EMSI Source Testing Department Form 

o Isokinetic Nozzle Calibration Form ST300-02 

Instructions: 

Make three inside diameter measurements with a NIST traceable micrometer at dimensions A, B, 

and C. Record the results and click the calculate button. Record the Nozzle ID number, mark 

free of damage, record name and date of calibration. 

Nozzle ID Number 

224090 Dimension A: .279 inches 

Glass Metal Dimension B: .279 inches 

Out of Service Dimension C: .280 inches 

Nozzle Edge Free of Damage 

NIST Master: 421002 Location: MN Glass Nozzle Box

 Find Save Next # Calculate Clear

Historical Data Path F:\EEQP\Historical Data\Nozzle.ini

Acceptance Criteria:

Average 0.279 inches Date: 

Maximum Diameter 0.28 inches 12/16/2008

Minimum Diameter 0.279 inches Technician: 

Difference 0.001 inches Nathan Traut 

Acceptable ≤ 0.004 Passed 

 

Comment: 

A B 

C 
EMSI Source Testing Department Form 

o Isokinetic Nozzle Calibration Form ST300-01 

Instructions: 

Make three inside diameter measurements with a NIST traceable micrometer at dimensions A, B, 

and C. Record the results and click the calculate button. Record the Nozzle ID number, mark 

free of damage, record name and date of calibration. 

Nozzle ID Number 

224005 Dimension A 0.281 inches 

 

Save Next # Find Dimension B 0.281 inches 

Glass Metal Dimension C 0.281 inches 

Out of Service

Nozzle Edge Free of Damage Calculate Clear

Acceptance Criteria:

Average 0.281 inches 

Maximum Diameter 0.281 inches 

Minimum Diameter 0.281 inches 

Difference 0 inches 

Acceptable ≤ 0.004 Passed

Technician Nathan Traut Date 1/21/2008

Historical Data Path F:\ASTM Compliance\Historical Data\Nozzle.ini

 

A B

C
EMSI Source Testing Department Form 

o Isokinetic Nozzle Calibration Form ST300-02 

Instructions: 

Make three inside diameter measurements with a NIST traceable micrometer at dimensions A, B, 

and C. Record the results and click the calculate button. Record the Nozzle ID number, mark 

free of damage, record name and date of calibration. 

Nozzle ID Number 

224042 Dimension A 0.282 inches 

 

Save Next # Find Dimension B 0.283 inches 

Glass Metal Dimension C 0.283 inches 

Out of Service

Nozzle Edge Free of Damage Calculate Clear

Acceptance Criteria:

Average 0.283 inches 

Maximum Diameter 0.283 inches 

Minimum Diameter 0.282 inches 

Difference 0.001 inches NIST Master 

Acceptable ≤ 0.004 Passed 421002

Technician Nathan Traut Date 1/31/2008

Historical Data Path F:\ASTM Compliance\Historical Data\Nozzle.ini

 

A B

C
EMSI Source Testing Department Form 

o Isokinetic Nozzle Calibration Form ST300-02 

Instructions: 

Make three inside diameter measurements with a NIST traceable micrometer at dimensions A, B, 

and C. Record the results and click the calculate button. Record the Nozzle ID number, mark 

free of damage, record name and date of calibration. 

Nozzle ID Number 

225073 Dimension A: .214 inches 

Glass Metal Dimension B: .214 inches 

Out of Service Dimension C: .214 inches 

Nozzle Edge Free of Damage 

NIST Master: 421002 Location: MN Metal Nozzle Box

 Find Save Next # Calculate Clear

Historical Data Path F:\EEQP\Historical Data\Nozzle.ini

Acceptance Criteria:

Average 0.214 inches Date: 

Maximum Diameter 0.214 inches 2010-01-28

Minimum Diameter 0.214 inches Technician: 

Difference 0 inches Randy Resch 

Acceptable ≤ 0.004 Passed 

 

Comment: 

A B 

C