NIST Measurement Services:

A Calibration Service for

Current Transformers

NlST

Special

Publication

250-36

John D. Ramboz and Oskars Petersons

U.S. Department of Commerce

National Institute of Standards and Technology

NIST MEASUREMENT SERVICES:

A Calibration Service for

Current Transformers

John D. Ramboz and Oskars Petersons

Electricity Division

Electronics and Electrical Engineering Laboratory

National Institute of Standards and Technology

Gaithersburg, MD 20899

June 1991

U.S. DEPARTMENT OF COMMERCE, Robert A. Mosbacher. Secretary

NATIONAL INSTITUTE OF STANDARDS AND TECHNOLOGY

John W. Lyons, Director
National Institute of Standards and Technology Special Publication 250-36

Natl. Inst. Stand. Technol. Spec. Publ. 250-36, 60 pages (June 1991)

CODEN: NSPUE2

U.S. GOVERNMENT PRINTING OFFICE

WASHINGTON: 1991

For sale by the Superintendent of Documents, U.S. Government Printing Office, Washington, DC 20402-9325
INTRODUCING THE ELECTRONICS AND

ELECTRICAL ENGINEERING LABORATORY

The Electronics and Electrical Engineering Laboratory (EEEL) is the component of the National

Institute of Standards and Technology charged with providing Institute activities and services

supporting electrotechnology. In response to identified and anticipated needs, Laboratory programs

provide national reference standards, measurement methods, supporting theory and data, and

traceability to national standards. Beneficiaries are U.S. industry, other Government organizations,

the academic community, and society at large. In fiscal year 1990, the Laboratory calibrated nearly a thousand transfer standards for some 200 customers. The Laboratory is organized into four technical divisions, with the Electricity and

Semiconductor Electronics Divisions at Gaithersburg, Maryland and the Electromagnetic Fields

and Electromagnetic Technology Divisions at Boulder, Coloraudo. The Laboratory provides

measurement research and services for:

• Conducted electrical signals and energy in the Electricity Division: — basic reference standards for electrical measurements — dielectrics for advanced power systems — conducted electromagnetic interference — waveform synthesis — signal acquisition and processing — automatic test equipment — calibration/testing strategies

• Radiofrequency signals and interference in the Electromagnetic Fields Division: — radiofrequency to millimeter-wave electromagnetic fields and signals — general microwave and millimeter-wave instruments, components — microwave and millimeter-wave noise — microwave and millimeter-wave antenna systems — radiated electromagnetic interference; electromagnetic compatibility — monolithic microwave/millimeter-wave integrated circuits — materials properties measurements at high frequencies

• Lightwaves, cryogenics, and magnetics in the Electromagnetic Technology Division: — optical fibers; integrated electro-optics — superconducting materials — superconducting electronics — magnetic materials

• Semiconductor Electronics, in the Semiconductor Electronics Division: — semiconductor materials, such as silicon and gallium arsenide — semiconductor materials modification — semiconductor devices — integrated-circuit test structures for rapid process assessment

For information, contact the Laboratory at 1-301-975-2220 or write to: Electronics and Electrical

Engineering Laboratory, Metrology Building Room B-358, National Institute of Standards and

Technology, Gaithersburg, MD 20899.

Hi
J
— PREFACE —

Calibrations and related measurement services of the National Institute of Standards and Technology

(NIST) provide the means for industry, government agencies, academia, and others to relate the

measurements they make to national standards and, by extension, to the measurement systems

of other countries throughout the world. It is of crucial importance for the improvement of U.S. productivity, efficiency, and quality—in short our competitive stance—that measurements of

appropriate accuracy can be made at all stages of product development, production, and marketing

and that it is possible to refer the results of these measurements to each other in a meaningful

way. NIST services provide essential support for satisfying these needs, as well as measurement

needs arising from societal concerns such as health, safety, natural resources, and defense. The

requirements of the NIST clientele range from the highest levels of accuracy realizable from advanced

science and technology to the practical levels of accuracy needed to support routine production. A

reference- level measurement may be the result of days of painstaking individual effort or may be

provided by a semi-automatic measuring system in minutes. The variation in customer needs and

hence in the NIST services responsive to those needs is great.

The more than 300 different calibration services, measurement assurance services, and special

tests that are offered by NIST are described iri NIST Special Publication 250, NIST Calibration

Services Users Guide. The Guide provides essential technical details of NIST calibration and related measurement services, such as levels of accuracy provided by NIST and the requirements to be met

by customers' transfer standards. It also provides information needed for placing orders for these

services and identifies technical and administrative contacts for each service.

Technical descriptions in the Users Guide are intended to be restricted to the material needed

by a potential customer to decide if a given service will meet that customer's needs. Frequently, a

customer may be interested in more detailed and extensive information, such as the way in which the

errors associated with a measurement are assessed. Beginning in 1987, NIST established a Special

Publication 250 series which supplements the Guide. Each publication in this series provides a

detailed technical description of a specific NIST calibration service, or closely related set of services,

and includes:

• specifications for the service

• design philosophy and theory

• description of the NIST measurement system used to provide the service

• NIST operational procedures

• measurement uncertainty assessment, including error budget, and identification of systematic

and random errors

• NIST internal quality control procedures

Special Publication 250-36, A Calibration Service for Current Transformers, describes

services offered under NIST Test Numbers 54520C through 54522C (current transformers). These

services are provided by the Applied Electrical Measurements Group in the Electricity Division of

the Electronics and Electrical Engineering Laboratory as part of a series of services offered by the

Division in support of the transmission and distribution of electric power.

V
\
Table of Contents

1 INTRODUCTION 1

1.1 General Requirements 1 1.2 Principles of Transformers and Nature of Their Errors 3

1.3 Terms Related to Transformer Calibration 5

1.4 Other Precision Transformer-Like Devices 7

1.4.1 Current Comparator 8

1.4.2 Two-Stage Transformer 9

1.4.3 Amplifier-Aided Transformer 9

1.4.4 Compensated Current Comparator 9

2 CALIBRATION SYSTEM AT NIST 12

2.1 Calibration Circuits 12

2.2 Details of Compensated Current Comparators and Calibration

Hardware 14

2.2.1 Low-Range Current Comparator 17

2.2.2 Mid-Range Current Comparator 17

2.2.3 High-Range Current Comparator 17

2.2.4 Sensitivity Considerations 18

2.2.5 Accuracy Evaluation of Current Comparators 19

2.3 Test Set 20

2.4 Burdens 21

2.5 Sources of Excitation 22

3 CURRENT TRANSFORMER CALIBRATION PROCEDURES 22

3.1 Summary of Procedures 22

3.2 Initial Preparation 23

3.3 Demagnetization 23

3.4 Burden Selection 25

3.5 Current Ratio and Phase-Angle Data Collection 28

3.6 Summarization of the Measurement Results 29

3.7 Historical Data and Expected Performance 34

3.8 The Report of Calibration 34

4 MEASUREMENT UNCERTAINTY ANALYSIS 38

4.1 General Considerations 38

4.2 Accuracy Evaluation of NIST Standards and Equipment 38

4.2.1 Current Comparators 38

4.2.2 Test Set 41

4.3 Errors Originating in Transformers Being Tested 41

vii
4.4 Overall Measurement Uncertainties 44

4.5 Special High-Accuracy Transformers 45

4.6 International Comparisons 46

5 ACKNOWLEDGMENTS 46

6 APPENDIX 47

7 REFERENCES 51

viii
List of Figures

1. Two-Winding Transformer 3

2. Current Transformer Connected to Source and Burden 4

3. Phasor Diagrams Illustrating Terms Used With Instrument

Transformer Ratios 7

4. Current Comparator Used to Calibrate a Current Transformer .... 8

5. Two-Stage Current Transformer 10

6. Amplifier- Aided Transformer 11

7. Compensated Current Comparator 11

8. Current Transformer Calibration With a Compensated Current

Comparator, Secondary Feed 12

9. Current Transformer Calibration With a Compensated Current

Comparator, Primary Feed 15

10. Calibration of a Current Transformer Having a Primary-to- Secondary

Current Ratio Smaller Than One 16

11. Test Set Used to Sense Secondary Current in the Current Comparator

and to Generate a Correction Current 20

12. Demagnetization Circuit 24

13. Current Transformer Calibration Typical Data Sheet 30

14. Current Transformer Calibration Spreadsheet for Calculation of Final

Results 32

15. Ratio and Phase-Angle Uncertainty as a Function of Secondary Current 33

16. Historical Data of Transformer Calibration 34

17. Typical Report of Calibration 36

ix
List of Tables

1. Turns Arrangement in the Low-Range Compensated Current

Comparator 17

2. Turns Arrangement in the Mid-Range Compensated Current

Comparator 18

3. Turns Arrangement in the High-Range Compensated Current

Comparator 19

4. Standard ANSI/IEEE C57. 13-1978 Burdens 26

5. Systematic Uncertainties in NIST Equipment 39

6. Estimated Possible Instabilities in a Typical Transformer Under Test 42

7. Calibration History of a Typical Laboratory Grade Current

Transformer, Marked Ratio 10 A:5 A 44

X
A CALIBRATION SERVICE FOR CURRENT

TRANSFORMERS

A calibration service at the National Institute of Standards and Technology

(NIST) for laboratory-quality current transformers is described. The service

provides measurements of the current ratio and the phase angle between the

secondary and primary currents. In the Report of Calibration or Test, the

measured ratio is reported as the product of the marked (nominal) ratio and

the ratio correction factor. The measured phase angle is reported directly in

miUiradians (mrad) and is positive if the secondary current leads the primary.

The range of primary-to- secondary current ratios that can be measured

with the equipment at NIST extends from 0.25 A:5 A to 12000A:5A. The

maximum current at the present time is about 20 000 A. Estimates of calibration

uncertainties, including their sources, are given and quality control procedures

are described. For routine calibrations, uncertainties of ±0.01% for the ratio

and ±0.1 mrad for the phase angle are quoted. However, lower uncertainties

to ±0.0005% or 5 parts per million (ppm) for ratio and ±0.005 mrad or 5 //rads

for phase angle—are possible under the provisions of Special Tests. Key words: calibration; current transformers; electric power; electric standards;

NIST calibration service.

1.1 General Requirements

The accurate measurement of electric energy at currents typically found in utility

distribution systems relies on three key components: a watthour meter, a voltage

transformer, and a current transformer. The voltage and current transformers used in

this application are known as instrument transformers. For polyphase power systems,

multiple transformers and meters may be used. Errors in any of these devices can lead

to energy measurement errors. The National Institute of Standards and Technology

John D. Ramboz

Oskars Petersons

Abstract

1 INTRODUCTION
2 INTRODUCTION

(NIST) provides calibration services for laboratory standards for each of these devices.

This Special Publication describes the service offered for the calibration of current

transformers. Other NIST publications describe calibration services for wattmeters

and watthour meters [1] and voltage transformers [2].

In measurement applications, one must know the true current ratio of the current

transformer and also the phase angle between the current phasors. In the Report of

Calibration (or the Report of Test) [3],^ the measured ratio is reported as a product

of the marked (nominal) ratio and the ratio correction factor (RCF). The measured

phase angle is reported directly in milliradians (mrads). The phase angle is positive if the secondary current leads the primary current. Depending upon the quality of the

current transformer and its intended use, the calibration results can be used to apply

the corrections to the nominal ratio, or to verify that the true ratio of the transformer

is sufficiently close to the nominal ratio that no corrections are required. Under ideal

conditions, the actual current ratio is exactly equal to the marked (nominal) ratio

and there is no phase angle displacement between the primary and the /secondary

currents.

The need exists to measure currents in power circuits over a wide range, typically from

a few amperes to several thousand amperes and even higher in special applications.

Most current transformers operate adequately over a 10-to-l dynamic range. In

addition, multiratio transformers that cover a large range of current ratios are in use by electric utilities. Many 50-Hz and 60-Hz transformers are used for metering and

relaying applications. These can be calibrated at NIST for the range of the primaryto-secondary current ratios from 0.25 A:5 A to 12 000A:5A. The maximum available

current at the present time is 20 000 A, limited by the power supply. Military, marine,

and aerospace applications, often employ 400- Hz instrument transformers. These can be calibrated to the maximum current of about 1200 A.

The relevant American National Standard, ANSI/IEEE C57. 13-1978 [4], mentions

relatively modest accuracies, with the calibration uncertainties to ±0.1% for current

ratios and iblmrad for phase angle and corrections to ±0.3% and ±5mrad

for instrument transformers in actual metering applications. Lower calibration

uncertainties, typically ±0.01 % and ±0.1mrad, are required for instrument

transformers that are used as standards to calibrate other transformers. Besides

for metering and control applications, instrument transformers and similar devices

are used in impedance bridges and test sets to measure the efficiency of electrical

equipment. The accuracy requirements in those applications may extend to

uncertainties as low as a few parts per million (ppm) and microradians (/^rad).

This Special Publication discusses the NIST facility for the calibration of current

transformers, the calibration procedures used, the methods employed to verify proper

functioning of the calibration system, and the errors in the calibration process. In

addition, this Introduction contains a brief presentation of transformer principles,

^Consult the cited reference for the differences between the two types of reports
1.2 Principles of Transformers and Nature of Their Errors 3

a discussion of the various types of precision transformers and the nature of their

errors, and a summary of the terminology pertaining to current transformers. The

Appendix presents information for submitting transformers to NIST for caUbration.

