DATE DUE

m 2 2

UNIVERSITY PRODUCTS, INC. #859-5503
I

I

i

I

I"

I
7niQ3 oasfifi^

OF %

(0 Q NBS SPECIAL PUBLICATION 260-64

U.S. DEPARTMENT OF COMMERCE/ National Bureau of Standards

Standard Reference Materials:

A Fluorescence

Standard Reference Material:

Quinine Sulfate Dihydrate
NATIONAL BUREAU OF STANDARDS

The National Bureau of Standards' was established by an act of Congress on March 3, 1901. The Bureau's overall goal is to strengthen and advance the Nation's science and technology

and facilitate their effective application for public benefit. To this end, the Bureau conducts

research and provides: (1) a basis for the Nation's physical measurement system, (2) scientific and technological services for industry and government, (3) a technical basis for equity in trade, and (4) technical services to promote public safety. The Bureau's technical work is per- formed by the National Measurement Laboratory, the National Engineering Laboratory, and

the Institute for Computer Sciences and Technology.

THE NATIONAL MEASUREMENT LABORATORY provides the national system of

physical and chemical and materials measurement; coordinates the system with measurement

systems of other nations and furnishes essential services leading to accurate and uniform

physical and chemical measurement throughout the Nation's scientific community, industry,

and commerce; conducts materials research leading to improved methods of measurement,

standards, and data on the properties of materials needed by industry, commerce, educational

institutions, and Government; provides advisory and research services to other Government

agencies; develops, produces, and distributes Standard Reference Materials; and provides

calibration services. The Laboratory consists of the following centers:

Absolute Physical Quantities2 — Radiation Research — Thermodynamics and

Molecular Science — Analytical Chemistry — Materials Science. THE NATIONAL ENGINEERING LABORATORY provides technology and technical ser- vices to the public and private sectors to address national needs and to solve national problems; conducts research in engineering and applied science in support of these efforts; builds and maintains competence in the necessary disciplines required to carry out this research and technical service; develops engineering data and measurement capabilities;

provides engineering measurement traceability services; develops test methods and proposes

engineering standards and code changes; develops and proposes new engineering practices;

and develops and improves mechanisms to transfer results of its research to the ultimate user. The Laboratory consists of the following centers:

Applied Mathematics — Electronics and Electrical Engineering2 — Mechanical

Engineering and Process Technology2 — Building Technology — Fire Research —

Consumer Product Technology — Field Methods.

THE INSTITUTE FOR COMPUTER SCIENCES AND TECHNOLOGY conducts

research and provides scientific and technical services to aid Federal agencies in the selection, acquisition, application, and use of computer technology to improve effectiveness and

economy in Government operations in accordance with Public Law 89-306 (40 U.S.C. 759), relevant Executive Orders, and other directives; carries out this mission by managing the Federal Information Processing Standards Program, developing Federal A DP standards

guidelines, and managing Federal participation in ADP voluntary standardization activities; provides scientific and technological advisory services and assistance to Federal agencies; and

provides the technical foundation for computer-related policies of the Federal Government.

The Institute consists of the following centers:

Programming Science and Technology — Computer Systems Engineering.

'Headquarters and Laboratories at Gaithersburg, MD, unless otherwise noted; mailing address Washington, DC 20234. 'Some divisions within the center are located at Boulder, CO 80303.
Standard Reference Materials:

A Fluorescence Standard Reference Material:

Quinine Sulfate Dihydrate

8 1980

R. A. Velapoldi and K. D. Mielenz

National Measurement Laboratory

National Bureau of Standards

Washington, DC 20234

U.S. DEPARTMENT OF COMMERCE

Luther H. Hodges, Jr., Under Secretary

Jordan J. Baruch, Assistant Secretary for Science and Technology

NATIONAL BUREAU OF STANDARDS, Ernest Ambler, Director

Issued January 1980
Library of Congress Catalog Card Number: 79-600119

National Bureau of Standards Special Publication 260-64

Nat. Bur. Stand. (U.S.), Spec. Publ. 260-64, 139 pages (Jan. 1980)

CODEN: XNBSAV

U.S. GOVERNMENT PRINTING OFFICE

WASHINGTON: 1980

For sale by the Superintendent of Documents, U.S. Government Printing Office Washington, D.C. 20402

Stock No. 003-003-02148-2 Price $4.25 (Add 25 percent additional for other than U.S. mailing).
PREFACE

Standard Reference Materials (SRM's) as defined by the

National Bureau of Standards are well-characterized materials

,

produced in quantity and certified for one or more physical

or chemical properties. They are used to assure the accuracy

and compatibility of measurements throughout the nation.

SRM's are widely used as primary standards in many diverse

fields in science, industry, and technology, both within the

United States and throughout the world. They are also used

extensively in the fields of environmental and clinical analysis. In many applications, traceability of quality control

and measurement processes to the national measurement system

are carried out through the mechanism and use of SRM's. For

many of the nation's scientists and technologists it is

therefore of more than passing interest to know the details

of the measurements made at NBS in arriving at the certified

values of the SRM's produced. An NBS series of papers, of

which this publication is a member, called the NBS Special

Publication - 260 Series is reserved for this purpose.

This 260 Series is dedicated to the dissemination of

information on different phases of the preparation, measurement, certification and use of NBS-SRM's. In general, much

more detail will be found in these papers than is generally

allowed, or desirable, in scientific journal articles. This

enables the user to assess the validity and accuracy of the

measurement processes employed, to judge the statistical

analysis, and to learn details of techniques and methods

utilized for work entailing the greatest care and accuracy.

These papers also should provide sufficient additional information not found on the certificate so that new applications

in diverse fields not foreseen at the time the SRM was originally issued will be sought and found.

Inquiries concerning the technical content of this paper

should be directed to the author(s). Other questions concerned with the availability, delivery, price, and so forth

will receive prompt attention from:

Office of Standard Reference Materials

National Bureau of Standards

Washington, D.C. 20234

George A. Uriano, Chief

Office of Standard Reference Materials
OTHER NBS PUBLICATIONS IN THIS SERIES

Catalog of NBS Standard Reference Materials

(1979-80 edition), R. W. Seward, ed., NBS

Spec. Publ. 260 (April 1979) $3.00* SN003-003-

02048-6

Michaelis, R. E., and Wyman, L. L., Standard

Reference Materials: Preparation of White Cast

Iron Spectrochemical Standards, NBS Misc.

Publ. 260-1 (June 1964). COM74-11061**

Michaelis, R. E., Wyman, L. L., and Flitsch, R., Standard Reference Materials: Preparation of

NBS Copper-Base Spectrochemical Standards,

NBS Misc. Publ. 260-2 (October 1964). COM74-

11063**

Michaelis, R. E., Yakowitz, H., and Moore, G. A.,

Standard Reference Materials: Metallographic

Characterization of an NBS Spectrometric LowAlloy Steel Standard, NBS Misc. Publ. 260-3

(October 1964). COM74-1 1060**

Hague, J. L., Mears, T. W., and Michaelis, R. E., Standard Reference Materials: Sources of

Information, NBS Misc. Publ. 260-4 (February

1965). COM74-1 1059

Alvarez, R., and Flitsch, R., Standard Reference

Materials: Accuracy of Solution X-Ray Spectrometric Analysis of Copper-Base Alloys, NBS

Misc. Publ. 260-5 (March 1965). PB168068**

Shultz, J. L, Standard Reference Materials:

Methods for the Chemical Analysis of White

Cast Iron Standards, NBS Misc. Publ. 260-6

(July 1975). COM74-11068**

Bell, R. K., Standard Reference Materials:

Methods for the Chemical Analysis of NBS

Copper-Base Spectrochemical Standards, NBS

Misc. Publ. 260-7 (October 1965). COM74-

11067**

Richmond, M. S., Standard Reference Materials:

Analysis of Uranium Concentrates at the

National Bureau of Standards, NBS Misc. Publ.

260-8 (December 1965). COM74-1 1066**

Anspach, S. C, Cavallo, L. M., Garfinkel, S. B., Hutchinson, J. M. R., and Smith, C. N., Standard Reference Materials: Half Lives of Materials

Used in the Preparation of Standard Reference

Materials of Nineteen Radioactive Nuclides

Issued by the National Bureau of Standards,

NBS Misc. Publ. 260-9 (November 1965).

COM74-I1065**

Yakowitz, H., Vieth, D. L., Heinrich, K. F. J., m

Michaelis, R. E., Standard Reference Materials: Homogeneity Characterization on NBS Spectre!

metric Standards II: Cartridge Brass and Low

Alloy Steel, NBS Misc. Publ. 260-10 (Decembet

1965). COM74-11064**

Napolitano, A., and Hawkins, E. G., Standard Reference Materials: Viscosity of Standard

Lead-Silica Glass, NBS Misc. Publ. 260-11 (November 1966). NBS Misc. Publ. 260-11**

Yakowitz, H., Vieth, D. L., and Michaelis, R. E Standard Reference Materials: Homogeneiti

Characterization of NBS Spectrometric Stand ards III: White Cast Iron and Stainless Stee Powder Compact, NBS Misc. Publ. 260-1! (September 1966). NBS Misc. Publ. 260-12**

Spijkerman, J. L., Snediker, D. K., Ruegg, F. C

and DeVoe, J. R., Standard Reference Mate

rials: Mossbauer Spectroscopy Standard for the Chemical Shift of Iron Compounds, NBS Misc, Publ. 260-13 (July 1967). NBS Misc. Pub

260-13**

Menis, O., and Sterling, J. T., Standard Reference Materials: Determination of Oxygen in Ferrous Materials - SRM 1090, 1091, and 1092, NBS

Misc. Publ. 260-14 (September 1966). NBS Misc Publ. 260-14**

Passaglia, E., and Shouse, P. J. Standard Refer ence Materials: Recommended Method of Usi of Standard Light-Sensitive Paper for Calibrat ing Carbon Arcs Used in Testing Textiles for Colorfastness to Light, NBS Misc. Publ. 260-1!! (June 1967). (Replaced by NBS Spec. Publj

260^1.)

Yakowitz, H., Michaelis, R. E., and Vieth, D. I] Standard Reference Materials: Homogeneity!

Characterization of NBS Spectrometric Stand- ards IV: Preparation and Microprobe Charac

terization of W-20% MO Alloy Fabricated b)J Powder Metallurgical Methods, NBS Spec

'

Publ. 260-16 (January 1969). COM74-11062** I Catanzaro, E. J., Champion, C. E., Garner, E. L„j Marinenko, G., Sappenfield, K. M., and Shields, W. R. Standard Reference Materials: Boric! Acid; Isotopic and Assay Standard ReferenccJ Materials, NBS Spec. Publ. 260-17 (February.

1970). Out of Print

IV
Geller, S. B., Mantek, P.A., and Cleveland, N.

G., Standard Reference Materials: Calibration of NBS Secondary Standard Magnetic Tape

(Computer Amplitude Reference) Using the

Reference Tape Amplitude Measurement

"Process A, "NBS Spec. Publ. 260-18 (November

1969). (See NBS Spec. Publ. 260-29.)

Paule, R. C, and Mandel, J., Standard Reference

Materials: Analysis of Interlaboratory Measurements on the Vapor Pressure of Gold (Certification of Standard Reference Material 745). NBS

Spec. Publ. 260-19 (January 1970). PB190071**

Paule, R. C, and Mandel, J., Standard Reference

Materials: Analysis of Interlaboratory Measurements on the Vapor Pressures of Cadmium and

Silver, NBS Spec. Publ. 260-21 (January 1971).

COM74-11359**

Yakowitz, H., Fiori, C. E., and Michaelis, R. E., Standard Reference Materials: Homogeneity

Characterization of Fe-3 Si Alloy, NBS Spec.

Publ. 260-22 (February 1971). COM74-1 1357**

Napolitano, A., and Hawkins, E. G., Standard

Reference Materials: Viscosity of a Standard

Borosilicate Glass, NBS Spec. Publ. 260-23

(December 1970). COM7 1-001 57**

Sappenfield, K. M., Marineko, G., and Hague, J. L., Standard Reference Materials: Comparison

of Redox Standards, NBS Spec. Publ. 260-24

(January 1972). COM72-50058**

Hicho, G. E., Yakowitz, H., Rasberry, S. D., and

Michaelis, R. E., Standard Reference Materials:

A Standard Reference Material Containing

Nominally Four Percent Austenite, NBS Spec.

Publ. 260-25 (February 1971). COM74-1 1356**

Martin, J. F., Standard Reference Materials:

National Bureau of Standards-US Steel Corportion Joint Program for Determining Oxygen and

Nitrogen in Steel, NBS Spec. Publ. 260-26

(February 1971). 85 cents* SN003-O03-O0786-2

Garner, E. L., Machlan, L. A., and Shields, W. R.,

Standard Reference Materials: Uranium

Isotopic Standard Reference materials, NBS

Spec. Publ. 260-27 (April 1971). COM74-

11358**

Heinrich, K. F. J., Myklebust, R. L., Rasberry, S. D., and Michaelis, R. E., Standard Reference

Materials: Preparation and Evaluation of

SRM's 481 and 482 Gold-Silver and GoldCopper Alloys for Microanalysis, NBS Spec.

Publ. 260-28 (August 1971). COM71-50365**

Geller, S. B., Standard Reference Materials: Calibration of NBS Secondary Standard Magnetic

Tape (Computer Amplitude Reference) Using

the Reference Tape Amplitude Measurement

"Process A-Model 2," NBS Spec. Publ. 260-29

(June 1971). COM7 1-50282

Gorozhanina, R. S., Freedman, A. Y., and

Shaievitch, A. B. (translated by M. C. Selby),

Standard Reference Materials: Standard

Samples Issued in the USSR (A Translation

from the Russian). NBS Spec. Publ. 260-30 (June

1971). COM71-50283**

Hust, J. G., and Sparks, L. L., Standard Reference

Materials: Thermal Conductivity of Electrolytic Iron SRM 734 from 4 to 300 K, NBS Spec. Publ.

260-31 (November 1971). COM7 1-50563**

Mavrodineanu, R., and Lazar, J. W., Standard

Reference Materials: Standard Quartz Cuvettes,

for High Accuracy Spectrophotometry, NBS

Spec. Publ. 260-32 (December 1973). 55 cents* SN003-003-0 1213-1

Wagner, H. L., Standard Reference Materials:

Comparison of Original and Supplemental

SRM 705, Narrow Molecular Weight Distri- bution Polystyrene, NBS Spec. Publ. 260-33

(May 1972). COM72-50526**

Sparks, L. L., and Hust, J. G., Standard Reference

Materials: Thermoelectric Voltage, NBS Spec.

Publ. 260-34, (April 1972). COM72-50371**

Sparks, L. L., and Hust, J. G., Standard Refer- ence Materials: Thermal Conductivity of

Austenitic Stainless Steel, SRM 735 from 5

to 280 K, NBS Spec. Publ. 260-35 (April 1972.)

35 cents* COM72-50368**

Cali, J. P., Mandel, J., Moore, L. J., and Young, D.

S., Standard Reference Materials: A Referee

Method for the Determination of Calcium in Serum, NBS SRM 915, NBS Spec. Publ. 260-36

(May 1972). COM72-50527**

Shultz, J. I. Bell., R. K. Rains, T. C, and Menis,

O., Standard Reference Materials: Methods of

Analysis of NBS Clay Standards, NBS Spec.

Publ. 260-37 (June 1972). COM72-50692**

Richmond, J. C, and Hsia, J. J., Standard Refer- ence Materials: Preparation and Calibration of

Standards of Spectral Specular Reflectance,

NBS Spec. Publ. 260-38 (May 1972). COM72-

50528**

Clark, A. F., Denson, V.A., Hust, J. G., and

Powell, R. L., Standard Reference Materials

The Eddy Current Decay Method for Resistivity

Characterization of High-Purity Metals, NBS

Spec. Publ. 260-39 (May 1972). COM72-50529**
McAdie, H. G., Garn, P.D., and Menis, O., Standard Reference Materials: Selection of Thermal

Analysis Temperature Standards Through a Cooperative Study (SRM 758, 759, 760), NBS

Spec. Publ. 260-40 (August 1972.) COM72-

50776**

Wood, L. A., and Shouse, P. J., Standard

Reference Materials: Use of Standard LightSensitive Paper for Calibrating Carbon Arcs

Used in Testing Textiles for Colorfastness to

Light, NBS Spec. Publ. 260^1 (August 1972)

COM72-50775**

Wagner, H. L. and Verdier, P. H., eds., Standard

Reference Materials: The Characterization of

Linear Polyethylene, SRM 1475, NBS Spec.

Publ. 260-42 (September 1972). COM72-

50944**

Yakowitz, H., Ruff, A. W., and Michaelis, R. E., Standard Reference Materials: Preparation and

Homogeneity Characterization of an Austenitic

Iron-Chromium-Nickel Alloy, NBS Spec. Publ.

260^3 (November 1972). COM73-50760**

Schooley, J. F., Soulen, R. J., Jr., and Evans, G. A.,

Jr., Standard Reference Materials: Preparation

and Use of Superconductive Fixed Point

Devices, SRM 767, NBS Spec. Publ. 260-44

(December 1972). COM73-50037**

Greifer, B., Maienthal, E. J. Rains, T. C, and

Rasberry, S. D., Standard Reference Materials:

Powdered Lead-Based Paint, SRM 1579, NBS

Spec. Publ. 260-45 (March 1973). COM73-

50226**

Hust, J. G., and Giarratano, P. J., Standard Refer- ence Materials: Thermal Conductivity and

Electrical Resistivity Standard Reference Materials: Austenitic Stainless Steel, SRM's 735 and

798, from 4 to 1200 k, NBS Spec. Publ. 260^6

(March 1975). SN003-OO3-0 1278-5

Hust, J. G., Standard Reference Materials: Electrical Resistivity of Electrolytic Iron, SRM 797,

and Austenitic Stainless Steel, SRM 798, from 5 to 280 K, NBS Spec. Publ. 260-47 (February

1974). COM74-50176**

Mangum, B. W., and Wise, J. A., Standard Refer- ence Materials: Description and Use of Precision

Thermometers for the Clinical Laboratory,

SRM 933 and SRM 934, NBS Spec. Publ. 260^8

(May 1974). 60 cents* SN003-O03-0 1278-5

Carpenter, B. S., and Reimer, G. M., Standard

Reference Materials Calibrated Glass Standards for Fission Track Use, NBS Spec. Publ.

260^9 (November 1974). SN003-003-0 1344-7

Hust, J. G., and Giarratano, P. J., Standard Refer- ence Materials: Thermal Conductivity and Electrical Resistivity Standard Reference Materials:

Electrolytic Iron, SRM's 734 and 797 from 4 to 1000 K, NBS Spec. Publ. 260-50 (June 1975).

$1.00* SN003-O03-0 1425-7

Mavrodineanu, R., and Baldwin, J. R., Standard

Reference Materials: Glass Filters As a Standard

Reference Material for Spectrophotometry;

Selection; Preparation; Certification; Use-SRM

930, NBS Spec. Publ. 260-51 (November 1975).

$1.90* SN003-003-01481-8

Hust, J. G., and Giarratano, P. J., Standard Refer- ence Materials: Thermal Conductivity and Electrical Resistivity Standard Reference Materials

730 and 799, from 4 to 3000 K, NBS Spec. Publ.

260-52 (September 1975). $1.05* SN003-003-

01464-8

Durst, R. A., Standard Reference Materials:

Standardization of pH Measurements, NBS

Spec. Publ. 260-53 (Under Revision).

Burke, R. W., and Mavrodineanu, R. Standard

Reference Materials: Certification and Use of

Acidic Potassium Dichromate Solutions as an

Ultraviolet Absorbance Standard, NBS Spec.

Publ. 260-54 (August 1977). $3.00* SN003-003-

01828-7

Ditmars, D. A., Cezairliyan, A., Ishihara, S., and

Douglas, T. B., Standard Reference Materials:

Enthalpy and Heat Capacity; Molybdenum

SRM 781, from 273 to 2800 K, NBS Spec. Publ. 260-55 (September 1977). $2.20* SN003-003-

01836-8

Powell, R. L., Sparks, L. L., and Hust, J. G.,

Standard Reference Materials: Standard

Thermocouple Materials, Pt.67: SRM 1967, NBS

Spec. Publ. 260-56 (February 1978). $2.20*

SN003-003-0 18864

Cali, J. P. and Plebanski, T., Guide to United

States Reference Materials, NBS Spec. Publ.

260-57 (February 1978). $2.20* SN003-003-

01883-0

Barnes, J. D., and Martin, G. M., Standard Refer- ence Materials: Polyester Film for Oxygen Gas

Transmission Measurements SRM 1470, NBS

Spec. Publ. 260-58 (June 1979) $2.00* SN003-

003-02077

Chang, T., and Kahn, A. H., Standard Reference

Materials: Electron Paramagentic Resonance

Intensity Standard: SRM 2601, NBS Spec

Publ. 260-59 (August 1978) $2.30* SN003

003-01975-5

vi
Velapoldi, R. A., Paule, R. C, Schaffer, R.,

Mandel, J., and Moody, J. R., Standard Refer- ence Materials: A Reference Method for the

Determination of Sodium in Serum, NBS Spec.

Publ. 260-60 (August 1978). $3.00* SN003-003-

01978-0

Verdier, P. H., and Wagner, H. L., Standard Refer- ence Materials: The Characterization of Linear

Polyethylene (SRM 1482, 1483, 1484), NBS

Spec. Publ. 260-61 (December 1978). $1.70*

SN003-003-02006-1

Soulen, R. J., and Dove, R. B., Standard Reference

Materials: Temperature Reference Standard

for Use Below 0.5 K (SRM 768), NBS Spec.

Publ. 260-62 (April 1979). $2.30* SN003-003-

02047-8

Velapoldi, R. A., Paule, R. C, Schaffer, R.,

Mandel, J., Machlan, J. L., and Gramlich, J. W.,

Standard Reference Materials: A Reference

Method for the Determination of Potassium in Serum, NBS Spec. Publ. 260-63 (May 1979).

$3.75* SN003-003-02068

Velapoldi, R. A., and Mielenz, K. D., Standard

Reference Materials: A Fluorescence Standard

Reference Material Quinine Sulfate Dihydrate

(SRM 936), NBS Spec. Publ. 260-64 (in press).

Marinenko, R. B., Heinrich, K. F. J., and Ruegg,

F. C, Standard Reference Materials: MicroHomogeneity Studies of NBS Standard Reference

Materials, NBS Research Materials, and Other

Related Samples, NBS Spec. Publ. 260-65

(September 1979). $3.50* SN003-003-02 114-1 Venable, W. H., Jr. and Eckerle, K. L., Standard

Reference Materials: Didymium Glass Filters for

Calibrating the Wavelength Scale of Spectrophotometers (SRM 2009, 2010, 2013), NBS Spec.

Publ. 260-66 (October 1979). $3.50* SN003-003-

02127-0

Velapoldi, R. A., Paule, R. C, Schaffer, R., Mandel,

J., Murphy, T. J., and Gramlich, J. W., Standard

Reference Materials: A Reference Method for the

Determination of Chloride in Serum, NBS Spec.

Publ. 260-67 (in press).

