MONITOR -

MINIMUM DETECTABLE SIGNAL
Frequency Responce Setting - LINEAR MODE

Date Time
Mar-26-91 11:31AM
Threshold Current mA
Trial :
Number

wenn
ees
BSSNR
22323

NOTE: Needle moves about 0.005mG

Frequency Responce Setting - FLAT MODE

Date Time
Mar-26-91 11:37AM,

Threshold Current mA
1 0.20 mA
Trial 2 0.12 mTM
Number 3 0.10 m
4 0.09 mA
5 0.11 mA

METER RESOLUTION DATA

Frequency Responce Setting - LINEAR MODE

Date Time
Mar-26-91 11:45AM

Drive Current Start A
1 1.4720 A
Trial 2 1.4720 A
Number: 3 1.4720 A
4 1.4720 A
5 1.4720 A

Frequency Responce Setting - FLAT MODE

Date Time
Mar-26-91 11:40AM

Orive Current Start A
1 1.4800 A
Trial 2 1.4800 A
Number 3 1.4800 A
4 1.4800 A
5 1.4800 A

Temperature
59° F

Threshold Field

Temperature

59° F

Threshold Field

10.79
6.47
5.40
4.86
5.93

6.69

Temperature
70° F

Threshold Current

1.4860
1.4860
1.4890
1.4850
1.4840

Temperature
60° F

Threshold Current

1.4930
1.4930
1.4930
1.4930
1.4930

Appendix VIA

MODEL 42b-1 - SN 1286

Meter Background

min 0.01 mG

max 0.01 mG

uG

uG

uG

uG

uG

uG

UG AVG THRESHOLD FOR SENSITIVITY
Meter Background

min 0.01 mG

max 0.01 mG

uG

uG

uG

uG

uG

uG

uG AVG THRESHOLD FOR SENSITIVITY
Meter Background

min 0.01 mG

max 0.01 mG

A Resolution uG Resolution %

A 755.29 uG 0.95%

A 755.29 uG 0.95%

A 917.13 UG 1.15%

A 701.34 uG 0.88%

A 647.39 UG 0.81%

755.29 uG 0.95%

Meter Background

min 0.01 mG

max 0.01 mG

A Resolution uG Resolution

A 701.34 uG 0.87%

A 701.34 UG 0.87%

A 701.34 UG 0.87%

A 701.34 UG 0.87%

A 701.34 uG 0.87%

701.34 uG 0.87%


6S

Responce (dB)

40 =

20 -

Ss
|

1
8
l

-40 4

Wi
Frequency Responce - Meter VA nm ae ateve
IRC Model IRC-109 + Sine Weve
— Bezier Fit

TTT

10

T TTT TT TTT TTT TTT]

10? 10° 104

Frequency (Hz)


HOLADAY - HI 3604 - SN 76111

FREQUENCY RESPONCE PRIMARY DATA

Date Time Temperature Meter Background
Feb-14-91 10:04AM 75° F 0.182 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
1 Hz 2m 1006.18 mG 0.22 mG +73.09 dB
2 Hz 2 mG 1006.18 mG 0.27 mG +71.20 dB
5 Hz 2 mG 1006.18 mG 0.63 mG -64.01 GB
10 Hz 20 mG 1006.18 mG 7.00 mG 743.15 cB
20 Hz 200 mG 404.30 m6 150.00 mG -8.61 GB
50 Hz 200 mG 145.38 mG 150.20 mG
60 Hz 200 mG 144.93 mG 150.00 mG
100 Hz 200 mG 146.50 mG 150.00 mG
200 Hz 200 mG 147.37 mG 150.10 mG
500 Hz 200 mG 149.30 mG 150.00 mG
1 kHz 200 mG 159.14 mG 150.00 mG
2 kHz 200 mG 250.89 mG 150.10 mG
5 kz 200 mG 861.90 mG 68.00 mG +22.06 dB
10 kHz 20 mG 551.71 mG 4.55 mG +41.67 dB
FREQUENCY RESPONCE AUXILIARY DATA
Date Time Temperature Meter Background
Feb-14-91 10:45AM 75° F 0.172 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
30 kz 200 mG 169.72 mG 150.00 mG +1,07 dB
40 Hz 200 mG 148.24 mG 150.10 mG 0.10 oB
3 kHz 200 mG 515.57 mG 150.00 mG -10.72 dB
15 Hz 200 mG 969.65 mG 89.20 mG 20.72 dB

Appendix IXA


eet

OB MoazZOVOMSD

0.44

Susceptibility - Meter II1
SAFE PROMETER SN 1263

60 Hz
E-FIELO 10kKV At

4 MHz 10 MHz 6100 Hz
UNHOD

4 MHz

10 MHz 100 MHz
Genz HOD

CORED OFF
FELD ON


Se

Accuracy and Linearity - Meter VIA - Linear Mode

MONITOR Model 42b-1 SN 1286

Pet ht

10 =
5 4
5
§ IA
2
3
&
-5 4
-10
197

TTT es TTT TTT TTT TTT]

1 10 10? 107
Applied Field (mG)


CODE

11

12
TA]
1A2
Ill
112
IIIA
ITIB
IVA
IVB
VA
VB
VIA
VIB
VIIA
VIIB
VITIA
VIIIB
IXA
IXB

XA
XIVA
XIVB
XVA
XVB
XVIA
XVIB
XVITA
XVIIB

MANUFACTURER

WRI
WRI

WRI

WRI

SAFE
SAFE
EXPAN
EXPAN
MSI

MST

IRC

IRC
MONITOR
MONITOR
SYDKRAFT
SYDKRAFT
EMD

EMD
HOLADAY
HOLADAY
WALKER
WALKER
ELF

ELF
DINDIMA
DINDIMA
MEDA
MEDA
MAG TECH
MAG TECH

MODEL #

" 670

670

660

660

Pro Meter
Pro Meter
1A

1A

MSI-20
MSI-20
IRC-109
IRC-27
42B-1
42B-1
88-05
88-05
ACGM-1
ACGM-1
HI-3604
HI-3604
ELF-50D
ELF-50D
Metex-3800
Metex-3800
Gaussmaus
Gaussmaus
PLM-100
PLM-100
ELF 12
ELF 12

SERIAL #

G1257
G1259
G1241
G1242
1263
1264
1103
1104
79011626
79010854

1286
1287

01-0116
01-0117
76111
76112
71181-205
71181-204
542513
542832
GM0400
GM0401
1380

1381

2

4

TABLE 3
154

MIN DET SIGNAL RESOLUTION

103
44
116
152
187
145
87
58
186
182
76
392

11

158
324
22
12
2751
1596
173
167
284
335
25
78
36146
8696

uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG
uG

=
>

22% @ 10
-85% @ 7
-18% @ 7
-41%@ 8
-82% @ 93
-57% @ 94
-90% @ 20
-46% @ 19
22% @103
-20% @102
-37% @101
17% @ 94
-87% @ 80
0.37% @ 78
-81% @ 97
-42% @ 97
«33% @105
-31% @104
23% @ 97
36% @100
-93% @104
-90% @105
-36% @ 99
-30% @100
-93% @ 77
-11% @ 77
18% @ 10
-13% @ 10
-45% @ 48
24% @ 52

—- en
a)

ecooeo eC COC OO OOO OC CO coooeoaea aoe oO

> >
=“ oO

mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG
mG


6¢1

On MnzotUumMs

e.64

0.4

fi at a

Susceptibility - Meter XVIA
MEDA Model PLM-100 SN 1380

e-Fiehe Ven

HB

4 iz 10MHz 108 Hz

19 fits tee Mz


Mention of trade names,
products, or services does not
convey, and should not be
interpreted as conveying,
official EPA approval,
endorsement, or recommendation.


6IT

OB Mazo vKOMs

N

O

Susceptibility - Meter I1
WRI Model 670 SN G1257

60 Hz
E-FIELD 1oKv mt

1 MHz 16 MHz «6160 MHz
UNMOD

4 MHz

16 MHz 408 MHz
6e@Hz MOD

(COFIELD OFF
FELD ON


89.

40 -

20 4

8
|

Responce (dB)
%
Ss
|

~40 4

-60

Frequency Responce - Meter VIIIB
EMD Model ACGM-1 SN 01-0117

TT TTT TTT TTY i TTT TTT]

10. 10? 3? at
Frequency (Hz)

[+b0

Square Wave
Santooth Wave
Sine Wave
Bezier Fit


MONITOR - MODEL 42b-1 - SN 1287
METER ANGULAR RESPONCE DATA

Frequency Responce Setting - LINEAR MODE

Date Time Temperature Meter Background
Jan-31-91 08:40AM 74° F 0.2 mG
ACTUAL FIELD MEASURED FIELD ANGULAR RESPONCE

147.24 mG 150.00 mG 1.00

149.85 mG 150.00 mG 0.98

157.63 mG 150.00 mG 0.93

170.67 mG 150.00 mG 0.86

195.55 mG 150.00 mG 0.75

228.96 mG 150.00 mG 0.64

291.13 mG 150.00 mG 0.50

459.96 mG 150.00 mG 0.32

919.00 mG 150.00 mG 0.16

1116.00 mG 20.00 mG 0.01

Frequency Responce Setting - FLAT MODE

Date Time Temperature Meter Background
dan-29-91 03:14PM The F 0.11 mG
Angle - ACTUAL FIELD MEASURED FIELD ANGULAR RESPONCE

o° 150.00 mG 1.00

10° 150.00 mG 0.99

20° 150.00 mG 0.93

30° 150.00 mG 0.86

40° 150.00 mG 0.76

50° 150.00 mG 0.63

60° 150.00 mG 0.49

70° 150.00 mG 0.32

80° 150.00 mG 0.11

Appendix VIB


2g

os mozotums

100+

a+

Om

Susceptibility - Meter XVIIB
MAG TECH ELF 12 SN4

ian |

iml

ial

60 Hz
E-FIELD 10k

4 MHz

4eMHz 1
UNHOD

ee MHz

4 MHz

10. MHz 108 MHz
Genz MOD

OOFIELD OFF
FIELD ON


6L

40 -

20 >

o-

-20 4

Responce (dB)

-40 4

Frequency Responce - Meter XVIA 1 Square Wave
A Sastooth Wave

MEDA Model PLM-100 SN 1380 + Sine Wave

— Bezier Fit

TT 1-111] rE T1117]

10 . . 10? 107

Frequency (Hz)


WALKER - ELF-50D - SN 71181-205

FREQUENCY RESPONCE PRIMARY DATA

Date Time Temperature Meter Background
Feb-07-91 09:07AM . 74° F tC) mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
10 Hz Lo 884.72 mG 144.00 mG 715.07 cB
20 Hz 447.44 0G 146.00 mG 9.73 cB
50 Kz 182.91 mG 150.00 mG 71.72 oB
60 Hz 151.96 mG 150.00 mG 70.11 oB
100 Hz 90.66 mG 150.00 mG 4.37 cB
200 Hz 44.49 mG 150.00 mG 10.56 dB
500 Hz 16.10 mG 150.00 mG 19.39 cB
1 kHz 13.86 mG 150.00 mG 20.69 dB
2 khz 32.07 mG 150.00 mG 13.40 dB
5 kkz 115.45 mG 150.00 mG 2.27 dB
00 mG 150.00 mG 78.16 cB

10 kHz 384.

FREQUENCY RESPONCE AUXILIARY DATA

NO AUX TESTS REQUIRED

Appendix x


SI

Percent Error

160 —

140 +

120 4

100 +

80
60
40

20

Accuracy and Linearity - Meter IAl
WRI Model 660 SN G1241

1g?

T rn T TTT TTT T TTT TTT]

! 10 10?
Applied Field (mG)


METER CODE

VIB

MONITOR - MODEL 42b-1 - SN 1287

MANUFACTURER:

Monitor Industries
6112 Fourmile Canyon
Boulder, CO 80302

METER NAME

Mil ligauss

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Frequency Responce Setting - LINEAR MODE

Date

dan: 29-91

RANGE SETTING

1000
1000
250
500
250
100
100
100
100
25
25

BBRRRRR RRS

Time

08:30AM

ACTUAL FIELD

488.65
193.12
194.88

22338828888

Frequency Responce Setting - FLAT MODE

Date

dan-29-91

RANGE SETTING

1000
1000
250
500
250
100
100
100
100
50
50

BBRBRABBRRR

Time

08:30AM

ACTUAL FIELD

B2RRR ARB RR

Temperature
74° F

MEASURED FIELD

500.00
200.00
200.00
150.00
150.00
75.00
50.00
75.00
50.00
20.00
10.00

Temperature
74° F

MEASURED FIELD

500.00
200.00
200.00
150.00
150.00
75.00
50.00
75.00
50.00
20.00
10.00

Appendix VIB

BARRA BB ABBR

Beane aaa RAR

MODEL #
428-1

Meter Background

0.2

% ERROR

2.32%
3.56%
2.63%
4.9%
1.68%
2.22%
2.91%
2.06%
1.43%
3.06%
3.15%

2.73%

Meter Background

0.11

‘% ERROR

2.58%
2.51%
2.52%
3.80%
1.63%
2.32%
3.07%
2.45%
2.52%
3.84%
4.72h

2.91%

mTM%

mG

SERIAL #

1287

AVG % ERROR

AVG % ERROR


WALKER - ELF-50D - SN 71181-204

METER CODE MANUFACTURER METER NAME
XA Walker Scientific Inc. ELF Field
Rockdale St.

Worcester, MA 01606-9988

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Date Time Temperature
Feb-07-91 11:00AM Th? F
RANGE SETTING LO ACTUAL FIELD MEASURED FIELD
2000 mG 502.48 mG 500.00 mG
251.74 mG 269.50 mG
2000 me 153.08 mG 150.00 mG
77.15 mG 75.00 mG
2000 mG 27.68 mG 25.00 mG
11.03 mG 10.00 me

METER ACCURACY AND LINEARITY AT 6OHz - TEST LOCATION REMOTE

Date Time Temperature
Mar-29-91 10:59AM 76° F
RANGE SETTING LO ACTUAL FIELD MEASURED FIELD
103.31 mG 100.00 mG
103.31 mG 100.00 mc
53.41 mG 50.00 mG
53.41 mG 50.00 mG
12.41 mG 10.00 mG
2000 12.41 mG 10.00 mG
5.40 mG 3.00 mG
5.40 mG 3.00 mG

Appendix XA

MODEL #
ELF-50D

SERIAL #

71181-204

Meter Background- mG

0

‘% ERROR

“4.19%

Meter Background

i)

‘% ERROR

+18.35%

mG

AVG % ERROR

AVG % ERROR


EFM - Metex-3800 - SN 542832

MINIMUM DETECTABLE SIGNAL

Date Time Temperature Meter Background
Mar-28-91 02:28PM 72° F min 0.1 mG
max 0 mG

Threshold Current mA = Threshold Field uG

1 2.90 mA 156.46 uG
Trial 2 3.20 mA 172.64 uG
Number 3 3.10 mA 167.25 uG
4 3.20 mA 172.64 uG
5 3.10 mA 167.25 uG
NOTE: Meter readings increase by about 0.1mG 167.25 uG AVG THRESHOLD FOR SENSITIVITY
METER RESOLUTION DATA
Date Time Temperature Meter Background
Mar-28-91 02:33PM 72° F min 70.1 mG
max 0 mG
Meter Range Auto Range - 100mG
Drive Current Start A Threshold Current A Resolution uG Resolution %
1 1.8550 A 1.8610 A 323.69 uG 0.32%
Trial 2 1.8550 A 1.8600 A 269.75 uG 0.27%
Number 3 1.8550 A 1.8600 A 269.75 uG 0.27%
4 1.8550 A 1.8610 A 323.69 uG 0.32%
5 1.8550 A 1.8610 A 323.69 uG 0.32%
NOTE: Meter readings increased about 0.25mG 302.11 uG 0.30%

Appendix XIVB


EXP - ELF - MODEL 1A - SN 1104

METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date Time Temperature

Feb-27-91 12:24PM 78° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.0 kV/M +-5%
E Field OFF Reading(mG) E Field ON Reading(mG)

Meter Range Meter Range
3G 3G
1 0.10 0.10
Trial 2 0.10 0.10
Number 3 0.10 0.10
4 0.10 0.10
5 0.10 0.10
0.10 AVG 0.10

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Date Time Temperature
Feb-20-91 12:05PM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
3G 3mG
1 MHz 0.10 0.10
10 MHz 0.10 0.10
100 MHz 0.10 0.10

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date Time Temperature
Feb-20-91 12:19PM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
3G 3 mG
1 MHz 0.11 0.11
10 MHz 0.11 0.11
100 MHz 0.11 0.19

Appendix I11B


MEDA - Model PLM-100 - SN 1380

METER 60Hz NON-SINUSOIDAL RESPONCE DATA
Date Time Temperature
May-14-91 03:40PM TF

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD
200 mG 166.15 mG 150.00
165.82 mG 149.95

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD
200 me 182.12 mG 150.10
181.92 mG 150.05

SINE WAVE TESTING

RANGE SETTING | ACTUAL FIELD MEASURED FIELD
200 mG 149.83 mG 149.95
150.30 mG 150.00

METER ANGULAR RESPONCE DATA

Date Time Temperature
May-15-91 10:40AM 77° F
Angle ACTUAL FIELD MEASURED FIELD
o 149.98 mG 149.90
10° 152.71 mG 150.05
20° 159.71 mG 149.95
30° 172.88 mG 149.95
40° 195.95 mG 150.05
50° 234.20 mG 150.05
60° 293.77 mG 150.05
70° 413.09 mG 149.95
80° 777.67 mG 150.00
90° 1026.92 mG 15.40

Appendix XVIA

ms
ms

mm
mG

mG
mG

3BRRBBARRB

Meter Background
0.1 mG

‘% ERROR
9.72%
“9.57%
79.64%

‘% ERROR

717.58%

717.52%

17.55%

‘% ERROR
0.08%

70.19%

70.05%

Meter Background
0.4 mG

METER RESPONCE

eececec9e-
QSKLRARASS

RELATIVE RESPONCE
0.88 cB

-0.87 cB

+0.88 db

RELATIVE RESPONCE

71,68 dB

+1.67 cB

+1.68 db

RELATIVE RESPONCE

0.00 dB

-0.01 cB

-0.00 db


821

os MnzOTUMD

Susceptibility - Meter IVB
MSI Model MSI-20 SN 79010854

ee eee

60 Hz
E-FIELD 10kV"4

4 Mz 1@ MHz «6180 Hz
UNHOD

4 MHz

10 HH
GHz MOD

COFIELD OFF
WIFIELD ON


DINDIMA - GAUSSMAUS - SN GM0401
FREQUENCY RESPONCE PRIMARY DATA

Frequency Responce Setting - WIDE BAND

Date Time Temperature Meter Background
May- 17-91 09:06AM 74° F 1 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
5 kz 99 1G 182.63 mG 19.00 mG 719.66 dB
10 Hz 99 mG 182.63 mG 99.00 mG +5.32 dB
20 Hz 99 mG 104.36 mG 99.00 mG 0.45 oB
50 Hz 99 mG 111.81 9G 99.00 mG +1.06 GB
60 Hz 99 1G 110.76 mG 99.00 mG 0.97 4B
100 kz 99 mG 112.58 mG 99.00 mG 1.42 eB
200 Hz 99 mG 113.95 mG 99.00 mG 1,22 dB
500 Hz 99 1G 117.39 9G 99.00 mG “1.68 oB
1 kHz 99 mG 129.06 mG 99.00 mG +2.30 4B
2 kHz 99 mG 173.98 mG 99.00 mG 24.90 dB
5 kHz 99 1G 908.53 mG 99.00 mG 19.25 dB
10 kHz 99 mG 105.56 mG 99.00 mG “0.55 B
Frequency Responce Setting - BAND PASS
Date Time Temperature Meter Background
May- 16-91 08:21AM 74° F 1 7G
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
5S hz 99 mG 1341.29 mG 4.00 mG +50.51 dB
10 Hz 99 mG 1341.29 mG 35.00 mG -31.67 dB
20 Hz 99 mG 1628.12 mG 73.00 mG +25.83 dB
50 Hz 99 mG 437.33 mG 86.00 mG 14.13 dB
60 Hz 99 mG 118.06 mG 99.00 mG 21,53) dB
100 Hz 99 1G 467.66 mG 37.00 mG “21.65 dB
200 Hz 99 0G 406.26 mG 12.00 mG -30.59 dB
500 Hz 99 mG 466.26 mG 3.50 m6 742.49 dB

Appendix  XVB


vel

as mazovumsa

@.4+

ll

Susceptibility ~ Meter VIB - Flat Mode
MONITOR Model 42b-1 SN 1287

68 Kz
E-FIELD 10k

(ii (i ef Ce

4 MHz = 16 MHz = 6180 MHz Hz ‘9 oft ree HHz
UNHOD

ORELD OFF
GFIELD ON


EMD - MODEL ACGM-1 - SN 01-0116

MINIMUM DETECTABLE SIGNAL

Date

Mar-29-91

Trial
Number

wanna

NOTE: Meter reading

Time

12:26PM

increases by 0.1mG

METER RESOLUTION DATA

Date

Mar-29-91

Meter Range

Trial
Number

vnuns

NOTE: Meter reading

Time

12:32PM

Auto Range
Drive Current Start

1.9470.
1.9470
1.9670
1.9470
1.9470

increases by 0.3mG

mTM
mTMm
mTM 156.46
mTM!
mA
mA

Temperature
73° F

Threshold Field
172.64
161.85

151.06
151.06

158.61

Temperature
76° F

A Threshold Current.

