# EMSI ST-627-01

Low Flow Meter Post Calibration

AutoBox Number: AB-1 Date: 5/13/2010 Reference Std: Bios Definer 220

Pump Number: ABP-1 Technician: NP Ref Std Corr Factor: 1.0000

Master Factor: 0.98 Std Temp (F): 68

Pbar (inHg) 29.07 Std Pres (inHg): 29.92

Run Avg Initial Final Initial Final Orifice Orifice Initial Final Actual Calculated

Time Flow Volume Volume Outlet Outlet Number Factor Ambient Ambient Vacuum Gamma

min L/min L L F F Metric F F inHg Yd

10 1.0163 0 10.163 78.8 79.2 K4-14D 2.2000E-02 79 79.2 16 0.999471

10 1.0112 0 10.112 79.2 79.7 K4-14D 2.2000E-02 78.9 79.3 16 1.000305

10 1.0123 0 10.123 79.5 80.1 K4-14D 2.2000E-02 79.4 79.7 16 1.000119

Average 0.999965

coff*press*time/sqrt(Temp)

0.989 1.11 Autobox Data Reference Standard Data

PreTest Gamma % DeviationUSING CALIBRATED CRITICAL ORIFICES3-POINT ENGLISH UNITSCalibration ConditionsConsole Model NumberAB-2Date Time4/29/10 00:00Std Temp528oRConsole Serial Number Barometric Pressure28.9in HgStd Press29.92in HgDGM Model NumberTheoretical Critical Vacuum113.6in HgK117.647oR/in HgDGM Serial Number Calibration TechnicianJW/NK1For valid test results, the Actual Vacuum should be 1 to 2 in. Hg greater than the Theoretical Critical Vacuum shown above.2The Critical Orifice Coefficient, K', must be entered in English units, (ft3*oR1/2)/(in.Hg*min).Calibration DataRun Time Metering ConsoleCritical OrificeDGM Orifice Volume Volume Outlet Temp Outlet Temp Serial Coefficient Amb Temp Amb Temp ActualElapsed∆HInitial Final Initial Final Number see list on right Initial Final Vacuum(Θ)(Pm) (Vmi) (Vmf) (tmi) (tmf)K'(tamb) (tamb)minin H2Ocubic feet cubic feetoFoFsee above2oFoFin Hg8.0 2.0 0.000 6.462 91 88 63 0.5939 77 75 189.0 2.0 0.000 7.229 88 85 63 0.5939 75 75 188.0 2.0 0.000 6.369 85 83 63 0.5939 75 75 18Calibration Factor FlowrateValue Variation Std & Corr 0.75 SCFM Variation(Vm(std)) (Qm(std)) (Vcr(std)) (Qcr(std))(Y)(∆Y)(Qm(std)(corr))(∆H@) (∆∆H@)cubic feet cfm cubic feet cfm cfm in H2O6.027 0.753 5.931 0.741 0.984 -0.002 0.741 1.872 -0.0076.780 0.753 6.678 0.742 0.985 -0.001 0.742 1.879 -0.0016.001 0.750 5.936 0.742 0.989 0.003 0.742 1.888 0.008Pretest Gamma0.988% Deviation0.2 0.986Y Average1.880∆H@ AverageNote: For Calibration Factor Y, the ratio of the reading of the calibration meter to the dry gas meter, acceptable tolerance of individual values from the average is +-0.02.I certify that the above Dry Gas Meter was calibrated in accordance with USEPA Methods, CFR Title 40, Part 60, Appendix A-3,Method 5, 16.2.3Signature _______________________________________Date _____________________________ResultsAPEX INSTRUMENTS METHOD 5 POST-TEST CONSOLE CALIBRATIONFactors/ConversionsDry Gas Meter Critical OrificeStandardized DataDry Gas Meter∆H @Meter Console InformationUnit 1 ST604-04 Auto M23.xlsUSING CALIBRATED CRITICAL ORIFICES3-POINT ENGLISH UNITSCalibration ConditionsConsole Model