolution of the 

spectrometer for the determination of manganese. Molybdenum serves to indicate oxide effects 

on cadmium isotopes. The other components are present to evaluate the ability of the 

measurement system to correct for various molecular-ion isobaric interferences. The SIC is 

used to verify that the interference levels are corrected by the data system within appropriate QC 

limits. 

NOTE: The final SIC solution concentrations in Table 1 are intended to evaluate corrections for 

known interferences on only the analytes identified in Sec. 1.0. If the test method is to 

be used to determine other element(s), it is the responsibility of the analyst to modify the 

SIC solution accordingly, or prepare an alternative SIC solution, so as to allow adequate 

verification of interference corrections on the additional element(s) (see Sec. 9.9). 

 7.23.1 Mixed stock SIC solutions - Prepare the SIC stock solutions using only 

ultra-pure reagents. They can be obtained commercially or prepared using the following 

procedures: 

 7.23.1.1 Mixed SIC stock solution I - Prepare by adding 13.903 g 

Al(NO3)3•9H2O, 2.498 g CaCO3 (previously dried at 180 ΕC for 1 hr), 1.000 g Fe, 

1.658 g MgO, 2.305 g Na2CO3 and 1.767 g K2CO3 to 25 mL of reagent water. 

Slowly add 40 mL of (50%) HNO3. After dissolution is complete, warm the solution

to degas. Cool and dilute to 1 L with reagent water. 

 7.23.1.2 Mixed SIC stock solution II - Prepare by slowly adding 7.444 g 

85% H3PO4, 6.373 g 96% H2SO4, 40.024 g 37% HCl, and 10.664 g citric acid 

(C6O7H8) to 100 mL of reagent water. Dilute to 1 L with reagent water. 

 7.23.2 Mixed working SIC solution - Prepare by combining 10.0 mL of SIC stock 

solution I, 2.0 mL each of 100-μg/mL titanium stock solution and 100-μg/mL molybdenum 

stock solution, and 5.0 mL of SIC stock solution II. Dilute to 100 mL with reagent water. 

Prepare fresh weekly. 

 7.24 Initial calibration verification (ICV) standard - Prepare by combining compatible 

metals from standard stock solution sources that differ from those used for the preparation of the 

calibration standards. The ICV should be prepared so as to contain metal concentrations that 

are near, but not equal to, the midpoint concentration level of the calibration curve. 

 7.25 Continuing calibration verification (CCV) standard - Prepare using the same acid 

matrix and stock standards employed when preparing the calibration standards. The CCV 

should be prepared so as to contain metal concentrations equal or nearly equivalent to the 

midpoint concentration of the calibration curve. 

 7.26 Mass spectrometer tuning solution - Prepare so as to contain elements that 

represent all of the mass regions of interest (i.e., 10 μg/L Li, Co, In, and Tl) in order to verify that 

the resolution and mass calibration of the instrument are within the designated specifications 

(see Sec. 10.1). 

7.27 If the determination of one or more metals using a non-aqueous solvent is required, 

then all standards and quality control samples must be prepared on a weight/weight basis in the 
SW-846 Update V 6020B - 13 Revision 2 

October 2012 

non-aqueous solvent since the density of non-aqueous solvents is not unity. Standards and 

quality control materials containing organometallic materials that are soluble in non-aqueous 

solvents are available from a variety of vendors. 

8.0 SAMPLE COLLECTION, PRESERVATION, AND STORAGE 

Sample collection, preservation and storage requirements may vary by EPA program and 

may be specified in a regulation or project planning document that requires compliance 

monitoring for a given contaminant. Where such requirements are specified in the regulation, 

follow those requirements. In the absence of specific regulatory requirements, use the following 

information as guidance in determining the sample collection, preservation and storage 

requirements. 

See Chapter Three, Inorganic Analytes, for sample collection and preservation 

instructions. 

9.0 QUALITY CONTROL 

9.1 Refer to Chapter One for guidance on quality assurance (QA) and QC protocols. 

When inconsistencies exist between QC guidelines, method-specific QC criteria take 

precedence over those criteria given in Chapter One. Any effort involving the collection of 

analytical data should include development of a structured and systematic planning document, 

such as a Quality Assurance Project Plan (QAPP) or a Sampling and Analysis Plan (SAP), which 

translates project objectives and specifications into directions for those that will implement the 

project and assess the results. Each laboratory should maintain a formal quality assurance 

program. The laboratory should also maintain records to document the quality of the data 

generated. All data sheets and QC data should be maintained for reference or inspection. 

9.2 Refer to Methods 3005, 3010, 3015, 3031, 3040, 3050, 3051, 3052, 7000, and 

6800 for QC procedures to ensure the proper operation of the various sample preparation 

techniques. Any more specific QC procedures provided in this method will supersede those 

noted in Methods 3005, 3010, 3015, 3031, 3040, 3050, 3051, 3052, 7000, and 6800. 

9.3 Instrument Detection Limits 

Instrument detection limits (IDLs) are useful means to evaluate the instrument noise level 

and response changes over time for each analyte from a series of reagent blank analyses to 

obtain a calculated concentration. They are not to be confused with the lower limit of 

quantitation, nor should they be used in establishing this limit. It may be helpful to compare the 

calculated IDLs to the established lower limit of quantitation, however, it should be understood 

that the lower limit of quantitation needs to be verified according to the guidance in Sec. 9.8. 

