NAT'L INST. OF STAND & TECH RI.C,

A111D4 TiSm?

MIST

PUBIICATJONS

"iPiiiliiiilia^^

NisrUnited States Department of Commerce

Technology Administration

National Institute of Standards and Technology

NIST Technical Note 1375

Band-Limited, White Gaussian Noise

Excitation for Reverberation

Chambers and Applications to

Radiated Susceptibility Testing

Myron L. Crawford

Thomas A. Loughry

Michael O. Hatfield

Gustav J. Freyer

NIST Technical Note 1375

Band-Limited, White Gaussian Noise

Excitation for Reverberation

Chambers and Applications to

Radiated Susceptibility Testing

Myron L. Crawford

Thomas A. Loughry

Michael O. Hatfield

Gustav J. Freyer

Electromagnetic Fields Division

Electronics and Electrical Engineering Laboratory

National Institute of Standards and Technology

325 Broadway

Boulder, Colorado 80303-3328

January 1996

^^AT^S O^

U.S. DEPARTMENT OF COMMERCE, Ronald H. Brown, Secretary

TECHNOLOGY ADMINISTRATION, Mary L. Good, Under Secretary for Technology

NATIONAL INSTITUTE OF STANDARDS AND TECHNOLOGY, Arati Prabhakar, Director
National Institute of Standards and Technology Technical Note

Natl. Inst. Stand. Technol., Tech. Note 1375, 116 pages (January 1996)

CODEN:NTNOEF

U.S. GOVERNMENT PRINTING OFFICE

WASHINGTON: 1996

For sale by the Superintendent of Documents, U.S. Government Printing Office, Washington, DC 20402-9325
FOREWORD

The research described in this technical note represents the results of a demonstration of the

characteristics of the electromagnetic environment (EME) produced in a reverberation chamber by

Band-Limited, White Gaussian Noise (BLWGN) excitation followed by a comprehensive test program. The three-phase test program was specifically planned to address the feasibility of using the

BLWGN technique for conducting system susceptibility and shielding effectiveness tests.

The primary objectives of the research were to:

• characterize the EME produced by exciting a reverberation chamber with BLWGN,

• evaluate sensor and probe performance with BLWGN excitation, and

• demonstrate the feasibility of using BLWGN for radiated susceptibility and shielding

effectiveness testing.

The tests were conducted at the reverberation and anechoic chamber facilities of the National

Institute of Standards and Technology (NIST) and the reverberation chamber of the Naval Surface

Warfare Center, Dahlgren Division (NSWCDD). The tests used much of the hardware and software

developed by the USAF Phillips Laboratory for implementing the BLWGN excitation technique.

The reverberation chambers were excited both conventionally with CW and swept frequency

signals and mechanical stirring, and with the BLWGN technique. The BLWGN excitation included

noise bandwidths of 2, 4, 10, 20, and 50 MHz. The frequency range covered was 200 MHz to 18

GHz. The NIST reverberation chamber typically is not used below 200 MHz as a mode-stirred

chamber because of the falloff in mode density below this frequency. Limited data were obtained,

however, as low as 50 MHz where only a few modes exist in the NIST chamber.

A variety of sensors and probes were used for chamber EME characterization as well as for

probe evaluations. These included calibrated log periodic and dual ridged horn antennas, an 8 mm

dipole antenna, surface mounted D-Dot and B-Dot probes, and a free field B-Dot probe.

The response of several test articles including standard TEM cells, a special common test article (CTA) for electromagnetic vulnerability tests, an instrumented 7.0 cm (2.75 in) folding-fin

aircraft rocket, and a panel for shielding effectiveness measurements were evaluated to satisfy the test objectives.

The test team consisted of Captain Thomas A. Loughry, SRA David Little, and SRA Chris

VanZandt of the USAF Phillips Laboratory; Michael O. Hatfield of the Naval Surface Warfare

Center, Dahlgren Division; Myron L. Crawford and William Riddle of the National Institute of

Standards and Technology; Gustav J. Freyer of Universal Systems Incorporated; and Tony Keyes of

EG&G.

Ill

CONTENTS

Page

FOREWORD iii

LIST OF TABLES vi

LIST OF FIGURES ; vi

1. INTRODUCTION AND BACKGROUND 1

2. BAND-LIMITED, WHITE GAUSSIAN NOISE EXCITATION 3

2.1 Mode Mixing Through Frequency Agility 3

2.2 Implementing BLWGN Excitation 4

3. PHENOMENOLOGY 5

3.1 Electromagnetic Sensors 5

3.1.1 Background 5 3.1.2 Theoretical Transfer Functions 5

3.1.3 Inherent Sensor Uncertainties 6

3.1.4 Problems Associated with Baluns 6

3.1.5 Experimental Results 7

3.2 Field Uniformity 7

3.2.1 Theory 7

3.2.2 Experimental Results 9

3.3 Peak-to-Average Ratio 9

3.4 Power Density Enhancement Using Multiple Sources 11 3.5 Time to Achieve Statistical Averaging 12

4. EQUIPMENT UNDER TEST RESPONSES 13

4.1 TEM Cells 14

4.1.1 Theory of TEM Cells 14

4.1.2 Experimental Results 15

4.1.3 Explanation of Results and Coupling Cross Sections 15

4.2 Electromagnetic Vulnerability Testing 17

4.2.1 Test Item Description 17

4.2.2 Experiment and Results 18

4.2.2.1 Band-Limited, White Gaussian Noise Excitation 19

4.2.2.2 Continuous Wave Excitation Using Mechanical Stirring 20

4.2.2.3 Comparison of Noise Data to Continuous Wave Data 20

4.2.2.4 Comparison of Chamber Input Power Required for Upset 21

4.3 Hazards of Electromagnetic Radiation to Ordnance Testing 21

4.3.1 Test Item Description 21

4.3.2 Instrumentation 22

4.3.3 Experiment and Results 23

4.3.3.1 Band-Limited, White Gaussian Noise Excitation 23

4.3.3.2 Continuous Wave Excitation Using Mechanical Tuning 23

4.3.3.3 Comparison of Noise Data to Continuous Wave Data 24

4.3.3.4 Anechoic Chamber Data 24

4.3.3.5 Comparison to Anechoic Chamber Data 25
4.4 Shielding Effectiveness Testing of Materials 25

4.4.1 Shielding Theory 25

4.4.2 Test Item Description 26

4.4.3 Experiment and Results 26

4.4.3.1 Band-Limited, White Gaussian Noise Excitation 26

4.4.3.2 Continuous Wave Excitation Using Mechanical Tuning 27

4.4.3.3 Comparison of BLWGN Data to Continuous Wave Data 28

5. SPECIAL CONSIDERATIONS ASSOCIATED WITH TESTING WITH

BLWGN EXCITATION 28

5.1 Instrumentation and Probes 28

5.2 Mixers 29

6. CONCLUSIONS 30

6.1 Chamber Environmental Characterization 30

6.1.1 Effective Mode Density 30

6.1.2 Field Uniformity 30

6.1.3 Field Peak-to-Average Ratio 30

6.1.4 Time Between Peak Field Excitation 31

6.2 Test Applications 31

6.2.1 Peak Power Sensitive EUTs 31

6.2.2 Average Power Sensitive EUTs 31

6.2.3 Shielding Effectiveness Measurements 31

6.3 Test Considerations 31

7. RECOMMENDATIONS 32

8. REFERENCES 32

VI
LIST OF TABLES

Table Page

3.1 Standard deviation of field ratios for different number of effective modes 8

4.1 Mode cutoff and resonant frequency analysis of 12 cm x 12 cm TEM cell 18

4.2 Mode cutoff and resonant frequency analysis of 3 cm x 6 cm TEM cell 18

4.3 Frequency ranges for swept BLWGN excitation testing of folding-fm aircraft rocket. 24

LIST OF FIGURES

Figure Page

2-1. Generating and measuring NBWGN excited fields 35

2-2. Theoretical spectrum generated by NBWGN method 35

2-3. Actual output of NBWGN excitation method with excitation method with a center frequency of 1.5 GHz and four agility bandwidths 36

3-1. Probe—balun configuration 37

3-2. Field uniformity with 10 MHz of NBWGN over 1 to 2 GHz 37

3-3. Field uniformity Relative Frequency Distribution for 2 MHz of NBWGN

at 0.5 to 1 GHz 38

3-4. Field uniformity Relative Frequency Distribution for 10 MHz of NBWGN

at 0.5 to 1 GHz 38

3-5. Field uniformity Relative Frequency Distribution for 20 MHz of NBWGN

at 0.5 to 1 GHz 39

3-6. Field uniformity Relative Frequency Distribution for 50 MHz of NBWGN

at 0.5 to 1 GHz 39

3-7. Field uniformity Relative Frequency Distribution for 2 MHz of NBWGN

at 1 to 2 GHz 40

3-8. Field uniformity Relative Frequency Distribution for 10 MHz of NBWGN

at 1 to 2 GHz 40

3-9. Field uniformity Relative Frequency Distribution for 20 MHz of NBWGN

at 1 to 2 GHz 41

vu
Figure Page

3-10. Field uniformity Relative Frequency Distribution for 50 MHz of NBWGN

at 1 to 2 GHz 41

3-11. Field uniformity Relative Frequency Distribution for 2 MHz of NBWGN

at 2 to 4 GHz 42

3-12. Field uniformity Relative Frequency Distribution for 10 MHz of NBWGN

at 2 to 4 GHz 42

3-13. Field uniformity Relative Frequency Distribution for 20 MHz of NBWGN

at 2 to 4 GHz 43

3-14. Field uniformity Relative Frequency Distribution for 50 MHz of NBWGN

at 2 to 4 GHz 43

3-15. Field uniformity Relative Frequency Distribution for 2 MHz of NBWGN

at 4 to 8 GHz 44

3-16. Field uniformity Relative Frequency Distribution for 10 MHz of NBWGN

at 4 to 8 GHz 44

3-17. Field uniformity Relative Frequency Distribution for 20 MHz of NBWGN

at 4 to 8 GHz 45

3-18. Field uniformity Relative Frequency Distribution for 50 MHz of NBWGN

at 4 to 8 GHz 45

3-19. Field uniformity Relative Frequency Distribution for 2 MHz of NBWGN

at 8 to 18 GHz 46

3-20. Field uniformity Relative Frequency Distribution for 10 MHz of NBWGN

at 8 to 18 GHz 46

3-21. Field uniformity Relative Frequency Distribution for 20 MHz of NBWGN

at 8 to 18 GHz 47

3-22. Field uniformity Relative Frequency Distribution for 50 MHz of NBWGN

at 8 to 18 GHz 47

3-23. Average measured Q for the NIST chamber 48

3-24. Equivalent modes per megahertz of NBWGN available for mode mixing 48

3-25. Chamber power density peak-to-average ratio measurement for NBWGN excitation. 49

viu
Figure Page

3-26. Chamber power density peak-to-average ratio measurement for CW

(mechanical tuning) and 10 MHz BW NBWGN excitation as a function

of absorber loading (Q) 49

3-27. Block diagram of system for increasing test field inside reverberation

chamber using a single noise source and mixer and two rf amplifiers 50

3-28. Block diagram of system for increasing test field inside

reverberation chamber using two noise sources and mixers and two rf amplifiers. . . 51

3-29. Power density measured inside NIST reverberation chamber excited using single

noise source, (a) power density from single and dual inputs, (b) increase

in power density from single input to dual input (Pd dual - Pd single) 52

3-30. Power density measured inside NIST reverberation chamber excited using two

noise sources, (a) Power density from single and dual inputs, (b) Increase in

power density from single input to dual input 53

3-31. Block diagram of system for measuring transient response and field uniformity

of NBWGN excited chamber 54

3-32. BLWGN 10 MHz BW time domain data for NIST reverberation chamber at 1 GHz. 55

3-33. BLWGN 50 MHz BW time domain data for NIST reverberation chamber at 1 GHz. 56

4-1. Geometry for an apertured TEM cell 57

4-2. SE measurements of NIST 12 cm TEM cell obtained using NIST reverberating

chamber with mechanical mode-mixing 58

4-3. SE measurements of NIST 6 cm TEM cell obtained using NIST reverberating

chamber with mechanical mode-mixing 58

4-4. Effect of cross-section bandwidth smaller than bandwidth of gating function 59

4-5. Effect of cross-section bandwidth greater than bandwidth of gating function 59

4-6. Narrow cross-section resonance of 12 cm TEM cell measured using BLWGN 60

4-7. Wide cross-section resonance of 6 cm TEM cell measured with BLWGN 60

4-8. Common test article showing interconnections and functions 61

4-9. Susceptibility profile of common test article obtained using CW excitation

in the NSWCDD mechanically mode-mixed reverberation chamber 61

IX
Figure Page

4-10. Set-up for performing susceptibility tests of common test article in a reverberation chamber 62

4-11. Block diagram of system for performing susceptibility tests of common test article in NIST reverberation chamber using BLWGN excitation 62

4-12. Block diagram of system for performing susceptibility tests of common test article in NIST reverberation chamber using CW excitation with mechanical

mode-mixing 63

4-13. Susceptibility profile of common test article obtained with

BLWGN (5 bandwidths) 64

4-14. Peak and average power susceptibility levels of common test article obtained with CW excitation and mechanical mode mixing 64

4-15. Difference between peak and average power susceptibility levels of

common test article 65

4-16. Comparison of BLWGN and average CW (mechanical mode mixing) power

densities data for common test article upset 66

4-17. Difference between CW (mechanical mode mixing) average and BLWGN

data for common test article upset 66

4-18. Difference between CW (mechanical mode mixing) peak and BLWGN

data for common test article upset 67

4-19. Difference between input power to NIST reverberation chamber required for

upset of common test article determined from average of all probe and

antenna measurements and from each individual sensor 68

4-20. Diagram of modified folding-fm aircraft rocket 69

4-21. Diagram of MetriCor system 70

4-22. Diagram of instrumented BED 70

4-23. Folding-fin aircraft rocket mounted on styrofoam block in reverberation chamber 71

4-24. Block diagram - BLWGN set-up for testing susceptibility of folding-fm

aircraft rocket in NIST reverberation chamber 72

4-25. Folding-fin aircraft rocket response - 0.05 to 0.5 GHz data 73

4-26. Folding-fin aircraft rocket response - 0.5 to 1 GHz data 74

X
Figure Page

4-27. Folding-fm aircraft rocket response - 1 to 2 GHz data 75

4-28. Folding-fin aircraft rocket response - 2 to 4 GHz data 76

4-29. Block diagram of CW, mechanical mode-mixing set-up for testing

susceptibility of folding-fm aircraft rocket in NIST reverberation chamber 77

4-30. Response of folding-fm aircraft rocket to CW excitation with

mechanical mode mixing 78

4-31. Average CW response of folding-fm aircraft rocket vs. noise response

(2 MHz bandwidth) 78

4-32. Folding-fm aircraft rocket in anechoic chamber 79

4-33. Diagram of anechoic chamber discrete frequency set-up 80

4-34. Folding-fm aircraft rocket response to vertically polarized illumination

at 553.5 MHz and 631 MHz 81

4-35. Folding-fm aircraft rocket response to horizontally polarized illumination

at 553.5 MHz 82

4-36. Folding-fm aircraft rocket response to horizontally polarized illumination

at 555.5 MHz 83

4-37. Folding-fm aircraft rocket response to horizontally polarized illumination

at 600 MHz 84

4-38. Folding-fm aircraft rocket response to horizontally polarized illumination

at 631 MHz 85

4-39. Diagram of anechoic chamber swept frequency set-up 86

4-40. Anechoic chamber response of folding-fm aircraft rocket 87

4-41. Folding-fm aircraft rocket anechoic chamber data vs. reverberation chamber

2 MHz noise data 87

4-42. Folding-fm aircraft rocket anechoic chamber data vs. reverberation chamber

2 MHz noise data corrected for polarization factor 88

4-43. "NESTED" chamber technique for collecting SE data 89

4-44. Photo of wire mesh window 89

XI
Figure Page

4-45. Photo of test fixture used for measuring the shielding effectiveness of

shielded window using the nested reverberation chamber technique 90

4-46. Shielded window mounting for test fixture 90

4-47. Window SE test fixture mounted inside NIST reverberation chamber 91

4-48. Reverberation chamber CW (mechanical mode mixing) test set-up

for SE measurements 92

4-49. Reverberation chamber BLWGN test set-up for SE measurements 93

4-50. Results of window SE measurements made using BLWGN technique

with 2 MHz bandwidth 94

4-5 1 . Results of window SE measurements made using BLWGN technique

with 10 MHz bandwidth 94

4-52. Results of window SE measurements made using BLWGN technique

with 50 MHz bandwidth 95

4-53 . Comparison of baseline data collected from the test fixmre and

from the reverberation chamber 96

4-54. Results of window SE measurements made using BLWGN excitation

with and without window mounted to the test fixture 96

4-55. Results of window SE measurements made using CW (mechanical mode

mixing) excitation with and without window mounted to the test fixture 97

4-56. Difference in SE obtained using BLWGN and mechanical mode mixing 97

4-57. Difference in shielding after smoothing 98

5-1. Improved modulator configuration 98

Xll
BAND-LIMITED, WHITE GAUSSIAN NOISE EXCITATION OF

REVERBERATION CHAMBERS AND APPLICATIONS TO

RADIATED SUSCEPTIBILITY TESTING

Myron L. Crawford

National Institute of Standards and Technology

Boulder, CO 80303

Captain Thomas A. Loughry

USAF Phillips Laboratory

Kirtland AFB, NM 87117

Michael O. Hatfield

Naval Surface Warfare Center, Dahlgren Division

Dahlgren, VA 22448

Gustav J. Freyer

Universal Systems Incorporated

Monument, CO 80132

This report gives the results of demonstration tests conducted to (1) evaluate the electromagnetic

environment (EME) produced by band-limited, white gaussian noise (BLWGN) excitation of a reverberation chamber and to verify its applications to susceptibility and shielding effectiveness

(SE) testing. Data were collected to compare the EME produced in a reverberation chamber by

CW and swept frequency excitation using both mechanical stirring and BLWGN to excite the

cavity mode structure. The feasibility of using the BLWGN technique for radiated susceptibility

testing was evaluated by comparison with mechanical stirring in a reverberation chamber and with

anechoic chamber results. Within normal measurement uncertainties, the response of both types

of systems were the same for mechanical stirring and for BLWGN excitation and were consistent

with the results obtained in the anechoic chamber.