1.2 Principles of Transformers and Nature of Their Errors

A typical laboratory-grade current transformer consists of a toroidal, ferromagnetic

core on which are placed the primary and the secondary windings with A'^i and N2

turns, respectively, as in figure la. The core is constructed by winding a continuous

ferromagnetic tape in a toroidal form. High permeability is desirable over a wide

range of flux densities to ensure linear operation and low errors. Hence, nickel-iron

alloys are used as the core materials in the most demanding applications. The design

objective is to achieve in a practical transformer the characteristics that approach

those of an ideal transformer, namely, the actual current ratio being nearly equal to

the reciprocal of the turns ratio, i.e.,

With modern materials and construction techniques, the ideal characteristics can be approached to a high degree to within ±0.01% and even better in some cases. However, imperfect magnetic coupling between the windings and energy losses in the

windings and the core prevent attaining fully the conditions of an ideal transformer.

/,//2«Arj/Ar,. (1)

(a) Circuit Diagram

(b) Equivalent Circuit

Figure 1. Two-Winding Transformer
4 INTRODUCTION

The conventional equivalent circuit of an actual transformer based on an ideal

transformer and a T-network, figure lb, explains why the actual transformer deviates

from ideal. While the T-network representation applies to any network with two

pairs of terminals, it is particularly useful for low-frequency transformers, since the

equivalent impedances have meaningful physical counterparts. Zm is the magnetizing

impedance; Zi and Z2 are the primary and secondary leakage impedances; and

n = N2/N1 is the ratio of the ideal transformer. While Zi and Z2 are reasonably

constant, Zm is not as a result of the nonlinear nature of the magnetic core. The actual operation of a current transformer is illustrated in figure 2a. At the left is a current source whose output is being measured. At the right is a measuring device

with its own internal impedance, Zb, referred to in instrument transformer practice

(a) Circuit Diagram

^ 1 ^ 2 I 2

1 : n

(b) Equivalent Circuit

Figure 2. Current Transformer Connected to Source and Burden

as the burden impedance. Substituting the equivalent circuit of figure lb for the

transformer in figure 2a, the equivalent circuit is obtained and shown in figure 2b. In

that circuit, the actual current ratio becomes

h/h = (iV2/iVi)[l -f- {Z2 + ZB)/Zm]. (2)

The above is a complex ratio having real and imaginary parts accounting for the

ratio correction factor and the phase angle. The above equation indicates that the

magnetizing, the secondary leakage, and the burden impedances all affect the actual
1.3 Terms Related to Transformer Calibration 5

ratio. The primary leakage impedance, Zi, has no effect on the current ratio. In

practice, the designer of a current transformer will minimize the ratio corrections and

phase angle subject to other construction, 6perational, and economic constraints.

In operation, the smallest possible burden impedance is desirable. Because of the

nonlinear nature of the transformer {Zm being current dependent), the corrections are affected by the current and the burden impedance.

1.3 Terms Related to Transformer Calibration

When employing an instrument transformer in critical measurement applications, the

user is interested in the magnitude of the true ratio of the primary to the secondary

currents and the phase angle between them. There are several ways to express such a complex ratio. Furthermore, there are variations in the use of terms that relate to

measurement errors and corrections. In this section, several frequently used terms

are explained and defined. Inasmuch as possible, the terminology developed through

consensus standards is retained.

The following terms are consistent with the formally defined terms of ANSI/IEEE

C57.13-1978:

Marked ratio. The ratio of the rated primary current to the rated secondary current

as stated on the name plate.

Ratio correction. The difference between the ratio correction factor and unity. It

is usually expressed in percent, but can also be expressed as a fraction (in per

unit) or, for very accurate transformers, in parts per million (ppm).

Ratio correction factor (RCF). The ratio of the true transformer ratio to the marked

ratio. The primary current is equal to the secondary current multiplied by the

marked ratio times the ratio correction factor.

Phase angle. The phase displacement between the primary and the secondary

currents. It is positive when the phasor of the secondary current leads that

of the primary current.

True ratio. The ratio of the primary current to the secondary current under specified

conditions.

Turns ratio. The ratio of the primary winding turns to the secondary winding turns.

The turns ratio may or may not be numerically equal to the reciprocal of the

marked ratio. To bring the ratio correction factor nearer to unity, turns may

be added to or deleted from one of the windings.
6

Besides the above, the following terms are also found in the technical literature:

Nominal ratio. This term is usually used as a simple number and is sometimes

substituted for the marked ratio. For example, the marked ratio of a current

transformer might be 1200 A:5 A, the corresponding nominal ratio is 240 or

240:1.

In-phase correction. The in-phase component between the phasors representing the

secondary current and the nominal secondary current (primary current divided

by the nominal ratio). This correction is normalized by dividing it by the

secondary current and it is nearly the same as the ratio correction defined

previously.

Quadrature correction. The normalized quadrature component between the phasors

representing the secondary current and the nominal secondary current. For

small phase angles, the quadrature correction is nearly numerically equal to the

phase angle in radians.

The preceding definitions can be illustrated by the use of phasor diagrams. The

mathematical ratio of the primary and secondary current phasors of a current

transformer is illustrated in figure 3 both in polar and Cartesian coordinates. In

polar coordinates the relationship is given by

/i//2 = n(l + a)e-^ (3)

where Ii and I2 are the primary and secondary currents, respectively, n is the marked

ratio; (1 + a) is the ratio correction factor; a is the ratio correction; and b is the

phase angle in radians. The phasor diagram of figure 3a illustrates the quantities in

eq. (3). The Cartesian form of the relationship is readily obtained from eq. (3)

/1//2 = n(l + a)(cos 6 — j sin6) = n(l 4- fli — i^i), (4)

where Oi and bi are usually denoted as in-phase and quadrature corrections,

respectively. For small angles, a and ai are nearly equal, as are b and 61. The

diagram of figure 3b illustrates the Cartesian coordinate representation.

The transformer-specific definitions for the general terms "error" and "correction"

need clarification. In the technical literature on instrument transformers, the term

"error" has two meanings-(a) an uncertainty in the knowledge of the true ratio;

and, (b) a known deviation from the marked (nominal) ratio or the turns ratio.

In this report, the term "error" used in the former sense, i.e., for uncertainties in

the measurement is always referred to as an "uncertainty." For known differences

between true and nominal ratios, the term "correction" is used in the sense that

the correction is added to the nominal value to obtain the true value. There are

cases in instrument transformer application where it is not practical or necessary to
1.4 Other Precision Transformer-Like Devices 7

Ij = nl 2 (l+a)e

I

2 ( 1 + a ) e

(a) Polar Coordinate Representation

I 1 = n I

2

( 1 + a J- j b j)

I 2 ( 1 + a 1 - j b

(b) Cartesian Coordinate Representation

Figure 3. Phaser Diagrams Illustrating Terms Used With Instrument Transformer Ratios

apply known corrections to the measurement or calibration results, because either

the corrections are small and thus relatively insignificant or the application is too

complicated. In those cases, the nominal ratio is used but the error of the transformer

is estimated by also considering the known but unapplied corrections.

1.4 Other Precision Transformer-Like Devices

The technology of precision transformers used as ratio devices for voltage and current

in various applications has been highly developed. In fact, transformers are generally

considered the most accurate scaling devices at power and audio frequencies with

stabilities in the range of parts in 10^ and corrections in the range of parts in

10^. These highly accurate transformers and transformer-like devices have influenced

current transformer calibration practice. Standard transformers and calibration

equipment incorporate such devices.

Precision transformers have increased the accuracy demands on calibration

laboratories by one to two orders of magnitude. A large number of publications

on specialized transformers are available in the technical literature. A reference book
8

TRANSFORMER

BEING

CALIBRATED

CURRENT

COMPARATOR

MAGNETIC

SHIELD

I1/I2 = /N21

Figure 4. Current Comparator (at Right) Used to Calibrate a Current Transformer (at

Left)

by Moore and Miljanic [5] describes some of them and also provides further references.

Four specialized types of transformers are summarized here.

1.4.1 Current Comparator

The current comparator is a three-winding current transformer (figure 4 at right)

operated with zero flux in the core. The current is supplied to two of the windings,

A^2i and N22, and the presence of magnetic flux is monitored via the induced voltage

in the winding 7V23. At the ampere-turn balance {I1N21 = I2N22) there is zero voltage

and current at the detector (D). The correspondence between the current ratio and

turns ratio is very high, at least within a 1 ppm for well-designed devices. To insure

that the windings A^2i and N22 are coupled to N2Z only through the flux in the core

(a necessary condition for high accuracy), magnetic shielding is required to eliminate

the adverse effects of the leakage flux between the windings as shown.

A current comparator can be used to calibrate a regular two-winding current

transformer as shown in figure 4. Additional circuitry is required to balance the

calibration circuit if the current transformer at the left has ratio and phase angle

corrections.
1.4 Other Precision Transformer-Like Devices 9

1.4-2 Two-Stage Transformer

A regular current transformer can be combined with a three-winding transformer

operated in a differential mode (figure 5a). This transformer can then be used as a

reference standard. The three-winding transformer supplies the error current of the

regular current transformer and the total secondary current approaches that value

which is dictated by the turns ratio. A separate burden is required for the threewinding transformer. The three-winding transformer is magnetically shielded with

the considerations for shielding identical to those of the current comparator. The

entire device can be constructed as a single transformer with the two cores placed

adjacent to each other (figure 5b) and by winding A'^i and 7V2 on both cores and

A^3 only on one core. In the diagram, a winding links all the cores to the right of

the winding but not those cores to the left of the winding. The two-stage technique

results in very accurate transformers that have corrections of a few parts per million

or even smaller.

1.4.3 Amplifier-Aided Transformer

Numerous circuits combining transformers with amplifiers have been used for

improving the accuracy of transformers. An elementary circuit is shown in figure 6. The amplifier senses the residual flux in the core via the winding A^3 and then aids the

transformer by forcing the current through the secondaxy winding to achieve ampereturn balance. In this particular circuit the amplifier supplies nearly all the power

to the secondary winding and the burden. In other, more complicated schemes, the

amplifier supplies only the correction current. The amplifier- aided transformer may

be used as a reference standard. Separate provisions must be made to balance the

calibration circuits.

1.4-4 Compensated Current Comparator

This transformer is constructed by adding a detection winding to the two-stage

current transformer as well as a source of power in series with the compensation

winding (figure 7). The core at the right is operated in zero-flux condition, hence,

the instrument is a true ampere-turn balance sensor. The adjustable voltage source,

S, is required in the compensation winding in order to supply the current to this

winding and balance the null detector, D. The power to the primary and secondary

windings of the compensated current comparator can be supplied from a source

in the primary winding or from a source in the secondary winding. The device

is very accurate, versatile, and well-suited for calibration of current transformers.

Current transformer calibrations at NIST employ compensated current comparators

as reference standards.
10 INTRODUCTION

TR 1

TR2

Nll/Ni2= N21/N22

N

2 2 = N

2 3 ll/ (I 2 +I3) = N2 2/^21= ^12^^ 11

(a) As Two Separate Transformers

N2 = N

3

I 1 / ( I 2 + I 3) = N 2/ N

1

(b) As One Unit

Figure 5. Two-Stage Current Transformer
1.4 Other Precision Transformer-Like Devices 11
12 CALIBRATION SYSTEM AT NIST

2 CALIBRATION SYSTEM AT NIST

2.1 Calibration Circuits

Current transformers at NIST are calibrated by comparing the customer's transformer

(referred to as the test transformer) to a standard current comparator whose

corrections are negligibly small. In routine calibrations, the compensated current

comparator shown schematically in figure 7 is used as the standard. A complete

calibration circuit is shown in figure 8. In the circuit of figure 8, the primary windings

of the test transformer, A^i, and the current comparator, A^2i? are connected in series

opposition. The secondary windings, N2 and N22, are also in series opposition. In

COMPENSATED

CC

2 1

TEST SET

0 . in

TEST

CT

BURDEN

"23

N22

"22

N 2 / N 1 = n

n [ l-I c /( I 22 + 1 23 n(l-I, /I,]

) ]

-- CAPACITIVE ERROR COMPENSATION

Figure 8. Current Transformer Calibration With a Compensated Current Comparator,

Secondary Feed
2.1 Calibration Circuits 13

addition, the compensation winding and the output of the test set are connected

to the junction of A^2 and A^22- The power source and a current- sensing shunt for

the input of the test set are in series with A^22- For a test transformer having no

corrections, the secondary current of the test transformer is equal to the combined

secondary and compensation currents of the compensated current comparator,

h = hi + -^23, (5)

and the correction current, /c, of the test set must be zero for the null condition at D, the detector. If the test transformer has a significant correction, eq (5) must be

modified,

h = hi + /23 + h- (6)

In accordance with eqs (3) and (4) defining transformer corrections

h = hlHl + a)e-i'']h « /i/[n(l + a - jh)]. (7)

The combined currents /22 and hi represent the nominal secondary current. Thus

h2-\-h3 = Ii/n. (8)

The correction current, h, is a small, adjustable fraction of the nominal secondary

current. It can be defined as

/e = (/i/n)(-a + i^). (9)

Substituting eqs (7-9) in eq 6, yields

l/(l + a-j6) = l-a + i/9. (10)

If the correction terms are small,

a ^ a, and 6 « /?. (11)

The adjustments of the test set, a and are designed to be direct reading for the

ratio correction and phase angle of the transformer.

The circuit shown in figure 8 is generally used for primary currents ranging between

5 and 1200 A (nominal ratios between 1:1 to 240:1). This configuration in which the

supply is connected in the transformer's secondary circuit, is known as the secondary

feed configuration. The supply is required to provide only the secondary current.