IDLs in μg/L can be estimated as the mean of the blank result plus three times the standard 

deviation of 10 replicate analyses of the reagent blank solution. (Use zero for the mean if the 

mean is negative). Each measurement should be performed as though it were a separate 

analytical sample (i.e., each measurement must be followed by a rinse and/or any other 

procedure normally performed between the analysis of separate samples). IDLs should be 

determined at least once using new equipment, after major instrument maintenance such as 
SW-846 Update V 6020B - 14 Revision 2 

October 2012 

changing the detector, and/or at a frequency designated by the project. An instrument log book 

should be kept with the dates and information pertaining to each IDL performed. 

9.4 Initial demonstration of proficiency 

Each laboratory must demonstrate initial proficiency with each sample preparation and 

determinative method combination by generating data of acceptable precision and bias for target 

analytes in a clean matrix. If an autosampler is used to perform sample dilutions, before using 

the autosampler to dilute samples, the laboratory should satisfy itself that those dilutions are of 

equivalent or better accuracy than is achieved by an experienced analyst performing manual 

dilutions. It is recommended that the laboratory should repeat the demonstration of proficiency 

whenever new staff members are trained or significant changes in instrumentation are made. 

9.5 Initially, before processing any samples, the analyst should demonstrate that all 

parts of the equipment that come into direct contact with the sample and reagents are 

interference-free. This is accomplished through the analysis of a method blank. As a continuing 

check, each time samples are digested and analyzed, and when there is a change in reagents, a 

method blank should be prepared and analyzed for the compounds of interest as a safeguard 

against chronic laboratory contamination. If an interference is observed that would prevent the 

determination of the target analyte, determine the source and eliminate it, if possible, before 

processing the samples. The method blank should be carried through all stages of sample 

preparation and instrument determination procedures. When new reagents or chemicals are 

received, the laboratory should monitor the preparation and/or analysis blanks associated with 

samples for any signs of contamination. It is not necessary to test every new batch of reagents 

or chemicals prior to sample preparation if the source shows no prior problems. However, if 

reagents are changed during a preparation batch, separate blanks need to be prepared for each 

set of reagents. 

9.6 Linear range 

The linear range establishes the highest concentration that may be reported without 

diluting the sample. Following calibration, the laboratory may choose to analyze a standard at a 

higher concentration than the high standard in the calibration. The standard must recover within 

10% of the true value, and if successful, establishes the linear range. The linear range standards 

must be analyzed in the same instrument run as the calibration they are associated with (i.e., on 

a daily basis) but may be analyzed anywhere within that run. If a linear range standard is not 

analyzed for any specific element, the highest standard in the calibration becomes the linear 

range. 

9.7 Sample QC for preparation and analysis 

 The laboratory must also have procedures for documenting the effect of the matrix on 

method performance (precision, bias, and sensitivity). At a minimum, this should include the 

analysis of QC samples including a method blank, a matrix spike (MS), a laboratory control 

sample (LCS), and a duplicate sample in each analytical batch. Any method blanks, LCS, MS 

samples, and duplicate samples should be subjected to the same preparatory and instrument 

determination procedures as those used on actual samples (see Sec. 11.0). 
SW-846 Update V 6020B - 15 Revision 2 

October 2012 

 9.7.1 For each batch of samples analyzed, at least one method blank must be 

carried throughout the entire sample preparation and instrument determination process, as 

described in Chapter One. The importance of the method blank is to aid in identifying 

when and/or if sample contamination is occurring. The method blank is considered to be 

acceptable if it does not contain the target analytes at concentration levels that exceed the 

acceptance limits defined in the project-specific DQOs. The laboratory should not subtract 

the results of the method blank from those of any associated samples. Such "blank 

subtraction" is not reliable because it is based on a single method blank value rather than a 

statistically determined blank concentration. 

 In the absence of project-specific DQOs, the method blank may be considered to 

be acceptable with respect to each target element, if it is less than the LLOQ, 50% of the 

regulatory limit, or 10% of the lowest sample concentration, whichever is greater. If the 

method blank fails to meet the necessary acceptance criteria, it should be re-analyzed 

once. If still unacceptable, then all samples associated with the method blank must be reprepared and re-analyzed, along with all other appropriate analysis batch QC samples. If 

the method blank results do not meet the acceptance criteria and reanalysis is not 

practical, then the laboratory should report the sample results along with the method blank 

results, and provide a discussion of the potential impact of the contamination on the 

sample results. If the method blank exceeds the acceptance criteria, but the sample 

results are all below the reporting level, regulatory limit or other applicable DQOs, then the 

sample data may be considered to be valid, despite the contamination indicated in the 

method blank. Refer to Chapter One for additional guidance regarding the proper protocol 

when analyzing method blanks. 

 9.7.2 Documenting the effect of the matrix should include the analysis of at 

least one MS and one duplicate unspiked sample or one matrix spike/matrix spike 

duplicate (MS/MSD) pair for each batch of samples processed, at a minimum frequency of 

one per every 20 samples, as described in Chapter One. An MS/MSD pair is used to 

document the bias and precision of a method in a given sample matrix. The decision on 

whether to prepare and analyze duplicate samples or an MS/MSD pair must be based on 

knowledge of the samples in the analysis batch. If samples are expected to contain target 

analytes above the LLOQ, laboratories may choose to use an MS and a duplicate analysis 

of an unspiked field sample. If samples are not expected to contain target analytes above 

the LLOQ, the laboratories should use an MS/MSD pair. 