Key words: electromagnetic environment; mode stirring; noise; quality factor; reverberation

chamber; shielding effectiveness

1. INTRODUCTION AND BACKGROUND

Mode mixing a shielded enclosure to improve the electromagnetic compatibility (EMC) test results obtained using an enclosure was first proposed in 1968 [1]. Since then measurement procedures have

been developed for implementing mechanical mode-mixing techniques for both radiated susceptibility

and emission testing [2-6] with considerable success.

During the 1980s and early 1990s, agencies including the U.S. Air Force's Phillips Laboratory

(PL) (and its predecessor the Air Force Weapons Laboratory), Kirtland AFB, NM; the National

Institute of Standards and Technology (NIST), Boulder, CO; and the Naval Surface Warfare Center,

Dahlgren Division (NSWCDD), Dahlgren, VA, were investigating various low cost methods of

1
performing radio frequency precharacterization of systems identified for high power microwave

testing. Specifically, techniques that would quickly and economically allow researchers to identify the

frequencies and power densities at which a system would fail or upset were sought. It was hoped that EMC, EM vulnerability (EMV), and high power microwave vulnerability (HPMV) test costs could be

reduced. During this same period, NIST and NSWCDD were developing the use of large

reverberation chambers with mechanical tuners for reducing the costs of EMC/EMV testing.

The reverberation chamber technique offered several advantages over anechoic chamber or open

range testing. These include: (1) the ability to generate high intensity radiated fields over large

volumes using modest amounts of rf power, (2) the elimination of polarization and angle of incidence

dependence, hence removal of physical rotation requirements of the equipment under test (EUT), (3)

potential use of the test facility for both radiated susceptibility and emission testing with minor

instrumentation changes, and (4) the resulting significant reduction in test time requirements.

These advantages are derived from the inherent characteristics of a reverberation chamber (simply

a high quality factor (Q), metal enclosure) that acts as a multimoded resonant cavity with increased

test fields. (The increase in the fields is due to the stored energy within the high Q cavity.) The

chamber's tuner perturbs the fields inside the cavity, thus resulting in a statistically averaged isotropic

impinging field at the EUT. One difficulty with this technique, however, is that dependence on the

mechanical tuner requires that data be taken over many positions of the tuner to ensure that the EUT

is exposed to the maximum field levels for the given test parameters. Further, a real-time

homogeneous field within the chamber can only exist on the same time scale as the rotation rate of the

tuner.

In an attempt to remove these lunitations, the PL conducted an Independent Laboratory Initiated

Research (ILIR) program to investigate the likelihood of mixing the modes within the chamber

electronically. It was anticipated that this technique would provide real-time field uniformity within

the chamber and eliminate dependence on the mechanical tuner. By November of 1991, the PL had

demonstrated that electronic mode-mixing could be accomplished in a practical and economical manner by up-converting band-limited, white gaussian noise to microwave frequencies [7]. By using

up-converted white gaussian noise to spread the energy over a narrow bandwidth, the technique

randomly excites the reverberation chamber's resonant modes that lie within this bandwidth. This

causes a spatial averaging out of the "hot" and "cold" spots that would typically exist in a mechanically mned high-Q reverberation chamber at specific tuner positions. Since the bandwidth of

the resulting signal is relatively narrow compared to the center frequency, the test results can still be

considered characteristic of a discrete frequency. This technique is expected to reduce precharacterization test times by as much as 1000 over anechoic chamber or open range testing and by a

factor of about 200 over mechanically mode-mixed reverberation chambers. Section 2 of this report

contains a description of the band-limited, white gaussian noise (BLWGN) technique including some

information for implementing this type of excitation in a reverberation chamber.

Demonstration tests were performed at NIST in January 1992 by the PL with representatives from

the NSWCDD present. This then led to a fiscal year 1993 cooperative research project among the

three organizations to investigate and validate the technique as an EMC/EMV testing tool. The

research was conducted at NIST, Boulder, CO.

The project had two experimental phases. The first phase investigated the response characteristics

of probes and instrumentation with noise excitation versus mechanical mode mixing in the NIST

reverberation chamber. The results of these measurements are shown in section 3.1 of this report.
The purpose of these measurements was to quantify and gain understanding of the electronically

mode-mixed electromagnetic fields environment (EME) inside the reverberation chamber using

theoretically characterizable devices. These results are needed to appropriately interpret results obtained with electronic mode-mixed fields in relation to more conventional field parameters. A

second goal of phase I was to measure the field uniformity within the chamber as a function of

frequency and modulation or agility bandwidth. These data were then compared with the

mechanically mode mixed chamber data. The results of these measurements are contained in sections

3.2 and 3.3 of this report. The effects of chamber Q were also investigated for the two methods of

mode mixing. The final objective of phase I was to characterize some calibrated, apertured, TEM

cells as a function of frequency and compare the results to the theoretically calculated cell resonant modes. These results are contained in section 4.1 of this report.

Phase II was an in depth susceptibility evaluation of three test articles again using both mechanical

and electronic mode-mixing techniques. The results of these measurements are contained in sections

4.2, 4.3, and 4.4 of this report. The three test articles are unclassified and therefore lend themselves

well to unrestricted reporting of research results. The test articles were provided by the NSWCDD.

The first is a generic common test article (CTA). The CTA consists of two separate units that communicate with each other over a series of cables. The first unit generates a clock signal which is fed to the other unit, which then generates two separate binary counts for comparison. If the first unit detects any error in the counts, it reports the discrepancy by optical fiber link to the operator. A

more detailed description of the CTA can be found in section 4.2.1. The CTA has been extensively

characterized in both anechoic and reverberation chambers. The second test article is the Hazards of

Electromagnetic Radiation to Ordnance (HERO) rocket. The HERO rocket is a 7.0 cm diameter,

folding-fin aircraft rocket with a fiber optically instrumented electro-explosive device (EED). The

last test article is a wire mesh shielded window used as a standard for comparing shielding

effectiveness (SE) measurements. Like the CTA, the latter two test articles have been extensively

characterized.

Conclusions resulting from this study and recommendation for future work are contained in sections 6 and 7 of this report.

2. BAND-LIMITED, WHITE GAUSSIAN NOISE EXCITATION

2.1 Mode Mixing Through Frequency Agility

The conventional method of mode-mixing uses a metallic paddle wheel to continuously or

incrementally change the boundary conditions within the reverberation chamber while maintaining a

constant excitation frequency. Field homogeneity is achieved by averaging the contribution of many

different eigenfunctions to a given point in the chamber. An equivalent result can be obtained by

maintaining the boundary conditions constant while the frequency is allowed to vary over a narrow

interval about some center frequency. Then the contribution of each eigenfunction to a given spatial

point in the chamber, in that narrow band of frequency, is averaged to provide the field homogeneity.

The advantage of this method over mechanical mode mixing is that the eigenfunctions corresponding

to that band can be stimulated simultaneously and randomly by spreading the power over a band of

frequencies. Thus, because the energy stored in the fields in an arbitrarily small volume centered at any given spatial point is the sum of all the energies in the individual eigenmodes, field homogeneity

is achieved in real time. Real-time homogeneity can drastically reduce test times. Even if individual

eigenfunctions are changed by adding a test object or moving an existing test object within the
chamber, the mean over the frequency band will remain constant (within the limits of the sampling

variance).

Several techniques might be considered to provide the source frequency agility required. At

minimum, however, the method used should meet the following criteria:

a. The power spectrum should be flat across the agility bandwidth to simplify data interpretation.

b. The signal should be ergodic (or at least loosely time stationary over the maximum averaging

period of the test article and/or the power measuring instrument), so that real-time uniformity can be achieved.

c. The center frequency and bandwidth agility of the source should be variable over a wide

parameter set to facilitate test flexibility.

d. The average output power should be continuously variable or variable in 3 dB or less increments, from 1 W to a couple of hundred watts to allow for a wide range of cavity quality

factors and test article upset levels.

2.2 Implementing BLWGN Excitation

As shown in figure 2-1, BLWGN excitation uses band-limited, white gaussian noise (WGN) upconverted by performing double-sideband, suppressed-carrier modulation with an rf signal from a synthesized sweeper. The output is then amplified with a traveling wave tube (TWT) amplifier to meet the higher power requirements. Center frequencies between 0.1 and 50 GHz and frequency

agility bandwidths of less than one to as high as several hundred megahertz can be easily realized. In

addition, the power output from the sweeper can be varied in 1 dB increments from -120 dBm to approximately -1-15 dBm. Therefore, for practical purposes, the dynamic range of the technique is identically the dynamic range of the amplifier. Figure 2-2 shows the theoretical spectrum of the

signal at each stage of the signal processing. The synthesized sweeper is used to generate the

monochromatic signal which determines the center frequency of the output while the WGN source and

low pass filters provide the frequency agility. WGN will maintain its statistical properties after having passed through a linear, invariant system such as a low pass filter [8].

In the set-up of figure 2-1, the instrumentation is controlled by a central computer and data

plotting is available from a standard plotter. The probes in the chamber provide a method of

monitoring field strength. The instrumentation used to measure the power delivered by the probes

must be capable of averaging over the agility bandwidth of the BLWGN. Scalar analyzers or

standard power meters can be used; however, narrow band instruments such as spectrum analyzers

and vector analyzers would require additional data reduction.

Figure 2-3 shows the actual output of the BLWGN at a fixed center frequency of L5 GHz,

immediately after up-conversion as measured by a spectrum analyzer. The spectrum analyzer is used

here to demonstrate the spreading of the energy across the agility bandwidth but should not be used as a measuring device during actual EMC testing. In practice, the scalar analyzer integrates over the

spectrum shown in figure 2-3 to give a single power measurement at the center frequency.

The configuration shown in figure 2-1 allows three different test modes to be accomplished.

First, input power supplied to the chamber can be held constant while the frequency is swept at a
predetermined rate. This is particularly useful when measuring electromagnetic shielding as a function of frequency. Second, the frequency can be held constant while the power into the chamber

is incrementally increased. This has its strongest application in determining a system's susceptibility

at a given frequency. The system of interest is placed in the chamber and is operated as the power

density in the chamber is slowly increased until the system malfunctions. At this point, the power

density in the chamber is measured using one of the probes to determine the upset threshold. Finally,

both power and frequency can be held constant. This would typically be used for performing

diagnostics on the chamber or providing long term exposure fields.

3. PHENOMENOLOGY

3.1 Electromagnetic Sensors

3.1.1 Background

The accurate measurement of microwave frequency electromagnetic fields is fundamental to the

broadband electromagnetic characterization of any system. Measurements of continuous-wave

coupling, shielding effectiveness, and narrow band electromagnetic vulnerability all depend directly on field measurements. It is common practice when conducting electromagnetic measurements in the

reverberation chamber to use standard antennas such as log periodic or horn antennas to monitor

fields. Because these antennas have large effective apertures at low frequencies (less than a few

hundred megahertz), they can significantly load the chamber and reduce overall chamber Q [7]. Such

antenna also tend to be bulky and limit overall usable chamber volume. These limitations can be overcome with the use of simple sensors such as electrically short dipole probes or the B-dot and Ddot probes. The B-dot consists of a small cylindrical loop antenna while the D-dot is composed of a small asymptotic dipole above a ground plane. They offer much smaller volumes as well as relatively

smaller effective apertures. In fact, surface mount probes can be used in the chamber thus, requiring

much less use of the test volume.

3.1.2 Theoretical Transfer Functions

Theoretical sensor transfer functions are often used when measuring narrow band electromagnetic

fields at microwave frequencies. These functions convert sensor outputs to incident field values.

Standard sensors such as the B-dot probe, D-dot probe, and antennas like standard gain horns are well

understood [9-11]. Small dipole probes which use high impedance transmission lines typically are precalibrated in TEM cells or anechoic chambers to provide their transfer function.

When properly polarized to a free-space, planar incident wave, the transfer function relates probe

output power to incident field power density. It can be expressed as follows for the B-dot and D-dot

sensors or for a horn antenna:

Pj 50 c

B-dot:

(0-, (3-1) « ,.2
D-dot:

''"-'0^.^'..^

(3-2)

Pd. <=

Horn:

0)-,

(3-3)

where and ?„, is the power delivered to a 50 Q instrument by the sensor, Pd„ is the incident power

density, Ag is the sensor's equivalent area, ii^ is the permeability of free space, e^ is the permittivity

of free space, c is the speed of light in a vacuum, G is a horn's gain, and co is the narrow band center radian frequency, respectively. It should be noted that these equations only apply well below the

cutoff frequency of the B-dot and D-dot and within the design bandwidth of the horn antenna. When

one uses these probes and antennas in a reverberation environment, a polarization factor of 2 must be

accounted for [12] and their directivity simply becomes 1.

3.1.3 Inherent Sensor Uncertainties

Measurement uncertainties can be attributed primarily to two sources, those errors introduced by

the power measuring instrumentation and those introduced by the sensor itself. Power measurement

errors appear both in the acmal power measurement and system component calibrations, but these

uncertainties can usually be controlled to within ±1 dB. Sensor uncertainties include opposing field or sensor leg electric field pickup, deviation from the ideal equations presented earlier, and slight

uncertainties in the mechanical dimensions that determine equivalent area and gain. In practice, the

first two sources of error dominate the last. The opposing field and leg pickup can be reduced by

using multielement probes that provide a differential output for the desired field and a common mode

output for extraneous pickup [12]. Thus, by using a 180° summation device such as a hybrid coupler

or a balun, common mode errors are reduced to within the limitations of the summation component.

As will be shown shortly, a summation component such as a balun can introduce more error than it alleviates and therefore as not used in these experiments. Finally, the other significant source of error appears when the sensors are used near their cutoff frequencies. The B-dot probe, for example,

introduces two poles in the transfer function due to the inductance and the transient time associated

with the sensor loop [9] . The D-dot has a similar problem due to the asymptotic dipoles capacitance

with the ground plane [10]. By definition, at cutoff, a 3 dB error will occur.

3.1.4 Problems Associated with Baluns

As previously mentioned, it is common practice to use a 180° summation device to eliminate

common mode problems associated with opposing field and sensor electric field leg pick up. One of

the most popular summation devices is the balun, see figure 3-1. A seldom recognized problem with

the balun, however, is errors created by leakage between the two input ports. A small amount of

power leaks from one leg to the other and is transmitted back toward the sensor. A B-dot probe acts
as a short while a D-dot probe acts like an open when viewing the Thevenin or Norton equivalent source impedance. Thus, power transmitted back towards the sensor will be reflected back again

towards the balun. Reflections back from the sensor will add vectorially with the original signal

according to their amplitude and phase and will create a ripple in the balun output as the frequency is swept across a wide bandwidth. As shown in figure 3-1, the source match of the probe can be

improved by the introduction of in-line attenuators. Employing a geometric series identity,

accounting for both legs of the probe, and using a small argument approximation for the logarithm

allow the error associated with the balun leakage to be expressed as:

^^ hi(lO)

-{.B,,*S^,*2A) 20 (3-4)

where B,l, Srl, and A are the balun leg-to-leg insertion loss, the sensor return loss, and the attenuator

value, respectively, all in decibels. For example, if it is assumed that the sensor has a return loss of

dB (short or open) and no attenuators are used, a balun with a leakage insertion loss of greater than

15 dB must be used to achieve an uncertainty of less than 3 dB. Many actual baluns cannot meet this specification. If a 10 dB attenuator is added to each leg, the same balun can achieve an error of less than 0.3 dB; however, the sensor sensitivity is reduced by 10 dB.

3.1.5 Experimental Results

Fields were generated in the chamber using discrete frequencies for both CW (mechanical mode

mixing) and 2, 10, 20, and 50 MHz of BLWGN excitation. The fields were then measured using a

surface mount D-dot, free field B-dot, surface mount B-dot, horn probe, NIST reference horn, and an

8 mm dipole with a resistive lead. The discrete frequencies measured included 0.5, 0.6, 0.8, 1.0,

1.5, 2.0, 3.0 4.0 4.5, 5.0, 5.5, 6.0, 6.5, 7.5, 8.0, 10.0, 12.0, 14.0, 16.0, and 18.0 GHz. Each

sensor was used in the appropriate frequency range matching its usable bandwidth. In all cases except

for the 8 mm dipole with resistive lead, all sensors measured the fields accurately to within the field uniformity as described in section 3.2. The 8 mm dipole with high resistance leads uses a diode

detector and a low pass filter circuit which caused erroneous readings when used with the BLWGN

excitation technique. The low pass filtering associated with the 8 mm dipole probe's high resistance

lead and low pass filter attenuated the upper frequency spectrum of the down converted noise and thus

tended to lower the probe's indicated field measurements as compared with the other sensors.

3.2 Field Uniformity

3.2.1 Theory

The purpose of mode mixing in a reverberation chamber is to provide a statistically averaged

uniform or homogeneous field throughout the volume of the chamber. A measure of uniformity \p can be expressed as

where w^^ and w,^ are the maximum and minimum possible local average energy densities found

anywhere in the chamber. Because power density and energy density are directly related, minimum

w

t|/ - 10 log^^, (3-5)
Table 3.1. Standard deviation of field ratios for different number of effective modes.

N o N a N CT

2 11.35 16 2.54 350 0.515

3 8.47 18 2.35 600 0.380

4 6.58 20 2.23 850 0.323

5 5.61 25 2.03 1100 0.286

6 4.72 29 1.84 1350 0.260

7 4.49 35 1.65 1600 0.238

8 3.88 45 1.46 1850 0.220

10 3.47 57 1.26 2100 0.200

12 3.15 82 1.02 2350 0.190

14 2.84 100 0.934 2600 0.186

and maximum average power density can replace average energy density in eq (3-5) with equal

validity.