* Send order with remittance to: Superintendent

of Documents, US Government Printing

Office, Washington, DC 20402. Remittance

from foreign countries should include an

additional one-fourth of the purchase price for postage. ** May be ordered from: National Technical

Information Services (NTIS), Springfield,

Virginia 22151. v i
TABLE OF CONTENTS

PAGE

I. General 2

II. SRM Use and Summary 5

A. Use of the SRM Quinine Sulfate Dihydrate. . . 5

III. Fluorescence Standards 9

IV. Glossary 11

V. Experimental Section 12

A. Materials 12

B. Instrumentation 15

C. Instrument Calibration 20

D. Measurement Procedures 22

VI. Quinine Sulfate 25

A. Memory Effect 25

B. Purity 27

C. Stability 40

D. Oxygen Quenching 41

E. Polarization 42

F. Emission Spectrum as a Function of

Excitation Wavelength (Xi) 48

G. Photon Yields 50

H. Emission Spectrum as a Function of Acid ... 52

I. Emission Spectrum as a Function of QS

Concentration 53

J. Emission Spectrum as a Function of

Temperature 60

VII. Corrections for Instrumental and Experimental

Parameters 70

A. Photomultiplier Tube Linearity 70

B. Emission Spectrum as a Function of

Bandwidth 71

C. Wavelength Correction 73

D. Dispersion Correction 78

E. Solvent Refractive Index 79

F. Cuvette Window Tr ansmittance 80

G. Composite Correction 80

viii
TABLE OF CONTENTS (continued)

PAGE

S

VIII. The Spectral Responsivity M, of the Detection

A 2 rp System, the Technical Emission Spectrum E (A)

and the Molecular Spectrum E(A) of Quinine

Sulfate in 0.105 mol/L HCIO^ 84

A. Measurement and Averaging of Data 85

B. Data Smoothing 85

S

C. Spectral Responsivity M, 86

A 2 X s

D. Compensated Emission Spectrum (Y 7J 7 c, ) . . 93

j4 / . b , A 2 T

E. Technical Emission Spectrum E (A) of

Quinine Sulfate 93

F. Molecular Emission Spectrum E(A) 93

IX. Composite Errors and Precisions 101

A. Errors 101

B. Precision 106

References 107

Appendix A 116

Appendix B 120

ix
LIST OF TABLES

TABLE NO. PAGE

1. The Molecular Emission Spectrum E(A) of Quinine

Sulfate Dihydrate in 0.105 mol/L HClOi* 7

2. Organic Compounds Proposed as Fluorescence

Standards 10

3. Comparison of SRM Grade Quinine Sulfate with a Purified Quinine Sulfate Preparation Using

Thin Layer Chromatography 29

4. Equivalence of SRM Grade Quinine Sulfate and a Purified Quinine Sulfate Sample as Shown by

Measurements of Specific Weight Absorbances

W.A., Relative Photon Yields Q, and

Fluorescence Lifetimes x 32

5. Percent Water in SRM Quinine Sulfate* 2H 2 0 as

determined by Karl-Fischer Titration and

Weight Loss Procedures 34

6. Specific Weight Absorbances for Twenty Solutions

of Quinine Sulfate Dihydrate in 0.1 mol/L

HCIO^ at 365.0, 347.5, and 250.0 nm 35

7. Ratios of the Absorbances for Several Peaks of

Quinine Sulfate in Sulfuric and Perchloric

Acids 38

8. Specific Weight Absorbances of Quinine Sulfate

in Sulfuric and Perchloric Acids 39

9. The Dichroic Ratio of Emission D and the

Emission Anisotropy r of Quinine Sulfate in

0.1 mol/L HClOit as a Function of Quinine

Sulfate Concentration 43

10. Comparative Photon Yields Using Quinine Sulfate

as the Reference Material 50

11. Fluorescence Photon Yields Q and Lifetimes t of

Quinine Sulfate in Various Acids [88] 52

12. Average Percent Change Per Degree Celsius for

the Uncorrected Values (AY)^^^ g at 25 nm

Wavelength Intervals 61

x
LIST OF TABLES (continued)

TABLE NO. PAGE

13. Emission Peak Maxima and (Y)„ 0 £ Values for

Quinine Sulfate in 0.1 mol/L HCIO^ at

Various Temperatures 64

14. The Percent Change per Degree Celsius for the

Emission Spectrum of Quinine Sulfate in 0.1

mol/L HC10 4 at 25 °C at Selected Wavelengths. . 70

15. Quinine Sulfate Emission Values (Y) as a Function of Wavelength 76

16. Typical Intrarun Instrument Stabilities as

S Determined by (M, ) Values for the Standard

A 2 Lamp at the Two Operating Currents 87

17. Measured and Calculated Data 88

thru 90

18. The Technical and Molecular Emission Spectra of

Quinine Sulfate in the Various Radiometric 95

and Photon Units of E

T (A), E ( A) , E (A), E(v), $

and E (v) ? 96

p

19. The Average Relative Standard Errors (RSE) in

E(A) at Selected Signal Levels 99

20. Interpolated E (v) Values at Even Wavenumber

Intervals . ? 99

21. Percent Error Limits (PEL) at the 95 Percent

Confidence Level for the Spectral

Responsivity of the Detection System 102

22. Summary of Estimated Component Errors and Percent

T Systematic Error Limits in E (A) and E(A)

Values 104

xi
LIST OF FIGURES

FIGURE NO. PAGE

1. The NBS reference spectroradiomet er 17

2. Photograph of the NBS reference

spectroradiometer 18

3. Teletype output for the operation of the

reference spectroradiometer in the three

operating modes 20

4. Alignment of the calibrated vacuum tungsten

ribbon lamp using the alignment laser AL 2 ... 22

5. Time symmetrical measurement procedure used for

determining effect of temperature on the

emission spectrum 23

6. Time symmetrical measurement procedure used for

determining the spectral responsivity of the

detection system and the uncorrected emission

spectrum (Y) for quinine sulfate 24

7. The structure, fluorescence spectra, and relative

photon yields as a function of wavelength for

quinine in acid media 26

8. Emission anisotropy r (8a) and Dichroic ratio of

emission D (8b) as functions of wavelength for

quinine sulfate in 0.105 mol/L HC1CK 44

9. The sensitivity of the detection system to

radiation emitted by the calibrated tungsten

ribbon lamp that is parallel (P orientation)

and perpendicular (S orientation) to the

grating grooves in the detection

monochromator 47

10. The ratio of the uncorrected emission spectra

of quinine sulfate in 0.105 mol/L HCIO^

using exciting radiation at 347.5 and 366 nm

to demonstrate the 'red edge' shift 49

11. The ratio of the uncorrected emission spectra

of quinine sulfate in 0.1 mol/L HCICK and

0.5 mol/L H2SO4 to demonstrate the effect

of acid on the emission spectrum 54

xii
LIST OF FIGURES (continued)

FIGURE NO. PAGE

12. Relative fluorescence flux * and relative

fluorescence flux corrected for inner filter

effects (absorption of exciting radiation)

,

X as a function of quinine concentration. ... 57

13. Uncorrected emiss ion_ spectra for 1.3x10 6 mol/L

(1 ppm) and 1.3xl0~ 2 mol/L (10,000 ppm)

quinine sulfate in 0.105 mol/L HCIO^ 58

14. Ratio of the emission spectral data for various

concentrations of quinine sulfate 59

15. The geometrical representation of the sample

cell and illuminated volume element in

considering the distortion of the emission

spectrum by absorption of the emitted

radiation followed by re-emission ....... 60

16. The compensated fluorescence signal (Y) at

selected wavelengths as a function of

temperature 62

17. The values of the ratios of the compensated

fluorescence signals (Y) at various

temperatures to those at 23.8 °C as a function of wavelength 63

18. The values of the ratios of the normalized

compensated fluorescence signals (Y) at

various temperatures to those at 23.8 °C

as a function of wavelength 65

19. Standardized residuals for a fifth order

polynomial fit (19a) and a seventh order

polynomial fit (19b) for the ratios of the

normalized compensated fluorescence signals

(Y) at various wavelengths 66

20. The values of { [ (Y)

t

/ (Y) 23 g

]-l}10 2 at selected

wavelengths as a function of temperature. ... 68

21. The average percent error per °C observed in the

emission spectrum of quinine sulfate over the

temperature range 25±5 °C 69

xiii
LIST OF FIGURES (continued)

FIGURE NO. PAGE

22. Relative photomult iplier tube linearity

correction factors for amplifier gains of

10 7 (22a) and 10 6 (22b) volts/amp applied to

the compensated fluorescence signal (Y) .... 72

23. Bandwidth correction in compensated fluorescence

units (Y) (23a) and the relative correction

factor applied to the quinine sulfate emission

(23b) as a function of wavelength 74

24. Wavelength deviations of the emission

monochromator at 34 wavelengths using line

sources 75

25. Relative correction factor applied to the

compensated fluorescence signal (Y) to

correct for the wavelength error 78

26. Relative correction factor applied to the

emission spectrum to compensate for errors

introduced by the refractive index of the

solvent ...... 81

27. Relative correction factor applied to the

emission spectrum to compensate for the

variation of cell transmit tance as a function of wavelength 82

28. Composite errors that occur in emission spectra

measurements (28a) and the composite

correction factors applied to the emission

spectrum of quinine sulfate to correct for

various instrumental and sample parameters

(28b) 83

S 29. The spectral responsivity (M^ ) for the

detection system of the NBS reference

spectroradiometer 91

30. The relative standard error in the values for

spectral responsivity of the detection

system 92

31. The compensated fluorescence emission spectrum

(Y) and the corrected emission spectrum E(A)

for quinine sulfate 97

xiv
LIST OF FIGURES (continued)

FIGURE NO. PAGE

32. The relative standard errors in the values for

the corrected emission spectrum E(A) of

quinine sulfate 98

33. The corrected emission spectra of quinine

sulfate in E(A) and E^Cv) units 100

34. Percent estimated errors in the reported values

at the 95 percent confidence level due to the

calibration of the standard lamp, the 0.2 nm

uncertainty in the emission peak position,

and the errors estimated at 30 percent of the

individual corrections for systematic error . . 103

35. The percent systematic error limits in the

values for the corrected emission spectrum

of quinine sulfate in 0.105 mol/L HClOt* at

the 95 percent confidence level 105

xv
". . . The solution of quinine 3 though it appears to be

perfectly transparent and colourless 3 like water, when

viewed by transmitted light 3 exhibits nevertheless in

certain aspects 3 and under certain incidences of the light,

a beautiful celestial blue colour. ..."

Sir George Gabriel Stokes, 1852 [l] 1

Figures in brackets indicate references at the end of this

paper

xvi
Abstract

The need, material selection, characterization, certification, and uses of the fluorescence Standard Reference

Material, quinine sulfate dihydrate, have been discussed.

The emission spectrum for quinine sulfate dihydrate in 0.1

mol/L HCIO^ has been measured on the NBS reference spectroradiometer. The spectrum has been reported every 5 nm from

375.0 to 675.0 nm as the relative technical emission spectrum

T

E (X) and the relative molecular emission spectra represented

by E(A), E (A) , E(v), and E^ (v) . The technical emission

spectrum has been corrected for the instrumental parameters

of the spectral responsitivity of the detection system,

photomult iplier tube non-linearity, bandpass, monochromator

wavelength error, and further corrections for the sample

parameters of solvent refractive index and cell window

transmi ttance were applied to obtain the molecular emission

spectra. Estimates of the 95 percent systematic error limits

of the emission spectrum values have been made as a function

of wavelength. These values are ^2 percent at the peak maximum 0455 nm) and ^5 percent and ^3 percent at the blue (395

nm) and red (565 nm) one -tenth power points. The experimental

replication precision is ^0.2 percent over the total spectral

range. The purity and the stability of the SRM quinine sulfate

dihydrate and the effect of solute concentration, solvent acid,

acid concentration, excitation wavelength, oxygen quenching,

temperature, and polarized exciting radiation on the emission

spectrum, and in part, on the absorbance spectrum, photon

yield, and fluorescence lifetime also have been discussed.

Key Words: Calibration of spectrofluor imeters ; corrected

fluorescence spectra; fluorescence; fluorescence

lifetimes; fluorescence standards; molecular

emission spectrum; photon yields; quinine sulfate

dihydrate; spectrofluorimetry ; Standard Reference

Material; technical emission spectrum; transfer

standards

1
I. General

The photoluminescence of molecules in solution was noted

by Monaides in the early 1500 's [2]. In 1852 Stokes correctly

described this phenomenon and named it fluorescence [1] . It

was not until 1955, however, when Bowman, Caulfield, and

Udenfriend published a paper [3] describing a two monochromator, photomultiplier equipped instrument that the

fluorescence spectra of small quantities of materials could

be measured easily. Since that time, the use of the fluorescence technique in diverse areas such as health, environmental

pollution, molecular biology, forensic sciences, species

identification and quantitation, etc. , has increased significantly [see for example, ref. 4].

Laboratories can report fluorescence spectra that are

either uncorrected or corrected for instrument and sample

parameters. The uncorrected spectra, although useful for

intralaboratory measurements of changes in fluorescence

characteristics, should not be used for interlaboratory

comparisons since the correction factors for each instrument

can be quite different.

Corrected spectra may be determined in two ways. First,

in absolute radiometric units, and second, by measuring the

spectrum for the compound under investigation relative to

that obtained for a standard on the same instrument under

similar experimental conditions. Very few laboratories are

equipped to make absolute radiant energy measurements; however,

comparative spectra obtained from standards run under the same

experimental conditions on the same instrument can be done

easily. This latter method, which effectively corrects for

instrument parameters that affect the fluorescence spectra,

is used most widely in practice but is severely limited by

the lack of reliable standards. Thus, spectra corrected in

this manner by different laboratories cannot be compared with

2
confidence. This is demonstrated by the large discrepancies

in the spectra obtained by different authors for the same

substance. For example, the values reported for the corrected

emission spectrum of quinine sulfate varied by as much as

50 percent [5- 8]

This variability underlined the need for standardization

of fluorescence measurements and data presentation [9-11]

and in 1963 several investigators collectively published a paper entitled "Proposal for Standardization of Methods of

Reporting Fluorescence Emission Spectra" [12]. This proposal

established a firm basis for the intercomparisons of data

among laboratories, but it did not fulfill the need for

fluorescence standards with certified values for the emission

and excitation spectra.

The National Institute for General Medical Sciences,

National Institutes of Health, recognizing this need and

seeing the impact on clinical and biochemical applications

of a similar program in spectrophotometry [13], provided

partial funding for the development of fluorescence Standard

Reference Materials (SRM's) by the Center for Analytical

Chemistry of the National Bureau of Standards.

The various types of SRM's available from NBS have been

discussed in several publications [14-18], These SRM's are

well characterized, stable materials that are produced in

quantity and are available for purchase over a long time

period (6-10 years). Their functions are to: (a) help

develop reference methods of analyses or tests; (b) calibrate

measurement systems; and (c) assure long term adequacy and

integrity of a quality control process [14]. SRM's are

available for uses ranging from the steel industry to clinical

laboratories and include metals as well as organic and

inorganic compounds. Meinke [18] discussed the importance

of SRM's in Clinical Chemistry, and initiated work at NBS

with the objective of providing fluorescence SRM's.

3
Because of its wide applicability and additional requirements that will be discussed later, the first fluorescence SRM

developed is quinine sulfate. This SRM was developed primarily

for the calibration of the emission detection systems of

luminescence spectrometers. If necessary, the excitation

spectrum of quinine sulfate and absolute values of the

spectral photon yield could be determined at a later time.

In the development of this SRM, the four categories of

need, material selection, material characterization, and

emission spectrum certification were considered. The first,

need, has already been discussed. The second, SRM selection,

depends on the prerequisites and imposed constraints. The

third, characterization, compares measured values for photophysical and chemical properties to those of purified materials. Lastly, certification, includes correction of the

emission spectrum for instrument and sample parameters so

that the corrected emission spectrum is independent of these

parameters. Since this study addresses many facets of

fluorescence standards, a summary of this work and directions

for the use and storage of SRM 936 Quinine Sulfate Dihydrate

are presented first in Section II. Additional information

on the measurement of the physical and chemical properties

and certification of this SRM is included in later sections.

In Section III we present a brief discussion of the

suitability of organic and inorganic fluorophors in various

matrices as fluorescence standards. Section IV is a glossary

of terms and Section V is a general experimental section;

however, where relevant and necessary, experimental conditions

or data handling techniques are given in each specific section.

In Section VI we present results of studies on absorbance and

fluorescence parameters of quinine sulfate under various

experimental conditions with the objective of further characterizing this material. In this way we hope to provide

information on the use of quinine sulfate as a standard and,

4
more importantly, to make the user aware of the errors,

associated with making measurements of corrected luminescence

spectra. The errors introduced into the emission spectrum

by various experimental parameters are discussed in Section

VII. In Section VIII we present the values for the normalized, corrected emission spectrum of SRM grade quinine

sulfate dihydrate in 0.105 mol/L HCIO4 followed by a discussion of the precision and accuracies of these values in

Section IX.

I II. SRM Use and Summary

A. Use of the SRM Quinine Sulfate Dihydrate

The Certificate issued with SRM 936 is reproduced in

Appendix A. To assure the most reliable and accurate

results, all glassware used in solution preparation should

be "class A" or equivalent. The distilled water used to

prepare the 0.105 mol/L HCIC^ should have a negligible signal

due to fluorescence under the given experimental conditions.

The molecular weight of^-quinine sulfate dihydrate,

(C2oH2 lfN202) 2 *H 2 S0 lt *2H 2 0, is 782 . 947 3

.

1 . Solution Preparation

Prepare a 1.28x10 6 mol/L Ol ppm) solution of quinine

sulfate dihydrate (QS'2H 2 0) in 0.105 mol/L HCICU in the

following manner:

(a) weigh using an analytical balance 0.100 g QS»2H 2 0

and quantitatively transfer it to a 1000 mL

volumetric flask;

(b) dilute to the mark with 0.105 mol/L HCIO4. giving

a solution 1.28x10"" mol/L (100 ppm) in QS-2H 2 0;

3 Atomic weights obtained from Pure and Applied Chem. 47,

75-95 (1976). —

5
(c) Transfer 10 mL of solution (b) to a 1000 mL flask

and dilute to the mark using 0.105 mol/L HClOi* to

give a final solution that is 1.28xl0~ 6 mol/L

(1 ppm) solution of QS»2H 2 0.

Store solution (b) and (c) in the dark. Solution (c)

should be prepared fresh monthly while (b) should be prepared

fresh every three months. For critical measurements, all

solutions should be prepared fresh from the solid material

that has been stored in the dark.

2. Instrumental Conditions

Generally, the instrumental conditions used when recording emission spectra are variable and do affect the spectral

distribution of the emitted radiation. Thus, it is recommended to measure the emission spectrum under the following

instrumental and experimental conditions

(a) solution temperature: 25.0 °C;

(b) excitation wavelength A i : 347.5 nm;

(c) bandpass of excitation and emission monochromator

: 5.3 nm; and

(d) geometry: 90° viewing, "right angle".

If wide deviations from the above conditions occur, it

is recommended that the user refer to the specific section

in this report to determine the appropriate corrections to

be applied to the emission spectrum.

After obtaining the emission spectrum of 1.28x10 6 mol/L

QS«2H 2 0 in 0.105 mol/L HCIO^ under the above conditions, the

correction function for your instrument over this wavelength

region may be determined by:

S

=

where S is the instrumental response,

is the instrumental reading at A, and

E(A) are the values from Table 1.

6
Table 1. The Molecular Emission Spectrum E(A) of Quinine

Sulfate Dihydrate in 0.105 mol/L HC1CK .

A, nm E (A) A , nm

375.0 0.005 475.0

380.0 .012 480.0

385.0 .028 485.0

390.0 .057 490.0

395.0 .103 495.0

400.0 .170 500.0

405.0 .257 505.0

410.0 .359 510.0

415.0 .471 515.0

420.0 .586 520.0

425.0 .694 525.0

430.0 .792 530.0

435.0 .874 535.0

440.0 .940 540.0

445.0 .984 545.0

450.0 .999 550.0

455.0 .997 555.0

460.0 .982 560.0

465.0 .947 565.0

470.0 .897 570.0

E (A) A, nm E (A)

0.838 575.0 0.076

.782 580.0 .065

.719 585.0 .057

.659 590.0 .050

.595 595.0 .043

.541 600.0 .037

.486 605.0 .032

.434 610.0 .028

.386 615.0 .024

.342 620.0 .021

.302 625.0 • .018

.264 630.0 .016

.231 635.0 .014

.201 640.0 .011

.175 645.0 .010

.153 650.0 .009

.132 655.0 .008

.116 660.0 .007

.101 665.0 .006

.008 670.0 .005

675.0 .004

7
The corrected spectra for other fluorescent materials

that emit in the same general spectral region may thus be

determined by dividing the R values at particular wavelengths by the instrument response function S determined

from the above equation at the corresponding wavelength.

It must be emphasized that this procedure can be used

for materials with the same spectral characteristics as quinine sulfate. If the spectral characteristics are quite

different, e.g., if the emission peak half-height bandwidth

is much smaller (10 nm rather than 90 nm) the monochromato

bandpass must be decreased accordingly.

3 . Summary

The need, material selection, characterization, and

certification of the fluorescence Standard Reference Material

Quinine Sulfate Dihydrate are discussed in Sections III-X.

The emission spectrum for quinine sulfate dihydrate in 0.1

mol/L HClOit has been measured on the NBS reference spectroradiometer. The spectrum has been reported every 5 nm from

375.0 to 675.0 nm as the relative technical emission spectrum

T

E (A) and the relative molecular emission spectra represented

by E(A), E (A), E(v), and E (v) . The technical emission

spectrum is calculated by correcting the raw emission data

for the instrumental parameters of the spectral responsitivity

of the detection system, photomul tiplier tube non-linearity,

bandpass, monochromator wavelength error; further corrections

for the sample parameters of solvent refractive index and

T cell window transmit tance are applied to E (A) to obtain the

molecular emission spectra. Estimates of the 95 percent

systematic error limits of the emission spectrum values are

made as a function of wavelength. These values are ^2 percent

at the peak maximum (^455 nm) and ^5 percent and ^3 percent

at the blue (395 nm) and red (565 nm) one-tenth power points.

The experimental replication precision is ^0.2 percent over

the total spectral range. The purity and the stability of

8
the SRM quinine sulfate dihydrate and the effect of solute

concentration, solvent acid, acid concentration, excitation

wavelength, oxygen quenching, temperature, and polarized

exciting radiation on the emission spectrum, and in part, on

the absorbance spectrum, photon yield, and fluorescence

lifetime also are discussed.

III. Fluorescence Standards

The requirements for an ideal fluorescence standard are

that it should: (a) have 'broad' fluorescence spectra;

(b) be easily purifiable; (c) be stable in solution or as

the solid; (d) have little overlap between the excitation

(absorbance) and emission spectra; (e) not be subject to

oxygen quenching; (f) have a constant quantum efficiency as

a function of exciting wavelength; (g) have isotropic emission; (h) have the same emission spectral shape independent

of exciting wavelength; (i) be soluble in aqueous and organic

solvents; and (j) absorb and emit in the same general regions

as the compound under study.

No single compound exhibits all these characteristics.

In fact, the last requirement precludes the use of a single

standard, since the emission spectra of most fluorophors are

limited to relatively small wavelength intervals. For this

reason, it is necessary to have available a series of fluorescence standards covering the wavelength range of interest.

With the increased use of lasers and interest in radiative

measurements in the near - infrared , this range now extends

from 250 nm to 1100 nm.

Many types of materials have been proposed or used as

fluorescence standards including organic molecules in solvents

or plastics, scintillators in radioactive solvents, and

inorganic ions in glass and crystal matrices [7,9,11,19-32].

Although we investigated solutions of organic molecules and

inorganic ions in glass and crystalline matrices [28,33-36]

9
as possible candidates for SRMs , only organic molecules in

aqueous based solvents were given in-depth consideration

because: (a) they can be used in both static and flow systems;

(b) a material soluble in an aqueous based solvent was desired;

(c) individual calibration is not required (statistical sample

testing permissible) ; and (d) quality control of the material

is simplified. Organic molecules in plastic are undesirable

since they exhibit large emission anisotropies [37] , oxygen

diffusion and quenching of fluorescence may be variable,

stability of the plastic matrix is questionable, and surface

scratching may occur.

Several organic compounds proposed as standards are

listed in Table 2.

Table 2. Organic compounds proposed as fluorescence standards.

Compound Emission Maximaa

, nm

3-naphthol 354 , 402

anthracencec '

384, 404, 428, 454

c ,d pyrene ' 390

c d diphenylanthracene * 408 , 427 , 458 e

, 490 e quinine sulfate 455

3- aminophthal imide 506 i fluorescein^ 518

m- dime thy laminonitrobenzene 542

rhodamine 6G 557

rhodamine B 572 i

! aluminum (III)- PBBR chelate f 635

methylene blue ^708

4- dimethyl aminonitros t ilbene 742

i

1

a The maxima reported here are approximate.

c Unstable in solution.

d Narrow emission or excitation spectra.

Oxygen quenched.

£ Shoulder.

PBBR = pontachrome blue black R.

10
Considering their characteristics listed in table 1,

most of these materials were eliminated as potential SRMs

Of the remaining materials, only quinine sulfate in an

aqueous-acidic solvent had been extensively studied and used

as a fluorescence standard. Based on the requirements and

preliminary investigations [33] quinine sulfate was chosen

as the first material to be issued as a fluorescence SRM.

With the recent advent of dye lasers, and research of

suitable laser dyes, several additional compounds have been

suggested as fluorescence standards [38] including coumarins,

oxazines, and two carbocyanines (hexamethyl indodi -, and

hexamethylindotricarbocyanine) . In general, these materials

have the best fluorescence characteristics when dissolved in

non-aqueous solvents and should certainly be investigated as

standards for use in organic solvents.

IV, Glossary

In general, the nomenclature followed in this publication will be a combination of that used by Melhuish [39],

Mielenz [40], and Winefordner [41]. Selected terms and

definitions used in this text are as follows:

P S' , = Compensated fluorescence signal obtained when

1

' 2 instrument is operated in the ratio mode (see

Section V-B2). The ls_t superscript refers to the

orientation of the excitation polarizer, (polarizer) , the 2nd to the orientation of the emission

polarizer (analyzer) where P and S represent

geometrical orientations perpendicular and parallel

to the grating grooves of the detection monochromator. The letter X denotes the absence of a polarizer. The subscripts Xi and X 2 refer to the

excitation and emission wavelengths, respectively.

PS Where convenient, Y '

. will be abbreviated (Y)

A i , A 2

11
c

R, = Emission detector signal obtained for the instrument

2 operated in the detection or 1 B" mode, usually in

conjunction with values obtained by placing the

standard lamp in the sample position, S, (see figure

1 in Section V-B2) . The superscript and subscript

have the same meanings as previously defined.

c M, = Spectral responsivity of the emission detection

2 system. This includes the ellipsoidal mirror

EL2, the analyzer, the detection monochromator

,

and the detection photomultiplier tube (see

figure 1)

T

E (A) = The emission spectrum in radiometric and wavelength

units corrected for instrumental parameters.

E(A) = The emission spectrum in radiometric and wavelength

units corrected for instrumental and sample

parameters

E (A) = The emission spectrum in photon and wavelength units

P corrected for instrumental and sample parameters.

E(v) = The emission spectrum in radiometric and wavenumber

units corrected for instrumental and sample

parameters

E (v) = The emission spectrum in photon and wavenumber

P units corrected for instrumental and sample

parameters

V. Experimental Section: Materials, Instrumentation,

and Procedures [42]

A . Materials

1. Reagents

The following reagents were used in this study:

Perchloric Acid : National Bureau of Standards, purified

reagent, double distilled (non-ebullient distillation process)

lot 111103 [43]

Sulfuric Acid : National Bureau of Standards, purified

reagent, double distilled (non-ebullient distillation

12
process) lot 219 [43] and ultrex grade, J. T. Baker Chemical

Company, Phillipsburg , New Jersey.

Distilled Water : Three types of distilled water were

produced and tested: (a) distilled water was passed through

an ion exchange column (IWT Research Column, Rockford,

Illinois) followed by double distillation from a quartz still;

(b) distilled water was redistilled from a hot potassium

permanganate - NaOH mixture; and (c) distilled water was

redistilled by the NBS non-ebullient distillation process

[44]. The three water samples, upon excitation at 347.5 nm

under high sensitivity detection conditions, gave a small

peak at ^395 nm attributable to the Raman spectrum of the

solvent [45]. The signal at this wavelength was ^1 percent

of a 1 ppm quinine sulfate signal (ym „ v at A 2m = 100) . At

max max

other wavelengths, however, the signals attributable to the

blank were several orders of magnitude less than those of

the sample. The water discussed under (a) or (c) above was

used to prepare the various acid concentrations needed in

these studies.

Dilute Acids : Dilute perchloric acid was prepared by

mixing appropriate volumes of concentrated perchloric acid

and water. A standard solution of potassium acid phthalate

(NBS SRM 84H) was used to determine the concentration of a sodium hydroxide solution (J. T. Baker, Analyzed Reagent,

M g/L) . The concentration of the dilute perchloric acid

used in the certification process was 0 . 1049±0 . 0001 mol/L.

A similar process was used to determine the concentration of the various sulfuric and perchloric acid concentrations used in these studies.

Ethylene Glycol : Fisher Chemical Company, Fairlawn,

New Jersey, "Certified Reagent" used as received.

Rhodamine B : Exciton Chemical Company, Inc., Dayton,

Ohio. (Laser grade, Rhodamine 610), used as received.

13
Solvents : All solvents used including chloroform,

methanol, acetic acid, n-propanol and formic acid were of

reagent grade quality.

2. Quinine Sulfate

(a) General : Samples of quinine or its derivatives

were obtained from various suppliers and purified by several

recrystallizations from warm water or warm ethanol [33].

