NAT'L INST. OF STAND & TECH NJST

PUBLICATIONS

REFERENCE

AlllDS TbfiTlD o

J

^'^rcso*''

x? NIST TECHNICAL NOTE 1333

U.S. DEPARTMENT OF COMMERCE / National Institute of Standards and Technology
NISTTechnical Publications

Periodical

Journal of Research of the National Institute of Standards and Technology—Reports NIST research

and development in those disciplines of the physical and engineering sciences in which the Institute

is active. These include physics, chemistry, engineering, mathematics, and computer sciences. Papers cover a broad range of subjects, with major emphasis on measurement methodology and

the basic technology underlying standardization. Also included from time to time are survey articles on topics closely related to the Institute's technical and scientific programs. Issued six times a year.

Nonperiodicals

Monographs—Major contributions to the technical literature on various subjects related to the

Institute's scientific and technical activities. Handbooks—Recommended codes of engineering and industrial practice (including safety codes) de- veloped in cooperation with interested industries, professional organizations, and regulatory bodies.

Special Publications—Include proceedings of conferences sponsored by NIST, NIST annual reports,

and other special publications appropriate to this grouping such as wall charts, pocket cards, and

bibliographies.

Applied Mathematics Series—Mathematical tables, manuals, and studies of special interest to physicists, engineers, chemists, biologists, mathematicians, computer programmers, and others engaged in

* scient^c and technical work.

National Standard Reference Data Series—Provides quantitative data on the physical and chemical

properties of materials, compiled from the world's literature and critically evaluated. Developed un- der a worldwide program coordinated by NIST under the authority of the National Standard Data

Act (Public Law 90-396). NOTE: The Journal of Physical and Chemical Reference Data (JPCRD)

is published quarterly for NIST by the American Chemical Society (ACS) and the American Insti- tute of Physics (AIP). Subscriptions, reprints, and supplements are available from ACS, 1155 Six- teenth St., NW., Washington, DC 20056.

Building Science Series—Disseminates technical information developed at the Institute on building

materials, components, systems, and whole structures. The series presents research results, test methods, and performance criteria related to the structural and environmental functions and the

durability and safety characteristics of building elements and systems.

Technical Notes—Studies or reports which are complete in themselves but restrictive in their treat- ment of a subject. Analogous to monographs but not so comprehensive in scope or definitive in treatment of Uie subject area. Often serve as a vehicle for final reports of work performed at NIST

under the sponsorship of other government agencies.

Voluntary Product Standards—Developed under procedures published by the Department of Commerce in Part 10, Title 15, of the Code of Federal Regulations. The standards establish nationally

recognized requirements for products, and provide all concerned interests with a basis for common

understanding of the characteristics of the products. NIST administers this program as a supplement

to the activities of the private sector standardizing organizations.

Consumer Information Series—Practical information, based on NIST research and experience, covering areas of interest to the consumer. Easily understandable language and illustrations provide useful background knowledge for shopping in today's technological marketplace.

Order the aboTe NISTpublications from: Superintendent of Documents, Government Printing Office,

Washington, DC 20402.

Order the following NISTpublications—FIPS and NISTIRs—from the National Technical Information

Service, Springfield, VA 22161.

Federal Information Processing Standards Publications (FIPS PUB)—Publications in this series col- lectively constitute the Federal Information Processing Standards Register. The Register serves as the official source of information in the Federal Government regarding standards issued by NIST

pursuant to the Federal Property and Administrative Services Act of 1949 as amended, Public Law

89-306 (79 Stat 1127), and as implemented by Executive Order 11717 (38 FR 12315, dated May 11, 1973) and Part 6 of Title 15 CFR (Code of Federal Regulations).

NIST Interagency Reports (NISTIR)—A special series of interim or final reports on work performed

by NIST for outside sponsors (both government and non-government). In general, initial distribution is handled by the sponsor; public distribution is by the National Technical Information Service,

Springfield, VA 22161, in paper copy or microfiche form.
Coaxial Intrinsic

Impedance Standards

Robert T. Adair

Eleanor M. Livingston

Electromagnetic Fields Division

Center for Electronics and Electrical Engineering

National Engineering Laboratory

National Institute of Standards and Technology

Boulder, Colorado 80303-3328

U.S. DEPARTMENT OF COMMERCE, Robert A. Mosbacher, Secretary

NATIONAL INSTITUTE OF STANDARDS AND TECHNOLOGY, Raymond G. Kammer, Acting Director

Issued October 1989
National Institute of Standards and Technology Technical Note 1333

Natl. Inst. Stand. Technol., Tech Note 1333, 32 pages (Oct. 1989)

CODEN:NTNOEF

U.S. GOVERNMENT PRINTING OFFICE

WASHINGTON: 1989

For sale by the Superintendent of Docunnents, U.S. Governnnent Printing Office, Washington, DC 20402-9325
CONTENTS

Page

FIGURES V

TABLES Vi

ABSTRACT 1

1.0 INTRODUCTION 1

2.0 GENERAL BACKGROUND 2

3.0 BASIC TRANSMISSION LINE EQUATIONS 4

4.0 THE MEASUREMENT OF IMPEDANCE 7

5.0 TYPICAL IMPEDANCE STANDARDS 8

6.0 BASIC DESCRIPTION OF SIX- PORT ANAs 11

7.0 DUAL SIX-PORT ANA CALIBRATION TECHNIQUES 13

8.0 NEED FOR MORE PRECISE IMPEDANCE STANDARDS 14

9.0 DEPENDENCE OF COMPUTED ELECTRICAL IMPEDANCE

UPON PHYSICAL DIMENSIONS 15

10.0 SUMMARY 20

11.0 CONCLUSIONS 20

12.0 ACKNOWLEDGEMENTS 21

13.0 REFERENCES 21

1 1

FIGURES

Page

Figure 1 Typical discrete-parameter circuit. 2

Figure 2 Equivalent circuit of a transmission line showing

distributed circuit parameters in discrete form

per unit length of line. 3

Figure 3 Distributed inductance, resistance, conductance, and

capacitance of a coaxial transmission line. 4

Figure 4 Incident and reflected wave of a transmission line

terminated in a mismatched condition showing the

resulting standing wave. 6

Figure 5 Dimensional relations for an ideal coaxial air line. ... .9

Figure 6 Typical circuit port configurations. 11

Figure 7 Basic six-port network analyzer. 11

Figure 8 Single six-port vector network analyzer for measuring

one-port parameters. 12

Figure 9 Dual six-port vector network analyzer for measuring

two-port parameters. 13

Figure 10 Values of reflection coefficient magnitude versus

frequency for existing and desired precision 7 mm

coaxial air-line impedance standards. Typical NIST

six-port ANA resolution and uncertainties for reflection coefficient measurements on 7 mm coaxial devices