1.9510
1.9540
1.9530
1.9560
1.9540

PPrr>

Appendix VIIIA

>

> ere

Meter Background

ms
mG

AVG THRESHOLD FOR SENSITIVITY

Meter Background

Resolution

215.80
377.64
323.69
485.54
377.64

356.06

mG
mG

uG

uG
uG
uG
ut

Resolution %

0.20%
0.36%
0.30%
0.46%
0.36%

0.33%


EXP - ELF - MODEL 1A -SN 1103

METER CODE MANUFACTURER METER NAME

THA Expan Test Inc. ELF Sense
232 St. John St. ,ste.316

Portland, ME 04102

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Date Time Temperature

Jan-31-91 01:00PM 74° F

RANGE SETTING
1000 mG
1000 mG
100 mG
30 mG

ACTUAL FIELD

sayw
BSS
Bas
ue

SBS

115.2:
76.1!

w
we
Ra

20.2%
10.0:

222282888

aN

MEASURED FIELD

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date

Mar-18-91

RANGE SETTING

100 mG

3 mG

Time

12:31PM

ACTUAL FIELD

100.89 mG
100.89

5
2
2

50.2

ww
3s
w

ico an Aah

Dgerenner-e-F-T
eeseaasexeayes
BRRRRRRRBR BBR BB

Temperature
66° F

MEASURED FIELD

1.50

Appendix 111A

BRAABZRRARRRRR ESR

MODEL # SERIAL #

1A 1103

Meter Background- inG
0.2

‘% ERROR

3.96%

Meter Background

0.1

% ERROR
-0.87%

+1,50%

mG

AVG % ERROR

AVG % ERROR


WRI - Model 660 - SN G1242

MINIMUM DETECTABLE SIGNAL

Date Time

Mar-14-91 02:15PM
Threshold Current m
1 2.80 m
Trial 2 2.80 m
Number: 3 2.90 m
4 2.80 mA
5 2.80 mTM”

NOTE: Fairly Stable at 0.19mG

METER RESOLUTION DATA

Date Time
Mar-14-91 02:19PM
Meter Range Auto Range - 100mG
Drive Current Start A
1 0.1570 A
Trial 2 0.1490 A
Number 3 0.1550 A
4 0.1490 A
5 0.1470 A

Temperature

71° F min 0.1
max 0.3

Threshold Field uG

151.06 uG
151.06 uG
156.46 uG
151.06 uG
151.06 uG

Meter Background
mG

ms

152.14 uG AVG THRESHOLD FOR SENSITIVITY

Temperature Meter Background
7° F min 0.1 mG

max 0.3 mG

Threshold Current A Resolution uG
0.1800 A 1240.83 uG

0.1740 A 1348.73 uG

0.1750 A 1078.98 uG

0.1690 A 1078.98 uG

0.1680 A 1132.93 uG

1176.09 uG

Appendix 1A2

Resolution %

14.65%
16.78%
12.90%
13.42%
14.29%

14.41%


‘74

Ss W
Frequency Responce - Meter XIVB rn curtooth wave
EFM Metex-3800 SN 542832 -+ Sine Wave
40 — — Bezier Fit
20 -
o 4
g
8
5 -20 4
oe
2
-40 4
-60 TTT TTT TTT TT TTT TTT TTT TTT]
1 10 10? 107 10

Frequency (Hz)


EMD - MODEL ACGM-1 - SN 01-0117

MINIMUM DETECTABLE SIGNAL

Date Time

Mar-29-91 12:51PM

Temperature
76° F

Threshold Current mA Threshold Field

1 5.80 mA 312.91
Trial 2 6.00 mA 323.70
Number 3 6.20 mA 334.49
4 5.90 ma 318.31
5 6.20 ma 334.49

NOTE: Meter reading increases by 0.1mG

METER RESOLUTION DATA

Date Time
Mar-29-91 12:56PM
Meter Range Auto Range - 100mG

Drive Current Start

1 1.9310

Trial 2 1.9310
Number 3 1.9310
4 1.9310

5 1.9310

NOTE: Meter reading increases by 0.2mG

324.78

Temperature
76° F

A Threshold Current

1.9390
1.9370
1.9360
1.9360
1.9370

> PEe>

Appendix VIIIB

min
max

uG
uG

min
max

P>ere

Meter Background

0.1
0.1

mG
mG

AVG THRESHOLD FOR SENSITIVITY

Meter Background

0.1
0.1

Resolution

431.59
323.69
269.75
269.75
323.69

323.69

mG
mG

uG

uG
uG
uG

Resolution %

0.61%
0.31%
0.25%
0.25%
0.31%

0.31%


2e

Percent Error

10-

Accuracy and Linearity - Meter VIIIB
EMD Model ACGM-1 SN 01-0117

T TTT TTT TY T TTT TTTTYy T TTT TTT}

10 10? 10?
Applied Field (mG)


MONITOR - MODEL 42b-1 - SN 1286
FREQUENCY RESPONCE PRIMARY DATA

Frequency Responce Setting - LINEAR MODE

Date Time Temperature Meter Background
Feb-28-91 09:45AM. 74° F 0.25 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
1 Hz 17 1003.15 mG 0.52 mG -65.71 dB
2 Hz 1 mG 1003.15 mG 0.56 mG *65.06 dB
5 Hz 1 mG 1003.15 mG 0.80 mG -61.97 dB
10 Hz 25 mG 1003.15 mG 8.25 mG +41,70 dB
20 Hz 250 mG 489.07 mG 150.00 mG *10.27 dB
50 Hz 250 mG 179.86 mG 150.00 mG -1.58 dB
60 kz 250 mG 149.33 mG 150.00 mG 0.03 dB
100 Hz 250 mG 89.00 mG 150.00 mG 4.53 dB
200 Hz 250 mG 44.59 mG 150.00 mG 10.54 oB
500 Hz 250 mG 18.21 mG 150.00 mG 18.32 dB
1 kHz 250 mG 9.49 mG 150.00 mG 23.98 dB
2 kHz 250 mG 5.64 mG 150.00 mG 28.50 dB
5 kez 250 mG 4.94 mG 150.00 mG 29.64 dB
10 kHz 250 mG 9.26 mG 150.00 mG 24.19 dB
Frequency Responce Setting - FLAT MODE
Date Time Temperature Meter Background
Feb- 28-91 10:25AM The F 0.12 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
2 Wz 0.5. mG 1001.74 mG 0.28 mG -71.07 dB
5 Hz 1m 1001.74 mG 0.62 mG -64.17 dB
10 Hz : 10 mG 1001.74 mG 4.60 mG -46.76 dB
20 Hz 250 mG 273.99 mG 150.00 mG *5.23 OB
50 Hz 250 mG 150.32 mG 150.00 mG -0.01 GB
60 Hz 250 mG 149.43 mG 150.00 mG 0.03 dB
100 Hz 250 mG 148.36 mG 150.00 mG 0.09 dB
200 Hz 250 mG 148.44 mG 150.00 mG 0.09 dB
500 Hz 250 mG 149.80 mG 150.00 mG 0.01 dB
1 kHz . 250 mG 157.25 mG 150.00 mG 70.41 dB
2 kHz 250 mG 191.18 mG 150.00 mG 72.11 GB
5 kHz 100 mG 143.67 mG 28.00 mG -14.20 dB
10 kHz 2.5 mG 4.00 mG 0.37 mG 20.56 dB
FREQUENCY RESPONCE AUXILIARY DATA
Frequency Responce Setting - LINEAR MODE
Date Time Temperature Meter Background
Feb-28-91 10:21AM Th? F 0.25 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
20 kHz 250 mG 30.35 mG 150.00 mG 13.88 dB
30 kkz 250 mG 71.86 mG 150.00 mG 6.39 dB
40 kHz 250 mG 141.98 mG 150.00 mG 0.47 @B
15 Hz 250 mG 1036.39 mG 125.00 mG +18.37 dB
Frequency Responce Setting - FLAT MODE
Date Time Temperature Meter Background
Feb-28-91 11:29AM 74° F 0.12 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
15 Hz 250 mG 1036.12 mG 150.00 mG +16.79 dB

Appendix VIA


£01

Responce

Angular Responce - Meter VIIIB
EMD Model ACGM-1 SN 01-0117

10 20 30 40 50 60 70

Rotation Angle

8&0

90

-++ Meter Responce
— Cosine Curve


6€

Percent Error

Accuracy and Linearity - Meter XVA - Wide Band

DINDIMA - GAUSSMAUS - SN GM0400
o-

-10 4

AZ

-20 T TTT T1117 T TTT 1717] T TT T1117]
1 10 10? 10?

Applied Field (mG)


METER 6OHz £ FIELD SUSCEPTIBILITY DATA
Date Time
Feb-25-91 02:47PM

EFM Metex

Trial
Number

wnuNne

E Field OFF Reading
0.00

E Field OFF Reading(mG)
Meter Range
Auto
0.00
0.00
0.00

0.00
0.00

0.00

MODEL ACGM-1-- SN 01-0116

Temperature
78° F

E Field ON Reading
10.0 kV/M +-5%

E Field ON Reading(mG)
Meter Range
Auto

0.00
0.00
0.00
0.00
0.00

AVG 0.00

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Date

Feb-22-91

1 MHZ
10 MHz
100 MHz

RADIO FREQUENCY

Date

Feb-22-91

1 MHZ
10 MHz
100 MHz

Time
11:494M
RF OFF Reading(mG)
Meter Range
Auto
0.00

0.00
0.00

ICEPTIBILITY MEASUREMENT DATA - 60 Hz

Time
12:05PM
RF OFF Reading(mG)
Meter Range
Auto
0.00

0.00
0.00

Temperature
79° F
RF ON Reading(mG)
Meter Range
Auto
0.00

0.00
0.30

ATED

Temperature
WF
RF ON Reading(mG)
Meter Range
Auto
0.00

0.00
29.10

Appendix VITIA


MEDA - Model PLM-100 - SN 1381

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date Time Temperature Meter Background

May- 15-91 10:04AM 77° F 0.1 mG

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELO ‘% ERROR RELATIVE RESPONCE
200 mG 166.77 mG 150.00 mG +10.05% -0.92 dB

166.54 mG 150.00 mG 9.93% -0.90 cB
+9. 99% 70.91 db

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR RELATIVE RESPONCE
200 mG 182.61 mG 150.15 mG +17.78% +1.70 dB

182.54 mG 150.15 mG 17.74% “1.70 dB
17.76% +1.70 db

SINE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR RELATIVE RESPONCE
200 mG 150.55 mG 150.20 mG 0.23% +0.02 dB

150.60 mG 149.95 mG 70.42% -0.03 dB
0.33% +0.02 db

METER ANGULAR RESPONCE DATA

Date Time Temperature Meter Background
May- 15-91 10:15AM 7 F 0.1 mG
Angle ACTUAL FIELD MEASURED FIELD METER RESPONCE

o° 150.57 mG 150.00 mG 1.00

10° 153.92 mG 149.95 mG 0.97

20° 160.41 mG 149.95 mG 0.93

30° 176.22 mG 149.95. mG 0.86

40° 198.61 mG 149.95 mG 0.75

50° 240.31 mG 150.05 mG 0.62

60° 314.01 mG 150.05 mG 0.48

70° 468.46 mG 150.05 mG 0.32

80° 922.03 mG 150.00 mG 0.16

90° 1023.82 mG 10.40 mG 0.01

Appendix XVIB


SYDKRAFT - MODEL 88 05 - VIIB

METER 60Hz E FIELO SUSCEPTIBILITY DATA
Date Time

Feb-27-91 09:46AM

E Field OFF Reading

EFM “etex 0.00

E Field OFF Reading(mG)

Meter Range

Trial
Number

wun
eeesse
RRREE

AVG

Temperature
The F

E Field ON Reading
10.0 KV/M +-5%

E Field ON Reading(ma)
Meter Range

mG

eeceo
RRREE

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMOOULATED

Date Time

Feb-25-91 10:05AM
RF OFF Reading(mG)

Meter Range

2mG

1 mHz 0.01
10 mHz 0.01
100 mHz 0.01

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DAT:

Date Time

Feb-25-91 10:05AM

RF OFF Reading(mG)

Meter Range
2m
1 mz 0.01
10 mHz 0.01
100 mHz 0.01

Temperature
78° F
RF ON Reading(mG)
Meter Range
2mG
0.01

0.01
0.01

60H. ATED

Temperature
78° F
RF ON Reading(mG)
Meter Range
2mG
0.01

0.01
0.01

Appendix VIIB


WRI - Model 670 - SN G1259
FREQUENCY RESPONCE PRIMARY DATA

Date Time Temperature Meter Background
Feb-07-91 03:02PM 77° F 2 mG

FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

1H. 100*1 mG 882.02 mG 1.50 mG “55.39 dB

2 Hz 100*1 mG 882.02 mG 2.30 mG +51.67 oB

5 Hz 100*1 mG 882.02 mG 21.00 mG 32.47 GB

10 Hz 100*1 mG 882.02 mG 100.00 mG 718.91 dB

20 Hz 100*10 mG 550.20 mG 200.00 mc 8.79 dB

50 Hz 100*10 mG 174.99 mG 200.00 mG 1.16 cB

60 Hz 100*10 mG 144.01 mG 200.00 mG 2.85 dB

100 Hz 100*10 mG 86.02 mG 200.00 mG 7.33 dB

200 Hz 100*10 mG 48.41 mG 200.00 mG 12.32 dB

500 kz 100*10 mG 28.59 mG 200.00 m6 16.90 dB

1 kHz 100*10 mG 25.86 mG 200.00 mG 17.77 oB

2 kHz 100*10 mG 33.73 mG 200.00 mG 15.46 dB

S kkz 100*10 mG 65.05 mG 200.00 mG 9.76 OB

10 kHz 100*10 me 128.44 mG 200.00 mG 3.85 dB

FREQUENCY RESPONCE AUXILIARY DATA

WO AUX TESTS

Appendix 12


DINDIMA - GAUSSMAUS - SN GMO401

METER 6OHz NON-SINUSOIDAL RESPONCE DATA
Frequency Responce Setting - WIDE BAND
Date Time
May-17-91 09:37AM

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD
99 mG 110.93 mG
110.86 mG

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD
99 mG 73.03 mG
74.40 mG

SINE WAVE TESTING

RANGE SETTING ACTUAL FIELD
99 mG 109.17 mG
109.95 mG

Frequency Responce Setting - BAND PASS
Date Time
May- 16-91 10:28AM

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD
99 mG 113.67 mG
120.94 mG

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD
99 mG 127.30 mG
122.00 me

SINE WAVE TESTING

RANGE SETTING ACTUAL FIELD
99 mG 110.26 mG
104.82 mG

Temperature
The F

MEASURED FIELD

99.00 mG
99.00 mG

MEASURED FIELD

99.00 mG
99.00 mG

MEASURED FIELD

99.00 mG
99.00 mG

Temperature
76° F

MEASURED FIELD

99.00 mG
99.00 mG

MEASURED FIELD

99.00 mG
99.00 mG

MEASURED FIELD

99.00 mG
99.00 mG

Appendix xVB

Meter Background

1

% ERROR
+10.76%
+10.70%
+10.73%
‘% ERROR
35.56%
33.07%
34.31%
‘% ERROR
+9.32%

9.96%

9.96%

mG

RELATIVE RESPONCE
-0.98 dB
-0.98 dB
-0.98 db
RELATIVE RESPONCE
2.64 dB
2.48 dB
2.56 db
RELATIVE RESPONCE
0.85 dB

0.91 dB

-0.88 db

Meter Background

1

‘% ERROR
12.91%

718.14%
+15.52%
‘% ERROR
22.23%
-18.85%
20.54%
‘% ERROR
710.22%

5.55%

+5.55%

ms

RELATIVE RESPONCE

RELATIVE RESPONCE
2.18 dB
71.81 dB
72.00 db

RELATIVE RESPONCE
0.93 dB
0.49 dB

-0.71 db


+9

40 -

20

-20 +

Responce (dB)

-40 4

Frequency R

- - (1 Square Wave
esponce - Meter VIB - Flat Mode A Sautooth Wave
MONITOR Model 42b-1 SN 1287 + Sine Wave

— Bezier Fit

10

TT TTT TTT TTT TTT TTT}

10? 107 10¢
Frequency (Hz)


HOLADAY - HI 3604 - SN 76112

MINIMUM DETECTABI MA

Date Time

Mar-28-91 11:23AM

Threshold Current ma

Trial
Wumber

wnune
e0000

NOTE: Meter readings increased by 0.002mG

METER Ri ATA
Date Time
Mar-28-91 11:27AM
Meter Range Auto Range - 100mG
Drive Current Start A
1 1.8660 A
Trial 2 1.8660 A
Number 3 1.8660 A
4 1.8660 A
5 1.8660 A

NOTE: Meter readings increased by 0.5mG

Temperature
70° F

Threshold Field

15.65
12.41
10.25

7.01
15.11

12.08

Temperature
70° F

Threshold Current

1.8720
1.8720
1,8720
1.8740
1.8740

Appendix —IXB

min
max

685555 &

min
max

>> >>e >

Meter Background

0.134
0.136

mG
mG

AVG THRESHOLD FOR SENSITIVITY

Meter Background

0.134
0.136

Resolution

323.69
323.69
323.69
431.59
431.59

366.85

mG
mm

uG

85565

&

Resolution %

0.32%
0.32%
0.32%
0.42%
0.42%

0.36%


MSI - MODEL MSI 20 - SN 79011626
RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Date Time Temperature
Feb-21-91 12:07PM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
ALL ALL
1 MHz 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 0.00

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date Time

Temperature
Feb-21-91 12:26PM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range a

ALL ALL
1 MHZ 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 0.00

NOTE: Meter tested without extension cable.

With cable attached readings changed
by as much as 20mG just by moving cable

‘Appendix IVA


DINDIMA - GAUSSMAUS - SN GM0401

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Frequency Responce Setting - WIDE BAND

Date Time Temperature
May-17-91 14:47AM 7 F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
9 ”
1 MHz 1.00 1.00
10 MHz 1.00 1.00
100 MHz 1.00 1.00

Frequency Responce Setting - BAND PASS

Date Time Temperature
May-17-91 11:59AM Tre
RF OFF Reading(mG) RF ON Reading(ma)
Meter Range Meter Range
9 9
1 MHz 1.00 1.00
10 MHz 1.00 1.00
100 MHz 1.00 1.00

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Frequency Responce Setting - WIDE BAND

Date Time Temperature
May-17-91 11:57AM 77° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
9 9
1 MHZ 1.00 1.00
10 MHz 1.00 1.00
100 MHz 1,00 1.00

Frequency Responce Setting - BAND PASS

Date Time Temperature

May- 17-91 11:59AM 7? F

RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
9 9
1 Mz 1.00 1.00
10 MHz 1.00 1.00
100 MHz 1.00

1.00
Appendix XVB


8b

on mazovama

1.2

0.4.

|

Susceptibility - Meter XVB - Band Pass

DINDIMA GAUSSMAUS SN GM0401

60 Hz
E-FIELD 10kV At

1 Maz

10 MHz «6160 Miz
‘UNMOD

18 MHz
GeHz MOD

OFELD OFF
HFELDON


Se

Percent Error

10-

-10 4

-20 +

-30 4

-40 4

Accuracy and Linearity - Meter XA
WALKER ELF-S0D SN 71181-204

T T1117 T TT T1117 T TT T1117]

10 10? 107
Applied Field (mG)


19

Responce (dB)

_ (Square Wave
Frequency Responce Meter VIIIA A. Saxtooth Wave
EMD Model ACGM-1 SN 01-0116 + Sine Wave
40 - — Bezier Fit
20 4
o-
-20 -|
-40 4
-60 T T rrrrny T T TUT TTTTy T T TUTTI T T TUTTTT)
1 10 10? 107 10

Frequency (Hz)


EXP - ELF - MODEL 1A - SN 1103

FREQUENCY RESPONCE PRIMARY DATA

Date Time
Jan-31-91 01:50PM
FREQUENCY RANGE SETTING
2 Hz 3
5 Hz 3
10 Hz 3
20 Hz 100
50 Hz 1000
60 Hz 1000
100 Hz 1000
200 Hz 1000
500 Hz 1000
1 kHz 1000
2 kkz 1000
5 kHz 1000
10 kkz 1000

FREQUENCY RESPONCE AUXILIARY DATA

Date Time
Jan-31-91 02:30PM

FREQUENCY RANGE SETTING
80 Hz 1000
150 Hz 1000
40 Hz 1000
30 Hz 1000
60 Hz 1000
300 Kz 1000
25 Hz 1000
35 Hz 1000

BRABRBAARBRBA

BaBaRRRR

Temperature

The F

ACTUAL FIELD

Temperature

74° F

ACTUAL FIELD

Appendix

TIA

Meter Background

0.2

m

MEASURED FIELD

0.20
0.40
1.20
30.00
150.00
150.00
150.00
150.00
*50.00
150.00
150.00
450.00
125.00

Meter Background

0.2

Lis

MEASURED FIELD

150.00
150.00
150.00
150.00
150.00
150.00

50.00
150.00

BaBABBAR

BARRRRRRRRRRS

RELATIVE

RELATIVE

RESPONCE

+72.66
766.64
-57.10
730.63
70.24
0.28
0.39
0.16
70.18

dB
dB
cB
dB

SSSSS3888S

RESPONCE

0.16
0.32
70.49
12.56
0.00
0.01
21.61
+4.79

dB
dB
BB
dB
dB
dB
Bo
dB


DINDIMA - GAUSSMAUS - SN GMO401

FREQUENCY RESPONCE AUXILIARY DATA

Frequency Responce Setting - WIDE BAND
NO AUX TESTS REQ.