(b) SRM Grade : A sample of the bulk, purified, analyzed

quinine sulfate obtained from the J. T. Baker Company, Phillipsburg, New Jersey (Appendix B) was compared to purified

samples of quinine sulfate by measuring various optical and

chemical properties (Section V-A2b) . Without further purification, the bulk material was bottled in ^1.1 g quantities

in amber, screw-capped bottles that were numbered consecutively from 1-500. Ten random numbered samples (numbers 27,

96, 122, 177, 230, 284, 311, 370, 421, and 453) were chosen

for testing of homogeneity and subsequent certification

measurements

(c) Solution Preparation : Samples of quinine sulfate

dihydrate from the ten randomly chosen vials were weighed to

±0.00001 g in glass weighing containers on a calibrated

[46a] semi-micro analytical balance. No changes in weight

due to water adsorption were noted. The solid samples were

rinsed quantitatively with 0.105 mol/L HC10i+ into previously

weighed 100-mL class "A" volumetric flasks. Solvent was

added to the calibrated mark, the flasks were tightly stoppered using polyfluoroethylene stoppers, and the flasks plus

solutions were weighed on either the semi-micro balance or

on a calibrated top loading analytical balance (to 0.01 g,

[46b]) to determine by difference the weight of solvent added

and thus, the weight concentration of the quinine sulfate

dihydrate. All subsequent dilutions of these quinine - stock

solutions were made by weight. Buoyancy corrections were

made for the differences in densities of the quinine sulfate

14
dihydrate, water, and "brass" weights. The stock solutions

were ^1 x 10 3 mol/L in quinine sulfate dihydrate. A fiftyfold dilution was made to prepare ^2 x 10 5 mol/L (20 ppm)

quinine sulfate solutions for absorbance measurements. A

further 15-fold dilution was made to prepare ^1.3 x 10 5 mol/L (1 ppm) quinine sulfate solutions for fluorescence

certification measurements.

For certification, one series of ten quinine sulfate

solutions was prepared in this manner and the emission spectra

were measured during week one. A second series of solutions

was prepared during week three by mixing the same ten sample

vials, choosing the vials randomly, weighing the sample,

diluting, and then, as before, the emission spectra were

measured. The emission spectra of the solutions in series

one were remeasured during week six.

B . Instrumentation

1. General

Absorbance values were obtained using Cary 14 and 16

spectrophotometers and a NBS high accuracy spectrophotometer

[47] . The absorbance scales for the Gary 14 and 16 spectrophotometers were calibrated using NBS SRM 930b, Glass Filters

for Spectrophotometry [48] . Comparative and certified fluorescence data were obtained on the NBS Reference Spectroradiometer [49] (hereafter called Reference Spectrof luorimeter)

and relative quantum yields were measured on a Turner '210'

Corrected Spectrofluorimeter [50] . The wavelength scale of

the Turner '210' was calibrated as previously described [33].

The fluorescence lifetimes were measured using a TRW

Model 75A Decaytime Fluorimeter System or an Ortec Model

9200 Nanosecond Single - Photon Photometer.

Matched spectrophotometer cells of known pathlength

[51] were used for absorbance measurements and a single

unblackened quartz spectrofluorimeter cell was used for all

fluorescence certification measurements.

15
All instruments except the Ortec system were equipped

with constant temperature cell blocks for sample temperature

control to ±0.1 °C. The temperature in the cuvette was

measured relative to the bath temperature with a thermocouple

type probe (YSI Co., Yellow Springs, Ohio) that was calibrated j against a NBS platinum resistance thermometer. All spectra

were measured at 25.0±0.1 °C unless otherwise noted.

2. NBS Reference Spectrof luor imeter

Although detailed accounts of the NBS reference spectrofluorimeter [49] and its calibration [52] are being written,

a brief description of it will be given here. A scale drawing

of the spectrof luor imeter in the right-angle fluorescence mode

and an overall photograph of the instrument are presented in

figures 1 and 2, respectively.

Excitation radiation is produced by a 450 W xenon source

(XL), directed by an ellipsoidal mirror (ELi) to a 0.3 m

Czerny-Turner crossed beam grating monochromator (blaze at

300 nm) equipped with a prism predisperser (PD) , stepping

motor (SM) , and absolute shaft encoder (SE) . The slits were

2.0 mm wide (bandpass 5.3 nm) unless noted otherwise. The

selected excitation radiation is focussed on the sample (S)

by a Cassegrain system (CS, 1:1 magnification, [53]). A

small portion of the exciting radiation is reflected by a beam splitter (BS) at a 30° enclosed angle and focussed at

the reference detector consisting of a quantum counter (8 g/L

1 rhodamine B in ethylene glycol [54], cell at 15° to the face

of the photomult iplier tube) (QC) and an EMI 9659QB extended

red photomult iplier tube (RD) . A Glan-Thompson polarizer can

|

be placed directly before the sample (S) and a second GlanThompson polarizer, the analyzer (P) , can be placed between

the ellipsoidal mirror (EL 2 ) and the detection monochromator.

Both polarizers were calibrated to an absolute angular orientation of 0.1° with a Rudolph Research Thin Film Ellipsometer

Model 43702-200E using a helium-neon laser as a source [55].
g i o3

p

CD P -H 03 ,0 O £ i_> p o Pi

CO fn rH

g

PQ •H 03 i • ~ o3 CJ fn rH . IS H CD

CD iH

O +-) rH 03 rH . ~ +_> o m t3 o fH 3 pj m •H XI O rC CD

aj

O 03 P CO

X o co P U i i 1 P 03 pq •H ftPM X

rH .1 CD • *\ rH < H") Jh CD Ph fH

CD 1 0 CD o

t/) o3 pq f-i

p o ,—

CD CD

U

H • - tsl fn P

CD -H CD CD

d*—

fn 5h

m 1—,

CD P o3 CD P o3

+-> rH fn <4H (j

C2f—

fH

O i o3

H

bO CD QhQ r& fn •H P I Pi CO CD r—

03 Ph i nj P.

CD fn CO (/) r—(

rH ^ CD l

*H

X

u +-> "• Ph 3 03 P fH CO O fH 3 o

T3 +->O P

5h

u

u fH CD < rH CD O Q CD 6 P P

CD 4-> 3 CD o • CD e o S 6 PJ •H CD O 03 rH bo "d p -H 3 03 P w p cr p •H x o i bO P o CO bOU -H P. 1 C/ to CD P H 1 P

(J •H CJ ••>(=) CO CD 03 fH CO i p. fH CD 2

CO bO bO (/) ••> CO

CD !=! fn CD

CD C/5 -H CD rH

CO -M

p 03 o3 co 6 6

CD

U fH •H 03 o3 P i bOT3 co rH

CD CO i CD i m U CD fH CO rH

CD 03 p 1 • •» fH

fH fH

Q CD P

co CD CD Ph bo

u

PQ P +-> 03 CD P rH • «n-H Ph •H -H CO fH CO CD rH MH fH fH 1 Ph I O Cti PI

E-i CO Ph fH

U CO

CDPP

bO H

Ph 17
t/1

CD

(/>

X

O• H

03

o3

5hbOO

+->

M

O

•h x:- u P. x

CD <d

CD 4->

E-h <-HO

CD P

CD6o

•HUO

4-> U

CD

P-. in

CD T3

•H

l/i

bOO

CD -H UC CD

CD "13 5i -H

CD m

CD

5-i

CO

PQ

CD

+->

4-HO

c/)

PSo

•H

</)

(/) - He

nd

PS

oj

+->

X <4H CU CD Oj i—

?H00 CD

+->

PiO

CDU

bO

18
The emitted radiation is focussed by the ellipsoidal

mirror EL 2 at the slit of the detection monochromator which

is identical to the excitation monochromator except that the

grating is blazed at 500 nm. The detection photomultiplier

is an EMI 9659QB extended red photomultiplier tube (SD)

,

thermoelectrically cooled to -12 °C. The instrument uses

analog detection in conjunction with a d.c. amplifier and a digital voltmeter. The instrument is interfaced to a digital computer. Shaft encoders (0.005 nm angular resolution)

are used to supply digitized wavelengths of the source

monochromator (A.i) and the detection monochromator (A2) to

the digital computer. The instrument can be operated in

single mode ("B mode" for obtaining emission spectrum, "A

mode" for obtaining an excitation spectrum) or ratio mode

("B mode" signal/reference signal). In single mode operation,

the signal is sampled one hundred times in 10 s and the values

are relayed to the computer where the average and the standard

error are calculated. In "R mode" operation (R=B/A) using

the reference detector, the emission signal (B) is sampled

once, then the reference signal (A) is sampled once, the

values are relayed to the computer where the ratio R is

calculated and the values B, A and R are stored. This procedure is repeated 50 times in 10 s after which B, A, R, and

the standard error in R are calculated. These values and

additional parameters are transmitted to a teletype or plotter.

An example of the teletype output is given in figure 3. The

'X' column represents the operating mode; 'N' represents the

number of points (or ratios) measured; 'A'; 'B'; 'R' the

computed averages; S.E.(X) the standard error of the value

designated in the 'X' column, and WVL(EX) and WVL(EM) the

excitation and emission wavelengths. After data are taken

at a particular position, the scanning monochromator is

changed by a preset wavelength interval (either to longer or

shorter wavelengths) and the measurement process repeated

until the total spectrum is recorded. The raw data are stored

19
on magnetic tape and transferred to the NBS Univac 1108

computer where data reduction and analyses are performed.

RUN 9023

X N A B R S.E.CX) VVL(EX) WVL ( EM

)

R 49 1.014757 .708699 . 698447 .188D-02 440.000 530.000

B 100 .704714 .991D-03 440.000 530.000

A 100 1.013186 .824D-03 440-000 530.000

END OF RUN

Figure 3. Teletype output for the operation of the reference

spectroradiometer in the three operating modes:

R-ratio, B-emission; A- excitation

The programs used for data reduction, statistical analysis, curve fitting, and smoothing were written in 0MNITAB,

a language developed at NBS that is well suited for manipulating columnar data [56]

C . Instrument Calibration

1. Wavelength

Both monochromators were calibrated for wavelength accuracy over the 202.5 to 894.4 nm wavelength range by measuring

37 lines from mercury, zinc, helium, cadmium, neon, cesium,

and rubidium low-pressure discharge sources. For calibration

of the detection monochromator , the sources were placed at

the sample position or attached to the detection monochromator.

For calibration of the excitation monochromator, the sources

were placed at the xenon lamp position or attached to the

excitation monochromator and a mirror was placed in the sample 1 position. The monochromators were stepped in ascending and

descending 0.005 nm wavelength intervals over the line position and the wavelength values obtained at the maximum for

ascending and descending wavelength intervals were averaged.

20
The differences between the average measured values and the

true values [57] were determined.

2. Detection System

The spectral respons ivity , M, , of the emission detection

system (ellipsoidal mirror EL 2 , analyzer, detection monochromator, and detection photomult iplier) was determined in

the following manner (refer to figure 1) . The detection

photomultiplier was replaced by the alignment laser AL 2 located ^15 cm from the exit slit of the monochromator . Masks

with central holes of ^5 mm diameter were placed over the

paraboloidal mirrors in both monochromators . Both alignment

lasers (AL 2 and ALi) were aligned such that the two beams

intersected at 90° at the sample position 30.5 cm (12 inch)

above the table.

A NBS calibrated vacuum tungsten ribbon lamp #A7 [58]

was placed in the sample position such that the flat portion

of the tungsten ribbon was perpendicular to the beam from

AL 2 and the alignment notch in the ribbon was on the optic

axis, figure 4. The position of the lamp was adjusted until

the beam from AL 2 just passed through the notch in the

ribbon and contacted the point of an arrow inscribed on the

quartz envelope of the lamp. At this point in the alignment

procedure, the beam from ALi was vertically bisected by the

edge of the tungsten ribbon. The masks were removed and the

lamp was activated and adjusted to an operating current of

^8A. A pin diode was placed at the exit slit of the detection monochromator without disturbing AL 2 and the signal

maximized using only the horizontal adjustment of EL 2 . The

pin diode was removed and the lamp current reduced to ^3A.

The beam from AL 2 impinged on the tungsten ribbon as in

figure 4. This process was repeated several times with the

same results and the lamp was considered to be aligned. The

spectra of the lamp at two operating currents were recorded

21
(see Procedures, Section IV-D) and the spectral responsivity

of the detection system was calculated using the calibrated

spectral radiance values (see Section VII).

Tungsten Ribbon

AL

?

(r 3> < Calibrated Area

Figure 4. Alignment of the calibrated vacuum tungsten

ribbon lamp using the alignment laser AL 2 .

3. Photomult iplier Tube Linearity

A tungsten lamp was placed in the sample position S with

a double aperture device (DA) [52] located between the sample

position and EL 2 , figure 1, such that both apertures A and B

were included in the 14° angle subtended by EL 2 . The photocurrents were measured at varying incident radiant fluxes

with aperture A open and aperture B closed, with aperture B

open and aperture A closed, and with both apertures open.

D . Measurement Procedures

The measurement procedure used in determining the

effect of temperature on the emission spectrum is depicted

in figure 5. The temperature of the QS solution was set to

T : , the spectrum recorded in ascending wavelength intervals,

the temperature was set to T 2 , 30 min were allowed for

equilibration, the spectrum recorded in descending wavelength intervals, etc. The spectra were corrected for photomultiplier non-linearity, then averaged as indicated by the

22
dotted lines, thus eliminating the effects of long term

instrumental drift as well as any backlash in the monochromator wavelength drive.

725 nm

360 nm

Bl T, 2 T3 T4

T. T, T2T. Bl

Figure 5. Time symmetrical measurement procedure used for

determining effect of temperature on the

emission spectrum.

In general, measurements were made with the analyzer

(detection polarizer) in the vertical mode (designated S) to

avoid "Woods" anomalies from the grating observed with the

polarizer in the horizontal mode (designated P) (see Section

VI-E)

The measurements on a single series of ten quinine

sulfate solutions for the correction and certification of

the emission spectrum takes three days as outlined in figure

6. On day 1, the standard lamp was placed in the sample

compartment, aligned (Section V-C2) , and R, values were

A 2 obtained at lamp operating currents of 8 . 80 30± 0 . 0008 and

c 6. 0000±0. 0006A from which the spectral responsivities M., of

A 2 the detection system were calculated over the 350.0 to 425.0

nm and 400.0 to 900.0 nm ranges, respectively (Section VIII),

23
Standard Lamp Quinine Sulfate Standard Lamp

/VO. A

1 r amps 8.803

"

6.000

i i

f ' i w w i r

i r 8.803 B

Q, — Q

B

10

At, At,

¥ ¥ " 8.803

^~

j ~ 8.803

6.000

Day 1 Day 2 Day 3

At,= At2

Figure 6. Time symmetrical measurement procedure used for

determining the spectral responsivity of the

detection system and the uncorrected emission

spectrum (Y) for quinine sulfate.

The standard lamp was then replaced by the normal,

right-angle (sideview) , cuvette sample holder. A clean,

quartz, unblackened cuvette was inserted, rinsed a minimum

of five times [60] with the blank solution (each rinse soluX S tion was removed by suction), and the blank signal Y 7 !J 7 c

, was measured every 5 nm over the wavelength region of

interest (360-725 nm) . The blank solution was removed by

suction without cell removal, the cell was rinsed a minimum

of five times with solution QSi , filled with QSi , and the

compensated fluorescence signal Y^'A 5 c

, was measured every

5 nm from 360 to 725 nm. The cuvette was then rinsed and

filled with QS2 and (Y) for QS2 was recorded every 5 nm.

This procedure was followed until the second blank was measured. Each spectrum took approximately 25 min to record.

On day 3 as on day 1, the spectral responsivity M, of the

A 2 24
detection system was determined to assure that the instrument

calibration did not change and to obtain a measure of the

stability of the instrument.

A spectrum was recorded in either increasing or decreasing

wavelength intervals as designated by the arrow direction in

figure 6. The time intervals, Ati and At 2 were kept constant

so that time symmetry was maintained. The spectra were

corrected for photomul t iplier non-linearities and then were

averaged according to the dashed lines.

VI. Quinine Sulfate

Quinine, an alkaloid extracted from cinchona bark, is a nitrogen heterocyclic organic species soluble in acid, alcohol,

and ether. It exhibits relatively broad fluorescence and

absorbance spectra (figure 7)

Quinine and its derivatives have been the most widely

used fluorescence standards but do exhibit some anomalous

characteristics. Due to these characteristics, inaccurate

values and wrong information on the fluorescence of quinine

sulfate can be found in the literature. This, in turn, has

cast doubt on using quinine as a fluorescence standard.

Therefore, we include here discussions of previous data

which suggested problems in the use of quinine sulfate. At

the same time, additional results and discussions are included

to characterize further the SRM grade quinine sulfate.

A. Memory Effect

It had been stated [6] that the absorbance spectral

characteristics depended on whether quinine, quinine bisulfate

or quinine sulfate was the original solute used to prepare the

solution. Fletcher [25] showed that dissolution of several

different samples of quinine and its salts gave equivalent

absorbance and excitation spectra [61]. The memory effect

25
r-l

5h > O

3

i-H «+H m

o

<u o P • H

4-1 u U

£

+-> 4h

to

a3 CD

t/> H CO

simp Ajgjjjqjv 'xn|-j uojoq,-)

26
observed earlier was attributed to the presence of an impurity,

varying waters of hydration, or slit width problems [26,27,33].

B . Purity

1. General

The purity and number of waters of hydration in a material

are important factors to consider when measurements of specific

molar absorbances and absolute quantum yields are made. However, they are not critical to the determination of a relative

spectral photon yield or a corrected spectrum unless the

impurity absorbs some of the emitted radiation or itself emits

in the same wavelength region as the standard. In both cases,

however, the presence of up to 0.1 percent impurity would not

significantly affect the measured data if the specific molar

absorbance and photon yield of the impurity are similar to

those of the standard since the effect is less than the

measurement precision and accuracy.

For example, consider an impurity (0.1 percent) that

absorbs at 347.5 nm and/or 450 nm with a specific molar

absorbance of 10** L mol 1 cm 1 and photon yield of 0.5. The

measured fluorescence signal (Y) from the quinine sulfate

^ e 347 5 ~ L mo ^

1 cm lf ^

~ ^ n dilute solutions [63]

would be decreased by the absorbance of the impurity in

solution to give a new measured fluorescence signal (Y) '

CY) ' = (Y) 10" ecb

(4)

where

£ = specific molar absorbance (^lO 1* L mol 1 cm x ) b = cell path length (0.5 cm)

c = impurity concentration (^1.3 x 10 9 mol L

1 [62])

Upon substitution of these values into equation 4, the fluorescence signal would be decreased by approximately 0.001

percent, a value that is two orders of magnitude smaller

27
than the observed measurement precision and approximately

three orders of magnitude smaller than the stated measurement

accuracy (Section IX) . Similarly, for two emitting species, the total fluorescence reading (Y) T

in terms of the signal for the pure

compound (Y) i is:

mT

- orha §ftf|> i, (5)

where Q is the photon yield and the subscripts 1 and 2 refer

to the quinine sulfate and impurity, respectively. Substitution of typical values for Qi,2, £1,2 and Ci , 2 show that a negligible error of ^0.1 percent would be incorporated in

the emission spectrum.

Under extreme conditions in which the impurity has a maximum e of ^10 5 [64] , the errors in the absorbance or

fluorescence readings increase an order of magnitude to 0.01

and 1 percent, respectively. It is more likely, however,

that impurities in quinine sulfate would have specific molar

absorbances and photon yields comparable to those of quinine

resulting in errors of 0.1 percent or less in absorbance or

fluorescence measurements.

2. SRM Grade Quinine Sulfate Dihydrate, 0S-2H 2 0:

SRM grade QS-2H20 was compared to highly purified samples

of quinine and quinine sulfate [33] . No impurities that

affected the optical characteristics of the material were

found in the SRM grade QS*2H 2 0 within experimental error as determined by thin layer chromatography (TLC) , relative photon

yield, fluorescence lifetime, specific weight absorbances,

and water content. High performance liquid chromatography

(HPLC) did show the presence of a second peak (^1.71) that

was shown to be dihydroquinine sulfate by high-resolution

mass spectrometry; however the optical characteristics of the

dihydro species were similar to those of quinine sulfate.

28
(a) TLC : Fluorescent and non- fluorescent silica gel

plates were spotted with eight samples of SRM grade QS*2H 2 0

and two samples of purified quinine sulfate. Two different

solvent systems (Solvent 1 = CHC1 3 :MeOH : HOAc -75:20:5;

Solvent 2 = i-PrOH:HCOOH-80 : 20) were used to develop the

plates [65] . Both types of plates were irradiated using 265

and 366 nm UV radiation. The fluorescent plates showed a single blue fluorescent spot for each sample (normal green

fluorescence of the plate was quenched) and the non- fluorescent

plate also showed a single, blue fluorescent spot for each

sample. The r^ values for the purified and SRM grade quinine

sulfate samples were the same, Table 3. The plates then were

developed by spraying with a ten percent H2SO4. solution.

Again only a single spot was observed for each sample with

the same r^ values as those in Table 3.

Table 3. Comparison of SRM Grade Quinine Sulfate with a Purified Quinine Sulfate Preparation Using Thin

Layer Chromatography.

- - - - R.£ Values - - - -

c d Solvent 1 Solvent 2 SRM Grade a 0.516±0.02 0.625±0.02

Purifiedb 0.515 0.626

Average of "eight samples.

Average of two samples.

C By volume: 75 CHC1 3 :20 meOH:5 HOAc.

d By volume: 80 i-PrOH:20 HCOOH.

(b) HPLC : The presence of dihydroquinidine in commercially prepared quinidine has been reported [66] and thus

suggests the presence of dihydroquinine in commercial quinine.

HPLC analysis of SRM grade quinine sulfate showed the presence

of a second constituent that was approximately two percent of

29
the total as determined by UV and fluorescence detection [67]. j H

A microparticulate reversed phase column was used with mobile

phases buffered at two pH ' s . Two of the mobile phases were

mixtures of 70 percent aqueous buffer (H 2 0:1.36 g/L KH 2 P(K:0.15

mol/L H 2 S0O and 30 percent acetonitrile at pH's of 4.3 and

2.3, respectively. A third mobile phase was a mixture of 85

percent buffer and 15 percent acetonitrile at a pH of 2.3.

Use of all three mobile phases with UV (254 nm) and fluorescence (X = 250, 317, or 350 nm, X = 450 nm) showed

ex ' ' 'em

the presence of a second constituent.

The major peak constituent (labelled A) and the minor

peak constituent (labelled B) were individually trapped in a flow cell in an uncorrected spectrof luor imeter . Uncorrected

emission and excitation spectra of constituent A and constituent B were identical indicating that constituent B was

optically equivalent within experimental error to quinine

sulfate. The spectra obtained from the two constituents

using the mobile phase buffer at a pH of 4.3 were quite

complex. This is not unexpected since 'optical' titration

curves, monitored by UV absorbance and fluorescence emission

spectra, have mid-points at about pH 4.3 suggesting the

presence of complex emitting species [68]

(c) High- resolution Mass Spectrometry : Several samples

of SRM grade quinine sulfate were separated by chromatography

under preparative HPLC conditions, each time separating the

materials responsible for producing peaks A and B. The

fractions of 'peak A' were combined and the fractions of

'peak B' were combined. Both fractions were neutralized with

sodium hydroxide and the free bases were extracted into

methylene chloride. The methylene chloride fractions were

separated from the aqueous phases and each fraction was

concentrated at least 10-fold by solvent evaporation. Both

'A' and 'B' fractions were analyzed by high resolution,

30
electron impact mass spectrometry [69] . The mass spectrum of

the material in fraction A was identical to the literature

spectrum of quinine with molecular ion at m/e 324 [70]. The

mass spectrum of the material in fraction 'B' showed a molecular ion at m/e 326. The exact mass measured for constituent

B was 326.1990 and the mass required for C20H26N2O2 is

326.1994; this is consistent with the interpretation that the

material separated in peak B is dihydroquinine . It is possible

that the peak B constituent could be dihydroquinidine , however,

the measured optical rotation (Appendix B) supports the

presence of the levorotatory species, dihydroquinine.

(d) Absorbance and Fluorescence Parameters : The optical

equivalence of the SRM grade QS-2H 2 0 and the purified QS'2H 2 0

was demonstrated by determining the specific molar absorbances

at 347.5 nm, the relative photon yield Q and the fluorescence

lifetimes t in 0.05 mol/L H2SCK. These results are summarized in Table 4 [33]

The normally accepted values for Q are 0.508 and 0.546

in 0.05 and 0.5 mol/L H 2 S0 4 , respectively [7,74]. The

generally accepted value for t in 0.05 mol/L H2SCK is approximately 19.2 ns [reference 75 and references therein]. As can

be seen, excellent agreement within experimental error was

obtained for all the parameters measured.

Ce) Water Content : As mentioned, waters of hydration,

if variable, can be considered an impurity when absolute

photon yields or specific molar absorbances are measured.