are also included. 19
TABLES

TABLE 1 Sununary of the improvement in connector performance

over the past four decades,

Page

14

TABLE 2 Summary of precision impedance standards (coaxial

air lines) characteristics. . . 16

TABLE 3 Typical and desired dimensional tolerances in the

fabrication of precision coaxial air-line conductors,

where D is the inner diameter of the outer conductor

and d is the outer diameter of the inner conductor. 16

TABLE 4 Calculated characteristic impedance (Z^) , reflection

coefficient magnitude (|r|), and return loss (RL)

for precision 14 mm, 7 mm, and 3 . 5 mm coaxial air lines

using ideal, existing, and desired fabrication toler- ances for conductor diameters D and d. . 17

VI
COAXIAL INTRINSIC IMPEDANCE STANDARDS

Robert T. Adair

and

Eleanor M. Livingston

This paper discusses how impedance standards are derived from the

basic definition of impedance, constructed and used in metrology

with coaxial air- line systems. Basic transmission line equations

are reviewed with emphasis given to intrinsic or derived standards

for obtaining the impedance in low- loss transmission line systems.

A brief description is given of how impedance standards are used

to calibrate the vector automatic network analyzer, and specifically, the six-port system automatic network analyzer used at the

National Institute of Standards and Technology for calibration

services in the radio frequency, microwave, and millimeter wave

areas. Measurement uncertainties are given for 7 mm coaxial

devices measured with the National Institute of Standards and

Technology six-port system. The resolution of our six-port system

is several orders more precise than that of the present impedance

standards from which it is calibrated. Required improvements in

the physical dimensions of air- line standards which permit the

automatic network analyzer's capability to be more fully utilized

are given.

Key words: automatic network analyzer; calibration services;

coaxial line; impedance; intrinsic; measurement uncertainties;

metrology; microwave; radio frequency; reflection coefficient;

scattering parameters; six-port systems; standards; transmission

line; 7 mm coaxial devices.

1.0 INTRODUCTION

Although the resolution of today's state-of-the-art automatic network

analyzers (ANAs) is approximately two orders of magnitude greater than what

can be used, the measurement accuracy of these ANAs have been limited because

of the lack of well-defined impedance standards. Impedance is one of the

basic electrical parameters used to describe and quantify electrical systems

and components. Historically, impedance standards have been one of the most

important and widely used standards in radio frequency (rf ) , microwave (/iw) , and millimeter wave (mmw) metrology. Sections of precision air line are the

basis for calculable impedance standards. They are nearly reflectionless and

represent the ultimate in adherence to design principles in maintaining a
constant characteristic impedance (Z^) throughout the precision air- line

sections. Today, ANAs are widely used in metrology because of their

versatility, sensitivity, resolution, and potential accuracy. Moreover, their

accuracy is directly dependent upon the quality of the impedance standards

used to characterize and evaluate the system parameters of the ANAs. Therefore, it is necessary to develop, characterize, and propagate a new class of

rf, juw, and mmw impedance standards.

2.0 GENERAL BACKGROUND

Coaxial impedance standards can be used from dc to above 60 GHz , but are widely used from audio frequencies up through 50 GHz. Propagation is

ordinarily restricted to the TEM (transverse electromagnetic) mode. Impedance

(Z) in a linear constant-parameter system can be defined [1,2] as the ratio of

the phasor equivalent of a steady-state sine-wave voltage or voltage-like

quantity (driving force) to the phasor equivalent of a steady- state sine -wave current or current-like quantity (response). The real part of impedance is

the resistance (R) . The imaginary part is the reactance (X) . For dc and low

frequencies below a few megahertz, impedances are easily modeled as shown in

Figure 1 by discrete parameters. However, at higher frequencies, such

impedances are more difficult to define since they do not maintain their low

frequency characteristics and therefore have to be represented as distributed

parameters for a unit length of line as shown in Figure 2.

I:,

0' Z|„- E,„

Z

n

^{ >{ Eout \

Figure 1. Typical discrete -parameter circuit
r~r —S"i

' —

G/2 : Y/2

o

o

Figure 2. Equivalent circuit of a transmission line showing distributed

circuit parameters in discrete form per unit length of line.

In the rf, /zw, and mmw frequency ranges where the wavelengths are relatively short, the electric and magnetic fields vary over small distances.

For coaxial transmission line systems, the electric field is restricted to the

region between the inner and outer conductors. When the dominant mode is the

TEM mode, the electric fields are radial and the magnetic fields concentric

circles. The impedance can be defined as the ratio of the electric field to

the magnetic field as they cross a plane perpendicular to the direction of

propagation [3]. This impedance is often called the intrinsic or characteristic impedance of the device and is designated Z^ . The term

"intrinsic" or "derived" is used to describe the impedance of a device by

itself and is determined by the physical properties of that device [4] . In

the case of a coaxial transmission line, Z^ is determined from the diameters

of the inner and outer conductors.