Frequency Responce Setting - BAND PASS

Date Time Temperature
May-16-91 10:23AM 76° F
FREQUENCY RANGE SETTING ACTUAL FIELD
40 kz 99 mG 383.52
30 Hz 99 mG 500.60
70 Hz 99 mG 258.32
80 kz 99 mG 372.42
71 Hz 99 mG 342.03
51 Hz 99 mG 410.28
52 Hz 99 mG 310.94
53 Hz 99 mG 199.56
34 Hz 99 mG 161,02
55 Hz 99 mG 155.12
56 Hz 99 1G 133.90
57 Wz 99 mG 123.89
58 Hz 99 mG 123.82
59 Hz 99 mG 116.20
60 Hz 99 1G 118.06
61 Hz 99 1G 113.11
62 Hz 99 mG 115.04
63 Hz 99 mG 114.06
64 Hz 99 mG 113.99
65 Hz 99 mG 120.73
66 Hz 99 mG 126.91
67 Hz 99 mG 136.43
68 Hz 99 mG 149.18
69 Hz 99 mG 180.62
Appendix XVB

BARBBABRBRRARRRAABRRRARBBRBRA

Meter Background

1

MEASURED FIELD RELATIVE

45.00
29.00
99.00
63.00
95.00
93.00
99.00
99.00
99.00
99.00
99.00
99.00
99.00
99.00
99.00
99.00
99.00
99.00
99.00
99.00
99.00
99.00
99.00
99.00

mm

mG

BRRBARRAAABRARRRBRABBBRBRR

RESPONCE

18.61
+24.76
8.33
+15.43
711.13
712.89
9.96
6.09
4.23
+3.90
72.62
71.95
71.96
1.39
“153,
71.16
71,30
“1.23
1.22
4,72
+2.16
72.79
73.56
75.22

SSSSSSSSSSS8SSRR8R8


EXP - ELF - MODEL 1A - SN 1104

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date Time Temperature Meter Background

Feb-05-91 11:35AM The F 0.1 mG

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR RELATIVE RESPONCE
1000 mG 198.73 mG 200.00 mG 0.63% 0.05 dB

198.81 mG 200.00 mG 0.60% 0.05 dB
9.61% 0.05 db

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR RELATIVE RESPONCE
1000 mc 199.63 mG 200.00 mG 0.18% 0.01 dB

201.99 mG 200.00 mG 70.98% 0.08 dB
70.39% +0.03 db

SINE WAVE TESTING

RANGE SETTING ‘ACTUAL FIELD MEASURED FIELD "% ERROR RELATIVE RESPONCE
1000 me 200.05 m6 200.00 mc -0.02%
202.29 mG 200.00 me “1.13%
+0.57% +0.05 db
METER ANGULAR RESPONCE DATA
Date Time Temperature Meter Background
Feb-05-91 12:10PM Th? F 0.1 1G
Angle ACTUAL FIELD MEASURED FIELD ANGULAR RESPONCE
o° 147.09 mG 150.00 mG 1.00
10° 149.95 mG 150.00 me 0.98
20° 154.84 mG 150.00 me 0.95
30° 168.11 mG 150.00 mG 0.87
40° 192.50 mG 150.00 mG 0.76
50° 226.79 mG 150.00 mG 0.65
60° 285.89 mG 150.00 mG 0.51
70° 412.49 mG 150.00 mG 0.35
80° 789.29 mG 150.00 mG 0.18
90° + 1311.33 mG 25.00 mG 0.01

Appendix ITI


MEDA - Model PLM-100 - SN 1380

MINIMUM DETECTABLE SIGNAL

Date Time

Aug-29-91 12:14PM

Threshold Current mA

1 0.52 mA
Trial 2 0.46 ma
Number: 3 0.48 mA
4 0.46 ma
5 0.46 ma

METER RESOLUTION DATA

Date Time
Aug-29-91 12:19PM
Meter Range 20 mG

Drive Current Start A

1 0.1857 A

Trial 2 0.1857 A
Number 3 0.1857 A
4 0.1857 A

5 0.1857 A

Temperature
97° F

Threshold Field
28.05

25.68

Temperature
97° F

Threshold Current

0.18612
0.18594
0.18614
0.18590
0.18615

Appendix XVIA

min

>

>Pr>>

Meter Background

0
0.01

mm
me

AVG THRESHOLD FOR SENSITIVITY

Meter Background

0
0.01

Resol

22.66
12.95
23.76
10.79
24.28

18.88

mG
mG

uc

65585

Resolution %

0.22%
0.12%
0.23%
0.10%
0.24%

0.18%


+01

Responce

Lo
ae
a6 5
a4

a2 -

a0-

10

Angular Responce - Meter IXA
HOLADAY HI-3604 SN 76111

20 30 40 50 60

Rotation Angle

70

&0

90

++ Meter Responce

— Cosine Curve


WRI - Model 670 - SN G1259

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMOOULATED

Date

Feb-21-91

1 MKz
10 MHz
100 MHz

1 MHz
10 MHz
100 MHz

FREQUENCY.

Date

Feb-21-91

1 MHz
10 MHz
100 MHz

1 MHZ
10 MHz
100 MHz

Time Temperature

09:11AM 80° F

RF OFF Reading(mG) RF ON Reading(mG)

Meter Range Meter Range
100*1 100*1
2.10 2.10
2.10 2.10
2.10 2.10

Meter Range Meter Range
100*10 100*10
1.80 1.80
1.80 1.80
1.80 1.40

PTIBILITY MI MENT DATA - ULATE!
Time Temperature
09:41AM

RF OFF Reading(mG) RF ON Reading(mG)

Meter Range Meter Range
100*1 100*1
0.22 0.22
0.22 0.22
0.22 3.15

RF OFF Reading(mG) RF ON Reading(mG)

Meter Range Meter Range
100*10 100*10
2.00 2.00
2,00 2.00
2.00 4.96

Appendix 12


92

OB MazZoTUMD

s

ll

Susceptibility - Meter IIIB
EXPAN Model 1A SN 1104

rn ae wn

60 Hz
E-FIELD 10kVAt

4 Miz 18 Miz 400 HAE Wz 19 HH

CORIELD OFF
FIELD ON


EMD - MODEL ACGM-1 - SW 01-0116

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date

Feb-08-91

SQUARE

RANGE SETTING

FULL mG

‘SAWTOOT!

RANGE SETTING

FULL mG

SINE

RANGE SETTING

FULL mG

METER ANGULAR RESPONCE DATA

Date

Feb-08-91

Angle

0°
10°
20°
30°
40°
50°
60°
70°
80°
90°

Time

12:44PM

WAVE TESTING

ACTUAL FIELD

300.12 mG
300.27 mG

H WAVE TESTING

ACTUAL FIELD
297.09 mG

297.19 mG

WAVE TESTING

ACTUAL FIELD

264.43 mG
264.43 mG

Time

12:56PM

ACTUAL FIELD

3Bge
SSESE

358.0!

Bass
BRRRBRRBRA

eur
3885

1016.67

Temperature
76° F

MEASURED FIELD

250.00 mG
250.30 mG

MEASURED FIELD

251.00 mG
251.30 mG

MEASURED FIELD

250.00 mG
250.00 mG

Temperature
76° F

MEASURED FIELD

250.00 mG
250.00 mG
250.00 mG
250.10 mG
250.20 mG
250.00 mG
250.00 mG
250.00 mG
108.90 mG
49.40 mG

Appendix VIIIA

Meter Background

0

‘% ERROR

+16.70%
716.64%

+16.67%

‘% ERROR

+15.51%
+15.44%

715.48%

‘% ERROR

75.46%
46%

75.46%

Meter Background

0

ANGULAR RESPONCE

S2ResuRess

eccoeoo0o-

RELATIVE RESPONCE

“1.59 cB
+1.58 dB

+1.58 db

RELATIVE RESPONCE

+1.46 OB
+146 dB

*1.46 db

RELATIVE RESPONCE

-0.48 dB
70.48 dB

°0.48 db


St

2 -

10 +

is)
!

Percent Error
ru
cS)
i

~20 +

Accuracy and Linearity - Meter XVIIA
MAG TEC * Model ELF 12 SN 2

T T TT TT TTT T T TTT TT]

10 10?
Applied Field (mG)


18

Responce

Angular Responce - Meter IA2

WRI Model 660 SN G1242
10 - '
+
a8
a6 -
a4
42-5
a0 Pd RA CE |
oa 10° 20 30 40 50 60 70 80 90

Rotation Angle

+

Meter Responce

Cosine Curve


HELMHOLTZ COIL SYSTEM
REMOTE: LOW LEVEL TESTING

HP 33254
SYNTHESIZER

FLUKE. 51014
CALIBRATOR
(2 amps maximum)

FLUKE . HELMHOLTZ COIL
8060A . : 0.5 m RADUIS
DIGITAL 3 TURNS. _
CURRENT . B=I (53.95 milligauss/amp)
METER

100 FEET

10 AWG

CABLE


Project Description

Recently, exposure to magnetic fields related to electrical power
equipment has been implicated as a risk to the public. Many commercially
available instruments may be used by the public to survey these fields in the
environment. This testing effort was designed to determine the
characteristics of these commercially available magnetic field survey meters.

This test series is part of an overall EPA effort to evaluate EMF
instrumentation, establish standard measurement -protocols, and work toward
mitigation strategies. This project was performed in cooperation with the
IEEE Magnetic Fields Task Force for the purpose of evaluating the
performance of readily available single-axis, coil-type power frequency
magnetic field meters. The project was centered at the EPA Electromagnetic
Fields Laboratory in Las Vegas which developed the test systems used in this
evaluation.

The test results show a broad range of meter characteristics and should
be helpful for evaluation of a meter for various different applications.
This report includes graphic representations of the results and has raw
numerical data listed in the appendix.



HOLADAY - HI 3604 - SN 76112

METER _60Hz NON-SINUSOIDAL RESPONCE DATA

Date

Feb-14-91

RANGE SETTING

200 ms

RANGE SETTING

200 mG

RANGE SETTING

200 mG

Time
01:26PM
SQUARE WAVE TESTING
ACTUAL FIELD

151.29 mG
151.22 mG

SAWTOOTH WAVE TESTING
ACTUAL FIELD

153.15 mG
153.18 mG

SINE WAVE TESTING

ACTUAL FIELD

149.13 mG

149.18 mG °

METER ANGULAR RESPONCE. DATA

Date

Feb-14-91

Angle

0°
10°
20°
30°
40°
50°
60°
70°
80°
90°

Time

01:39PH

ACTUAL FIELD

s
38
BRBRRRBRRR

Temperature
76° F

MEASURED FIELD

150.10 mG
150.05 mG

MEASURED FIELD

149.95 mG
150.05 mG

MEASURED FIELD

149.95 mG
150.00 mG

Temperature

76° F

MEASURED FIELD

150.00
18.00

g
a
BaaaaR RS

“Appendix 1XB

Meter Background
0.166 mG

‘% ERROR

0.78%
-0.77%

0.78%

% ERROR

+2.09%
72.06%

72.07%

‘% ERROR

0.54%
0.54%

0.54%

Meter Background

0.172 mG

ANGULAR RESPONCE

1.00
98

0.
0.9
0.8;
0.7%
0.6
0.4
0.3
0.1
0.0

Seasaasa

RELATIVE RESPONCE

+0.06 dB
70:06 dB

70.06 db

RELATIVE RESPONCE

0.18 dB
70.17 dB

70.18 db

RELATIVE RESPONCE

0.04 db


OST

On mozorumD

0.44

Susceptibility - Meter XVIB
MEDA Model PLM-100 SN 1381

OL

Ha |

frie 10m 100 MHz
E-FIELD ‘TORU At

4 MHz = 10 MHz
Genz MOD

Id

OOFIELD OFF
FELD ON


EFM - Metex-3800 - SN 542832
FREQUENCY RESPONCE PRIMARY DATA

Date Time Temperature Meter Background
May-15-91 01:16PM 77° F 0.1 mG

FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

2 Hz 200 mG 712.99 mG 20.20 mG +30.95 dB

5 Hz 200 mG 712.99 mG 90.60 mG +17.92 cB

10 Hz 200 mG 712.99 mG 151.50 mG 713.45 dB

20 Hz 200 mG 378.84 mG 150.40 mG +8.02 dB

50 Wz 200 mG 171.01 mG 150.05 mG 1.14 GB

60 Hz 200 mG 149.88 mG 149.95 mG 0.00 dB

100 Kz 200 mG 112.20 mG 149.95 mG 2.52 dB

200 Hz 200 mG 93.32 mG 150.05 mG 4.13 dB

500 Hz 200 mG 97.09 mG 150.05 mG 3.78 dB

1 kHz 200 mG 127.22 mG 150.05 mG 1.43 GB

2 kHz . 200 mG 211.95 mG 150.05 mG +3.00 dB

5 kHz 200 mG 531.40 mG 150.05 mG -10.98 dB

10 kHz 200 mG 553.95 mG 71.30 mG -17.81 GB

FREQUENCY RESPONCE AUXILIARY DATA

Date Time Temperature Meter Background
May- 15-91 01:35PM 77° F 0.1 mG

FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

80 Hz 200 me 125.31 mG 149.95 mG 1.56 dB

150 Hz 200 mG 98.14 mG 150.00 mG 3.69 oB

300 Hz 200 mG 91.63 mG 150.00 mG 4.28 dB

Appendix XIVB


- ey

-10 4

Percent Error

-{5 4

Accuracy and Linearity - Meter XVB - Band Pass

DINDIMA - GAUSSMAUS - SN GMO401

T TT TITTY T TT TTT TTT T TT T1TT37)

10 10? 107
Applied Field (mG)


WRI - Model 670 - SN G1257

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Date Time Temperature
Feb-20-91 02:55PM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
100*1 100*1
1 MHZ 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 0.00
Meter Range Meter Range
100*10 100*10
4 MHz 0.00 0.00
10 MHz 0,00 0.00
100 MHz 0.00 0.00

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date Time Temperature
Feb-20-91 02:55PM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
100"1 40081
1 Miz 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 3.00
Meter Range Meter Range
100*10 100*10
1 MHZ 0.00 0.00
10 Hz 0.00 0.00
100 MHz 0.00 3.00

Appendix 11


Responce . (dB)

20 -

-20 4

-40 4

- (Square Wave

Frequency Responce - Meter XIVA T_ Sastooth Wave
EFM Metex-3800 SN 542513 + Sine Wave

— Bezier Fit

TT TTT re a TTT TTT TTT),

10 10? 107 104

Frequency (Hz)


28

Responce (dB)

Frequency Responce - Meter XVIIB 1 Square Wave
A Saitooth Wave
MAG TEC Model ELF 12 SN 4 + Sine Wave
40 — — Bezier Fit
20 4
A
i_—
-20 5
-40 4
$60} ep a
1 10 10? 10? 104

Frequency (Hz)


United ata tes Air and Radiation sor bees ote
Enviro ntal Protectio (NAREL)

Laboratory Testing of
Commercially Available Power

Frequency Magnetic Field
Survey Meters

Final Report


801

Responce

Ae

Las

a6
a4-

a2

Angular Responce - Meter XIVA
EFM - Metex-3800 - SN 542513

40 RRA AT I

Oo

10

20

30

4050
Rotation Angle

60

70

80

90

+

Meter Responce
Cosine Curve


SAFE - SAFE PRO METER - SN 1263

METER 60Hz E FIELD SI PTIBILITY DATA

Date Time Temperature

Feb-27-91 12:50PM 78° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 : 10.0 kV/M +-5%
E Field OFF Reading(mG) E Field ON Reading(m

Meter Range Meter Range
Auto Auto
1 0.25 0.80
Trial 2 0.25 0.80
Number 3 0.20 0.75
4 0.20 0.55
5 0.20 0.75
0.22 AVG 0.73

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Date Time Temperature
Feb-21-91 10:22AM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Auto Auto
1 MHz 0.20 . 0.20
10 MHZ 0.20 0.20
100 MHz 0.20 . 0.40

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date Time Temperature
Feb-21-91 10:22AM 80° F
RF OFF Reading(mG) RF ON Reading(me)
Meter Range Meter Range
Auto ‘Auto
1 MHZ 0.20 0.20
10 MHz 0.20 0.30
100 MHz 0.20 0.40

Appendix 111


T€1

OB mazovuma

44

ee

6.24

Susceptibility - Meter VIA - Linear Mode
MONITOR Model 42b-1 SN 1286

MALT

68 Hz
E-FIELD 10k¥ 1

4 MHz = = 10 MHz «188 MHz 1 HHz = 610 KHz 480 MHz
UNMOD Genz MOD

OORELD OFF
WFIELDON


WRI - Model 670 - SN G1257

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date

Time

Feb-08-91 09:18AM

‘SQUARE

RANGE SETTING
1000 mG

WAVE TESTING

ACTUAL FIELD

154.72 mG
157.77 mG

SAWTOOTH WAVE TESTING

RANGE SETTING
1000 mG

ACTUAL FIELD

102.33 mG
101.76 mG

SINE WAVE TESTING

RANGE SETTING
1000 mG

METER ANGULAR RESPONCE DATA

Date

Feb-08-91

Angle

0°
10°
20°
30°
40°
50°
60°
70°
B0°
90°

ACTUAL FIELO

174.34 mG
177.55 mG

Time

09:26AM

ACTUAL FIELO

174.34 mG
178.44
189.62
203.58
229.11
274.24
335.77

BABBBBBASR

Temperature

74° F

MEASURED FIELD

200.00
200.00

MEASURED FIELD

200.00
200.00

MEASURED FIELD

200.00
200.00

Temperature
76° F

MEASURED FIELD

200.00
200.00
200.00
200.00
200.00
200.00
200.00
200.00
200.00

20.00

Appendix 11

BRRRRR BBR

Meter Background

0.5 mG

‘% ERROR

29.27%
26.76%

28.01%

% ERROR

95.44%
96.53%

95.99%

‘% ERROR

14.72%
12.65%

13.68%

Meter Background
0.5 mG

ANGULAR RESPONCE

RELATIVE RESPONCE

2.23 dB
2.06 dB

2.14 db

RELATIVE RESPONCE

5.82 dB
5.87 dB

5.84 db

RELATIVE RESPONCE

1.19 dB
1.03 dB

4.11 db


Table of Contents

Project Description... 2... .... ee ee ee 1
Project Organization .. 2... 2... Ce ee ee ee 2
Quality Assurance 2... ee 3
Sampling Procedures... 2... .......00. 0004 Ne ee eee 4
Sample Custody 2... . ee es 4
Test Procedures . . . Se 6
Internal Quality Control Checks... 2... 0... ee ee ee eee 11
Meter 60 Hz Accuracy and Linearity ....... bem ee we tw es 12
Meter Frequency Response Measurements... ...........2.0004 47
Meter 60Hz Non-Sinusoidal Response Measurements. .............. 47
Meter Angular Response Measurements... .............0000- 83
Meter 60Hz E-Field Susceptibility Measurements Be ee 118
Minimum Detectable Signal and Resolution Measurements ............ 153
SUMMARY © 2. eee 155


EXP - ELF - MODEL 1A - SN 1103

METER 60Hz NON-SINUSOIDAL RESPONCE DATA
Date Time

Feb-01-91 09:00AM

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD

1000 mG 207.40 mG

207.28 mG

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD

1000 mG 208.77 mG

206.23 mG

SINE WAVE TESTING

RANGE SETTING ACTUAL FIELO

1000 mG 202.14 me

208.20 ma

METER ANGULAR RESPONCE DATA

Date Time
Feb-01-91 09:25AM
Angle ACTUAL FIELD

o° 156.09 mG

10° 158.12 mG

20° 159.84 mG

30° 175.34 mG

40° 198.14 mG

50° 227.52 mG

60° 299.43 mG

70° 417.66 mG

80° 764.16 mG
1380.77 m6

90°

Temperature Meter Background
74° F 0.2 mG
MEASURED FIELD ‘% ERROR
200.00 mG 3.57%
200.00 mG 3.51%
+3.54%
MEASURED FIELD ‘% ERROR
200.00 mc 4.20%
200.00 mG 3.02%
+3.61%
MEASURED FIELD ‘% ERROR
200.00 mG +1.06%
200.00 ma 3.94%
+2.50%
Temperature Meter Background
74° F 0.2 mG

MEASURED FIELD ANGULAR RESPONCE

150.00 mG 1.00
150.00 mG 0.98
150.00 mG 0.97
150.00 mG 0.89
150.00 mG 0.78
150.00 m6 0.68
150.00 mG 0.52
150.00 mG 0.37
150.00 mG 0.20
25.00 mG 0.01
Appendix ILIA

RELATIVE RESPONCE

0.31 GB
0.31 oB

70.31 db

RELATIVE RESPONCE

0.37 GB
70.26 cB

+0.32 db

RELATIVE RESPONCE

70.09 dB
+0.34 dB

+0.22 db


METER COOE

VA

CURACY AND.