Predicting field uniformity for a given reverberation chamber has been previously treated [7]. Table 3.1 shows the correlation between the number N of modes and the standard deviation o of the

ratio of two average power density measurements taken at two random positions. Traditionally, N

has been calculated using the theoretical mode density expressed as

N = ^f df, (3-6) c

where V is the cavity volume, c is the speed of light, f is the center frequency, and df is the

bandwidth over which the modes are stirred. This, however, does not account for the overlapping of

modes due to their non zero bandwidth. This can be, at least heuristically, corrected for by taking

the ratio of the noise bandwidth to the bandwidth BWq of the resonant modes giving the resulting

reduced number Nq of effective modes. The resonant mode bandwidth is calculated as

BW^ = /. (3-7)

If Nq is less than N, then Nq should be used in place of N in table 3.1. As indicated in ref. [7], the

actual uniformity as defined above can be related to the field standard deviation as
ilf = 2.5 o. (3-8)

3.2.2 Experimental Results

The uniformity of fields in the chamber can be measured by placing the same probe in two

different locations and measuring the difference in field intensity as a function frequency. It is assumed that the statistics of the fields vary little over small changes in frequency. Thus, sampling

the fields over a small frequency span at a single location in the chamber is equivalent to sampling

several chamber locations at a single frequency. Typical data are shown in figure 3-2 for the 1 to 2 GHz band using 10 MHz of BLWGN. A more demonstrative way to represent the same data is to use a relative frequency distribution. The ordinate is broken down into 100 equal-sized cells of A dB

and the total number of data points falling within each cell is then plotted to form a relative frequency

distribution (also referred to as a probability distribution). Figures 3-3 through 3-22 show the field uniformity relative frequency distribution for the 0.5 to 1, 1 to 2, 2 to 4, 4 to 8, and 8 to 18 GHz for

2, 10, 20, and 50 MHz of BLWGN.

The number of modes expected in the NIST chamber at these frequencies is very high. For

example, at 1 GHz and with 2 MHz of BLWGN, the total modes available for stirring can be

calculated from eq (3-6) as approximately 74. Equation (3-8) and table 3.1 show that the fields at these input parameters will not to vary more than ±2.5 dB. The data in figure 3-3 indicate much

more variation in the average power density. However, the field uniformity stays relative the constant as the center frequency of the BLWGN is increased as indicated in figures 3-11 and 3-15.

Equation (3-6) and table 3.1 suggest the opposite. This implies that some other mechanism may be at work to affect the actual number of modes available for mode mixing.

This effective lowering of the mode density may be explained by considering the finite Q of the

chamber as discussed above. The Q for the NIST chamber was measured using the methods

described in refs. [7] and [13]. Figure 3-23 shows average results for 2, 10, 20, and 50 MHz of

BLWGN. Of more interest, however, is the ratio of BLWGN bandwidth to Q bandwidth as described

in eq (3-7). Figure 3-24 demonstrates this ratio N^ per megahertz of BLWGN. For example, at 1 GHz, Nq is approximately 4 as read from figure 3-24. Hence, for 2 MHz of BLWGN at 1 GHz, the

equivalent modes available for mode mixing would be 8. This corresponds to a uniformity of

approximately ± 10 dB according to table 3.1 and eq (3-8). This is a much better predictor of the

data in figure 3-23 . The dropoff of uniformity with frequency is also explained by the dropoff of Nq

with frequency as shown in figure 3-24. Thus, the chambers Q can and does have a profound effect on the ability of BLWGN (and mechanical mode mixing as well) to stir the fields in the chamber.

3.3 Peak-to-Average Ratio

The peak-to-average ratio of the fields in a mechanically stirred chamber is typically in the range

7 to 8 dB [6]. The same ratio can be demonstrated using BLWGN excitation. Figure 3-1 shows the

fields measured using a B-dot probe and diode detector operating in its square law region and

therefore the voltage shown on the vertical axis is directly proportional to power. Two traces are shown on the oscilloscope. The first is a real-time field variation, while the second is a ensemble

average of many such traces. Hence, the peak-to-average ratio can be calculated by the ratio of the

peak value of trace 1 (6.3 mV) to the average value of trace 2 (1.3 mV) or 6.8 dB).
The theoretical calculation of the peak-to-average ratio for the BLWGN excitation method follows

the mechanical mode-mixing calculation [6] identically and hence accounts for experimental results indicating the same peak-to-average ratio in both cases. The mechanical mode-mixing calculation assumes that each position of the paddle wheel contributes a random independent sample of a mode

structure defined as

(^\x,y,z) = sin^C^/) sin^C^^) sin^C^^), (3-9)

k^ = ^^, k^ = rL!L, k^ = P-^, (3-10)

where m, n, and p are integers, and a, b, and d are the dimensions of the chamber. The simplified

eq (3-1) represents only single mode, and the multimode, vector namre of the field should be included

in a future analysis. The paddle wheel directly perturbs the mode indices k^, ky, and k^ causing the

sine arguments to behave like random variables. Here it is assumed that the amplitude of the mode

does not change with the tuner position although in practice it will because of changing cavity input

impedance mismatch. At this point, ref. [6] lets x, y, and z become the variable of integration and

finds the expected value of <^^(x,y,z). In reality, the probe being used to measure the fields is fixed at a point (Xp, yp, Zp) and the mode indices are the random variables. Relative to the math, it does not matter which is considered the random variable provided that the total normalized argument of the

sine is a uniform random variable on the interval -tt to -t- tt or equivalently because the sin? repeats

every tt, the interval to tt. The average value of the fields at the point (Xp, yp, Zp) can be expressed

equivalently as

^^ = £ [sin^Cvj) sin^Cvj) sm\v^)\, (3-11)

where v^, V2, and v^ are random variables with the following probability distributions

y(vj) = — ; O^Vj^TT,

jiv^) = 1 ; O^Vj^Tt, (3-12)

/(V3) = — ; O^Vj^TT,

Their joint distribution is

AVi,V2,V3) = \ (3-13)

71

and hence,

10
7C Jt Tt J2 ^ |||-L sin^Cvj) sin2(v2) sin^Cvj) dv^dv^dv^ = -, (3-14)

r»000n n 1^

which is equivalent to the ref. [6].

The frequencies f at which resonant modes exist in the

cavity are defined by

/ = ^=^\l^x + ^y + ^z- (3-15)

Thus, if the frequency f is changed randomly over a small interval of frequency, as is the case in BLWGN excitation, k^, ky, and k^ must change randomly but in accordance with eqs (3-15) and (3-

16). Therefore, the same resuhing peak-to-average value would be expected (8 to 1, or roughly 9 dB

for a single mode). In practice, the peak-to-average ratio rarely exceeds 8 dB for either mechanical

or BLWGN mode mixing, although values as high as 9.8 dB have been observed in very large data

samples.

As the Q of the chamber is reduced by loading it with rf absorber, the peak-to-average value in general decreases as shown in figure 3-26. As the Q decreases, the bandwidth of each resonant mode

increases. Eventually, several resonant modes overlap so the same excitation frequency will excite

many modes instead of one. This has no effect on the average value but does tend to reduce the peak

value. For example, if 10 modes are simultaneously stimulated, the maximum value can only be

achieved if the peaks from each resonant mode are coincident at the same spatial location. The lower

the Q, the greater the number of overlapping modes and the less likely the maximums will coincide at the same locations. However, if the effectiveness of the mode mixing can be increased, for example,

by increasing the noise bandwidth, then the typical peak-to-average ratios can be achieved even for

low Q cavities [3-14].

3.4 Power Density Enhancement Using Multiple Sources

An interesting feature of a reverberation chamber is the possibility of increasing the test fields inside the chamber by combining the output from rf sources. These sources may be operating at the same frequency or at different frequencies. If BLWGN stirring techniques are used, a single source can be used to drive multiple amplifiers with the same increase in test fields as if two incoherent sources were used. Block diagrams of the systems used to verify these possibilities are shown in

figures 3-27 and 3-28. Figure 3-27 is for two coherent sources and figure 3-28 is for two incoherent

sources. The results of the measurements are given in figures 3-29 and 3-30. Figures 3-29 and 3-30

show the power density, in dBm, measured inside the chamber using 10 MHz bandwidth filters with

the BLWGN system for coherent and incoherent sources, respectively. Figures 3-3(b) and 3-4(b)

show the difference between the power densities measured inside the chamber with a single source and with two sources, each having the same output as the single source, for the coherent and

incoherent cases. (The (b) figures were obtained from the (a) data.)

11
The above measurements were acquired from single locations of the transmitting and receiving

antennas and reflect the effects of field uniformity. Based on field uniformity measurements,

averaging of measurements from multiple locations should yield a flat 3 dB increase.

3.5 Time to Achieve Statistical Averaging

One of the major advantages of reverberation chamber testing is that an EUT is exposed to a

statistically isotropic and randomized field. As a result of the isotropic illumination, the EUT need

not be rotated to obtain all aspect angle susceptibility data.

However, the deterministic nature of the field as a function of tuner position for mechanical mode

mixing could present a problem for testing some EUTs. For a fixed mner position, the reverberation

chamber EME will be characterized by a series of nodes and hot spots. These field variations are averaged and the field becomes statistically isotropic over a complete rotation of the tuner. However

for some EUTs, particularly large EUTs, the response could be effected by the nonuniform

illumination at a given tuner position. The effect is most pronounced for mode-tuned operations.

In the mode-tuned procedure, the tuner is stepped successively through a number of discrete

positions. The tuner remains stationary at each step for an interval equal to or somewhat longer than

the EUT response time. Thus, in the mode-tuned procedure any EUT response effects due to nonuniform illumination would be most apparent.

In the mode-stirred procedure, the tuner is rotated continuously. In most cases, the rotation rate is slow enough that the dwell time of a field configuration is equal to or greater than the EUT response

time. Thus, for slow tuner rotation rates the EUT response should be the same as for the mode-tuned

procedure.

To avoid the problem of a series of nonuniform configurations some testers use very high rotation

rates in the mode-stirred procedure. The concept is to produce the field uniformity over a time

interval which is short compared to the EUT response time. For example, if the EUT response time

is 1 s, then one might try to establish field uniformity on a time scale of 0.1. This would require a

rotation rate of 10 rev/s. Some testers do, in fact, routinely operate at 1 to 5 rev/s [3-15]. However,

the high rotation rates necessary to cover a wide range of EUT response times could pose operational

problems for the chamber.

Since with BLWGN excitation the mode stirring is electronic, it seemed appropriate to investigate

the time scales necessary to establish field uniformity. This investigation requires the measurement

system to have a fast response time. For these measurements the scalar network analyzer was

replaced with a fast digitizing signal analyzer. The test set-up is shown in figure 3-31. A typical data

trace is shown in figure 3-32. The figure is a "snapshot" of the time dependence of the power density

as well as a running average of power density integrated over the noise BW. The issue is how long an interval (the EME refresh time) is required before the noise excitation randomizes the chamber

EME. The indication of randomization will be the recurrence of the expected peak power density as referenced to the average power density. Typical values of 7 to 8 dB are observed for mechanical

mode-mixing in reverberation chambers. We will use a peak-to-average ratio of 7 dB as the refresh

criterion.

To investigate the amplimde of the peak-to-average ratio for BLWGN and the average refresh

time, data were collected for an ensemble of 4026 runs. Averaging provides better statistical data.

12
However, the number of runs averaged, although a common value, was too large for proper

evaluation of the refresh time. Since the data collected were the peak within a particular sampling

time window, an event which exceeded any specified level (in this case, the 7 dB criterion) but which was less than the maximum value collected in that time window for any of the 4026 runs, would not be recorded. This saturation effect was demonstrated in one data set where all the measured values

exceeded the 7 dB criterion. Therefore, these data can only define an upper bound on the refresh

time.

Peak data at 1 GHz for 10 MHz noise BW are shown in figure 3-2. The average value of another

4026 runs completed under the same test conditions is also shown in the figure.

Data collection is initiated by the trigger signal at time zero. The settling time of about 2.5 /^s is apparent and is disregarded in the analysis.

The solid line in the figure is the 7 dB criterion for the peak power density. Using the data points

which exceed the 7 dB reference line, the single sweep time, and the number of runs averaged, we

can determine that the refresh time has an upper bound of less than 200 /xs. A similar analysis of 50

MHz BW noise data at 1 GHz, shown in figure 3-33, yielded an upper bound of about 150 ixs which

is the limit of resolution for the number of data runs averaged.

Analysis of a limited number of "snapshot" runs suggests that the refresh time may be less than

50 ^s. Further data should be collected and evaluated to obtain better resolution of the refresh time.

An BUT in a BLWGN-excited reverberation chamber will be effectively immersed in an isotropic

and randomly polarized field on time scales of the BLWGN refresh time. Therefore, if the BUT

response time is longer than the refresh time, it will not "see" the modal hot spots and nodes which

occur with most mechanical mixing operations.

4. EQUIPMENT UNDER TEST RESPONSES

To demonstrate the feasibility of using BLWGN excitation of a reverberation chamber for radiated

susceptibility and SB testing, several issues needed to be investigated. One issue was the ability of

the BLWGN excitation to excite frequency-dependent or resonant responses in a BUT. This included

the ability to excite responses from BUTs which have either narrow or broad susceptibility

bandwidths. Another issue was how both peak and average power sensitive BUTs would respond to BLWGN excitation. Also BUTs whose susceptibility had an aspect angle dependence needed to be

investigated. Finally shielding-effectiveness measurements with BLWGN excitation needed to be

compared to results obtained with mechanical tuning.

Frequency-dependent responses were primarily investigated using standard transverse electromagnetic (TBM) cells. The theory of TBM cells, a description of the experimental set-up, and

the results are discussed in section 4.1. A generic electronic system called the CTA (common test

article), developed by NSWCDD and extensively tested using several test techniques, was used for the

radiated susceptibility demonstration for peak power sensitive systems. A description of the CTA, the

experimental set-up, and the results are covered in section 4.2. To demonstrate the response of an

average power sensitive system, an instrumented 7 cm (2.75 in) folding-fin aircraft rocket (FFAR)

was tested. The BUT description, the experimental set-up, and the results from BLWGN excitation

and mechanical tuning in a reverberation chamber as well as anechoic chamber data are presented in

13
section 4.3. Section 4.4 describes the SE tests and compares the results obtained from BLWGN and

mechanical tuning in the reverberation chamber.

4.1 TEM Cells

4.1.1 Theory of TEM Cells

A cross section of a TEM cell is shown in figure 4-1. A circular aperture is centered in the upper

wall, and the cell is tapered down to a coaxial line at each end. The line is terminated in a 50 fi impedance matched load at one end, and the power is received in an impedance matched, calibrated,

power measurement detector at the other end.

Theory describing the aperture coupling of TEM cells is contained in refs. [16] and [17]. Briefly,

for aperture coupling, this includes defining shielding effectiveness (SE) as

SE = -10 Log — , dB,

A.

(4-1)

where Ae is the effective area of the apertured TEM cell (defined as the ratio of the received power P,

picked up by the TEM cell to the incident power density P^ to which the TEM cell is exposed, A^ =

Pf/Pd) and Af is the effective area of a reference antenna with known efficiency. The apertured TEM

cell does not permit a rigorous analysis for determining its effective area because the external

geometry is too complicated to compute the current density (tangential magnetic field) and charge

density (normal electric field) incident on the aperture.

An approximate analysis for circular apertures [16] has been performed assuming that the fields incident on the aperture are equal to the free-space incident fields. Polarizability theory described in

ref. [17] was used, along with a similar theory for excitation of the internal TEM mode by the

aperture dipole moments. No averaging over incidence angles was done, but an expression for the

maximum received power was derived. The maximum occurs when the incident field is propagating

along the z axis with the electric field polarized in the y direction. The maximum effective area of

the TEM cell is

e(niax)

Tl

16r^

X asinh(7rfe/2fl)

(4-2)

where the dimensions a and b are shown in figure 4-1. Since this theory deals only with the

polarization matched case, A^ is normalized by the effective area XMtt of an isotropic antenna to get

shielding effectiveness

SE = -10 Log (t:ZJx\)

32rX^asmhiTzb/la)

,dB. (4-3)

The theory for calculating the cavity resonances of TEM cells is given in ref. [18]. A summary

of the modal equations for calculating the frequencies for the first few higher order modes is given in

14
table I, page 126 of ref. [18]. Once the cutoff frequency of a particular mode has been determined,

the associated resonance frequencies f. can be calculated from

*(m.n.p)

where f^ is the cutoff frequency of the mode, L^ is the effective length of the cell for the particular

mode, and m,n, and p are integers.

The quality factor Q of the resonances can be determined from their +3 dB bandwidths. If Vf

equals distance between points on the response curve for which the amplitude response is down to IA/2 of its maximum value (corresponding to -3 dB), then Q is given as

Q = fj^f, (4-5)

where f^ is the resonant frequency. Af = Bandwidth (BW), so eq (4-5) equals eq (3-7).

Results of calculations using table 1 of ref. [18] and eq (4-4) for the 12 cm x 12 cm and 3 cm

X 6 cm apertured TEM cells are given in tables 4.1 and 4.2.

4.1.2 Experimental Results

Experimental results obtained using the NIST reverberating chamber with mechanical stirring for the two apertured TEM cells are shown in figures 4-2 and 4-3. The top curve on each figure is the

measured SE of the particular cell with its aperture open. The smooth curves are the theoretically

predicted SE calculated from eq (4-3). The lower curve on each figure is the measured SE with the

cell's aperture covered with conductive tape. These data were taken to be sure the significant

coupling was via the apertures. The sharp resonance spikes seen with the top and bottom measured

curves are due to cavity resonances that occur in the particular cell. These can be compared with the resonance frequencies shown in the appropriate tables, 4.1 and 4.2. The agreement between

measurement and theory is excellent for determining the resonant frequencies. The theoretical SE

curves, however, do not agree with the measured values as well, probably because of the effects of

the external geometry of the cells on the fields exciting their apertures. This seems apparent since the

agreement improves as the size of the cell decreases.

The + 3 dB bandwidth and quality factor of the measured resonances associated with the cells are

also shown in tables 4.1 and 4.2. These data are given for comparison with the results of

measurements made using the BLWGN technique to evaluate the cells as discussed in section 4.1.3.

4.1.3 Explanation of Results and Coupling Cross Sections

Coupling cross section is an important parameter that relates the power density incident on a system to the power delivered to a 50 Q instrument connected to a node of interest in the system

under investigation. Coupling cross section is also often called effective area A^ or effective aperture

and it typically has the units of square centimeters. Cross section is defined as [20]:

15
where a(f), Pl, Hv(f), Rl. and Sinc(f) are the coupling cross section, power delivered to measuring

instrument, voltage transfer function (voltage impressed at test point divided by the magnitude of the

incident E field), input resistance of measuring instrument, and power density incident on system.

The first observation that can be made about eq (4-6) is that it assumes that the test object is being

radiated with a plane wave. Secondly, it also assumes that the source of radiation is at a single

frequency. Neither of these assumptions is correct in the case of BLWGN excitation. A B-dot probe

or other polarized electromagnetic sensor inside a cavity with an isotropically impinging field will on

the average measure only half of the power density associated with the field due to its polarization.