Finally, definitions [1] of accuracy, precision and resolution are presented as follows: accuracy - the degree of correctness with which a measured value agrees with the true value; precision - the degree of mutual

agreement among individual measurements, namely repeatability and reproducibility; resolution - the degree to which nearly equal values of a quantity can

be discriminated.
3.0 BASIC TRANSMISSION LINE EQUATIONS

Traveling waves are set up when voltage, £,„, is applied to the input of a transmission medium and input current, I|„, flows in the line as shown in

Figure 2 [5] . If the transmission line is sufficiently long then Z, Y, L, R,

G, and C are defined as uniformly distributed constants of the complex series

impedance Z in ohms, complex shunt admittance Y in Siemens, inductance L in

henries, dc resistance R in ohms, series conductance G in Siemens and capacitance C in farads, respectively, per unit length. These components are shown

in Figure 2 and presented for a coaxial transmission line in Figure 3. The

series resistance and inductance are shown in the equivalent

Center

Conductor

^Dielectric

Figure 3. Distributed inductance, resistance, conductance,

and capacitance of a coaxial transmission line.

circuits in Figure 2 and can be distributed equally or unequally in the transmission line [6]. Under these conditions, Z^ of the line is equal to Z.^ where

z,, = E,yi,, = Jin. (1)

Since Z = R + jwL

and Y = G + jwC,

then

R + jwL

G + jwC (2)
The input voltage, E^^ , requires a finite time to propagate along the

transmission line. The complex propagation constant, 7, [6] describes the

effect of the transmission line on the propagation of the traveling voltage

wave where a = attenuation constant in nepers per unit length and /3 - phase

constant in radians per unit length.

7 = yZY = J(R + jwDCG + jwC) = a+j^. (3)

The velocity of propagation [7] of the voltage wave is the phase

velocity, u^, where

i/p = w/^ = 27rf/^. (4)

The voltage and current at any distance, i, along the line are given

respectively by E^ = Ei^e"'"^ and I^ = I.^e'"^ , and they are related by the

characteristic impedance, Z^, of the line:

Z = Ein e-''Vl' in e-^*. (5) \ -^ /

Standing waves [6] result when we do not have the ideal transmission line

terminated in its characteristic impedance, Z^ . A portion of the incident

voltage wave is reflected at the line's terminating load, Z|_ , and reacts or interferes with the incident wave to set up a standing wave as shown in

Figure 4. The reflected wave carries energy that is not delivered to the

load. Reflection coefficient, T [3], is the term used to quantify this

reflected voltage wave. Its magnitude is the ratio of the amplitude of the

reflected voltage wave, E^, to that of the incident voltage wave, E|

Thus |r| = EyEj. (6)

The complex reflection coefficient can be expressed in terms of Z^ and Z^ [3].
Transmission

Line ('^^; System

Traveling

Waves

Standing

Waves

$$^^lncident Wave>$^^^^$^$$$^;

Z, +Zr

E^in

/T\

Jr\vty

\o' z

Terminating

Plane

Figure 4. Incident and reflected wave of a transmission line terminated in a mismatched condition showing the resulting standing wave.

Thus

or

r = (z, - z,)/(z, + zj

r= |r| A

where is the angle by which the reflected wave is displaced from the

incident wave.

Equation (7) can be rewritten in a different form which is also useful:

(7)

(8)

z, = z„ (1 + D/d - D (9)

Voltage standing wave ratio (VSWR) is defined as the resultant peak-totrough variation of the amplitude of the periodic standing wave created when a line is not terminated in Z^, . Specifically, VSWR, p, is the ratio of E^^^ to

E^jin of the standing wave [8]. VSWR can be expressed in terms of F, so

p = (1 + |r|)/(i - |r|) (10)
Equation (10) can be solved for |r| and yields a useful equation for

calculating |r| from VSWR:

I ^ VSUR - 1

' I VSWR + 1

• ^-^^^

Return loss, RL, is the ratio of the incident power, P., to the reflected

power, P^ , at a point on the transmission line, expressed in decibels.

Pi RL = 10 log -p^. (12)

r

Equation (12) can also be expressed in terms of |r| and VSWR:

RL = 20 log -r^ = 20 log ^f^^. (13)

4.0 THE MEASUREMENT OF IMPEDANCE

Historically, slotted lines and ref lectometers have been the most

accurate means available to measure impedance.

Precision slotted transmission lines [6] are excellent means of impedance

measurement. They are designed to measure the standing-wave pattern of the

electric field intensity as a function of the longitudinal position along a line. They determine the maximum and minimum magnitudes of voltage from which

VSWR is calculated. A matched impedance standard (such as a sliding load) is

used to calibrate (or evaluate) the slotted line since any residual VSWR due

to the slotted line must be separated from the value measured by the slotted

line. The impedance of the device under test (DUT) can then be calculated.

However, the effects of connectors on the measurement results cannot be

separated from those of the slotted line or of the DUT.

Tuned reflectometers [9,10] and untuned (broadband) reflectometers [11] are typically more precise (and more complex) than slotted lines for measuring

the magnitude and phase of the characteristic impedance. Their use requires a precision section of transmission line, a precision quarter-wave short-circuit
termination, and a precision quarter-wave offset open-circuit termination as the impedance standards. Very precise measurements can be performed on these

systems but typically these are performed manually and hence are timeconsuming and cumbersome.

Current and future methods of precise impedance measurements most

certainly lie in the realm of the computer-controlled ANA. In order to treat

the difficult area of rf , /iw, and mmw parameters more effectively both

magnitude and phase information is included. The following discussion is

based on the vector ANA rather than the scalar ANA. The recent publication of

the IEEE standard on network analyzers [12] is proof of the wide acceptance

and present use of vector ANAs for the major portion of complex measurements.

The National Institute of Standards and Technology (NIST) has done

considerable work in the design, development, and refinement of the six-port

automatic network analyzer [13,14]. The six-port ANA incorporates two sixport reflectometers , one on either side of the measurement insertion point.

The high precision and resolution of these systems demands a more accurate

standard than those currently available. The absolute accuracy of the

impedance standards used to calibrate these systems appears to be the limiting

factor in ANA measurement abilities.

Several commercially available ANAs demonstrate excellent complex

measurement capabilities, and they also require more accurate standards than

those currently available. See for example, [15,16].

5.0 TYPICAL IMPEDANCE STANDARDS

Important impedance standards can be placed into three major categories:

(1) matched terminations; (2) mismatched terminations; and (3) precision

sections of air-dielectric transmission lines.

Matched terminations are designed and constructed to match as precisely

as possible the characteristic impedance of the system in which they are to be

used. These are usually broadband devices. They can be fixed or they can be

adjustable terminations such as sliding loads [17, 18].