Date
Feb-11-91

RANGE SETTING
2000 mG
200 mG
20 mG

IRC - MODEL IRC 109

MANUFACTURER

Integrity Research Corp.
266 Elmwood Ave., ste.489
Buffalo, NY 14222

METER NAME
60 Hz H Field

NEARITY AT 60Hz - TEST LOCATION LAB

Time

12:40PM

ACTUAL FIELD

503.73 mG
251.59 mG
151.94
50.49

5.04

mG
mG
+04 mG
10.00 mG

Temperature
76° F

MEASURED FIELD

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date

Mar-21-91

RANGE SETTING
200 mG
20 mG
2G

Time

01:10PM

ACTUAL FIELD

101.43 mG
101.43
10.07
10.04

mm
mi
mG
1.00 mG

Temperature
51° F

MEASURED FIELD

100.00 mG
100.00 mG
10.00 mG
10.00 mG

1.00 mG

Appendix VA

MODEL # SERIAL #

TRC-109

Meter Background- mG
0.2

% ERROR

70.74%
70.63%
71.28%
0.96%
70.25%
0.10%

0.66% AVG % ERROR

Meter Background

0.072 mG

‘% ERROR

“1.01%
“1.41%
70.71%
0.39%
70.34%

0.85% AVG % ERROR


601

Responce

10

Angular Responce - Meter XIVB
EFM - Metex-3800 - SN 542832

i

20 30 40 50 60 70

Rotation Angle

80

90

-++ Meter Responce
— Cosine Curve


0€

Accuracy and Linearity - Meter VIIB
SYDKRAFT Model 88-05

10
54
| + t+
fal o-
uw
»
§
oc
a
5 4
-10 eels A
10” 1 10 10? 10?

Applied Field (m6)


IRC - MODEL IRC 109

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date Time

Feb-12-91 09:12AM

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD
. 200 mG 166.17 mG
166.10 mG

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD
200 mG 185.52 mG
185.69 mG

SINE WAVE TESTING

RANGE SETTING ACTUAL FIELD
200 mG 151.74 mG
151.81 mG

METER ANGULAR RESPONCE DATA

Date Time
Feb-12-91 09:32AM
Angle ACTUAL FIELD
o° 151.66 mG
10° 154.07 mG
20° 161.38 mG
30° 177.25 mG
40° 200.20 mG
50° 245.38 mG
60° 311.70 mG
70° 480.97 mG
80° 1006.95 mG
90° 1076.45 mG

Temperature
76° F

MEASURED FIELD

150.00
150.00

MEASURED FIELD

150.00
150.00

MEASURED FIELD

150.00
150.00

Temperature
76° F

MEASURED FIELD

150.00
150.00
150.00
150.00
150.00
150.00
150.00
150.00
146.80

24.00

Appendix VA

BaRaR ARBAB

Meter Background
0.2 mG

‘% ERROR

9.73%
9.69%

-9.71%

‘% ERROR

+19.15%
19.22%

+ 19,18%

‘% ERROR

71.15%
“1.19%

“1.17%

Meter Background
0.13 mG

ANGULAR RESPONCE

RELATIVE RESPONCE

0.88 dB
88 dB

0.88 db

RELATIVE RESPONCE

“1.85 dB
71.85 cB

+1.85 db

RELATIVE RESPONCE

-0.10 dB
-0.10 dB

-0.10 db


68

10

Angular Responce - Meter I12
SAFE PRO METER SN. 1264

20 30 40 350 60
Rotation Angle

70

80

-++ Meter Responce
— Cosine Curve


MSI - MODEL MSI 20 - SN 79011626

METER CODE MANUFACTURER METER NAME
IVA Magnetic Sciences Int'l MSI, MM

2425 B Channing Way ste. 489
Berkley, CA 94704

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Date Time Temperature

Jan-25-91 09:13AM Th? F

RANGE SETTING ACTUAL FIELD MEASURED FIELD

2000 mG 510.36 mG 500.00

205.79 mG 200.00

200 mG 153.30 mG 150.00

102.38 mG 100.00

200 mG 76.50 mG 75.00

51.10 mG 50.00

200 mG 20.56 mG 20.00

10.36 mG 10.00
METER At IND _LINEARITY AT 60Hz + TEST LOCATION REMOT!

Date Time Temperature

Mar-18-91 02:12PM 66° F

RANGE SETTING ACTUAL FIELO MEASURED FIELD

103.58 mG 100.00

200 mG 103.58 mG 100.00

20.93 mG 20.00

20.93 mG 20.00

10.52 mG 10.00

10.52 mG 10.00

2.21 mG 2.00

2.21 mG 2.00

1.16 mG 1.00

6415 mG 1.00

aaRRRRRR

BaRaRRARRR

MODEL # SERIAL #
MSI-20 79011626

Meter Background: mG

72.46% AVG % ERROR

Meter Background

0 mG

7.18% AVG % ERROR


+8

Responce

“10

Angular Responce - Meter [1
WRI Model 670 SN G1257_

20 30 40 50 60

Rotation Angle

70

80

90

-++ Meter Responce
— Cosine Curve


Meter 60 Hz Accuracy and Linearity

The meters were tested for absolute accuracy in a 60 Hz magnetic field.
Two data points were recorded in each amplitude range available on the test
meter. To perform the test the Helmholtz coil was configured for 60 Hz
operation. Since the test meters vary in design, the exact amplitude
settings were determined at test time. Additionally two different
experimental setups were used in two different locations to complete the full
range of accuracy. and linearity testing. Low level testing.was performed at
a remote testing location where the ambient magnetic fields are minimized.
High level (background magnetic fields less than 1% of the applied field)
were performed in the Jab. The remote testing was performed with the
Helmholtz coil energized. by a calibrated current source. The Jab testing
utilized.a function generator/laboratory amplifier to energize the coil.

One graph representing accuracy and linerarity.was created for each
meter tested. These graphs, included on the following pages, combine all
remote test data points and any lab test data points with an applied field
value higher than the maximum value for the remote test. In addition any
data points with: actual field values within 10% of each other were combined
as one average data point. The chart label ’Applied Field’ refers to the
calculated actual field. “The formulas. used to calculate these data points
were as follows using terms described in the test plan (QAPP):

1) HILLEVEL_ACTUAL_FIELD. = ACTUAL_V / 0.0004026

2] LOW_LEVEL_ACTUAL_FIELD = DRIVE_I / 0.018536
3] ERROR = (MEASURED_FIELD - ACTUAL_FIELD) / ACTUAL_FIELD

12


MSI - MODEL MSI 20 - SN 79010854

METER CODE MANUFACTURER METER NAME

1VB Magnetic Sciences Int'l
2425 B Channing Way, ste.489
Berkeley, CA 94704

MSI,MM

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Date Time Temperature

Jan-28-91 08:56AM Th? F

RANGE SETTING ACTUAL FIELO MEASURED FIELD

2000 mG 502.19 mG 500.00
202.24 mG 200.00
200 mG 150.97 mG 150.00
100.72 mG 100.00
200 mG 58 mG 75.00
50.44 mG 50.00
200 mG 20.24 mG 20.00
10.25 mG 10.00

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date * Time Temperature

Mar-18-91 02:56PM 68° F

RANGE SETTING ACTUAL FIELD MEASURED FIELD

102.51 mG 100.00
200 mG 102.56 mG 100.00
20.66 mG 20.00
20.66 mG 20.00
10.41 mG 10.00
10.41 mG 10.00
2.21 mG 2.00
2.21 mG 2.00
1.13 mG 1.00
1.13 mG 1.00
Appendix VB

mTM
mG
mi

mG
mG
mG
mG
mG

322223328288

MODEL #
MSI-20

SERIAL #
79010854

Meter Background: mG

0

% ERROR

0.43%
“1.11%

71.02%

Meter Background

0

% ERROR

2.44%
72.50%
3.21%
3.21%
3.96%
+3.96%
+9.58%
+9.58%
11.31%
11.31%

76.11%

mG

AVG % ERROR

AVG % ERROR


%S

&
!

Responce (dB)

- (1 Square Wave

Frequency Responce - Meter II2 1 Savtooth Wave
SAFE PRO METER SN 1264 + Sine Wave

— Bezier Fit

T1911 1114111 111-1141
T T 1

10? 107 10¢
Frequency (Hz)


METER 60H2 NON-SINUSOIDAL RESPONCE DATA

Date

Jan-03-91

RANGE SETTING

100 mG

RANGE SETTING
100 mG

RANGE SETTING
100 mG

Time

12:03PM

SQUARE WAVE TESTING

WRI -.Model 660 - SN G1241

ACTUAL FIELD

64.85 mG
69.82 mG

SAWTOOTH WAVE TESTING

ACTUAL FIELD

47.67 mG
46.90 mG

SINE WAVE TESTING

ACTUAL FIELD

75.68 mG
69.60 mG

METER ANGULAR RESPONCE DATA

Date

Jan-03-91

Angle

o°
10°
20°
30°
40°
50°
60°
70°
80°
90°

Time

01:00PM

ACTUAL FIELD

BARR 2888 88

Temperature

Th? F

MEASURED

MEASURED

MEASURED

Temperat
76° F

MEASURED

FIELD

100.00
100.00

FIELD

100.00
100.00

FIELD

100.00
100.00

ure

FIELD

100.00
100.00
100.00
100.00
100,00
100.00
100.00
100.00
100.00
100.00

mG

mm
mm

BaaaRR RRR

Meter Background
0.5 mG

% ERROR

54.19%
43.22%

48.71%

‘% ERROR

109.80%
113.24%

111.52%

‘% ERROR

32.13%
43.68%

37.91%

Meter Background

0.5 mG

ANGULAR RESPONCE

eocecpe0os>
LNREUYR SSCS
SBEBARBRSS

RELATIVE RESPONCE

3.76 dB
3.12 dB

3.44 db

RELATIVE RESPONCE

6.44 dB
6.58 dB

6.51 db

RELATIVE RESPONCE

2.42 dB
3.15 dB

2.78 db


DINDIMA - GAUSSMAUS - SN GM0400

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Frequency Responce Setting - WIDE BAND

Date

May-17-91

RANGE SETTING

99 mG

RANGE SETTING

99 mG

RANGE SETTING

99 mG

Time
08:28AM
SQUARE WAVE TESTING

ACTUAL FIELD

113.74 mG
112.54 mG

SAWTOOTH WAVE TESTING

ACTUAL FIELD

78.58 mG
76.54 mG

SINE WAVE TESTING

ACTUAL FIELD

111.53 mG
113.53 mG

Frequency Responce Setting - BAND PASS

Date

May- 16-91

RANGE SETTING
99 mG

RANGE SETTING
99 mG

RANGE SETTING

99 mG

Time
12:28PM
SQUARE WAVE TESTING
ACTUAL FIELD

117.75 mG

118.90 mG

SAWTOOTH WAVE TESTING

ACTUAL FIELD

127.62 mG
127.51 mG

SINE WAVE TESTING
ACTUAL FIELD

107.28 mG
110.44 mG

Temperature

74° F

MEASURED FIELD

MEASURED FIELD

99.00 mG
99.00 me

MEASURED FIELO

99.00 mG
99.00 mG

Temperature
76° F

MEASURED FIELD
99.00 mG

99.00 mG

MEASURED FIELD

99.00 mG
99.00 mG

MEASURED FIELD

99.00 mG
99.00 mG

Appendix XVA

Meter Background
1 mG

‘% ERROR

‘% ERROR

25.99%
29.34%
27.66%
‘% ERROR
711.23%
12.80%
12.80%

Meter Background
1 mG

‘% ERROR
15.92%
16.74%
7 16.33%
‘% ERROR
7 22.42%
22.36%
722.39%
‘% ERROR
+7.72%

=10.36%

9.04%

RELATIVE RESPONCE

71.21 dB
“1.11 dB

+1216 db

RELATIVE RESPONCE

2.01 dB
2.23 dB

2.12 db

RELATIVE RESPONCE

-1.03 dB
“1.19 GB

“1.11 db

RELATIVE RESPONCE

71.51 GB
+1.59 cB

*1.55 db
RELATIVE RESPONCE
72.21 GB
2.20 oB
+2.20 db
RELATIVE RESPONCE
70.69 cB

-0.95 OB

+0.82 db


Angular Responce - Meter VIA - Linear Mode

++ Meter Responce
MONITOR Model 42b-1 SN 1286

Lo — Cosine Curve

a8 4

a6 5

96

04-4

Responce

42-5:

a0-
Oo 10 20 30 40 50 60 70 80 90

“Rotation Angle



METER CODE

XVB

DINDIMA - GAUSSMAUS - SN
MANUFACTURER:

The Dindima Group

PTY LTD

METER NAME

GAUSSMAUS

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Frequency Responce Setting - WIDE BAND

Date

Mar-28-91

RANGE SETTING

Time

01:12PM

ACTUAL FIELD

Frequency Responce Setting - BAND PASS

Date

Mar- 28-91

RANGE SETTING

Time

01:18PM

ACTUAL FIELD

Temperature
Te F

MEASURED FIELD

Temperature
66° F

MEASURED FIELD

Appendix  XVB

GM0401

MODEL #

Meter Background

0.1

% ERROR

+ 10.18%
10.30%
11.44%
11.46%
-10.18%

710.71%

Meter Background

0.1

% ERROR

7.92%
7.92%
11.80%
-10.96%
6.92%

“9.11%

mG

mG

SERIAL #
GM0401

AVG % ERROR

AVG % ERROR


SAFE - SAFE PRO METER - SN 1264

FREQUENCY RESPONCE PRIMARY DATA

Date Time
Feb-06-91 09:58AM
FREQUENCY RANGE SETTING
10 Hz Full 200 m6
20.Hz Full 200 mG
50 Hz Fult 200 mG
60 Hz Full 200 m6
100 Hz Full 200 mG
200 Hz Full 200 mG
500 Hz Full 200 mG
1 kkz Fult 200 mG
2 kHz Full 200 mG
5 kHz Full 200 mG
10 kHz Full 200 mG
FREQUENCY RESPONCE AUXILIARY DATA
Date Time
Feb-06-91 10:30AM
FREQUENCY RANGE SETTING
45 Hz Full 200 me
90 Hz Full 200 m6
120 Hz Full 200 mG
80 Hz Full 200 me
300 Hz Full 200 mG
700 Hz Full 200 mG

Temperature

76°

F

ACTUAL FIELD

791.88 mG
239.22
147.37
142.65
137.08
143.39
244.91
989.47
1195.80

BARR RBBB

2
3
Ss
o

u
2
3
a

Temperature

76°

F

ACTUAL FIELD

Appendix

155.07 mG
139.94 mG
139.92 mG
141.88 mG
163.86 mG
1017.24 mG

m2

Meter Background

0.2

Lit}

MEASURED FIELD

150.60
150.80
150.50
150.70
150.10
150.00
150.10
46.20
8.80
5.20
6.20

Meter Background

0.2

BaRBRRRRABR

mG

MEASURED FIELD

151.50
150.10
150.00
150.60
150.10

97.70

3222838

RELATIVE RESPONCE

+14.42 dB
-4.01 dB
0.18 dB
0.47 dB
0.78 dB
0.39 dB
+4.25 dB
+26.62 dB
42.66 cB
44.77 cB
739.29 dB

RELATIVE RESPONCE


EMD - MODEL ACGM-1 - SN 01-0117

FREQUENCY RESPONCE PRIMARY DATA

Date Time

Feb-11-91 09:44AM

FREQUENCY RANGE SETTING
THz FULL

FREQUENCY RESPONCE AUXILIARY DATA

Date Time
Feb-11-91
FREQUENCY RANGE SETTING
80 Hz FULL mG
150 Hz
300 Hz
35 Hz

Temperature
74° F

ACTUAL FIELD

Temperature
74° F

ACTUAL FIELD

256.09 mG
312.34 mG
930.28 mG
320.79 mG

Appendix VIIIB

Meter Background

0.1

mG

MEASURED FIELD

3.10
13.30
73.00
230.40
250.20
250.20
250.00
251.50
250.30

67.50

8.80

1.00

0.10

Meter Background

0.1

BARB RBBZRR RRB

mm

MEASURED FIELD

250.10
250.10
250.20
250.00

mG
mG
mG
im

RELATIVE RESPONCE

50.18 dB
37.53 cB
+22.74 dB
712.76 cB
5.77 dB
-0.80 dB
-0.39 GB
0.32 dB
74.63 cB
+23.02 dB
44.10 cB
60.06 dB
79.18 dB

RELATIVE RESPONCE

0.20 dB
+1.93 oB
711.41 dB
+2.17 dB


HOLADAY - HI 3604 - SW 76111

MINIMUM DETECTABLE SIGNAL

Date Time

Mar-28-91 10:55AM

Threshold Current mA

1 0.60
Trial 2 0.46
Number 3 0.46
4 0.33
5 0.21

NOTE: Meter reading increases by 0.003mG

METER RESOLUTION DATA

Date Time

Mar-28-91 11:02AM

Drive Current Start

1 1.8020

Trial 2 1.8020
Number 3 1.8020
4 1.8020

5 1.8020

NOTE: Meter reading increases by 0.2mG

A

>> >>>

Temperature

70° F

Threshold Field

Temperat
70° F

Threshold Ci

Appendix

32.37
24.82
24.82
17.80
11.33

22.23

ure

urrent

1.8060
1.8060
1.8060
1.8060
1.8070

IXA

min
max

55555

min
max

>

>> >>>

Meter Background

0.134
0.136

mG
mi

AVG THRESHOLD FOR SENSITIVITY

Meter Background

0.134
0.136

Resolution

215.80
215.80
215.80
215.80
269.75

226.59

mG
mG

uG
uG
uG
uG
uG
uG

5

Resolution %

0.22%
0.22%
0.22%
0.22%
0.27%

0.23%


WRI - Model 660 - SN G1241
FREQUENCY RESPONCE PRIMARY DATA

Date Time Temperature Meter Background

Jan-02-91 01:51PM 78° F 0.5 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

Taz 100 mG 832.76 mG 2.00 mG

2 Hz 100 mG 832.76 mG 6.00 mG

5 Wz 100 mG 832.76 mG 21.00 mG

10 Hz 100 mG 832.76 mG 100.00 mG

20 Hz 100 mG 249.88 mG 100.00 mG

50 Hz 100 mG 85.57 mG 100.00 mG

60 Hz 100 mG 72.13 mG 100.00 mG

100 Hz 100 mG 46.77 mG 100.00 mG

200 Hz 100 mG 33.16 mG 100.00 mG

500 Hz 100 mG 38.08 mG 100.00 mG

1 kz 100 mG 60.80 mG 100.00 mG

2 kHz 100 mG 106.68 mG 100.00 mG

5 kHz 100 mG 267.88 mG 100.00 mG

10 kHz 100 mG 508.32 mG 100.00 mG

FRI CY. PONCE AUXILIARY DATA
Time Temperature Meter Background
dan-03-91 10:18AM 78° F 0.5

FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELO RELATIVE RESPONCE
30 Hz 100 me 151.84 mG 100.00 mG 3.63 dB
40 Hz 100 mG 110.31 mG 100.00 mG 70.85 dB
300 Hz 100 mG 34.55 mG 100.00 mG 9.23 dB
5 kz 100 me 820.72 mG 20.00 mG 32.26 cB

Appendix IA


SYDKRAFT - MODEL 88_05 - VIIA

METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date Time.