Under normal circumstances, this does not exceed 5 A. However, the voltage might

be relatively high, especially for the larger ratios and higher primary currents. The

mid-range compensated current comparator that is mainly used with the secondary

feed can be excited up to 200 V at 50 and 60 Hz when used in this configuration.

This is generally adequate for the largest ratios and the largest primary currents.
14 CALIBRATION SYSTEM AT NIST

The compensated current comparator serves both as a precision ratio device and as a power transformer, transferring in this case the power from the secondary to the

primary winding. The power requirements in the primary circuit are usually larger

than in the secondary circuit. Hence, the compensated current comparator in the

secondary feed configuration has to transfer more power. This, in turn, could lead

to larger error in the measurement system. Relatively large voltage in the secondary

winding causes capacitive currents that result in an error in the current ratio of a

current comparator. The compensation current in the secondary-feed configuration can be high causing significant voltage at the junction of the secondary windings. This

voltage imposes an additional burden on the test transformer. Nevertheless, these

shortcomings can be controlled in current comparators that have moderate ratios as indicated previously.

For the primary currents greater than 1200 A, the circuit of figure 9 is used in which

the power source is connected in series with the primary windings. This is known as the primary feed configuration. It is more accurate and is the only viable configuration

for large currents in excess of a few thousand amperes. Calibration of transformers

whose ratios are smaller than one presents special problems because of the large

impedance that is reflected into the primary winding and potentially large capacitive

currents. The difficulty is overcome by adding an auxiliary current transformer to

the circuit as shown in figure 10. This transformer supplies most of the power to

the secondary circuit thus improving the performance of the compensated current

comparator. Calibrations are routinely performed at 50, 60, and 400 Hz. They can be performed at other frequencies on a Special Test basis. Three compensated current

comparators are employed in routine calibrations covering the range of ratios from

0.25 A:5 A to 12 000 A:5 A. The maximum current is limited by the power supplies to

about 20 000 A for 50 and 60 Hz, and 1200 A for 400 Hz. The transformers are usually

calibrated with the secondary currents between 0.5 A and 5.0 A (10% to 100% of the

rated current), but calibration with lower currents is possible, sometimes at reduced

accuracy. Short-term overcurrent tests usually can be performed at a 10-A secondary

current.

Details of the test set and burdens used at NIST are discussed in the following sections.

2.2 Details of Compensated Current Comparators and

Calibration Hardware

Three compensated current comparators are available for routine calibration of ratios

that are normally requested by NIST clients. These are low-range, mid-range, and

high-range current configurations. All the current comparators are constructed in

accordance with the diagram of figure 7. All are rated for a secondary current of 5 A,

but for different primary currents. Brief descriptions follow.
2.2 Details of Compensated Current Comparators and Calibration Hardware 15

POWER

SUPPLY

COMPENSATED

CC

2 1

TEST SET

0 . in

TEST

C T

BURDEN

N22/N2i= ^2 /Nj = n

Ij / I2 ^ n [ 1 - I f / ( I 77 + I 2 2^^ 23' n[l-I

^ /I2)

Figure 9. Current Transformer Calibration With a Compensated Current Comparator,

Primary Feed
16 CALIBRATION SYSTEM AT NIST

POWER

SUPPLY

AUXILIARY COMPENSATED

CT CC

TEST

CT

2 1

TEST SET

0 . Ifl

BURDEN

N22 /N21= N32/N 31 = N 2 /N 1

II / I2 ^ n [ 1 -I c

/ ( I 22 + I 23) 1 ^ n [ 1 - I /I

c 2

= n

Figure 10. Calibration of a Current Transformer Having a Primary-to- Secondary Current

Ratio Smaller Than One
2.2 Details of Compensated Current Comparators and Calibration Hardware 17

Table 1. Turns Arrangement in the Low-Range Compensated Current Comparator

The primary winding has switch- selectable taps. The full-scale secondary current is 5 A.

FULL-SCALE

PRIMARY CURRENT

(Amperes)

NUMBER OF TURNS

Primary Winding Secondary Winding

0.25 240 12

0.5 120 12

0.75 80 12

1.0 60 12

1.25 48 12

1.5 40 12

2.0 30 12

2.5 24 12

3.0 20 12

3.75 16 12

4.0 15 12

5.0 12 12

2.2.1 Low-Range Current Comparator

This current comparator is a "step-up" device, having a primary current that is smaller than the secondary current. The secondary and compensation windings each

have 12 fixed turns. The primary winding has switch-selectable taps with the number

of turns ranging from 12 to 240. The available turns ratios and primary currents are

listed in table 1. The auxiliary transformer has the same turns ratio, but the number

of turns is eight times greater with 96 in the secondary winding.

2.2.2 Mid-Range Current Comparator

The number of selectable turns in the secondary and compensation windings are 160,

200, and 240. The permanently placed primary turns range from 8 to 160 and are made available through series-parallel connection. Additionally, feed-through primary

turns (turns through the window) are used for larger ratios. The turns arrangements

and typical primary currents are tabulated in table 2.

2.2.3 High-Range Current Comparator

This current comparator is designed for primary currents rated to 12000 A having a maximum of 2400 secondary turns and from 1 to 24 primary turns. The compensation

winding is fixed at 1200 turns. A large number of nominal ratios and primary
18 CALIBRATION SYSTEM AT NIST

Table 2. Turns Arrangement in the Mid-Range Compensated Current Comparator

The secondary and compensation windings can have 160, 100, or 240 turns each. The

number of turns in the fixed primary winding can be selected between 8 and 260 by means

of series-parallel interconnections. Additional primary turns can be added through the

window in the transformer. The full-scale secondaxy current is 5 A. Typical rated primary

currents are given in the table below.

SECONDARY WINDING

PRIMARY WINDING (Number of Turns)

(Number of Turns) 160 200 240

FuU- Scale Primary Current

Fixed Winding (Amperes)

160 5 7.5 80 10 12.5 15

40 20 25 30

32 37.5

20 40 50 60

16 75

10 80 100 120

8 125 150

Turns Through

Window

5 160 200

4 250 300

2 400 500 600

1 800 1000 1200

currents are available as shown in table 3 by utilizing the different combinations

of the number of turns in both windings. A small current transformer is cascaded to

the compensation winding to obtain an effective number of compensation turns from

1200 to 2400.

2.2.4 Sensitivity Considerations

The sensitivity of current comparators can be expressed in a form of a transimpedance

per turn, defined as the ratio of the voltage in the detection winding to the ampereturns of excitation in the ratio windings. For the sizes and types of magnetic cores that are used in typical current comparators, the transimpedance is about 10 ohms

per turn (10 volts per ampere-turn). The actual measured values are as follows: for

the low-range current comparator, 6.5 ohms per turn; mid-range, 11 ohms per turn;

high-range, 10 ohms per turn.
2.2 Details of Compensated Current Comparators and Calibration Hardware 19

Consider the following example: assume the sensitivity figure of 10 volts per ampereturn, one ppm unbalance in a 0.5- A secondary current, and a 100-turn secondary

winding. These quantities yield a 500-/iV signal at the terminals of the detection

winding. This signal is about two orders of magnitude larger than the noise and

interference level in a typical laboratory operation of this type. Obviously, the

sensitivity of current comparators is not a problem in current transformer calibration.

Table 3. Turns Arrangement in the High-Range Compensated Current Comparator

The secondary winding has a switch- selectable number of turns from 1200 to 2400 in 20-turn

increments. The same effective number of turns is obtained in the compensation winding

through a fixed 1200-turn winding and a cascaded transformer. The primary winding can have 1 to 24 turns as shown in the table. They are selectable through series- parallel

interconnection. The full-scale secondary current is 5 A. Typical primary currents are tabulated below. A larger number of other ratios are possible with different secondary

number of turns.

PRIMARY WINDING

(Number of Turns)

SECONDARY WINDING

(Number of Turns)

1200 1600 1800 2400

Full-Scale Primary Current

(Amperes)

1 6000 8000 9000 12000

2 3000 4000 4500 6000

3 2000 3000 4000

4 1500 2000 2250 3000

6 1000 1500 2000

8 750 1000 1125 1500

12 500 750 1000

24 250 375 500

2.2.5 Accuracy Evaluation of Current Comparators

The current comparators themselves are the minor contributors to the overall

uncertainty. They are designed to have corrections that are negligible for routine

calibrations. The performance of the current comparators have been evaluated by

indirect tests of the magnetic and capacitive errors. All three current comparators

described above have been rigorously calibrated at selected ratios to vahdate the

results of these indirect tests. The results are presented in section 4 of this report.
20 CALIBRATION SYSTEM AT NIST

RATIO CORRECTION ADJUSTMENT o VW ^ WV -O' + 1 5 V

MULTIPLYING

DAC VOLTAGE-TOSUMMING CURRENT AMPLIFIER AMPLIFIER

3^)

o

CURRENT

HON ITOR

I V = I A

15 V O V/V O +15 V PHASE ANGLE ADJUSTMENT

Figure 11. Test Set Used to Sense Secondary Current in the Current Comparator and to

Generate a Correction Current

2.3 Test Set

The calibration technique requires the injection of a small known current, equal to the

difference between the currents of the current comparator and the test transformer,

at the junction of the secondary windings. This is shown in figures 8, 9, and 10. An

electronic test set is used to sense the secondary current in the current comparator and

generate a correction current given by eq. (9) [6]. Figure 11 shows a block diagram of

the test set. The secondary current, I22, is passed through a precision four-terminal

shunt, thus generating a voltage signal at the input of the test set. The test set input impedance of 20 kO presents negligible loading on the shunt. The test set then generates in-phase and quadrature signals. By the use of multiplying digital-toanalog converters, each signal is separately scaled by an adjustable amount selected

by variable controls on the test set. The signals of two channels are then summed and

converted to a current which is used to bring the current comparator into balance as indicated by a null condition in the detector. With this arrangement, the correction

current is always proportional to the secondary current of the current comparator,

I22. Small changes and drifts in the secondary current therefore do not influence the

balance. Furthermore, I22 is sufficiently close to the nominal secondary current, /i/n,

which satisfies the balanced conditions of eq (10) and the approximation of eq (11).
2.4 Burdens 21

Selectable ranges are available (not shown in diagram) which generate full-scale

balancing currents of 100 ppm, 1000 ppm, and 10000 ppm of the nominal secondary

current /i/n, and phase angles of 100 //rad, 1000 /zrad, and 10000 //rad. The phase and

amplitude of the quadrature current is direct reading for frequencies of 50, 60, and 400

Hz. The frequency selection is made by means of a front panel switch. Transformer

calibrations at other frequencies may be done by scaling the phase reading by the ratio

of the measurement frequency to the frequency indicated by the dial setting. The test

set has two three-digit panel meters which indicate the test transformer correction

and phase angle directly in parts per million and microradians, respectively.

Test set components are selected and internal adjustments made so that the error

in Ic is less than ±0.5% at frequencies of 50 and 60 Hz, and less than ±1% at a frequency of 400 Hz. Because the test set must generate only a current equal to the

correction of the test transformer, the test set output need not be very accurate. If, for example, a test transformer has a ratio error of 100 ppm, and the test set current

is known to within 0.5%, an overall measurement uncertainty of only 0.5 ppm results.

There is a small fixed uncertainty of about 1 ppm attributable to the test set.

2.4 Burdens

Burdens are selected using stable values of resistance and inductance combined in a burden box made at NIST. It is capable of carrying currents up to 10 A. Incremental

values of resistance in 0.01-1) steps up to 1 Q can be selected. External resistance

of 1, 2, and 3 ft can be added for burdens requiring larger impedances. Incremental

values of inductance in 0.01-mH steps up to 1 mH can be similarly selected. External

inductance of 1, 2, 5, and 10 mH can be added for larger burdens.

The value of the burden in terms of resistance and inductance is measured for every

burden selected for calibration. This process is discussed in section 3.4. The secondary

connecting leads and contact resistance contribute to the impedance of the burden

and can change if the transformer is disconnected and reconnected. This is especially

true for near-zero burdens. As the impedance of the burden becomes larger, the

effects of leads and contacts diminish.

The accuracy to which the burden is set varies with the magnitude of the burden. At

very low-impedance burdens, the accuracy is Hmited by the resolution of the burden

box (0.01 n resistance and 0.01 mH inductance). The measurement accuracy for the

resistive part is ±(0.002 Q + 2% of the value). Similarly for the inductive part it is ±(0.002 mH + 2% of the value). For most transformer calibrations, this is sufficient

and will contribute only small errors which are accounted for ia the overall uncertainty

statement assigned to the Report of Cahbration (or Report of Test).
22 CURRENT TRANSFORMER CALIBRATION PROCEDURES

2.5 Sources of Excitation

The test currents are derived from one of two sources, depending on the current

required. For most of the calibrations performed, electronically generated sinusoidal

voltages are obtained from a 1500 volt-ampere (VA) electronic power supply. It has

four selectable voltage ranges of 30, 60, 120, and 240 volts using series and parallel

combinations of transformer windings contained within the supply. It is used at

power frequencies and at audio frequencies up to 1 kHz with low range and midrange current comparators. An electronic counter and a phase-lock circuit are used

to synchronize the generator with the power-line frequency for tests performed at

60 Hz. This helps to reduce beat-frequency problems.

The second excitation source is a 54-kVA electronic power amplifier and special highcurrent supply transformer capable of delivering 20 000 A in its secondary. This

combination drives a coaxial current cage and the high-range current comparator

which operates over current ratios of 250A:5A to 12 000A:5A. The power source can be operated over a frequency range from about 40 to 5000 Hz. At higher

frequencies, the current capability is greatly reduced because of circuit inductances.