Mismatched terminations designed for specific values of impedance mismatch are available. They are used to test the measurement system's ability
to correctly measure impedances other than the characteristic impedance of the

system. These terminations [19] typically fall into three categories:

(1) short-circuit terminations (either flat or quarter-wavelength) ; (2) opencircuit terminations (either flat or quarter -wavelength offset), and (3) mismatches with fixed precise values of VSWR such as 1.2, 1.5, and 2.0. These

standard mismatch terminations furnish precisely calculated values of reflection coefficient magnitude and phase. When connected to the system of

interest they provide a means of testing the measurement system's- accuracy

.

Precision air-dielectric transmission line sections, commonly called "air

lines," provide calculable values of Z^ based on physical dimensions of the

individual air lines [20], The dimensions of interest for ideal coaxial

transmission lines are (1) the inner diameter, D, of the outer conductor and

(2) the outer diameter, d, of the inner conductor. Figure 5 shows the

relationship of the conductor's dimensions to the characteristic impedance of

a coaxial transmission line [20].

7 ~ 60 p„ D

Figure 5. Dimensional relations for an ideal coaxial air line,

The following equations provide a rapid and accurate means of determining

the electrical parameters of precision, air-dielectric, coaxial transmission

lines. For a dielectric medium of air, the relative permeability, n, = fi^, and
the relative permittivity, e^ = 6^ . When the line can be considered lossless

(R = G = 0), eq. (2) becomes

Z„ = yiTc" (14)

where L = (nJ27r) in (D/d)

and C = (27reJ(€/eJ/(£n(D/d)).

Then Z„ « (60/ye7) £n(D/d ) ,

where Z^ is in ohms. The value of the coefficient of £^(D/d) has been

determined at NIST to seven decimal places [20], so

for /i„ = 47r X 10"^ H/m,

and e„ = 8.854 192 x 10"'' F/m,

then Z„ = (59.958 4916/767) £n(D/d) Q. (15)

For e^ = 1.000 649,

Z„ = 59.939 0446 £n(D/d)

(16)

The value of e^ is computed from the refractive index of air [17] for ambient

conditions of 23°C, 50% relative humidity and an atmospheric pressure of

1.013 25 x 10^ Pa (760 Torr) . The value of e^ varies as a function of the

ambient conditions. For example, in Boulder, Colorado, where the atmospheric

pressure is 0.903 92 x 10^ Pa (678 Torr), the value for e, becomes 1.000 558.

Accurate, stable impedance standards are critically needed to characterize, calibrate, and support all ANAs including six-port ANAs . The equations

above help to illustrate the requirements for precision in the physical

dimensions of impedance standards, to produce the corresponding electrical

precision for the new generation of ANAs.

10
6.0 BASIC DESCRIPTION OF SIX-PORT ANAs

Single-port and six-port ANAs can be assembled from readily available

commercial components and instruments. Typical port configurations are shown

in Figure 6. The basis of the system can be thought of as a directional

coupler and three voltage probes connected as shown in Figure 7.

in

"in

One-Port Device

"in Y/2 Y/2

Port 1 Two-Port Device

"out U .

Port 2

Figure 6. Typical circuit port configurations

G>

voltage probes

000

Port 1

. A . . A , 6 6 X9

directional coupler

^yVA^

Port 2

Figure 7. Basic six-port network analyzer

11
A six-port network analyzer is a linear passive microwave network with

several ports. It is used to measure power and F. These parameters are measured at one port when a signal is applied to a second port, and the

remaining sidearm ports are terminated with power detectors. The power and

the reflection coefficient at the measurement port are calculated from the

sidearm power detector readings. Usually four sidearm detectors are used, so the network has six ports in all and is called a six-port reflectometer . A

typical six-port network analyzer consists of components such as directional

couplers or probes that couple in different ways to the incident and reflected

waves in a transmission line. Other designs are reported in the literature

[22].

When power detectors are connected to each of the ports as shown in

Figure 8, this device becomes a vector network analyzer. A major advantage of

this technique is that both amplitude and phase information can be obtained

from only amplitude information from the set of power detectors.

Power Detectors

From

Signal

Source

Measurement Port

Power Detectors

Figure 8. Single six-port vector network analyzer

for measuring one-port parameters.

12
7.0 DUAL SIX-PORT ANA CALIBRATION TECHNIQUES

The addition of a second six-port reflectometer to the system, as shown

in Figure 9, provides the means to measure two-port parameters in addition to

one-port parameters. Suitable six-port calibration techniques have been

sought for the past decade. Many techniques have been developed by many

different scientists [23-30] to calibrate six-port reflectoraeters . As in most

other developing technologies, the more we learn the more insights we have in

ways to improve upon previous work.

Signal Input Signal Input

H

Measurement L

Ports I I

Figure 9. Dual six-port vector network analyzer for

measuring two-port parameters.

Two distinctly preferred techniques for the calibration of six-port ANAs

have evolved from the research work in this area: (1) the Thru-Reflect-Line

(TRL) technique and (2) the Line-Reflect-Line (LRL) technique [31]. The LRL

technique has been expanded from (1) the One -Line Technique to (2) the TwoLine Technique to (3) the Five -Line Technique [32].

Typically, the procedure for calibration of a dual six-port ANA consists

of observation and analysis of the six-port's response to a set of suitably

chosen, known excitation and reflection conditions at the measurement ports.

This procedure yields a series of complex simultaneous equations which are then solved for the desired parameters in terms of scattering parameters [24]

13
8.0 NEED FOR MORE PRECISE IMPEDANCE STANDARDS

The advances in technology and metrology over the past decade have

greatly improved the measurement capabilities and performance of ANAs . The

precision and resolution of the current state-of-the-art computer-controlled

network analyzers are sufficient to detect and quantify the effects of individual circuit components. The component of variation due to inadequacies of

coaxial connectors and waveguide flanges can be seen and isolated from the

rest of the measurands . The difficulty lies in the fact that the certifiable

absolute accuracy of tnese ANAs is approximately two orders of magnitude less

than their precision, based upon present impedance standards. Also, greatly

improved connectors are becoming available. Table 1 summarizes the improvements in coaxial connectors over the past four decades [33]. The newer highprecision connectors allow more precise measurements than were possible in the

past, and therefore require more accurate impedance standards than those

currently available. Impedance standards with reflection coefficients less

than 0.0002 are needed if ANA accuracies with two additional orders of

magnitude are to be achieved.