 MS/MSD samples should be spiked with each target element at the project-specific 

action levels, or, when lacking project-specific action levels, between the low- and midlevel standards, as appropriate. Acceptance criteria should be set at laboratory-derived 

limits, developed through the use of historical analyses, for each matrix type being 

analyzed. However, historically derived acceptance limits must not exceed ± 25% recovery 

of the target element spike values for bias, and ≤ 20 relative percent difference (RPD) for 

precision. In the absence of historical data, MS/MSD acceptance limits should be set at ± 

25% recovery and ≤ 20 RPD. Refer to Sec. 4.0 of Chapter One for further guidance. If the 

bias and precision indicators in an analytical batch fail to meet the acceptance criteria, then 

the interference test discussed in Sec. 9.10 should be performed. Refer to the definitions 

of bias and precision, in Chapter One, for the proper data reduction protocols. 

 Calculate the RPD between duplicate or MS determinations as follows: 
SW-846 Update V 6020B - 16 Revision 2 

October 2012 

100

2

D D

D D

RPD

1 2

1 2

×















 +

−

=

where: 

RPD = relative percent difference 

D1 = MS or first sample analysis value 

D2 = MSD or duplicate sample analysis value 

 9.7.3 At least one LCS should be prepared and analyzed with each batch of 

analytical samples processed, at a minimum frequency of one LCS per every 20 samples, 

as described in Chapter One. The LCS consists of an aliquot of a clean (control) matrix 

similar to the sample matrix and of the same weight or volume. The LCS should be spiked 

at the same levels and using the same spiking materials as the corresponding MS/MSD 

(see above Sec. 9.7.2). When the results of the MS analysis indicate a potential problem 

due to the sample matrix itself, the LCS results are used to verify that the laboratory can 

acceptably perform the analysis in a clean matrix. 

 LCS acceptance criteria should be set at laboratory-derived limits, developed through the 

use of historical analyses. However, historically derived acceptance limits must not exceed 

± 20% of the target element spike values. In the absence of historical data, LCS 

acceptance limits should be set at ± 20%. If the result of an LCS does not meet the 

established acceptance criteria, it should be re-analyzed once. If still unacceptable, then 

all samples associated with the LCS must be re-prepared and re-analyzed, along with all 

other appropriate analysis batch QC samples. 

 9.7.4 Reference materials containing known amounts of target elements are 

recommended when an appropriately similar medium of interest are available as one type 

of QC after appropriate sample preparation. The reference material may be used as the 

LCS. For soil reference materials, the manufacturers’ established acceptance criterion 

should be used. For solid reference materials, ± 20% (see Sec. 9.7.3) recovery of the 

reported manufacturers’ target element values may not be achievable. Refer to Chapters 

One and Three for additional information. 

 9.8 Lower Limit of Quantitation (LLOQ) check standard 

9.8.1 The laboratory should establish the LLOQ as the lowest point of 

quantitation which, in most cases, is the lowest concentration in the calibration curve. The 

LLOQ is initially verified by the analysis of at least 7 replicate samples, spiked at the LLOQ 

and processed through all preparation and analysis steps of the method. The mean 

recovery and relative standard deviation of these samples provide an initial statement of 

precision and accuracy at the LLOQ. In most cases the mean recovery should be +/- 35% 

of the true value and RSD should be < 20%. In-house limits may be calculated when 

sufficient data points exist. Monitoring recovery of LLOQ over time is useful for assessing 
SW-846 Update V 6020B - 17 Revision 2 

October 2012 

precision and bias. Refer to a scientifically valid and published method such as Chapter 9 

of Quality Assurance of Chemical Measurements (Taylor 1987) or the Report of the 

Federal Advisory Committee on Detection and Quantitation Approaches and Uses in Clean 

Water Act Programs (http://water.epa.gov/scitech/methods/cwa/det/index.cfm) for 

calculating precision and bias for LLOQ. 

9.8.2 Ongoing LLOQ verification, at a minimum, is on a quarterly basis to 

validate quantitation capability at low analyte concentration levels. This verification may be 

accomplished either with clean control material (e.g., reagent water, method blanks, 

Ottawa sand, diatomaceous earth, etc.) or a representative sample matrix (free of target 

compounds). Optimally, the LLOQ should be less than the desired regulatory action levels 

based on the stated project-specific requirements. 

 

 9.9 Verify the magnitude of elemental and molecular-ion isobaric interferences and the 

adequacy of any corrections at the beginning of an analytical run or once every 12 hours of 

continuing sample analysis, whichever is more frequent. Do this by analyzing the SIC solution. 

Results for the unspiked elements in the SIC solution should be less than 2 times the LLOQ. 

Note that it may not be possible to obtain SIC spiking solutions that are completely free of the 

unspiked elements. If the presence and concentration of an unspiked element can be confirmed 

via vendor documentation and/or determination of multiple isotopes of the element in the correct 

ratios, the concentration actually present may be subtracted from the determined value prior to 

comparing to the LLOQ limits. Refer to Sec. 4.0 for a discussion on interferences and potential 

solutions to those interferences if additional guidance is needed. 

 

 9.10 The intensities of each internal standard must be monitored for every analysis to 

ensure that it does not decrease below 30%, with respect to its intensity during the initial 

calibration. If this occurs, a significant matrix effect must be suspected. Under these conditions, 

the IDL has degraded, and therefore the correction capability of the internal-standardization 

technique must then be questioned. If this happens, perform the following procedure: 

 9.10.1 Make sure the instrument has not drifted by observing the internal 

standard intensities in the nearest clean matrix, i.e., the calibration blank. If the low 

internal standard intensities are also observed in the nearby calibration blank, terminate 

the analysis, correct the problem, recalibrate the instrument, verify the new calibration, and 

reanalyze the affected samples. 