Further, any directivity associated with the sensor will become 1. In general this can also be said of

a node inside the system being investigated [20]. Thus, the coupling cross section measured in a reverberation chamber will be reduced by both its directivity and polarization factor associated with

the system regardless of how the modes are stirred. Although the polarization factor is easily

corrected for, the directivity is much more difficult. For simple systems such as long slender rockets,

the maximum directivity can be approximated from simple dipole models. If the maximum directivity

at a given frequency caimot be modeled or measured in some other fashion, the coupling cross section

as measured in a reverberation chamber should be considered as an average cross section over 4x sr.

Next, we consider the effect of spreading the spectrum of the excitation source. Because the

measuring instrument effectively measures the total average power being received by the node over

the agility bandwidth, measuring the cross section using this method is equivalent to convoluting a

gating function g with the narrow band cross section and then dividing by the power density, i.e.,

f

g(f) o(f'-f) df

otf) = ^-^^^ . (4-7)

f

8(f) df

fc-BWI2

This has the same effect as mathematically smoothing a(f) over the same bandwidth. The primary

difference is, however, that the smoothing is done in real time and is independent of the sampling

resolution. In other words, if the sample interval were greater than the smoothing window,

mathematical smoothing would not work, this method would, however.

As shown in figures 4-4 and 4-5, details in the cross section that span frequency intervals less than the agility bandwidth will tend to be washed out, while those that span greater intervals will

remain. In fact, when the bandwidth of the resonance is less than the excitation bandwidth, the resonance will be reduced by the ratio of the resonance to the source bandwidth. This can be further seen in figures 4-6 and 4-7, which show the coupling cross section of the NIST standard 12 cm and

6 cm TEM cells, respectively [16]. Each cell has a small aperture through which energy couples.

The cross section of each cell was measured with the BLWGN excitation technique using four

16
different agility bandwidths (2, 10, 20, and 50 MHz). Note the resonance in figure 4-6 at approximately 0.925 GHz. This resonance is very narrow (on the order of 1 MHz). As the agility

bandwidth is widened, the resonance appears to widen. In this case, because the cross-section

bandwidth BW^ is much less than the agility bandwidth BWg the resonance appears to have Hx

bandwidth equal to the agility bandwidth, while its amplitude is decreased by the same amount the

agility bandwidth is increased. The 6 cm TEM cell (fig. 4-7), on the other hand, has a resonance at approximately 6.45 GHz with a bandwidth of around 100 MHz. Here, because all four agility

bandwidths are less than the resonance bandwidth, the cross section as measured by each agility

bandwidth is essentially the same and accurately reflects the true cross section less the polarization

factor and directivity.

A comparison of figures 4-6 and 4-7 with figures 4-2 and 4-3 shows a match in the measured

resonances for both cells between BLWGN and mechanical stirring excitation data, however, with some limitations. First, the BLWGN technique demonstrated the capability to find resonant modes

with bandwidths much smaller than the BLWGN agility bandwidth although their amplitude is reduced. In figure 4-6, all the noise bandwidths located the narrow resonance identified in table 4.1

at 929 MHz and shown in figure 4-2 for mechanical mode mixing. The dependence of the resonance

amplitude on noise bandwidth is also evident in figure 4-6. Thus, assuming the measurement system

has sufficient dynamic range to compensate for the decrease in response amplitude with bandwidth, an BUT resonance can be found for a wide range of bandwidths. Second, when the cross-section resonant bandwidth is greater than that of the BLWGN agility bandwidth, the spatial average cross

section can be accurately measured. This can be seen in a comparison of the BLWGN at 6.4 GHz in figure 4-7 with the mechanical mode-mixed data in figure 4-3 . For this case there is excellent

agreement between both the frequency and amplitude of the resonance between mechanical and

BLWGN excitation.

4.2 Electromagnetic Vulnerability Testing

4.2.1 Test Item Description

The NSWCDD developed a simple electronic system to evaluate various susceptibility test techniques for repeatability and comparability. The system, referred to as the common test article (CTA), consists of three shielded boxes whose functions and interconnects are shown in figure 4-1. Boxes 1 and 2, which are battery powered and connected together via a combination of fiber optic

cables and unshielded copper wires, are intended to be exposed to various electromagnetic

environmental effects (E^). Box 3, which is connected to boxes 1 and 2 via fiber optic cables and is ac powered, is used to monitor the "health" of the system and remotely test and/or reset the system.

Box 1 consists of a 100 kHz clock, a comparator circuit to compare two, 8-bit digital words, an error detection circuit which trips when the comparator circuit detects a mismatch, and a reset circuit that resets the counters and error detection circuit whenever an upset or test function occurs. Box 2

consists of two, 8-bit counters which simultaneously count the clock signal sent from box 1 and a test circuit which grounds one of the output lines of one counter, which causes the two counts to fail to match whenever the test function is activated. Box 3 consists of two fiber optic transmitters and one

fiber optic receiver. The fiber optic receiver circuit monitors the system health and activates a green

LED on the front of the box to indicate proper operation. If an upset occurs then a red LED is activated. The two fiber optic transmitter circuits send signals to the reset and test circuits whenever

the corresponding button of the front of the box is pushed. Box 3 is also equipped with TTL

compatible I/O to enable the system to be monitored, tested, and reset under computer control.

17
Table 4.1. Mode cutoff and resonant frequency analysis of 12 cm x 12 cm TEM cell.

Mode fc(m,n) Lmn fr(m,n,l)

(MHz) (m) Freq.

(MHz)

BW

(MHz)

Q

TEOl 728 0.26 929 1.2 761

TEIO 1250 0.18 1502 3.2 470

TEll 1563 0.16 1823 3.0 606

Table 4.2. Mode cutoff and resonant frequency analysis of 3 cm x 6 cm TEM cell.

Mode fc(m,n) Lmn fr(m,n,l)

(MHz) (m) Freq.

(MHz)

BW

(MHz)

Q

TEOl 3833 0.092 3269 5.9 556

TEIO 2679 0.080 3270 4.3 761

TEll 4951 0.068 5420 3.0 1813

TE20 5357 0.080 5676 2.0 2835

TE21 6207 0.080 6484 2.0 3230

Before this test, the susceptibility of the CTA was determined using CW excitation in the

NSWCDD reverberation chamber. The results of the NSWCDD testing are shown in figure 4-9.

The susceptibility of the CTA is low at 200 MHz, and the susceptibility level increases with

frequency until about 2 GHz. Above 2 GHz, the CTA can withstand power densities above the

arbitrarily unposed test limit of 10.61 mW/cm^ which is equivalent to a field strength of 200 V/m.

Figure 4-10 shows the CTA set-up in a reverberation chamber with 2 m of cable between boxes 1 and 2. Previous testing of the prototype CTA have shown that variations in box position, orientation,

and cable runs do not significantly affect the susceptibility test results obtained using a reverberation

chamber.

All data signals, which pass from box 1 to box 2 by unshielded copper wires, pass through

74LS373 line drivers. These line drivers have a data set-up time of 5 ns which allows them to respond to inputs of very short duration.

4.2.2 Experiment and Results

The CTA was placed inside the NIST reverberation chamber on a styrofoam block as shown in

figure 4-10. The measurement system was configured as shown in figure 4-11 for BLWGN excitation

18
and as shown in figure 4-12 for CW excitation with mechanical stirring. The testing conducted

determined the power density at which system upset would occur for both CW excitation and for

noise excitation utilizing five different bandwidths of noise.

Since the previous susceptibility testing revealed that the susceptibility of the CTA increased with

frequency, attempting to sweep would have resulted in upsets occurring at different power densities over the frequency sweep. This would result in having to "track" the vulnerability by altering both

start frequency and power density. To eliminate this problem, testing was conducted at discrete

frequencies for both methods of excitation. Testing was conducted at 19 frequencies over the

frequency range from 100 MHz to 2 GHz and was limited to power densities not to exceed 10.61

mW/cm^. At each of the discrete frequencies tested, the power density was initially set below the

expected upset level, which was based on the previous CW testing conducted at NSWCDD. Once the

initial power density was established, the power was increased in steps until an upset occurred or the

power density limit was reached. For each failure the chamber input power was recorded and the

average power density measured by each of the four probes (see section 3.1) shown in figures 4-11

and 4-12. A scalar network analyzer was used to measure the probe responses because its broad

frequency response allowed it to respond accurately to both CW and narrow band noise stimulus. For

CW excitation, the peak power density was also measured using a spectrum analyzer and a receiving

antenna.

Before and after each test was conducted, the CTA was cycled by activating the test circuit and

then executing a reset. This ensured that the CTA was still functioning properly and had not been

damaged during the test.

4.2.2.1 Band-Limited, White Gaussian Noise Excitation

Testing began at 100 MHz and proceeded upward in frequency. Once the initial power level was

established, the power level was increased in 0.5 dB steps, with a 1 s dwell time at each step, until an

upset occurred or the test limit was reached. At each failure the chamber input power was recorded

and the average power density was measured by each of the four probes shown in figure 4-11. Only

average power density was measured for the BLWGN measurements due to equipment limitations.

Data were collected for noise bandwidths of 2 MHz, 4 MHz, 10 MHz, 20 MHz, and 50 MHz.

The average upset power density as measured by the four probes for each of the five bandwidths

tested is shown in figure 4-13. The data show that the upset level varied as much as ±4 dB over the

five bandwidths tested, with the average variation about ±2 dB. These variations are a sum of the

input power variations caused by slight frequency dependent variations in the attenuations of the

filters, cables, and couplers used and the field uniformity of the chamber for the various bandwidths.

The largest variation was found to be at 100 MHz, while the smallest variations occurred at 500 MHz

and 1.4 GHz. At 1.8 GHz an upset occurred using only the 20 MHz bandwidth. The upset occurred

at a power density of about 5 mW/cm^ which is 3 dB below the maximum permitted power density.

Based on the variation of the preceding data points it is possible that the failure levels for the other

bandwidths could have been slightly above the maximum permitted power density without having

variations greater than those already experienced. The data also show that the susceptibility levels were randomly distributed among the five bandwidths tested. This indicates no systematic bandwidth

bias in the susceptibility results.

19
4.2.2.2 Continuous Wave Excitation Using Mechanical Stirring

To check the resuhs obtained using BLWGN excitation, data were collected using CW excitation

and mechanical stirring at the same center frequencies used to collect BLWGN data. To collect the

data, the test set-up was reconfigured as shown in figure 4-12. Data were collected by exciting the

chamber while rotating the mechanical tuner 360° over a 10 s period. The rotation time was selected

to ensure that the time rate of change of the fields was slower than or equal to the response time of

the spectrum analyzer used to measure the peak value of the fields. This also ensured that the

effective dwell time afforded by the tuner rotation rate would be greater than the response time of the

CTA. The power density was increased in 0.5 dB steps after each rotation of the mechanical tuner

until an upset occurred or the 10.61 mW/cm^ limit was reached. Initial power densities values were

determined by subtracting 5 dB from the upset levels found using the BLWGN technique. When an

upset occurred, the chamber input power was recorded, the peak power density was measured using a spectrum analyzer, and the average power density was measured using the four probes shown in

figure 4-12. Each of the three readings was taken with the mechanical tuner in motion over a period

corresponding to at least one complete rotation of mechanical tuner.

The peak and average power densities at which upset of the CTA occurred are shown in figure 4-

14. The peak upset levels are consistent with the upset levels shown in figure 4-2, which were

previously obtained using the NSWCDD reverberation chamber. The difference between the peak

and average data is shown in figure 4-8. The difference between the peak and average upset levels varies from a high of 12.4 dB to a low of 3 dB. The largest differences occur at the lower

frequencies as should be expected since chamber uniformity decreases with frequency. The average

difference is 7.7 dB, which is consistent with the 7 to 8 dB variations previously discussed and with

findings presented in refs. [6,19,21,22,23].

4.2.2.3 Comparison of Noise Data to Continuous Wave Data

Figure 4-16 combines the average of the BLWGN data shown in figure 4-13 with the CW data

from figure 4-14. The data show an excellent correlation between the BLWGN upset levels and the

average upset levels obtained using CW excitation and mechanical stirring. The difference between

the CW average and the average BLWGN responses is shown in figure 4-19. The data indicate that the average BLWGN upset level was generally slightly higher than for the CW average. However

two-thirds of the data points are within ±3 dB and only one data point above 500 MHz exceeded 3

dB. These variations are within the expected chamber field uniformity as outlined in section 3.2.

The peak power upset level obtained using mechanical stirring is approximately 7 dB higher than

the average power upset level using either mechanical stirring or the BLWGN. Since no peak data were collected for the BLWGN excitation, there are no data available for a direct comparison.

However, as was discussed in section 3.3, the peak-to-average ratio for both BLWGN and mechanical

mode mixing is typically 7 to 8 dB. The difference between the peak of mechanically mode-mixed

data and the average BLWGN data is shown in figure 4-18. The difference between the peak

mechanical and average noise data varies from a high of 10.4 dB to a low of 3 dB. The largest

differences occur at the lower frequencies as should be expected since chamber uniformity decreases

with frequency. The average difference is 6.5 dB, which is consistent with the 7 to 8 dB variations

previously discussed. The data are consistent with data presented in section 4.2.2.2, which were

obtained using mechanical tuning alone. This indicates that the peak levels generated by the BLWGN

are equivalent to the peak levels generated using mechanical tuning and that the CTA responded to the

peak levels generated by the BLWGN.

20
Testing was conducted for 1.8 GHz using CW excitation, however no upset occurred. As stated

in 4.2.2.1, an upset occurred when the chamber was excited with 20 MHz noise bandwidth and no

upset occurred at the other four bandwidths tested. Since the CW excitation did not cause upset at the

10.61 mW/cm^ limit, the possibility that the upset level is within the variations previously discussed

cannot be ruled out.

In summary, the fields generated by both methods have the same peak-to-average ratio and the

two methods of excitation yield the same results when the data are properly interpreted. For this test the BLWGN excitation method was a factor of 10 faster than the mechanically tuned method.

However, since the response time of the CTA was much shorter than the dwell times used for either

excitation, and since the speed at which the fields can be changed is greater for the BLWGN than for

mechanical mode mixing, it is likely that BLWGN excitation could be significantly faster than

mechanically tuning when used for EMV testing.

4.2.2.4 Comparison of Chamber Input Power Required for Upset

Another way to determine the variation in the upset levels of the CTA is to examine the variation

in chamber input power at which upset occurred for each excitation used. To accomplish this the

average of the input power(s) at which upset occurred for each test frequency and bandwidth was

calculated. The difference between the average input power and each individual input power was

calculated. The results of these calculations are shown in figure 4-17. The data show that 97 percent

of the data points fall within +3 dB and that 100 percent of the data fall within ±4 dB. The data

also show no systematic bias due to either excitation method or bandwidth variations in BLWGN.

The data did not converge or diverge with frequency indicating that the CTA responded the same to both excitation methods.

4.3 Hazards of Electromagnetic Radiation to Ordnance Testing

4.3.1 Test Item Description

The item tested, shown schematically in figure 4-1, was a modified, 7 cm (2.75 in) folding-fin

aircraft rocket (FFAR) which consisted of a MK 66 MOD 1 rocket motor, MK 1 MOD electric

squib, and a warhead/fuse assembly. The electric squib is located in the forward end of the motor

section and is enclosed in a thin-walled steel housing. Its function is to ignite the rocket motor

propellent. The squib firing leads exit the igniter housing through a glass-insulating terminal. One

firing lead connects to a wire that passes through the grain perforation, nozzle throat, and weather

seal to connect with the contact band on the nozzle/fin assembly, while the other firing lead attaches

to a spring assembly that grounds the lead to the motor case. The standard FFAR was modified for ease of instrumentation and safety as follows:

a. Rocket Motor—The MK 66 rocket motor propellant was removed and replaced with a spacer

to maintain firing lead position. All other hardware was the same as the tactical rocket motor.

b. Electric Squib—The MK 1 MOD electric squib was replaced by a MK 17 MOD electric

squib because of its known electrical/firing characteristics.

c. Warhead/Fuse—The warhead/fuse assembly was removed and replaced with a metal end cap

which simulated the warhead/fuse attachment.

21
d. Firing Lead—The firing lead was disconnected from the contact band and extended out of the

nozzle. This increased the EM coupling to the rocket, which allowed all reverberation chamber

testing to be conducted using only 1 W of input power to the chamber.

4.3.2 Instrumentation

For this test the MK 17 MOD squib, which will be referred to as an FED, was monitored using

a fiber optic multisensor system. The system, shown in figure 4-21, consisted of a measurement

instrument, temperature sensor probes, fiber optic cables, and personality keys. The instrumented

FED is shown in figure 4-22.

The measurement instrument may be viewed as a light transmitter and light receiver. The

instrument sends light out four sensor ports simultaneously. The light propagates down an optical

fiber until it reaches the sensor probe assembly's sensing element. The color of the transmitted light

is altered by the sensing element as a function of the parameters it was designed to sense and reflected

back to the instrument. The change in color (spectral change) of the light is measured by the

instrument as it monitors the reflection from the sensor. The amount of color change is then

compared to a calibration table stored in a memory device called a personality key. The resultant

measurement information is presented on the front panel and is provided to an analog output on the

rear of the instrument.

The measurement system has several advantages when used for monitoring FEDs which utilize hot bridgewires such as the one used in this test. These advantages include:

a. The measurement system is a self-contained unit which can be easily transported and operated.

b. The fiber optic cables can be of lengths approaching 2 km and can be used in most

applications. The optical cables do not perturb the electromagnetic field nor does the electromagnetic

field have an effect on the fiber cables.

c. The sensor probe assembly and sensing element are non-metallic and small compared to the

FED. Therefore, the sensing element does not affect the energy induced into the EFD's bridgewire.

In addition, the sensing element is small enough that it may be placed in close proximity to or

actually touching the bridgewire as shown in figure 4-22.

d. The measurement system can measure currents as low as 10 mA which is approximately

40 dB below the maximum no fire current (MNFC) of most FEDs.

e. The instrumentation's dynamic range is limited by the FED itself and the maximum repeated

current allowed before damage will occur to the bridgewire of the FED. In this application, the

dynamic range was approximately 34 dB.

f. The measurement system response time varies between 25 and 90 ms, which is slightly longer

than the typical FED which has a thermal time constant of approximately 12 to 15 ms.