TABLE 1

Summary of the improvement in connector performance over the past four decades.

Connector

Type

Approximate

Date of

Introduction

VSWR at 5 GHz

(Typical)

Approximate

Maximum Usable

Frequency (GHz) Symmetrical

Well-Defined

Mating Plane

GR874 1948 1.035 7 Yes No

OR 900 1963 1.005 8.5 Yes Yes

Free. N 1967 1.02 18 No No

APC7 1968 1.007 18 Yes Yes

APC 3.5 1976 1.006 32 No No

K2.92 1983 1.01 40 No No

PC 2.4 1986 1.01 50 No No

PC 1.85 1987 < 1.016 65 No No

14
9.0 DEPENDENCE OF COMPUTED ELECTRICAL IMPEDANCE UPON PHYSICAL DIMENSIONS

Lengths of precision air-dielectric coaxial transmission lines have

constant impedance and are nearly ref lectionless . They actualize ideal design

principles and are the bases for calculable impedance standards. The electrical performance of these air lines depends primarily on the diameters of

the two conductors as illustrated in Figure 5. The effect of irregularities

in the diameters of these conductors on the characteristic impedance of a 50 H

coaxial line can be computed from AZ^/Z^, . Differentiation of (15) yields

AZo AD _ Ad 1

Z^ ' D d ' £n (D/d) '

where AZ^ is the change in characteristic impedance, D is the inner diameter

of the outer conductor, AD is the deviation of D, d is the outer diameter of

the inner conductor, and Ad is the deviation of d [34] . The currently achievable dimensional tolerances in the fabrication of

coaxial conductors in various sizes are listed in Table 2. The desired

dimensional tolerances and frequency ranges are also listed. Table 3 shows

typical and desired dimensional tolerances of coaxial conductors. The effect

of air-line conductor's dimensional tolerances on Z^ and on |r| can be

determined with equations (16) and (7), respectively (assuming that Z^ is

constant), and with the dimensional tolerances found in Table 3.

The outer conductor tolerances, AD, are added to the outer conductor

dimension, D, and the center conductor tolerances, Ad, are subtracted from the

center conductor dimension, d, to obtain the "worst- case" effect on Z^ of the

precision air-line impedance standards. Equation (16) then takes the

following form for these calculations:

Z„ = 59.939 0446 £n [ (D+AD)/(d- Ad) ] . (18)

Table 4 is a tabulation of the changes in Z^ and |r| as a function of AD and

Ad for 14 mm, 7 mm, and 3 . 5 mm air lines, respectively, using a value of 50 n

for Z, .

15
TABLE 2

Summary of precision impedance standards (coaxial air lines) characteristics.

Air-Line Diameter

(mm)

Connector

Type

Frequency Range

(GHz)

Tolerance (uin.)*

Typical Desired

14

7

7

3^

3.5

2.92

2.4

1.85

PC 1.85 N

PC 7

SMA

PC 3.5 K

PC 2.4 PC 1.85

0-8.5

0-18

0-18

0-25

0-32

0-40

0-50

0-65

100-200

250-500

100-200

100-500

100-200

50-100

100-200

15-25

15-25

15-25

10-20

10-20

25-50

10-20

5-15

TABLE 3

Typical and desired dimensional tolerances in the fabrication of precision coaxial air-line conductors, where D is the inner diameter of the outer conductor and d is the outer diameter of the inner conductor.

Air-Line

Diameter (in.)*

Dimensions (in.)

(Typical) (Desired)

14 mm

D = 0.562 500*

d = 0.244 255

7 mm

D = 0.275 591

d = 0.119 670

3.5 mm

D = 0.137 795

d = 0.059 835

2.4 mm

D = 0.094 490

d = 0.041 064

Ad = +0.000 100*

AD = ±0.000 200

Ad = +0.000 100

AD = ±0.000 200

Ad = +0.000 100

AD = ±0.000 200

Ad = +0.000 100

AD = ±0.000 200

+ 0.000 015*

±0.000 025

+ 0.000 015

±0.000 025

+ 0.000 010

±0.000 020

+ 0.000 010

±0.000 020

*Note: Dimensions are those given by the manufacturers and are not converted to SI

to avoid round-off errors.

16
TABLE 4

Calculated characteristic impedance (Z^), reflection coefficient magnitude ( F ), and return

loss (RL) for precision 14 mm, 7 mm, and 3.5 mm coaxial air lines using ideal, existing and

desired fabrication tolerances for conductor diameters D and d. D = Inner diameter of outer conductor,

d = Outer diameter of inner conductor.

Z,, = 59.939 0446 In (D + AD/d - Ad). r = (z„ - 50.000 ooo)/(z„ + 50.000 ooo) . RL = Return Loss = 20 log (1/ T ).

D d

Tolerance (in.) (in.) AD

(Min.)

Ad

(urn.)

^0n r RL

(dB)

14 mm diameter air line

Ideal 0.562 500 0.244 255

Typical 0.562 700 0.244 155

Desired 0.562 525 0.244 240

+ 200

+ 25

-100

-15

49.999 85

50.045 71

50.006 20

0.000 001

0.000 457

0.000 062

120.0

66.8

84.2

7 mm diameter air line

Ideal 0.275 591 0.119 670

Typical 0.275 791 0.119 570

Desired 0.275 616 0.119 655

+ 200

+ 25

-100

-15

49.999 95

50.093 54

50.012 90

0.000 0005

0.000 935

0.000 129

126.0

60.6

77.8

3.5 mm diameter air line

Ideal 0.137 795 0.059 835

Typical 0.137 995 0.059 735

Desired 0.137 815 0.059 825

+ 200 + 20

-100

-10

49.999 73

50.186 92

50.018 45

0.000 003

0.001 866

0.000 184

110.5

54.6

74.7

17
Figure 10 illustrates the relative resolution and uncertainties of the

NIST six-port ANA measurements of |r| on 7 mm coaxial devices. The random

uncertainty includes the effect of the connectors on the standards and on the

DUT. The resolution for the measurement of |r| by the NIST six-port ANA

ranges from a value of 0.000 02 at 2 GHz to a value of 0.000 05 at 18 GHz.

