## EMSI Source Testing Department Form

o Isokinetic Nozzle Calibration Form ST300-02

Instructions: Make three inside diameter measurements with a NI ST traceable micrometer at dimensions A, B, and C. Record the results and click the calculate button. Record the Nozzle ID number, mark free of damage, record name and date of calibration.

Nozzle ID Number

224090 Dimension A: .279 inches

Glass Metal Dimension B: .279 inches Out of Service Dimension C: .280 inches Nozzle Edge Free of Damage

NIST Master: 421002 Location: MN Glass Nozzle Box

Find Save Next # Calculate Clear

Historical Data Path F:\EEQP\Historical Data\Nozzle.ini

Acceptance Criteria:
Average 0.279 inches Date:

Maximum Diameter 0.28 inches 12/16/2008

Minimum Diameter 0.279 inches Technician:

Difference 0.001 inches Nathan Traut

Acceptable ≤ 0.004 Passed

Comment: A B

C

## EMSI Source Testing Department Form

o Isokinetic Nozzle Calibration Form ST300-01

Instructions: Make three inside diameter measurements with a NI ST traceable micrometer at dimensions A, B, and C. Record the results and click the calculate button. Record the Nozzle ID number, mark free of damage, record name and date of calibration.

Nozzle ID Number

224005 Dimension A 0.281 inches

Save Next # Find Dimension B 0.281 inches

Glass Metal Dimension C 0.281 inches Out of Service

Nozzle Edge Free of Damage Calculate Clear

Acceptance Criteria:
Average 0.281 inches

Maximum Diameter 0.281 inches

Minimum Diameter 0.281 inches

Difference 0 inches
Acceptable ≤ 0.004 Passed

Technician Nathan Traut Date 1/21/2008

Historical Data Path F:\ASTM Compliance\Historical Data\Nozzle.ini A B

C

## EMSI Source Testing Department Form

o Isokinetic Nozzle Calibration Form ST300-02

Instructions: Make three inside diameter measurements with a NI ST traceable micrometer at dimensions A, B, and C. Record the results and click the calculate button. Record the Nozzle ID number, mark free of damage, record name and date of calibration.

Nozzle ID Number

224042 Dimension A 0.282 inches

Save Next # Find Dimension B 0.283 inches

Glass Metal Dimension C 0.283 inches Out of Service

Nozzle Edge Free of Damage Calculate Clear

Acceptance Criteria:
Average 0.283 inches

Maximum Diameter 0.283 inches

Minimum Diameter 0.282 inches

Difference 0.001 inches NIST Master
Acceptable ≤ 0.004 Passed 421002

Technician Nathan Traut Date 1/31/2008

Historical Data Path F:\ASTM Compliance\Historical Data\Nozzle.ini A B

C

## EMSI Source Testing Department Form

o Isokinetic Nozzle Calibration Form ST300-02

Instructions: Make three inside diameter measurements with a NI ST traceable micrometer at dimensions A, B, and C. Record the results and click the calculate button. Record the Nozzle ID number, mark free of damage, record name and date of calibration.

Nozzle ID Number

225073 Dimension A: .214 inches

Glass Metal Dimension B: .214 inches Out of Service Dimension C: .214 inches Nozzle Edge Free of Damage

NIST Master: 421002 Location: MN Metal Nozzle Box

Find Save Next # Calculate Clear

Historical Data Path F:\EEQP\Historical Data\Nozzle.ini

Acceptance Criteria:
Average 0.214 inches Date:

Maximum Diameter 0.214 inches 2010-01-28

Minimum Diameter 0.214 inches Technician:

Difference 0 inches Randy Resch

Acceptable ≤ 0.004 Passed

Comment: A B

C