It was reported that quinine sulfate contained variable water

content [25,71] but drying at 50-55 °C under vacuum brought

the material to the dihydrate, QS*2H 2 0 [25]. Thermogravimetric analysis showed that a weight loss of M.5 percent

occurred upon heating. This was interpreted to mean that

the quinine sulfate originally contained 4-5 waters of hydration [33]. However, preliminary mass spectrometry data

showed water loss at ^50-80 °C followed by decomposition

31
q_l Mh

S-H rj (J CD ,—

!=: K

4_) CO CD

nj

d S_, r-1 (4_| * -(-1 r—

rH • ,—CO 3 <r' o3

r/> . UJ " 1—

,—

pi

O

(/) CD «H 0 ,—| d CJ <4H

.1-1

d

d

r—

cr CD

H

ry 0 to CO 5h #\ O

CO o3

<D W)

H Tj LO

•H ,Q I—I q_l 4H CO CD O +1 O

•t—1 CD •H _1 ^j- " Pi >H \ <^f

1—1 t-*—1 ' 1 I—

LO LO

O £ 0 •H •H CD O 6 O 4_)

CT3 CD '-H +-> OS 0 ~ *H

i-C j- Mh b>r~

d 0 Cu

O CD

o3 -H CD CO TH

MH

O CD <M LO 1 1 I—* >

+->

O CD •H LO

03 CDU +->

O o3

cLi p, >f) o3

O T3

t—1 CO CD 1—

O _(-|d

CD OO t-o o3 CO 0 (X O 4_) 0 p LO LO CO

CD rH d C/) P-i

OO CD •H +J f—

d d Ti 4_) •H CD C! j- 3 E= 03 O CD * ' \U 1—

}_, o3

cd 3 ry ' » h3 co <-i >—

o3 o3 co 1 CD

$_l 0 rrj 6 0 b> n r—

w n u CO •H

CD 1 m It; M—

CNl vO

ro ^> rH

O

dd rH

O | d>—< M-t |5 O O • •H Q O +-> CO O O o3 d n +1 -l-l , 1 4_) CNl 5h

*—/ 1—

to )—

CO CO CD Vh L/J O }—

(-< CO

CD cd CD CO LOO 0 d

d

W) I—

,—1 0 •H •H r—

Zi CD «H +J •H r!>— 11—1' JJ1 03 0 oi o3 +J <-|_| • H CH( Ctf o3 Q

i-) Pi <^ < r* to •H •H

L-) LU 4** •H to Pi " H CO m r/i ry^ r-M UJ Mh f_, fn CD

rrj CD CD }h

CD CD 4_> 4_) CD <-« i—

rjf—

(j CD CD +j CO

CD •H >—

nH CD oi H3 O t3 CD TJ3 CD >w CD c\i o3 j_i CD PI

g_j •H

X j_i CD CD ri CD JH 0 <4H CNJ CJ cxi +J <4H bo $-1 CD

03 • H • • CD •H

O

H fH CO CO

QWH 3 cy ex

a. CO 03 ^2

u CD MH

32
[72] indicating that the water of hydration was removed at

temperatures similar to those suggested for drying the

quinine sulfate [25]

The water content of six and four samples of the SUM

grade QS were determined by Karl-Fischer (KF) titration and

weight loss procedures, respectively. The KF titrations

were made using varying amounts of distilled water as the

standard and varying amounts of "as is" SRM grade quinine

sulfate so that linear regression analyses could be performed

on the analytical curves [73] . The weight loss procedure

consisted of drying the quinine sulfate at 50-55 °C under

vacuum until a constant weight was achieved. The values

obtained by these two procedures are summarized in Table 5.

The average water content as determined by the KF procedure,

4.74 percent, is a little higher than the 4.57 percent

determined by the weight loss procedure. Both values bracket

the theoretical value of 4.60 percent. The KF and weight

loss values correspond to 2.06 and 1.99 molecules of water

per molecule of quinine sulfate.

The dried quinine sulfate samples were found to be

hygroscopic when allowed to stand at room temperature in

^35 percent humidity. After absorbing water and reaching

constant weight (^0.75 h) , the weight gained by the four

samples corresponded to the weight lost by heating in vacuum

to within 0.3 percent by all samples. This indicates that

the "before drying stoichiometric, two water molecules per

one quinine sulfate molecule ratio" was again reached.

These data show that drying the SRM quinine sulfate at

50-55 °C for 12 hours produces the anhydrous rather than

the di-aquo species. Additional evidence for this conclusion

is given in the homogeneity and specific weight and molar

absorbances sections.

33
Table 5. Percent Water in SRM Quinine Sulfate* 2H 2 0 as

determined by Karl-Fischer Titration and Weight

Loss Procedures.

uaJiilJ ^ani"n 11 pL #" r L U C c LI LI I fc; X

311 242 .7

o KF 4.77

17-L Li Lt2 l ^ \ ft A 7ft 284 80.0 KF 4.64 4 74

177 187.6 KF 4.76

96 3Q3.

KF 4.73

453 320.8 KF 4.73

27 147.56 WLb

' C 4.52

311 591. 76 WL 4. 59 4.57

453 236. 59 WL 4.62

96 273.73 WL 4.55

The Karl-Fischer titrations were performed by S. Margolis,

Organic Analytical Research Division, National Bureau of

Standards

.

Semi-micro balance calibrated to ±.01 mg using NBS #62730

calibrated weights.

Dried at 55 °C under vacuum for 12 h, put while warm in

desiccator over drierite, covered, and weighed when cool.

(f) Homogeneity : The specific weight absorbances WA

of twenty quinine sulfate solutions (^2 x 10 5 mol L

l ) were measured on an NBS reference spectrophotometer to

determine the homogeneity of the SRM. The values obtained

are summarized in Table 6.

The average WA values are ^4 percent lower than those

reported earlier [33] i.e., 9.24±0.09-, 14.34±0.12-, and

77.82±0.86 x (10

_5

) g solv (g sol

)

1 cm" 1

. If it is assumed,

however, that the weighed quinine sulfate was the dihydrate,

and the dried material in reference 33 was anhydrous, the

WA values in Table 6 adjusted to the anhydrous state become

34
Table 6. Specific Weight Absorbances for Twenty Solutions

of Quinine Sulfate Dihydrate in 0.1 mol/L HC1CU

at 365.0, 347.5, and 250.0 nm.

- (W.A. ^solv^sol^

1 cm jxlO - 3

A = 365 • 0 nm A = 347 . 5 nm A = 250 .0 nm

Sample S-I a S-II S-I S -II S-I S-II

27 8. 8509 8 . 8854 1 X Q X f\ 7 1 X 8217 72 . 9934 72 .7913

96 8. 8504 8 . 8233 11 J

X . O07A7 JO /1 X 7926 73 .0886 73 .3980

122 8.8561 8 . 8578 1 X Q C X A 1 X 8487 73 . 0102 72 .9393

177 8. 8382 8 . 8543 13. 8377 13. 7977 73 .0297 72 . 5286

230 8.8426 8 .8743 13.8546 13. 8656 72 .9184 72 .9920

284 8. 8568 8 . 8160 13.8547 13. 7726 72 . 9882 72 .8797

311 8. 8324 8 . 8426 13.8306 13. 7796 73 . 1418 72 . 5563

370 8.8698 8 . 8248 13. 8737 13. 7797 73 . 0159 72 .9112

421 8.8600 8 . 8823 13. 8798 13. 8457 73 . 2624 72 .4397

453 8. 8806 8 . 8762 13. 8936 13. 8246 72 .9898 72 . 8001

x

b

8. 854(6 .918) C 13.834(10. 809) c 72.934(56 .986)

s b .015 . 022 .125

% s b. 169 . 159 .171

Corrected for instrumental bandpass error.

The two values for each sample were averaged and these

averages were used to determine x, s, and % s. C

e (L, mol" 1

, cm" 1 ) x 10" 3

.

35
9.287-, 14.512-, and 76.507 x (10" 5 ) S solv (g sol

)~ 1 cm' 1 at

365.0, 347.5 and 250.0 nm, which are in good agreement

with those reported earlier. This provides additional

evidence that drying quinine sulfate*x hydrate at 50-55 °C

under vacuum produces the anhydrous and not the di-aquo

species. The relative standard deviations calculated for

the data at each of these wavelengths were comparable .although

they might have been expected to decrease as the WA values

increased. This may be offset by the added difficulty in

making accurate absorbance measurements in the UV region.

If, in the worst case, the total standard deviation s for the 347.5 nm data is due to sample inhomogenei ty (assuming

no experimental error) , the inhomogeneity of the SRM quinine

sulfate can be expressed in terms of a tolerance interval,

represented by x ± Ks where x is the average value for the

specific weight absorbance, s is the standard deviation of

the measurements, and K is a factor for two-sided tolerance

limits of normal distributions which depends on the number

of samples analyzed n, the probability value or confidence

level desired y, and the percent of the 500 samples included

in the tolerance interval P. In this case the tolerance

interval ± Ks is designed to cover at least 95 percent of

the population with a probability of 95 percent [14]. Thus,

for n=10, P=95, and Y=95, the factor K is 3.379 [76], and

the inhomogeneity in the samples would be on the order of

0.5 percent since

x ± Ks = 13.834±0 . 075 g , (g ,

)

1 cm" 1 & . solv^ & sol^

However the total percent standard deviation is composed

of the contributions due to the inhomogeneity of the SRM and

the contributions due to experimental errors:

36
where

aWA

WA the overall percent standard deviation (0.159

from Table 6)

the percent due to inhomogeneity ; and

the percent due to the experimental measurements.

a

E Estimates of the individual components comprising are:

concentration (several weighings and dilutions) a

c /C = 0.07

percent; absorbance - = 0.1 percent [77]; and cell length

a^/b = 0.005 percent. Additional errors in the measurement

process are temperature variation a^/T = 0.15 percent and

cuvette placement cTp/p = 0.03 percent. These- values , added

in quadrature, give an estimated experimental error for o^/E

of 0.196 percent. Thus,

and the estimate of SRM inhomogeneity (±Ks) drops to a more

realistic 0.26 percent. One major factor in this 'inhomogeneity' could be slightly varying waters of hydration which

do affect the measurement of specific weight absorbance

values but not the normalized, certified emission spectrum.

Thus, although in the worst case, we assign the inhomogeneity

of quinine sulfate a tolerance level of 0.5 percent, it is

more likely to be on the order of 0.3 percent or less.

The values reported in Table 6 are indicative of vial

to vial homogeneity and are not to be used as certified

values of the specific weight absorbances. Although preliminary data showed no fluorescence effect when measuring the

absorbance of fluorescent solutions on the NBS Reference

0. 078

37
Spectrophotometer [33] . Aditional verification would be

necessary before certification of absorbance values could

be accomplished.

Cg) Absorbance Spectrum : It has been reported that

the absorbance spectrum of quinine sulfate is independent of

the sulfuric acid concentration [75]. This appears to be the

case within experimental error when peak ratios are measured

over a hydrogen ion concentration range of 0.01 to 1.0 mol/L

for H 2 SCU or HC1CU, Table 7. (Note: Ratio values that are

within 1 percent of each other are considered equivalent

under the measurement conditions if the absorbance values for

the individual peaks are approximately equal. Thus, if the

values of the ratio of the absorbances at 347.5 nm and 317.5

nm agree to within one percent, the ratios are considered

equivalent On the other hand, one could expect somewhat

larger errors for measurements of the absorbances for two

disparate peaks such as the 250 nm and 347.5 nm peaks and

still consider the ratios to be equivalent.)

Table 7. Ratios of the Absorbances for Several Peaks of

Quinine Sulfate in Sulfuric and Perchloric Acids.

i

A

250 /A347.5 A

347.5 /A 317 A

347.5 /A 365

H 2 SCU HC1CU HaSCK HC1CU HaSCK

(H

+ )mol/L

0.01 5.317 5.400 1.238 1.235 1.561

0.1 5.382 5.382 1.250 1.256 1 .648

1.0 5.418 5.404 1.235 1.235 1 .570

5.0 5.427 5.626 1.217 1.193 1.510

10. 0 5.427 5.558 1.179 1.003 1.414

I

38
As can be seen, the range of ratios for a hydrogen ion

of 0.01 to 1.0 mol/L for the two acids are 1.2 and 1.6 percent

respectively (essentially equivalent) for the A-^y 5/^317

ratios, however, a larger percent is observed for the ratio

at these wavelengths over the total hydrogen ion concentration range of 0.01 to 10 mol/L; i.e., 5.7 and 20.2 percent,

respectively. A slight increase (^2 percent) is observed

when the A-250^347 5 ra

"t i° s are calculated over the 0.01 - 10.0 mol/L hydrogen ion concentration range.

This trend was also observed in a separate experiment

in which the specific weight absorbances were found to be

equivalent experimentally in 0.1 and 1.0 mol/L perchloric

acid and 0.5 mol/L sulfuric acid, Table 8. However,

increasing the HCIO4 concentration to 3.5 mol/L resulted in

a 4.8 and 5.7 percent increase at 317.5 and 365.0 nm,

respectively while the specific molar absorbances at 250.0

and 347.5 nm show only a 0.9 and 0.7 percent decrease,

respectively.

Table 8 . Specific Weight Absorbances of Quinine Sulfate

in Sulfuric and Perchloric Acids.

- " " ("-A., gsdvCSsoir'cm-MxlO-' - -

-

(H )mol/L 365 nm 347.5 nm 317.5 nm 250 nm

0.1 H 2 SO H 9.25 14.20 11.45 77.85

0.1 HC1CU 9.10 14.20 11.35 76.45

1.0 HC1CU 9.25 14.20 11.35 76.15

3.5 HC1CH 9.70 14.10 11.90 75.55

39
Thus it does appear that the absorbance spectrum of

quinine sulfate is dependent on acid concentration at high

[H

+

] with the shoulder at 317.5 nm and possibly a peak at

^365 nm (under the major 347.5 nm absorbance peak) increasing

as the H

+ concentration increases. This observation could

support the presence of a second emitting level (at 365 nm

or higher excitation) that is responsible for the "red edge"

effect, [Section VI-F)

.

C. Stability

It has been reported that quinine sulfate exhibits

stability in solution [7,33,71,78]. Several reports, however,

have shown that quinine sulfate is unstable in solution upon

irradiation [79], but it must be emphasized that in all

cases except that reported by Melhuish, the instabilities

were observed under extreme conditions of intense radiation,

excessive excitation bandpasses (^30-40 nm) , very dilute

solutions, and small sample sizes. Melhuish, on the other

hand, reported the instability using an excitation wavelength

of 313 nm, a location where Dawson and Windsor [80] reported

variable quantum yields indicating that excitation at 313 nm

may cause problems.

We found [33] that quinine sulfate in 0.05 mol/L H 2 S0u

was stable to ±1 percent when stored in the dark and measured

periodically for six months. With similar storage over a six week period, the spectral emission envelope of quinine

sulfate in 0.105 mol/L HC1CK did not change by more than

±0.2 percent (equal to the measurement imprecision) at the

67 percent confidence level (see Section VIII on the corrected emission spectrum)

.

Additionally, studies of 1 ppm quinine sulfate in

0.105 mol/L perchloric acid show stability under normal

irradiation (450 W xenon lamp, 5.3 nm bandpass, 347.5 nm

excitation) at the ±0.2 percent level for short periods of

40
time (2 h) and at the ±0.8 percent level over longer time

periods (12 h) . This latter observation may indicate degradation over long irradiation times.

Nevertheless, to avoid any possible photodegradation,

all prepared solutions of quinine sulfate should be stored

in the dark and removed only for measurement purposes.

Additionally, freshly prepared solutions of quinine sulfate

should be used for critical measurements.

D. Oxygen Quenching

Oxygen quenches the fluorescence of many organic aromatic

species and may be considered to be an impurity molecule.

It is soluble in aqueous solutions to ^1 x 10 k mol/L and

aliphatic solvents to ^2 x 10 3 mol/L [81] . Every collision

of an excited molecule and oxygen leads to quenching by the

viscosity dependent equations [82]

a QS* +

3 0 2

-* J QS +

3 0 2 (7a)

2 QS* +3 0 2 +

3 QS*+ 3 0 2 (7b)

with the overall effect of reducing the photon yield:

n = kFM r7 s Q ~ kFM

+ k IM+ V02 ) C?C)

and thus n 2 , the spectral photon yield i.e.:

Q = n 2 dA 2 .

No change in the spectral intensity was observed after ten

freeze-vacuum- thaw cycles performed in the "Schlenk" adapted

[83] spectrof luorimeter cell [33] which agrees with earlier

reports [7,26] of non-oxygen quenching of quinine sulfate in

1.0 mol/L H 2 S0k .

41
E . Polarization

It has been shown that errors in fluorescence measurements can be as large as 80 percent and that errors of 10-20

percent are common under normal operating conditions (90°

geometry) if the radiation emitted by a fluorophor is anisotropic [84]. Stokes reported in 1852 [1] that quinine appeared

to be an isotropic emitter in sulfuric acid. However, quinine

sulfate in glycerol containing sulfuric acid exhibits polarization which is a function of both excitation and emission

wavelengths [85]

.

The dichroic ratio of emission D and the emission

anisotropy r, indicative of the polarization of a material,

can be determined by measuring the four fluorescence fluxes

yF

,P

, , Y^ ,S, , Y? ,P

, , and YP,S

, and substituting these

A i , A 2 Ai,A2 Ai,A2 Ai,A2

values in the following equations [84]

rYs,s I rY

p, P

-I

J) = L A 1 » A 2J L Ai ,A 2 J (g-j

[Ys,p

]

rY p,s 1

and

r = 5li (9) D+2 1 }

where the superscripts and subscripts have the same meaning

as in Section IV.

The ranges of values for D and r are:

1/2 < D < 3

-1/5 < r < 2/5

and for a material that is not polarized and emits isotropically, D = 1 and r = 0.

42
We determined r and D for quinine sulfate in 0.1 mol/L

HC1CU at 25 °C using Xi = 347.5 nm and a X 2 range of 360 to

725 nm in 5 nm intervals. The average r and D values obtained

by three, time symmetrical determinations are graphed every

25 nm in figure 8a and b. The one percent signal limits are

designated by the vertical dashed lines at ^390 and 615 nm.

The averages for all the r and D values obtained between the

wavelength range 380 to 630 nm were 0 . 00354±0 . 00088 and

1. 0107±0. 0027, respectively.

Additionally, r and D values were determined as a function of quinine sulfate concentration at X i = 347.5 nm

and X 2 = 460 nm, Table 9.

Table 9. The Dichroic Ratio of Emission D and the Emission

Anisotropy r of Quinine Sulfate in 0.1 mol/L

HCIO4 as a Function of Quinine Sulfate Concentration.

Concentration

,

ppm

0.297

0.513

1.087

1.121

5 .460

10. 315

15.877

D

1.0167

0. 9966

1.0023

1. 0029

1. 0023

0. 9934

1.0026

D = 1. 0024

r

0. 00554

- .00113

.00077

. 00087

.00076

- . 00221

.00192

r = .00093

Although care was taken to align the excitation

polarizer and analyzer optically so that shifts of excitation and emission radiation were minimized upon rotation

of the polarizers, small differences were observed for the

average values of D and r in Table 9 and those average

values reported over the total emission range for one quinine

sulfate concentration (D = 1.0107 and r = 0.00354). These

43
0.008

0 .006 -

0.004 -

0.002

OhO

Ph

E-O

CO

I—

<

oI—

CO

CO22 o.ooo -

-0.002

400 450 500 550 600

WAVELENGTH, nm

1 .04

O

CO 1 -03

i—

w

Cm

o

1 .02 -

1 .01 -

S3 1 .00 - O

Ph

O

650

0.99 Q ~ "350 400 450 500 550 600

WAVELENGTH, nm

650

Figure 8. Emission anisotropy r (8a) and Dichroic ratio

(a,b) of emission D (8b) as functions of wavelength

for quinine sulfate in 0.105 mol/L HCIO^.

44
differences are considered to be due in large part to a small

optical misalignment in the excitation polarizer and not

isotropic emission. Thus the D and r values determined in

these two studies are equivalent and demonstrate the isotropy

of the radiation emitted by a solution of quinine sulfate in

0.1 mol/L HC1CU . However, if the values for the emission anisotropy were

real and due to a true residual polarization [86] , the error

introduced in a fluorescence measurement using an instrument

with no polarizers, can be calculated by [reference 84,

equation 2 7]:

Y wJ^ 1 + r 3cos 2 q - (1 + F + G - 2FG)

(1 + F) (1 + G) ]}

(10b)

/ X X \ where

\ A i A / t ^ie S ^-S na l recorded without polarizers,

X U

(Y, ' , ) would be the signal if the sample emitted isotropiA i , A 2 O

cally, F is the polarization ratio of the exciting radiation,

G is the polarization ratio of the detection system, and a

is 90° for right angle viewing.

In the worst case in which measurements are made with

no polarizers, F and G - 10 \ and r - 0.002, then:

(Y) = (Y)

Q

[1 + (.002) (-2)] and the error in the measurement

is approximately -0.4 percent. Usually, the G and F values

are - 0.6 and the error would be on the order of + 0.1

percent which is negligible.

In certification measurements, we wanted to remove the

excitation polarizer and allow the analyzer to remain in

the ' S' orientation to avoid alignment problems with the

excitation polarizer and to avoid the Wood's anomaly caused

by the grating in the detection monochromator . The

45
uncorrected spectra for the standard tungsten lamp (unpolarized) with the analyzer in the S and P positions are given

in figure 9. The "P" spectrum shows a "cusp" attributable

to the Wood's anomaly of the grating. Since, the anomaly,

and thus possible decreased measurement accuracy would be

present if the analyzer were absent or in the "P" orientation, it was decided to make all certification measurements

with the analyzer in the "S" orientation [52].

In this case [reference 79, equations 25a and b modified

as in reference 49], the two signal components are:

Y x;!x 2 - v> [

Y

x;!* 2] o ^ ^ j

where (Y) corresponds to the signal unbiased by polarization,

F is the polarization ratio of the exciting radiation, and the remaining symbols are as previously

designated.

The total signal observed is:

yX,S _ yP>S + y^'^ (12a)

A i , A 2 A i , A 2 A i , A 2

.(

y

a;!-a 2 )o]

(1 + r * (12b)

1

where t^;!x 2

) o 1/3 < F+1) t

Y x;!x 2 / p ) + 2Y

x;!xJ-

^

46
400 450 500 550 600 650 700 750 800 850 900

Wavelength, nm

Figure 9. The sensitivity of the detection system to

radiation emitted by the calibrated tungsten

ribbon lamp that is parallel (P orientation)

and perpendicular (S orientation) to the

grating grooves in the detection monochromator

47
The relative error introduced by removal of the excitation

polarizer is [49]

yX , S

(Y X,S

a i , A i o

A 2 ^o

AYX,S

A i , A 2 2F-1

F + l

(12d)

The average value of .s^J/^fi m 's 9 xJ 0

> ^ere

380 nm < A 2 < 615 nm {1% signal limits) is 0.9998±0.Q06 which

would, in the worst case, introduce a negligible error of

0.06 percent due to residual polarization into the certification measurements. Thus the certification data for quinine

sulfate dihydrate in 0.105 mol/L HCIO^ were obtained without

an excitation polarizer and the analyzer in the "S" orientation

F . Emission Spectrum as a Function of Excitation

Wavelength (Ai)

The luminescence spectrum [87] of quinine sulfate shifts

to longer wavelengths if the excitation wavelength is on the

"red edge" (greater than 360 nm) of the first absorbance

peak [66,85,88-90]. This anomalous effect also has been

observed with various fluorescent materials that have similar

(quinoline, 6 -methoxyquinoline , 6 -ethoxyquinoline) and

dissimilar ( 2 -aminopurine , luminol) molecular structures

compared to quinine sulfate.

The red shift does not appear to involve a change in

the emission shape [85,90] nor photon yields (Section I-G),

but does give a fairly significant change in the location of

the emission peak maximum going from an emission maximum of

453 nm using excitation wavelengths from 230 to 360 nm to an

emission maximum of 470 nm using an excitation wavelength of

440 nm [88] . We found that changing the excitation wavelength from 347.5 nm to 366 nm shifts the emission maximum

1.8 nm (451.5 nm to 453.1 nm) and errors of up to 20 percent

in fluorescence measurements on the "blue edge" of the

48
emission peak were observed as shown by the ratio

(Y^jJ? r , )/(Y?£? -x ) plotted as a function of A 2 , Figure 10. j4 / . J

, A2 jOO , A2

0.9

Figure 10.

350 400 450 500 550 600 650

WAVELENGTH, nm

The ratio of the uncorrected emission spectra of

quinine sulfate in 0.105 mol/L HClOf* using

exciting radiation at 347.5 and 366 nm to

demonstrate the 'red edge' shift.

This anomalous effect has been attributed to emissions

from two excited singlets [85], a rotatable auxochrome - solvent

effect [88] , and emission from several molecular configurations differing in solvent orientation [90] . Although the

reason for the shift has not been unambiguously established,

care must be exercised in using literature values for the

emission spectrum since they obviously depend on excitation

wavelength.

The certification values of the emission spectrum given

in this report (Section VIII) are for 347.5 nm excitation and

49
the emission spectrum should be determined using this excitation wavelength.

G. Photon Yields

1. Absolute and Relative Values

The photon yield values for quinine sulfate of 0.508 in

0.05 mol/L and 0.546 in 0.5 mol/L H 2 SCU first measured by

Melhuish [7,74] and later by Dawson and Windsor [80] have

been challenged [5,6,8,85,91,92]; however, the values still

appear to be the best determined photon yields and those

most used as relative standard values [25,26,71,75]. An

intercomparat ive study using quinine sulfate as the basic

reference standard and fluorescein, pyrene, and rhodamine B

as the comparison materials was reported [33] and some of

the data are summarized in Table 10. These data show that

using the photon yield of 0.508 for quinine in 0.05 mol/L

H 2 SCU results in good agreement between the values measured

for fluorescein, pyrene, and rhodamine B and those average

values reported in the literature.

Table 10. Comparative Photon Yields Using Quinine Sulfate

as the Reference Material.

Compound Solvent

QS 0. 05 mol/L HaSO*

Fluorescein 0.1 mol/L NaOH

Rhodamine B Ethanol

Pyrene Toluene

J2. Q (Lit.)

0.508[7]

0.87+0.03 0.88b

[17]

0.72±0.02 0.69[11], 0.71[26]

0.57±0.04 0.60[80]

Standard, 0.508 as photon yield.

Average of 16 values Csee reference 33)

.

50
Other intercomparative studies have been performed using

quinine sulfate as the standard with a photon yield of 0.508

with excellent results [75, 78b, 80], all of which support the

values reported by Melhuish [7,74] and Dawson and Windsor [80].

2. Photon Yield as a Function of Excitation Wavelength

It was reported that the photon yield of quinine sulfate

in sulfuric acid is dependent on excitation wavelength

[80,85,91]; however, careful measurements have shown that the

photon yield of quinine sulfate in 0.05 mol/L I^SOi* [25 , 26 , 33,71,92] or 0.5 mol/L H2SCU [21] is constant within experimental error (±5%). Our results [33] are summarized in figure

7 and show that the relative photon yield is essentially

constant over the 224-390 nm excitation wavelength range

which agrees with the summary of the past data [26]

Thus, although the emission spectrum of quinine sulfate

shifts to longer wavelengths upon excitation at wavelengths

longer than 360 nm, the quantum yield remains constant, an

important fact when considering quinine as a standard.

3. Photon Yield as a Function of Acid Concentration

It was reported that the photon yield of quinine sulfate

changes with varying sulfuric acid strength [24,80]; an

observation that has since been verified [33,75]. Additionally, Chen reported that the fluorescence lifetime was also

dependent on the sulfuric acid concentration [75]

In a recent study [93] we found that although the photon

yield and fluorescence lifetime were functions of the sulfuric

acid concentration attributable to quenching by the sulfate

ion, use of an acid with a "non-quenching" anion such as

perchloric, phosphoric, or nitric resulted in a photon yield

and lifetime that were essentially independent of acid concentration, Table 11. Increasing the HC10i+ concentration to

10 mol/L may result in a slight increase in Q (0.59 to 0.62),

but over the acid concentrations that are commonly used

51
(0.1 to 1.0 mol/L) , Q is constant and does not exhibit a 10

percent increase as is observed over the same acid concentration range in H2SO4

.