22
4.3.3 Experiment and Results

The FFAR was placed inside the NIST reverberation chamber on a styrofoam block and

connected to the fiber optic measurement system as shown in figure 4-23. The testing conducted

measured the amount of current induced in the bridgewire of the rocket's BED for approximately 1 W

of rf energy injected into the chamber. Data were collected using the reverberation chamber for two

types of excitation: (1) BLWGN, and (2) CW with mechanical tuning. Data were also collected using

the NIST anechoic chamber with CW illumination for the purpose of comparing the reverberation

chamber data to anechoic chamber data.

Before and after the test, the FFAR was calibrated by injecting a known direct current (dc) signal

into the bridgewire leads and monitoring the output signal. This ensured that the characteristics of the

EED did not change during the course of the test.

4.3.3.1 Band-Limited, White Gaussian Noise Excitation

Testing began with swept BLWGN excitation over the frequency ranges listed in table 4.3, using

the test set-up shown in figure 4-24. Sweep time for each frequency range was 200 s. This ensured

that the EED would have time to respond to any current induced in the bridgewire. Testing was

conducted using 2 MHz, 4 MHz, 10 MHz, 20 MHz, and 50 MHz noise bandwidths. The data

collected are shown in figures 4-25 through 4-28. The data show multiple narrow, well defined

resonances. The amplitudes of the responses vary greatly with bandwidth at frequencies below 600

MHz. At frequencies above 600 MHz the amplitude variations caused by changing the bandwidth

decrease as frequency increases. This indicates that the bandwidths of the resonances at the lower

frequencies, such as at 450 MHz, are narrower than the noise bandwidths. However, the bandwidths

of the resonances at the higher frequencies, above 2.1 GHz, are increasing with frequency

approaching the width of the noise bandwidths. These bandwidth effects are consistent with the

theory outlined in chapter 3 and section 4.1.

The data also show a resonance at approximately 150 MHz, which is below the chamber's lowest

recommended usable frequency. At this frequency, the chamber has only 28 modes, considerably less than the 60 or more modes generally accepted [6]. This suggests the possibility that the lower

frequency limit of a reverberation chamber may be extended using BLWGN excitation as opposed to CW excitation with mechanical tuning for determining resonance characteristics of the device under

test.

4.3.3.2 Continuous Wave Excitation Using Mechanical Tuning

To check the results obtained with BLWGN excitation, data were collected over a limited

frequency range using CW excitation with mechanical stirring. The frequency range investigated was

from 500 MHz to 650 MHz and was selected based on the BLWGN data. The test set-up was

reconfigured as shown in figure 4-29. Data were collected by sweeping over the frequency range at each of 201 discrete tuner positions required to rotate the tuner through 360° of rotation. Sweep time was maintained at 200 s per sweep. After all 201 sweeps had been collected and saved on disk, the

data were analyzed to determine the maximum and average responses of the rocket. These responses

are shown in figure 4-30. The difference between the maximum and average data is approximately 7

dB. This is consistent with data shown in ref. [6] and earlier in this report.

23
Table 4.3. Frequencies ranges for swept BLWGN excitation testing of folding-fin aircraft rocket.

Frequency (GHz)

0.05 - 0.5

0.5 1

1 2

2 4

4.3.3.3 Comparison of Noise Data to Continuous Wave Data

The average data obtained using CW excitation with mechanical tuning and 2 MHz BLWGN are compared in figure 4-3 1 . The data show excellent correlation. Again, the maximum response

obtained using the mechanical method is approximately 7 dB higher than the BLWGN response. This

is as expected assuming the BED response time is less than the test field exposure dwell time. (The

EED's response time is approximately 15 ms). The mechanical stirring method causes frequency

dependent maximums to exist at various locations through the test volume of the chamber as a function of tuner position. Each of these maximums exists for a period of time consistent with the

dwell time afforded by the frequency sweep time. If this is greater than the EED's response time, the

EED will respond to the maximum field. The maximum fields generated by BLWGN excitation exist

for time intervals measured in microseconds, as discussed in section 3.3. Thus, the EED responds to the BLWGN averaged field present within the chamber.

In summary, the two methods of excitation yielded the same results when the data are properly

interpreted. The BLWGN excitation method was on the order of 200 times faster than the

mechanically tuned method. However, for a given chamber at a given input power, the mechanically

tuned CW excitation method will effectively deliver approximately 7 dB more power to an average

power sensitive device than can be obtained using BLWGN.

4.3.3.4 Anechoic Chamber Data

To complete the characterization in an anechoic chamber of the 2.75-in rocket, an evaluation was

conducted over the same 500 MHz to 650 MHz frequency range. The rocket was mounted in the

NIST anechoic chamber as shown in figures 4-32. The block diagram of the test system is shown in

figure 4-33. Initially, discrete frequency data were collected at four frequencies between 500 MHz

and 650 MHz. Two frequencies were selected from the first peak, one frequency from between the

peaks (a low point) and one frequency from the second peak. Data were collected at each of the four

frequencies using horizontal polarization at five roll angles, between and 180°, as the rocket was

rotated 360° in azimuth. Data were also collected at two frequencies corresponding to the "peaks"

using vertical polarization at roll angles of 0, 45, and 90°. The power density in the anechoic

chamber was set the same as the average power density obtained in the reverberation chamber using 1 W of input power. These data are shown in figures 4-34 through 4-38. As would be expected, the

data show that minimal response was obtained when using vertical polarization and that maximum

response was obtained using horizontal polarization. Based on the limited number of aspect angles

evaluated, 0° roll at an azimuth angle of 65° was determined to be the maxunum response

orientation. The measurement system was reconfigured as shown in figure 4-39, to collect swept

frequency data at the maximum response orientation. Again, the power density in the anechoic

24
chamber was the same as the average power density obtained in the reverberation chamber. The

response obtained is shown in figure 4-40.

4.3.3.5 Comparison to Anechoic Chamber Data

The anechoic chamber data verify the presence of the two resonances found in the reverberation

chamber between 500 and 650 MHz. A comparison of the anechoic chamber response to the 2 MHz

bandwidth noise response is shown in figure 4-41. As expected, the anechoic chamber response was

higher than the reverberation chamber response [6] . If we apply a polarizability factor of 2 to the

rocket's response data, as discussed in section 3.1, and correct for the reverberation chamber data

accordingly, as shown in figure 4-42, we see that the response obtained from the reverberation

chamber is still less than the anechoic chamber response. The remaining difference is attributed to the directivity of the device [6]. In this case, the difference was about 1.8 dB, or roughly equivalent

to a dipole antenna which has a directivity of 2 dB. As previously described, the configuration of the

firing lead was approximately a short dipole. Thus, this example demonstrates the loss of directivity

in a reverberating chamber and illustrates that if the directivity of a test item is known or can be

estimated, the responses obtained using the two types of facilities can be correlated.

An anechoic chamber test is the best simulation for an item immersed in a plane wave, free field environment, while a reverberation chamber best represents an item immersed in a complex

environment such as that encountered in an enclosed cavity.

Also, the power necessary to generate the required power density in the anechoic chamber was

approximately 10 W or 10 dB more than for the reverberation chamber.

4.4 Shielding Effectiveness Testing of Materials

4.4.1 Shielding Theory

Shielding effectiveness measurements using reverberation chambers are typically performed using

nested chambers, that is, a reverberation chamber within a reverberation chamber. The technique is shown in figure 4-43.

Two approaches have been used for SE measurements. In one approach, the ratio of the received

power in the fixture to the received power in the large chamber yields the SE. In the second

approach, the ratio of the received power in the fixture with the sample removed to the received

power in the fixture with the sample in place yields the SE.

Several factors that depend on the test fixture itself could influence the measured SE values.

These factors include the fixture size and Q, aperture effects, and the sample mounting techniques.

Work is in progress at NIST, NSWCDD, and PL to address these issues both theoretically and

experimentally.

To investigate the applicability of the BLWGN technique to SE measurements, a direct

comparison was performed to the SE results obtained with mechanical mode mixing for a wire mesh

window assembly.

25
4.4.2 Test Item Description

SE measurements were conducted on a 17.8 cm x 10.2 cm shielded window shown in figure 4-

44. Shielding was provided by a fme wire mesh embedded in the window which was terminated to a wire mesh gasket. The window was mounted over a 25.4 cm x 15.2 cm opening in the top of a 45.7

cm X 45.7 cm x 45.7 cm aluminum box test fixture shown in figure 4-45. The test fixture is sectioned into two compartments to provide isolation between the receiving antenna, which was

mounted in the larger upper compartment, and a dc motor, which was mounted in the lower

compartment. The drive shaft of the dc motor protrudes into the upper compartment where it connects to the paddle wheel tuner.

Since the shielded window is smaller than the aperture in the top of the test fixture, an adaptor

plate was used to mount the window to the fixture. The window was attached to the adaptor plate

using fourteen 6-32 UNC screws as shown in figure 4-46 and remained attached to the adapter plate

for the duration of the test. This was done to prevent mounting repeatability problems from affecting

the test results. (The actual SE measured was that of the window assembly and not necessarily that of

the window itself. References to the SE of shielded window refer to the SE of the entire assembly.)

4.4.3 Experiment and Results

The test fixture was placed in the NIST reverberation chamber on a styrofoam block as shown in

figure 4-47. The measurement system block diagrams are shown in figures 4-48 and 4-49 for CW

excitation (mechanical stirring) and for BLWGN excitation. The tests required measuring the power

density inside both reverberation chambers (the NIST chamber and the test fixture) with and without

the shielded window installed. The tuners were stationary for the BLWGN measurements. The data

collection process consisted of three separate measurements. They were:

1Baseline Data: These were data collected with essentially no shielding of the receiving

antenna present. For this test the shielded window was removed from the test fixture.

2. Shielding Data: These were data collected with the shielded window mounted to the test fixture. The difference between these data and the baseline data is the SE.

3. Dynamic Range Data: These were data with the shielded window covered with aluminum foil and aluminum tape. The difference between these data and the baseline data should be greater than

the difference between the shielding data and the baseline data. This ensured that the detected signal

entered through the shielded window or its associated gasket, and not through leakage through the test fixture or associated connectors, cabling, etc.

4.4.3.1 Band-Limited, White Gaussian Noise Excitation

Testing began with collection of data at 2 MHz, 10 MHz, and 50 MHz noise bandwidths over the

frequency range of 1.25 GHz to 6 GHz to determine what effect various bandwidths of noise would

have on shielding effectiveness measurements. For completeness, one data run to 18 GHz was

performed using 10 MHz noise bandwidth. The data shown in figures 4-50 through 4-52 reveal that the shielding effectiveness smooths out as noise bandwidth increases. This "smoothing" is due to averaging the shielding over the bandwidth and the improving field uniformity with increasing

bandwidth. (The contribution attributed to each effect is an area for further study.) Q The sharp

decrease in shielding at approximately 3 GHz was detected by all three bandwidths, but is most

26
pronounced in the 10 MHz bandwidth data. These data give the best comparison with data obtained

previously for the shielded window [24]. Unfortunately, the mating surface of the adaptor plate had

not been properly prepared for the BLWGN measurements which resulted in less shielding being

provided by the window. For this reason, these data should be used only to examine bandwidth

effects and should not be compared in an absolute sense to other data presented for the shielded

window.

Measurements with 10 MHz bandwidth BLWGN were repeated after properly preparing the

mating surface of the adaptor plate and reattaching the shielded window.

Baseline data for the measurements were collected from both the large chamber and from the test fixture. As stated in section 4.4.1, SE measurements can be made using baseline data obtained from

either the large chamber or from inside the test fixture. The results of the baseline measurements are shown in figure 4-53. These data show an offset of approximately 4 dB between the received powers

of the chamber and the test fixture with the aperture open. This is due to the differences in cable

attenuations and the efficiencies of the two chambers' receiving antennas. This indicates that if the

shielding effectiveness were determined by taking the difference between the chamber antenna and the

test fixture antenna when the shielded window was installed, a correction factor of approximately 4

dB would need to be added to the shielding data to correct for the differences between the two

receiving systems. The data also indicate that the data collected by the test fixture antenna were not

affected by waveguide below cutoff effects caused by the aperture in the test fixture.

SE data for the window were obtained by measuring the difference between the power received

by the test fixture anteima with and without the window mounted. The results of these measurements

corrected for the approximately 4 dB difference previously mentioned are shown in figure 4-54.

These data agree well (+2 dB) with previously obtained results.

Dynamic range data were collected by covering the shielded window with aluminum foil and

taping the edges to the adaptor plate using aluminum tape. This yielded a minimum of 10 dB

reduction in received power across the frequency range tested indicating that the shielding measured was due to coupling through the window not leakage through cables and other ports of entry.

The average data collection time using the BLWGN method was about 1 min per sweep.

4.4.3.2 Continuous Wave Excitation Using Mechanical Tuning

To check the results obtained using BLWGN excitation, data were collected using CW excitation

and mechanical tuning over the 2 to 18 GHz frequency range. To collect the data, the test set-up was

reconfigured as shown in figure 4-48. The rotation rate of the test fixture tuner was set to 1 rev/s

and the rotation rate of the chamber tuner was set to 6 rev/min. The spectrum analyzer was set to sweep the 2 to 18 GHz frequency range over a 60 ms time and placed in a "max-hold" mode while

the rf synthesizer was set to continuously sweep over the 2 to 18 GHz frequency range at the rate of

200 s per sweep.

Baseline data were collected by removing the window from the test fixture and collecting data for a period of approximately one hour.

Shielding data were collected by installing the shielded window/adaptor plate to the test fixture

and repeating the data collection process for a period of approximately one hour.

27
The difference between the baseline data and the shielding data yields the SE of the window

shown in figure 4-54.

4.4.3.3 Comparison of BLWGN Data to Continuous Wave Data

The difference between the CW and BLWGN data is shown in figure 4-55. The data indicate

reasonable agreement for the SE measured by both techniques. The difference between the CW data

(smoothing factor 5 percent) and the BLWGN data is shown in figure 4-56. Although the maximum

difference in the data shown in figure 4-55 is about 20 dB, the smoothed data in figure 4-56 show

variations on the order of ±2 dB. The data also show an offset of about 2 dB, indicating that the

BLWGN yielded slightly higher SE than did the CW excitation with mechanical mixing. This offset could be caused by several factors and should be studied further. In addition, the smoothing of the

CW data assumes that the majority of the structure in the CW data is due to variations in field uniformity. Determining whether smoothing should be applied, as well as how much, is also a

subject for further study.

In summary, SE data obtained using BLWGN or CW excitation with mechanical mode mixing

yield comparable results. The notable difference is the "smoothing" effect of the BLWGN technique.

The time required to collect SE data using the BLWGN technique however, is about one one- hundredth of that required for CW excitation using mechanical tuning. With the two-anterma method,

differences in the efficiency of the receiving antennas and cable losses used in the test fixture and

outer chamber must be properly corrected for.

5. SPECIAL CONSIDERATIONS ASSOCIATED WITH TESTING WITH BLWGN

EXCITATION

5.1 Instrumentation and Probes

The EME associated with mechanically stirred or BLWGN excited reverberating chambers are

quite different. Time constants associated with variations in the test fields are determined by either

the rotation rates and hence mechanical redistribution of the fields stirred by tuners, or the modulation

rates of the BLWGN system. In each technique, the fill or relaxation time of the reverberation

chamber cavity can also affect the test field EME. Some types of instrumentation are capable of

responding to very short time constants, for example probes such as B-dot or D-dot, and diode

detectors/sensors. However, some types of probes, for example, electrically short dipoles with high

resistance lines and distributed filters, have relatively slow response times (milliseconds). Also, some

instrumentation has limited resolution bandwidth detection systems, for example, spectrum analyzers,

and hence, cannot be used with BLWGN bandwidths that are significantly greater. This is the reason

scalar analyzers were used with the BLWGN excitation measurement system. The need to understand

and determine the impact of these measurement limitations was the motivation for the first phase of

this study. The results of that study verified that only calibrated rf probes such as the B-dot, D-dot,

or log periodic or horn antennas should be used to measure the test field for the BLWGN excitation

technique. Essentially all types of good quality, calibratable probes can be used to measure the test field using mechanical stirring. The study also verified that spectrum analyzers with sufficient

resolution bandwidths (bandwidths greater that the BLWGN excitation bandwidths) could be used, for

example, a 3 MHz resolution bandwidth for BLWGN bandwidths less than 2 MHz. Otherwise,

spectrum analyzers should not be used.

28
The limitations discussed above raise the question, "How will the EUT respond in the EME of the

two techniques?" If the EUT's response is very fast (microseconds or less) it will track the peak of

the exposure EME while the BLWGN excitations instrumentation will measure average values.

Hence, this effect must be accounted for in evaluating EUT responses.

5.2 Mixers

Since microwave mixers are fundamental to the generation of BLWGN excitation, consideration

must be given to the way they are used. Improper use of mixers will give erroneous results which

are difficult to distinguish from real EUT responses. The mixers used in this experimental series were a special type called up-converters or balanced modulators. They were selected for their good

carrier suppression, low VSWR, and low conversion loss. Practical considerations for using these

mixers include input dynamic range, intermediate frequency (IF) to local oscillator (LO) isolation, and

IF contained on rf output.

It is desirable to vary the level of the BLWGN excitation signal over a large range so that EUTs

with varying degrees of upset sensitivity can be tested. The test configuration described earlier for

generating BLWGN excitation relies on the variability of the synthesized sweeper to accomplish this. In other words, the LO input to the modulator is varied to create corresponding variation in output

power level. Practical modulators, however, rely on the nonlinear properties of diodes to accomplish

frequency conversion. The diodes must be operated in a region defined at their low sensitivity end by

the noise level of the diode and at their high output end where they respond in a linear fashion.

Hence, the synthesized sweeper must maintain an input level to the modulator that does not vary

outside this range. The modulators used in this series were designed to operate best with an IF input

of + 10 dBm and a LO input of typically -5 dBm. The IF input from the noise source was held

constant by setting the noise source's internal attenuation using the following equation:

A = 20-10 LOG ^^ , (5.2-1)

100x10^

This equation was determined by noting that the full power output of the noise source is 30 dBm and

that this power is reduced by the ratio of the low pass filter bandwidth LPg^ to the full bandwidth of

the source, 100 MHz. For example, if the BLWGN excitation bandwidth is 10 MHz, the low pass

filter would have a bandwidth of 5 MHz and, thus, the internal attenuator would be set to 7 dB. The

LO input was varied over a relatively large range (-20 to dBm), but care was taken to monitor the

BLWGN excitation signal to ensure good suppression of the rf carrier and good SNR. A better

system would use a step attenuator and small signal amplifier after the mixers to allow the LO input

to be maintained at the optimal level for the mixer while the output power is controlled by the step

attenuator. This would allow maximum dynamic range.