Feb-27-91 09:05AM

E Field OFF Reading

EFM Metex 0.00

E Field OFF Reading(mG)
Meter Range
2mG

0.03
0.03
0.03
0.03
0.03

Trial
Number

Wann

AVG

Temperature
74° F

E Field ON Reading

10.0 kV/M +-5%

E Field ON Reading(mG)
Meter Range
2mG
0.03
0.03
0.03

0.03
0.03

0.03

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMOOULATED

Date Time
Feb-25-91 09:20AM
RF OFF Reading(mG)

Meter Range

2mG

1 MHz 0.03
10 MHz 0.03
100 MHz 0.03

Temperature
76° F
RF ON Reading(mG)
Meter Range
2mG
0.03

0.03
0.03

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date Time
Feb-25-91 09:20AM
RF OFF Reading(mG)

Meter Range

2mG

1 MHZ 0.03
10 MHz 0.03
100 MHz 0.03

Temperature
76° F
RF ON Reading(mG)
Meter Range
2G
0.03

0.03
0.03

Appendix VIIA


“GL

Frequency Responce ~ Meter XVA - Wide Band mn sae weve
DINDIMA . GAUSSMAUS SN GM0400 + Sine Wave
40 — — Bezier Fit
ao a
o-4
a +
Re]
g -20 4 +
ao
£
-49 4
-60 oT]
1 10 10? 10? 10

Frequency (Hz)


MONITOR - MODEL 42b-1 - SN 1287

MINIMUM DETECTABLE SIGNAL

Frequency Responce Setting - LINEAR MODE

Date Time
Mar-26-91 12:36Pm

Threshold Current mA
1 0.10 m
Trial 2 0.12 ma
Number 3 0.11 ma
4 0.08 ma
5 0.09 ma

Frequency Responce Setting - FLAT MODE

Date Time
Har-26-91 12:30eM
Threshold Current mA
Trial :
Nunber

vinune
e2000

NOTE:Meter reading increased by about 0.002mG

METER RESOLUTION DATA
Frequency Responce Setting - LINEAR MODE

Date Time
Mar-26-91 12:50PM
Meter Range 100mG

Drive Current Start A

1 1.4580 A

Trial 2 1.4580 A
Number 3 1.4580 A
4 1.4580 A

5 1.4580 A

Frequency Responce Setting - FLAT MODE

Date Time
Mar-26-91 12:55PM
Meter Range 100mG

Orive Current Start A

1 1.4580 A

Trial 2 1.4580 A
Number 3 1.4580 A
4 1.4580 A

5 1.4580 A

Temperature
6 F

Threshold Field

5.40
6.47
5.93
4.32
4.86

5.40

Temperature

61° F

Threshold Field

9.17
6.47
5.40
5.40
5.40

6.37

Temperature
59° F

Threshold Current

1.4650
1.4650
1.4650
1.4660
1.4660

Temperature
57° F

Threshold Current

1.4640
1.4630
1.4640
1.4630
1.4630

Appendix VI6

min
max

uG
uG
uG
UG
uG
uG

min
max

uG

55585

5

min
max

A

>> >>>

min
max

A

>> >>>

Meter Background

0.01
0.01

mG
mm

AVG THRESHOLD FOR SENSITIVITY

Meter Background

0.008
0.008

mG
mG

AVG THRESHOLD FOR SENSITIVITY

Meter Background

0.01 mG
0.01 mG
Resolution uG Resolution %

377.64 uG 0.48%
377.64 uG 0.48%
377.64 uG 0.48%
431.59 uG 0.54%
431.59 uG 0.54%
399.22 uG 0.50%

Meter Background

0.008 =m
0.008 =m

Resolution uG Resolution %
323.69 uG 0.41%
269.75 uG 0.36%,
323.69 UG 0.41%
269.75 uG 0.36%
269.75 uG 0.34%
291.32 uG 0.37%


FREQUENCY RESPONCE PRIMARY DATA

Date Time
Feb-12-91
FREQUENCY RANGE SETTING
5S kz 200 mG
10 Hz 200 mG
20 Hz 200 mG
50 Hz 200 mG
60 Hz 200 mG
100 Hz 200 mG
200 kz 200 mG
500 kz 200 mG
1 kHz 200 mG
2 kHz 200 mG
5 kHz 200 mG
10 kHz 200 mG
FREQUENCY RESPONCE AUXILIARY DATA
Date Time
Feb-12-91 02:06PM
FREQUENCY RANGE SETTING
30 Hz 200 mG
40 Hz 200 mG
55 Hz 200 mG
65 Hz 200 mG
80 Hz 200 mG

IRC - MODEL IRC 27

Temperature
76° F

ACTUAL FIELD
1017.7'

1
1

aaaARAAR ARR

Temperature
76° F

ACTUAL FIELD

1010.16 mG
1011.95 mG
226.25 mG
179.16 mG
395.18 mG

Appendix vB

Meter Background

0

mm

MEASURED FIELD

1.00
5.20
23.30
149.90
150.00
150.00
137.50
64.40
31.70
11.30
2.30
0.20

Meter Background

ty)

aRARRAAR BAAR

mG

MEASURED FIELD

61.90
144.80
150.00
150.00
150.00

BaBBB

RELATIVE RESPONCE

RELATIVE RESPONCE


9€

Percent Error

10 -

-10 4

-20 4

-30 4

~40 +

Accuracy and Linearity - Meter X
WALKER ELF-SOD SN 71181-205

TTT Trt] T TTT TITTY T TT TTTTT

10 10? 107
Applied Field (mG)


SAFE - SAFE PRO METER - SN 1263
METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date Time Temperature Meter Background
Feb-05-91 02:55PM 76° F 0.2 mG

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR RELATIVE RESPONCE
200 mG 1275.06 mG 95.00 mG 92.55% 22.56 dB

SAWTOOTH WAVE TESTING
RANGE SETTING ACTUAL FIELO MEASURED FIELD ‘% ERROR RELATIVE RESPONCE

200 mG 204.64 mG 150.10 mG 26.65% +2.69 dB

SINE WAVE TESTING
RANGE SETTING “ACTUAL FIELD MEASURED FIELD ‘% ERROR RELATIVE RESPONCE

200 mG 140.91 mG 150.20 mG 6.59% 0.55 dB

METER ANGULAR RESPONCE DATA

Date Time Temperature Meter Background
Feb-06-91 08:00AM 7o° F 0.2 mG
Angle ACTUAL FIELD MEASURED FIELD ANGULAR RESPONCE

o° 139.92 mG 150.00 mG 1.00

10° 142.23 mG 150.00 mG 0.98

20° 149.28 mG 150.30 mG 0.93

30° 162.44 mG 150.40 mG 0.86

40° 184.58 mG 150.60 mG 0.76

50° 218.90 mG 150.20 mG 0.64

60° 281.45 mG 150.40 mG 0.49

70° 415.08 mG 150.00 mG 0.33

80° 771.68 mG 150.60 mG 0.18

90° 969.92 mG 10.40 mG 0.01

Appendix 111


IRC - MODEL IRC 27

METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date Time Temperature
Feb-25-91 ‘ 01:58PM 78° F
E Field OFF Reading £ Field ON Reading
EFM Metex 0.00 10.0 kV/M +-5%
E Field OFF Reading(mG) E Field ON Reading(mG)
Meter Range Meter Range
Auto Auto
1 0.10 0.10
Trial 2 0.10 0.10
Nunber 3 0.10 0.10
4 0.10 0.10
5 0.10 0.10
0.10 ANG 0.10

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Date Time Temperature
Feb-21-91 02:38PM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Auto Auto
1 MHz 0.20 0.20
10 MHz 0.20 0.20
100 MHz 0.20 0.40

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA 60_Hz MODULATED

Date Time Temperature
Feb-21-91 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Auto Auto
1 MHz 0.20 0.20
10 MHz 0.20 0.00
100 MHz 0.20 0.00

Appendix VB



the meter is relatively wide band, then it may show a higher reading in a
square wave or sawtooth wave environment, even if the root mean square
amplitudes are the same as for the sine wave testing.

Results of the non-sinusoidal response tests are included with the
frequency response graphs on the following pages. The square wave response
is designated by a box, the sawtooth response is designated by a triangle.
The formulas used to calculate these data points were the same as the
formulas used for the frequency response data.

48


dIl

Responce

00

Angular Responce - Meter XVIIB
MAG TEC Model ELF 12 SN 4

+f

oO

AURA) CAAA AR EI |

10 20 30 40 50 60 70 80 90

Rotation Angle

-++ Meter Responce
— Cosine Curve


€s

Responce (dB)

40

Square Wave
Frequency Responce - Meter II1 in Sutooth Wave
SAFE PRO METER SN 1263 + Sine Wave
: — Bezier Fit

“10

TT TTT] Ty TTT T1917]

0? 007 10¢
‘Frequency (Hz)


MSI - MODEL MSI 20 - SN 79011626

MINIMUM DETECTABLE SIGNAL

Date Time

Mar-18-91 02:41PM

Threshold Current mA

1 3.50 mA
Trial 2 3.50 ma
Number 3 3.40 ma
4 3.40 ma
5 3.50 ma

NOTE: Meter changed to 0.1mG

METER RESOLUTION DATA

Date Time
Mar- 18-91 02:48PM
Meter Range Auto Range - 100mG
Drive Current Start A
1 1.9190 A
Trial 2 1.9190 A
Number 3 1.9190 A
4 1.9190 A
5 1.9190 A

NOTE: Meter readings changed by 0.2mG

Temperature

66° F

min
max

Threshold Field uG

188.83
188.83
183.43,
183.43
188.83

186.67

Temperature

Threshold Current

Appendix

1.9220
1.9260
1.9230
1.9240
1.9240

IVA

uG

uG
uG
uG
uG

Meter Background

0
0

mG
mG

uG AVG THRESHOLD FOR SENSITIVITY

PEer> >

Meter Background

Resolution

161.85
269.75
215.80
269.75
269.75

_ 237.38

ms
m

55585 &

5

Resolution %

0.15%
0.26%
0.20%
0.26%
0.26%

0.22%


sel

os MozZo.uMD

24

124

a

Susceptibility - Meter VIIIB
EMD Model ACGM-1 SN01-0117

60 Hz
E-FIELD 1ekV

ee et

10 MHz «6108 MHz
UNHOD .

4 MHz

19 MHz $00 MHz
6euz MoD

COFIELD OFF
FELD ON


MEDA - Model PLM-100 - SN 1381

METER 60H2 E FIELD SUSCEPTIBILITY DATA

Date Time Temperature
May-20-91 09:52AM 74° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.00 KV/M +
— Field OFF Reading(mG) E Field ON Reading(mG)
Meter Range Meter Range
20 20
1 0.06 0.12
Trial 2 0.07 0.22
Number 3 0.08 0.23
4 0.10 0.14
5 0.09 0.17
0.08 AVG 0.17

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Date Time Temperature
May-17-91 . 02:43PM 74° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
20 20
1 MHz 0.08 0.08
10 MHz 0.08 0.08
100 MHz 0.08 0.27

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date Time Temperature
May-17-91 02:48PM Te
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
20 20
+ MHz 0.08 0.08
10 MHz 0.08 0.08
100 Miz 0.08 0.48

Appendix XVIB


6b

Responce (dB)

40 -

Frequency Responce - Meter I! [Square Wave
A Savtooth Wave

WRI Model 670 SN G1257 ++ Sine Wave

— Bezier Fit

rn TTT rr)

10 10? 10? 104

Frequency (Hz)


IRC - MODEL IRC 109
RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Date Time Temperature
Feb-22-91 01:00PM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
2mG 2m
1 MHZ 0.08 0.08
10 MHz 0.08 0.08
100 MHz 0.08 0.08

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date Time Temperature
Feb-22-91 02:05PM 80° F
RF OFF Reading(mG) RF ON Reading(mc)
Meter Range Meter Range
2mG 2mG
1 MHz 0.08 0.08
10 MHz 0.08 0.08
100 MHz 0.08 0.08

Appendix VA


DINDIMA - GAUSSMAUS - SN GMO4O1

METER 60H2 E FIELD SUSCEPTIBILITY DATA

Frequency Responce Setting - WIDE BAND

Date Time Temperature

May- 17-91 09:27AM Th? F
& Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.00 KV/M +-5%
E Field OFF Reading(mG) E Field ON Reading(mG)

Meter Range. Meter Range
9.9 9.9
1 0.30 1.10
Trial 2 0.30 1.10
Number 3 0.20 1.00
4 0.20 1.10
5 0.20 4.10
0.24 AVG 1.08

Frequency Responce Setting - BAND PASS

Date Time Temperature

May-20-91 09:21AM 74° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.00 KV/M +-5%
E Field OFF Reading(mG) E Field ON Reading(mG)

Meter Range Meter Range
99 9.9
1 0.30 1.10
Trial 2 0.20 1.10
Number 3 0.20 1.10
4 0.30 0.60
5 0.20 4.10
0.26 AVG 1.00

Appendix  XVB


MSI - MODEL MSI 20 - SN 79010854

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Date Time

Temperature
Feb-21-91 12:49PM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
ALL ALL
1 Maz 0.00 0.00
10 MHz 0.00 0.00
100 MHZ 0.00 0.00

NOTE: Meter tested without extension cable.

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date Time

Temperature
Feb-21-91 12:55PM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
ALL ALL
4 MHZ 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 0.00

Appendix 1VB


HINIMUM DETECTABLE SIGNAL

TRC --MODEL IRC 109

Date Time Temperature
Mar-21-91 01:30PM 53° F
Threshold Current mA Threshold Field
1 1.50 ma 80.93
Trial 2 1.70 m 91.72
Number 3 1.20 m 64.74
4 1.20 ma 64.74
5 1.50 mA 80.93
NOTE: Meter readings increase by about 0.04mG 76.61
METER RESOLUTION DATA
Date Time Temperature
Har-21-91 01:37PM 53° F
Meter Range Auto Range - 100mG

Drive Current Start

1 1.8840
Trial 2 1.8840
Number 3 1.8840
4 1.8840
5 1.8840

NOTE: Heter readings increase by 0.3mG

A Threshold Current

1.8910
1.8910
1.8910
1.8910
1.8910

PP reer

Appendix VA

min
max.

uG

$5555

uG

min
max

>>eer> >

Meter Background

0.058
0.075

mG
mG

AVG THRESHOLD FOR SENSITIVITY

Meter Background

0.2
0.3

Resolution

377.64
377.64
377.64
377.64
377.64

377.64

mG
mG

uG

Resolution %

0.37%
0.37%
0.37%
0.37%
0.37%

0.37%


82

- _ (1 Square Wave
Frequency Responce - Meter XVB ~- Band Pass A Sawtooth Wave
DINDIMA GAUSSMAUS SN GMO401 + Sine Wave
40 - — Bezier Fit
a
o-
8
8 -20 4
+
&
-40 =
a
+
-60 T TT T1117 TT—TTTTT1T7 TTT TTT]
1 10 10? 10°

Frequency . (Hz)


MEDA - Model PLM-100 - SN 1380
METER CODE MANUFACTURER: MODEL # SERIAL #

XVIA Macintyre Electronic Design PLM- 100 1380

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Date Time Temperature Meter Background: mG
Jan-14-91 01:14PM . 77° F 0.1
RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR
200 mG 149.88 mG 150.00 mG 0.08%

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date Time Temperature Meter Background
Aug-29-91 11:21AM 97° F -0.005 mG
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
200 mG 100.24 mG 100.00 mG 0.23%
200 mG 101.05 mG 100.00 mG +1.04%
200 mG 50.07 mG 50.00 mG 0.13%
200 mG 50.07 mG 50.00 mG 70.13%
20 mG 20.04 mG 19.99 mG 0.26%
20 mG 20.06 mG 19.99 mG 0.34%
20 mG 5.01 mG 5.00 mG 0.13%
20 mG 5.01 mG 5.00 mG 70.13%
20 mG 1.01 mG 1.00 mG -0.87%
20 mG 1.01 mG 1.00 mG 0.87%
0.41% ‘AVG % ERROR

Appendix XVIA


88

Responce

L105

10

Angular Responce - Meter IT1
SAFE PRO METER SN 1263

2 32 40 50 60

Rotation Angle

70

8&0

90

-++ Meter Responce
— Cosine Curve


ey

Accuracy and Linearity - Meter XVIA
MEDA Model PLM-100 SN 1380

10-5
54
5
& a q - t ++
2
oa
8
a
-5 4
-10 nS | TTY 4417]

1 10 10? 10°
Applied Field (mG)


621

os Mazo TOMsD

0.64

6.4

il

Susceptibility - Meter VA
IRC Model IRC-109

na ane

Hat a

60 Hz
E-FIELD 10k¥/1

4 MHz to nus 108 MHz

1 HHz = =10 MHz 400 MHz
6e@Hz HOD

(FIELD OFF
WIFIELD ON


ell

Responce

Lo-

ae

06 -

a4-

a2 5

0.0

Angular Responce ~ Meter XVB ~ Band Pass
DINDIMA - GAUSSMAUS - SN GMO0401

10

20

30

40 50

Rotation Angle

4

60

70

80

90

+

Meter Responce

Cosine Curve


WALKER - ELF-50D - SW 71181-205
METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date Time Temperature Meter Background
Feb-07-91 10:09AM 74° F 0 1G

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR RELATIVE RESPONCE
LO mG 145.21 mG 149.00 mG 2.61% 0.22 4B

144.61 mG 149.00 mG 3.04% 0.26 dB
2.82% 0.24 db

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR RELATIVE RESPONCE
LO mG 133.73 mG 149.00 mG 11.42% 0.93 dB

133.01 mG 149.00 mG 12.02% 0.98 dB
11.72% 0.96 db

SINE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELO % ERROR RELATIVE RESPONCE
LO mG 151.59 mG 150.00 mG 71.05% -0.09 dB

151.32 mG 150.00 mG 0.87% -0.07 dB
70.95% -0.08 db

METER ANGULAR RESPONCE DATA

Date Time Temperature Meter Background
Feb-07-91 10:20AM 74° F 0 mG
Angle ACTUAL FIELD MEASURED FIELD ANGULAR RESPONCE

o° 151.89 mG 150.00 mG 1.00

10° 154.35 mG 150.00 mG 0.98

20° 162.57 mG 150.00 mG 0.93

30° 173.99 mG 150.00 mG 0.87

40° 196.25 mG 150.00 mG 0.77

50° 233.21 mG 150.00 m6 0.65

60° 304.12 mG 150.00 mG 0.49

70° 448.58 mG 150.00 mG 0.33

g0° 864.16 mG 150.00 mG 0.17

90° 959.54 mG 13.00 mG 0.01

Appendix X


ze

On MOZOTOMD

nN

Susceptibility - Meter IA2
WRI Model 660 SN Gt242

ian Geek Eee!

an el

68 Hz
E-FIELD 10k

1 MHz 18 MHz «6108 MHz
UNHOD

4 MHz

10 HHz 100 MHz
6eHz MOD

COFIELO OFF
FIELD ON


1 + MODEL MSI 20 - SN 79010854

MIN DETECTAI AL

Date Time Temperature
Mar-18-91 03:06PM 68° F min
max
Threshold Current mA = Threshold Field uG
1 3.40 m 183.43 uG
Trial 2 3.30 m 178.04 uG
Number 3 3.50 m 188.83 uG
4 3.40 m 183.43 UG
5 3.30 m 178.04 uG
WOTE: Meter changed to 0.1mG 182.35 uG
METER RESOLUTION DATA
Date Time Temperature
Mar-18-91 03:15PM 6 F min
max
Meter Range 200 mG
Drive Current Start A Threshold Current A
1 1.9010 A 1.9050 A
Trial 2 1.9010 A 1.9040 A
Number 3 1.9010 A 1.9050 A
4 1.9010 A SO A
5 1.9010 A -9050 A

WOTE: Meter changed about 0.25mG

Appendix IVB

Meter Background

mG
mG

AVG THRESHOLD FOR SENSITIVITY

Meter Background

0 mG
0 mG
Resolution uG Resolution %

215.80 uG 0.21%
161.85 uG 0.15%
215.80 uG 0.21%
215.80 uG 0.21%
215.80 uG 0.21%
205.01 uG 0.20%


Let

on MazZOTHMD

Susceptibility - Meter IVA
MS! Model MSI-20 SN 79011626

—

1.64
4.24
0.8.
e.44
eo = — — =
60 Hz 7 Hz 1@ MHz = 108 MHz 1 MHz = 10 MHz tee MHz
E-FIELD 16k¥/4 UNHOD 6eHz HOD

COFIELD OFF
FIELD ON


MONITOR - MODEL 42b-1 - SN 1287

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Frequency Responce Setting - LINEAR MODE
Date Time Temperature Meter Background
Jan-29-91 02:06PM Ta? F 0.2 mG
SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELO ‘% ERROR RELATIVE RESPONCE

250 m6 40.94 mG 150.00 m6 266.67% 11.29 dB
41.18 mG 150.00 mG 264.23% 11.23 dB
265.45% 11.26 db
SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR RELATIVE RESPONCE
250 me 72.43 mG 150.00 mG 107.10% 6.32 dB

71.58 mG 150.00 mG 109.54% 6.43 dB

108.32% 6.37 db

SINE WAVE TESTING .