Table 11. Fluorescence Photon Yields Q and Lifetimes t of

Quinine Sulfate in Various Acids [88]

.

(H

+

) H 2 soi, HC10 k HNO 3 H 3 P(H

mr>1 / T JUL) J. / Li n t , ns t , ns Q T,ns Q x , ns

.01 0.52 20.1

. 1 0.51 19.2 0. 59 21.6 0.61 22.0 0.59 22.1

1.0 0.55 20.2 0. 59 21.2 0.61 22.4 0.58 22.0

3.5 0.57 20.5 0.61 21.7

5.0 0.58 20.8 0 . 61 21.7

10.0 0.60 21.9 0.62 21.8 0.60 22.1 0.60 21.8

Thus

,

perchloric acid was chosen over nitric and phosphoric acids as the solvent for use in certification measurements

because: (a) it forms one anion in aqueous solution as opposed

to phosphoric acid which forms two; (b) phosphoric acid is

somewhat viscous and difficult to work with; (c) the large

absorbance tail due to the nitrate ion where quinine absorbs

necessitates large corrections in absorbance measurements

when relative photon yields are determined; and (d) nitric

acid is a fairly strong oxidizing agent.

H . Emission Spectrum as a Function of Acid

In these discussions, it usually is assumed that changing

from H2SCK to HClOit does not affect the envelope of the

emission spectrum. To test this, measurements of the emission

spectra were made in the usual time symmetrical mode using

the same concentration of quinine sulfate (^1 ppm) in 0.1 mol/L

HCIO4 and 0.5 mol/L H 2 S0 4 under exactly the same instrumental

conditions. A small difference in the maxima for both curves

52
in the uncorrected spectra Y^y ^ ^ was observed: 451.5 nm

for the 0.1 mol/L HC10 4 and 452.7'nm for the 0.5 mol/L H 2 S0 4 . Additionally, plotting the ratio of the normalized uncorrected

spectra, (Y^y

5 ^ ) HClCU/CY^y 5 ^ ) H 2 S(H shows differences

from approximately +3 percent towards the blue region to -4

percent towards the red region, figure 11. Thus the emission

spectrum of quinine in the two acids is different. This

observation is not important when measurements of photon yield

are made relative to a standard since the area under the

emission curves are used; however, when using the emission

spectrum to check instrumental operation, correct instruments,

or determine the emission spectrum of other materials relative

to the standard, the emission spectrum of the standard is

important and care should be exercised in the choice of the

acid as solvent and in data comparisons .to reported spectra.

I . Emission Spectrum as a Function of QS Concentration

Two phenomena that can affect fluorescence spectra are

quenching and inner filter effects. Luminescence quenching

is a fundamental effect that diverts the absorbed radiant

energy such that the photon yield is less than one. In

general, it does not affect the shape of the emission spectrum

unless the fluorescence lifetime approaches picoseconds or

unless a complex is formed with the quencher that emits over a different wavelength region than the uncomplexed fluorophor.

Inner filter effects, conversely, are instrumental and

experimental conditions that do affect the fluorescence

spectra or measured flux in three ways: (a) the exciting

radiation can be absorbed; (b) the emitted radiation can be

absorbed; and (c) the absorbed radiation in (b) can be

emitted at different or the same wavelengths.

Assuming there are no impurities at the 0.01 percent

level (Section VI-B) and the solvent does not absorb, then

the concern is mainly with the effect of increasing solute

concentration. In general, absorbance of the excitation

53
1 .07

1 .05 h

I 1.03 I- S3

S^ 1 -01 - or—

o

EC

d

0.99 -

0.97

0.95 -

0.93

350 400 450 500 550 600 650

WAVELENGTH, nm

Figure 11. The ratio of the uncorrected emission spectra of

quinine sulfate in 0.1 mol/L HClOi* and 0.5 mol/L

H 2 S0 4 to demonstrate the effect of acid on the

emission spectrum.

54
radiation by the solute (or impurities or solvents) results

in errors in the determination of fluorescence flux, excitation spectra, and photon yields, since, as the absorbance of

the radiant flux I increases, the dilute solution approximation, equation 4, no longer applies. This effect is depicted

in figure 12 as a plot of the compensated fluorescence

X S reading (Y-^y

^ 450^ versus quinine sulfate concentration.

Deviations from linearity of greater than 1 percent are not

observed until the concentration of quinine sulfate is ^2-3

ppm. A least squares fit of this data using (—) as weighting

factors produced a straight line in which the standard predicted values differed from the experimental values by an

average of ^0.8 percent over the 1 to 0.001 ppm range. The

problem of excitation radiation absorption by the solute has

been discussed previously and the following equation was

proposed to correct this effect [94,95]:

, 2.303A(d 2 -d 1 ) n

_. VF " 1Q -Adi . 1Q -Ad 2 U33

where

F

q

= true Y (no inner filter effect)

,

F = measured Y,

A = absorbance in the solution at X 1

, and

di,d 2 = the distances the two sides of the emission

monochromator slit images are from the cell face.

The absorbance of a 1 ppm solution of QS*2H20 at X\ = 347.5 nm is 0.0138 (WA = 13.834xl0" 3 g solv (g sol

)

1 cm

~

1

> Table 6) . Assuming that the emission monochromator slits

are 2 mm and a 1 cm cell is used, di = 0.4, and d 2 = 0.6.

Substitution of these values into equation 13 yields a value for F /F of 1.016 which shows that the observed Y

o should be ^1.6 percent low as compared to the emitted radiation if no effect were present.

The data for the 5, 10, and 100 ppm solutions were corrected according to equation 13 and these values are plotted

55
in figure 12 as hourglasses that now fall on the linear

portion of the curve, thus demonstrating the usefulness of

equation 13.

Re-absorption of the emitted radiation by the solute in

optically dense solutions [80,96-99], plane parallel luminophors [100], and by other species [94] has been discussed and

correction factors have been derived in some of these papers;

however, in the case of solute re-absorption, the emission of

secondary fluorescence occurs which also distorts the emission

spectrum in the red region as well as in the overlap region.

These effects are demonstrated by the uncorrected emission

spectra for 1 and 10,000 ppm solutions of quinine sulfate,

figure 13. A large effect due mainly to re- absorption with

some compensation of the effect by re-emission resulting in

decreased 'Y' values is observed in the overlap region (350-

425 nm) , while a smaller effect due to re-emission resulting

in increased ' Y' values is observed at longer wavelengths.

These and additional data are presented as the ratio of the

spectra for X ppm QS*2H 2 0 to that for 1 ppm QS«2H 2 0 in figure

14 from which the extent of the effect is readily apparent.

Equation 13 can be used as a first approximation to

determine the absorption of the emitted light in the overlap

region. The one percent fluorescence signal level (relative

X S

to the maximum for CY347 5 \

} occurs at A 2 - 390 nm. The

absorbance at this wavelength is ^1/10 that at 347.5 nm, and

considering the crosshatched illuminated volume element in

figure 15 as a source and the absorbing depth as (di+ d 2 )/2,

then A ^0.00069. Substitution of these values into equation

13 gives an F /F ratio = 1.00098 or an error of approximately

0.1 percent which, at the 1 percent signal level, is considered negligible compared to the imprecision and errors

in the overall measurement process.

56
o

55

100 E

10 -

1 - Q

<!

go- 1 >

£10 -2 _

05 10 -3 _

10

10

* EXPERIMENTAL DATA

E CORRECTED DATA

» i 1

10"2 0.1 1 10 100 10 3

CONCENTRATION [QS-2H^0] , ppm

Figure 12. Relative fluorescence flux * and relative

fluorescence flux corrected for inner filter_

effects (absorption of exciting radiation) , X

as a function of quinine concentration.

57
58
o

o

o o o°r- LO

o o

o

oloo

O O «

ID

LO

eg

LO

E

c

LO 5>

>

CD

LO

CM

00oddCD CM

d

LO

p 00

d

CD H

o

•H t f J

II •H '"O H " * CD

C7" i—3

1 H

m i—03 r> o o +* o o CO O 1 TO •H O +->

+-* !~H TO

*—

TO X

i i

r-i 1

0

be M •

i C i—

o CD

CD • H rH

r \ II CO

(/) >

O ^ •H to

1 u s Is CD

CO 1—] f 1 1 -J /i^ QJ CJ O rH i •H O +-* H O

co" O •H

+JO

M O

O rH 1

1 H

f | j LO (—

to • O nJ

+-> vo e

CO

II cS

i—1 o CO CD

TO rH 4-*

f-< •

*> X ctj

-t-> 1 1 f i i H—

u ^ * rH

CD r—

rH

P-i o CO CO II CDc O i •H

•H O

in rH i—•H

CO X 3 J- •H tO rH cro (

1 o v—

CD T—

CD II rH rC i O ,-1

4-> • o e

I—

rH

( 1 i X«> o

O CD K) I e

4-> o

O TO" i—

rH LO •H X o

4-> rH II ro rH

CTJ 3

C£ CO 4 rH o

CD

Sh

•H

59
1 '°

I

T2 - —Ud 2 —

Ul -—

—1

1

Figure 15. The geometrical

M

representation of the sample

cell and illuminated volume element in

considering the distortion of the emission

spectrum by absorption of the emitted

radiation followed by re-emission.

We therefore recommend that emission spectra be obtained

for quinine sulfate solutions that have concentrations of

1.28xl0" 6 mol/L (1 ppm) or less.

J . Emission Spectrum as a Function of Temperature

Melhuish reported [7] that a 5 x 10 3 mol/L solution of

quinine bisulfate in 0.5 mol/L H 2 SCU viewed in a front

surface mode exhibited a change in "fluorescence intensity"

of -0.2 to -0.3 percent per degree in the +10 to +40 °C

range. This value represents an average change per degree

for the total emission spectrum and not the variation as a function of wavelength since in that case a rhodamine B

quantum counter was used to measure the emitted radiation.

60
The uncorrected emission spectrum of a 1.28 x 10 6 mol/L

Ol ppm) in 0.1 mol/L HC1CU in the side view mode was measured at 16.0, 23.8, 27.8, and 34.6 °C. Plots of the compensated fluorescence signal (Y) at selected wavelengths are

given in figure 16 and the CO x °c/ 00 23 3 °q ratios are

given in figure 17. The average change, (AY) per degree at

25 nm intervals, is summarized in Table 12. The average

percent change per degree using the 375-600 nm data from

Table 12 (useful wavelength region for rhodamine B - acriflavine quantum counter 220-600 nm) is -0.21 percent/°C, in

good agreement with the previously reported value of -0.2 to

-0.3 percent [7]. These calculations were made assuming

that the change at individual wavelengths is linear as a function of temperature. This assumption generally holds

from +20 to +30 °C over the wavelengths of interest,

figure 16.

Table 12. Average Percent Change Per Degree Celsius for the

Uncorrected Values (AY)^^^ g at 25 nm

Wavelength Intervals.

A,nm (AY)/°C

375 +.40

400 -.40

425 -.48

450 -.62

475 -.33

500 -.25

525 -.19

550 -.14

575 -.05

600 -.01

625 -.02

61
0.95

-a 0.61

£ 0.60

S 0.59

E 0.58

<S 0.57

Figure 16

20 25 30

Temperature, °C

The compensated fluorescence signal

(Y) at selected wavelengths as a function of temperature.

62
r T" I —i i i

X- •4-X M

• 4X M

MK «• ft MX «4- SZ 4- M •ft M S 4- M X ft KM 4- * MX 4» M X«4- M X ft 4- M X ft 4- M X ** M X # 4- M X ft 4- M X ft4- M X ft 4- M X ft 4- M X ft 4- M X ft 4- M X # 4- M X ft 4- M X ft ft rw* Tw> t

T WJ M X ft 4- e 0 o 0 M X ft 4- oo oo eo OO M X ft 4- pi n CO ri M X ft 4- cm N 01 M X ft ft u U CJ M X ft ft 0 0 e e M X ft ft on © (0 M X ft ft fA

S3

i M X ft ft «H CM M X ft ft M • M X ft ft M X ft ft M X ft ft M X ft 4- M•"1 X ft ft M X ft 4- M X ft ft M X ft 4- M X ft ft M X ft ft M X ft ft M X ft ft M X ft ft M X 4-ft

4- x m «• X ft M

1 1 • i 1 1 1

o

ID

CD

LD

CM

COO

o

CD

IDLDO

LDLD

LD

CMLDO

LD

O

LD

LD

CNJ

OO

LDO

CO

O

LD

B

d

a

o

w

>

o OO CO 1* CM O 00 CO

• o• o• o• O O CD CD CO ^4 • •—4 O O o

Do 9^(A)/^ X(A) OLLVH

3

W)

•H

63
We are concerned, however, with the temperature effect

on the normalized spectrum measured at 25 °C and not changes

in absolute values. The spectra at the four temperatures

were normalized to the (Y) value at their peak maximum, Table

13, and ratios of the normalized readings (Y) ^/ (Y) ^=23 8 °C

were plotted as a function of wavelength, figure 18. The

solid lines represent the best fit to these data using 7th

order polynomials [56] . Other fits using lower order polynomials gave plots of the standardized residuals which showed

cyclical variations as a function of wavelength, figure 19a,

while the 7th order polynomial gave random standardized

residuals, figure 19b, indicating a reasonable fit.

Table 13. Emission Peak Maxima and (Y)^ 0(~. Values

for Quinine Sulfate in 0.1 mol/L HC1CK

at Various Temperatures.

X S Temperature Emission Maximum (Y c

, ) L A , nm 7

16.0 461.4 1.0683

23.8 461.8 1.0330

27.8 461.9 1.0140

34.6 462.1 0.9940

It is apparent that the ratios and thus the percent error

observed is a function of wavelength as well as temperature.

This error is experimentally determined and incorporates

various parameters that also change as a function of temperature, i.e., solvent refractive index, absorbance, and solvent

expansion with concomitant decrease in solute concentration.

With normalization, the last contribution is eliminated.

Although as demonstrated by figure 16 the plots of the

changes in the absolute values as a function of wavelength

and temperature are essentially linear, normalizations of

the emission spectra to peak maxima and 23.8 °C magnify small

64
o

LOCD

LOCM

COoo

CD

ID

LDO

LDLD

LD

CMLD sOW

•JW> o

LD

LD

CM

-3" OO

LD

COO

LO o CO CO CM o CO CD CMfO • o o o o o cn CD CD• cn o o o o

Pio

•H

+-> U

P!

cd

03 0

CDo

PiCDU

•P

PJ •HO o3 10 +-> 03 U bfl T3 o 3

CD

O

•M oo 5h 03 • XI

t/) to +-> PJ c^J CD +-> P.-P "H E 03 m oU CD 4-> V) C/) 13 O OJ CD X

•H CD O rC

03eo

PJ

(A CD

03

CD

CDSi

03 CD

P!

CD • H

4H PhiH o 6

CD

H

75

O 5 (/) oH CD

CD o3

H

X >

«4H 03

X

O 4-> (/)>-<£

CD

v

—-' CD P rH H VI D

03 rH > > 03 03

CD W) •h

m

to

O

Do 9^(^/0, X(

A) oixvh aaznvRHON

oo

<DpH

65
3.75

75 2.25

3

o

'</> 0> oc .75

D

Q)N

1 -.75

(0 •ac

ro ^ -2.25

-3.75

3.75

I 2.25

3

"O

W

cc .75

0)N

1 "-75 1 o

c

CO 55

-2.25 -

-3.75

i 1 r

• • •

» • • • • • • • • • • • • •• • • •

— •

— •

375 500

Wavelength, nm

625

, 1 I I

1 I:

B •

• • •

•

-•

•

• • •

•

••• •

• •• • • •• •

• • •• •

• • • • • • •

• •

• •

• • —

• • •

•

•

•

-

i ill

• •

• •

1 1 375 500

Wavelength, nm

625

Figure 19

(a,b)

Standardized residuals for a fifth order

polynomial fit (19a) and a seventh order

polynomial fit (19b) for the ratios of

the normalized compensated fluorescence

signals (Y) at various wavelengths.

66
deviations from linearity and plots of the normalized ratios

{ [ (Y) T

/ (Y) T _23 g-' -1 ^ versus temperature at selected wavelengths produce figures 20a and b that demonstrate this

magnification.

Since a 10-15 percent error in linearity produces a small

error in a small correction, linearity can be assumed between

20 and 30 °C and the percent change per degree Celsius

[ (1/Y,p) (AY^/AT) ] -j, = 25 oq can be graphically approximated by

obtaining the slope at 25 °C for each wavelength depicted in

figures 20a and b in the following way:

Let

Y

23.8

= Vrp and graphically determine

V

26 2 and for each wavelength. Thus

V

26.2

Y 26.2

23.8

-1

Yx Y 26.2- 23.8

Adding 1 to gives

V

25

+ 1 = 25

23.8

Y

23. 8

+ 1

26. 2

23.8

AY,

23.8

(14a)

(14b)

Dividing (14a) by (14b) and dividing by 2.4(AT), we find

that

T=25.0

(14c)

Values, calculated using eq. 14c at selected wavelengths,

are summarized in Table 14 and are presented graphically in

figure 21 and represent the percent error per degree Celsius

that would be observed for the normalized emission spectrum

of QS in 0.1 mol/L HC1CK as a function of wavelength over

the temperature range designated by 25±5 °C.

67
k, nm

20 25 30 35 Temperature, °C Figure 20. The values of {[(Y)

T /(Y) 23 g

]-l}10 2 at selected

( a >b) wavelengths as a function of temperature.

68
.50 roo

to

CM

ii H

<

-0.10-

400 450 500 550 600

Wavelength, nm

650

Figure 21 The average percent error per °C observed in the

emission spectrum of quinine sulfate over the

temperature range 2 5 ± 5 °C.

69
Table 14. The Percent Change per Degree Celsius for the

Emission Spectrum of Quinine Sulfate in 0.1

mol/L HC1CU at 25 °C at Selected Wavelengths.

A / ayt\1

v at

/J

380 + .05

385 + .03

390 + .01

395 - . 03

400 - .07

415 - .11

425 - .14

435 - .08

JT=25 °C

A r/i \ / ayt\1

l AT

J

450 - . 01

475 + .08

500 + .14

525 + .17

550 + .22

575 + .28

600 + .32

620 + .36

T=25 °C

VII. Corrections for Instrumental and Experimental Parameters

A. Photomult iplier Tube Linearity

The measurement of the non-linearity of the photomultiplier tube response caused by a dependence of the photomultiplier gain on the incident radiative flux has been reviewed

[47,101,102]. The double aperture device and "cascade"

measuring procedure was used to determine the non-linearity

of the total detection system of the reference spectroradiometer. The photocurrent measured with "A" open plus the

photocurrent measured with "B" open should equal the photocurrent measured with "A and B" open (see Section V-C3).

The differences obtained between these measurements

defines the non-linearity of the detection system, and for

amplifier gains of 10 6 and 10 7 V/A, the following equations

represent the relative correction factors applied to ( Y ) m

> respectively

Of' .'.1 (15)
(Y) = (Y) il+Tl-CY) /(Y) 1.0028 (Y) (, (16a)

(Y) = (Y) il+Tl-CY) /(Y) 1.00028 (Y) I . (16b) ^

; corr v ^m \ [_ ^

J w v -'maxj c J max j

J

The correction factors were as large as 0.3 and 0.03 percent

for the photomul tiplier tube used at amplifier gains of 10 6 and 10 7

, respectively and are given in figures 22a and b.

The non-linearity of the photomultiplier tube was determined at 450.0 and 695.0 nm and results verified that the

correction is wavelength independent for this type of photomultiplier tube at the 1 part in 10 1* measurement level

[49,102]. These corrections were applied to all measurements

made on the the NBS Reference Spectroradiometer

B . Emission Spectrum as a Function of Bandwidth

A rule of thumb in spectrophotometry that is also applied

in spectrofluorimetry is that if the bandpass of a monochromator is 1/10 to 1/20 that of the measured peak half-height

bandwidth (HHBW) , the measured values are accurate to ^0.1

percent at the peak maximum but only to 5 percent at the wings.

A procedure to correct spectra for bandpass error was

developed [49] using numerical differentiation of the

emission spectrum based on an earlier method of Hardy and

Young [103], The derived equation is:

Y(A) = Y

m

(X) - 1/12 (AA) 2 Y^' (A) + ^(AA) 1* Y^" ' ( A) (17)

where: Y(A) = true readings,

Y (A) = measured readings.

AA = bandpass of the monochromator , and

Y

m '(A) and Y^' «

1 (A) are the second and fourth

derivatives of the emission spectrum calculated

by the method of Rutledge [104].

i 71
1 .00030

>>

E-i—i 02<

E0*^"oi—

E-O

W

02

028 1.000051-

1 .00025

1 .00020

1.00015

1 .00010

>I—

w

02

1.00000-

0.999951-

E-" 02

<;w

i—

J

EOh

o

E-O

02

02O 1 .COOSE- w

>

1 .002E1 .002C1 . 00 1 E1 . 00 1 CE02

1 -000C0.9993-

xm

xx

**

x*

XX

9K V

AMP. GAIN. 10 V/A

350

1 .0030—

400 450 500 550 600

WAVELENGTH, nm

650 700

X

XXX

XX

X

x

x

x

x

AMP. GAIN, 10 V/A

xx

350 400 450 500 550 600

WAVELENGTH, nm

650 700

gure 22. Relative photomult iplier tube linearity

(a,b) correction factors for amplifier gains

of 10 7 (22a) and 10 6 (22b) volts/amp

applied to the compensated fluorescence

signal (Y)

.

72
The odd derivatives, Y ' (A) , Y

1 ' ( A) . . . , cancel out due to the '

jti ' m

v - 1 triangular bandpass function produced if the width of the

two monochromator slits are equal. The bandpass used in the

measurement of the tungsten lamp spectral radiant flux was

^1/450 of the HHBW for the "measured peak" and no correction

was necessary. However, the bandpass used in the measurement

of the emitted flux from the quinine sulfate in solution was

^1/18 of the HHBW for the emission peak and the correction

factor by the numerical integration procedure was calculated

and applied. Equation (17) was truncated after the second

order term since the fourth derivative term contributed less

than 0.1 percent to the overall correction.

A typical correction derived from the second derivative

function of the emission spectrum is plotted in absolute (Y)

values versus wavelength in figure 23a and the relative correction applied to the spectrum is illustrated in figure 23b.

C . Wavelength Correction

The detection monochromator wavelength errors (differences

between the measured wavelengths and the true wavelengths for

the line sources, Section IV-C) and the best polynomial fit

for these errors are graphed in figure 24. Selected (Y) data

at small wavelength increases (^0.03 nm) are summarized in

Table 15 for the "blue edge" of the uncorrected emission

spectrum of quinine sulfate. The difference in (Y) , A(Y),

for a total wavelength change of 0.295 nm is 0.005614 units.

From this, the error in (Y) was calculated to be 4.6 percent

per nanometer in the wavelength reading. Considering the

wavelength region of interest for the quinine sulfate emission

spectrum, the largest errors in the wavelength calibration

occur at 375 nm (-0.04 nm) and 715 nm ( + 0. 077 nm) . Errors

in QO would, therefore, range from -0.18 to +0.36 percent.

We are, however, interested in the error produced in the

final reported value of E(A) and not the error produced in

(Y) since the wavelength error associated with the measured

m

73
0.003

m

E-O 0.002

S5 0.001 - Oi—

E-O

W 0-000

PhOh

o

O^ -0.001

Q

S -0.0021-

03 -o.oo:

)50 400 450 500 550 600 650 700

WAVELENGTH, nm

E-Qi—

&Q

m

. o

1—1 o

So

o o o

02

1 .00

t r t r

98 -

96 -

94 - >

E-5

K 0

0.92 -

90

J*^ ^^^"^"IWWtfK Hi a» X

400 450 500 550 600 650 700

WAVELENGTH, nm

Figure 23. Bandwidth correction in compensated

Ca,b) fluorescence units (Y) (23a) and the

relative correction factor applied

to the quinine sulfate emission

(23b) as a function of wavelength.

74
0.21

0.18

0.15 -

0.12 -

0.03 -

0.00 -

-0.03

-0.06 -

-0.09

-1 I 1 1 1 1 1 X

*

Y Zn

X Hg m

i X He

"T" + Rb

a Cs X

_ £. La

+ i

Ne

X QUADRATIC FIT

IK

A A,

1 1^ ( Z *x

X

z

X

A

A X

X X

Y X * * A

X X

X X

v

Y XX

z

X

X

1 1 1 1 1 1 1 150 250 350 450 550 650 750 850 950

WAVELENGTH, nm

Figure 24. Wavelength deviations of the emission monochromator at 34 wavelengths using line sources.

The stars represent the quadratic fit and these

values were used to correct the quinine sulfate

emission spectrum.

75
error in the emission detector signal (R) , used in calculating

M-^ may be either compensative or additive.

Table 15. Quinine Sulfate Emission Values (Y)

as a Function of Wavelength.

A, nm m

415

.

0\J %J5 5 411944

41*T -L %J5 • X X «J • *t41X L,76\j 7I O5

yi i r 415

.

1 A P 14 5 At 1 *7 O A

. 412784

415. 205 . 415470

415. 235 .415675

415. 260 .415808

415. 290 .416142

415. 370 .417509

415. 350 .417558

For measurements of (Y) and (M) we find that:

/d(Y) \

( Y

>T = &\ +

( A E) ( -dxf) ^a)

and

/d(M) \

(M) T

= (M) m

* (AE) C18b)

where 00 -j- and (M)^ are true values,

00 is the measured compensated fluorescence signal,

CM) is the value calculated from the measured

detector signal ( R ) m

>

Ag is the wavelength error, and

dOO d(M]

and —g-r—- are the changes in the (Y) and (M) dA 2 dA 2 — — ^ Jja —

values as a function of A2 at the

wavelength measurement.

76
The corrected emission spectrum in radiometric units E (A) is:

E

A (A) =

(Y)

d(Y)

(Y) + (AJ —rr-

_ ^m _ E

y dA 2 d(M)

ID

m

m

^ ^m

m

1 * (X

E

)

1 + (A £

)

dA

d(y)

m

dA

d(M)

m

dA

(18c)

By Taylor expansion and truncation, using only terms of first

order in A^ (A* is ^30 times smaller than the Ag terms)

,

the relative error in E (A) is:

AE (A) = T

R

00

(Y)

m

CM)

(Y)

m

dm

m

0O m A dXa

i yw«m

. (18d)

The errors in E (A) were calculated using equation (18d)

and were subtracted from 1.0 to give the relative correction

T values, figure 25, that were applied to the E (A) values.