Mixers designed to have low input VSWRs typically do so at the expense of IF to LO isolation. Some of the IF power is transmitted back out the LO input and ends up at the synthesized sweeper

output port. If internal leveling is being used on the sweeper, the IF signal being fed back can be

interpreted as sweeper output power causing the sweeper to generate less rf. One way to reduce this problem is to place some attenuation between the mixer and the sweeper and increase the sweeper

output by the same amount. For example, if a 10 dB attenuator is placed between the synthesizer and

the modulator, the synthesizer output can be increased to +5 dBm (to give -5 dBm at the mixer).

29
The IF being transmitted back will be reduced by 10 dB while the sweeper output was increased by

10 dB, thus, allowing a 20 dB improvement in SNR for the leveling circuits. In this experimental

series, a 6 dB splitter was placed between the sweeper and the mixer to monitor the sweeper output.

Thus, a 12 dB improvement was inherent in the set-up.

The final issue that must be considered when using mixers for BLWGN excitation is the IF and,

to a lesser extent, the higher harmonics of the LO present on the rf output port. The harmonics

generated by the TWT amplifiers are usually much more significant. When the signal is fed through

a band-limited amplifier, such as a TWT, and then through a band-limited antenna, the IF is inherently filtered out. When measuring the signal directly out of the mixer or when using it for

external leveling, however, the IF can have a significant effect. The detector heads used on the scaler

analyzer have a lower frequency range of 10 MHz and, hence, for wider bandwidth BLWGN

excitation they can pick up the IF signal which results in erroneous readings. Likewise, detector

diodes used for external leveling experience the same problem. Certain low power applications where

very broad band amplifiers with low cutoff frequencies are used can further aggravate the problem by

amplifying the IF. This problem can be corrected by using a high pass filter with a lower cutoff

range below the desired BLWGN excitation frequency but above the IF bandwidth.

Figure 5-1 shows the suggested improvements mentioned above. The 10 dB attenuator improves

isolation between the IF signal and the synthesized sweeper. The high pass filter eliminates the IF

signal from appearing in the output. It should be noted that a high pass filter could also be used in

place of the 10 dB attenuator if sufficient power is not available from the sweeper. And finally the

small signal amplifier boosts the signal for input to a TWT while the computer or voltage controlled

attenuator allows varying the output level over any desired dynamic range.

6. CONCLUSIONS

6.1 Chamber Environmental Characterization

6.1.1 Effective Mode Density

At low frequencies where the total number of modes within the span of the Q bandwidth is small,

the effective mode density is essentially equivalent to the theoretical mode density. At higher

frequencies, where many modes are encompassed by the span of the Q bandwidth, effective mode

density becomes dependent upon the ratio of noise bandwidth to Q bandwidth.

6.1.2 Field Uniformity

In general, as described above, a reverberation chamber has two separate frequency regimes that determine the ability of either BLWGN excitation or mechanical mode perturbation to provide field uniformity. In both regimes, mechanical mode mixing relies upon physical stirrer size, while

BLWGN excitation depends on noise bandwidth. For a given mechanical tuner and test configuration, an equivalent field uniformity can be achieved by selecting an appropriate noise

bandwidth.

6.1.3 Field Peak-to-Average Ratio

The peak-to-average field ratio for both mechanical and BLWGN excitation are equivalent and are approximately 7 to 8 dB for sufficiently high-Q cavities. For a given level of mode mixing with

30
either mechanical mode mixing or BLWGN, the peak-to-average ratio in a heavily loaded (low Q)

chamber can be significantly less than the typical range of 7 to 8 dB.

6.1.4 Time Between Peak Field Excitation

For mechanical mode mixing the EME refresh time is determined by the tuner rotation rate and is typically on the order of seconds. In the NIST reverberation chamber, the peak field refresh time for

BLWGN excitation was less than 150 ms and could be as short as a few tens of microseconds.

6.2 Test Applications

Based on the three EUTs tested, the following conclusions were reached.

6.2.1 Peak Power Sensitive EUTs

Failure levels for BLWGN excitation are equivalent to failure levels for mechanical mode mixing

for peak power sensitive EUTs.

6.2.2 Average Power Sensitive EUTs

For average power sensitive EUTs there is an apparent 7 to 8 dB difference in failure levels. This is based on the difference in cavity EME for the two test procedures. During a mode-tuned

procedure, for some tuner position(s) the EUT will be exposed to the maximum field for the tuner dwell time. Thus, an average power sensitive EUT will fail at a level determined by the peak field in

the chamber when a mode-tuned procedure is used.- When the chamber is excited with BLWGN

whose bandwidth is less than the susceptibility bandwidth of the EUT, the failure level will be

determined by the average power in the chamber. As noted earlier, the peak-to-average power ratios for the two techniques are equivalent and are approximately 7 to 8 dB. If the EUT susceptibility

bandwidth is narrower than the noise bandwidth, the measured failure level will be increased by the

ratio of the noise to susceptibility bandwidth.

6.2.3 Shielding Effectiveness Measurements

Shielding effectiveness measurement results for BLWGN excitation are equivalent to mechanical

stirring results provided resonances in the shielding transfer function are wider than the noise

bandwidth. Shielding transfer function features which are narrower than the noise bandwidth will be

averaged over the noise bandwidth.

6.3 Test Considerations

1Testing using noise excitation requires consideration of several instrumentation and procedural

issues which may be different from the issues associated with mechanical stirring.

2. Using BLWGN test time requirements using noise excitation are reduced by a factor up to 200

compared to a conventional mechanical stirring method.

3. Test EUTs are exposed to fields which are uniform on time scales on the order of tens of

microseconds.

31
4. Most methods of measuring chamber field strength may be used for either technique.

However, high resistance line sensors should not be used with BLWGN excitation. In lieu of

typically used reference antennas such as log periodics and horns, smaller magnetic and electric field sensors such as B-dot and D-dot probes can be used to reduce chamber loading.

5. Power stacking or combining is achievable using both mechanical stirring and BLWGN

excitation techniques.

7. RECOMMENDATIONS

Extend the comparison of mechanical and BLWGN mode-mixing techniques to further investigate

the characterization of cavity uniformity by the two techniques.

Investigate how devices with narrow susceptibility bandwidths affect "effective" cavity uniformity.

Continue the validation of the BLWGN technique for radiated immunity testing by investigating a broader class of test systems.

Extend the comparison of the BLWGN technique to additional established radiated techniques

such as open area test sites, shielded enclosures, GTEM cells, semi-anechoic and anechoic chambers.

Investigate the ability to employ a broad range of modulations with the BLWGN technique.

Investigate the application of BLWGN to small and/or densely packed cavities in which it is impractical to accommodate an adequately sized mechanical tuner.

Continue the theoretical and experimental investigation necessary to resolve the effects of aperture

size, window-to-aperture interface, and fixture size and Q effects on shielding effectiveness

measurements using nested reverberation chambers.

Evaluate the feasibility of simulating the effects of plane wave illumination of large test articles using multiple antennas driven by incoherent BLWGN.

8. REFERENCES

[1] Mendez, H.A.; A new approach to electromagnetic field-strength measurements in shielded

enclosures. Wescon Technical Papers, Western Electronic Show and Convention: Los Angeles,

CA; Aug. 20-23, 1968.

[2] Bean, J.L.; Hall, R.A.; Electromagnetic susceptibility measurements using a mode stirred

chamber. IEEE Intl. Symp. on Electromagnetic Compatibility; Atlanta, GA; June 1978.

[3] Corona, P.; Latmiral, G.; Paolini, E.; Piccioli, L.; Use of a reverberating enclosure for

measurement of radiated power in the microwave range. IEEE Trans. Electromagn. Compat.

EMC-18(2): 54-59, May 1976.

32
[4] Cummings, J.R.; Translational electromagnetic environment chamber, A mew method for

measuring radiated susceptibility and emissions. IEEE Intl. Symp. on Electromagnetic

Compatibility; San Antonio, TX; Oct. 1975.

[5] Roe, J.M.; An improved technological basis for radiated susceptibility and emissions. IEEE Intl. Symp. on Electromagnetic Compatibility; San Antonio, TX; Oct. 1975.

[6] Crawford, M.L.; Koepke, G.H.; Design evaluation and use of a reverberation chamber for

performing electromagnetic susceptibility/vulnerability measurements. Nat. Bur. Stand (U.S.)

Tech. Note 1092; April 1986.

[7] Loughry, T.A.; Frequency stirring: An alternate approach to mechanical mode-stirring for the

conduct of electromagnetic susceptibility testing. Phillips Laboratory, Kirtland AFB, NM; Report

No. PL-TR-91-1036; Nov. 1991.

[8] Carlson, A.B.; Communication Systems, An Introduction to Signals and Noise in Electrical

Communication. McGraw-Hill, New York, NY; 1975.

[9] Mory, R.; et al.; Development and production of multi-gap loop (MGL) series EMP B-dot.

Report No. AFWL-TR-70-153; Air Force Weapons Laboratory; Feb. 1971.

[10] Olsen, S.; Asymptotic conical dipole D-dot sensor (ACD-Sl(R)) development. Report No.

AFWL-TR-75-263; Air Force Weapons Laboratory; May 1977.

[11] Pozar, D.M.; Microwave Engineering. Addison-Wesley Publishing Co., Reading, MA; 1990.

[12] Whiteside, H.; King, W.P.; The loop antenna as a probe. IEEE Trans. Antennas Propagat.

AP-12(3): 291-297; May 1964.

[13] Richardson, R.E.; Mode-Stirred Chamber Calibration Factor, Relaxation Time, and Scaling

Laws. IEEE Trans. Instrum. Meas. IM-34(4); Dec. 1985.

[14] Hatfield, M.O.; Freyer, G.J.; Johnson, D.M.; Farthing, C.L.; Demonstration test of the

electromagnetic reverberation characteristics of a transport size aircraft. Naval Surface

Warfare Center, Dahlgren Division, Dahlgren, VA; Report No. NSWCDD/TR-93/339.

[15] Jackson, G.A.; Smithers, B.W.; Review of EMC methods using mode stirred enclosures.

Electromagnetic Compatibility Conf., University of Surrey, London; Sept. 1984.

[16] Hill, D.A.; Crawford, M.L.; Kanda, M.; Wu, D.I.; Apermre coupling to shielded

transmission lines: theory and experiment. Natl. Inst. Stand. Technol. NISTIR 3988; Apr.

1992.

[17] Wilson, P.F.; Ma, M.T.; A study of techniques for measuring the electromagnetic shielding

effectiveness of materials. Nat. Bur. Stand. (U.S.) Tech Note 1095; May 1986.

[18] Wilson, P.F.; Ma, M.T.; Simple approximate expressions for higher order mode cutoff and

resonant frequencies in TEM cells. IEEE Trans. Electromagn. Compat. EMC-28(3): 125-

130; Aug. 1986.

33
[19] Department of Defense Methodology Guidelines for High Power Microwave (HPM)

Susceptibility Assessments, Office of the Secretary of Defense; Jan. 1990.

[20] Tai, C.T.; On the definition of the effective aperture of antennas. IEEE Trans. Antennas

Propagat. AP-9: 224-225; Mar. 1961.

[21] Crawford, M.L.; Koepke, G.H.; Electromagnetic radiation test facilities: Evaluation of

reverberation chambers located at NSWCDD, Dahlgren, VA. Nat. Bur. Stand. (U.S.)

NBSIR 86-3051; June 1986.

[22] Crawford, M.L.; Koepke, G.H.; Ladbury, J.M.; EMR test facilities - Evaluation of

reverberating chamber located at RADC, Griffiss AFB, Rome, NY. Nat. Bur. Stand. (U.S.)

NBSIR 87-3080; Dec. 1987.

[23] Crawford, M.L.; Ma, M.T.; Ladbury, J.M.; Riddle, B.F.; Measurement and evaluation of a TEM/reverberating chamber. Natl. Inst. Stand. Technol. Tech. Note 1342, July 1990.

[24] Hatfield, M.O.; Shielding effectiveness measurements using mode-stirred chambers: A

comparison of two approaches. IEEE Trans. Electromagn. Compat. EMC-30(3): 229-238;

Aug. 1988.

34
1

,—

1

fLDTriR

1

-| r.irrH ri.c

1 II

( , 1 IJ

f —.-] (XxfuraR

synthesized

sweeper

1 .x<—r-—H> 1 /

/

1 '< 1

. i-i.cn foacs

1• J

scalar

analyzer iicrc :ajKcz

f \ ( kiiis:

\-

, >lv£f-jo?/.rtDr( cwMf€S ^\ ,

l)

Figure 2-1. Generating and measuring NBWGN excited fields.

1 i

J v:c^f LPP %"^K

©

fc

y

Y(f) ©

ysy yETf

^ «

Figure 2-2. Theoretical spectrum generated by NBWGN method.

35
HoJuUCir Output (1Q H*z)

'rl

-75

1

1 1

1

1 II

1 (I II

j

1

1 1 1 \

1

1 7 M, 1 -^ r-r 1 , ,

, r-1

1 .ZS 1 .50 1.75

Mooul sl<: • IIPUC ( 20 Mu: ) -25

1 ^

6 -vO / \\, 1 w f \

1 1 r

1 X

1

.. z V 1 -75 1 ^' :- : 1

1.2S l.SQ 1.75

Noduisccr Output ( 50 wHz; -25

1

1

1 *^

1 1

i \ / 1 \

9 -.U _L/ \^

M ^:=r^ 1s X ' ^

^ 1 N

1 .

III 1 1

J 1 .10 1 7

f r*.ju*ricy C'^nO

Wodui«tor Ou tput ( IQQ WHI) -JO M* "mt

J

. "<n / \ -S*; /1 N

-t;o ^ 1 WJ

-dS 1 1 1 i I 1 1

1. :5 1 .50 1.75

Figure 2-3. Actual output of NBWGN excitation method with excitation method with a center frequency of 1.5 GHz and four agility bandwidths.

36
Vo

MULTIGAP B-DOT PROBE

<8)

ATTENUATORS

Figure 3-1. Probe—balun configuration.

Figure 3-2. Field uniformity with 10 MHz of NBWGN over 1 to 2 GHz.

37
0. Q30

0. D25

0. 020

P

F 0.015

0. Q-IO

0. 005

D. Qoa

Figure 3-3. Field uniformity Relative Frequency Distribution for 2 MHz of NBWGN at 0.5 to 1 GHz.

a. D30

0. 025

0. 020

. D [I 5

C . [I [I

Figure 3-4. Field uniformity Relative Frequency Distribution for 10 MHz of NBWGN at 0.5 to 1 GHz.

38
Figure 3-5. Field uniformity Relative Frequency Distribution for 20 MHz of NBWGN at 5 to 1 GHz.

0.030

0.02S

0. 02Q

R

F 0.013

0. 010

0.005

0. 000 flUliTmOirffi

0.5 0.0 1 . 2.

Figure 3-6. Field uniformity Relative Frequency Distribution for 50 MHz of NBWGN at 0.5 to 1 GHz.

39
Q. Q3S

0. i:3Q

0. 025

R 0.020

f

0. 015

0. 010

0. 005

O.OOO h-i n n p

Figure 3-7. Field uniformity Relative Frequency Distribution for 2 MHz of NBWGN at 1 to 2 GHz.

0.030

.02S

. 020

R

F

. 01S

0.010

. 0Q5

. OOO

- 1

d9

Figure 3-8. Field uniformity Relative Frequency Distribution for lU MHz of NBWGN at 1 to 2 GHz.

40
.030

0.025

.020

F 0.015

D

.010

.005

0.000

-2.0 D .0 2. 5 dB

Figure 3-9. Field uniformity Relative Frequency Distribution for 20 MHz of NBWGN at 1 to 2 GHz.

0. 030

0.025

Q. Q2D

F 0.013

0. 01O

0. 005

0. OOO

1.5 - 1 .0 1 . 5

Figure 3-10. Field uniformity Relative Frequency Distribution for 50 MHz of NBWGN at 1 to 2 GHz.

41
0. 040

0. 035

0. 030

0. 025

a

f 0.020

D

0. 015

0. 010

0. 005

0. 000

15 - 10

d8

10 15

Figure 3-11. Field uniformity Relative Frequency Distribution for 2 MHz of NBWGN at 2 to 4 GHz.

o.ana

0.a35

0.030

o.a2s

n

F 0.020

O.01S

0.010

0.005

o.ooo

Figure 3-12. Field uniformity Relative Frequency Distribution for 10 MHz of NBWGN at 2 to 4 GHz.

42
Figure 3-13 . Field uniformity Relative Frequency Distribution for 20 MHz of NBWGN at 2 to 4 GHz.

Figure 3-14. Field uniformity Relative Frequency Distribution for 50 MHz of NBWGN at 2 to 4 GHz.

43
. D30 r

.025

. 020

f 0. 015

D

. 010

. 005

. DOQ

15 10

Figure 3-15. Field uniformity Relative Frequency Distribution for 2 MHz of NBWGN at 4 to 8 GHz.

Figure 3-16. Field uniformity Relative Frequency Distribution for 10 MHz of NBWGN at 4 to 8 GHz.

44
o.ooo

Figure 3-17. Field uniformity Relative Frequency Distribution for 20 MHz of NBWGN at 4 to 8 GHz.

0.03Q

0.023

O.O20

R

F 0.015

0.010

0.00 5

0. OOO

-4 - 1 dB

Figure 3-18. Field uniformity Relative Frequency Distribution for 50 MHz of NBWGN at 4 to 8 GHz.

45
Figure 3-19. Field uniformity Relative Frequency Distribution for 2 MHz of NBWGN at 8 to 18 GHz.

Figure 3-20. Field uniformity Relative Frequency Distribution for 10 MHz of NBWGN at 8 to 18

GHz.

46
. 035

0.030

. 025

. 020

D 0.015

0.0-10

. OGS

O.OOO

Figure 3-21. Field uniformity Relative Frequency Distribution for 20 MHz of NBWGN at 8 to 18

GHz.

Figure 3-22. Field uniformity Relative Frequency Distribution for 50 MHz of NBWGN at 8 to 18

GHz.