RANGE SETTING . ACTUAL FIELD MEASURED FIELD x ERROR RELATIVE RESPONCE
250 mG . 167.86 mG 150.00 m6 1.66% 0.12 dB

467.42 mG 150.00 mG 1.96% 0.16 oB

. , 1.70% 0.14 @B

Frequency Responce Setting - FLAT MODE
Date Time - Temperature Meter Background
Jan-29-94 01:37PM . 74? F “ 0.11 6G

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELO ‘% ERROR RELATIVE RESPONCE
250 mG 148.04 mG 150.00 mG 1.33% 0.11 dB

147.86 mG 150.00 mG 1.46% 0.12 GB
1.39% 0.11 db

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR RELATIVE RESPONCE
250 mG 147.67 mG 150.00 mG 1.58% 0.13 dB

148.66 mG 150.00 mG 0.90% 0.07 aB

1.26% 0.10 db

SINE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR RELATIVE RESPONCE
250 mG 147.27 mG 150.00 mG 1.86% 0.16 dB

147.24 mG 150.00 mG 1.87% 0.16 dB
1.86% 0.16 db

Appendix VIB


MONITOR - MODEL 42b-1 - SN 1286
METER CODE MANUFACTURER METER NAME MODEL # SERIAL #
VIA Monitor Industries Milligauss 428-1 1286

6112 Fourmile Canyon
Boulder, CO 80302

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Frequency Respuice Setting - LINEAR MODE

Date Time Temperature Meter Background- mG
Feb-28-91 08:40AM 74° F 0.25
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
1000 mG 493.69 mG 500.00 mG 1.28%
250 200.67 mG 200.00 mG 70.33%
250 mG 149.45 mG 150.00 mG 0.36%
500 143.19 mG 150.00 mG 4.75%
100 mG 74.96 mG 75.00 mG 0.04%
100 49.18 mG 50.00 mG 1.67%
1.30% AVG % ERROR

Frequency Responce Setting - FLAT MODE

Date Time Temperature Meter Background- mG
Feb-28-91 09:42AM The F 0.12
RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR
1000 mG 493.72 mG 500.00 mG 1.27%
250 200.62 mG 200.00 mG 70.31%
500 mG 142.95 mG 150.00 mG 4.93%
250 149.33 mG 150.00 mG 0.44%
100 mG 74.99 mG 75.00 mG 0.01%
100 49.16 1G 50.00 mG 1.72%
1.35% AVG % ERROR

‘Appendix VIA


IST

os moazovuma

214

12.6

8.44

if

Susceptibility - Meter XVIIA
MAG TECH ELF 12 SN2

WN

Ria

60 Hz
E-FIELD teKust

4 Maz

40 His

108 MHz

4 MHz

10 His

fee Miz

OFELD OFF
FIELD ON


SUMMARY

This project has attempted to evaluate the performance of available
single-axis, coil-type power frequency magnetic field meters. This document
is intended to present the data collected in this project. The numerical
test data is included in the appendix. There is one appendix for each meter
tested designated by the meter code assigned during the initial check-in.
Each of these includes results data for each of the test catagories described
in Table 2 on page 6.

155


Final Report

Laboratory Testing of Commercially
Available Power Frequency Magnetic Field
Survey Meters

Prepared by:

SC&A, Inc.
1311 Dolley Madison Boulevard
McLean, Virginia 22101
and
Science Applications International Corporation
1710 Goodridge Drive
McLean, Virginia 22102

Contract No.: 68090170
Work Assignment: Laboratory Testing ...

Prepared for:
U.S. Environmental Protection Agency
Office of Radiation Programs
401 M Street S.W.
Washington, DC 20460

E. Manti ply
Work Assignment Manager

R. Lyon
Quality Assurance Coordinator

September 30, 1991


EMD - MODEL ACGM-1 - SN 01-0116

METER CODE MANUFACTURER

VITIA Electromagnetics Design

METER NAME

Gaussmeter

9100 Bloomington Pkwy. ste. 183

Bloomington, MN 53431

METER ACCURACY AND LINEARITY AT 60Hz- TEST LOCATION LAB

Date Time

Feb-08-91 11:22AM

RANGE SETTING ACTUAL FIELD

FULL

Temperature
76° F

MEASURED FIELD

500.00
250.00
150.00
75.00
50.00
25.00
10.00

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date Time

Mar-29-91 11:21AM

RANGE SETTING ACTUAL FIELD

FULL

NOTE:

Temperature
72°F

MEASURED FIELD

100.00
100.00
100.00
50.00
50.00
10.00
3.00

This meter reads significantly different values

Appendix VIIIA

MODEL # SERIAL #

ACGM-1 01-0116

Meter Background: mG

0

‘% ERROR

75.58%

BaaR RRR

4.00% ° AVG % ERROR

Meter Background

0 mG

% ERROR

5.91%

BA28888

5.54% AVG % ERROR

between Sun & Shade


EMD - MODEL ACGM-1 - SN 01-0117
METER CODE MANUFACTURER METER NAME MODEL # SERIAL #
VILIB Electromagnetics Design Gaussmeter ACGM-1 01-0117

9100 Bloomington Pkwy. ste. 183
Bloomington, MN 53431

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Date Time Temperature Meter Background- m¢
Feb-11-91 09: 254M 7h F 0.1
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
FULL mG 519.75 mG 500.00 mG +3.80%
259.76 mG 250.00 me -3.76%
155.79 mG 150.00 mG 3.71%
77.70 mG 75.00 mG 3.47%
51.72 mG 50.00 mG -3.33%
25.68 mG 25,00 mG +2.65%
10.16 mG 10.00 mG “1.59%
+3.19% AVG % ERROR

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date Time Temperature Meter Background

Mar-29-91 12:45PM 76° F 0.1 mG
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
102.72 mG 100.00 me +2.65%
FULL 51.16 mG 50.00 mG -2.26%
10.09 mG 10.00 m6 -0.87%
3.18 mG 3.00 mG -5.75%

+2,88% AVG % ERROR
NOTE: Keep Meter in Shade

Appendix VITIB


EFM - Metex-3800 - SN 542513

FREQUENCY RESPONCE PRIMARY DATA

Date Time Temperature Meter Background
May-15-91 11:39AM . 77° F 0.1 mG

FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

2 Hz 200 mG 687.48 on” 24.30 mG -29.03 dB

5 Kz 200 mG 687.48 mG 95.20 mG 717.17 GB

10 Hz 200 mG 687.48 mG 150.60 mG +13.19 dB

20 Kz 200 mG 367.11 mG 150.70 . mG +7.73 GB

50 Hz 200 mG 169.27 mG 150.00 mG +1.05 dB

60 Hz 200 mG 149.53 mG 149.95 mG 0.02 dB

100 Hz 200 mG 115.18 mG 149.95 mG 2.29 dB

200 Hz 200 mG + 98.39 mG 149.95 mG 3.66 dB

500 Hz 200 mG 103.58 mG 150.00 mG 3.22 dB

1 kHz 200 mG 134.80 mG 150.05 mG 0.93 dB

2 kHz 200 mG 222.58 mG 150.05 mG 73.42 cB

S kHz 200 mG 555.27 mG 149.95 mG +11.37 cB

10 kHz 200 mG 547.62 mG 57.75 mG 719.54 dB

FREQUENCY RESPONCE AUXILIARY DATA

Date Time Temperature Meter Background
May- 15-91 12:14PM 77° F 0.1 mG

FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

80 Hz 200 mG 126.97 mG 150.00 mG 1.45 dB

150 Hz 200 m6 102.61 mG 150.05 mG 3.30 dB

3.75 dB

300 Hz 200 mG 97.39 mG 149.95 mG

Appendix XIVA


So

a8 mazotums

1.6

0.4

{]

Susceptibility - Meter XVA - Wide Band

DINDIMA GAUSSMAUS SN GM0400

am ee

OFELD OFF
MFELD ON

60 Hz
E-FIELD 10k¥1

10 MHz = 168 MHz
‘UNMOD

4 MHz

10 MHz 100 MHz
6eHz MOD


Angular Responce - Meter XA
WALKER ELF-5OD SN 71181-204

-+- Meter Responce

90

Responce

LO-

a8 -

06 —-

04 -

a2

40

Oo

ARIA) RARE UE RE A RE

10

20

30

40 50

Rotation Angle

60

70

80

90

Cosine Curve


FREQUENCY RESPONCE PRIMARY DATA

Frequency Responce Setting - WIDE BAND

Date Time
May- 16-91 01:00PM
FREQUENCY RANGE SETTING
2 Hz 99 mG
5 Wz 99 1G
10 Hz 99 mG
20 kz 99 mG
50 Hz 99 mG
60 Hz 99 1G
100 Hz 99 mG
200 Hz 99 mG
500 Hz 99 mG
1 kHz 99 mG
2 kHz 99 mG
5 kHz 99 mG
10 kHz 99 mG

Frequency Responce Setting - BAND PASS

Date Time
May- 16-91 11:18AM
FREQUENCY RANGE SETTING
5 uz 99 mG
10 Hz 99 mG
20 Hz 99 1G
50 Hz 99 mG
60 Hz 99 mG
100 Hz 99 mG
200 Hz 99 mG
500 Hz 99 mG
1 kHz 99 mG

DINDIMA - GAUSSMAUS - SN GMO4O0

Temperature
76° F

ACTUAL FIELD

a

Ss

®
GES

aaasss
ayryespeya
SRARABRRRRRAS

GOe

Temperature
76° F

ACTUAL FIELD

1428.65
1428.65
1262.55
421.86
115.43
554.06
378.53
360.05
175.56

2223222833

‘Appendix XVA

Meter Background

1

Lt

MEASURED FIELD

2.00
21,00
99.00
99.00
99.00
99.00
99.00
99.00
99.00
99.00
99.00
99.00
99.00

Meter Background

1

BABRRABBRABBBBB

Lic}

MEASURED FIELD

7.00
37.00
72.00
81.00
99.00
40.00
13.00

5.00

2.00

BARBBRBBB

RELATIVE

RELATIVE

RESPONCE

38.56 dB
18.14 dB
4.67 cB
70.56 aB
70.93 dB
1,03 dB
1.17 GB
1.22 dB
+1.30 dB
+4.40 dB
+5.82 cB
+20.40 dB
-1.48 dB

RESPONCE

+46.20 dB
-31.73 GB
+24.76 GB
-14.33 GB
1.33 dB
-22,83 dB
29.28 dB
-37.15 dB
-38.87 dB


EFM - Metex-3800 - SN 542832

METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date Time Temperature

May-20-91 08:40AM Tae F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.0 KW/M +-5%
E Field OFF Reading(nG) £ Field ON Reading(mG)

Meter Range Meter Range
200 200
1 0.20 2,00
Trial 2 0.20 1.75
Number 3 0.20 2125
4 0.20 4.00
5 0.20 3135
0.20 WG 2.67

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Date Time Temperature
May-17-91 01:22PM _ Te
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
200 200
1 MHz 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 0.00

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date Time Temperature
May-17-91 01:11PM TF
RF OFF Reading(ms) RF ON Reading(mG)
Meter Range Meter Range
200 200
1 MHz 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 0.00

‘Appendix XIVB


99

. _y (Square Wave
Frequency Responce Meter VIIB T_ Saxtooth Wave
SYDKRAFT Model 88-05 + Sine Wave
40 - : — Bezier Fit
20 +
o-
8
8
5 -20-
o
e
40 —
-60 TTT TTT TTT T TTT TITTY TTT TTTT1T
1 10 10? 107

Frequency (Hz)


2g

Responce (dB)

- (1 Square Wave
Frequency Responce - Meter IA2 1 Savtooth Wave
WRI Model 660 SN G1242 + Sine Wave
— Bezier Fit

T TT TrTtTtTyT T TOT TTiTtTeT T TT TTttTyT ~T Tor TrTyt
T T T l

10 10? 10? 1o*
Frequency (Hz)


WALKER - ELF-50D - SN 71181-204
METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date Time Temperature
Feb-25-91 03:05PM 78° F
& Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.0 kV/M +-5%

E Field OFF Reading(mG) E Field ON Reading(mG)

Meter Range Meter Range
Lo Lo
1 0.00 6.00
Trial 2 0.00 6.00
Number 3 0.00 6.00
4 0.00 6.00
5 0.00 6.00
0.00 AVG 6.00
 E Field OFF Reading(mG) E Field ON Reading(mG)
Meter Range . Meter Range
HI HI
1 0.00 20.00
Trial 2 0.00 20.00
Number 3 0.00 20.00
4 0.00 20.00
5 0.00 20.00
0.00 AVG 20.00
RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED.
Date Time Temperature
Feb-22-91 . 09:30AM 79° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Lo to
1 MHz 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 8.00
Meter Range Meter Range
Ls HI
1 MHz 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 60.00

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DA’ 60_Hz MODULATED

Date Time Temperature
Feb-22-91 09:00AM 79° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Lo Lo
1 MHz 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 . -8.00
Meter Range Meter Range
HI K
4 MHz 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 60.00

Appendix XA


WRI - Model 660 - SN G1241

MINIMUM DETECTABLE SIGNAL

Date Time

Mar-14-91 01:49PM
Threshold Current m4
i 1.90 mA
Trial 2 2.30 mA
Number 3 2.50 mA
4 2.00 m
5 2.10 mA

METER RESOLUTION DATA

Date Time

Mar-14-91 01:40PM

Meter Range Auto Range - 100mG

Drive Current Start

1 0.1310 A
Trial 2 0.1360 A
Number 3 0.1400 A
4 0.1370 A
5 0.1390 A

Temperature

64° F min
max

Threshold Field uG

102.51 uG
124.09 uG
134.88 uG
107.90 uG
113.30 uG

Meter Background

0.3
0.3

mi
mG

116.53 uG AVG THRESHOLD FOR SENSITIVITY

Temperature

60° F min
max

A Threshold Current A

0.1620 A
0.1590 A
0.1540 A
0.1520 A
0.1520 A

Appendix 1A1

Meter Background

0.35
0.35

Resolution

1672.42
1240.83
755.29
809.24
701.34

1035.82

mTM
mG

uG

uG
uG
uG

Resolution %

23.66%
16.91%
10.00%
10.95%

9.35%

14.18%


0

Responce

LOA

a8 3

06 -

04 -

a2-

00-
oO

_ Angular Responce - Meter VIIB

10

20

SYDKRAFT

30

Model 88-05

40 50

Rotation Angle

60

70

80

90

+

Meter Responce
Cosine Curve


HOLADAY - HI 3604 - SN 76111
METER CODE MANUFACTURER METER NAME MODEL # SERIAL #
XA Holaday Industries ELF Field H1-36046 76111

14825 Martin Drive
Eden Prairie, MN 55344

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Date Time Temperature Meter Background- mG
Feb-14-91 09:05AM TS° F 0.192
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
2000 mG 929.68 mG 999.50 mG 7.51%
460.51 mG 500.00 me 8.58%
200 m6 144.88 mG 150.10 mG 3.60%
47.96 mG 50.00 mG 4.25%
200 mG 23.63 mG 24.95 mG 5.59%
14.46 mG 15.03 mG 3.91%
20 mG 9.61 mG 10.02 mG 4.28%
4.88 mG 5.10 mG 4.48%
5.28% AVG % ERROR

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date Time Temperature Meter Background

Mar-28-91 10:47AM 69° F 0.135 mG
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
97.65 mG 100.00 mG 2.41%
Auto Range 97.22 mG 100.00 mG 2.86%
71 mG 10.00 mG 2.98%
0.95 mG 1.00 mG 4.66%

3.23% AVG % ERROR

Appendix XA


92

10 -

Percent Error

Accuracy and Linearity - Meter VIA - Flat Mode

MONITOR Model 42b-1 SN 1286

10

at

| TT TTY TT ry TTT]

! 10 10? 107
Applied Field (mG)


MONITOR - MODEL 42b-1 - SN 1286

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Frequency Responce Setting - LINEAR MODE

Date Time Temperature
Feb-28-91 03:40PM 7h? F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
0.581 0.581
1 Miz 0.16 0.16
10 MHz 0.16 0.16
100 MHz 0.46 0.16

Frequency Responce Setting - FLAT MODE

Date Time Temperature
Feb-28-91 04:00PM 74° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
0.51 0.5*1
1 MHz 0.11 0.11
10 MHz 0.11 0.11
100 MHz 0.11 0.1

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Frequency Responce Setting - LINEAR MODE

Date - Time Temperature
Mar-01-91 08:15AM 74° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
0.541 0.541
1 MHz 0.14 0.14
10 MHz 0.14 0.14
100 MHz . 0.14 0.14

Frequency Responce Setting - FLAT MODE

Date Time Temperature
Mer-01-91 08:30AM The F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
0.541 0.581
1 Miz 0.11 on
40 Miz ont ot
100 Miz on ot

Appendix VIA


MONITOR - MODEL 42b-1 - SN 1287

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Frequency Responce Setting - LINEAR MODE

Date Time Temperature
Feb-22-91 02:25PM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
0.5*1 0.5"1
1 MHz 0.12 0.12
10 MHz 0.12 0.12
100 MHz 0.12 0.12

Frequency Responce Setting - FLAT MODE

Date Time Temperature
Feb-25-91 08:15AM 74° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
0.581 0.5*1
1 MHz 0.05 0.05
10 MHz 0.05 0.05

100 MHz 0.05 0.05

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED
Frequency Responce Setting - LINEAR MODE

Date Time Temperature
Feb-22-91 02:55PM 80° F
RF OFF Reading(ms) RF ON Reading(mG)
Meter Range Meter Range
0.5*1 0.5*1
1 MRz 0.12 0.12
10 MHz 0.12 0.12
100 MHz 0.12 0.12

Frequency Responce Setting - FLAT MODE

Date Time Temperature
Feb-25-91 08:15AM 74° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
0.5¢1 0.5*1
1 MHz 0.05 0.05
10 MHz 0.05 0.05
100 Miz 0.05 0.05

Appendix VIB


74

Accuracy and Linearity ~ Meter IVB
MSI Model MSI-20 SN 79010854

5 5
o +
5
c Wo
uw
2
3
é
-10 4
aS T TTT TTT T TT TTT TTT T TT T1111]
1 10 10? 107

Applied Field (mG)


$01,

Responce

10-

ae

a6-

a4-

42-

a0 -

10

Angular Responce - Meter IXB

HOLADAY

20

30

HI-3604

40 50

Rotation Angle

SN 76112

60

70

80

90

+

Meter Responce
Cosine Curve


MINIMUM DETECTASLE SIGNAL

Date

Mar-14-91

Trial
Number

wnuns

METER RESOLUTION DATA

Date

Mar-14-91

Meter Range

WRI: Model 670 - SN G1257

Threshold Current mA

sseex
22222

Drive Current Start

Trial
Number

wane

0.1800
0.1690
0.1730
0.1930
0.1820

Temperature
58° F

Threshold Field

118.69
97.11
97.11

102.51

102.51

103.58

Temperature
58° F

Threshold Current

A

A 0.2130
A 0.2060
A 0.2030
A 0.2000
A 0.2000

Appendix 11

min

c
5

55855

&

min

Pre>> >

Meter Background

0.1
0.2

mG
mG

AVG THRESHOLD FOR SENSITIVITY

Meter Background

0.1
0.25

Resolution

1780.32
1996.12
1618.47
377.64
971.08

1348.73

mG
mm

uG

uG
uG

uG
ug

Resolution %

18.33%
21.89%
17.34%
3.63%
9.89%

14.22%


MEDA - Model PLM-100 - SN 1381
METER CODE MANUFACTURER MODEL # SERIAL #

XVIB Macintyre Electronic Design PLM-100 1381

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Date Time Temperature Meter Background- mG
May-15-91 08:13AM 76° F 0.1
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
200 mG 150.70 mG 150.00 mG 0.46%

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date Time Temperature Meter Background
Aug: 29°91 12:54PM 99° F 0.03 mG
RANGE SETTING ACTUAL FIELO MEASURED FIELD % ERROR
200 mG 100.51 mG 100.00 mG +0.50%
200 mG 100.51 mG 100.00 mc 0.50%
200 mG 50.44 mG 50.00 mG -0.87%
200 mc 50.50 mG 50.00 mG 0.98%
200 mG 20.11 mG 19.99 mG 0.58%
200 mG 20.10 mG 19.99 mG 70.55%
200 mG 5.03 mG 5.00 mG 0.66%
200 mG 5.03 mG 5.00 mG 0.56%
200 mG 1.03 mG 1.00 mG 72.44%
200 mG 1.02 m6 1.00 mG 71.93%
0.96% AVG % ERROR

Appendix  XVIB


WRI - Model 670 - SN G1259

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date

Feb-08-91

RANGE SETTING
100*10 mG

RANGE SETTING
10010 mG

RANGE SETTING
100*10 mG

Time
01:58PM

SQUARE WAVE TESTING

ACTUAL FIELL

135.52 mG
136.19 mG

SAWTOOTH WAVE TESTING

ACTUAL FIELD

86.91 mG
85.89 mG

SINE WAVE TESTING

ACTUAL FIELD

148.48 mG
151.04 mG

METER ANGULAR RESPONCE DATA

Date

Feb-08-91

Angle

0°
10°
20°
30°
40°
50°
60°
70°
80°
90°

Time

2:08PM

ACTUAL FIELD

Temperature
76° F

MEASURED FIELD

200.00
200.00

MEASURED “FIELD

200.00
200.00

MEASURED FIELD

200.00
200.00

Temperature
76° F

MEASURED FIELD

200.00
200.00
200.00
200.00
200.00
200.00
200.00
200.00
200.00
200.00

Appendix 12

mG
mm

mG
mG

mG
mG

BABAR BRBBS

Meter Background
2 mG

‘% ERROR

47.58%
46.85%

47.22%

% ERROR

130.12%
132.85%

131.49%

% ERROR

34.69%
32.41%

33.55%

Meter Background
2 ms

ANGULAR RESPONCE
1.00

©
RSFBARSR

RELATIVE RESPONCE

3.38 dB
3.34 dB

3.36 db

RELATIVE RESPONCE

7.24 dB
7.34 dB

7.29 db

RELATIVE RESPONCE

2.59 dB
2.44 dB

2.51 db


WRI - Model 660 - SN G1241

METER CODE MANUFACTURER METER NAME MODEL # . SERIAL #
1A Widerange Instruments Magalert 660 1261
110 Shelter Lagoon
Santa Cruz, CA 95060