NumberAB-3Date Time11-Mar-10 10:00Std Temp528oRConsole Serial Number Barometric Pressure28.9in HgStd Press29.92in HgDGM Master Factor1.39Theoretical Critical Vacuum113.6in HgK117.647oR/in HgVersion NumberST626-01Calibration TechnicianNic Presutti1For valid test results, the Actual Vacuum should be 1 to 2 in. Hg greater than the Theoretical Critical Vacuum shown above.2The Critical Orifice Coefficient, K', must be entered in English units, (ft3*oR1/2)/(in.Hg*min).Calibration DataRun Time Metering ConsoleCritical OrificeDGM Orifice Volume Volume Outlet Temp Outlet Temp Serial Coefficient Amb Temp Amb Temp ActualElapsedΔHInitial Final Initial Final Number Initial Final Vacuum(Θ)(Pm) (Vmi) (Vmf) (tmi) (tmf)K'(tamb) (tamb)minin H2Ocubic feet cubic feetoFoFsee above2oFoFin Hg10.0 1.2 0.000 5.785 55 56 WY55 0.455 51 51 1710.0 1.2 0.000 5.795 56 57 WY55 0.455 51 51 1710.0 1.2 0.000 5.798 57 58 WY55 0.455 51 52 17Calibration Factor FlowrateValue Variation Std & Corr 0.75 SCFM Variation(Vm(std)) (Qm(std)) (Vcr(std)) (Qcr(std))(Y)(ΔY)(Qm(std)(corr))(ΔH@) (ΔΔH@)cubic feet cfm cubic feet cfm cfm in H2O5.729 0.573 5.811 0.581 1.014 0.000 0.581 2.021 0.0085.730 0.573 5.810 0.581 1.014 0.000 0.581 2.001 -0.0115.725 0.572 5.808 0.581 1.015 0.000 0.581 2.016 0.004Pretest Gamma1.011% Deviation0.3 1.014Y Average2.013ΔH@ AverageNote: For Calibration Factor Y, the ratio of the reading of the calibration meter to the dry gas meter, acceptable tolerance of individual values from the average is +-0.02.I certify that the above Dry Gas Meter was calibrated in accordance with USEPA Methods, CFR Title 40, Part 60, Appendix A-3,Method 5, 16.2.3Signature _______________________________________Date _____________________________ResultsAPEX INSTRUMENTS METHOD 5 POST-TEST CONSOLE CALIBRATIONFactors/ConversionsDry Gas Meter Critical OrificeStandardized DataDry Gas MeterΔH @Meter Console Informationpost cal AB3 3.11.10USING CALIBRATED CRITICAL ORIFICES3-POINT ENGLISH UNITSCalibration ConditionsConsole Model NumberAB 4Date Time3-24-10 13:00Std Temp528oRConsole Serial Number232004Barometric Pressure29.1in HgStd Press29.92in HgDGM Model NumberTheoretical Critical Vacuum113.7in HgK117.647oR/in HgDGM Serial Number Calibration TechnicianJW1For valid test results, the Actual Vacuum should be 1 to 2 in. Hg greater than the Theoretical Critical Vacuum shown above.2The Critical Orifice Coefficient, K', must be entered in English units, (ft3*oR1/2)/(in.Hg*min).Calibration DataRun Time Metering ConsoleCritical OrificeDGM Orifice Volume Volume Outlet Temp Outlet Temp Serial Coefficient Amb Temp Amb Temp ActualElapsed∆HInitial Final Initial Final Number Initial Final Vacuum(Θ)(Pm) (Vmi) (Vmf) (tmi) (tmf)K'(tamb) (tamb)minin H2Ocubic feet cubic feetoFoFsee above2oFoFin Hg10.0 1.1 0.000 5.373 47 48 55 0.4319 38 38 1810.0 1.1 0.000 5.361 48 49 55 0.4319 38 37 1810.0 1.1 0.000 5.355 49 50 55 0.4319 37 37 18Calibration