The desired |r| of 0.0002 for the new generation of precision coaxial air

lines is compared to the value of |r| of 0.001 for the existing air- line

impedance standards. The present estimated uncertainty in the measurement of

|r| by the NIST six-port ANA is a few parts in 10^.

The total uncertainty, U^., is the sum of the random and systematic

uncertainties. The NIST six-port ANA value of Uj for the measurement of |r|

of 7 mm coaxial devices can be determined using the equation

Ut = 3L3, +fV + A. (19)

S|«j|gT is the random uncertainty associated with the calibration of the NIST

dual six-port ANA and is given as one standard deviation. S^ is the standard

deviation computed from n connections of the DUT, and A represents the systematic uncertainty of the NIST six-port ANA. Three standard deviations are taken as the overall random uncertainty; hence the factor of 3.

Typical values of Uj, S|^isj, and A are plotted in Figure 10. Values of S^. are not plotted since they are frequency dependent and are different for each

DUT. Typical values of S, range from 0.0001 at 1 GHz to 0.001 at 18 GHz under

normal conditions

The measurement of conductor diameters to achieve the desired accuracy is

typically a tedious and difficult process. Air gauges [35,36], capacitance

gauges [37], and laser micrometer gauges [38] provide excellent techniques to

measure these diameters, but a nxomber of problems must be solved during the

measurement process. These include mechanical standards of diameter, un- certainties in the measuring system, measurement environment, and handling

techniques, all of which directly affect the measurements.

18
0.01

Typical Systematic

Uncertainty (A) for NIST Six-Port ANA

0.001

n

0.0001

0.00001

r| of Existing Air- Line Impedance

Standards

Typical Random Uncertainty

(One Standard Deviation, Snist)

for NIST Six-Port ANA

I|r|of Desired Air-Line Impedance Standards

Resolution of NIST

Six-Port ANA'

Note: S„ is the connector standard deviation based on n connections during the calibration —

of the device under test (DUT). S^^ is not shown

on this graph since it is different for each DUT.

8 12

Frequency (GHz)

16 20

Figure 10. Values of reflection coefficient magnitude versus frequency

for existing and desired precison 7 mm coaxial air- line

impedance standards. Typical NIST six-port ANA resolution

and uncertainties for reflection coefficient measurements on

7 mm coaxial devices are also included.

19
10.0 SUMMARY

Transmission line concepts are used in electronic measurement systems to

provide an integral, vital part of rf, /iw, and mmw measurements. Precision

sections of coaxial air-dielectric transmission lines are the most precise

impedance standards in existence [39]. They are necessary for the calibration

and support of ANAs which now make up a majority of the active measurement

systems. The accuracy and capability of ANAs depend directly upon the

precision and accuracy of the physical dimensions of these air lines. The

electrical parameters of these impedance standards are calculated from their

physical properties and dimensions, and consequently depend directly upon the

quality of their fabrication and the measurement of their physical properties.

The results presented in this report are best shown in terms of the

parameter of interest which must be improved to achieve these goals -- the

reflection coefficient of the impedance standard. The degree of improvement

in the reflection coefficient is directly dependent upon improvements in the

physical dimensions of the impedance standards. Data which quantitatively

define the degree of uncertainty in reflection coefficient produced by

specific values of uncertainty in each physical dimension of the impedance

standard are presented.

11.0 CONCLUSIONS

The resolution of state-of-the-art ANAs now is approximately two orders

of magnitude greater than their accuracy. Therefore, significantly greater

accuracies in impedance standards than those currently available are needed to use the full capability of these ANAs. Impedance standards with reflection

coefficients of less than 0.0002 are required to utilize the existing ANA

resolution. The dimensional tolerances of the impedance standards necessary

to achieve this level of reflection coefficient are now known, as presented in

this document. Thus, a new generation of impedance standards which will

provide the required accuracy and precision not currently available in the

impedance standards of today must be developed. The results presented in this

report provide a quantitative analysis of the methods needed to achieve the

full capability of the present ANAs.

20
12.0 ACKNOWLEDGEMENTS

The authors sincerely thank William E. Little for his encouragement and

helpful suggestions. Also the technical assistance of William C. Daywitt,

Robert M. Judish, and Kurt M. Phillips is greatly appreciated as are the

contributions of the technical readers, Cletus A. Hoer, Ramon L. Jesch, and

Mark T. Ma. Special thanks go to Terry Yenser and Karen Martin for their

tireless and cheerful efforts in preparing this manuscript.

13.0 REFERENCES

[I] Jay, F. , ed. in chief. IEEE standard dictionary of electrical and

electronic terms, ANSI/IEEE Std. 100-1984. 1174 pp.

[2] Jones, R.N. The measurement of lumped parameter impedance: A Metrology

Guide. Natl. Bur. Stand. (U.S.) Monogr. 141; 1974 June. 211 pp.

[3] Kerns, D.M. Definitions of v, i, Z, Y, a, b, T, and S. Proceedings of

the IEEE. 55(6): 892-900; 1967 June.

[4] Michels, W.C., ed. in chief. The international dictionary of physics

and electronics . Second edition. D. Van Nostrand Company, Inc. 1961.

1356 pp.

[5] Beatty, R.W. Microwave impedance measurements and standards.

Natl. Bur. Stand. (U.S.) Monogr. 82; 1965 August. Reprinted June 1969

with Errata. 32 pp.

[6] Lance, A.L. Introduction to microwave theory and measurements . New York: McGraw-Hill Book Company; 1964. 308 pp.

[7] Gardiol, F.E. Introduction to microwaves. Artech House, Inc.; 1984.

495 pp.

[8] Jesch, R.L. ; Jickling, R.M. Impedance measurements in coaxial

waveguide systems. Proc . IEEE. 55(6): 912-923; 1967 June.

[9] Engen, G.F.; Beatty, R.W. Microwave reflectometer techniques. IRE

Trans. Microwave Theory Tech., MTT-7 (3): 351-355; 1959 July.