METER ACCURACY AND LINEARITY 60Hz_- TEST LOCATION LAB

Date Temperature Meter Background

dan-02-91 78° F 0.5 mG
RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR
100 mG 79.06 mG 100.00 mG 26.48%
73.17 mG 100.00 mG 36.66%
100 mG 62.30 mG 100.00 mG 60.53%
67.04 mG 100.00 mG 49.17%
100 mG 37.76 mG 50.00 mG 32.40%
38.84 mG 50.00 mG 28.72%
100 mG 16.18 mG 20.00 mG 23.57%
14.03 mG 20.00 mG 42.58%
100 mG 7.55 mG 10.00 mG 32.40%
7.55 mG 10.00 mG 32.40%

36.49% AVG % ERROR
METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE
Date Time Temperature Meter Background
Mar-14-91 01:11PM 64° F 0.35 mG
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
75.53 mG 32.40%
100 mG 75.53 mG mG 32.40%
31.83 mG mG 57.08%
31.56 mG mG 58.42%
13.38 mG 0 mG 49.48%
13.49 mG 0.00 mG 48.29%
7.12 mG 10.00 me 40.42%
7.34 0G 10.00 mG 36.29%
4.01 mG 5.00 mG 24.74%
3.98 mG 5.00 mG 25.75%
1.56 mG 2.00 mG 28.27%
1.60 mG 2.00 mG 25.26%
0.69 mG 1.00 mG 44.81%
0.65 mG 1.00 mG 51.93%
0.25 mG 0.50 mG 93.08%
0.20 me 0.50 mG 150.48%
49.96% AVG % ERROR

Appendix 1A1


ve

10 -

Percent Error

Accuracy and Linearity - Meter IXB
HOLADAY HI-3604 SN 76112

LR TT TTT TITY TT TTT TT TTrTrT

1 10 10? 10°
Applied Field (mG)


WRI - Model 660 - SN G1242
FRI NCY_RESP( PRIMARY DATA

Date Time Temperature Meter Background
Jan-26-91 10:50AM 72° F 0.5 mG

FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

THz 100 mG 927.12 mG 1.50 mG +55.82 dB

2 Hz 100 mG 927.12 mG 3.00 mG 49,80 dB

5 kz 100 mG 927.12 mG 21.00 mG 32.90 dB

10 Hz 100 mG 927.12 mG 100.00 mG 719.34 dB

20 Hz 100 mG 291.46 mG 100.00 mG 9.29 cB

50 Hz 100 mG 102.24 mG 100.00 mG +0.19 cB

60 Hz 100 mG 85.89 mG 100.00 mG 1.32 cB

100 Kz 100 mG 52.58 mG 100.00 mG 5.58 dB

200 kz 100 mG 38.62 mG 100.00 mG 8.26 dB

500 Hz 100 mG 41.78 mG 100.00 mG 7.58 dB

1 kHz 100 mG 65.28 mG 100.00 mG 3.70 dB

2 kHz 100 mG 125.29 mG 100.00 mG 1.96 dB

5 kkz 100 mG 278.74 mG 100.00 mG +8.90 dB

10 kHz 100 mG 495.08 mG 100.00 mG 713.89 dB

FREQUENCY RESPONCE AUXILIARY DATA
Date Time Temperature Meter Background
dan-24-91 10:50AM 72° F 0.5

FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

30 Hz 100 mG 174.76 mG 100.00 mG 74.85 dB

40 Hz 100 mG 124.29 mG 100.00 mG 1,89 dB

300 Hz 100 mG 38.43 mG 100.00 mG 8.31 dB

5 Hz 100 mG 967.54 mG 20.00 mG +33.69 dB

Appendix 1A2


WALKER - ELF-50D - SN 71181-2046

MINIMUM DETECTABLE SIGNAL

Date Time

Mar-29-91 11:03AM

Temperature
80° F

Threshold Current ma Threshold Field

1 51.00 ma 2751.45
Trial 2 53.00 ma 2859.35,
Number 3 50.00 ma 2697.50
4 50.00 m 2697.50
5 51.00 ma 2751.45
NOTE: Meter reading increses 1mG 2751.45
METER RESOLUTION DATA
Date Time Temperature
Mar-29-91 11:07AM 76° F
Meter Range Auto Range - 100mG

Drive Current Start

1 1.9450

Trial 2 1.9450
Number 3 1.9450
4 1.9450

5 1.9450

NOTE: Meter reading increses 1mG

Threshold Current

1.9610
1.9610
1.9620
1.9630

A
A
A
A
A
A 1.9660

Appendix XA

65655 & 83.

5

x3

Meter Background

0
0

m
ms

AVG THRESHOLD FOR SENSITIVITY

Meter Background

°°

Resolution

863.19
863.19
917.13
971.08
1132.93

949.50

mm
mG

uG
uG
uG

uG

Resolution %

0.82%

0.90%


MONITOR - MODEL 42b-1 - SN 1287

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Frequency Responce Setting - LINEAR MODE

Date Time Temperature Meter Background

Mar-26-91 12:11PM 60° F 0.011 mG
RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR
100.0 mG 78.66 mG 80.00 mG 1.70%
50.0 m6 39.28 mG 40.00 mG 1.84%
25.0 mG 19.48 mG 20.00 mG 2.69%
10.0 mG 7.75 mG 8.00 mG 3.19%
5.0 mG 3,88 mG 4.00 mG 3.12%
2.5 mG 1.96 mG 2.00 mG 2.98%
1.0 mG 0.77 mG 0.80 mG 2.90%
0.5 mG 0.39 mG 0.40 mG 2.69%

2.64% AVG % ERROR
Frequency Responce Setting - FLAT MODE
Date Time Temperature Meter Background
Mar-26-91 12:20PM 60° F 0.008 mG
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
100.0 mG 78.66 mG 80.00 mG 4.70%
50.0 mG 39.33 mG 40.00 mG 1.70%
25.0 mG 19.58 mG 20.00 mG 2.42%
10.0 mG 7.79 mG 8.00 mG 2.69%
5.0 mG 3.88 mG 4.00 mG 2.98%
2.5 mG 1.95 mG 2.00 mG 2.63%
1.0 mG 0.77 mG 0.80 mG 3.12%
0.5 mG 0.39 mG 0.40 mG 2.12%
2.38% AVG % ERROR

Appendix VIB


SIt

Responce

Lor

08 5

a6-

a4-

a2-

0a0-

10

Angular Responce - Meter XVIB
MEDA Model PLM-100 SN 1381

20 30 40-50 60 70

Rotation Angle

90

-+ Meter Responce
— Cosine Curve


HOLADAY - HI 3604 - SN 76112
FREQUENCY RESPONCE PRIMARY DATA

Date Time Temperature Meter Background

Feb-14-91 12:52PM 76° F 0.188 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

1 Wz 2 mG 1006.66 me 0.23 mG

2 hz 2 0G 1006.66 mG 0.27. mG

5 Hz 2. mG 1006.66 mG 0.59 mG

10 Hz 20 mG 1006.66 mG 6.9% mG

20 Hz 200 mG 421.11 mG 150.20 mG

50 Hz 200 mG 148.88 mG 150.00 mG

60 Hz 200 mG 148.88 mG 150.00 mG

100 Hz 200 mG 151.09 mG 150.00 mG

200 Hz 200 mG 152.43 mG 150.10 mG

500 Hz 200 mG 156.42 mG 150.10 mG

1 kuz 200 mG 164.06 mG 150.10 mG

2 kz 200 mG 256.09 mG 150.10 mG

5 kz 200 mG 897.76 mG 69.30 mG

10 kHz 20 mG 582.64 mG 4.68 mG

FREQUENCY RESPONCE AUXILIARY DATA

Date Time Temperature Meter Background

Feb- 14-91 01:17PM 76° F 0.169 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED’ FIELD RELATIVE RESPONCE
15 Hz 200 mG 1004.62 m6 88.50 mG -21.10 dB
30 Hz 200 mG 173.20 m6 150.10 mG -26 dB
40 Hz 200 m6 151.47 mG 150.10 mG +0.07 @B
3 kz 200 m6 526.65 mG 150.00 mG +10.91 dB

t t

Appendix 1X8


EXP - ELF - MODEL 1A - SN 1104
METER CODE MANUFACTURER METER NAME MODEL # SERIAL #
TB Expan Test, Inc. ELF Sense 4A 1104

232 St. John St.,ste 316
Portland, ME 04102

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Date Time Temperature Meter Background- mG
Feb-05-91 09:24AM 76° F 0.1
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
1000 m6 483.81 mG 500.00 m6" 3.35%
267.42 mG 250.00 mG ~ 1.04%
147.69 mG 150.00 mG 1.56%
100 mG 72.48 mG 75.00 mG 3.48%
47.84 mG 50.00 mG 4.52%
30 mG 19.05 mG 20.00 mG 4.98%
9.71 mG 10.00 mG 2.97%
100 mG 72.83 mG 75.00 mG 2.98%
47.80 mG 50.00 mG 4.60%
30 mG 19.21 mG 20.00 mG 4.13%
9.67 mG 10.00 mG 3.42%
3.37% AVG % ERROR
TER ACCURACY AND LINEARITY AT 60Hz - TEST 1 MOT
Date Time Temperature Meter Background
Mar-18-91 01:33PM 67° F 0.05 TM
RANGE SETTING ACTUAL FIELO MEASURED FIELD % ERROR
98.73 mG 100.00 ma 1.29%
100 m6 98.73 mG 100.09 me 1.29%
48.56 mG 50.00 mG 2.98%
48.02 mG 50.00 me 4.13%
29.78 mG 30.00 mG 0.73%
30 mG 29.78 mG 30.00 mG 0.73%
14.24 mG 15.00 mG 5.32%
16.35 mG 15.00 mG 4.52%
9.98 mG 10.00 mG 0.19%
10 mG 9.93 mG 10.00 mG 0.73%
4.73 1G 5.00 m6 5.80%
4.69 mG 5.00 mG 6.65%
2.96 mG 3.00 mG 1.47%
3 mG 2.93 mG 3.00 mG 2.22%
1.43 mG 1.50 mG 4.92%
1.41 mG 1.50 mG 6.12%
3.07% AVG % ERROR

Appendix IIIB


HOLADAY - HI 3604 - SN 76112

METER CODE MANUFACTURER

1X8 Holaday Industries
14825 Martin Drive
Eden Prairie, MN 55344

METER NAME
ELF FIELD

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Date Time

Feb- 14-91 12:33PM

RANGE SETTING ACTUAL FIELD

2000 mG 936.74 mG
461.90 mG

200 mG 148.88 mG
49.09 mG

200 mc 24.17 mG
14.78 mG

20 mG 9.83 mG
4.88 mG

Temperature
76° F

MEASURED FIELD

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date Time

Mar-28-91 11:15AM

RANGE SETTING ACTUAL FIELD

Auto

Temperature
70° F

MEASURED FIELD

100.00 mG
100.00 mG
10.00 mG
10.01
1.01
1.01

ES)

Appendix XB

MODEL #
HI-3604

SERIAL #
76112

Meter Background- mG

0.172

‘% ERROR

6.75%
8.25%
0.75%
1.85%
3.02%
1.47%
1.68%
2.52%

3.29%

Meter Background

0.135

‘% ERROR

+1.56%
71.41%
70.56%
0.82%
0.90%
1.12%

70.38%

mG

AVG % ERROR

AVG % ERROR


EMD - MODEL ACGM-1 - SN 01-0117

METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date Time Temperature

Feb-25-91 02:36PM 78° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.0 kV/M +-5%
E Field OFF Reading(mG) E Field ON Reading(mG)

Meter Range Meter Range
Auto Auto
1 0.10 0.10
Trial 2 0.10 0.10
Number 3 0.10 0.10
4 0.10 0.10
5 0.10 0.10
0.10 AVG 0.10

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA ~ UNMODULATED

Date Time Temperature
Feb-22-91 11:32AM 79°F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Auto Auto
1 Mhz 0.10 0.10
10 MHz 0.10 0.10
100 MHz 0.10 8.15

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date Time Temperature
Feb-22-91 79° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Auto ‘Auto
1 MHZ 0.10 0.10
10 MHz 0.10 0.10
100 MKz 0.10 25.70

Appendix VITIB


WALKER - ELF-50D - SN 71181-205

MINIMUM DETECTABLE SIGNAL

Date

Mar-29-91

Trial
Number

wnune

Time

10:47AM

Temperature
75° F

Threshold Current mA Threshold Field

NOTE: Meter reading increased by 1m

METER RESOLUTION DATA

Date

Mar-29-91

Meter Range

Trial
Number

vinuna

NOTE: Meter reading

29.00 ma 1564.55
30.00 ma 1618.50
29.00 ma 1564.55
30.00 ma 1618.50
30.00 ma 1618.50
1596.92
Time Temperature
10:51AM TSF
Lo

Drive Current Start

1.9400
1.9400
1.9400
1.9400
1.9400

increased by 1nG

A Threshold Current

1.9600
1.9580
1.9590
1.9580
1.9560

>> >e>

Appendix xX

min
max

Meter Background

mG
ms

AVG THRESHOLD FOR SENSITIVITY

Meter Background

Ly)
)

Resolution

1078.98
971.08
1025.03
971.08
863.19

981.87

mG
mG

uG

uG
uG
uG
uG
uG

Resolution %

1.03%
0.92%
0.97%
0.92%
0.82%

0.93%


MONITOR - MODEL 42b-1 - SN 1286

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Frequency Responce Setting - LINEAR MODE

Date Time Temperature Meter Background
Feb-28-91 12:05PM 74° F 0.25 mG
SQUARE WAVE TESTING
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR RELATIVE RESPONCE
250 mG 41.38 m6 150.00 mG 262.48% 44.19 oB
41.55 mG 150.00 m6 260.97% + 14.15 8
261.73% 14.17 db
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR RELATIVE RESPONCE
250 mG 72.03 mG 150.00 mG 108.26% 6.37 dB
72.43 mG 150.00 mG 107.10% 6.32 aB
107.67% 6.35 db
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR RELATIVE RESPONCE
250 mG 148.88 mG 150.00 mG 0.75% 0.06 dB
148.83 mG 150.00 mG 0.78% 0.06 dB
0.76%
0.06 db
Frequency Responce Setting - FLAT MODE
Date Time Temperature Meter Background
Feb-28-91 11:578M 74° F 0.12 mG
SQUARE WAVE TESTING
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR RELATIVE RESPONCE
250 mG 148.36 m6 150.00 mG 1.12% 0.09 4B
149.11 mG 150.00 m6 0.60% 0.05 aB
0.86% 0.07 db
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR RELATIVE RESPONCE
250 mG 148.11 mG 150.00 m6 1.27% 0.11 aB
149.06 mG 150.00 m6 0.63% 0.05 dB
0.95% 0.08 db
RANGE SETTING ACTUAL FIELD MEASURED FIELD X ERROR RELATIVE RESPONCE
250 mG 148.61 mG 150.00 mG 0.93% 0.08 4B
148.73 mG 150.00 mG 0.85% 0.07 dB
0.89% 0.07 db

Appendix VIA


9€L

OB MazZOVvOMD

md

Susceptibility - Meter VIIB
SYDKRAFT Model 88-05

oe eee!

60 Hz
E-FIELD 1eKv At

4 MHz «10 MHz 100 MHz
Genz MOD

CRIED OFF
FELD ON


WRI - Model 660 - SN G1242
METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date Time Temperature
Feb-27-91 01:10PM 78° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10 KV/M +-5%
E Field OFF Reading(mG) E Field ON Reading(m
Meter Range Meter Range
Full Full
1 0.15 0.95
Trial 2 0.15 0.95
Number 3 0.15 0.95
4 0.15 0.95
5 0.15 0.95
0.15 AVG 0.95

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMOOULATED

Date Time Temperature
Feb-20-91 02:04PM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Full Full
1 MHz 0.10 0.10
10 MHz 0.10 0.10
100 MHz 0.10 0.10
RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hi: ATED
Date Time Temperature
Feb-20-91 02:24PM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Full Full
1 MHz 0.10 0.10
10 MHz 0.10 0.10
100 MHz 0.10 5.00

Appendix 1A2


DINDIMA - GAUSSMAUS - SN GM0400

FREQUENCY RESPONCE AUXILIARY DATA

Frequency Responce Setting - WIDE BAND

NO AUX TESTING REQ

Frequency Responce Setting - BAND PAS

Date Time . Temperature Meter Background
May- 16-91 11:40AM 76° F 1 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
30 Hz 99 mG 446.89 mG 31.00 mG +23.18 dB
40 Hz 99 mG 318.11 mG 43.00 mG 17.38 dB
70 Hz 99 mG 171.31 mG 99.00 mG 4.76 dB
80 Hz 99 mG 417.34 mG 68.00 mG +15.76 dB
74 HZ 99 mG 455.32 mG 89.00 mG -14.18 dB
51 Hz 99 mG 391.77 mG 87.00 mG +13.07 dB
52 Hz 99 mG 378.42 mG 94.00 mG 12.10 dB
53 Hz 99 mG 284.49 mG 99.00 mG -9.17 dB
54 Hz 99 mG 197.77 mG 99.00 mG +6.01 dB
55 Hz 9 mG 158.21 mG 99.00 mG -4,07 dB
56 Hz 99 mG 147.78 mG 99.00 mG -3.48 dB
57 Hz 99 mG 129.69 mG 99.00 mG +2.35 dB
58 Hz 99 mG 121.05 mG 99.00 mG “1.75 dB
59 Hz 99 mG 113.78 mG 99.00 mG “1.21 dB
60 Hz 99 mG 115.43 mG 99.00 mG +1,33 dB
61 Kz 99 mG 107.59 mG 99.00 mG -0.72 dB
62 Hz 99 mG 106.65 mG 99.00 mG -0.64 dB
63 Hz 99 mG 108.93 mG 99.00 mG -0.83 dB
64 Hz 99 mG 108.89 mG 99.00 mG 0.82 dB
65 Hz 99 mG 116.02 mG 99.00 mG 1,38 dB
66 Hz 99 mG 119.96 mG 99.00 mG 1,67 dB
67 Hz 99 mG 127.55 mG 99.00 mG -0.72 dB
68 Hz 99 mG 135.34 mG 99.00 mG -0.64 dB
69 Hz 99 mG 168.93 mG 99.00 mG -0.83 dB

‘Appendix XVA


” 9g

40

20

-20

Responce (dB)

-40 4

- (Square Wave

Frequency Responce - Meter IVB A Sautooth wave
MSI Model MSI-20 SN 79010854 + Sine Wave

— Bezier Fit

TTT TTT TTT TTT TTT TTT] TT TTT]

10 10? 107 1o*
Frequency (Hz)


ell

Responce

Lor

08 —

06-

a4—-

a2 -

0.0

Angular Responce - Meter XVB - Wide Band
DINDIMA - GAUSSMAUS - SN GM0401

10

20

30

40 50

Rotation Angle —

60

70

80

90

+

Meter Responce

Cosine Curve


92

Responce (dB)

- - (Square Wave
Frequency Responce - Meter XVA - Band Pass J Sastooth wave
DINDIMA GAUSSMAUS SN GM0400 + Sine Wave
40 — — Bezier Fit
20 4
o-
-20 4
-49 4
-60 T T1117 T ToT T TT T1117]
1 10 10? 107

Frequency (Hz)


MEDA - Model PLM-100 - SN 1381
MINIMUM DETECTABLE SIGNAL

Date Time Temperature Meter Background
Aug-29-91 12:47PM 9° F min 0.02 mG
max 0.03 mG
Threshold Current m4 = Threshold Field = uG
1 1.41 mm 76.07 uG
Trial 2 1.44 mA 77.69 uG
Number 3 1.45 mA 78.23 uG
4 1.50 ma 80.93 uG
5 1.50 ma 80.93 uG
NOTE: Meter readings increase by 0.5 mG 78.77 uG AVG THRESHOLD FOR SENSITIVITY
METER RESOLUTION DATA
Date ~ Time Temperature Meter Background
Aug: 29-914 12:40PM 99° F min 0.03 mG
max 0.03 mG
Meter Range 20 mG
Drive Current Start A Threshold Current A Resolution uG Resolution %
1 0.1863 A 0.18649 A 10.25 uG 0.10%
Trial 2 0.1863 A 0.18657 A 14.57 uG 0.14%
Number 3 0.1863 A 0.18656 A 14.03 uG 0.14%
4 0.1863 A 0.18656 A 14,03 uG 0.14%
5 0.1863 A 0.18659 A 15.65 uG 0.15%
NOTE: Meter readings increase by 0.015 mG 13.70 uG 0.13%

Appendix XVIB


82

10-

Percent Error

Accuracy and Linearity - Meter VIB - Flat Mode
MONITOR: .Model 42b-1 SN 1287

TTT TTI] TTT TTT I T

! 10 10?