Factor FlowrateValue Variation Std & Corr 0.75 SCFM Variation(Vm(std)) (Qm(std)) (Vcr(std)) (Qcr(std))(Y)(∆Y)(Qm(std)(corr))(∆H@) (∆∆H@)cubic feet cfm cubic feet cfm cfm in H2O5.451 0.545 5.632 0.563 1.033 -0.004 0.563 1.896 0.0065.428 0.543 5.635 0.563 1.038 0.000 0.563 1.890 0.0005.412 0.541 5.638 0.564 1.042 0.004 0.564 1.884 -0.006Pretest Gamma1.006% Deviation3.1 1.038Y Average1.890∆H@ AverageNote: For Calibration Factor Y, the ratio of the reading of the calibration meter to the dry gas meter, acceptable tolerance of individual values from the average is +-0.02.I certify that the above Dry Gas Meter was calibrated in accordance with USEPA Methods, CFR Title 40, Part 60, Appendix A-3,Method 5, 16.2.3Signature _______________________________________Date _____________________________ResultsAPEX INSTRUMENTS METHOD 5 POST-TEST CONSOLE CALIBRATIONFactors/ConversionsDry Gas Meter Critical OrificeStandardized DataDry Gas Meter∆H @Meter Console InformationM5 Box Standard Cal Ver 1.0.xlsUSING CALIBRATED CRITICAL ORIFICES3-POINT ENGLISH UNITSCalibration ConditionsConsole Model NumberAB 6Date Time15-Apr-10 0:00Std Temp528oRConsole Serial Number Barometric Pressure28.9in HgStd Press29.92in HgMaster FactorTheoretical Critical Vacuum113.6in HgK117.647oR/in HgDGM Serial Number Calibration TechnicianNT1For valid test results, the Actual Vacuum should be 1 to 2 in. Hg greater than the Theoretical Critical Vacuum shown above.2The Critical Orifice Coefficient, K', must be entered in English units, (ft3*oR1/2)/(in.Hg*min).Calibration DataRun Time Metering ConsoleCritical OrificeDGM Orifice Volume Volume Outlet Temp Outlet Temp Serial Coefficient Amb Temp Amb Temp ActualElapsed∆HInitial Final Initial Final Number see list on right Initial Final Vacuum(Θ)(Pm) (Vmi) (Vmf) (tmi) (tmf)K'(tamb) (tamb)minin H2Ocubic feet cubic feetoFoFsee above2oFoFin Hg10.0 1.1 0.000 5.693 76 77 55 0.4319 80 78 1710.5 1.1 0.000 5.996 77 79 55 0.4319 78 77 1711.5 1.1 0.000 6.601 79 81 55 0.4319 77 77 17Calibration Factor FlowrateValue Variation Std & Corr 0.75 SCFM Variation(Vm(std)) (Qm(std)) (Vcr(std)) (Qcr(std))(Y)(∆Y)(Qm(std)(corr))(∆H@) (∆∆H@)cubic feet cfm cubic feet cfm cfm in H2O5.427 0.543 5.376 0.538 0.991 0.000 0.538 2.048 0.0115.700 0.543 5.653 0.538 0.992 0.001 0.538 2.037 -0.0016.252 0.544 6.194 0.539 0.991 0.000 0.539 2.027 -0.010Pretest Gamma0.980% Deviation1.1 0.991Y Average2.037∆H@ AverageNote: For Calibration Factor Y, the ratio of the reading of the calibration meter to the dry gas meter, acceptable tolerance of individual values from the average is +-0.02.I certify that the above Dry Gas Meter was calibrated in accordance with USEPA Methods, CFR Title 40, Part 60, Appendix A-3,Method 5, 16.2.3Signature _______________________________________Date _____________________________Dry Gas Meter Critical Orifice∆H @APEX INSTRUMENTS METHOD 5 POST-TEST CONSOLE CALIBRATIONMeter