[10] Anson, W.J. A Guide to the use of the modified reflectometer technique

of VSWR measurement. J. Res. Natl. Bur. Stand. (U.S.) Engineering and

Instrumentation. 65C (4): 217-223; 1961.

[II] Little, W.E. ; Ellerbruch, D.A. Precise reflection coefficient

measurements with an untuned refectometer . J. Res. Natl. Bur. Stand.

(U.S.) Engineering and Instrumentation. 70C (3): 165-168;

1966 July-September.

21
[12] IEEE standard on network analyzers (100 kHz to 18 GHz). ANSI/IEEE Std.

378-1986. 15 pp.

[13] Hoer, C.A. A network analyzer incorporating two six-port

reflectometers. IEEE Trans. Microwave Theory Tech., MTT-25 (12):

1070-1074; 1977 December.

[14] Juroshek, J.R. A technique for extending the dynamic range of the dual

six-port network analyzer. IEEE Trans. Microwave Theory Tech., MTT-33

(6) : 453-459; 1985 June.

[15] Hewlett-Packard. Millimeter wave vector measurements using the HP 8510A

network analyzer. Product Note No. 8510-1. 1984 November. 15 pp.

[16] Wiltron Company. Wiltron 5600 Series automated network analyzer

systems. 10 MHz to 40 GHz. Production Note. 1984 January. 14 pp.

[17] Little, W.E.; Wakefield, J. P. A coaxial adjustable sliding termination.

IEEE Trans. Microwave Theory Tech., MTT-12 (2): 247-248; 1964 March.

[18] Weinschel, B.O.; Sorger, G.U.; Raff, S.J.; Ebert , J.E. Precision

coaxial VSWR measurements by coupled sliding- load technique. IEEE

Trans. Instrum. Meas ., IM-13 (4): 292-300; 1964 December.

[19] Beatty, R.W. ; MacPherson, A.C. Mismatch errors in microwave power

measurements. Proc . I.R.E. 41 (9): 1112-1119; 1953 September.

[20] Nelson, R.E.; Coryell, M.R. Electrical parameters of precision coaxial,

air-dielectric transmission lines. Natl. Bur. Stand. (U.S.) Monogr . 96;

1966 June. 104 pp.

[21] Essen, L. ; Froome , K.D. Dielectric constant and refractive index of air

and its principal constituents at 24 GHz. Nature. (167): 512;

1951 March.

[22] McGraw-Hill Yearbook of Science and Technology: NY 1986; 289-292.

[23] Engen, G.F.; Hoer, C.A. "Through-reflect-line" : an improved technique

for calibrating the dual six-port automatic network analyzer. IEEE

Trans. Microwave Theory Tech., MTT-27 (12): 987-993; 1979 December.

[24] Hoer, C.A. Calibrating two six-port reflectometers with an unknown

length of precision transmission line. IEEE MTT-S International

Microwave Symposium Digest. June 27-29, 1978: 176-178.

[25] Engen, G.F.; Hoer, C.A., and Speciale, R.A. The application of

"through-short-delay" to the calibration of the dual six-port. IEEE

MTT-S International Microwave Symposium Digest. June 27-29, 1978: 184-

185.

22
[26] Engen, G.F.; Hoer, C.A. "Through- load-delay" : An improved technique for

calibrating the dual six-port. IEEE MTT-S International Microwave

Symposium Digest. April 30-May 4, 1979: 53.

[27] Engen, G.F. On-line accuracy assessment for the dual six-port ANA:

background and theory. IEEE Trans. Instru. Meas . IM-36 (2): 501-506:

1987 June.

[28] Judish, R.M. On-line accuracy assessment for the dual six-port ANA:

statistical methods for random errors. IEEE Trans. Instrum. Meas.,

IM-36 (2): 507-513; 1987 June.

[29] Hoer, C.A. On-line accuracy assessment for the dual six-port ANA:

treatment of systematic errors. IEEE Trans. Instrum. Meas., IM-36 (2):

520-523; 1987 June.

[30] Juroshek, J.R. On-line accuracy assessment for the dual six-port ANA:

experimental results. IEEE Trans. Instrum. Meas., IM-36 (2): 524-529;

1987 June.

[31] Hoer, C.A.; Engen, G.E. On-line accuracy assessment for the dual ANA:

extension to non-mating connectors. (To be published.)

[32] Hoer, C.A. Some questions and answers concerning air lines as impedance

standards. 29th Automatic RF Techniques Group (ARFTG) Conference,

Las Vegas, NV. June 12-13, 1987. 161-173.

[33] Somlo, P.I.; Hunter, J.D. Microwave impedance measurement. Peter

Peregrinus Ltd., London, UK. 1985; 213 pp.

[34] MacKenzie, T.E.; Sanderson, A.E. Some fundamental design principles for

the development of precision coaxial standards and components. IEEE

Trans. Microwave Theory Tech., MTT-14 (1): 29-39; 1966 January.

[35] Kirk, David B. Introduction to principles of pneumatic gaging. Tech.

Paper 8101. American Society of Mechanical Engineers Annual Meeting,

December 5, 1952.

[36] Kennedy, C.W. Inspection and gaging. 3rd ed. New York: Industrial

Press, 1962. 580 pp.

[37] Sanderson, A.E.; Van Veen, F.T. The precise measurement of small

dimensions by a capacitance bridge. Gen. Radio Exper. 38 (2):

1964 February.

[38] Hewlett-Packard. Traceability and the HP 8510 Network Analyzer.

Publication No. 5954-1552. 1985 November. 11 pp.

[39] Weinschel, B.O. Air-filled coaxial lines as absolute impedance

standards. Microwave J. 7 (4): 47-50; 196^ April.