The magnitude of the corrections range from ^0.5 to ^0.2

percent with many being close to zero.

The source monochromator was calibrated in a similar

manner and found to be in error by +0.07 nm at 347.5 nm

which was taken into account when setting Ai on the source

monochromator

77
1 .001

W

§ 1 .000

a

> 0-999

§ 0.998

i—

E-» O

§ 0.997

PSO

°

0.996

0.995 -

05

0.994

350 400 450 500 550 600

WAVELENGTH, nm

650 700

Figure 25. Relative correction factor applied to the

compensated fluorescence signal (Y) to

correct for the wavelength error.

D . Dispersion Correction

The grating dispersion of a monochromator is wavelength

dependent and normally a correction [49] is applied to

spectral data since the bandpass, as defined by the grating

dispersion and slit width, is a function of wavelength. In

this case, however, the correction must be made to both

(Y^S c , ) and (R? ), and since E

T

(A) is:

78
where ML

= uh 9

the correction cancels out.

E. Solvent Refractive Index

Although the radiation is emitted isotropically within

the solution and within the cone of radiation subtended by

the collecting mirror EL2, figure 1, the fraction emerging

into the surrounding air depends on the square of the

solvent refractive index, n

2 [105] . In relative quantum

efficiency measurements, the solvent refractive index used is

generally the value at the emission maximum [26] ; however,

the refractive index is a function of wavelength [106,107]

and the emission spectrum should be corrected by multiplying

the compensated fluorescence signal (Y) by the value of n

2

at A2. The difference between the refractive indices of

water and 0.105 mol/L HCIO4 is ^0 . 1 percent [106]; thus the

index of refraction values for water [49] were used to correct

the emission spectrum since the total index of refraction

correction is ^2 percent over the emission spectrum of

quinine sulfate and the error introduced by this approximation is small (0.002 percent or less).

The values of n

2 for water at different A 2 's were

calculated from equation 20 [49] :

n

2 = 1.7604457 + 4.03368 x 10" 3 A - 1.54182 x 10

_2

A

2 + (20)

6.44277 x 10" 3

A

2

- 1.49119 x

10" 2

where A is in nm.

79
The relative values for the refractive index corrections

are presented graphically in figure 26.

F . Cuvette Window Transmittance

In a manner similar to the refractive index correction,

a correction was also made for the variation of the transmittance of the cuvette window as a function of wavelength.

The equation derived for this correction at a given emission

wavelength, X 2 , is [49]:

4nn'

(21)

[n 2 + (n') 2

] + n

! (1 + n

2

)

the transmissivity of the quartz window at Xz,

the refractive index of quartz at \z , and

the refractive index of water at A 2 .

where t' = n =

n' =

The refractive index of quartz as a function of wavelength was calculated using the equation given by Malitson

[108] and t,' was calculated by substituting the appropriate

A 2 n' and n values at given A 2 's. The correction is on the

order of 0.2 percent and the relative cell transmittance

correction is plotted versus A 2 in figure 27.

G . Composite Correction

The composite errors as a function of wavelength for

the preceding instrumental and experimental parameters are

given in figure 28a. To show the magnitude of the corrections applied to either (Y) or E ( A) in section VII, the

composite corrections [excluding the correction for spectral

responsivity) as a function of wavelength are given in

figure 28b.

80
1 .03

0.97

350 400 450 500 550 600

WAVELENGTH, nm

650 700

Figure 26. Relative correction factor applied to the

emission spectrum to compensate for errors

introduced by the refractive index of the

solvent

81
6

Sh

+-> C/) U oJ

cua. cu

t/i u

O +-> •H +->

C/) -H

•H t/) £ fi

CD o3

CD +->

4-> r-l i—

O <D

+-> U

CD O

•HH C

o

o3 +-> 03 ^ -H O ^

•(-> 03 U >

03 MH CD +> C -M O CD •H Jh iH

+-> O CD u m >

CD 03

CD ^ +-> O o3 Mh u tn O

CD CD > ao •h g H

-P O +-> 03 U o

i—lCD O

'X +-> <4H

•

CNI

T^I 'N0LL33HH0D 3AUV13H

CD

•H

82
1-10

PS

o

Ph 1 -07

Ph

r_T

w

E- 1 .04

i—

(/3 O

Oh

a 1 .01

o

o

o

O

W

Ph

PhO

o

w

Hi—

CO

o

Oh

O

0.98

350 400 450 500 550 600 650 700

WAVELENGTH, nm

1 .02

0.99 -

0.96 -

0.93

0.90

350 400 450 500 550 600 650 700

WAVELENGTH, nm

Figure 28.

(a,b)

Composite errors that occur in emission spectra

measurements (28a) and the composite correction

factors applied to the emission spectrum of

quinine sulfate to correct for various instrumental and sample parameters (28b)

83
VIII. The Spectral Responsivity M, of the Detection System,

A 2 rp the Technical Emission Spectrum E (X), and the

Molecular Spectrum E(X) of Quinine Sulfate in 0.105

mol/L HC1CU

S The determination of the spectral responsivity M, , the

rr. A 2 technical emission spectrum E (a) (the emission spectrum

corrected for instrument parameters) , and the molecular

emission spectrum E(X) (the emission spectrum corrected for

instrument and sample parameters) could be discussed in two

separate sections since M, is a correction factor for the

A 2 rp response of the detection system and E (X) and E(X) are

emission spectra for the quinine sulfate. However, we have

chosen to combine them because their determinations are

closely related with respect to measuring procedures and

data reduction.

In general, data reduction was performed with the objective of improving measurement precision by eliminating

instrumental drift or by assuring that a maximum measured

value of (R) or (Y) , used as a normalization factor, was not

in error so that the normalized data would not be biased by

one inaccurate data point. Additionally, the accuracy of the

data was increased by applying theoretically sound corrections

to the measured data for instrumental and sample related

errors. Attaining increased precision and accuracy are interdependent and it was not possible to average or smooth all

data first and then apply all corrections. For example, since

the correction for photomult iplier non-linearity must be made

on the measured values of (Y) and not on the normalized (Y)

readings, the correction was made before time symmetrical

averaging, which was, in turn, performed before data normalization. Thus, the following sections will be somewhat

chronological in order. Also, discussions will be included

where relevant.

84
A. Measurement and Averaging of Data

S Values of for the standard lamp at the two operating

X S currents (8.8030 and 6.0000 amps) and Y *

, for the three

A 1 , A 2 series of quinine sulfate solutions were obtained' as Runs I,

II, and III according to the time symmetrical measurement

sequence illustrated in figure 7. The data were corrected

for photomult iplier non-linearity and then were averaged as

indicated in figure 7.

B . Data Smoothing

To avoid errors in the (Y) or (R) value chosen for normalization, attempts were made to "fit" the emission envelope

data by polynomial equations (up to 12th order, reference 56),

a modified spline technique [109] , or an exponent ial -polynomial

combination [110] to obtain a predicted (Y) or (R) value at

the peak maximum, designated A . These procedures were

unsuitable since the imprecisions introduced by the fitting

techniques were larger than the intra- and inter-run

imprecisions

However, several data points brackeucing A

n

[111] could

be fitted using a 3rd order polynomial [112] yielding a calculated (Y) value that was accurate to better than 0.1 percent.

The predicted values of (Y) or (R) at A for the spectra from

n n n

Runs I, II, and III then were used to obtain initial spectral

normalizations. The normalized spectra were averaged and for

Runs I, II, and III the following data were available:

S

1. the averaged, normalized data (R, ) for the two

A 2 n

standard lamp currents before measuring the ten

quinine spectra;

2. the averaged, normalized, compensated values

X S

(Y, ' . ) for the ten quinine sulfate spectra; and

a i , A 2 n „ 3. the averaged, & 7 normalized data (R, ) for the two

A 2 n

standard lamp currents after measuring the ten

quinine spectra.

85
s C. Spectral Responsivity M,

The spectral responsivity of the detection system was

determined by:

S _ (R

A2 } n (2 2) M

A 2 L/L

n

where: L is the spectral radiance of the calibrated

tungsten lamp [58] in W (mm) 2 (sr)

1 (nm)

1

, and

L is the value of the spectral radiance at A . n r n

The values of the spectral responsivities obtained

S S before, ( M

^ 2 )g> an^ after, (M-^ measuring the quinine

spectra were compared as an indication of instrument stability

S S during a given run. The ratios ( M

^ 2

) A

/(M

A ) B for Run III

are summarized at selected wavelengths in Table 16 and show

agreement that is on the order of the precision for a single

data point during a particular run [113] . The ratios at

400 nm (6.0000 A data) and 350 nm (8.8030 A data) show a little more variation, however, this is due to the low signal

values at these wavelengths (II or less of the maximum signal),

S In order to obtain a single M, , the two sets of data

A 2 (350 to 425 nm and 400 to 900 nm) were spliced by using the

average ratio of the normalized values in the 400 to 425 nm

overlap region and spectral fitting procedures to produce a

S calculated M, for the wavelength interval 350 to 900 nm for

S each run. The M, values for Runs I, II, and III then were

A 2 averaged and evaluated statistically to give an average

S spectral responsivity for the detection system (M^ ) and

the relative standard errors (RSE) in these values. The

results are summarized in Table 17 and depicted graphically

in figures 29 and 30.

86
Table 16. Typical Intrarun Instrument Stabilities as

S Determined by (M, ) Values for the Standard

A 2 Lamp at the Two Operating Currents.

- - - rcMf

2

) A]/[CM^) B

] - - -

X , nm 6.0000 amp 8.8030 amp

350 0.9968

370 1. 0002

375 . 9985

400 . 9964 1.0003

410 . 9983 1. 0006

420 . 9982 1.0000

450 1.0025 —

475 1. 0002

500 1. 0003

525 1.0013

550 . 9997

575 . 9995

600 . 9998

625 . 9990

The shoulder observed at ^375 nm in figure 30 is not an

artifact of the splicing procedure since it is observable in

the 8.803A data. The RSE values average ^0.17 percent over

the total wavelength region measured (350-900 nm) and are

less than 0.1 percent over most of the wavelength region of

interest (375 to 675 nm) which demonstrates excellent system

stability.

87
m

wCO

to to 00 >st oo

r-» oo o oo tx i—i o~> ct> cxi OO 00 (XI (Nl «sj- tO O OO \0 "St

-st tO Ex "st CM to to to *st lo

CO N \0 LO to «st "st *t to to to cxi eg cxi CXI CXI (X] CXI CXI

U m

m

oi o to to cxi i—i o o o o

o o

cxi to to to to o o o o ootO CXI CXI i—l (—1 o o o o o

o

o o o o o

lo cxi go tx to O i—I CXI LO o O O O O i—

O N Ol H o NLf) l/l N OO

I—I CXI tO ^t LO

st CXI "sj- O "=t

o~> cn tx "st oo

CTi ix cxi tx ix

cn cn oo "st cri

CTi CTi CTi CTi OO

H

mCO

Ph

O "St CXI O VO

"st LO [x CXI vO

tx to cxi lo o i—i t— to CTi

CTi t—I Cx OO tO

CXI O [>- LO "st tO N WO NtO tO tO "st t

0O IX iO \£) W LO ^t "st tO tO to to CXI CXI CXI CXI CXI CXI CXI CXI

EL

w

cr> \D to to cxi i—I o o o o o o

cxi to to to to o o o o o

o

to (XI CXI i—| i—

o o o o o oo o o o o

o o

V I H

m

LO CXI 00 IX tO O r—I CXI LO O O O O O i—

at o r— at to \0 LO LO \0 00

l—I CXI tO "St LO

CXI o cxi oo to

CTi Ol ix IO O0

vO tx oo CTi CTi

00 CO tO CT) 00

cn CTi OO "st CTi

CTi CTi CTi CTi 00

as pq P CO

t*— oo to CXI CXI i—I o o o o o o o o

i—l CXI CXI CXI CXI o o o o o o

o o o o o o o

i—I O i—l i—I i—

o o o o o

to r- cx o o O O i—I to IX o o o o otO 00 «=t LO \0

exj cr> CTi o cxi i—I rH CXI "st LO

Cx CTi LO CTi 00 ^t LO LO CXI C vO tx OO CTi CTi

CTi LO CTi tO r—

CTi CTi CXI \0

CTi CTi CTi CTi 00

cq

i—I 00 LO LO 00

tO (XI tO tO rH o o o o o

o o o

O i—I "st O tx

i—l rH i—l i—I i—l O O O O O

O

vO LO CXI CTi CXI i—I i—I i—I O i—

o o o o o o o

i—l LO \o t~* oo

i—I o o o o o o o o

o o o

CO^

tO i—l CXI O "vt- vO "st 0O Cx vD

vO OO i—I VO to \0 \o X X oo

tO VO CTi CXI i—

st CTi O tx Cx O vO to oo to

CTi CTi O O i—l

vO "st CXI CTi tO

CTi tO IX CTi vO

IX rH tO <=t LO

i—I cxi cxi cxi cxi

>st rH O LO IX

CXI 00 "St VO IX O LO "st CXI O (XJ (XJ CXI (XI CX]

CD

i—l o o o o o o o o o o o o o o o o o o o o

LO O LO O LO

IX 00 OO CTi CTi to to to to to O LO O LO o O i—l i—l CXI

'St "St "St "St 'vt

LO O LO O LO

CX) tO tO "st "st

"St *t "St "St 't O LO O LO O LO LO vD vO tx

"St "St "St "St "St 88
EE

wCO

Ph

EE

i—I \D CTi CNJ to *D Mn IN CTi CNI Tj- rv n oo oo OO OO OO 01 CTi

CN1 CNI CN) CNI CNI

(NJ CN) (NJ CNJ tO o o o o o o o o o o

N (M M N N

CNI i—I tO tO CNI o o o o o o o

CNI (NJ CNJ CNJ (VI

i—I tO CNI CNI o o o o o o

<T> CTi CTi CTi CTi

CNI CNI CN] CNJ CNI

i—I i—I t—I CNI CN] o o o o o o o

T3

CDg

•Hp

§

OO CNI CTi CTi LO

tO 00 i—I LO CTi 00 [N IN VO LO H \£) vO Csl OO Kl OO

lo -3- to to

CNI ,—) i—I LO

tO CNI CNJ CNI t—

tO CNI \0 i—

t-O tO t—H O

OO

COO

H

EE

mCO

Ph

LO OO O r—I i—l L/l LO O O »0

i—I CNI LO \Q

vO vO *0 N CO CO OO COvO vO

CO IN \0 t^l \0 \Q \0 \D \D

CNJ CNI CNI CNI CN! CNI CNI CNI CNI CNI CNI CNJ CNJ CNJ CNJ CNJ CNJ CNI CNJ (NJ

H

EE

CNI CNJ CN) (Nl tO o o o o o o o o oCNI i—i tO tO CNJ o o o o o o

t—I tO tO CNJ CNJ o o o o o

l—I i—I r—I CNJ CNI o o o o o o o

E-i O i—l CTi OO

<vf OO CNJ LO CTi CO N N vO LO

tO 00 vO 00 ^ CO to CO ^J" LO H" <rfr to to

^j- \o to to IN O \£> to O IN

tO CNJ CNJ CNJ t—

tO IN (NJ CTl LO to i—I O 00

1—I I—I I—I T-{ O

cqCO

Pi

rH (NJ i-H CNI CNJ o o o o o o o

CNJ rH (NJ (NJ CNJ o o o o o o

I—I tO CNJ CNJ CNI o o o o o o o o o

CNJ r-1 I—I i—I CNJ o o o o o o

rH in i—l t— O CTi r—I ^1" vO CTl IN in LO "3-

(NJ (NJ 00 O 00

tO IN I—I IN CNJ tO tO CNJ CNJ

(NJ O to

CTi O to

CNI

CTi o

\0 CNJ (NJ tO LO

In vD lO"3- tO o o o o o

CDgH

pjou

CO

co^

Cn 00 vO tO CNJ o o o o o

o o o

OO CO \Q (NJ H"

I—I CO I—I CNI 0O LO H OO

r-H i—I rH O O

O tO CNJ LO IN o o o o o o o o

o o o o

00 OO vO IN LO o o o o o o o o o o o

CTi vO vO o o o o o o o o

IN OO CTi i—I i—

CO IN \0 LO tO

CTi LO i—I [n K)

CTi CTi CTi OO OO O CNJ CNI LO Cn

-vj- vO O LO i—

CTi LO (NJ OO LO

IN jn fN vO

LO LO I—I I—I LO

00 IN IN IN O rH 00 LO CNJ VO LO LO LO LO

CD rHsEH

o o o o o o o o o o o o o o o o o o o o

LO O LO O LO IN 00 OO CTi CTi H- H- H- O LO O LO o O r—I I—I (NJ LO LO LO LO LO

LO O LO O LO

CNJ to to H" H"

LO LO LO LO LO O LO O LO o LO LO vO tNLO LO LO LO LO 89
w

CO

CXI tO LO vO

CT) CT> cr> cn Ol N 1^- N OO OO

CT) CT> Ol CTi CT> 00 00 OO CT) t—

01 Ol Ol Cl o

tOOOHHNN t~~- i—I \£5 (XI CT)

cxi cxi cxi cxi co cxi cm cm cxi cxi (N (M M M M Kl K) Kl Kl Kl K)

K5 i—I tO «vt- o o o o o o o

vO N i£) K) H O O O O i—

O O O O O

to un <* n m

O i—I i—1 to r-l o o o o o

t-- lo to \o to LO

Csl tO l>- LO VO o o o o o o

v ' m

\OultNOtO N vO i/l i/) o o o o o

(N OO H

to to cxi cxi cxi o o o o o

OO VO -* H O I—I I—I I—I I—I I—

o o o o o

CT) OO Is- \Q LO ^3" o o o o o o o o o o o o

E-hEL

wCO

a,

l/l l/HO O >0 vO vO *0 VO LO Lf) <vf \0 v£> O \0

to to to to

vO O O vO

vO Ol (XI [>. to CT) \0 O N N OO OO

(xi (X) eg (x) (xi CXI (XI (XI (XI CXI (xi (xi eg (xi (xi (XI (XI (XI (XI CM (XI

» I —

to to i—I to "3- o o o o o o o o ovO CXI \0 to i—

O O O O i—

o o o o o O i—l i—I to i—o o o o o

MO tO \D tO Lfl

(xi to t>- to \o o o o o o o

N X 00 O ^ N LO IO t o o o o o

CO (XI 00 LO i—

tO tO CNI (XI (XI o o o o o

CT) \£5 «=d- (XI O i—I i—I i—I rH i—

o o o o o

O O0 N \D IO o o o o o o o o o o o o

(XI (XI rH IO tO o o o o o o o o o

LO (XI LO tO CTl o o o o o o

tO tO N H vO O i—l i—I to i—

o o o o o

vo lo \o lo o niovO'*'*^ o o o o o o

CT) *zf O vQ tO

(XI CXI (XI I—I I—

o o o o o

i—I CT> l>- vO LO

i—I o o o o o o o

^d- to to cxi cxi o o o o o

o

(XI i—| i—I r-1 t—I i—l o o o o o o o o o o o o

s

+->

I=! o

o

co^

CM

0O \D tx 0O o o o o o o

o o

N N OO \0 o o o o o

o o o oo

ix vO vO Is- to o o o o o o o o o

LO LO LO O LO o o o o o o o o o o o o o o o o o o

VO i—I CT) CM i—

CD LO tO tO

*3" "5f tO

\C> LO (XI LO LO

to ^ r-- o vO fXj ,—| CT> tO to tO tO (XI

CT) (XI i—( LO OO^ON"* N \o lo to rsi

(Xi cxi (xi (xi (xi

cxi lo i—i cxi r-~ to

i—I O 00 O LO

CXI (XI i—l i—l i—I i—l

rH8

o o o o o

LO O LO O LO

r~- 00 00 CT) CT) LO LO LO LO LO

o o o o o o o o o o o o o o o o

O LO O LO o O i—I i—I CXI vO vO vO *0

LO O LO O LO

(XI tO tO "3" vO \o *0 o

O LO O LO O LO

\0

90
o

CO

PQ

CD XI

+J mo6

CD

+->

(/)

toCOu

CD -M

CD

<U

5-i O><

CO r<

CD >N-P

•M CD •H g > o

•H -H P rt O U

P,

O

t/l

u

CD -P

5h

U

CDP<

03 C/) UMu u

CD P

Ph CD

(/) Sh

CD

CD <-H

AV( 8V

W)

cn

CN)

CD

5hP

W)

•H

91
-p •H>

•H

10o

D-i 10

<D

t—

<DPw

CO

o

to

i—

>

£

•H

5hO

f-i ?H • CD S

CD

+-> fH to

0$ CO

OS

(-> o

00 •Hu

> CD •H P

-t-> CD

i—

(

<L> CD

+->

o

H0HH3 aHVQNVXS 3AIXY13H

u

p

bo •H

92
X s D. Compensated Emission Spectrum (Y-^y

5 ^)

The compensated, normalized, uncorrected emission

spectra for Runs I, II, and III (i.e., (Y) (Y)j-j-, and

(Y)jjj) were averaged and the RSE's in these values were

calculated and are summarized in Table 17. As can be seen,

the RSE in (Y) varied from 0 percent at the peak maximum

values to as high as 6 percent at one of the wings. In

general, however, the RSE was less than 0.2 percent over the

major portion of the uncorrected emission spectrum.

T

E . Technical Emission Spectrum E (X) of Quinine Sulfate

The normalized (Y) values for Runs I, II, and III were

corrected for bandpass errors (Section VII-B) and monochromator wavelength errors (Section VIII-C) and the technical

T emission spectrum E (X) for each run was calculated by:

T

(Y) x

* Mx

= c (23)

<^x

where the subscript x represents the values from Runs I, II,

or III and the other symbols have the same representation as

stated previously.

T T T The values for E (X)j, E ( A

) j ^

, and E (^)jjj were

averaged to give E

T

(X). The E

T

(X) values, the RSE E

T

(X),

T and the percent systematic error limits PSEL E (X) (Section

IX-A-4) are summarized in Table 17.

F . Molecular Emission Spectrum E(X)

T The E (X) values for each run were then corrected for

the sample parameter effects of solvent refractive index

(Section VII-E) and cell window transmittance (Section

VII-F). The resulting values were averaged and the E(X)

values, the RSE in the E(X) values, and the PSEL E(X) values

(Section IX-A5) are given in Table 17.

93
In general, the E (A) values differ from the E(_A) values

by no more than 10 percent with most of the values differing

by much less than that.

The values of E (A) , E(v), and E (v] were calculated

according to [39,114]:

A

3 E(A) = A

2 E

p

(A) = AE(v) = E

p

(v) . (24)

T The normalized, calculated values for E (A), E(A), E (A) , E(v), and E (v) are summarized in Table 18. The values of

(Y) and E(A) are plotted versus wavelength in figure 31 so

that the extent of the corrections are evident. The relative

standard errors, a measure of imprecision in the E(A) values

are plotted in figure 32 and vary from ^1 percent at the

spectrum wings to less than 0.1 percent at the center of the

spectrum with most of the values being less than ^0.2 percent.

The signal level expressed as the percent of the maximum

signal and the RSE for E(A) values are summarized in Table

19. The observed RSE's are of the same order of magnitude

as the RSE's determined by the usual propagation of error

treatment used throughout the data handling and smoothing.

The emission spectra of quinine sulfate in E(A) and

Ep(v) units as functions of wavelength are also presented

in figure 33 to demonstrate the differences in spectral

distribution depending on the reporting units [39,78b].

Interpolated E

p

(v) values at even wavenumber intervals are

given in Table 20.