Console InformationFactors/ConversionsResultsStandardized DataDry Gas MeterST632-02 Auto SW0011 ERT .xlsUSING CALIBRATED CRITICAL ORIFICES3-POINT ENGLISH UNITSCalibration ConditionsConsole Model NumberAB 8Date Time3-26-10 08:10Std Temp528oRConsole Serial Number232008Barometric Pressure28.8in HgStd Press29.92in HgDGM Model NumberTheoretical Critical Vacuum113.6in HgK117.647oR/in HgDGM Serial Number Calibration TechnicianNT1For valid test results, the Actual Vacuum should be 1 to 2 in. Hg greater than the Theoretical Critical Vacuum shown above.2The Critical Orifice Coefficient, K', must be entered in English units, (ft3*oR1/2)/(in.Hg*min).Calibration DataRun Time Metering ConsoleCritical OrificeDGM Orifice Volume Volume Outlet Temp Outlet Temp Serial Coefficient Amb Temp Amb Temp ActualElapsed∆HInitial Final Initial Final Number Initial Final Vacuum(Θ)(Pm) (Vmi) (Vmf) (tmi) (tmf)K'(tamb) (tamb)minin H2Ocubic feet cubic feetoFoFsee above2oFoFin Hg14.0 0.5 0.000 5.795 40 41 48 0.3365 34 35 1713.0 0.5 0.000 5.385 41 43 48 0.3365 36 36 1713.5 0.5 0.000 5.618 43 44 48 0.3365 36 36 17Calibration Factor FlowrateValue Variation Std & Corr 0.75 SCFM Variation(Vm(std)) (Qm(std)) (Vcr(std)) (Qcr(std))(Y)(∆Y)(Qm(std)(corr))(∆H@) (∆∆H@)cubic feet cfm cubic feet cfm cfm in H2O5.891 0.421 6.101 0.436 1.036 0.000 0.436 1.388 -0.0095.458 0.420 5.657 0.435 1.036 0.001 0.435 1.418 0.0225.677 0.421 5.874 0.435 1.035 -0.001 0.435 1.384 -0.013Pretest Gamma1.024% Deviation1.1 1.036Y Average1.396∆H@ AverageNote: For Calibration Factor Y, the ratio of the reading of the calibration meter to the dry gas meter, acceptable tolerance of individual values from the average is +-0.02.I certify that the above Dry Gas Meter was calibrated in accordance with USEPA Methods, CFR Title 40, Part 60, Appendix A-3,Method 5, 16.2.3Signature _______________________________________Date _____________________________ResultsAPEX INSTRUMENTS METHOD 5 POST-TEST CONSOLE CALIBRATIONFactors/ConversionsDry Gas Meter Critical OrificeStandardized DataDry Gas Meter∆H @Meter Console InformationM5 Box Standard Cal Ver 1.0.xls

# EMSI ST-627-01

Low Flow Meter Post Calibration

AutoBox Number: AB-10 Date: 5/13/2010 Reference Std: Bios Definer 220

Pump Number: ABP-1 Technician: NP Ref Std Corr Factor: 1.0000

Master Factor: 1.00 Std Temp (F): 68

Pbar (inHg) 29.07 Std Pres (inHg): 29.92

Run Avg Initial Final Initial Final Orifice Orifice Initial Final Actual Calculated

Time Flow Volume Volume Outlet Outlet Number Factor Ambient Ambient Vacuum Gamma

min L/min L L F F Metric F F inHg Yd

10 0.9908 0 9.908 67.8 68.1 K4-14D 2.2000E-02 70.1 69.9 16 0.997505

10 1.0095 0 10.095 68.3 68.7 K4-14D 2.2000E-02 70.4 71 16 0.997227

10 1.0112 0 10.112 69.1 69.5 K4-14D 2.2000E-02 71.4 71.2 16 0.997608

Average 0.997447

coff*press*time/sqrt(Temp)

0.995 0.25 PreTest Gamma % Deviation Autobox Data Reference Standard Data