An electronic counter is used to measure the test frequency. The test frequency is usually maintained to within 0.1 Hz of the desired frequency during the calibration

process.

3 CURRENT TRANSFORMER

CALIBRATION PROCEDURES

3.1 Summary of Procedures

The procedure for the calibration of current transformers at NIST involves the

following steps:

• initial preparation for the calibration,

• demagnetization of the transformer,

• selection of the test transformer secondary burden,

• determination of ratio correction and phase angle,

• summarization of the measurement results and assessment of measurement

uncertainties,

• comparison of the results with historical and/or expected performance data,

and
3.2 Initial Preparation 23

• preparation of the Report of Calibration (or Report of Test), and final review

and approval.

Each of these is discussed below.

3.2 Initial Preparation

Upon receipt of the current transformer from a customer, the transformer is visually

inspected to ensure that no apparent damage was incurred during transit. If damage

is obvious, the customer is notified and steps are taken to return the device. If damage is suspected, the transformer's performance is verified. Connections and

terminal conditions are also noted. If no significant corrosion or oxide accumulations

on the terminals is obvious, then the calibration process is continued. If only a minor

amount of oxide is present, the terminals may be cleaned before proceeding. If excess oxide is present or corrosion is severe, the customer is notified and the transformer

is returned uncalibrated. No repairs or major cleaning are normally performed by

NIST. In some instances, NIST staff may have suggestions for customers regarding

possible corrective action.

Records are checked to determine if a particular transformer has been previously

calibrated at NIST. If so, the historical calibration data are reviewed to determine if any unusual conditions or performance may have existed. After the calibration has

been completed, the present results are compared to the historical data as discussed

later. The order is reviewed to ensure that the test conditions stated are understood

and can be satisfied; if not, the customer is contacted for clarification.

3.3 Demagnetization

To ensure a reliable calibration, current transformers must have no residual core magnetization. Generally, the demagnetization is a simple and quick procedure which

makes certain that no residual magnetic flux exists in the core. A circuit such as shown in figure 12 is used. An ac power source (variable voltage supply), ammeter,

and voltmeter are connected to the secondary terminals of the current transformer.

The primary circuit is left open.

The source is energized, but left at a zero output. The switch in series with the

ammeter and transformer is then closed, and the source voltage is slowly increased

while the ammeter is being carefully observed. As the voltage increases across the transformer winding, very little current exists because of the high magnetizing

impedance. Above some voltage level, however, the current will rapidly increase as the

transformer core reaches magnetic saturation. The source voltage is slowly increased

until the rated secondary current (usually 5 A) is noted on the ammeter. At the
24

TRANSFORMER

BE ING TESTED

Figure 12. Demagnetization Circuit

same time, the voltmeter reading is also noted. The source voltage is then slowly

decreased to zero, and the switch is opened. The source is then turned off. Both the

current and voltage values are recorded on the laboratory data sheet. This completes

the demagnetization procedure. The switch in the series circuit is a guard against

current transients or surges which might be generated by the power source when it

is turned on and off. If such a surge were to occur after the above demagnetization

procedure was completed, transformer remagnetization might occur. Opening the

switch prior to turning off the source prevents magnetizing surges from reaching the

transformer.

For some transformers, the secondary impedance may be extremely high and may

require several hundred volts for core saturation. Transformers may have multiple

secondary and primary windings, in which case high voltage may also be present at

other transformer terminals. Precautions must be exercised against the hazards of

electric shock.

Occasionally, the ac power source will not be able to develop a sufficiently high voltage

to drive the core into saturation. If additional windings are available, the selection of a winding having fewer turns (and hence a lower impedance) can sometimes be used.

For convenience, the frequency used for demagnetization is generally 60 Hz. However,

when insufficient supply voltage prohibits demagnetization, the source frequency can be reduced to 40 Hz, for example, and another attempt made. Lowering the frequency

lowers the reactive impedance of the transformer winding and the saturation voltage.

The winding and demagnetization frequency used are noted on the laboratory data

sheet.
3.4 Burden Selection 25

For some window-type transformers having a single- fixed secondary, demagnetization

at even 40 Hz may not be achievable within the limits of the source voltage. In this case, a special 40-conductor cable is fed several times through the window forming an

additional winding. Because of the small gauge wire, the current through the cable

is limited to about one ampere. For each turn through the window, 40 ampere-turns

link the transformer core. It is almost always possible to find a configuration which

saturates the core.

3.4 Burden Selection

The process of selecting and setting the burden for the secondary winding of the

current transformer must be completed prior to calibration data collection. The

customer specifies the burden or burdens at which the calibration is to be performed.

Burdens can be specified in any one of the following ways:

1. by specifying the series resistance and inductance,

2. by specifying the volt-ampere load and power factor (a 5-A secondary current

is assumed unless otherwise stated),

3. by requesting a standard ANSI burden as defined in ANSI/IEEE C57.13-1978

(see table 4), or

4. by supplying the burden and leads with the transformer.

If the burden is specified by either (2) or (3) above, the appropriate values of resistance

and inductance are calculated and used. When the burden is given in terms of voltamperes (or, more precisely, average phasor power) and power factor, the resistance

is calculated from the relationship

i2=[5//|](PF), (12)

and the inductance is calculated from the relationship

I = 1000[5M|][l-(PF)2]i, (13)

where

R is the series resistance of the burden (ohms),

L is the series inductance of the burden (millihenries),

S is the volt-ampere product or the average phasor power.
26 CURRENT TRANSFORMER CALIBRATION PROCEDURES

Table 4. Standard ANSI/IEEE C57.13-1978 Burdens

The following table is excerpted from the American National Standards Institute publication

C57.13-1978, American National Standard Requirements for Instrument Transformers [4]. Described are secondary burdens for current transformers with 5-A secondaries. If a current

transformer is rated at other than 5- A, ohmic burdens for specification and rating shall be

derived by multiplying the resistance and inductance of the table by [5/(ampere rating)].

The VA at rated current, power factor, and the burden designation remain the same. These

standard burden designations have no significance at frequencies other than 60 Hz.

STANDARD BURDEN FOR CURRENT TRANSFORMERS

BURDEN WITH 5-A SECONDARIES

DESIGNATION Resistance Inductance Impedance Volt-amperes Power

(mH) (ft) (at 5 A) Factor

Metering Burdens

B-0.1 0.09 0.116 0.1 2.5 0.9

B-0.2 0.18 0.232 0.2 5.0 0.9

B-0.5 0.45 0.580 0.5 12.5 0.9

B-0.9 0.81 1.04 0.9 22.5 0.9

B-1.8 1.62 2.08 1.8 45.0 0.9

Relaying Burdens

B-1 0.5 2.3 1.0 25 0.5

B-2 1.0 4.6 2.0 50 0.5

B-4 2.0 9.2 4.0 100 0.5

B-8 4.0 18.4 8.0 200 0.5

uj = 27rf, where f is the test frequency,

I2 is the rated secondary current (assumed to be 5-A, unless otherwise stated),

and

(PF) is the power factor.

If the burden is stated in terms of the standard ANSI/IEEE C57.13-1978 burden,

the values of resistance and inductance specified in the ANSI/IEEE document [4] are used. Table 4 is a copy of table 11 of ANSI/IEEE 057.13-1978 which gives

the characteristics of the specific standard burdens. The ANSI-specified burdens

are defined only at a frequency of 60 Hz. Occasionally a 50-Hz calibration will be

requested with an ANSI burden. In this instance, it is assumed that the values of

resistance and inductance apply as given in table 4. Sometimes a current transformer submitted for calibration will be accompanied with

a particular burden (for example, a shunt and the connecting leads, or a test set with

leads). The added minimum burden of the calibration facility can be maintained

within 0.03 O for resistance and 20 for inductance. These same minimum values
3.4 Burden Selection 27

also apply when "zero" burden is requested. They represent the added leads and

other impedances of the NIST equipment.

The NIST test set itself is used for accurate determination of the burden impedance,

including the contribution from leads. The procedure involves connecting a 1000-Q

resistor (or another resistance with appropriate value) directly to the secondary

terminals of the test transformer thus shunting some of the current that would

normally flow through the burden impedance and the test set. Such a shunting

operation introduces an apparent change in the ratio correction and phase angle. This

change can be readily measured by the test set and from it the burden impedance

calculated. It should be stressed that the addition of the shunting resistor does not

significantly change the overall burden impedance. The change in the test set readings

results from some of the secondary current being diverted from the test set and not

from a change in the secondary current.

In the actual burden selection procedure, the burden resistance and inductance are

first calculated. Once the proper ratios have been selected on the compensated current

comparator and the test transformer is connected, an approximate value of resistance

and inductance are selected on the burden box. The test current is then slowly

increased to a value of about 10% or 20% of the rated value (0.5 or 1 A). The test set "ratio error" and "phase angle" controls are adjusted to bring the null detector

to zero. At this point, the current of the test set is equal to the correction current of

the test transformer. The test set digital panel meters will indicate the ratio error in

ppm, and the phase angle in /irad. These readings are noted on the laboratory data

sheet.

Next, the lOOO-O precision resistor is placed directly across the secondary terminal

of the test transformer, and a second balance is obtained by adjusting the test set controls. The new test set readings are recorded. The burden may be calculated from

the following relationships:

Rb = {a2 - ai)RA X 10-\ (14)

and

Lb = {bi - b2){RAH X 10-^ (15)

where

Rb is the burden resistance in ohms,

Ra is the value of the resistance connected directly to the secondary terminals of

the test transformer in ohms, usually Ra = 1000 Q,

Lb is the burden inductance in henries,

fli and a2 are the ratio errors in ppm obtained in the first and second tests,

respectively,
28 CURRENT TRANSFORMER CALIBRATION PROCEDURES

bi and 62 are the phase angles in //rad obtained in the first and second tests,

respectively, and

uj = 2xf, where / is the signal frequency in hertz.

To further illustrate the procedure, consider the following example. With an unknown resistive-inductive burden, the test set readings at 60 Hz without Ra are: ai = 123 ppm, and 61 = 315/irad. With Ra = 1000 Q. connected to the secondary

terminals, the readings become a2 = 278 ppm, and 62 = -73//rad. Using eqs. (14)

and (15), we obtain

The proper burden is seldom achieved in the first setting. From the measurements

indicated, the amount of resistance and inductance to be increased or decreased can

readily be calculated. Incremental changes in the resistance and inductance are then

made, and the measurement process is repeated. Usually after a few iterations, the

proper value is obtained. The final value of the measured burden is recorded on the

laboratory data sheet.

3.5 Current Ratio and Phase-Angle Data Collection

After the appropriate burden has been selected, the secondary current is set to a value

of 0.5 A and the test set is adjusted for a null. The values of the ratio error and phase

angle are recorded as indicated by the two digital displays on the test set. The current

is then increased to 1 A and the process repeated. The remaining secondary currents

of 2, 3, 4, and 5 A are set, balances obtained, and the data recorded. Figure 13 is an

example of a typical laboratory data sheet.

After the first test run of secondary currents, the current is reduced again to 0.5 A

and a balance obtained. The remaining currents are set and all data obtained

a second time. The operator compares the values from the first run with those

of the second run. The values should repeat to within about one-fifth of the

measurement uncertainty. If agreement is not apparent, it may be an indication

that the transformer is magnetizing because of a poor connection somewhere in

the circuit, usually in the secondary current loop (including connections within the

transformer). Significant heating of the transformer can sometimes cause the values

to be unrepeatable.

If magnetization is occurring, the values for ratio error and phase angle will not repeat

between the first and the second runs at the low value of current, but may agree at the higher values of current. Even after the transformer has been permitted to cool,

Rb = (278 - 123) X 10"^ x 1000 = 0.155 fl, Lb = (315 + 73) x 10"^ x lOOOQ/377 = 1.029 mH.

(16)

(17)
3.6 SummasizatioD of the Measurement Results 29

the offset in the data will be apparent. To verify this as the cause of disagreement, the

transformer is again demagnetized and the test repeated. If the low-current data from

both runs of the second set do not agree, it is probable that the circuit is magnetizing

the transformer. In this case, the connections or leads must be cleaned, tightened, or replaced before repeating the test. If the lack of repeatability is caused by transformer

heating, the data should once again repeat after a brief period of no-current flow to

allow the transformer to cool.

For transformers having a window-type primary winding, tests are performed to

determine the sensitivity to the position of the conductor (or conductors) passing

through the window. The uncertainty assigned to the calibration results is adjusted

to account for this positional sensitivity. To minimize any loss of accuracy in the

calibration of the transformer and subsequent use, the primary link in the window

is provided with suitable spacers which position and hold the conductor in a fixed

window location. The transformer thus arranged can subsequently be used in the

same configuration as calibrated, minimizing the effect of the position of the primary

turn. The position of the primary return conductor may also affect the ratio error and phase angle, but it is usually less critical.

Once the data have been collected from one range and burden, the test is continued

to the remaining burdens, ranges, and frequencies. The trends and transformer

response are examined to determine if the results are representative for that type

of transformer.