47
19000

16000

14000

12000

Q 10000

6000

4000

:aaa

/

A / MA /

1 \

/

' \ ^^

-'^'

A /

^

/

/

/

,^/

(

E g 10

Figure 3-23. Average measured Q for the NIST chamber.

7

1

5

\ u1

<3

3 ^

2

il ^--^ \

\/^ A

\

' w 1

1 """^^^

.

: •< £ 6 10 i; 14 11 1d Fr*qu*ncy CGH:)

Figure 3-24. Equivalent modes per megahertz of NBWGN available for mode mixing.

48
)e 3

e ] U ]

'.( ]

3e ]

f ]

I e

:

Ge •

:3

sS

Figure 3-25. Chamber power density peak-to-average ratio measurement for NBWGN excitation.

a

c

Q.a

Jo

M

_/u

>C

\

X

Max/Five o-f Loss Wtth Rbsorber, 8GHz

Mechanical Tuning

:/flvc of Lors uuith fibsorber, 2GHz

Mechanical Tuning

riax/Hve Loss ulth Flbsorber, 2GHz

1 MHz No i se

2 4 6 No. of Pieces [2' Absorber

Figure 3-26. Chamber power density peak-to-average ratio measurement for CW (mechanical tuning)

and 10 MHz BW NBWGN excitation as a function of absorber loading (Q).

49
SHIELDED ENCLDSURE:

HORN

TRANSMIT

HORN 1

TRANSMIT

DRN 2

NDISE

SOURCE

RF

SOURCE

SCALAR

ANALYZER

IMIH6 dB

SPLITTER

RF AMP

DETECTORS

Figure 3-27. Block diagram of system for increasing test field inside reverberation chamber using a

single noise source and mixer and two rf amplifiers.

50
SHIELDED ENCLOSURE

^-7 TRANSMIT V HDRN 1

TRANSMIT

HDRN 2

NDISE

SOURCE •xlee:

NDISE

SOURCE -XLEB

RF

SOURCE

WIMB

WJME}

RF AMP

6 dB

SPLITTER

ANAl'y' ER hHP^TECTDR

Figure 3-28. Block diagram of system for increasing test field inside reverberation chamber using two

noise sources and mixers and two rf amplifiers.

51
-5

-10

-15

-20

^ -25

Bm -35

-40

-45

-50

( (a)

Suj- 1

k

ildlj I a- Dual input (1 source) (10 MHz)

^

ii 1 1 , ^^i]n\

A,*

^*i I

IIif^^ b„, t

. 1

im

T!|MWMi\ii Single input (1 source) (10 MHz)-^^ mm%rl

2 4 6 8 10 12 14 16

Frequency (GHz)

18

Power Increase (l Noise Source)

4.2

4.0

3.5

3.2

3.0

2.6

2.4

8 10 12

Frequency (GHz)

14 16

Figure 3-29. Power density measured inside NIST reverberation chamber excited using single noise

source, (a) power density from single and dual inputs, (b) increase in power density from

single input to dual input (Pd dual - Pd single).

52
-10

-15

-20

-25

B -35

m -40

-45

-50

-55

-60

h. (a)

fk

«It ^— Dual Input (lO MHz. dBm) ^^^

feftl^

^ 'l*ii f IS3ML W| ^ ta^f Lui

'r^

fti,

Singh5 Input (10 MHz. dBmi—-^' "W^Ww

2 4 6 8 10 12 1^ Frequency (GHz)1 IEi 18

Power Increase (2 noise sources)

8 10 12

Frequency (GHz)

14 16

Figure 3-30. Power density measured inside NIST reverberation chamber excited using two noise

sources, (a) Power density from single and dual inputs, (b) Increase in power density

from single input to dual input.

53
1

SHIELDED ENCLOSURE

DIGITIZING

SIGNAL

ANALYZER

NOISE

SOURCE

B-DOT D-DOT

PROBE PROBE

TRANSMIT

ANTENNA

COMPUTER

2-18 GHz

MIXER

SCALAR

ANALYZER

"DETECTOR<

DETECTDR<

DETECTOR

^DETECTOR

Figure 3-31. Block diagram of system for measuring transient response and field uniformity of

NBWGN excited chamber.

54
iii iii ii i|n I I II I I i |

I I

1 1 m+f4jjj |ii 1

1

1 II i iiiiii|iiii mii|iiii nil i|i II mil i

|

i ni l i n i

| Mi i Mi M I ^n^

5

M

2

N

I

cc

(11 CQ<

o

z

o

<

cc

Ui

CQ

cc>

Ui

cc

<0Z

I 1 1 1 1 1 1 I 1 1 I r I r r I 1 1 1 1 I n r f I 1 1 1 1 1 1 I iTTrfrt-i-t-M 1 1

1 11 1 ii r i u m 1 1i 1i 1^ I . , r u 1 1 1 1 I (-^ ooooooooooo OCDOOr^CDLO'^COCNJ-.-O

T- o o o o o o o o o o

LO

uo

©

o

o

E

E

iN

X

o

at

0)

•so

X!

(U>

C/3

-I—

-ag

s

o

e

N

K

o

en

I en3

(SO

(L-3)

(S}|0A) ieu6lS p8A!909y

55
Ill II 11 ii{ II II II 11 1| II II 11 iii| II II II II i| II II II II i{ II II II II i{ II II II II i|i II I UN 1(1 II II nil |i II II II II 5

QQ

N

X

o

N

DC

(J

<

<

<

Ui

a.

<

<

a

m

<

a:

LU>

<

cc

Ui<

o

z

o

H

<

cc

HI

CD

a:Ui>

Ui

cc

<0

C\J

LO

o

LO

I " 1 11 11 1 I I I I I I I r I I I I I I r I M [ I I r I I I 1

I I TTT^I i

i r i i i I i i i i i i i i i I i r i i r i i i r I i i i [

i r i i I i i i i i i i i i ^ i r i i i r r i I ooOCDC0r^CDL0'<;fC0C\jT-O ooooooooo TT- o o o o o o o o o o

o

N

o

^^

C!o

(0

O 2

i_

o on

• ^-^ l-H £

Z

Ui '^-^ a

E

i- i

o

(Us

N

o

o

2

PQ

en

enm

3

60

(s^ioa) |bu6!s peAj8oey

56
y

circular aperture,

radius r

Figure 4-1. Geometry for an apertured TEM cell.

57
-so

-80

J I I L.

Figure 4-2. SE measurements of NIST 12 cm TEM cell obtained using NIST reverberating chamber

with mechanical mode-mixing.

1 o

- 10

30

-so

-70

-I—1—1—I—1—I—I—I

\—1

I—I—1—1—I—I—1—I—I—1—I—I—1

\—I—q—1—1—I

I r

Figure 4-3 . SE measurements of NIST 6 cm TEM cell obtained using NIST reverberating chamber

with mechanical mode-mixing.

58
GATING FtlNCTION

n

— BVg —

Ag

I1_ COm'O LLrnON/(AgBWg) —BVg-BVr

I

ArBVr/BUg

CROSS SECTION

Ar

r

BVo + BVr

BV/r

Figure 4-4. Effect of cross-section bandwidth smaller than bandwidth of gating function.

GATEKG FUNCTION

-^9

—BVgH

CROSS SECTION

CONVOLUTION/(AgBWg)

BVr-BV/g

Ar

BVg + EWr

Figure 4-5. Effect of cross-section bandwidth greater than bandwidth of gating function.

59
TEM Ce I I 1 C2 MHz}

TEU Ce I I 1 C20 UHz]

TEU Ccl I 1 CIO WHz]

TEU Cel I 1 CSa WHzJ

Figure 4-6. Narrow cross-section resonance of 12 cm TEM cell measured using BLWGN.

TEM Ce I I 2 C2 UHz^

TEM Ce M 2 Q20 MHz}

TEK C«l I 2 CIO UHz3

TEU Ce I I 2 C50 UHz}

Figure 4-7. Wide cross-section resonance of 6 cm TEM cell measured with BLWGN.

60
COUNTER BOX

2 - 8 BIT COUNTERS

TEST CIRCUIT

16 DATA LINES

(COPPER)

CLOCK LINE

RESET LINE

(FIBERS)

TEST

MASTER BOX

CI DCK

ERROR DETECTION CIRCUIT

RESET CIRCUIT

(FIBER)

MODE STIRRED CHAMBER

FIBER OPTIC

LINES

(ERROR,RESET

& TEST)

MONITOR BOX

ERROR INDICATOR

RESET

ANTE-ROOM

Figure 4-8. Common test article showing interconnections and functions.

<E

u\

3

E

m

>-f-M

z

u

p

CLu

o

a.

0.2 0.4 0.6 O.B 1 1.2 1.4

FREQUENCY GHz

1 . B

Figure 4-9. Susceptibility profile of common test article obtained using CW excitation in the

NSWCDD mechanically mode-mixed reverberation chamber.

61
Figure 4-10. Set-up for performing susceptibility tests of common test article in a reverberation

chamber.

N^ a^

FIBER OPTIC

LINES

SPECTRUM

ANALYZER

REE

FIELD B-DOT NIST SURFACE PROBE MOUNT

B-DOT

CTA MONITOR BOX NETWORK

ANALYZER

RF SYNTH

LP

FILTER

NOISE SOURCE

Figure 4-11. Block diagram of system for performing susceptibility tests of common test article in NIST reverberation chamber using BLWGN excitation.

62
RF

SYNTH

NETWORK

ANALYZER

Figure 4-12. Block diagram of system for performing susceptibility tests of common test article in

NIST reverberation chamber using CW excitation with mechanical mode-mixing.

63
15 1

1 i

10 -

e\

£

50

-5

-10

-

e

ft G

i

It

e

t

G

e «

«• «. -

5 J. 6 ? c $ e "

f _

m

a

-15

-20

-

s FOUR PROBE RVERRGE DRTR**"

e - 2 MHz NOISE F3flNDWIDTH

» - 4 MHz NOISE BANDWIDTH

S - 10 MHz NOISE BANDWIDTH

V. -- - 20 MHz NOISE BANDWIDTH

-25

f i. - 50 MHz NOISE BANDWIDTH

-30 "" J

( t* - 2 PROBE AVG AT 350 MHz) _

-35 1 1 1 0.5 1 1.5 2 FREQUENCY (GHz)

Figure 4-13. Susceptibility profile of common test article obtained with BLWGN (5 bandwidths).

15

10

. y. X -

5

OJ£

1

\

£

-5 -

-^^ -10 _ y.

X -15 — m _ T! -20 •£

-25 -

-30

X -35

y. - PEAK POWER DENSITY AT UPSET

X - AVERAGE POWER DENSITY AT UPSET

0.5 1 FREQUENCY. ( GtLz )•

I .5

Figure 4-14. Peak and average power susceptibility levels of common test article obtained with CW

excitation and mechanical mode mixing.

64
-I i

1

i—^—I—1\ 11 ^\ r

if] _J

LI>

U

_J

I-u

U)

Q.

OJ

in

o

o

o

o

> Z

U

o

N

I

(J

(L> O

O

M

H

CE

q:

LI

CEu

>

CEO

H

Y

CEU

CL

CE

1-U

I I I I I J I L

in n en fN- in n n in 'o

u

o

Oh

0) z

u (U> o

Cj a

b. •^

(UOh

in

' g B U

o ^(U x>

(Uu

C

(U

u<

,1^

(SO C3Dby3AU 01 >lU3d Ull^Q ) ap

65
(\JE

\

3.

E

m

15 1

10 -

S -

>i

y.

Q -

y

''

>i y

R R -5 - y

^

ft X X

10 - y.

a

y.

X

flX

15 -

X

-30

-35

• fl A - AVG OF 4 PROBES FOR 2,4,10,20 & 50 MHz BWs

X - AVG OF 4 PROBES FOR CW (MECHANICAL TUNING)

\ - AVG OF PEAK OF 4 PROBES FOR CW (MECH. TUNING)

I JL 0.5 1 1 . 5 FREQUENCY (GHz)

Figure 4-16. Comparison of BLWGN and average CW (mechanical mode mixing) power densities data

for common test article upset.

u

(J

cr

q:u

>u

o

zo

IUJ

CJ

ITu

>

2

U

(T

h-

_lu

a

0. 5 1 rREOUENCY (GHz)

1 . 5

Figure 4-17. Difference between CW (mechanical mode mixing) average and BLWGN data for

common test article upset.

66
u

(J

CE

01U

>

CEU

cnH

o

z

m

>

(E

U

0.z

u

cno

L.O

M

ICEu

(T

>

CEO

ICEU

Q.

CEH

U

1—I—

r

"1—1—

r

aj

N

I

Z

Q

IDn

hCE

(J

>

CEU

m

o

CL

OJ

I *

in

N

I

(J

U

z

u

D

o

u

(K

ino

CL,

0)

o

o

o

l-l ,o

in n

J I I I I L_l I I '

I di r^ in n , ^ '-• cn m

o

O

S

O

oI

U

00

(DAU 3SI0N - >]U3d MO 01130) SP

67
m

n

CE

_JP

cru

z

o

a.

6

5

4

3

2

I

-1

-2

-3

-4

-5

X - CW

e - 2 MHz

# - 4 MHz

NOISE BRNDWIDTH - NOISE BRNDWIDTH "

s. If

? ^

f ?

S - 10 MHz NOISE BRNDWIDTH -

'^ - 20 MHz NOISE BANDWIDTH

8.-50 MHz NOISE BRNDWIDTH

1

0.5 1 1.5 2

FREQUENCY (GHz)

Figure 4-19. Difference between input power to NIST reverberation chamber required for upset of

common test article determined from average of all probe and antenna measurements and

from each individual sensor.

68
69
OHCE PLUGGED IHTO THE

PERSONALITY KEY COHHECTOR

THE INSTRUMENT READS THE

PERSOHALITY KEY POU EACH

OP THE SENSOR PROBE

CALIBRATION DATA .

PERSONALITY KEY

TEMPERATURE CONTROLLED

LED LIGHT SOURCE

OPTICAL

COUPLER

^in>

CHI ==>CM1

' CK 1

I ='cri t

^=1II>

^<2_

INSTRUMENT

REAR PANEL

PERSONAL ITY KEY CONNECTOR

B

SENSOR PROBE

ASSEMBLY

<Z

PlIOTODIODE

ft

D E AH SPLITTER

PMOTODIODE

INSTRUMENT

BULKHEAD

CONNECTOR

:=:m^

HETRI COR MODEL 14 FIBEROPTIC MULTISENSOR SYSTEM

Figure 4-21. Diagram of MetriCor system.

COPPER BRIDCEVIRE LEADS

IHSULATIOH

PROTECTIVE JXCKKT

FIBER OPTIC CABLE

Figure 4-22. Diagram of instrumented FED.

70
Si>>5;-:-;.':.v,..

- , -r': --v .-•':• r-i,-'.'-^^,.<TCi''r'..ft-f'-^^\,-s;»...

'' v..-.'

•'iy.;s<.;," ';i-'-'r< i'i^;;;^'^:--' <:-i; ^v;v"v;A:;^li^•^•

J3o

c

o

X)

(U>

o

o

co

o

uo

2

e3

C!

I 60Co

en

3

71
-sao>HCO

l-H 2

c4 O

STuCO

ca

<u fo^

1^o2

72
—I

1 1 1 1 1 r

I.7S- HOCKCT, »t»0 KSfOuSC

t riMi NOISC IMnOhIOTh

0.03 O.J 0.13 0.2 0.23 0.3 0.33 O.l rBCOUCNCY <CHi)

<0 1

1 1 1 r-- 1 T I

33

2.>3- HOCKCT. HC»0 «CS^ONSC

2a nHt NOISC IOnOuIDTH

-

30

J5 ^^f^f^f^^

M /*v. J^«»// ^

20 - -

IS - -

10 - -

3 r 1 1 r 1 r

1 0.03 0.1 0.13 0.2 0.23 0.3 0.33 razouCNCf (CMi)

0.< 0.<50.5

(a) (d)

5 . 0.03

Jl I 1 l_ 0.1 0. IS 0.2 0.23 0.3 0.33 0.< 0.430.3 rSCOOCNCY (CH:) (b)

0.03 0.1 0.13 0.2 0.23 0.3 0.3S rSCOUENCY (CHi)

0.< 0.«3O.3

(e)

<0 - —r-

' r - r i i i r —i

2.l'5- BOCKCT. HC»0 "CSPONSt

IS nut NOISC BANOHIOTm

n

" 30 2u

U iniWV*'^!^''^ • ^20 -

^ 13 o

K

a

10

3 1 1 1 1 1 I t 1 O.l 0.13 0.2 0.25 0.3 0.33 0.« 0.450.3 TRCOUEnCY ( Gwi ) (c)

INTENTIONALLY LEFT BLANK

Figure 4-25. Folding-fin aircraft rocket response - 0.05 to 0.5 GHz data.

73
(a) (d) —I ' 1

'

1

2.7S- ROOXT. HCnO aCCPONSC

4 r*« t^l9C nNOMIOTH

0.? o.e FREOUCNCT CCHi)

1 '

1

' 1 2.75' Rocxrr, HCRs «ccpaNsc 3a PHI HOISC tANDHIDTH

(b)

0.7 0.1 FREQUENCY (CHi) (e)

INTENTIONALLY LEFT BLANK

(c)

Figure 4-26. Folding-fin aircraft rocket response - 0.5 to 1 GHz data.

74
(a) (d)

(b) (e)

a

0

30

27

T II . 1 "^ -1 t.n- K>ox.r. MCKo narwsc

It nm HOlsc lAHCMtDTV

Z4 -

/I f\

.

** /

1 A

Zu

ITs

21

1*

q: 12

15 '^\/^rr^ V1^

.

h! * - -

C - - a

3 -

1 . r . 1 . 1 TReOUCNCTr (CMZ)

(c)

INTENTIONALLY LEFT BLANK

Figure 4-27. Folding-fin aircraft rocket response - 1 to 2 GHz data.

75
(a) (d)

t

(b) (e)

INTENTIONALLY LEFT BLANK

(c)

Figure 4-28. Folding-fin aircraft rocket response - 2 to 4 GHz data.

76
zsooOc

•aaOo3Oeo

"2ou -I 2 o

;s "-I ^ >

as

(U

l-c 3

op

77
50 "1 ^ MRXIMUM RESPONSE (201 SWEEPS)

20

500

RESPONSE (20 1 SWEEPS)

-I I 1- J I ^ 510. 580 620

FREQUENCY (MHz)

560

Figure 4-30. Response of folding-fin aircraft rocket to CW excitation with mechanical mode mixing.