 9.10.2 If drift has not been demonstrated to occur as outlined in Sec. 9.10.1, 

matrix effects need to be removed by diluting the affected sample. Dilute the sample fivefold (1:5), taking into consideration the need to add the appropriate amounts of internal 

standards, and reanalyze. If the first dilution does not eliminate the problem, repeat the 

dilution procedure in an iterative fashion, using ever-increasing dilutions, until the internalstandard intensities exceed the 30% acceptance limit. Correct the reported results using 

the appropriate dilution factors. 

 9.11 To obtain analyte data of known quality, it is necessary to measure more than the 

analytes of interest in order to apply corrections or to determine whether interference corrections 

are necessary. For example, tungsten oxide molecular-ion species can be very difficult to 

distinguish from mercury isotopes. If the concentrations of interference sources (such as C, Cl, 
SW-846 Update V 6020B - 18 Revision 2 

October 2012 

Mo, Zr, W) are such that, at the correction factor, the analyte is less than the LLOQ and the 

concentration of interferents are insignificant, then the data may go uncorrected. 

NOTE: Monitoring the interference sources does not inevitably necessitate monitoring of the 

interferant itself, but that a molecular species may be monitored to indicate the presence 

of the interferent. 

 When correction equations are used, all QC criteria must also be met. Extensive QC for 

interference corrections is needed at all times. The monitored masses must include those 

elements whose hydrogen, oxygen, hydroxyl, chlorine, nitrogen, carbon and sulfur molecular 

ions could impact the analytes of interest. Unsuspected interferences may be detected by 

adding pure major matrix components to a sample to observe any impact on the analyte signals. 

 When an interference source is present, the sample elements impacted must be flagged to 

indicate (a) the percentage interference correction applied to the data; or (b) an uncorrected 

interference, by virtue of the elemental equation used for quantitation. The isotope proportions 

for an element or molecular-ion cluster provide information useful for QA. 

NOTE: Only isobaric elemental, molecular, and doubly charged interference corrections, which 

employ the observed isotopic-response ratios or parent-to-oxide ratios (provided an 

oxide internal standard is used as described in Sec. 4.2) for each instrument system, 

are acceptable corrections for use in this method. 

 9.12 It is recommended that the laboratory adopt additional QA practices for use with 

this method. The specific practices that are most productive depend upon the needs of the 

laboratory and the nature of the samples. Whenever possible, the laboratory should analyze 

reference materials and participate in relevant performance evaluation (PE) studies. 

 9.13 If less than acceptable bias and precision data are generated for the matrix 

spike(s), the additional QC protocols in Sections 9.13.1 and/or 9.13.2 should be performed prior 

to reporting concentration data for the elements in this method. At a minimum these tests should 

be performed with each batch of samples prepared/analyzed with corresponding unacceptable 

data quality results. If matrix interference effects are confirmed, then an alternative test method 

should be considered or the current test method modified, so that the analysis is not affected by 

the same interference. The use of a standard-addition analysis procedure may also be used to 

compensate for this effect (refer to Method 7000). 

 9.13.1 Dilution test 

 If the analyte concentration is within the linear range of the instrument and 

sufficiently high (minimally, a factor of 25 times greater than the LLOQ), an analysis of a 

1:5 dilution should agree to within ± 20% of the original determination. If not, then a 

chemical or physical interference effect must be suspected. The matrix spike is often a 

good choice of sample for the dilution test, since reasonable concentrations of most 

analytes are present. Elements that fail the dilution test are reported as estimated values. 

 9.13.2 Post-digestion MS 

 If a high concentration sample is not available for performing the dilution test, then 

a post-digestion MS should be performed. The test only needs to be performed for the 
SW-846 Update V 6020B - 19 Revision 2 

October 2012 

specific elements that failed original matrix spike limits, and only if the spike concentration 

added was greater than the concentration determined in the unspiked sample. Following 

preparation, which may include, but is not limited to, pre-filtration, digestion, dilution and 

filtration, an aliquot, or dilution thereof, should be obtained from the final aqueous, 

unspiked-analytical sample, and spiked with a known quantity of target elements. The 

spike addition should be based on the indigenous concentration of each element of 

interest in the sample. The recovery of the post-digestion MS should fall within a ± 25 % 

acceptance range, relative to the known true value, or otherwise within the laboratoryderived acceptance limits. If the post-digestion MS recovery fails to meet the acceptance 

criteria, the sample results must be reported as estimated values. 

 9.14 Ultra-trace analysis necessitates the use of clean chemistry practices. Several 

suggestions for the reduction of contaminants in the analytical blank are provided in Chapter 

Three, Inorganic Analytes. 

10.0 CALIBRATION AND STANDARDIZATION 

 10.1 Conduct mass calibration and resolution verification checks in the mass regions of 

interest using the mass spectrometer tuning solution (Sec. 7.26). The mass calibration and 

resolution verification acceptance criteria must be met prior to the analysis of samples. If the 

mass calibration differs by more than 0.1 u from the true value, then the mass calibration must 

be adjusted to the correct value. The resolution must also be verified to be less than 0.9 u full 

width at 10% peak height. 

 10.2 At a minimum, the elements required for the project plus any required for 

interference correction must be calibrated. Recommended isotopes for the analytes in Sec. 1.2 

are provided in Table 2. Flush the system in between each standard and sample using the rinse 

blank (Sec. 7.22.3). The rinse time needs to be sufficient to ensure that analytes present in the 

linear range are effectively cleaned out prior to analysis of the subsequent sample. Use the 

average of at least three readings (of a single injection) for both calibration standard and sample 

analyses. 