3.6 Summarization of the Measurement Results

After the data have been collected and reviewed for consistency, the results are entered

into a spreadsheet program for analysis. Figure 14 shows a typical spreadsheet used

for the summarization. The averages of the current ratio errors and the averages of

the phase angles are determined, and the ratio correction factor (RCF) is calculated

from the relationship

RCF = 1-1- ratio correction. (18)

The scaling from units of ppm to ratio correction factor, and conversion of the

phase angles from microradians to milliradians is done in the spreadsheet. Values

of estimated measurement uncertainty at 50 and 60 Hz are also calculated using the

following equations:

Ur = ±[0.0075 -h 0.0025(5/72)]% (19)

Up = ±[0.09 + 0.01(5//2)](1 + |/?l/3) mrad (20)
30 CURRENT TRANSFORMER CALIBRATION PROCEDURES

CURRENT TRP

TEST NUnBER:7S2/JL^}'_^5_8_

DPTE:

CUSTOMER:

MfiKER

MODEL:

SERIAL NLWBER:

CURRENT RfiNGE : To -120(5 i5j5/*fr\p€rei

RPTED FREQUENCY: JL^i'-f£9Ji'T'^'t RftTED BURDEN : J(eMj /^"nvgcre «, DEMftG. : tL}f. ^Ou'^S, ftMPS,

HERTZ

NSFORMER C P LIBRfiTION

BURDEN CHECK

(SPECIFIED 1 MEASURED 1 IDE5IG R 1 14

' _-Al£7 R 1 L 1

' Pi ' O.ZO' 1 till 1 1

1 1 1 1 1 1

1 1 1 1 1 1

1 1 1 1 1 till 1

1 1

i SECONDARY INOM 1 NOM 1 CC

FRGI BR 1 AMP ICUR 1 RAT 1 RAT

RATIO ERhOR data

PPM 1 PPM 1 PPM

PHASE ANGLE DATA

uRAD 1 uRAD I uRAD 1

too /I L/ o 1 U t-. -zTLip, 1 - cr 1 a i_£rp/^ 1 j> 1 ~ ' ^ c. ^

1 1/1 II -^5 ' ' -8'=?

1 1 Z ' '

' -B'V 7 ' -37fc ' -376

1 131 II -33^- ;-3az

1 C/. 1 -/03 ' -103 ' -/o j -284'-280 '-28/

1 1 5- . .1 -Z^b • -Z'^h 1 -2 4-8

'100 '-'52, ' -"^ 3

1 1 ' '

' -ilO 1-/0^ '-//o -20b ' -207 ' -207

-83 ' -8Z ' -8 1

1 1 tr ' ' ' -yo3 '-yo4 ' -/0'4-

1^0^ A ' 6.5" ' 100 "ao' -84 |-8+ ;-84- -q.i|4 '-if3^ '-i/-'4o

1 1 tr ' '

' -iC? ^ 'lolo ' -yo3 -/^3 1 -/^ i

1 -201

Figure 13. Current Transformer Calibration Typical Data Sheet
3.6 Summarization of the Measurement Results 31

CURRENT TRONSFQRMER CQLIBROTION

ISECONDftRY INOi^ 1 NOn 1 CC

r HU 1 fK 1 Hrnh \ UUK 1 KH 1 1 KW 1 RATIO ERROR DOTft

i~i r"i iw» 1 P'l ivi 1 ^1 fi M« PHfibE 1 ONGLE DfiTrt 1 uRPD 1 uRWU 1 uKftD 1 —; —1 —1 —1

1 1

;

-8 8

' -8b ' - ^0 ' -29fc —1 __l— 1 -1 - 1- - 1 1 --|_ _ __! , -111 1-/7/ ' - i7/

-/^2. ' -/92 ' -/^ 2 —1_ _-| _ __|____|_ -|„_ 1 1 _ 1 |._ _ '__ 1 —1 1_—1 —1 — 1 1 \ _-_ 1

1 _| 1 —1_-_|___ 1 ;_ __ _ 1.

,'___l —1

1 1 1 —1 —I —i___i__- 1 ; 1 1

1 —1 —1 —; —1

1 1 1_ —1 —1 —1

1 1 \ 1

; —1 —1 —1 —1

1 1 \ 1 —; —1

1 1 1 1

Figure 13. Current Transformer Calibration Typical Data Sheet (Continued)
32 CURRENT TRANSFORMER CALIBRATION PROCEDURES

1 F 1 B

1 R 1 U

1 Q 1 R

1 R 1

1 A 1

1 T 1

1 SEC 1 CUR 1RATIO 1

1 « 1

1

1 AVER 1

1

R C F

1

1

1

II PHASE 1

AVER

1

1

1 PHASE

1

1 1

1 1 DATA 1 UNCER 1

i DATA 1 DATA

IIII DATA 1

1

DATA 1

1

1 DATA 1

1

1 &a 1 fl 1 2 1

1 d.5 1

1 -93 1

1 -91 1 -91 —1 3 1 -91.7 1 8.99991 18.83258

1 1-

II -548 1 -519 1 -528 1 -529.8 1 8.5290

1 1

1 8.2235 1

1 1 1 1 1 1 -93 1 -89 1 -89 3 1 -98.3 1 8.99991 18.82888 II -481 -454 1 -454 1 -463.8 1 8.4638 1 8.1616 1

1 1 1 1 2 1 -93 1 -93 1 -93 3 1 -93.8 1 8.99991 le. 81375 II -397 -376 1 -376 1 -363.8 1 8.383e 1 8.1297 1

1 1 1 1 3 1 -97 -94 1 -95 1 3 1 -95.3 1 8.99998 18.81187 II -334 -321 1 -322 -325.7 1 8.3257 1 e.ll82 1

1 1 1 1 4 1 -183 -183 1 -181 3 1 -182.3 1 8.99998 18.81882 II -264 -268 1 -281 -281.7 1 8.2617 1 8.1121 1

1 1 1 1 5 1 -189 -188 1 -189 1 3 1 -188.7 1 8.99989 18.81888 II -246 -246 1 -248 -246-7 1 8.2467 1 8. 1882 1

1 1

1 ee 1 A

1 1

1 4 1

1 e.5 1 -IW1 -92 1 -931 1

1 3 1

1 -95.8 1 8.99991

118.83258

11

II -418

1 -443 1 -446 -436.31

1 8.4363

1 1

1 8.2176 i

1 1 1 1 S 1 -lie -189 1 -lie i 3 1 -189,7 1 8.99989 18.81888 II -286 -287 1 -287 -286.7 1 8.2867 1 8.1869 1

1 1 1 1 1 1 1 1 ERR 1 ERR 11 1 ERR 1 1 ERR 1

1 £0 1 A 118 1 e.s 1 -83 -82 1 -81 i 3 -82.8 1 8.99992 18.83258 II -436 -441 1 -448 -439.8 1 0.4398 i 8.2178 1

1 1 1 ! 5 1 -183 -184 1 -184 1 3 -183.7 1 8.99998 le.eieee II -195 -195 1

1

-194 -194.7 1 8.1947 1 8.1865 1

1 1

1 &e 1 A

1 1 m 1

1 e.s 1 -W

1 -84 1 -84

1

1 3

1 -64.8 1 8.99992

118.83258II

II -444 -439 1 -448 -441.8

1

1 8.4418

1 1

1 8.2179 1

1 1 1 1 5 1 -187 -188 1 -183 1 3 -185.3 1 8.99989 ie.ei888 II -193 -191 1 -281 -195.8 1 8.1958 1 8.1865 1

1 1 1 1 1 1 1 1 ERR 1 ERR 1

1 1 1 1

1 U 1 A 12401 e.5 1 -98 1 -98 1 -92 1 3 -93.3 I 8.99591 18.83258 II -541 -568 1 -559 -553.3 1 8.5533 1 8.2258 1

1 1 1 1 5 1 -86 1 -88 1 -98 1 3 -86.8 8.99991 18.81888 II -254 -256 1 -243 -251.8 1 8.2518 1 8.1884 1

1 1 1 1 1 1 1 1 ERR 1 1 II 1 1 1 1

1 1 A 1 2 1 e.5 1 -171 1 -171 1 -171 1 3 -171.8 1 8.9996 18.83258 II -119 1 -118 1 -119 -118.7 1 8.1187 1 8.1975 1

1 1

1 1

1 1

1 1

5 1

1

-192 1 -192 1

1 1

-192 1 3

1

-192.8 1 8.9998 le. 81888

1 HII

-65 -85 1

1 1

-65 -85.8

1

1 8.8858

1

1 8.1828 1

1 1

Figure 14. Current Transformer Calibration Spreadsheet for Calculation of Final Results

where

Ur is the estimated uncertainty in the value for the ratio correction factor

Up is the estimated uncertainty in the value for the phase angle,

I2 is the nominal value of secondary current, and

1 is the absolute value of the phase angle in mrad.

Care must be exercised in interpreting the uncertainty values by noting that Ur is a

fraction of the ratio correction factor in percent while Up is a direct phase-angle value.

For example, the measured RCFof a transformer at 5 A is 1.000123 and the phase

angle is 0.315 mrad. From eqs (16) and (17), it is determined that Ur = ±0.01 %

and Up = ±0.101 mrad. Using these uncertainty values the probable upper and lower

bounds for RCF are

1.000123(1 ±0.0001) = 1.000223 and 1.000023, (21)

and for phase angle

0.315 ± 0.101 = 0.416 and 0.214mrad. (22)

The eqs (19) and (20) have been selected so that the limits of the uncertainty

represent a realistic characterization of the transformer being calibrated over a range

of secondary currents from 10% to 100% of the rated secondary current. The

uncertainties will not usually be valid for currents above the rated current or for
3.6 Summarization of the Measurement Results 33

Secondary Current, Amperes Secondary Current, Amperes

Figure 15. Ratio and Phase-Angle Uncertainty as a Function of Secondary Current

currents of less than 10% of rated current. For the ratio correction factor, the

uncertainty at a secondary current of 5 A is ±0.01%, while at a secondary current

of 1 A, the uncertainty doubles to ±0.02%. The uncertainty for the phase angle is not only a function of the secondary current, I2, but also of the absolute value of

the phase angle itself. For very small phase angles, the uncertainty is slightly greater

than O.lmrad at 5 A and almost doubles to a value of 0.19 mrad at 0.5 A. Figure 15

shows plots of these functions. For the phase angle plot, the phase angle of zero

is shown to represent the smallest possible uncertainty. The other values shown,

100 and 1000 //rad, are typical phase angles for which plots are also shown. It has

been determined that these uncertainty formulas are adequate for neaxly all of the

calibrations performed. In the instances where it is apparent that the calculated

uncertainties do not properly describe the characteristics of the current transformer,

the formulas are suitably altered to describe the transformer's performance. This is discussed further in chapter 4. The formulas and values of uncertainty are stated on

the Report of Cahbration (or Report of Test).
34 CURRENT TRANSFORMER CALIBRATION PROCEDURES

1 1

1 1

1 corn t MT

1 1

1 ScC 1

1 n lO 1

1 LUn 1

___________

BURDEN

nlOni/ 1 L\Un/

1 1 1

1 ao 1 c

1 a c

1 1 1 i.e 1 e.2 1 m

11 1 C* U

1 1 3.e e.2 1 140

1

1

1

1

1

1

1 s.a

1

1

1 1

1 (a 4

I

1 s.e

1

1

1

1 (A It

1

1 s.e

1

1

1

1 (A se

1

1 S.8

1

1

1

1 60

1

1 s.e

1

1

1

1 1

1 m 2 1 s.e

1

1 1

1 1

aTQBER 1970

-0.009 I ^m i -e.00a I -0.008 I -0.006 I -0.008 I

I -4.004 I

I -4.003 I

I -0.006 I

I 0.001 I

I -0.02 I

II

II -IIII -IIII

II

II

II

II

II

II

II

II

II

II

II

WVCH 1975

-0.008 I -0.008 I -0.008 I -0.009 I -«.009 I -«.010 I

I -4.010 I

I -4.009')

I -4.010 I

I

II -0.«

-0.02 I

I

I i COR'N I CUR

JULY 1980 II FEBRUARY 1986 I

11 .1

II 1 RAT II t COR'N I CUR RAT I

II 11

II -«.WSa I 0.99991 1

II -0.0030 I 0.99991 1

II -0.0090 I 0.99991 I

II -4.0100 I 0.99930 I

II -0.8100 I 0.99990 I

II -0.0110 I 0.99989 1

II I I

11 -4.0110 I 0.99989 I

II I I

II -4.0100 I 0.99990 I

II I I

II -4.0118 I 0.99989 I

II I I

II -4.0090 I 0.99991 I

II I I

II -0.0200 I 0.9998 I

I II I I

I -0.010 I

I -0.009 I

I -4.009 I

I -0.010 i

I -0.018 I

I -0.010 I

I I

1 -4.011 I

I I

I -0.009 I

I I

I -0.010 I

I I

I -0.008 I

I I

I -4.020 I

I

Figure 16. Historical Data of Transformer Calibration

3.7 Historical Data and Expected Performance

For high-quality current transformers, the values of the ratio correction factor and

the phase angle change very little, if at aU, over many years. This then makes the

calibration history for a particular transformer very valuable. When a transformer

has been previously calibrated at NIST, the previous data are compared to the present

calibration results. Figure 16 shows a typical "history" sheet. Very close agreement,

within two standard deviations, is expected if the transformer is tested under the

same conditions (i.e., same burdens, frequency, ratios, currents, etc.). If the results

do not agree with historical data, there is usually a problem in the transformer or the

test arrangement. When no previous calibration history is available, the calibration

results are compared to the expected performance for the type of transformer

being calibrated. Occasionally, a transformer of unusual design and performance

is encountered. This occurs when a transformer is heavily compensated, for example,

and the results may appear different from the typical response. In such instances, the

transformer manufacturer is usually consulted to verify that the observed response is representative for that particular design.