94
r

Table 18. The Technical and Molecular Emission Spectra of

Quinine Sulfate in the Various Radiometric and

Photon Units of E

T (A), E(A), E f A) , E(v), and E (v)

A , nm E

T

(A) a E(A) a

E

D

(A) a v,

P

cm 1 (10" 4

) E(v) a E

D

(v)

P

O 7/ ^ n 0 . U n oo ; oU. UUJ o o ; n no a 7 ftft 7 0 0 0'^ 0 0 0 X JOU ^ ro • ou .. U ± L m 7 01 0 7 ftU J^ i»7 . 00Q u u y . 0u 0u 7/ OxrkO O • nU • nU 7L ro < 0U 7L RO 07 A 7 ^Q 7 • 0U L7 0U . U01J. 7/ 390.0 .057 .057 . 049 2. 564 . 042 . 036

395. 0 . 103 . 103 . 090 2. 532 .078 .067

a4 oU oU . oU 1 ft Q • JLi 7/ Un .-LOU 1 KO 7 JUU 1 x i . _L1 1-L AH

. 7^6 i DO • Z7^7O 1 • 7L L7 QV 7 Aft Q 7 0"^

. z u o • I/O 1 7 R a4 i o n X U . u X K 7 X K Q . OX Zr7 A4 £ • AtoyX Q . z7 Qy 0u 7 ; Q . z o y 415. 0 .469 .471 . 430 2 . 410 .390 . 352

420.0 . 583 . 586 . 542 2. 381 .497 .453

a 7 ; n AO? . fto Qy A4 . ftu o^ 0u Z . X K X ft 0 X .33/

4 jU . u

7 q n . / y u7 Q 7 7^0 Z . JiO7 ft 7 n ; . / U 3ft Q ft 4 O O . U

8 7 7 • O / Z R 7 A . o / 4 R ^ 7 7 z7 Q Q y y . / y o . 7^7 / 3 L 440.0 . 938 .940 . 911 2. 273 . 875 . 836

445. 0 . 983 .984 .965 2. 247 .93 7 . 906

a c: n n 4 O U . UQ Q ft

. y y o . QQQ y y y qq n . y y u

9Z . ZinZ Z Q 7 7 . y / z Q k 1

. y o i a q £ n Q Q ft

, y y o . QQ7 y y / . QQQ y y y Z . j.1 Qy Ro . Qy Qy oX Q R 7 . y o z

4 ft n n 4 o u . u Q ft X . y o oQ ft 7 . y o z QQC . y y o Z . 11/47A . QQQ y y y . QQQ y y y

465.0 . 949 . 947 .970 2. 151 .985 .995

470.0 . 898 . 897 . 929 2. 128 .953 .973

A 7 t; n 4 / o . u RAO ft X ft • O JO .oil R 7 7 7Z . in; Q 1 0 . y x uQ X Q . y o y

aro 4 o u . nu . 7/ R0 A4 .IOC. 7 R 7 ROiU7 ft 7 0 R X .out) Rft ft

. Qy 0u oX ARK o 7 1 Q . / J. y . 7/ j.1 Qy ./DO 7 ft R 7 Oft u u 7z R 1 ^ R c; 7 AQO 4 j u nu . fto j^ Qy ft .03/c; 7 . 70Q / u y z7 . OAU f 1J. . 7/ jKQy . ROo u Ro 495.0 .598 . 595 . 649 2. 020 . 702 . 755

500.0 . 543 . 541 . 596 7Z . 000 .651 . 707

505 .0 .488 . 486 . 540 1. 980 . 596 . 653

510.0 .436 .434 . 487 i. 961 . 542 .601

515.0 . 388 . 386 . 438 i. 942 .492 .551

520 .0 . 344 . 342 . 392 i. 923 .445 . 503

aPeak maxima: ET

(A) = 452.4 nm; E ( A) = 452.2 nm; E (X) = 456.0 nm; E(v) = 2 . 178 cm" 1 (xl0~ ") ; E (v) = 2.165 cm" 1 (xlO "

*

)

continued

95
Table 18 continued.

\A TVTT1

2 11111 F^ f A ") F f > 1 F (\ ~\ V } Vw 111 1 fl 0 1 F f\H

525. 0 0. 304 0. 302 0. 349 1. 905 0.400 0.456

530.0 . 266 . 264 . 308 1. 887 .356 .410

5 3 5 n • 2Li 3 .J3 • 2Li O31J- m Li2 71 Li2 X1 • 869 O VJ 17 31 8 370

«J540*+ vj • nVJ • Li \J O 2 01 • 2L, o3 QV

1X • 8O «JS 2Lj t 282 Li U Li 3 31

•J54*T «JS B 0VJ • 1X 77/ / 1 7 R o 211 L, ± ±

1X e 83O .J u5 • 2L J511 • Lt297Z> I

550. 0 .154 .153 . 185 1. 818 .223 . 266

555 .0 .133 . 132 . 162 1. 802 . 196 . 236

560 0 • 117 -L X / • 116 X X VJ • X143T sj 1X • 786 . 174 .212

*-J S6S\J -J m 0\J • 102 X VJ £j • 101 X VJ X 126 1X • 770 .155 • 191 X -/ X

5 70 0 a 0VJ 89\J Zf • 088 u u u • 110 J—L VJ X • 7 54 • 137 X «J / 170

575.0 . 077 . 076 . 096 1. 739 .121 . 151

580.0 . 067 . 065 . 085 1. 724 . 107 .135

58 s o • 0VJ *JS8O 0 S 7 • 0VJ 74/ *T 1X • 709 / VJ 17 • 094 VJ .7 *+ • 119 X X -7 S90 n . 0u oSOyj • nu os ou • 06VJ VJ «JS 1X • 69VJ .7 «_> 5 « 084 VJ Q *-t • 10X VJ 7/

«JS9J %Js m 0VJ « D44 VJ T" *T • nVJ 4H «J3 • 0VJ OS6VJ iX 681 VJ O X • 0VJ 7/ ,J3 • 09VJ +J5

600. 0 .038 . 037 . 049 1. 667 . 064 .084

605 .0 . 032 . 032 . 043 1. 653 . 057 .074

610 *J X VJ • 0VJ • 028 VJ Li U • 028 VJ Lt O 0 38 1X • 639 U J J • 0VJ *JSOVJ • 066 VJ VJ VJ 61VJ X «J5 • 0\J 02 5 • 024 \J T" • 033 VJ «J »J X1

» V626Li U • 044 VJ T" *T .059

620 0 021 W Li X o 021 vj t-> J. • V/029L* «J X1

• 61\J X «J3 039 .052

625.0 . 019 . 018 . 025 1. 600 . 035 . 047

630.0 . 016 . 016 . 022 1. 587 .029 . 041

6 3 S f) 014 01 4 • 01VJ X 9J

1X • S 7 S • 0VJ 2L 7I 0 3 7

vj *+ u • vj 01 2 01\J X 1X 016 1X • J56U J3 • U02Li J3 • 031 VJ J X 64 5 n mo • 01VJ X 0VJ • 01VJ X J5 1X • J^50J U • 0VJ 21Lj X • 0VJ 2£i 9

U «_> vj • U DDQ • 009 VJ VJ »/ 01 3 1X • J^38J U • 01VJ X 8o > 02U Li J5 6vj +J5 .J5 • 0VJ • 008 \J VJ KJ 008 \J \J K> • 011 \J X X X1 • 527 . 016 .02 2 660.0 . 007 . 007 . 010 1. 515 . 014 .020

665. 0 . 006 . 006 . 009 1. 504 . 013 . 018

670. 0 . 005 . 005 . 007 1. 493 . 011 . 015

675.0 . 004 . 004 . 007 1. 481 . 010 . 014

96
ooo + * *

£4'

CNJ

O 00 \ +* * + + $ + * +

HC + & +

CO

(y)a jo (a)

CMO oo

LD o U

P! • o

cu +j o

aj • >-. 4_| LT) v—' 1—I (NI

II

5h •MU

CDao

CD

H

PiH - C 6 H C o

t/i

m to

(A •h > II S

CD

CDU

CDU

toCD

5hO

t—

<4H

r<^ X H CD r< 63 •«

rH * M OU rH

CD

U

P! \O rH

T3

CD P

n5

t/> P!

CDPU CD 6 +-* ou

CD

U P! CD CD

rH > U rH o o U CO

rH

to

CDU• H

97
H0HM3 dHVCINVIS 3AIXY13H

ft

•H

98
Table 19. The Average Relative Standard Errors (RSE)

in E(A) at Selected Signal Levels.

A , nm

Signal Level as

^Percent of Maximum RSEa

375 .0 0 . 5 0.0194

380 . 0 1 .

0

. 0097

395 . 0 10. 0 . 0022

420 . 0 50.0 . 0030

455.0 100.0 .0008

505. 0 50.0 .0019

565. 0 10.0 .0018

645. 0 1.0 .0266

670.0 0.5 .0549

Average of RSE values for A, A + 5 nm, and A -5 nm except

for 375 nm data which is the average of A and A+5 nm

data

Table 20. Interpolated E (v) Values at Even Wavenumber

Intervals. ^

Wavenumber

cm" 1 (10 _lt

) E (v)

P

70

75

80

1.85

90

95

00

05

10

0.111

. 164

. 233

. 327

.444

.572

. 707

. 830

.931

Wavenumber

cm'^lO" 1*)

15

20

25

30

35

40

45

50

E (v)

P

0.995

.980

.899

. 748

. 568

.385

.228

.114

99
1 .2

400 600 650 700

WAVELENGTH, nm

Figure 33. The corrected emission spectra of quinine

34 7.5 nm,

sulfate in E(X) and E (v) units.

Solvent =0.1 mol/L HC1CU;

T = 25.0 °C.

ex

100
IX. Composite Errors and Precisions

A. Errors

Three basic sources of error have to be considered when

T estimates of the total errors in the E (A) and E(A) values

are determined. These are: (1) the errors in the values

determined for the spectral responsivity of the detection

system; (2) the error in the wavelength position of the peak

maximum; and (3) systematic errors such as those presented

in Sections VII-A-F for which corrections can be made.

1 . Errors in the Spectral Responsivity of the

Detection System

The relative errors in the spectral responsivity for

the detection system are summarized in table 20 and consist

of (1) the quadrature addition of individual errors in the

various components in the calibration procedure of the

standard lamp (e.g., uncertainties in black body quality,

temperature determination of the black body, current measurement, etc.) (2) spectral radiance changes as a function of

operating time and current; and (3) laboratory lamp transfer

(e.g., lamp alignment, quartz envelope reflections, convection air currents, soldered socket contacts, etc.).

In the calculation of the spectral responsivity of the

detection system, the spectral radiance values reported for

the standard lamp were normalized at 500 nm, equation (22).

Thus, the error associated with the spectral radiance value

at 500 nm was propagated through all the spectral radiance

values. This additional error component in each normalized

value was added in quadrature to the error for the individual

L value to give the normalized errors NE:

101
NE = #)'(£)'

The NE values for the spectral responsivity are given in

Table 21 and depicted graphically in figure 34. [Note: In

T order to calculate the relative errors in E (X) and E(X),

Table 17, the estimated percent error in the spectral

radiance values were determined at 5 nm wavelength intervals

by use of an Omnitab interpolation routine [56].]

Table 21. Percent Error Limits (PEL) at the 95 Percent

Confidence Level for the Spectral Responsivity

of the Detection System.

Wavelength PEL

nm

350 2.28

400 2.70 a

450 1.80

550 1.30

654.6 1.10

750.0 1.08

850.0 1.56 a

Normalized PEL

2. 70

3. 06

2. 32

1. 96

1. 84

1. 82

2. 14

a Value includes the averages of the errors for the 8.803

and 6 . 000 A data.

2 . Error in the Emission Peak Maximum

Due to the broad character of the quinine sulfate emisT sion peak, the computed peak maxima for E (X) and E(X) have

an uncertainty of ±0.2 nm in wavelength position. This

uncertainty in wavelength gives rise to possible errors in

the E (X) and E(X) values which were calculated at selected

wavelengths by determining AE(X)/AX at intervals bf 0.02 nm.

102
These estimated errors are summarized in Table 22 and plotted

in figure 34. They vary from a high of 2.44 percent at

375 nm to a low of ^0.01 percent near the peak maximum.

350 400 450 500 550 600 650 700

WAVELENGTH, nm

Figure 34. Percent estimated errors in the reported values

at the 95 percent confidence level due to the

calibration of the standard lamp, the 0.2 nm

uncertainty in the emission peak position, and

the errors estimated at 30 percent of the

individual corrections for systematic error.

103
0

•H

If) 4-* •H

•HHo

5h5iU

•HP

03>—

<D P

If)

COpo

CDU

5-i <D Pi

-dC

ctj

?Ho

Sh5h wpc

CDco

OhO

CJ • in

CD CD P rH

o3 03

j—

S !>

•H

•P ' * If) V ' w O Tj0

o3

Sh

o3 /—\ 6 ^< 0 H

CO

m

CXI CX]

CD iHH

•HmH

CO

Ph

to t-> OO (XI to to tO VO t-» vo

OO

H

NN OO K) Oi

O) Ol O) OlO

N

CO

W Kl

NN

NNNNN K) (O

O

too

00 r-. o tO rH CXI vO CT> CX] "=3- CO

1—1 o o o i—

(XI (XI CXI CXI to to to o o o o o O o o o o o o

LOo

to

LO vO iH as 00 (XI o to SO o

00 00 o vO rH vO rH LO 00 tH vO

I—

o o o rH rH CXI CXI CXJ to to to o

«=3-

N Ol

M

rH OJ tO LOrHI^OOrfrvOtOvDCTi LOvOvDvOvOvOvOvOCO

00 LO to CXI CXI CX] CXI CXI CXI CXI (X) >0o

OO vO vO o ^t- 00 rH Oi o to

p

LO CXI o rH rH

OOO CXI CXI

u to

rH o o o o o o O o o o O o CD 1 CO

VII-A

i—

O

tern i—

>£

VIIcxi vO o o o CXI LO 00 rH cti to m O

r-~ to o vo o to vO

O 00 to LO 0 •H

LO o o o rH rH rH CXI rH rH cxi to in

OH Ct o

•H

CXI o t

ion

Sect ,

Se -Pu • CD m co 00 OO O CXI vO 00 00 00 CO OO 00 u • 5h • i u o o o o o O o o o o o O o CD CXI O ^IH H o o o o o o o o o o o o o •P i >x Sh vO IH CD S CD < p $-1 cxi > U (XI o •CT i i—

M-H rH ar i CD ^

PQ rO 0

on

tan 0 S O IH CD i -P O •H +-> o > 0

•H

ih bO-H

rH

O P

Ct •H6

•H

4-> (XI «=3- LO CT> rH rH o -=3- o *3- CTi to CTi •P c rH > cd o3 CD m 03 to O (XI "vf LO LO LO LO LO LO vO •H o 0 rH 0 CO 0 0 o > -H o o cd cr o3 cr

+1 cxi rHO •H 4-> o O > CD Sh CD

n 10

U •H 03X

p S O CD S-H -p £ g CD S

Ph

O CO Ph

V)

g

lie o o

CD f-H f-i CO O M-l

Ind dow fro

cxi vO "=3" CXI o OO CX] vO (X] o vO ^±

O CD 0 Oh P 0

03 o rH to (XI

O o CTi CTI CTi 00 oo 00 ^ 6 •H •n n3 t3 CD •H

•H

p

X 6 <D > £ CD CO to to cxi CX] cxi CXI cxi rH rH rH rH rH rH rH X rH POO • H 0

o3 03 P CD •H

5h S 6 •h ,o e P 6 P o S U Sh 03 P Sh U ^ PI) o o Sh CD CD s CD 0} o O 0 +J <P. > P c LO o LO o LO o LO O LO O LO O LO Oh CD o3 O CD CD o CD o cxi LO o CXI LO t-- O CXI LO CO Ph CnfflSQ oS u Q *\ to "St H- «=* LO LO LO LO vO vO vO vO

03 43 CJ

rCT CD MH

H

104
3 . Errors in the Applied Correction Factors

The corrections calculated from theoretical considerations for the systematic errors attributable to experimental

and instrumental parameters (Sections VII-A-F) involve

assumptions that introduce errors into the corrections and

T thus into the E (A) and E(A) values. These errors are

estimated to be ^30 percent of the correction and are listed

in Table 22 at 25 nm intervals. The error for all the

corrections were obtained by addition of the absolute values

of the individual errors and are plotted in figure 35. (See

Section IX-A-5 below).

10

350 400 450) 500 550 £ WAVELENGTH, nm

600 650 700

Figure 35. The percent systematic error limits

in the values for the corrected

emission spectrum of quinine sulfate

in 0.105 mol/L HC1C\ at the 95

percent confidence level.

105
4 . Percent Systematic Error Limits in the E^A)

Values, PSEL E

T

(A)

T T The uncertainties in the E (A) values, PSEL E (A), were

calculated by adding the absolute values of the uncertainties

in the spectral respons ivity , the peak maximum, and the

applied corrections, for the instrumental factors only,

equation (26)

T ASL APM APMT . ABW AA

E ro ,. PSEL E (A) = ~~SL ~~PM TMT + "BW

+ ~XZ (26 ^ E

T The PSEL E (A) values are summarized in column 7 of Table 22

at 25 nm intervals and in Table 17 at 5 nm intervals.

5 . Percent Systematic Error Limits in the E(A) Values , PSEL E (

) The percent systematic error limits in the E(A) values

were determined by the addition of the errors used in deterT mining PSEL E (A) plus the additional error estimates in the

applied correction factors for the sample parameters of

refractive index and cuvet window transmittance

PSEL[E(A)J = PSEL E

T

(A) + ^ + ^ t 27 )

These values are summarized in the last column of Table

22 at 25 nm intervals and in Table 17 at 5 nm intervals and

also are plotted in figure 35b.

B . Precision

The relative standard errors calculated for the E(A)

values are experimental precision values at the 67 percent

confidence level and are given in Table 17 and plotted in

figure 24. They are compatible with the usual propagation

of error and precision treatments (see discussion in Section

VIII-C and D)

.

106
References

[1] Stokes, G. G., On the Change of Refrangibility of

Light. Phil. Trans., Roy. Soc, London A142 , 463

(1852); ibid A143 , 385 (1853).

[2] Harvey, E. Newton, "A History of Luminescence" The

Amer. Phil. Soc. , Independence Square, Philadelphia,

PA, 391 (1957) .

[3] Bowman, R. L., Caulfield, P. A. and Udenfriend, S.,

Science 122 , 32 (1955) .

[4] Passwater, R. "Guide to Fluorescence Literature", Vols.

I, II, III, 1967, 1970, 1974. Plenum Press, New York,

N.Y.; Udenfriend, S., "Fluorescence Assay in Biology

and Medicine," Vols. I, II, 1961, 1969, Academic Press,

New York, N.Y.

[5] Drobnik, J. and Yeargers, E . , J. Mol. Spectry. 1_9, 454

(1966) . [6] Rusakowicz, R. and Testa, A.C., J. Phys. Chem. 72_, 793

(1968) .

[7] Melhuish, W. H., J. Phys. Chem. 65, 229 (1961).

[8] Scott, T. G., Spencer, R. D., Leonard, N. J. and

Weber, G., J. Amer. Chem. Soc, 92^, 687 (1970).

[9] Lippert, E., Nagele, W. , Seibold-Blankenstein , I.,

Staiger, W. , and Voss, W., Z. Anal Chem. 170_, 1 (1959).

[10] Melhuish, W. H., J. Phys. Chem. 64, 762 (1960).

[11] Parker, C. A. and Rees, W. T., Analyst 8S_, 587 (1960).

[12] Chapman, J. H., Forster, T., Kortiim, G., Lippert, E.,

Melhuish, W. H., Nebbia, G., and Parker, C. A., Z. Anal

Chem. 197_, 431 (1963) . [13] Mavrodineanu , R. and Baldwin, J., NBS Special Publication 260-51 SD Catalog No. C 13.10:260-51, U.S.

Government Printing Office, Washington, DC 20402.

107
[14] Cali, J. p;, Mears, T. W., Michaelis, R. E. , Reed, W. P.

Seward, R. W. , Stanley, C. L., Yolken, H . T. , and

Ku, H. H., "The Role of Standard Reference Materials

in Measurement Systems", NBS Monograph 148, 1975.

Available from the U.S. Government Printing Office,

Washington, D.C. 20402, SD Catalog No. C 13.44:148.

Cali, J. P., Anal Chem. 4_8 (11), 802A (1976).

Report for Analytical Chemists, Anal. Chem. 3_6 (12),

23A (1964); ibid, Anal. Chem. 3J8 (8), 27A (1966).

Meinke, W. W. , Materials Research and Standards, 15

October 1969.

Meinke, W. W. , Anal Chem. £3(6), 28A (1971).

Birks, J. B., J. Lumin. 9, 311 (1974).

Vavilo, S. I., Z. Phys. 22_, 266 (1924); idem 42, 311

(1927) . Melhuish, W. H . , J. Opt. Soc. Amer. 54, 183 (1964).

Cehelnik, E. D., Cundall, R. B., Lockwood, J. R. and

Palmer, T. F. , Chem. Phys. Letters 27, 586 (1974).

Weber, G. and Teale, F. W. J., Trans Faraday Soc. 53

,

646 (1957).

Eisenbrand, J., Z. Anal. Chem. 179 , 170 (1961).

Fletcher, A. N., Photochem. Photobiol. 9, 439 (1969).

Demas, J. N. and Crosby, G. A., J. Phys. Chem. 75 , 991 (1971) . Birks, J. B., J. Res. Nat. Bur. Stand. (U.S.), 80A,

389, May-June (1976) . Reisfeld, R., Honigbaum, A., and Velapoldi, R. A.,

J. Opt. Soc. Amer. 61, 1422 (1971).

Reisfeld, R., J. Res. Nat. Bur. Stand. (U.S.), 76A (6),

613, Nov-Dec (1972) . Arthur, E. P., Ceramic Bull. 47_, 183 (1968).

Brody, M. and Brody, S. S . , Photochem. Photobiol. 13

,

293 (1971).

108
[32] Hastings, J. W. and Weber, G., Photochem. Photobiol. 4,

1049 (1965).

[33] Velapoldi, R. A., J. Res. Nat. Bur. Stand. (U.S.) 76A

No. 6., 641 (1972).

[34] Velapoldi, R. A., Reisfeld, R. , and Boehm, L. , Phys

Chem. of Glass 14(6) , 101 (1973).

[35] Reisfeld, R. , Velapoldi, R. A., and Boehm, L., J.

Phys. Chem. 76, 1293 (1972).

[36] Reisfeld, R. , Greenberg, E. , Velapoldi, R. A. and

Barnett, B., J. Chem. Phys. 56 (4) , 1698 (1972).

[37] Velapoldi, R. A. and Mielenz, K. D., Unpublished Results.

[38] Drexhage, K. , J. Res. Nat. Bur. Stand. (U.S.) 8 OA , No. 3, 421 (1976).

[39] Melhuish, W. H . , J. Res. Nat. Bur. Stand. (U.S.) 76A , No. 6, 547 (1972).

[40] Mielenz, K. D. , Anal. Chem. 48 (7) , 1093 (1976).

[41] Winefordner , J., IUPAC Proposal on Nomenclature,

Private Communication.

[42] In order to describe materials and experimental procedures adequately, it was occasionally necessary to

identify commercial products by manufacturer's name or

label. In no instances does such identification imply

endorsement by the National Bureau of Standards nor does

it imply that the particular product or equipment is

necessarily the best available for that purpose.

[43] Kuehner, E. C, Alvarez, R. , Paulsen, P. J. and

Murphy, T. J., Anal Chem. 44, 2050 (1972).

[44] The authors acknowledge with thanks, J. Moody, Analytical Spectrometry Section, NBS for the preparation of

the water used for solution preparation in some of

these measurements

[45] Parker, C. A., Analyst, 8£, 446 (1959).

109
[46] (a) The semi -micro balance was accurate to ±0.00001 g

as determined by using NBS #6273D calibrated weights.

(b) The top loader balance was accurate to ±0.01 g as determined by using NBS #3617 calibrated weights.

[47] Mavrodineau, R. , J. Res. Nat. Bur. Stand. (U.S.) 76A

(5) , 405 (1972)

.

[48] Mavrodineanu, R. , "Glass Filters as a Standard Reference

Material for Spectrophotometry" NBS Special Publication

260-51, 1975, Washington, DC 20234.

[49] Mielenz, K. D., NBS 260 Special Publication, to be

published.

[50] Turner, G. K. , Science 146_, 183 (1964).

[51] Mavrodineanu, R. and Lazar, J. W. , "Standard Quartz

Cuvettes for High Accuracy Spectrophometry , NBS

Special Publication 260-32, Washington, DC 20234.

[52] Mielenz, K. D. and Velapoldi, R. A., NBS 260 Special

Publication, to be published.

[53] Mielenz, K. D., Appl. Opt. 13, 2580 (1974).

[54] Yguerabide, J., Rev. Sci. Instrum. 3_9 , 1048 (1968).

[55] The authors gratefully acknowledge R. Dehl , Polymer

Stabilities and Reactivities Section, NBS, for performing this calibration.

[56] Hilsenrath, J., Ziegler, G. G. , Messina, C. G.,

Walsh, P. J. and Herbold, R. J., "OMNITAB", NBS Handbook 101, 1966; Peavy, S. T., Varner, R. N. and

Hogben, D., "Source Listing of OMNITAB II Program",

NBS Special Publication 339, 1970; Hogben, D., Peavy,

S. T., and Varner, R. N., "OMNITAB II User's Reference

Manual", NBS Tech. Note 552, 1971.

[57] Kaufman, V., J. Opt. Soc . Amer. 5_2 , 866 (1962); Martin,

W. C, J. Res. Nat. Bur. Stds . (U.S.) 64A, 19 (1960);

Zaidel', A. N. , Prok of ev V. K. , Raiskii, S. M. , Slavnyi, V. A., Shreider, E. Ya., "Tables of Spectral

Lines" 3rd Edition, Plenum Press, New York, N.Y. (1970).

110
[58] The spectral radiances for the vacuum ribbon filament

lamp were determined by comparison to a high-quality

variable- temperature black body of approximately equal

spectral radiance. NBS Test No. 221 . 12/4B/75 . D. McSparron and J. Walker, Radiometric Physics Division,

National Bureau of Standards, Washington, DC 20234.

[59] A Tungsten lamp similar to the calibrated lamp was used.

[60] Burke, R. W. , NBS Private Communication. Rinsing a cuvette five times in situ provides for less than 0.01

percent error in absorbance values under normal

circumstances

[61] In this paper, weight absorptivity is replaced by

specific weight absorbance. The specific weight

absorbance, WA in g SQ , Cg t 3

2 cm 1 is calculated by:

A

WA = cb

where A is the absorbance

b = cell pathlength in cm

c = sample concentration in S sol (g solv

)

1

•

[62] Under typical conditions the emission spectrum of

quinine is measured at the 1 ppm (1 x 10 6 mol/L)

level. Assuming the impurity is homogeneously distri

buted in the original quinine sulfate material, is

present at the 0.1 percent level, and has a molecular

weight similar to quinine sulfate 0783 g/mol), then

the impurity concentration is:

C = (10" 9 g/mL) (10 3 g/L) (1 mL/g)(783 g/mol)" 1

C = 1.3 x 10~ 9 mol/L

[63] Parker, C. A., "Photoluminescence of Solutions'*

(Elsevier Publishing Company, New York, N.Y (1968).

164] Braude, E . A. , J. Chem. Soc. 379 (1950).

[65] Gill, J. E., J. Chrom. 26, 309 (1967).

Ill
[66] We would like to thank Dr. J. Flood, Hartford Hospital,

Hartford, Connecticut for bringing this to our attention; Crouthamel, W. G., Kowarski, B. , and Narang, P. K.

Clin. Chem. 2_3, 2030 Q-977) ; Hartel, G. and Korhoven, A.

J. Chromatogr. 37, 70 (1968).

[6 7] We would like to thank W. May and J. Brown, Organic

Analytical Research Division, NBS , for running the

HPLC analyses.

[68] Chen, R. F . , J. Res. Nat. Bur. Stand. (U.S.) 76A, 593

(1972)

.

[69] We would like to thank E. White, V, Organic Analytical

Research Division, NBS, for measuring and interpreting

the mass spectra.

[70] Fales, H. M. , Lloyd, H. A., and Milne, G. W. A., J.

Amer. Chem. Soc. 9_2, 1590 (1970).

[71] Gill, J. E . , Photochem and Photobiol. 9 , 313 (1969).

[72] Velapoldi, R. A., Unpublished Results.

[73] We thank S. Margolis, Bioorganic Standards Section,

National Bureau of Standards for making the Karl

Fischer measurements.

[74] Melhuish, W. H. , New Zealand J. Sci, Tech. 37, 142

(1955)

.

[75] Chen, R. F . , J. Res. Nat. Bur. Stand. (U.S.), 76A , 593, (1972).

[76] Natrella, M. G., "Experimental Statistics", NBS Handbook 91 Tables A-6, pages T10-11, (1963).

[77] Burke, R. W. and Mavrodineanu , R. , Private Communication

The estimated error in the measurement independent of

the magnitude of the absorbance value was ±0.0002

aA absorbance units and in our case ^— expressed in

percent is 0.1.

[78] (a) Birks, J. B., J. Res. Nat. Bur. Stand. (U.S.I 8 OA , No. 6, 389 (1976}, Cb) Heller, C. A., Henry, R. A.,

McLaughlin, B. A., and Bless, D. E., J. Chem. Eng.