23

ft us, GOVERNMENT PRINTING OFFICE 1969^acket 773-W3/25O01 Region 8

NBS-n4A (REV. 2-eci U.S. DEPT. OF COMM. BIBLIOGRAPHIC DATA

SHEET (See instructions)

1. PUBLICATION OR

REPORT NO. NIST/TN-1333

2. Performing Organ. Report NoJ 3. Publication Date

October 1989

4. TITLE AND SUBTITLE

COAXIAL INTRINSIC IMPEDANCE STANDARDS

5. AUTHOR(S)

Robert T. Adair and Eleanor M. Livingston

6. PERFORMING ORGANIZATION (l( joint or other tlian NBS, see mstruct/ons;

NATIONAL INSTITUTE OF STANDARDS AND TECHNOLOGY

U.S. DEPARTMENT OF COMMERCE

BOULDER, OOIDRADO 80303-3328

7. Contract/Grant No.

8. Type of Report & Period Covered

9. SPONSORING ORGANIZATION NAME AND COMPLETE ADDRESS (Street, City. State. ZIP)

10. SUPPLEMENTARY NOTES

I I Document describes a computer program; SF-I8S, FlPS Software Summary, is attached.

11. ABSTRACT (A 200-word or /ess factual summary of most si gnificant information. If document includes a si gnificant

bibliography or literature survey, mention it here)

This paper discusses how impedance standards are derived from the basic definition

of impedance, constructed and used in metrology with coaxial air-line systems.

Basic transmission line equations are reviewed with emphasis given to intrinsic

or derived standards for obtaining the impedance in low-loss transmission line

systems. A brief description is given of how impedance standards are used to

calibrate the vector automatic network analyzer, and specifically, the six-port

system automatic network analyzer used at the National Institute of Standards and

Technology for calibration services in the radio frequency, microwave, and millimeter wave areas. Measurement uncertainties are given for 7 mm coaxial devices

measured with the National Institute of Standards and Technology six-port system.

The resolution of our six-port system is several orders more precise than that of

the present impedance standards from which it is calibrated. Required improvements

in the physical dimensions of air-line standards which permit the automatic network analyzer's capability to be more fully utilized are given.

12. KEY WORDS (Six to twelve entries; alphabetical order; capitalize only proper names; and separate key words by semicolons)

automatic network analyzer; calibration services; coaxial line; impedance; intrinsic;

measurement uncertainties; metrology; microwave; radio frequency; reflection coefficient; scattering parameters; six-port systems; standards; transmission line; 7 mm

coaxial devices

13. AVAILABILITY

^g Unlimited Q^ For Official Distribution. Do Not Release to NTIS

JiY] Order From Superintendent of Documents, U.S. Government Printing Office, Washington, D.C.

20402.

\2Zi Order From National Technical Information Service (NTIS), Springfield, VA. 22161

14. NO. OF

PRINTED PAGES

32

15. Price

vj6COMM-DC <.v-4J-Pt

rhcNational Institute of Standards and Technology* was established by an act of Congress on March 3, 1901. The Institute's overall goal is to strengthen and advance the Nation's science and technology and

fecilitate their effective application for public benefit To this end, the Institute conducts research to assure intemadooal competitiveness and leadership of U.S. industry, science and technology. NIST work involves development

and transfer of measurements, standards and related science and technology, in support of continually improving

U.S. productivity, product quality and reliability, innovation and underlying science and engineering. The Institute's technical work is performed by the National Measurement Laboratory, the National Engineering Laboratory, the National Computer Systems Laboratory, and the Institute for Materials Science and Engineering.

The National Measurement Laboratory

Provides the national system of physical and chemical measurement;

coordinates the system with measurement systems of other nations and furnishes essential services leading to accurate and uniform

physical and chemical measurement throughout the Nation's scientific community, industry, and commerce; provides advisory and research

services to other Government agencies; conducts physical and chemical

research; develops, produces, and distributes Standard Reference

Materials; provides calibration services; and manages the National

Standard Reference Data System. The Laboratory consists of the following centers: The National Engineering Laboratory

Basic Standards^ Radiation Research

Chemical Physics

Analytical Chemistry

Provides technology and technical services to the public and private

sectors to address national needs and to solve national problems;

conducts research in engineering and applied science in support of these

efforts; builds and maintains competence in the necessary disciplines

required to carry out this research and technical service; develops engineering data and measurement capabilities; provides engineering measure- ment traceabflity services; develops test methods and proposes engineering standards and code changes; develops and proposes new

engineering practices; and develops and improves mecham'sms to trtmsfer results of its research to the ultimate user. The Laboratory

consists of the following centers: The National Computer Systems Laboratory

Computing and Applied

Mathematics

Electrom"cs and Electrical Engineering*

Manufacturing Engineering

Building Technology

Fire Research

Chemical Engineering'

Conducts research and provides scientific and technical services to aid Federal agencies in the selection, acquisition, application, and use of

computer technology to improve effectiveness and economy in Government operations in accordance with Public Law 89-306 (40 U.S.C. 759), relevant Executive Orders, and other directives; carries out this mission

by managing the Federal Information Processing Standards Program,

developing Federal ADP standards guidelines, and managing Federal

participation in ADP voluntary standardization activities; provides scientific and technological advisory services and assistance to Federal

agencies; and provides the technical foundation for computer-related

policies of the Federal Government The Laboratory consists of the following divisions: The Institute for Materials Science and Engineering

Information Systems

Engineering

Systems and Software

Technology

Computer Security

Systems and Network

Architecture

Advanced Systems

Conducts research and provides measurements, data, standards, refer- ence materials, quantitative understanding and other technical information fundamental to the processing, structure, properties and perfor- mance of materials; addresses the scientific basis for new advanced

materials technologies; plans research around cross-cutting scientific themes such as nbndestructive evaluation and phase diagram develop- ment; oversees Institute-wide technical programs in nuclear reactor

radiation research and nondestructive evaluation; and broadly disseminates generic technical information resulting from its programs. The

Institute consists of the following divisions:

Ceramics

Fracture and Deformation'

Polymers

Metallurgy Reactor Radiation

'Headquarter* ud Laboratork* at Oaithcnburg, MD, unle» otherwise noted; mailing addrea

Oaithenburt. MD 20899. *Some dirisloni within the center are located at Boulder, CO 80303.

* locattid at Boulder, CO, with aome elcmenta at Oaithenburg, MD.
U.S. DEPARTMENT OF COMMERCE

National Institute of Standards and Technology

(formerly National Bureau of Standards)

325 Broadway

Boulder, Colorado 80303-3328

OFFICIAL BUSINESS

PENALTY FOR PRIVATE USE, $300