3.8 The Report of Calibration

For each current transformer that is tested, a Report of Calibration (or Report of

Test) is issued. The Report gives the results of the measurements made by NIST on

that particular current transformer under the test conditions specified at the time of

test. An example of a Report of Cahbration for a current transformer is shown in
3.8 The Report of Calibration 35

figure 17. The Report of Calibration specifies the instrument that was calibrated,

and includes the manufacturer, model or type, and serial number. If the transformer

lacks a suitable unique serial number, NIST will mark and uniquely identify the

transformer. The current ratios and frequency range are stated. The company which

submitted the transformer for testing is clearly given by name and address.

Specific details of the test conditions are given, such as the current ratios, burden(s),

frequency, and current, and other information such as laboratory temperature, special

connections, or other pertinent conditions. Calibration results are given a^ ratio

correction factors (RCF) and phase angle (in milliradians).

The estimated values of uncertainty are also given for both the ratio correction factors

and for the phase angle. As discussed before and in chapter 4, allowances are made

for both the systematic and random components of the measurement uncertainty.

Values used for an individual calibration are available upon request.

The NIST test number and customer's order number are given. Questions regarding

the test or inquiries regarding the Report of Calibration (or Report of Test) should

be accompanied with the NIST test number, the instrument manufacturer, model,

and serial number.

Reports of Calibration are issued for calibrations that are considered routine in that

the specific device or others of its type have been calibrated previously at NIST. They

are characterized by their secondary currents being in the range from 0.5 to 5 A, and

the items under calibration are sufficiently stable to warrant uncertainty statements

of about ±100 ppm and ±100/zrad.

The Reports of Test, on the other hand, are issued for special tests where

the deviations from routine are significant. Special Tests may be performed to

accommodate unusual operating characteristics and unusual accuracy requirements,

e.g., to perform measurements beyond the determination of ratio correction and phcise

angle. Special Tests usually require additional experimental setups and additional

investigation and are performed on an "at cost" basis.
36 CURRENT TRANSFORMER CALIBRATION PROCEDURES

U.S. DEPARTMENT OF COMMERCE

NATIONAL INSTITUTE OF STANDARDS AND TECHNOLOGY

(formerly NATIONAL BUREAU OF STANDARDS)

GAITHERSBURG, MD 20899 REPORT OF CALIBRATION

CURRENT TRANSFORMER

10-1200 to 5 Amperes, 25-500 Hertz, 25 Volt-Amperes

Knun-Betare Model 3121/C, Serial Number 1234

Submitted by

Peyew Public Gas & Electric Co.

100 Main Street

Sommeware , Ohio 45891

Frequency Secondary

(hertz) Burden

Secondary

Current

(amperesCurrent Ratio

Phase Angle

(milliradian)

60 0.5

1.0

2.0

3.0

4.0

5.0

2 X 0.99991

0.99991

0.99991

0.99990

0.99990

0.99989

+0.53

+0.46

+0.38

+0.33

+0.28

+0.25

60 5.0 4 X 0.99989 +0.21

60 5.0 10 X 0.99990 +0.19

60 5.0 20 X 0.99989 +0.20

60 5.0 240 X 0.99991 +0.25

400 5.0 20 X 0.9998 +0.1

Date of test: January 30, 1990

Temperature: 23 "C

728/123456-90

Figure 17. Typical Report of Calibration
3.8 The Report of Calibration 37

CURRENT TRANSFORMER

Knun-Betare Model 3121/C, Serial No. 1234

Peyew Public Gas & Electric Co.

- 2

- Secondary burden A consisted of a resistance of 0.20 ohm in series with an

inductance of 140 microhenries.

At a frequency of 60 hertz, the uncertainty for the values of ratio does not

exceed ±[0.0075 + 0.0025(5/13)] percent, and for the values of phase angle does

not exceed ±[0.09 + 0.01(5/Is) ] • (1 + \fi\/2,) milliradians , where Ig is the

secondary current in amperes and p is the reported value of phase angle in

milliradians. At a frequency of 400 hertz, the uncertainty for the value of ratio

does not exceed ±2* [0.0075 + 0.0025(5/Ig) ] percent, and for the value of phase

angle does not exceed ±3.[0.09 + 0.01(5/Is) ] • (1 + |/3|/3) milliradians. These

figures include allowances (available on request) for both the random and the

systematic errors of the calibration process.

For the Director,

National Institute of Standards and Technology

Kalie Braetour, Group Leader

Applied Electrical Measurements

Electricity Division

Test No. 728/123456-90

Purchase Order No. 8-00331/A

Date: January 30, 1990

Figure 17. Typical Report of Calibration (Continued)
38 MEASUREMENT UNCERTAINTY ANALYSIS

4 MEASUREMENT UNCERTAINTY

ANALYSIS

4.1 General Considerations

Several considerations are important in preparing the uncertainty estimates for the

caHbration of current transformers. The total measurement uncertainty arises from

both (1) the NIST standards and test equipment and (2) the device under test. The

former source of error is well characterized; the latter, usually being the dominant one,

may not be. In addition, many transformers submitted to NIST for calibration are

built for special purposes and have unique characteristics. Even for the widely used

transformers, the accuracy of each unit is strongly affected by the conditions under

which it is used and calibrated, that is: the nominal ratio, the value of the burden,

the magnitude of the current, and the electromagnetic environment. To achieve

the maximum accuracy, many transformers and even test points would have to be

evaluated separately rendering the process prohibitively expensive and uneconomical

for the customers. An approach has been taken to develop an error budget that

includes most of the transformers. For exceptionally stable transformers, higher

accuracies are possible under Special Test provisions and additional cost. Accuracies

might be lowered for transformers whose stabilities are inferior.

4.2 Accuracy Evaluation of NIST Standards and

Equipment

4-2.1 Current Comparators

The customers' transformers are calibrated at NIST on a relative basis against

current comparators that are designed and built to have negligible corrections

when used for routine calibrations. To achieve this, the current comparators

themselves must be calibrated on a more fundamental or "absolute" basis. Complete

"absolute" calibration is an involved and lengthy undertaking for a multi-range

current comparator. Often the limits of errors of a current comparator can be

determined by indirect means. Both approaches have been used at NIST. For

example, as the current comparators were constructed, indirect evaluations and tests were used. Later, all finished devices were calibrated on a fundamental basis at several nominal ratios to ensure that the indirect approaches are valid and that

the corrections are negligible, as planned. Table 5 shows estimates of systematic

uncertainties for three of the NIST compensated current comparators and the test

set.
4.2 Accuracy Evaluation of NIST Standards and Equipment 39

Table 5. Systematic Uncertainties in NIST Equipment

a = ratio error, in ppm, and /3 = phase angle in /irad, both of which are test-set readings.

SOURCE OF UNCERTAINTIES Ratio

(ppm)

Phase Angle

(/-trad)

Tjjow-jxange nw 'Ra-ncro PrT"* v^v^\^ -1-8 ito

Mid-Range CCC

Primary feed ±3 ±3

Secondary feed ±8 ±8

High-Range CCC ±5 ±5

Test Set ±(l-f0.005|a|) ±(1+0.005|/3|)

CCC-Compensated Current Comparator

The fundamental measurement of transformer ratios at NIST relies on a buildup ("bootstrap") technique whereby two transformers or current comparators with

known ratios are used to calibrate a third one [5] [7]. The starting point is a

1:1 current comparator that can be self- calibrated by connecting the primary and

secondary windings in series. Two calibrated 1:1 current comparators are then used

to calibrate a 2:1 current comparator.

The secondary windings of all three are connected in series. The primary currents

of the two 1:1 current comparators are added and applied to the primary winding of

the 2:1 current comparator. The new ratio is the sum of the two previously known

ratios. The process can be continued for higher ratios. Three identical multi-range

current comparators are required. In the literature the approach has been called the

m + n technique since the new ratio is the sum of the two previous ones, m -f n [5].

Another bootstrap technique, in which the new ratio is the product of two previously

known ratios (m x n technique), is also utilized at NIST [8]. Two multi-range current

comparators or current transformers are used, plus a small current comparator for

comparing two unequal secondary currents in the ratio of 2:1 and 2.5:1. The primary

windings of the full-range current comparators are connected in series while the

secondary windings are connected to the small current comparator. For example,

a 240:1 ratio can be calibrated with known 120:1 and 2:1 ratios.

Two amplifier- aided current transformers were calibrated by the m x n method.

These, in turn, were used to check at least a few ratios on each of the three current

comparators used for routine calibrations.

The build-up calibration process requires a large amount of time and equipment. Also

the uncertainties tend to accumulate with every step and may lead to unrealistically

high values at large ratios. Nevertheless, it is a rigorous approach and therefore
40

a highly desirable means for checking other approaches, including the indirect

techniques discussed in the following paragraphs. Fortunately, the problem is mitigated by the fact that current comparators and similar devices have permanently

stable ratios. Usually a minimum number of transformers are then calibrated on a fundamental basis and others calibrated on a relative basis.

There are two types of intrinsic errors in current comparators: leakage flux causing

magnetic errors, and internal capacitances causing capacitive errors. Both types

of errors are well understood and can be controlled by the design of the current

comparator. The limits of both can be estimated from indirect measurements.

With respect to the magnetic error, the worst-case leakage flux can be simulated

and its effects measured. For example, two concentrated ratio windings with the

same number of turns can be placed at the diametrically opposite parts of the

core. Alternatively, one of the windings can be distributed along the core, the other

concentrated. Both configurations will result in large leakage flux. The two windings

are connected in series opposition and the device is calibrated on a fundamental basis

as a 1:1 current comparator. The number of turns in the windings can be small to

minimize the capacitive error. The test is performed with the worst-case and realistic

geometries of the ratio windings. Under all conditions, the magnetic error should be

insignificant for a well-designed current comparator.

Current comparators wound on toroidal cores are particularly sensitive to the leakage

fields with certain geometries, for example, when the magnetic field is radial with

respect to the toroid. External current loops can be used to simulate such fields.

High-permeability toroidal cores inherently minimize the relative magnitude of the

leakage flux. Further reduction is achieved by placing a magnetic shield between the

ratio and detection windings. The magnetic shield tends to short circuit the leakage

flux, thus preventing its entry into the toroidal core and the detection winding. The

aim is to provide sufficient magnetic shielding so that the error is negligible even in

adverse winding configurations.

Different types of magnetic shields, their effectiveness and the errors resulting from the

leakage flux have been thoroughly investigated in a current comparator designed for an impedance bridge [9]. The worst-case error for an unshielded current comparator

can be several tens of ppm. Yet a simple judiciously designed shield made of highpermeability material will reduce the error to a sub-ppm level. A multi-layer, highperformance shield reduces the error below a part in 10*.

The capacitive errors are caused by the turn-to-turn capacitances in the same ratio

winding, by the winding- to-ground capacitances, and by the winding-to-winding

capacitances. The latter are eliminated by electro-static shields, but the former two

may be dealt with as follows.

The turn-to-turn and winding-to-ground capacitances tend to divert some of the
4.3 Errors Originating in Transformers Being Tested 41

current from passing through all of the turns in a winding. They effectively produce

a capacitive shunt across the ratio windings. In the primary winding, the number

of turns in many cases is below 100 with negligibly small internal capacitances. The

secondary windings in mid- and high-range current comparators have significant

internal capacitances that could cause possible errors if there were voltage across these windings. However, when operated with the primary feed, there is a very small

voltage across the secondary windings and, hence, negligible capacitive error. This is not so when the secondary feed is used and up to 200 volts can be introduced in the

secondary winding producing capacitive error of the order of 10 ppm. The capacitive

error is partly compensated by the external capacitor, Cc, in figure 8. The low-range current comparators with a nominal ratio of less than unity present a special problem. The voltage in the primary winding is high regardless of whether

primary or secondary feed is used when the device is operated as a compensated

current comparator. The auxiliary current transformer in figure 10 removes the need

of the compensated current comparator to serve as a power transformer and thus

reduces the voltage across the windings. As a result, the capacitive error can be

greatly reduced.

4.2.2 Test Set

The test set is a minor contributor to uncertainties. Because of the added circuitry

it could act as a receiver of electromagnetic interference. The circuit that generates

the correction current has accuracv limitations. The test set has a fixed uncertainty

component of ±1 ppm and ±1 /zrad, and another uncertainty component that depends

on the magnitude of the injected current. The latter is ±0.5% of the ratio correction

and ±0.5% of the measured phase angle. Expressed algebraically the uncertainties

become ±(1 + 0.005 |a|) ppm for the ratio correction, and ±(1 + 0.005|/?|) //rad for

the phase angle, where a and /3 are the test set readings.

4.3 Errors Originating in Transformers Being Tested

The ratio in good instrument transformers does not have drift over time. However,

short- and long-term fluctuations are observed in the measured ratio correction and

phase angle. These can be treated as random errors if adequate time and resources

are available to obtain sufficient data. This is possible with some transformer types

that are submitted to NIST often, but not with others that are one-of-a-kind or are calibrated infrequently.

The observed instabilities in transformers have well-known physical bases, such

as magnetization, heating effects, burden sensitivity, and winding geometry. The

limits of variations due to the above causes can be estimated from the properties
42 MEASUREMENT UNCERTAINTY ANALYSIS

Table 6. Estimated Possible Instabilities in a Typical Transformer Under Test

Ratio Phase

SOURCE Correction Ancle

(ppm) (^rad)

1. Core magnetization ±5 to ±15 ±16 to ±48

2. Uncertainty in burden ±6 ±19

3. Transformer temperature ±6 ±19

4. Current value ±3 ±9

5. Primary winding position ±20 ±6

6. Electromagnetic interference ±10 ±10

of the device or can be determined from auxiliary tests. From a knowledge of the

transformers and the results of past tests, a list of errors, table 6, has been prepared.