10.3 Calibration standards should be prepared on an as-needed basis unless stability 

warrants preparing fresh daily, (or each time a batch of samples is analyzed). If the ICV 

standard is prepared daily and the results of the ICV analyses meet the acceptance criteria, then 

the calibration standards do not need to be prepared daily and may be prepared and stored for 

as long as the calibration standard viability can be verified through the use of the ICV. If the ICV 

fails to meet the acceptance criteria, trouble shoot the situation, and then prepare a new set of 

calibration standards if needed and recalibrate the instrument 

10.4 A calibration curve must be analyzed daily. The instrument may be calibrated 

using a single point standard and a calibration blank (ICB) or a multipoint calibration curve. If a 

multipoint curve is used a minimum of three standards are required and the correlation 

coefficient (r) should be > 0.995 or the coefficient of determination (r2

) should be > 0.990. 

Relative Standard Error may be used as an alternative to r or r2, and should be < 20%. If a 

multipoint calibration is used the low standard must be at or below the LLOQ. 

NOTE: Inversely weighted linear regressions or other methods may be used in order to minimize 
SW-846 Update V 6020B - 20 Revision 2 

October 2012 

curve fitting errors at the low end of the calibration curve. 

10.5 After the initial calibration is completed it is verified using several checks. 

10.5.1 Initial Calibration Verification (ICV) - The ICV is a standard prepared from 

a different source than the initial calibration standards. It is analyzed at approximately the 

mid-level of the calibration and serves as a check that the initial calibration standards are 

at the correct concentrations. The acceptance range is 90-110% of the true value. 

10.5.2 Low-level readback or verification - For a multi-point calibration, the low 

level standard should quantitate to within 80-120% of the true value. For a single point 

calibration, a standard from the same source as the calibration standard and at or below 

the LLOQ is analyzed and should recover within 80-120% of the true value. 

10.5.3 Mid-level readback or verification - For a multi-point calibration, the midlevel standard should quantitate to within 90-110% of the true value. For a single point 

calibration, a standard from the same source as the calibration standard and at the midpoint of the linear range is analyzed and should recover within 90-110% of the true value. 

10.5.4 Initial Calibration blank (ICB) - If a multi-level calibration is used, an ICB is 

analyzed immediately after the calibration (or after the ICV) and must not contain target 

analytes above half the LLOQ. If a single point calibration is used, the calibration is forced 

through the ICB, but a second ICB is analyzed as a check and must not contain target 

analytes above half the LLOQ. 

NOTE: After cleaning the sampler and skimmer cones, improved performance in calibration 

stability has been observed by method users if the instrument is exposed to the SIC 

solution. Improved performance has also been observed if the instrument is allowed to 

rinse for 5 - 10 minutes before starting the calibration process. 

 

 10.5.5 Verify the ongoing validity of the calibration curve after every 10 samples, 

and at the end of each analysis batch run, through the analysis of a CCV standard (Sec. 

7.25) and a CCB (Sec. 7.22.1). For the curve to be considered valid the analysis result of 

the CCV standard must be within ± 10% of its true value and the CCB must not contain 

target analytes above the LLOQ. If the calibration cannot be verified, sample analysis 

must be discontinued, the cause of the problem determined and the instrument 

recalibrated. All samples following the last acceptable CCV standard must be reanalyzed. 

 If the ICB consistently has target analyte concentrations greater than half the LLOQ, the 

LLOQ should be re-evaluated. Flow-injection systems may be used as long as they can 

meet the performance criteria of the method. 

11.0 PROCEDURE 

 11.1 Preliminary treatment of most samples is necessary because of the complexity and 

variability of sample matrices. Groundwater samples which have been pre-filtered and acidified 

will not need acid digestion. Samples which are not digested must either use an internal 

standard or be matrix-matched with the standards (i.e., acid concentrations should match). 

Solubilization and digestion procedures are presented in Chapter Three, Inorganic Analytes. 
SW-846 Update V 6020B - 21 Revision 2 

October 2012 

NOTE: If mercury is to be analyzed, the digestion procedure must use mixed nitric and 

hydrochloric acids through all steps of the digestion. Mercury will be lost if the sample is 

digested when hydrochloric acid is not present. If it has not already been added to the 

sample as a preservative, Au should be added to give a final concentration of 2 mg/L 

(use 2.0 mL of gold preservative stock (Sec. 7.20.11) per 100 mL of sample) to preserve 

the mercury and to prevent it from plating out in the sample introduction system. 

 11.2 Initiate an appropriate operating configuration of the instrument computer according 

to the instrument manufacturer's instructions. 

 11.3 Set up the instrument with the proper operating parameters according to the 

instrument manufacturer's instructions. 

 11.4 Operating conditions 

 Tune the instrument by following the instructions provided by the instrument manufacturer. 

 Allow at least 30 minutes for the instrument to equilibrate before analyzing samples. 

NOTE: The instrument should have features that protect it from high ion currents. If not, 

precautions must be taken to protect the detector. A channel electron multiplier or 

active film multiplier will suffer from fatigue after being exposed to high ion currents. 

This fatigue can last from several seconds to hours depending on the extent of 

exposure. During this time period, response factors are constantly changing, which 

invalidates the calibration curve, causes instability, and invalidates sample analyses. 

 11.5 Calibrate the instrument following the procedure outlined in Sec. 10.0. 

 11.6 Flush the system with the rinse blank solution (Sec. 7.22.3) until the signal levels 

return to the data quality objectives or method LLOQs (usually about 30 seconds) before the 

analysis of each sample. Nebulize each sample until a steady-state signal is achieved (usually 

about 30 seconds) prior to collecting data. 