50

'!5

1 ' ' ^

-NOISE DflTfl - 2 MHz BANDWIDTH

20

SCO

-flVERFlGE RESPONSE C201 SWEEPS) .

510 580 G20

FREQUENCY (MHz)

.J I 6S0

Figure 4-31. Average CW response of folding-fm aircraft rocket vs. noise response (2 MHz

bandwidth).

78
a> Io

<«2oaI c

"o

0)

ill

79
1-1000 MHz sow POWER AMPUFIER

600 MHz LOW PASS

DIRECTIONAL COUPLER

(20 dB)

INC.

HEF. POWERMETER COMPOTER

iL iL

TRANSMITTING ANTENNA

TURNTABLE

2^ METERS

ANECHOIC CHAMBER

FIBER OPTIC CABLE

V

EMC TEST SYSTEM VOLTMETER

STRIP CHART RECORDER

RF

DATA

IEEE /CONTROL

TEST ORIENTATION X

0-180 DEG. AZIMUTH < 1 0.45.-90. 135. 180 DEG. ROLL

^

'o^

A2IMUTVI

ROLL

Figure 4-33. Diagram of anechoic chamber discrete frequency set-up.

80
-1 1

. 1 r f - SS3.3 f«i POWCR OCNSirr - O.ZS „uy,^ < vcRT. pocnsiZflriOH HORH TO ROCKCr-CENTCR - Z.i

e OCC. ROLL

ll« J/0 ROrflTION RNCLE (DCC) (a)

T- (^ T^

r - C)| riMt POMCR oCHSirr - e.25 ^w^iq < VCRT. POLflRIZflTIOH WORN TO ROCKCT-CEXTCR - J . S e OCC. ROLL

ROTflTION RNCLE (OCC) (d)

-T- T (^ 553.3 MMr POWGR DCNSITT - 0.25 •»H^tq c«

VCRT. POUWIZSTIOH

HORN TO ROCKET-CENTCR - 2 . 5 «i 4 5 OCC. ROLL

JO liO 270 ROrflTION BNCLE (DCC) (b)

T- -r

f - S3I MHt POhCR OCHSITT - 8.25 nW^.q cia VCRT. POLftRIZPTIGM HORN TO ROCKET-CENTCR - 2 . 5 «i <5 OCC. ROLL

30 liO Z'O ROrflltON ANCLE (DEC) (e)

r - 553. S MHi POWER OENSITT - a. 25 ..Hx,q , VERT. POLBftlZflTIOK MORN TO ROCKET-CENTER -2.5

9a OEC. ROLL

110 110 ROTPITtON ANCLE (DCCl (C)

3 "

-] 1 I

,

;

r - t3i MHi POWER DCNSITT - 8.25 «>H/iq ca

VCRT. POLflRlIATIOM HORN TO ROCKET-CENTER - 2 . 5 n

99 OEC. ROLL

DO 2'0 ROTBTION RNCLE (OEC)

Figure 4-34. Folding-fin aircraft rocket response to vertically polarized illumination at 553 .5 MHz and

631 MHz.

81
r - S53.5 MHz POWCR OCNSITT - e.2S «H^eq ex

.HORIZ. POLAflrZHTtOH MO«N TO ROCKCT-CCNrCR - i.i r, e OCC ROLL

RorflTiON BNCLE (oec) (a)

-] , —I 1

1

r - SS3.3 ^»^I POWCR oCHSire - a.zs xw^.q ca

HORtZ. POLARIZflriOH HORN TO ROCKCT-CENTCR - 2.S n US OCC ROLL

10 110 270 RO TOT ION ANCLE (OEC

(d)

-I ' I ' I ' r - 353.3 nut POWCR OCMSirT - e.J3 «H^«q c> HORtZ. POLfiRIZRTtOM HORN TO ROCKCr-CENTCR - J . 3 n

<S OCC ROLL 0 <J

1^ 1^

110 ROTATION RNCLE (OEC) (b)

r - 533.5 MHt POWCR OCNSITT - e.J3 •>«/-. q em

HORIZ. POLflRIZBTIOH HORN TO fiOCKCT-CENTER - 2 . 5 »

I le OCC ROLL

90 110 no

ROTATION ANCLE (DEC) (e)

-r -r nr

r - 553.3 fiHt POWCR DCNSITT - e.2S »^H/.q c*

HORIZ. POLflRIZBTIOH HORN TO ROCKCT-CCNTCR -2.3"

Se OCC ROLL

ROTATION BNCLE (.DEC) (c)

Figure 4-35. Folding-fin aircraft rocket response to horizontally polarized illumination at 553 5 MHz.

82
zu

Q:: o

f - 355. S MHr

1 POWCfl OCNSITT - a. 25 «iH/<q c»

HORtZ pouwizRTroM

HORN ro Rooccr-cENrcR - 2.5 n

8 OCC L A /^

ROtL

\

ItO 2/0

iT

r - 555.5 MH:

powcft oCNSirr - a. 23 .>M^tq < MORIZ. POCflRIZTIOM HORN TO ROCKCr-CENTER - 2.5

135 OCC. ROLL

30 tlO £70 RornriON rnclc (DCO

(a) (d)

^30

T- l^ r - 555.5 «Ht POWER DCHSirT - a. 25 .>H/'«g < HORtZ. PO<.nR(ZflrroN HORN TO ROCKCr-CENTCR -2.3

«5 DEC ROLL

30 110 2;o ROTRTION fiNCLC (DCCJ (b)

-r T T 553.5 POWER OENSITT - a. 25 .^H.'.q c«

HORIZ. POLH^rZflTtOM HORN TO ROCKET-CENTER - 2.5 n

laa DEC. ROLL

30 ItO 2;o RorniiON ANCLE (oec) (e)

nr i^ r - 535.5 fiHi POWER OCMSITT - a. 25 rMy,,, < HORIZ. POLflRtZHTION HORN TO ROCKET-CENTER -2.5

93 DEC ROLL

30 110 2/0 ROTRTIOn RnCLE <0EC3 (c)

Figure 4-36. Folding-fin aircraft rocket response to horizontally polarized illumination at 555.5 MHz.

83
-r T r^ POUCn OCHSITX - e.23 "Wz-iq ea

MORIZ. cocnRlZflrtoti HORN TO Rocxcr-CENrcR - z.s «

e occ. RO(.L

Its 110 :<a ROrflTtON fiNCLE (GEO

(a)

-r -r

r - cea mmi POMCR OCHSITT - a.ZS »a<^.q < Ho«rz. pocflRiZflrroM HORN TO ROCKCr-CENTCR - 2 .

S

133 OCC. ROLL

110 zro RornitoN RNCuc (oeo

(d)

?3,

— 1• 1 1 1 r - tea nut POWER OCNSITY - a.JS inM/jq < HORIZ. POLBRtZRTIOH WORN TO ROCKCr-CENTCR - Z.S

4 5 OCC. ROLL TJ 3'

l{0 110 I'O ROTRTION PNCUE (DEC) (b)

r - cea mhi POWCR OCHSITT - e.2S "HXiq < H0R:Z. fOLflRtZRTtON HORN TO ROCKCT-CENTCR -2.5

laa OCC. ROLL

30 110 t)Q ROrPTION ANCLE (oec) (e)

^ I ' 1 I ' I f - (ea MHi POWCR OCnSITT - 8.25 «>H^.q HORIZ. POLflRrZRTtOM HORN TO ROCKCr-CENTCR - 2.3

3 a OCC. ROLL

1(0 Ito t'O ROTRTION RNCLE (OEC) (c)

Figure 4-37. Folding-fin aircraft rocket response to horizontally polarized illumination at 600 MHz.

84
r - (31 MWf powc« ocNStTr - 9.J5 -w'm < HO«tz. PO<.««tznrfON HORN TO ROCKCT-CCNTC« - Z.S

a occ. aou.

110 I'O ROTfinoN rnclc (dcci

"T 1 1 » - tj c nut foucR oCNsrir - e.zs " MOBn to BOCKCT-CCNTCR US occ. aocL

110 270 SOTRTION PNCLE (DCC) (a) (d)

T TPOWER OCnSITT - e.JS " MORir. POl.W»IZ«TI0M HORN TO ROCKET-CENTCR

4S OCC. ROLl.

)0 laO 270 ROTATION BnCLC (DCC) (b)

-1 1

' 1 r - 53 1 MHt cowen oCNStTr - a. 25 »H.'iq < Koorz. ^o<.««iznTroN MORN TO ROCKCT -CENTER - 2.3

lie occ. Ro<.i.

.90 ItO 270 ROTflTrON RNCUC (DECl (e)

-1 • 1 ' 1 ' f - t3 I riH, POWER OENStTT - 6.25 «M^fq c HO«rZ. PO(.W»lZ«r[0W HORN TO ROCKET-CENTER - 2.5

9a OCC. aoui.

30 110 2?0 RornrtON oncuE (DCCi (C)

Figure 4-38. Folding-fin aircraft rocket response to horizontally polarized illumination at 631 MHz.

85
1-1000 MHz sow POWER AMPUFIER

800 MHz LOW PASS

DIRECTIONAL COUPLER

(20 dB)

INC.

REF. POWER

METER

COMPLTTER

,i n

TRANSMITTING ANTENNA

TURNTABLE

2.50 METERS

ANECHOIC CHAMBER

FIBER OPTIC CABLE

>7

EMC TEST SYSTEM VOLTMETER

STRIP CHART RECORDER

RF

DATA

IEEE /CONTROL

TEST ORIENTATION X

AZIMUTH

ROLL

0-180 DEG. AZIMUTH C

. 45 , 90 , 135 , 180 DEG. ROLL V

1

Figure 4-39. Diagram of anechoic chamber swept frequency set-up.

86
500 S-'O 580 G20

FREQUENCY (MHz)

660

Figure 4-40. Anechoic chamber response of folding-fin aircraft rocket.

500 540 580 S20

FREQUENCY (MHz)

GSO

Figure 4-41

. Folding-fin aircraft rocket anechoic chamber data vs. reverberation chamber 2 MHz

data. noise

87
so

500

RNECHOIC CHAMBER DRTfl

540 580 G20

FREQUENCY (MHz)

660

Figure 4-42. Folding-fin aircraft rocket anechoic chamber data vs. reverberation chamber 2 MHz noise

data corrected for polarization factor.

88
STEPPER

MOTOR

CONTROL WINDOW

TEST

FIXTURE TUNER

MOTOR DC

POWER

SOURCE

Figure 4-43. "NESTED" chamber technique for collecting SE data.

Figure 4-44. Photo of wire mesh window.

89
^i

^ i.-^'.'-'^ir,

'--- ^-'^^

-•i^mi^

Figure 4-45. Photo of test fixture used for measuring the shielding effectiveness of shielded window

using the nested reverberation chamber technique.

Figure 4-46. Shielded window mounting for test fixture.

90
Figure 4-47. Window SE test fixture mounted inside NIST reverberation chamber.

91
CW SHIELDED WINDOW MEASUREMENTS

STEPPER f^TOR

co^rrROL WINDOW

DC POWER

SOURCE

LP

FILTER 6dB

SPLITTER SYNTH

POWER

METER

IEEE-488 COMPUTER

DETECTOR DETECTOR

SCALAR

ANALYZER

PLOTTER

lEEE-488 COMPUTER

Figure 4-48. Reverberation chamber CW (mechanical mode mixing) test set-up for SE measurements.

92
WINDOW

TEST

FIXTURE

"

jC

Tx

ANfT

10 dB ATTENUATOR10 dS ATTENUATOR

DETECTOR DETECTOR

20 dB COUPLER

PF AMP

6dB

SPLITTER

DUAL POWER

METER H^

POWER

METER

LP

FILTER

PLOTTER

SCALAR ANALYZER COMPUTER

NOISE SOURCERF SYNTH

IEEE-488

Figure 4-49. Reverberation chamber BLWGN test set-up for SE measurements.

93
Figure 4-50. Results of window SE measurements made using BLWGN technique with 2 MHz

bandwidth.

Figure 4-51. Results of window SE measurements made using BLWGN technique with 10 MHz

bandwidth.

94
60

55

50

AS

d ^° I

A

TK

/] V , B 35

30

V '^/l^ lin uwA

25 ^A J / "^A^ ^Aa

20

w ^n'^A^

^

'^Kr^ w-y15

1.0 1.5 2.0 2.5 3.0 3.5 4.0 4.5 5.0 5.5 6.0

Frequency (GHz)

Figure 4-52. Results of window SE measurements made using BLWGN technique with 50 MHz

bandwidth.

95
CHRMBER vs TEST FIXTURE (BRSELINE5)

15 "T 1 T -I 1 r

CHRMBER

FREQUENCY GHz

Figure 4-53. Comparison of baseline data collected from the test fixture and from the reverberation

chamber.

SHIELDING OF NINDON USING BLNGN METHOD

55 T T

15

DRTR FROM TWO RUNS - W & WO/WINDOW

USING ONE Rx RNTENNR INSIDE THE TEST FIXTURE

-I I L J I U ± -I 1 L. 14 18 2 6 10

FREQUENCY GHz

Figure 4-54. Results of window SE measurements made using BLWGN excitation with and without

window mounted to the test fixture.

96
SHIELDING OF NINDON USING MODE STIR METHOD

m

T5

(3

Z

M

n

_i

u

M

I

55

50

' I T I r 1 I r I

1 1 I r DRTR FROM TNO RUNS - N 8. NO/NINDOW

USING ONE Rx RNTENNR INSIDE THE TEST FIXTURE

6 10

FREQUENCY GHz

14 18

Figure 4-55. Results of window SE measurements made using CW (mechanical mode mixing)

excitation with and without window mounted to the test fixture. DELTA IN SE BETWEEN BLWGN HND MODE STIR METHODS

m

25

20

15

10

(J 5

z

M

P o

_l

u -5 M

I

CO -10

T I 1 T

DRTR FROM TWO RUNS-W & WO/WINDOW

USING ONE Rx RNTENNR INSIDE THE TEST FIXTURE

-15 --

-2

-25

B 10

FREQUENCY MHz

Figure 4-56. Difference in SE obtained using BLWGN and mechanical mode mixing.

97
fihinl^lirp p.'K" ' => ——

-a.

1.0

Figure 4-57. Difference in shielding after smoothing.

RF IN AT

+5 dBn

10 dB

ATTENUATOR

SMALL SIGNAL

BRDAD BAND

AMPLIFIER

IF

HIGH PASS

FILTER

PRE-FILTERED

AND +10 dBn

LEVELED NDISE

Figure 5-1. Improved modulator configuration.

98

COMPUTER DR

VOLTAGE' CONTROLLED

ATTENUATOR

• U.S, GOVERNMENT PRINTING OFFICE 1996-0-774-018/40000


NISTTechnical Publications

Periodical

Journal of Research of the National Institute of Standards and Technology—Reports NIST research

and development in those disciplines of the physical and engineering sciences in which the Institute is active. These include physics, chemistry, engineering, mathematics, and computer sciences. Papers

cover a broad range of subjects, with major emphasis on measurement methodology and the basic

technology underlying standardization. Also included from time to time are survey articles on topics

closely related to the Institute's technical and scientific programs. Issued six times a year.

Nonperiodicals

Monographs—Major contributions to the technical literature on various subjects related to the Institute's

scientific and technical activities. Handbooks—Recommended codes of engineering and industrial practice (including safety codes)

developed in cooperation with interested industries, professional organizations, and regulatory bodies.

Special Publications—Include proceedings of conferences sponsored by NIST, NIST annual reports, and

other special publications appropriate to this grouping such as wall charts, pocket cards, and

bibliographies.

Applied Mathematics Series—Mathematical tables, manuals, and studies of special interest to physicists,

engineers, chemists, biologists, mathematicians, computer programmers, and others engaged in scientific and technical work.

National Standard Reference Data Series—Provides quantitative data on the physical and chemical

properties of materials, compiled from the world's literature and critically evaluated. Developed under a worldwide program coordinated by NIST under the authority of the National Standard Data Act (Public

Law 90-396). NOTE: The Journal of Physical and Chemical Reference Data (JPCRD) is published bi- monthly for NIST by the American Chemical Society (ACS) and the American Institute of Physics (AIP).

Subscriptions, reprints, and supplements are available from ACS, 1155 Sixteenth St., NW, Washington,

DC 20056.

Building Science Series—Disseminates technical information developed at the Institute on building

materials, components, systems, and whole structures. The series presents research results, test methods, and performance criteria related to the structural and environmental functions and the durability

and safety characteristics of building elements and systems.

Technical Notes—Studies or reports which are complete in themselves but restrictive in their treatment of a subject. Analogous to monographs but not so comprehensive in scope or definitive in treatment of

the subject area. Often serve as a vehicle for final reports of work performed at NIST under the sponsorship of other government agencies.

Voluntary Product Standards—Developed under procedures published by the Department of Commerce

in Part 10, Title 15, of the Code of Federal Regulations. The standards establish nationally recognized

requirements for products, and provide all concerned interests with a basis for common understanding of

the characteristics of the products. NIST administers this program in support of the efforts of privatesector standardizing organizations.

Consumer Information Series—Practical information, based on NIST research and experience, covering

areas of interest to the consumer. Easily understandable language and illustrations provide useful background knowledge for shopping in today's technological marketplace.

Order the above NIST publications from: Superintendent of Documents, Government Printing Office,

Washington, DC 20402.

Order the following NIST publications—FIPS and NISTIRs—from the National Technical Information

Service, Springfield, VA 22161.

Federal Information Processing Standards Publications (FIPS PUB)—Publications in this series collectively constitute the Federal Information Processing Standards Register. The Register serves as the

official source of information in the Federal Government regarding standards issued by NIST pursuant to the Federal Property and Administrative Services Act of 1949 as amended. Public Law 89-306 (79 Stat. 1127), and as implemented by Executive Order 11717 (38 FR 12315, dated May 11, 1973) and Part 6 of

Title 15 CFR (Code of Federal Regulations).

NIST Interagency Reports (NISTIR)—A special series of interim or final reports on work performed by

NIST for outside sponsors (both government and non-government). In general, initial distribution is handled by the sponsor; public distribution is by the National Technical Information Service, Springfield,

VA 22161, in paper copy or microfiche form.
U.S. Department of Commerce

National Institute of Standards and Technology

325 Broadway

Boulder, Colorado 80303-3328

Official Business

Penalty for Private Use, $300