The estimated values in table 6 represent the limits of variations originating from

the indicated sources. The equivalent circuit of figure 2b and eq. (2) are useful for

estimating the source of error. It should be emphasized that the values in table 6 are approximations for typical transformers submitted to NIST for calibration. More

stable transformers, e.g., two-stage types, are available in which instabilities are lower by one or two orders of magnitude. Less stable transformers have also been

encountered by NIST. In both cases the stability and quality of the device usually

becomes apparent soon after the tests have been started.

For estimating the values in table 6, Zm has been assumed to be 1000 0, the series

inductive reactance being three times larger than the equivalent loss resistance. Thus,

Xm = 949 and = 316 fi. The secondary leakage impedance is assumed to be

resistive with a value of R2 = 0.5 0. Similarly, the burden impedance is assumed to

be resistive with Rb = 0.5 0.

1. The most serious source of error in current transformers is residual magnetization

of the core. Even if steps are taken to minimize it, some residual magnetization

will be generated by switching transients and small direct currents in the circuit

caused by rectification in oxidized connections or by thermal electromotive

forces. Magnetization reduces the effective permeability of the core and thus

the magnetizing impedance. A range of instabilities is given for this effect.

The estimated minimum effect is ±3% variation in the magnetizing impedance

which is achievable under laboratory conditions following demagnetization of

the core. The estimated majcimum variation in is ±10%, a value that could

be expected after a period of use without demagnetization.

2, 3. The burden uncertainty arises from the accuracy to which the burden can be

set and then maintained due to temperature changes. A 2% allowance is made

for the initial setting of the burden and 2% for the heating effects. The changes
4.3 Errors Originating in Transformers Being Tested 43

in the burden directly affect the indicated ratio corrections and phase angles by

changing the ratio Rg/Zm (see figure 2). In the absence of the external burden,

temperature changes affect the ratio Z2/Zm. A 4% uncertainty is assumed due

to ambient temperature and self-heating effects in R2/Zm4. The nonlinearity of the core causes changes in Zm with current, especially at low

permeabilities and currents. It is assumed that Zm changes by 10% between the

secondary currents of 0.25 and 0.75 A. If in the calibration the 0.5-A value can be maintained within ±0.1 A, the uncertainty from this source will be ±3 ppm

and ±9 //rad. Note that the previously discussed sources of error produce larger

effects in the pha:Se angle and are related to the variations in (Z2 + ZB)/Zm.

5. In the window-type transformers, an uncertainty is caused by the position of the

conductor in the window. The geometry of the primary winding affects both

the primary and the secondary leakage inductance, thus causing an uncertainty

in the ratio correction and, to a lesser extent, in the phase angle. The estimated

values in table 6 are based on previous observations.

6. Finally, the electromagnetic interference from the supply transformer, the power

lines in the building, and ground currents can also cause measurement errors.

The estimated instabilities tabulated in table 6 will lead to a lack of repeatability

or long-term random error as the transformer is recalibrated. It is instructive to

compare such predicted lack of repeatability with actual performance of a transformer

with similar electrical characteristics. Such a comparison was made using a current

transformer that had been calibrated 15 times over a 55-month period. The

calibration data are shown in table 7, with the standard deviation of 14 ppm for

the ratio correction and of 25 //rad for the phase angle.

The estimated errors of table 6 are combined as if they were true, independent random

errors. A uniform distribution is assumed for each of them. For a uniform distribution

where s is the standard deviation, and d is the maximum value of the estimated error. The combined standard deviation is the square root of the sum of the squaxes of the

individual standard deviations. It is calculated for the values of table 6 using the

mean value of the magnetization effect (±10.5 ppm, ±32/zrad). Such a calculation

yields 15.2 ppm for the ratio correction and 25.6 //rad for phase angle.

The transformer whose calibration data are shown in table 7 has a fixed primary

winding and it was calibrated without an external burden and at relatively high

secondary current, thus, the items 2, 4, and 5 of table 6 do not apply. Calculating
44 MEASUREMENT UNCERTAINTY ANALYSIS

Table 7. Calibration History of a Typical Laboratory Grade Current Transformer, Marked

Ratio 10A:5A.

Mean values and standard deviations are the cumulative values to and including the date

of calibration.

RATIO CORRECTION (ppm) PHASE ANGLE (^rad)

Date Measured Mean Std Measured Mean Std.

Value Dev. Value Dev.

16 Nov 81 1 OK too

20 Jan 82 1 KR 91 Q d.7n t < u 1IZ.U 9 n 22 Feb 82 130 137 0 16.6 434 452 3 18 0

23 Feb 82 131 135.5 13.9 439 449.0 16.1

11 Mar 82 145 137.4 12.8 403 439.8 24.9

17 Jan 83 161 141.3 14.9 437 439.3 22.3

27 Apr 83 148 142.3 13.9 495 447.2 29.3

30 Sep 83 128 140.5 13.8 451 447.7 27.1

27 Oct 83 120 138.2 14.6 444 447.3 25.4

2 Feb 84 141 138.5 13.8 466 449.2 24.7

3 Jul 84 120 136.8 14.2 442 448.5 23.5

27 Sep 84 163 139.0 15.5 484 451.5 24.6

17 Sep 85 142 139.2 14.9 487 454.2 25.6

7 Jan 86 141 139.4 14.3 488 456.6 26.2

17 Jun 86 144 139.7 13.8 459 456.8 25.2

the standard deviation for the remaining items yields 9.1 ppm for the ratio correction

and 22.2 //rad for the phase angle. The calculated standard deviations from the

estimated error components are in reasonable agreement with the actually observed

long-term standard deviation of the transformer indicating that the instabilities of a

transformer can be predicted from its characteristics.

4.4 Overall Measurement Uncertainties

In estimating the overall uncertainty, the random and systematic uncertainty

components are combined using the guidelines of the BIPM Committee [10]. The

two components are combined on a root- sum-squared basis at the three sigma level.

A uniform distribution is assumed for systematic uncertainties:

l/ = (952-h3<f2)?, (24)

where U is the overall uncertainty, s is the long-term standard deviation, and d is the

systematic uncertainty of NIST equipment.

In the absence of sufficient data to determine long-term standard deviation, it
4.5 Special High-Accuracy Transformers 45

is estimated from the type of information in table 6 computed for the specific

transformer.

Consider an example. The mid-range compensated current comparator is used with

the test set having the total systematic uncertainty of 6 ppm and 10 ^rad. Using the

cumulative standard deviations from table 7 (June 1986 values), we obtain

C/r = 43.3 ppm, (25)

t/p = 77.0/irad (26)

as the uncertainties in ratio correction and phase angle.

Consider another example where the long-term standard deviation is not available,

but instead the combined uncertainties from table 6 are used (15.2 ppm and

25.6 //rad). The overall uncertainties become

Ur = [(3 X 15.2)2 + (V^ X 6)2] 2 = 46.7 ppm, and (27)

C/p[(3x 25.6)2 -K(V3xlO)2]i = 78.7 //rad. (28)

For this type of transformer the uncertainties quoted in NIST reports are ±100 ppm

and ±100 /zrad. In both cases the quoted uncertainty for ratio correction is somewhat

larger then the observed value, while the phase angle uncertainty is about the same

magnitude as the observed value. There is a certain advantage to keeping accuracy

statements simple and as inclusive as possible. Hence, the rounded off figure of

100 ppm (or fiT&d) is used for both error components.

4.5 Special High-Accuracy Transformers

The discussion in this chapter so far has dealt with the transformers that constitute

the majority of the NIST calibration workload. These are relatively old designs having

corrections within 0.1% and 1 mrad and stabilities about an order of magnitude better.

Advances in technology enable the construction of simple, two-winding transformers

that are about an order of magnitude more accurate. Two-stage transformers can be constructed that have corrections and instabilities within a few ppm (or /zrad).

The sources of error as listed in table 6 are greatly reduced or even eliminated. The

accuracy of these specialized transformers is comparable to that of NIST standards.

Calibration of high-accuracy ratio devices on a Special Test basis is provided. The

charges are based on the actual cost rather than a fixed fee. Special Tests are subject

to constraints due to scheduling and availability of equipment. The accuracies for

such tests are evaluated on an individual basis. Uncertainties in the range of ±5 to

10 ppm (or //rad) are possible.
46 ACKNOWLEDGMENTS

4.6 International Comparisons

When a laboratory, such as NIST, attempts to verify the accuracy of its current

ratio standards, a vast amount of time and effort is required. One has to be

always cautious in guarding against undetected errors. Inasmuch as possible,

redundant techniques are employed to reduce the possibility of undetected errors. Interlaboratory comparisons among the national standards laboratories of other

countries provide excellent means for checking the highest accuracy measurements.

Two international comparisons have been conducted over the past 25 years in which

NIST directly participated.

In 1965 a comparison was conducted between NIST and the National Research

Council (NRC), Canada. Two multirange, audio-frequency current transformers were

used [11]. The nominal ratios ranged from 1:1 to 6:1. The tests were conducted

between 400 Hz and 10 kHz. At the frequencies of the primary interest, 400 and

1000 Hz, the agreement was within 0.4 ppm and 5.4/irad at 400 Hz, and 0.3 ppm

and 6.1 //rad at 1000 Hz. The greater than expected phase angle difference at 400 Hz

was attributed to the magnetization of the core.

During 1984-1985, NIST, NRC, and three European laboratories participated in

calibrating a multi-range, power frequency transformer at 50 and 60 Hz [13]. A range

of current ratios from 5:1 to 200:1 was checked. The secondary currents ranged from

200% of rated to 1% of rated. At the rated current the agreement of all participants

was within 1 ppm and 1.9/irad. At very low currents, 1% of rated, the spread

was 3 ppm and 5 /irad. The spread at low currents was partly attributed to the

electromagnetic pickup in the transformer under test. The above direct interlaboratory comparisons and others that have been indirect

(e.g., NRC with European laboratories) confirm that the state of the art is well

within the 100 ppm/100 ^rad uncertainty which is used by NIST in routine work. At

the 5 ppm/5//rad level there are differences. Such uncertainties are given by NIST

only in special cases in which there is ample evidence through independent checks that

the data are valid. Also the transformer under test must exhibit adequate stability.

5 ACKNOWLEDGMENTS

The authors would like to express their appreciation to the following staff members:

Mr. T. M. Souders who, on many occasions, guided and consulted in matters

regarding current transformers, Mrs. E. A. Matthews and Mr. Robert Palm who

prepared portions of the graphical data, and Mrs. R. Cummings, Mrs. D. Williams,

Mrs. B. Meiselman, Mrs. R. Von Feldt, and Mr. J. Greenberg who helped prepare

the manuscript.
47

6 APPENDIX

INFORMATION REGARDING

THE NIST CALIBRATION SERVICE FOR

INSTRUMENT CURRENT TRANSFORMERS

Al Technical Information

Only metering-type or laboratory-standard current transformers (including current

comparators) will be accepted for test. These tests determine the current ratio

correction factor and phase angle of the transformer "as received." NIST does not

undertake the cleaning of transformers and does not knowingly begin tests on faulty

transformers.

Before a test can be started, the test conditions must be completely specified by the

user. The current ratio (primary to secondary) available on a routine basis ranges

from 0.25:5 to 12000:5. Tables 1 to 3 in the body of this document give specific

ratios that are routinely available. Ratios other than those listed may be available

by special arrangements at an additional fee. Normal secondary current levels for

testing are 0.5, 1, 2, 3, 4, and 5 A. Over-current tests, up to but not exceeding 10 A,

are generally available within the limitations of NIST equipment. Burdens should be

specified either as:

1. Series resistance in ohms and inductance in milliheneries,

2. ANSI/IEEE C57.13-1978 standard burdens,

3. VA load at specified power factor, or

4. The user supplied burden and connecting leads.

Normal test frequencies are 50, 60, and 400 Hz. Other frequencies may be available

and inquiries should be made regarding special tests.

Uncertainties of the measurement are stated on the Report of Calibration (or Report

of Test) which is issued for each instrument calibrated.

A2 Calibration Request, Transformer Shipping, Insurance and Risk of Loss

A formal purchase order for the calibration or test should be sent before or at the time

the transformer is shipped. This should provide clear identification of the apparatus

being submitted and give separate instructions for the return shipment, mailing of the

report, and billing. To minimize the time during which equipment is out of service,

arrangements can be made to delay shipment to NIST until shortly before the test is scheduled to begin.
48 APPENDIX

Requests for calibrations or tests from Federal and state agencies, must be

accompanied either by a purchase order, or by a letter or other document authorizing

the costs to be billed to the agency. The NIST agreement to perform the calibration

does not imply acceptance of any provisions set forth in the order contrary to the

policy, practice, or regulations of the NIST or the U. S. Government. The purchase

order should clearly state the desired test conditions.