 11.7 Dilute and reanalyze samples that exceed the linear range for an analyte (or 

species needed for a correction) or measure an alternate, but less-abundant, isotope. The 

linearity at the alternate mass must be confirmed by appropriate calibration (see Sec. 10.4). 

Alternatively apply solid-phase chelation chromatography to eliminate the matrix as described in 

Sec. 4.3. 

11.8 Determination of percent dry weight 

When sample results are to be calculated on a dry-weight basis, a separate portion of 

sample for this determination should be weighed out at the same time as the portion used for 

analytical determination. 

CAUTION: The drying oven should be contained in a hood or vented. Significant laboratory 

contamination may result from a heavily contaminated hazardous waste sample. 
SW-846 Update V 6020B - 22 Revision 2 

October 2012 

11.8.1 Immediately after weighing the sample aliquot to be digested, weigh an 

additional 5- to 10-g aliquot of the sample to the nearest 0.01g into a tared crucible. Dry 

this aliquot overnight at 105 ΕC. Allow the sample to cool in a desiccator before weighing. 

 

11.8.2 Calculate the % dry weight as follows: 

This oven-dried aliquot is not used for the extraction and should be appropriately disposed 

of once the dry weight is determined. 

12.0 DATA ANALYSIS AND CALCULATIONS 

 12.1 If dilutions were performed, apply the appropriate corrections to the sample values. 

 

 12.2 If appropriate, or required by the project or regulation for data reporting, calculate 

results for solids on a dry-weight basis as follows: 

W S

C V

ConcentrationDW

×

×

=

 where: 

 ConcentrationDW = Concentration on a dry weight basis (mg/kg) 

C = Digest concentration (mg/L) 

V = Final volume after sample preparation (L) 

W =Wet sample mass (kg) 

S = % Solids/100 = % dry weight/100 

 Calculations must include appropriate interference corrections (see Sec. 4.2 for examples), 

internal-standard normalization, and the summation of signals at 206, 207, and 208 m/z for lead (to 

compensate for any differences in the abundances of these isotopes between samples and 

standards). 

13.0 METHOD PERFORMANCE 

Performance data and related information are provided in SW-846 methods only as 

examples and guidance. The data do not represent required performance criteria for users of 

the methods. Instead, performance criteria should be developed on a project-specific basis, 

and the laboratory should establish in-house QC performance criteria for the application of this 

method. These performance data are not intended to be and must not be used as absolute QC 

acceptance criteria for purposes of laboratory accreditation.

 x 100

 g of sample

 g of dry sample % dry weight = 
SW-846 Update V 6020B - 23 Revision 2 

October 2012 

Table 3 summarizes the method performance data for aqueous and sea water samples with 

interfering elements removed and samples preconcentrated prior to analysis. Table 4 

summarizes the performance data for a simulated drinking water standard. These data are 

provided for guidance purposes only. 

14.0 POLLUTION PREVENTION 

 14.1 Pollution prevention encompasses any technique that reduces or eliminates the 

quantity and/or toxicity of waste at the point of generation. Numerous opportunities for pollution 

prevention exist in laboratory operations. The EPA has established a preferred hierarchy of 

environmental management techniques that places pollution prevention as the management 

option of first choice. Whenever feasible, laboratory personnel should use pollution prevention 

techniques to address their waste generation. When wastes cannot be feasibly reduced at the 

source, the Agency recommends recycling as the next best option. 

 14.2 For information about pollution prevention that may be applicable to laboratories 

and research institutions consult Less is Better: Laboratory Chemical Management for Waste 

Reduction, a free publication available from the American Chemical Society (ACS), Committee 

on Chemical Safety, 

http://portal.acs.org/portal/fileFetch/C/WPCP_012290/pdf/WPCP_012290.pdf. 

15.0 WASTE MANAGEMENT 

 The Environmental Protection Agency requires that laboratory waste management 

practices be conducted consistent with all applicable rules and regulations. The Agency urges 

laboratories to protect the air, water, and land by minimizing and controlling all releases from 

hoods and bench operations, complying with the letter and spirit of any sewer discharge permits 

and regulations, and by complying with all solid and hazardous waste regulations, particularly the 

hazardous waste identification rules and land disposal restrictions. For further information on 

waste management, consult the ACS publication listed in Sec. 14.2. 

16.0 REFERENCES 

1. Horlick, G., et al., Spectrochim. Acta 40B, 1555 (1985). 

2. Gray, A.L., Spectrochim. Acta 40B, 1525 (1985); 41B, 151 (1986). 

3. Tan, S.H., and Horlick, G., Appl. Spectrosc. 40, 445 (1986). 

4. Vaughan, M.A., and Horlick, G., Appl. Spectrosc. 40, 434 (1986). 

5. Holden, N.E., "Table of the Isotopes," in Lide, D.R., Ed., CRC Handbook of Chemistry and 

Physics, 74th Ed., CRC Press, Boca Raton, FL, 1993.