Data, 19, 214 (1974)

.

112
[79] West, M. A. and Kemp, D. R. , International Laboratory,

May/June 27 (1976); White, J. U. , Pittsburgh Conference

Abstracts, paper 488 (1977); Perre, T., Personal

Communication

[80] Dawson, W. R. and Windsor, M. W. , J. Phys. Chem. , 72 , 3251 (1968).

[81] Pringsheim, P., "Fluorescence and Phosphorescence",

Interscience Publishers, New York, N.Y. 1949, p. 332.

[82] Birks, J. B., "Photophysics of Aromatic Molecules",

Wiley-Interscience, New York, N.Y. 1970, p. 499.

[83] Schlenk, W. and Thai, A., Ber. Deut. Chem. Ges . , 46_,

2840 (1913).

[84] Cehelnik, E. D., Mielenz, K. D., and Velapoldi , R. A.,

J. Res. Nat. Bur. Stand. (U.S.), 79A, No. 1, 1 (1975).

[85] Chen, R. F. , Anal Biochem. 1_9 , 374 (1967).

[86] Pesce, A. J., Rosen, C-G. and Pasby, T. L. , "Fluorescence Spectroscopy", Marcel Dekker, Inc., New York,

N.Y. , 1971, p. 96.

[87] Luminescence refers to both fluorescence and

phosphorescence

[88] Fletcher, A. N., J. Phys. Chem. 72_, 2742 (1968).

[89] Weber, G. and Shinitzky, M. , Proc. National Academy of

Science, U.S. 6_5 , 823 (1970).

[90] Itoh, K. and Azuni, T. , Chem. Phys. Letters 22 , , 395

(1973) . [91] Borreson, H. C, Acta Chem. Scand. L9 , 2089 (1965).

[92] Eastman, J. W. , Photochem. Photobiol. 6, 55 (1967).

[93] Velapoldi, R. A., Kenigsberg, D. R. , Menis, 0., and

Cehelnik, E. D., submitted for publication, Photochem.

and Photobiol.

[94] Reference 51, p. 222-225.

[95] Parker, C. A. and Barnes, W. J., Analyst 8_2 , 606 (1957).

[96] Gill, J, E., Applied Spectry. 24_, 588 (1970).

[97] Mode, V. A. and Sisson, D. H. , Anal. Chem. 46 , 200

(1974) . 113
[98] Forster, Th. , "Fluorescence Organischer Verbindungen"

Vandenhoeck and Ruprecht, Gottingen, 1951, p 35-42.

[99] Jablonski, A., Compt. Rend. Soc. Polon. Phys . 7_, 1

(1926)

.

[100] Grzywacz, J., J. Lumin. 4, 244 (1971).

[101] Clarke, F. J. J., J. Res. Nat. Bur. Stand. (U.S.), 7 6A

(5) 375 (1972) . [102] Mielenz, K. D. and Eckerle, K. L. , App. Oct. 11(10) , 2294 (1972).

[103] Hardy, A. C. and Young, F. M. , J. Opt. Soc. Airier. 39,

265 (1949).

[104] Rutledge, Phys. Rev. 40, 262 (1932).

[105] Lipsett, F. R. , Progr. in Dielectrics 7_, 217 (1967).

[106] Kohner, H. and Gressmann, M. L., Z. Physikal. Chem.

144(A) , 137 (1929); Faj ans , K. and Gressman, M. L.

,

Z. Physikal. Chem. 146(A) , 309 (1930); Hantsch, A.

and Durigen, F., Z. Physik. Chem. 144(A) , 147 (1929).

[107] Hellwege, A. M. , Landol t -Bornstein , II Band, 8 Teil,

5-565 (1962).

[108] Malitson, I. H. , J. Opt. Soc. Amer. 55 (10) , 1205

(1965) . [109] Rosenblatt, J., National Bureau of Standards, private

communication

.

[110] The authors gratefully acknowlege the efforts of

G. Ritter, National Bureau of Standards to fit this

dat a

.

[Ill] The normalization wavelengths were: quinine sulfate

data M50 nm, fitting interval 435 to 470 nm; 6.0000

amp standard lamp data - ^500 nm, fitting interval

475 to 525 nm; 8.8030 amp standard lamp data - 420 nm,

fitting interval 400 to 425 nm.

114
[112] The data were "fitted" using polynomials of the 3rd

to 6th order. The 3rd order polynomial was adequate

to smooth the data at A such that the standard

max

deviation of the precision for the predicted value of

(Vhich would be used to normalize the spectrum) was

better than the standard deviation in the data.

[113] In general, precisions of ^0.2 percent were obtained

for a single point and since four spectra were

combined, the precision should be 0.2//4" or ^0.1

percent

[114] Ejder, E. J., J. Opt. Soc. Amer . 59, 223 (1969).

115
U.S. Department of Commerce APPENDIX A Juanita M> Kreps

SgcretSfy?

National Bureaw of Standards Emest Ambler, Director Rational ^mmn af Standards

(tterttfttate

Standard Reference Material 936

Quinine Sulfate Dihydrate

R. A. Velapoldi and K. D. Mielenz

This Standard Reference Material is intended for use in the evaluation of methods and the calibration of

fluorescence spectrometers. It is certified for the relative molecular emission spectrum, E(X), in radiometric

units for a solution of 1.28 x 10 mol/L quinine sulfate dihydrate in 0.105 mol/L perchloric acid using an

excitation wavelength of 347.5 nm. The certified values of the molecular emission spectrum at 5 nm wavelength intervals from 375 to 675 nm are given in table 1. These values have been corrected for instrument and

sample parameters, including the spectral responsivity of the detection system, monochromator bandwidth,

photomultiplier tube nonlinearity, monochromator wavelength error, solvent refractive index, and cell window

transmittance. The relative standard error in E(X), RSE [E(X)j, is given in table 1. The estimate of the

relative systematic error limits in the molecular emission spectrum, RSEL [E(X)], is also given in table 1 and was determined by the addition of the absolute values of the estimated systematic errors. These relative error

limits include uncertainties in the calibration values for the spectral responsivity, the wavelength position of the emission peak maximum, and in the corrections applied for instrument and sample parameters.

From the certified values of E(X), values may be calculated for the molecular emission spectrum in the various

photon, radiometric, wavelength, and wavenumber units using the following equation: [1,2]

EP(X) E(V) Ep(sr)

E(X) = =—- = —

X x

2 X3

These values have been calculated and are given in NBS Special Publication 260-64. T The technical emission spectrum, E (X), i.e., the emission spectrum corrected for instrument parameters only,

is also given in SP 260-64. The quinine sulfate dihydrate used for SRM 936 was a special lot of material

obtained from the J. T. Baker Chemical Co., Phillipsburg, N.J. The technical and support aspects concerning the preparation, certification, and issuance of this Standard

Reference Material were coordinated through the Office of Standard Reference Materials by T. W. Mears and

R. W. Seward.

Washington, D.C. 20234 George A. Uriano, Chief

April 1, 1979 Office of Standard Reference Materials

(over)

116
Table I. The Molecular Emission Spectrum, E(A), of Quinine Sulfate Pihydrate in 0.105 mol/L HOQ4, the

Relative Standard Error, RSE, and the Estimated Relative Systematic Error Limits, RSEL, in the

E(A) Values.

X,nm E(\) RSE [E(A)] RSEL [E(A}J

17S 0 0 005 0 019 0 087

ISO 0 012 006 .078 385 0 .028 .003 .071 390.0 .057 .003 .064

395.0 .103 .002 .059 400.0 .170 .002 .054 405.0 .257 .003 .049

410.0 .359 .003 .045 415.0 .471 .003 .041 420.0 .586 .003 .037

4?S 0 694 014

410 O . / 7£. 00? Oil

415 0 874 00? 0?8

440.0 .940 .001 .026

445.0 .984 .001 .024

450.0 .999 .001 .023 455.0 .997 .001 * 023

460.0 .982 .001 .024

465.0 .947 .001 .024 470.0 .897 .001 .025

475 0 838 002 0?6

480.0 .782 002 027

485.0 .719 002 027

AS7 on? .uz / 495.0 .595 .003 .027

500.0 .541 .002 .027

505.0 .486 .001 .028 510.0 .434 .003 .028

515.0 .386 .003 .028 520.0 .342 .002 .028

SRM 936

Page 2

\,nm E(X) RSE [E(/\)j RSEL [E(A)j

525.0 0.302 0.001 0.029

530.0 .264 .003 .029

535.0 .231 .003 .029 540.0 ,201 .002 .029

545.0 .175 .002 .029 550.0 .153 .001 .029

555.0 .132 .001 .029 560.0 .116 .001 029

565.0 .101 .002 .029

570.0 .088 .002 .029

575.0 .076 .003 .029 580.0 065 .003 .029 585.0 .057 .001 .029

590.0 .050 .003 .030

595.0 .043 .004 .030 600.0 .037 .006 .030

605.0 .032 .002 .030

610.0 .028 .006 .030

615.0 .024 .003 .030

620.0 .021 .011 ,-OfJ

625.0 .018 .003 .030

630.0 .016 .015 .030

635.0 .014 .014 .030

640.0 .01.037 .030

645.0 .010 ,015 .030 650.0 .009 .027 .030

655.0 .008 .035 .03! 660.0 .007 .073 .031 665.0 .006 .046 .032

670.0 A:

r

> .053 .032 675.0 .004 .« >~> .033

117
SUPPLEMENTARY DATA

The following data for the specific molar absorbances, water content, photon yields, and fluorescence lifetimes are considered to be supplementary and are not to be considered certified values. The quinine sulfate dihydrate (QSD) used for SRM 936 was found to be homogeneous to better than 0.5% by

thin-layer chromatography with development by two solvent systems and the determination of specific molar

absorbances, €, at three different wavelengths. The SRM contains approximately 1.7% of an impurity as determined by high performance liquid chromatography using absorbance and fluorescence detection. This

impurity is believed to be dihydroquinine sulfate dihydrate, which has optical characteristics that are similar to those of the quinine sulfate dihydrate. The ultraviolet absorption spectrum of SRM 936 in 0. 105 mol/L HC IO4

exhibits the following absorption maxima:

250.0 nm, €max = 56,990 ± 90 Lmof'cm"1

and, on the side of a peak:

347.5 nm, €max — 10,810 ± 20 Lmof'cm" 1

365.0 nm, Cobs = 6,920 ± 10 L mol ' cm

1

The water content of this material was measured by two methods. The average of six determinations by the

Karl-Fischer method gave a value of (4.74 ± 0.05%), while the average of four determinations by a weight loss procedure gave a value of (4.57 ± 0.04%). The theoretical value for water in quinine sulfate dihydrate is 4.60%.

The photon yield, Q, and the fluorescence lifetime, r, of SRM 936 were compared to values obtained for a sample of purified quinine sulfate dihydrate and are summarized below:

Q r, ns

0.5 mol/L H2SO4 0.5 mol/L H2SO4

SRM 936, QSD 0.544 ± 0.03 19. 1 ± 0.

Purified QSD 0.546* 19.2 ±0.1

a Melhuish, W. H., J. Phys. Chem. 65, 229 (1961); ibid, New Zealand J. Sci. Tech. 37, 142

(1955).

SRM 936

Page 3

(over)

118 i
PREPARATION AND USE OF SRM 936

This Standard Reference Material is for "in vitro" diagnostic use as a clinical laboratory standard. A "stock"

standard solution containing 0. 1 mg/mL of quinine sulfate may be prepared as follows: Weigh O.lOOgofSRM

936 to the nearest one-tenth milligram and quantitatively transfer it to a 1 000-mL volumetric flask. Dilute to the

I calibrated volume with 0.1 05 mol/L HCIO4, to give a solution that is 1.28x10 mol/L (0.1 mg/mL) in quinine

sulfate. Store this solution in the dark in a well-stoppered, glass bottle. A "working" standard solution

containing 1 ng/mL may be prepared by transferring 10 mL of the above "stock" standard solution to a

1 000-mL volumetric flask and diluting to the calibrated volume with 0. 1 05 mol / L HC 1 04 to give a solution that

is 1.28 x 10 mol/L (1 /ig/mL) in quinine sulfate. Store this solution in the same manner as the above "stock"

standard solution.

! Several opinions regarding the stability of quinine sulfate solutions have appeared in the literature [3]. NBS

j considers the 0.1 mg/mL "stock" standard solution prepared from SRM 936 to be stable for 3 months when

! stored as specified; and the 1 jug/mL "working" standard solution to be stable for 1 month when so stored.

! SRM 936 should be kept in its original bottle and stored in the dark at room temperature (30 °C or less). It should not be subjected to heat or direct sunlight during storage. Experience at NBS indicates that under proper

storage this material is stable for at least 3 years. If this material degrades beyond the limits certified, purchasers

will be notified by NBS. It is recommended that the material not be used after 3 years from the date of purchase.

References:

[1] Ejder, E. J., J. Opt. Soc. Amer. 59, 223 (1969).

1 [2] Melhuish, W. H., J. Res. Nat. Bur. Stand. (U.S.) 76A, No. 6, 547 (1972).

[3] Melhuish, W. H., J. Phys. Chem. 65, 229(1961); Gill, J. E., Photochem. and Photobiol. 9, 313(1969); Birks,

J. B., J. Res. Nat. Bur. Stand. (U.S.) 80A, 389 (1976); Heller, C. A., Henry, R. A., McLaughlin, B. A., and

Bless, D. E., J. Chem. Eng. Data 19, 214 (1974); West, M. A., and Kemp, D. R., Int'l. Lab., p. 27 (May/ June

1976); and White, J. U., Pittsburgh Conf. Abstracts, Paper 488 (1977).

This Standard Reference Material has been measured and certified at the laboratories of the National Bureau

of Standards, Gaithersburg, Maryland. All inquiries should be addressed to:

Office of Standard Reference Materials

Room B311, Chemistry Building

National Bureau of Standards

Washington, D.C. 20234

The date of issuance and certification of SRM 936 was April 1, 1979

SRM 936

Page 4

119
I

Appendix B

Supplemental Analytical Data

for SRM Grade Quinine Sulfate Dihydrate

Product No. 4891

Lot Number: 517677

Date of Analysis: 6-23-75

Formula: ( C 20 H'24 N

2 O

2^ 2

* H

2 SCV 2H 2°

FW 7 8 2.95

Assay, Anhydrous Basis ( (^qI^^^C^) 2

6 H^SCO (by non-aqueous acidbase titration)

r Elemental Analysis (anhydrous basis)

Carbon (C) (by combustion)

Hydrogen (H) (by combustion)

Nitrogen (N) (by Kjeldahl)

Sulfur (S) (as sulfate by barium sulfate

gravimetry)

Absence of Other Cinchona Alkaloids

Fluorescence Excitation and Emission

Curves in 0.1 N Sulfuric Acid

Infrared Spectrum

Particulate Matter (after solution

in hydrochloric acid)

Residue after Ignition

Specific Rotation^

Thin-Layer Chromatography6

Ultraviolet-Visible Absorption

Spectrum in 0.1 N Sulfuric Acid

Water (H

2 0)

£

Weight Absorptivity at 25°C in 0.1 N

Sulfuric Acid (kg per g-cm)

at 250.0 nm

at 347.5 nm

120

ACTUAL ANALYSIS

100. 2% a

64. 50%

6. 741

7. 36%

12.83%

passes test

authentic

authentic

0.001%

0. 001%

-242°

no extraneous

fluorescent spot

authentic; smooth

curve

4.5%

76. 3

14.2
ACTUAL ANALYSIS

NON-METALLIC IMPURITIES

in parts per million (ppm)

Boron (B) g 0.6

Halide (as CI) 12

Silicon (Si) g 0.3

METALLIC IMPURITIESg

in parts per million (ppm)

Aluminum (Al) 0.4

Bismuth (Bi) < 0.1

Cadmium (Cd) < 2 Calcium (Ca) < 1

Chromium (Cr) 0.1

Cobalt (Co) < 0.1

Copper (Cu) 0.03

Iron (Fe) 0.6

Lead (Pb) 0.2

Magnesium (Mg) 0.00 7 Manganese (Mn) 0.03

Molybdenum (Mo) < 0.02

Nickel (Ni) 0.2

Silver (Ag) 0.04

Sodium (Na) 5 Tin (Sn) < 0.1

Titanium (Ti) 0.03

Vanadium (V) < 0.0 2 Zinc (Zn) < 2

121
aAnalytical data supplied by J. T. Baker Chemical Company,

Phillipsburg, New Jersey 08865.

Average value for 2 determinations: 100.2% and 100.2%.

rAverage value for duplicate determinations: carbon,

64.58% and 64.421; hydrogen, 6.721 and 6.75%; nitrogen,

7.32% and 7.39%; and sulfur (as sulfate), 12.83% and

12 . 83%

.

d. 2 0

[a] -p, anhydrous basis, c=2 in 0.1 N hydrochloric acid.

e R

No extraneous fluorescent spot on Baker-flex Silica Gel

IB2 TLC flexible sheets on separate development with

1 -propanol- 88% formic acid (80:20 v/v) and chloroformmethanol -acetic acid (75:20:5 v/v) and ultraviolet

illumination

.

£By Karl Fischer titration; average value for 2 determinations: 4.48% and 4.59%.

^Average value for duplicate samples analyzed by dc-arc

spectrography (after 50-fold concentration by heating

with sulfuric acid followed by ashing with addition of

lithium carbonate; against commercial standards in lithium

carbonate, reading of lines in 2450-3875 A region); key

elements found absent are reported as < (less than) the

detection limit.

122

«U.S. GOVERNMENT PRINTING OFFICE: 1979 311-046/350 1-3
NBS-114A (REV. 9-78)

U.S. DEPT. OF COMM. 1. PUBLICATION OR REPORT NO. 2. Gov't Accession BIBLIOGRAPHIC DATA SP 260-64

SHEET

4. title and subtitle Standard Reference Materials:

A Fluorescence Standard Reference Material

Quinine Sulfate Dihydrate

5. Publication Date January 1980

6. Performing Organization Code

7. AUTHOR(S) R. A. Velapoldi and K. D. Mielenz

8. Performing Organ. Report No.

9. PERFORMING ORGANIZATION NAME AND ADDRESS

NATIONAL BUREAU OF STANDARDS

DEPARTMENT OF COMMERCE

WASHINGTON, DC 20234

10. Project/Task/Work Unit No.

11. Contract/Grant No.

12. SPONSORING ORGANIZATION NAME AND COMPLETE ADDRESS (Street, City, state, ZIP)

Same as item 9.

13. Type of Report & Period Covered Final

14. Sponsoring Agency Code"

15. SUPPLEMENTARY NOTES

Library of Congress Catalog Card Number:

I

I Document describes a computer program; SF-185, FIPS Software Summary, is attached.

79-600119

16. ABSTRACT (A 200-word or less [actual summary of most significant information. If document includes a signi ficant bibliography or literate survey, mention n here.) The need, material selection, characterization, certification,

and uses of the fluorescence Standard Reference Material, quinine sulfate dihydrate,

have been discussed. The emission spectrum for quinine sulfate dihydrate in 0.1 mol/L

HCIO4 has been measured on the NBS reference spectroradiometer. The spectrum has been

reported every 5 nm from 375.0 to 675.0 nm as the relative technical emission spectrum

E*(A) and the relative molecular emission spectra represented by E(X) , Ep(A), E(v), and

Ep(v). The technical emission spectrum has been corrected for the instrumental parameters of the spectral responsitivity of the detection system, photomultiplier tube nonlinearity, bandpass, monochromator wavelength error, and further corrections for the

sample parameters of solvent refractive index and cell window transmittance were applied

to obtain the molecular emission spectra. Estimates of the 95 percent systematic error

limits of the emission spectrum values have been made as a function of wavelength.

These values are ^2 percent at the peak maximum (M55 nm) and ^5 percent and ^3 percent

at the blue (395 nm) and red (565 nm) one-tenth power points. The experimental replication precision is ^0.2 percent over the total spectral range. The purity and the

stability of the SRM quinine sulfate dihydrate and the effect of solute concentration,

solvent acid, acid concentration, excitation wavelength, oxygen quenching, temperature

and polarized exciting radiation on the emission spectrum, and in part, on the absorbance spectrum, photon yield, and fluorescence lifetime also have been discussed.

17. KEY WORDS ("six to twelve entries; alphabetical order; capitalize only the first letter of the first key word unless a proper name;

separated by semicolons) Calibration of spectrofluorimeters ; corrected fluorescence spectra;

fluorescence; fluorescence lifetimes; fluorescence standards; molecular emission

spectrum; photon yields; quinine sulfate dihydrate; spectrofluorimetry ; Standard

Reference Material; technical emission spectrum; transfer standards.

18. AVAILABILITY [X) Unlimited 19. SECURITY CLASS

(THIS REPORT)

21. NO. OF

PRINTED PAGES

1 I For Official Distribution. Do Not Release to NTIS UNCLASSIFIED 139

[X] Order From Sup. of Doc, U.S. Government Printing Office, 20402, SO Stock No. SN003-003-

Washington, DC

20. SECURITY CLASS

(THIS PAGE)

22. Price

1 I Order From National Technical Information Service (NTIS), VA. 22161

Springfield, UNCLASSIFIED USCOMM-DC
I

i

I
NBS TECHNICAL PUBLICATIONS

PERIODICALS

JOURNAL OF RESEARCH—The Journal of Research of the National Bureau of Standards reports NBS research and develop- ment in those disciplines of the physical and engineering sciences in which the Bureau is active. These include physics, chemistry,

engineering, mathematics, and computer sciences. Papers cover a broad range of subjects, with major emphasis on measurement

methodology and the basic technology underlying standardization. Also included from time to time are survey articles on topics

closely related to the Bureau's technical and scientific programs. As a special service to subscribers each issue contains complete

citations to all recent Bureau publications in both NBS and non- NBS media. Issued six times a year. Annual subscription: domestic

$17; foreign $21.25. Single copy, $3 domestic; $3.75 foreign. NOTE: The Journal was formerly published in two sections: Section A "Physics and Chemistry" and Section B "Mathematical

Sciences." DIMENSIONS/NBS—This monthly magazine is published to in- form scientists, engineers, business and industry leaders, teachers, students, and consumers of the latest advances in science and

technology, with primary emphasis on work at NBS. The magazine

highlights and reviews such issues as energy research, fire protection, building technology, metric conversion, pollution abatement,

health and safety, and consumer product performance. In addition, it reports the results of Bureau programs in measurement

standards and techniques, properties of matter and materials, engineering standards and services, instrumentation, and

automatic data processing. Annual subscription: domestic $11;

foreign $13.75. NONPERIODICALS

Monographs—Major contributions to the technical literature on

various subjects related to the Bureau's scientific and technical activities. Handbooks—Recommended codes of engineering and industrial practice (including safety codes) developed in cooperation with in- terested industries, professional organizations, and regulatory

bodies. Special Publications—Include proceedings of conferences sponsored by NBS, NBS annual reports, and other special publications

appropriate to this grouping such as wall charts, pocket cards, and

bibliographies. Applied Mathematics Series—Mathematical tables, manuals, and

studies of special interest to physicists, engineers, chemists,

biologists, mathematicians, computer programmers, and others engaged in scientific and technical work.

National Standard Reference Data Series—Provides quantitative data on the physical and chemical properties of materials, compiled from the world's literature and critically evaluated. Developed under a worldwide program coordinated by NBS under

the authority of the National Standard Data Act (Public Law

90-396).

NOTE: The principal publication outlet for the foregoing data is the Journal of Physical and Chemical Reference Data (JPCRD)

published quarterly for NBS by the American Chemical Society

(ACS) and the American Institute of Physics (AIP). Subscriptions,

reprints, and supplements available from ACS, 1 155 Sixteenth St., NW, Washington, DC 20056.

Building Science Series—Disseminates technical information developed at the Bureau on building materials, components,

systems, and whole structures. The series presents research results,

test methods, and performance criteria related to the structural and

environmental functions and the durability and safety characteristics of building elements and systems.

Technical Notes—Studies or reports which are complete in themselves but restrictive in their treatment of a subject. Analogous to monographs but not so comprehensive in scope or definitive in treatment of the subject area. Often serve as a vehicle for final reports of work performed at NBS under the sponsorship of other government agencies.

Voluntary Product Standards—Developed under procedures

published by the Department of Commerce in Part 10, Title 15, of

the Code of Federal Regulations. The standards establish nationally recognized requirements for products, and provide all concerned interests with a basis for common understanding of the

characteristics of the products. NBS administers this program as a supplement to the activities of the private sector standardizing

organizations. Consumer Information Series—Practical information, based on NBS research and experience, covering areas of interest to the consumer.' Easily understandable language and illustrations provide

useful background knowledge for shopping in today's tech- nological marketplace. Order the above NBS publications from: Superintendent of Documents, Government Printing Office, Washington, DC 20402. Order the following NBS publications—FlPS and NBSJR's—from

the National Technical Information Services, Springfield, VA 22161

Federal Information Processing Standards Publications (FIPS

PUB)—Publications in this series collectively constitute the Federal Information Processing Standards Register. The Register

serves as the official source of information in the Federal Govern- ment regarding standards issued by NBS pursuant to the Federal Property and Administrative Services Act of 1949 as amended.

Public Law 89-306 (79 Stat. 1127), and as implemented by Executive Order 11717 (38 FR 12315, dated May 11, 1973) and Part 6 of Title 15 CFR (Code of Federal Regulations).

NBS Interagency Reports (NBSIR)—A special series of interim or

final reports on work performed by NBS for outside sponsors

(both government and non-government). In general, initial dis- tribution is handled by the sponsor; public distribution is by the National Technical Information Services, Springfield, VA 22161,

in paper copy or microfiche form.

BIBLIOGRAPHIC SUBSCRIPTION SERVICES

The following current-awareness and literature-survey bibliographies

are issued periodically by the Bureau:

Cryogenic Data Center Current Awareness Service. A literature sur- vey issued biweekly. Annual subscription: domestic $25; foreign

$30. Liquefied Natural Gas. A literature survey issued quarterly. Annual

subscription: $20.

Superconducting Devices and Materials. A literature survey issued

quarterly. Annual subscription: $30. Please send subscription or- ders and remittances for the preceding bibliographic services to the National Bureau of Standards, Cryogenic Data Center (736) Boulder, CO 80303.
U.S. DEPARTMENT OF COMMERCE

National Bureau of Standards

Washington. D.C. 20234

OFFICIAL BUSINESS

Penalty for Private Use. $300

POSTAGE AND FEES PAID

U.S. DEPARTMENT OF COMMERCE COM—215

SPECIAL FOURTH-CLASS RATE BOOK