NIST staif will provide assistance for individual measurement problems where they

can. The NIST headquarters is located in Gaithersburg, Maryland, approximately 25

miles northwest of Washington, D. C. The calibration of instrument transformers is performed at the Gaithersburg site by the Electricity Division (811) of the Electronics

and Electrical Engineering Laboratory of NIST. Inquiries may be made by direct

correspondence to:

National Institute of Standards and Technology

Electricity Division (811)

Metrology Building, Room B344

Gaithersburg, MD 20899

The reader is encouraged to obtain a copy of the NIST Special Publication 250,

Calibration and Related Measurements Services of the National Institute of Standards

and Technology. This publication is available from the following sources:

Superintendent of Documents

Government Printing Office

Washington, DC 20402

National Institute of Standards and Technology

Office of Physical Measurement Services

Physics Building, Room B354

Gaithersburg, MD 20899

National Institute of Standards and Technology

Publications and Program Inquiries Office

Administration Building, Room El28

Gaithersburg, MD 20899

The appendix of SP 250 listing current services and fees is issued twice yearly (usually

in June and December) and is available at no charge from the above sources.

Scheduled work assignments for calibrations and other tests generally will be made

in the order in which confirmed requests are received. However, work for U.S.

Government agencies may be given priority. For the regular services, the workload is usually such that the turnaround interval, between the date a customer's apparatus

is received and the date it is prepared for return shipment, will not be more than 45

days. Some types of instruments may require a longer calibration time, particularly

if their abnormal behavior requires reruns to check reliability. The customer who
49

can spare his instrument for only a short time usually can arrange shipment of his

instrument to NIST in time to meet the scheduled start of calibration. Generally, the

acknowledgment of the purchase order gives the expected completion date.

NIST staff does not undertake repair services. Therefore, all apparatus submitted for

calibration should be free of defects and in proper working order. Electrical contacts

should be in proper condition both mechanically and electrically.

A report is issued upon the calibration of each current transformer. This report

contains the measured values of the transformer and their uncertainties. Reports of

Calibration for current transformers are discussed in section 3.8 of the body of this

document and an example of a typical report is given.

NIST neither requires nor recommends intervals between NIST calibrations for

current transformers. These calibration intervals depend upon the performance of

the individual standard and the accuracy requirements of its application. These must

both be determined by the user. Some state utility commissions have established

calibration intervals determined to be necessary for such instruments.

Shipment of apparatus to NIST for calibration or other test should be made only

after the customer has accepted the estimate of cost and the tentative scheduling.

Apparatus not in good condition will not be calibrated. If defects are found

after calibration has begun, the effort may be terminated and data will be issued

summarizing information collected to that point. A fee may be charged in accordance

with the amount of work done.

The customer should carefully pack apparatus sent to NIST to minimize the likelihood

of damage in shipping and handling. In every case, the sender should consider the

nature of the apparatus, pack it accordingly, and clearly label shipments containing

fragile instruments or materials. Care should be taken in selecting the best mode of

transportation.

To minimize damage during shipment resulting from inadequate packing, the use

of strong, reusable containers is recommended. As an aid in preventing loss of

such containers, the customer's name should be legibly and permanently marked

on the outside. In order to prolong the container's use, the notation REUSABLE

CONTAINER, DO NOT DESTROY should be marked on the outside.

Shipping and insurance coverage instructions should be clearly and legibly shown on

the purchase order. The customer must pay shipping charges to and from NIST;

shipments from NIST will be made collect. The method of return transportation

should be stated and it is recommended that return shipments be insured, since

NIST will not assume liability for their loss or damage. For long-distance shipping

it is found that air freight provides an advantage in reduction of time in transit. If return shipment by parcel post is requested or is a suitable mode of transportation,

shipments will be prepaid by NIST, but without insurance coverage. When no
50 APPENDIX

shipping or insurance instructions are furnished, return shipment will be made by

common carrier collect and uninsured.

Shipments of transformers to NIST for calibration should be directed to:

National Institute of Standards and Technology

Electricity Division, MET B165

Rt. 1-270 and Quince Orchard Road

Gaithersburg, MD 20899
REFERENCES 51

7. REFERENCES

[1] J. D. Ramboz and R. C. McAulifF, "A Calibration Service for Wattmeters and

Watthour Meters," Natl. Bur. Stands. (U.S.), Tech. Note 1179, July 1983.

[2] W. E. Anderson, "A Calibration Service for Voltage Transformers and HighVoltage Capacitors," Natl. Bur. Stands. (U.S.), Spec. Publ. 250-33, June 1988.

[3] J. D. Simmons, Ed., "NIST CaHbration Services-Users Guide," Natl. Inst. Stands, and Technol., Spec. Publ. 250, 1989.

[4] American National Standard Requirements for Instrument Transformers, Am.

Natl. Stands. Instit., (ANSI/IEEE) C57.13-1978 (reaffirmed, 1986), New York.

[5] W. J. M. Moore and P. N. Miljanic, "The Current Comparator," Peter Peregrinus

Ltd., London, 1988. [lEE Electrical Measurement Series; 4].

[6] R. L. Kahler, "An Electronic Ratio Error Set for Current Transformer

Calibration," IEEE Trans. Instrum. Meas., Vol. IM-28, No. 2, June 1979.

[7] N. L. Kusters and W. J. M. Moore, "The Development and Performance of

Current Comparators for Audio Frequencies," IEEE Trans. Instrum. Meas., Vol.

IM-14, No. 4, Dec. 1965.

[8] T. M. Souders, "Wide-Band Two-Stage Current Transformers of High Accuracy,"

IEEE Trans. Instrum. Meas., Vol. IM-21, No. 4, Nov. 1972.

[9] 0. Petersons and W. E. Anderson, "A Wide-Range High-Voltage Capacitance

Bridge with One PPM Accuracy," IEEE Trans. Instrum. Meas., Vol. IM-24, No.

4, Dec. 1975.

[10] Bureau International des Poids et Mesures (BIPM) Working Group on the

Statement of Uncertainties; Metrologia, Vol. 17, No. 2, p. 73, 1981.

[11] B. L. Dunfee and W. J. M. Moore, "An International Comparison of CurrentRatio Standards at Audio Frequencies," IEEE Trans. Instrum. Meas., Vol. IM-14,

No. 4, Dec. 1965.

[12] A. Braun, H. Danneberg, and W. J. M. Moore, "An International Comparison of

50-60 Hz. Current-Ratio Standards at Currents Up to 60,000 A," IEEE Trans.

Instrum. Meas., Vol. IM-27, No. 4, Dec. 1978.

[13] W. Schwitz, R. Kampfer, A. Braun, T. M. Souders, W. J. M. Moore,

B. R. Cassidy, and T. A. Decon, "International Comparison of Current

Transformer Cahbrations," IEEE Trans. Instrum. Meas., Vol. IM-34, No. 2, June

1985.

* U.S. G.P.O. :1991-281-557:40033

NIST.114A U.S. DEPARTMENT OF COMMERCE

(REV. 3-90) NATIONAL INSTITUTE OF STANDARDS AND TECHNOLOGY

BIBLIOGRAPHIC DATA SHEET

1. PUBUCATION OR REPORT NUMBER NIST/SP-250/36

2. PERFORMING ORGANIZATION REPORT NUMBER

3. PUBUCATION DATE June 1991

4. TITLE AND SUBTITLE

A Calibration Service for Current Transformers

S. AUTHOR(S)

John D. Ramboz and Oskars Petersons

6. PERFORMINQ ORGANIZATION OF JOINT OR OTHER THAN NIST, SEE INSTRUCTIONS)

U.S. DEPARTMENT OF COMMERCE

NATIONAL INSTITUTE OF STANDARDS AND TECHNOLOGY

GAITHERSBURG, MD 20S99

7. CONTRACT/GRANT NUMBER

8. TYPE OF REPORT AND PERIOD COVERED Final

9. SPONSORING ORGANIZATION NAME AND COMPLETE ADDRESS (STREET, CITY, STATE, ZIP)

Same as item #6 above.

10. SUPPLEMENTARY NOTES

11. ABSTRACT (A 200-WORD OR LESS FACTUAL SUMMARY OF MOST SIGNIFICANT INFORMATION. IF DOCUMENT INCLUDES A SIGNIFICANT BIBUOGRAPHY OR

LITERATURE SURVEY, MENTION IT HERE.) A calibration service at the National Institute of Standards and Technology

(NIST) for laboratory-quality current transformers is described. The service

provides measurements of the current ratio and the phase angle between the

secondary and primary currents. In the Report of Calibration or Test, the

measured ratio is reported as the product of the marked (nominal) ratio and

the ratio correction factor. The measured phase angle is reported directly in nuUiradians (mrad) and is positive if the secondary current leads the primary.

The range of primary- to-secondary current ratios that can be measured

with the eqtupment at NIST extends from 0.25 A:5 A to 12 000A:5A. The

maximum current at the present time is about 20 000 A. Estimates of calibration

uncertainties, including their sources, are given and quality control procedures

are described. For routine calibrations, uncertainties of ±0.01% for the ratio

and ±0.1 mrad for the phase angle are quoted. However, lower uncertainties

to ±0.0005% or 5 parts per million (ppm) for ratio and ±0.005 mrad or 5 /zrads

for phase angle—are possible under the provisions of Special Tests.

12. KEY WORDS (6 TO 12 ENTRIES; ALPHABETICAL ORDER; CAPITAUZE ONLY PROPER NAMES; AND SEPARATE KEY WORDS BY SEMICOLONS)

calibration; current transformers; electric power; electric standards; NIST calibration

service

13. AVAILABIUTY UNUMITEO

FOR OFFICIAL DISTRIBUTION. DO NOT RELEASE TO NATIONAL TECHNICAL INFORMATION SERVICE (NTIS). X

ORDER FROM SUPERINTENDENT OF DOCUMENTS, U.S. GOVERNMENT PRINTING OFFICE. WASHINGTON, DC 20402. ORDER FROM NATIONAL TECHNICAL INFORMATION SERVICE (NTIS). SPRINGRELD.VA 22161.

14. NUMBER OF PRINTED PAGES

60

IS. PRICE

ELECTRONIC FORM

THE SP 250 SERIES ON NIST MEASUREMENT SERVICES*

SP 250-1 Spectral Radiance Calibrations PB871 79883

SP 250-2 Far Ultraviolet Detector Standards PB87227609

SP 250-3 Radiometric Standards in the Vacuum

Ultraviolet PB87227625

SP 250-4 Fricke Dosimetry in High-Energy Electron Beams

PB881 10374

SP 250-5 Alpha-Particle Calibrations PB881 68620

SP 250-6 Regular Spectral Transmittance PB881 08550

SP 250-7 Radiance Temperature Calibrations PB881 23674

SP 250-8 Spectral Reflectance PB881 09905

SP 250-9 Calibration of Beta-Particle-Emitting

Ophthalmic Applicators

PB881 08535

SP 250-10 Radioactivity Calibrations with the "4it" Gamma Ionization Chamber and Other

Radioactivity Calibration Capabilities

PB881 23708

SP 250-1 1 Dosimetry for High Dose Applications

PB88201587

SP 250-12 Neutron Personnel Dosimetry

PB87227617

SP 250-13 Activation Foil Irradiation with Californium

Fission Sources PB88217443

SP 250-14 Activation Foil Irradiation by Reactor Cavity

Fission Sources PB88217435

SP 250-15 Photometric Calibrations PB881 53747

SP 250-16 Calibration of X-Ray and Gamma-Ray

Measuring Instruments PB88211826

SP 250-17 The NBS Photodetector Spectral Response

Calibration Transfer Program

PB88201595

SP 250-18 Neutron Source Strength Calibrations PB88211818

SP 250-19 Calibration of Gamma-Ray-Emitting

Brachytherapy Sources PB89 193858

SP 250-20 Spectral Irradiance Calibrations PB881 23781 SP 250-21 Calibration of Beta-Particle Radiation Instrumentation PB88201579

SP 250-22 Platinum Resistance Thermometer

Calibrations PB881 38367

SP 250-23 Liquid-in-Glass Thermometer

Calibration Sen/ice

PB891 28888

SP 250-24 Standard Cell Calibrations PB881 23690

SP 250-25 Calibration Sen/ice for Inductive Voltage Dividers SP 250-26 NBS Phase Angle Calibration Services PB88225636

SP 250-27 AC-DC Difference Calibrations PB892222616

SP 250-28 Solid-state DC Voltage Standard Calibrations PB881 68703

SP 250-29 Traceable Frequency Calibrations PB881 68364

SP 250-30 GOES Satellite Time Code Dissemination:

Description and Operation

PB881 68760

SP 250-31 Mass Calibrations PB891 53894

SP 250-32 A Calibration Sen/ice for 30 MHz

Attenuation and Phase Shift PB88238324

SP 250-33 A Calibration Service for Voltage Transformers and High-Voltage Capacitors

PB882252903

SP 250-34 High Vacuum Standard and Its Use

PB891 93841 SP 250-35 The Calibration of Thermocouples

and Thermocouple Materials PB89209340

SP 250-36 A Calibration Sen/ice for Current Transformers SP 250-40 Absorbed-Dose Calibration of

Ionization Chambers in a ^Co Gamma-Ray Beam

* Entries Entries containing a stock number (SN003-003-) and price can be purchased from the Superintendent of Documents, U.S. Government Printing Office, Washington, DC 20402. GPO will accept checks, money orders, VISA, and MasterCard. For more

infomation, or to place an order, call (202) 783-3238. Be sure to cite the stock number on all orders.

Entries containing PB numbers can be purchased from the National Technical Information Sen/ice, Springfield, VA 22161. NTIS will accept American Express in addrtion to the payment methods listed for GPO. For more information call (703) 487-4650; to place an order

call (800) 336-4700. Be sure to cite the PB number on all orders.

Entries without stock or PB numbers are in preparation.
U.S. Department of Commerce

National Institute of Standards and Technology

Gaitfiersburg, MD 20899

Official Business

Penalty for Private Use $300