6. Hinners, T.A., Heithmar, E., Rissmann, E., and Smith, D., Winter Conference on Plasma 

Spectrochemistry, Abstract THP18; p. 237, San Diego, CA (1994). 
SW-846 Update V 6020B - 24 Revision 2 

October 2012 

7. Lichte, F.E., et al., Anal. Chem. 59, 1150 (1987). 

8. Evans E.H., and Ebdon, L., J. Anal. At. Spectrom. 4, 299 (1989). 

9. Beauchemin, D., et al., Spectrochim. Acta 42B, 467 (1987). 

10. Houk, R.S., Anal. Chem. 58, 97A (1986). 

11. Thompson, J.J., and Houk, R.S., Appl. Spectrosc. 41, 801 (1987). 

12. Newberry, W.R., Butler, L.C., Hurd, M.L., Laing, G.A., Stapanian, M.A., Aleckson, K.A., Dobb, 

D.E., Rowan, J.T., and Garner, F.C., Final Report of the Multi-Laboratory Evaluation of Method 

6020 CLP-M Inductively Coupled Plasma-Mass Spectrometry (1989). 

13. Taylor, Daniel B., Kingston, H.M., Nogay, Donald J., Koller, Dagmar, and Hutton, Robert. OnLine Solid-phase Chelation for the Determination of Eight Metals in Environmental Waters by 

Inductively Coupled Plasma Mass Spectrometry. 

14. Kingston, H. M., Siriraks, A., and Riviello, J. M., Patent Number 5,126,272, "A Method and 

Apparatus for Detecting Transition and Rare Earth Elements in a Matrix", U.S. Patent, Filed 

U.S. Patent Office, March 1989, 31 pages, Granted June 30, 1992, Patent held by US 

Government. 

15. Kingston, H. M., Siriraks, A., and Riviello, J. M., Patent Number 5,244,634 , "A Method and 

Apparatus for Detecting Transition and Rare Earth Elements in a Matrix", U.S. Patent , Filed 

U.S. Patent Office, March 1989, 31 pages, Granted Sept. 14, 1993, Patent held by US 

Government. 

16. Dobb, David E., Rowan, J.T., and Cardenas, D., Lockheed Environmental Systems and 

Technologies Co., Las Vegas, NV; and Butler, L.C., and Heithmar, E.M., U.S.EPA, Las Vegas, 

NV; Determination of Mercury by ICP-MS. 

17. Taylor, J. K., “Quality Assurance of Chemical Measurements”, Lewis Publishers (1987). 

18. Federal Advisory Committee on Detection and Quantitation Approaches and Uses in Clean 

Water Act Programs, Final Report, http://water.epa.gov/scitech/methods/cwa/det/index.cfm 

(December 28, 2007). 

17.0 TABLES, DIAGRAMS, FLOW CHARTS, AND VALIDATION DATA 

 The pages to follow contain the tables, and figures referenced by this method. 
SW-846 Update V 6020B - 25 Revision 2 

October 2012 

TABLE 1 

RECOMMENDED SPECTRAL INTERFERENCE CHECK (SIC) SOLUTION 

COMPONENTS AND CONCENTRATIONS 

Solution 

Component 

SIC Concentration 

(mg/L) 

Al 100.0 

Ca 300.0 

Fe 250.0 

Mg 100.0 

Na 250.0 

P 100.0 

K 100.0 

S 100.0 

C 200.0 

Cl 2000.0 

Mo 2.0 

Ti 2.0 
SW-846 Update V 6020B - 26 Revision 2 

October 2012 

TABLE 2 

RECOMMENDED ELEMENTAL ISOTOPES FOR SELECTED ELEMENTS 

Element of Interest Mass of Isotope 

Aluminum 27

Antimony 121, 123

Arsenic 75

Barium 138, 137, 136, 135, 134 

Beryllium 9

Bismuth (IS) 209 

Cadmium 114, 112, 111, 110, 113, 116, 106 

Calcium (I) 42, 43, 44, 46, 48 

Chlorine (I) 35, 37, (77, 82)a

Chromium 52, 53, 50, 54 

Cobalt 59

Copper 63, 65

Holmium (IS) 165 

Indium (IS) 115, 113 

Iron (I) 56, 54, 57, 58 

Lanthanum (I) 139 

Lead 208, 207, 206, 204 

Lithium (IS) 6b, 7 

Magnesium (I) 24, 25, 26

Manganese 55

Mercury 202, 200, 199, 201 

Molybdenum (I) 98, 96, 92, 97, 94, (108)a

Nickel 58, 60, 62, 61, 64 

Potassium (I) 39

Rhodium (IS) 103 

Scandium (IS) 45 

Selenium 80, 78, 82, 76, 77, 74 

Silver 107, 109

Sodium (I) 23

Terbium (IS) 159 

Thallium 205, 203 

Vanadium 51, 50

Tin (I) 120, 118

Yttrium (IS) 89 

Zinc 64, 66, 68, 67, 70 

NOTE: Method 6020 is recommended for only those analytes listed in Sec.1.2. Other elements are 

included in this table because they are potential interferents (labeled I) in the determination of 

recommended analytes, or because they are commonly used internal standards (labeled IS). 

Isotopes are listed in descending order of natural abundance. The most generally useful isotopes 

are underlined and in boldface, although certain matrices may necessitate the use of alternative 

isotopes. 

a These masses are also useful for interference correction (Sec. 4.2). 

b Internal standard must be enriched in the 6

Li isotope. This minimizes interference from 

indigenous lithium.
SW-846 Update V 6020B - 27 

TABLE 3 

METHOD PERFORMANCE DATA FOR AQUEOUS AND SEA WATER SAMPLES a

WITH INTERFERING ELEMENTS REMOVED AND SAMPLES PRECONCENTRATED PRIOR TO A

CONCENTRATION (ng/mL) b

ELEMENT ISOTOPE 9.0 mL 27.0 mL 

Manganese 55 1.8±0.05 1.9±0.2 Nickel 58 0.32±0.018 0.32±0.04 Cobalt 59 0.033±0.002 0.028±0.003 Copper 63 0.68±0.03 0.63±0.03 Zinc 64 1.6±0.05 1.8±0.15 Copper 65 0.67±0.03 0.6±0.05 Zinc 66 1.6±0.06 1.8±0.2 Cadmium 112 0.020±0.0015 0.019±0.0018 Cadmium 114 0.020±0.0009 0.019±0.002 Lead 206 0.013±0.0009 0.019±0.0011 Lead 207 0.014±0.0005 0.019±0.004 Lead 208 0.014±0.0006 0.019±0.002 

NOTE: Data obtained from Ref. 12. 

a The dilution of the sea-water during the adjustment of pH produced 10 mL samples containing 9 mL of sea-w

samples containing 27 mL of sea-water. Samples containing 9.0 mL of CASS-2, n=5; samples containing 27

n=3. 

b 95% confidence limits 
SW-846 Update V 6020B - 28 Revision 2 

October 2012 

TABLE 4 

ANALYSIS OF NIST SRM 1643b - TRACE METALS IN WATER a

 CONCENTRATION (ng/mL) b

ELEMENT ISOTOPE DETERMINED CERTFIED 

Manganese 55 30±1.3 28±2 

Nickel 58 50±2 49±3 

Cobalt 59 27±1.3 26±1

Nickel 60 51±2 49±3 

Copper 63 23±1.0 21.9±0.4 

Zinc 64 67±1.4 66±2 

Copper 65 22±0.9 21.9±0.4 

Zinc 66 67±1.8 66±2 

Cadmium 111 20±0.5 20±1 

Cadmium 112 19.9±0.3 20±1 

Cadmium 114 19.8±0.4 20±1 

Lead 206 23±0.5 23.7±0.7 

Lead 207 23.9±0.4 23.7±0.7 

Lead 208 24.2±0.4 23.7±0.7 

NOTE: Data obtained from Ref. 12. 

a 5.0 mL samples, n=5 

b 95% confidence limits 
SW-846 Update V 6020B - 29 Revision 2 

October 2012 

TABLE 5 

COMPARISON OF TOTAL MERCURY RESULTS IN HEAVILY CONTAMINATED SOILS 

Mercury in μg/g 

Soil Sample ICP-MS CVAA 

1 27.8 29.2 

2 442 376 

3 64.7 58.2 

4 339 589 

5 281 454 

6 23.8 21.4 

7 217 183 

8 157 129 

9 1670 1360 

10 73.5 64.8 

11 2090 1830 

12 96.4 85.8 

13 1080 1190 

14 294 258 

15 3300 2850 

16 301 281 

17 2130 2020 

18 247 226 

19 2630 2080 

NOTE: Data obtained from Ref. 16. 
SW-846 Update V 6020B - 30 Revision 2 

October 2012 

METHOD 6020A 

INDUCTIVELY COUPLED PLASMA - MASS SPECTROMETRY
SW-846 Update V 6020B - 31 Revision 2 

October 2012 

Appendix A 

Summary of Revisions to Method 6020 (From Revision 1, February 2007):

1. Improved overall method formatting for consistency with new SW-846 methods style guidance. 

2. Section 1.6 - inserted references to additional 3000 series preparatory methods to ICP 

analysis. Also added method 6800 to sections 1.6 and 9.2 as a preparatory method. 

3. Inserted additional safety guidance regarding the use of HF. 

4. Inserted new section (7.27) regarding analysis of non-aqueous solvents. 

5. Reformatted certain paragraphs with the heading "NOTE" or "WARNING" to better denote the 

importance of the recommendations provided therein.

6. Extensively reformatted “REAGENTS AND STANDARDS” section and to meet current SW846 method guidelines. 

7. Significantly updated and expanded “QUALITY CONTROL” section for better adherence to 

current SW-846 method guidelines and for improved alignment with current universal 

practices for published analytical methods. 

8. Inserted new sections (Sections 7.23 and 9.9) to describe the preparation and use of the 

spectral interference check (SIC) solution; also added instructions to match the matrix of this 

solution to that of the calibration standards. 

9. Renamed "QC standard" as "ICV standard" in Sec. 7.24. 

10. Added new Sec. 7.25 describing the preparation of a "CCV" standard, consistent with the 

equivalent section in 6010. 

11. Removed all references to method 7000 except for guidance regarding the method of 

standard addition. 

12. The term “accuracy” was replaced by “bias” where appropriate. 

13. In Section 9.4, the requirement to repeat the demonstration of proficiency for new staff and 

instrumentation changes was changed to a recommendation. 

14. The section regarding analysis of reference materials (Sec. 9.7.4) was revised for clarity and 

the term “Standard Reference Material” was replaced with “reference material” throughout the 

method. 

15. Inserted new section (Sec. 9.8) describing the preparation and use of an LLOQ standard. This 

section includes two new references for guidance on assessing precision and bias. 

16. The section describing matrix interference check samples (Sec. 9.13) has been revised for 

clarity. The post-digestion MS is only recommended if a high concentration sample is not 

available for performing the dilution test. 

17. Substituted certain terms with new terms (i.e. “must” in place of “shall”) to conform with the 

Performance-based Methods Approach goal of flexibility. 

18. Removed reference to “linear dynamic range” as noted by the Inorganic Methods Work Group. 

Section 9.6 regarding the linear range was added. 

19. Mid-level read back or verification standard added to Section 10.5.3. 

20. Tables 3 and 4 from 6020A presenting example precision and accuracy data for aqueous and 

solid matrices were removed.