T TTT TT]
107
Applied Field (mG)


+6

Responce

10
ae
a6
a4
a2

a0 -

10

Angular Responce - Meter VA

20

IRC Model IRC-109

30

40 50

Rotation Angle

60

70

90

+

Meter Responce
Cosine Curve


E FIELD SUSCEPTIBILITY TEST SETUP

SCHEMATIC DIAGRAM
ELECTRIC FIELD

10 kV/mm
60 HZ
HP 33144
FUNTION
GENERATOR TRANSFORMER
‘ HI VOLT AGE
RED ° 5000 Vrms
CROWN M600 but ‘
AMPLIFIER RED osm
FOAM ce : |
\_BLOCK

PARALLEL PLATES

1m DIAMETER CIRCULAR
ALUMINUM DISKS

HP 3456A HI] @
DIGITAL VOLTMETER Lo 4.460 Vrms



il

_ _— We : (1 Square Wave
Frequency Responce - Meter XVB - Wide Band 7 Saitooth Wave
DINDIMA GAUSSMAUS SN GMO401 + Sine Wave
40 — — Bezier Fit
20 4
a5 4
8 oe /
8
2 -20- +
e
&
-49 4
-60 T T Terry | T T Le | T T VT VETTy
1 10 10? 10? 104

Frequency (Hz)


Meter Angular Response Measurements

The meters were normally tested with the coil axis oriented with the
magnetic field vector (for maximum coupling). In this test the meter, was
incrementally rotated off axis until it was oriented perpendicular to the
magnetic field vector. One expects to see a cosine type of dependence of the
meter reading vs. angle. The meter was rotated in 10° increments from its
normal position of 0°, to 90° off axis. At each angular position the meter
reading was recorded.

The graphs provided on the following pages show the angular response
normalized to the meter’s zero degree response versus the theoretical cosine
response. The formulas used to calculate these data points were as follows:

1) ACTUAL_FIELD = ACTUAL_V / .0004026
2] ANGULAR_RESPONSE = (ACTUAL_ZERO_DEGREE_FIELD / ACTUAL_FIELD) *
(MEASURED_FIELD / MEASURED_ZERO_DEGREE_FIELD)

83


MONITOR - MODEL 42b-1 - SN 1286
METER 60Hz E FIELD SUSCEPTIBILITY DATA

Frequency Responce Setting - LINEAR MODE

Date Time Temperature

Feb-28-91 01:53PM 74° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.0 kV/M +-5%
E Field OFF Reading(mG) E Field ON Reading(mG)

Meter Range Meter Range
0.581 0.5*1
1 0.29 0.29
Triat 2 0.29 0.29
Number 3 0.29 0.29
4 0.29 0.29
5 0.29 0.29
0.29 AVG 0.29

Frequency Responce Setting - FLAT MODE

Date Time Temperature

Feb- 28-91 01:40PM Th? F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.0 kV/M +-5%
E Field OFF Reading(m) E Field ON Reading(mG)

Meter Range Meter Range
O.5*1 0.5*1
1 0.12 0.12
Trial 2 0.12 0.12
Number 3 0.12 0.12
4 0.12 0.12
5 0.12 0.12
0.12 AVG 0.12

Appendix VIA


Magnetic Tech - Model ELF 12 SN 4

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date Time

May-14-91 11:40AM

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD
Auto mG 41.58 mG
45.65 mG

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD
Auto mG 54.17 mG
49.01 mG

SINE WAVE TESTING

RANGE SETTING ACTUAL FIELD
Auto mG 94.36 mG
93.84 mG

METER ANGULAR RESPONCE DATA

Date Time
May-14-91 12:34PM
Angle ACTUAL FIELD
or 96.99 mG
10° 96.94 mG
20° 102.09 mG
30° 109.61 mG
40° 125.93 mG
50° 156.11 mG
60° 203.40 mG
70° 300.55 mG
80° 669.03 mG
90° 1092.47 mG

Temperature
77° F

MEASURED FIELD

96.00 mG
96.00 mG

MEASURED FIELD

96.00 mG
96.00 mG

MEASURED FIELD

96.00 mG
96.00 mG

Temperature
77°F

MEASURED FIE!

8eseee0o000
ssssssssss

SFLELLELES

w
8

Appendix XVIIB

LD

332882238383

Meter Background

2 mG

% ERROR

130.88%
110.28%

120.58%

% ERROR

77.21%
95.89%

86.55%

‘% ERROR

1.74%
2.30%

2.02%

Meter Background
2 mG

METER RESPONCE

RELATIVE RESPONCE

7.27 dB
6.46 dB

6.86 db

RELATIVE RESPONCE

4.97 oB
5.84 dB

5.41 db

RELATIVE RESPONCE

0.15 dB
0.19 dB

0.17 db


liz
MA
HIB
IVA
VB
VA
VB
VIA
VIB
VIIA
VIB
VINA
VIB
IXA
XB
x
XA

, XIVA
XIVB
XVA
XVB
XVIA
XVIB
XVIIA
XVIIB

MANUFACTURER

Widerange Instruments’
Widerange Instruments‘
Widerange Instruments'
Widerange Instruments'
Safe Computing Company
Sate Computing Company
Expan Test iInc.'

€xpan Test, Inc.’

Magnetic Sciences Int'l
Magnetic Sciences Int'l’
Intergrity Research Corp.
Intergrity Research Corp.'
Monitor Industries‘

Monitor Industries'

Sydkraft

Sydkraft*

Electromagnetics Design”
Electromagnetics Design” :
Holaday Industries*

Holaday Industries

Walker Scientific Inc.”
Walker Scientific Inc.”
Electric Field Measurements'
Electric Field Measurements'
The Dindima Group*

The Dindima Group'
Macintyre Electronic Design'
Macintyre Electronic Design’
Magnetic Technology‘
Magnetic Technology’

METER NAME

Megalert 2
Megalert 2
Megalert
Megalert

Pro Meter
Pro Meter
ELF Sense
ELF Sense
MSI, MM
MSI,LMM

60 Hz H Field
60 Hz H Field
Milligauss
Milligauss

H Flux

H Flux

* Gaussmeter

Gaussmeter
ELF Field
ELF Field
ELF Field
ELF Field
Metex Digital
Metex Digital
Gaussmaus
Gaussmaus

NOTE: Superscript “jo” means instrument was purchased
by EPA; superscript "f means instrument was
Joaned to EPA by the manufacturer.

TABLE 1

5

MODEL #

670
670
660
660

1A

1A

MSI-20
MSI-20
IRC-109
IRC-27
42B-1
42B-1
88-05
88-05
ACGM-1
ACGM-1
HI-3604
HI-3604
ELF-50D
ELF-50D
Metex-3800
Metex-3800

PLM-100
PLM-100
ELF 12
ELF 12

SERIAL #

G1257-
G1259
G1241
G1242
1263
1264
1103
1104
79011626
79010854

1286
1287

01-0116
01-0117
76114
76112
71181-205
71181-204
542513
542832
GM0400
GM0401
1380

1381

2

4


Magnetic Tech - Model ELF 12 SN 4
METER CODE MANUFACTURER MODEL #

XVIIB Magnetic Technology ELF 12

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date Time Temperature Meter Background
‘Aug-29-91 14:05AM 95° F 2 m%

RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
96 mG 91.72 mG 96.00 mG 4.6%

96 mG 91.72 mG 96.00 me 4.67%

96 mG 36.69 mG 32.00 mG “12.7%

96 mG 36.69 mG 32.00 me “12.7%

96 mG 12.16 mG 12.00 mG =1.16%

96 me 12.14 mG 12.00 mG “1.16%

96 mG 3.67 mG 4.00 m6 9.03%

96 mG 3.61 mG 4.00 mG 10.66%

0.15%

FREQUENCY RESPONCE PRIMARY DATA

Date Time Temperature Meter Background
May- 14-91 11:14AM 77° F 2 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD

Vaz Auto mG 599.60 mG 2.00 mG

2 Hz Auto mG 599.60 mG 16.00 mG

5 Hz Auto m6 599.60 mG 64.00 mG

10 Hz Auto mG 599.60 mG 96.00 mG

20 Hz Auto mG 276.86 mG 96.00 mG

50 Hz Auto mG 109.69 mG 96.00 mG

60 Hz Auto mG 97.86 mG 96.00 mG

100 Hz Auto mG 62.39 mG 96.00 mG

200 Hz Auto mG 44.16 mG 96.00 mG

500 Hz Auto mG 35.54 mG 96.00 mG

1 kHz Auto mG 36.49 mG 96.00 mG

2 kHz Auto mG 43.17 mG 96.00 mG

5 kHz Auto mG 57.48 mG 96.00 mG

10 kHz Auto mG 65.97 mG 96.00 mG

FREQUENCY RESPONCE AUXILIARY DATA
Date Time Temperature , Meter Background
May- 14-91 11:33AM 77 F 2 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD

30 Hz Auto mG 181.89 mG 96.00 mG

80 Hz Auto mG 74.42 mG 96.00 mG

300 Hz Auto mG 38.08 mG 96.00 mG

15 Hz Auto mG 367.61 mG 96.00 mG

Appendix  XVIIB

SERIAL #
4

AVG % ERROR

RELATIVE RESPONCE

749.54 cB
731.47 GB
=19.43 dB
“15.91 cB
14 dB
+16 dB
16 dB

RERSaRN
SS38eaa

RELATIVE RESPONCE


Ut

Responce

Loo

4a8-

06 -

a4-

42-

a8 -

Angular Responce - Meter XVA - Band Pass
DINDIMA ~ GAUSSMAUS ~ SN GM0400

10

ao

30

40 50

Rotation Angle

60

70

80

90

+

Meter Responce

Cosine Curve


18

40 =

20 +

Responce (dB)
i
g
l

Square Wave

Frequency Responce - Meter XVIIA O
A Savtooth Wave
MAG TEC Model ELF 12 SN 2 + Sine Wave
— Bezier Fit

TT TTT TY Lr TTT TTT TTT TTT]

10 10? 107 10°
Frequency (Hz)


SAFE - SAFE PRO METER - SN 1264

MINIMUM DETECTABLE SIGNAL

Date Time Temperature

Mar-18-91 11:29AM or F min
max

Threshold Current mA Threshold Field uG

1 3.40 ma 183.43 uG

Trial 2 2.60 ma 140.27 uG
Number 3 2.30 mA 124.09 uG
4 2.70 mA 145.67 uG

5 2.50 ma 136.88 us

NOTE: Meter Reads 0.2mG at Threshold

METER RESOLUTION DATA

Date Time

Mar-18-91 11:35AM

Meter Range Auto Range - 200mG

Drive Current Start

1 1.7500
Trial 2 1.7500
Number 3 1.7500
4 1.7500
5 1.7500

NOTE: Threshold Reading is about 0.8mG Higher

0.1
0.1

Meter Background

mG

mm

145.67 uG AVG THRESHOLD FOR SENSITIVITY

Temperature

A Threshold Current

1.7630
1.7600
1.7590
1.7600
“1.7580

>>>
>P>r>

Appendix 112

a
0.2

A Resolution

701.34
539.49
485.54
539.49
431.59

539.49

Meter Background

m
ms

685555 &

&

Resolution %

0.74%
0.57%
0.51%
0.57%
0.45%

0.57%


£01,

Responce

10

a8 -

a6 -

a4-

42-7

a0-

10

Angular Responce - Meter x

WALKER

20

ELF-50D

30 40 50

' * Rotation Angle

SN 71181-205

60

70

80

+

Meter Responce
Cosine Curve


26

Responce

Angular Responce ~ Meter IVA

MSI Model MSI-20 SN 79011626 “F  Meter Responce
Lo- . . == Cosine Curve

a8 -
a6-
a4-

a2-

a0 PAA A I

oO 10 2 30 40,50 60 70. 80 9

Rotation Angle


SAFE - SAFE PRO METER - SN 1263
FREQUENCY RESPONCE PRIMARY DATA

Date Time Temperature Meter Background
Feb-05-91 01:13PM 76° F 0.2 7m

FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

10 Hz 200 mG 766.79 mG 150.50 mG 14.16 cB

20 Hz 200 mG 238.00 mG 155.00 mG 73 dB

50 Hz 200 mG 143.86 mG 150.50 mG 0.39 dB

60 Hz 200 mG 139.34 mG 150.30 mG 0.65 dB

100 Hz 200 mG 133.93 mG 150.40 mG 1.01 cB

200 Hz 200 mG 138.67 mG 150.20 mG 0.69 dB

500 Hz 200 mG 199.33 mG 150.10 mG 72.46 dB

1 kHz 200 mG 431.17 mG 49.00 mG +18.89 dB

2 kHz 200 mG 1097.69 mG 10.20 mG 40.64 oB

5 kHz 200 mG 870.42 mG 7.80 mG 40.95 dB

10 kHz 200 m6 576.63 mG 4.80 mG 41.59 dB

FREQUENCY RESPONCE AUXILIARY D/

Date Time Temperature Meter Background
Feb-05-91 01:53PM 76° F 0.2 mG

FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

45 Hz 200 mG 147.47 mG 150.00 mG 0.14 dB

90 Hz 200 mG 134.00 mG 149.90 mG 0.97 dB

120 Hz 200 mG 133.66 mG 150.10 mG 1.01 dB

80 Hz 200 mG 135.22 mG 150.20 mG 0.91 dB

90 Hz 200 mG 134.45 mG 150.30 mG 0.96 dB

300 Hz 200 mG 153.15 mG 150.10 mG 70.17 dB

700 Hz 200 mG 779.73 mG 108.50 mG “17.13 dB

500 Hz 200 mG 214.08 mG 150.10 mG 73.08 dB

300 Hz 200 mG 153.68 mG 150.10 mG 0.20 oB

Appendix 111


SAFE - SAFE PRO METER - SN 1263

MINIMUM DETECTABLE SIGNAL

Date Time Temperature Meter Background
Mar-18-91 11:58AM 64° F min 0.2 mG
max 0.2 mG

Threshold Current mA Threshold Field uG

1 3.40 ma 183.43 uG
Trial 2 3.40 mA 183.43 uG
Number 3 3.40 ma 183.43 uG
4 3.50 ma 188.83 uG
5 3.60 ma 194.22 uG
186.67 UG AVG THRESHOLD FOR SENSITIVITY
NOTE: Meter moves from 0.2mG 0.3mG
METER RESOLUTION DATA
Date Time Temperature Meter Background
Mar-18-91 12:05PM 64° F min 0.2 mG
max 0.2 mG
Meter Range Auto Range - 200mG
Drive Current Start A Threshold Current A Resolution uG Resolution %
1 1.7200 A 1.7370 A 917.13 uG 0.98%
Trial 2 1.7200 A 1.7340 A 755.29 uG 0.81%
Number 3 1.7200 A 1.7340 A 755.29 uG * 0.81%
4 1.7200 A 1.7350 A 809.24 uG 0.87%
5 1.7200 A 1.7310 A 593.44 uG 0.64%
NOTE: Threshold reading about 0.9mG Higher 766.08 uG 0.82%

Appendix 111


09

Responce (dB)

40 -

20 4

8
!

1
g
i

-40 4

- (1 Square wave
Frequency Responce - Meter VB D. Saitooth vave
IRC Model IRC-27 ++ Sine Wave
; — Bezier Fit

TTT TIT TT TTT A |

10 10? ; 107 104

‘Frequency (Hz)


95°

Responce: (dB)

40 -

20 +

-20 +

-40 4

Frequency Responce - Meter IIIB

EXPAN

Model 1A SN 1104

TTT TTT T

10

TOTTI TT TTT TTT T1117],

10? 10° 104

Frequency (Hz)

1+bO0

Square Wave
Sawtooth Wave
Sine Wave,
Bezier Fit


81

Accuracy and Linearity - Meter IIe
SAFE PRO METER SN 1264 -

10 -
5 -— + ++
&
& 04
»
&
-5 4
-10 TT TTT TTT TTT TTT TTI TT T1117}
1g? v7 10 10? 107

Applied Field (mG)


MONITOR - MODEL 42b-1 - SN 1287
METER 60Hz E FIELD SUSCEPTIBILITY DATA

Frequency Responce Setting - LINEAR MODE

Date Time Temperature

Feb-27-91 08:14AM T3° F
€ Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.0 kV/M +-5%
E Field OFF Reading(mG) E Field ON Reading(mG)

Meter Range Meter Range
0.5*1 0.5*1
1 0.25 0.25
Trial 2 0.25 0.25
Number 3 0.25 0.25
4 0.25 0.25
5 0.25 0.25
0.25 AVG 0.25

Frequency Responce Setting - FLAT MODE

Date Time Temperature
Feb-27-91 08:25AM 73° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.0 kV/M +-5%
E Field OFF Reading(mG) E Field ON Reading(me)
Meter Range Meter Range
0.5*1 0.5*1
1 ot 0.11
Trial 2 o.11 Ot
Number 3 0.11 0.11
4 O11 0.11
5 0.11 o.11
o.1t AVG 0.11

‘Appendix VIB


Meter Frequency Response Measurements

The magnetic field meters were tested using the Helmholtz coil up to
its full operational bandwidth (10kHz). The low end starting frequency was
1Hz. The general approach was to test each meter using two frequency sweeps
of the Helmholtz coil. The first sweep provides the primary data set,. which
was designed to detect gross behaviors, resonant frequencies, and regions of
special interest. The second pass consisted of taking data at discreet
frequencies of special interest throughout the band 1Hz - 10kHz. The
standard data set for the first pass was a log sweep throughout the test band
(1Hz, 2Hz, 5Hz, ... 5kHz, 1OkHz). If the first sweep identified resonances
or areas of special concern then the density of data points in the second
sweep was locally increased. It was the test operators judgement as to how
many data points to add in areas of special interest.

One Graph representing frequency response was created for each meter
tested. These graphs, included on the following pages, combine all data
points from both passes. Any data points with frequency values within 10% of
each other were combined as one average data point. The line drawn shows a
200 point bezier fit to the combined data set. In addition, results of the
non-sinusoidal tests, as described below, were included on these graphs. The
formulas used to calculate these data points were as follows using terms
described in the test plan (QAPP):

1] ACTUAL_FIELD = ACTUAL_V / .0004026
2] RESPONSE (dB) = 20 * LOG(MEASURED_FIELD / ACTUAL_FIELD)

Meter 60Hz Non-Sinusoidal Response Measurements

The meters were tested in the Helmholtz coil driven by square wave and
sawtooth waves at a repetition rate of 60 Hz. Both of these waveforms have
Fourier components beyond the fundamental frequency of 60Hz. The data from
the frequency response testing gives an indication of whether a given meter
will display augmented readings when driven by square waves or sawtooth waves
as apposed to sine waves. If the frequency response testing indicated that

47


WALKER - ELF-50D - SN 71181-
METER CODE MANUFACTURER: METER NAME
x Walker Scientific Inc. ELF Field
Rockdale St.

Worcester, MA 01606-9988

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Date Time Temperature

Feb-07-91 08:35AM The F

RANGE SETTING LO ACTUAL FIELD MEASURED FIELD

2000 mG 501.19 mG 500.00 me
250.52 mG 250.00 mG

2000 mG 151.89 mG 150.00 mG
76.61 mG 75.00 m6

2000 mG 26.60 mG 25.00 mG
11.89 mG 10.00 mG

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date Time Temperature

Mar-29-91 10:40AM 75° F

RANGE SETTING ACTUAL FIELD MEASURED FIELD

105.20 mG 100.00 mG

Lo 105.20 mG 100.00 mG
55.57 mG 50.00 mG

55.57 mG 50.00 mG

14.30 mG 10.00 mG

14.30 mG 10.00 mG

5.40 mG 3.00 mG

5.40 mG 3.00 mG

Appendix xX

205
MODEL #

ELF-50D

SERIAL #
71181-205

Meter Background- mG

4.28%

Meter Background

0 mG

‘% ERROR

4.95%
“4.95%
-10.02%
-10.02%
+30.05%

+22.35%

AVG % ERROR

AVG % ERROR


Magnetic Tech - Model ELF 12 SN 2

METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date

May-20-91

EFM Metex

Trial
Number

wnune

Time

09:59AM

E Field OFF Reading
0.00

E Field OFF Reading(mG)
Meter Range
Auto
2.00
2.00
2.00

2.00
2.00

2.00

AVG

Temperature
73° F

E Field ON Reading
10.00 KV/M +

E Field ON Reading(mG)
Meter Range
Auto
2.00
2.00
2.00

2.00
2.00

2.00

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Date

May-17-91

1 MHz
10 MHz
100 MRz

Time
01:34PH
RF OFF Reading(mG)
Meter Range
Auto
2.00

2.00
2.00

Temperature
77 F
RF ON Reading(mG)
Meter Range
Auto
2.00

0.00
2.00

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date

May-17-91

4 MHz
10 MHz
100 MHz

Time
01:55PM
RE OFF Reading(mG)
Meter Range
Auto
2.00

2.00
2.00

Temperature

77° F
RF ON Reading(mG)

Meter Range
Auto
12.00
0.00
2.00

Appendix XVIIA


1s

Responce (dB)

- (Square Wave
Frequency Responce - Meter JAl ‘A. Saitooth Wave
WRI Model 660 SN 61241 + Sine Wave
— Bezier Fit
40 -
20 4
0 -
-20 4
-40 4
-60 - LEELA DL ALL ee |
1 10 10? 10? 104

Frequency (Hz)


HOLADAY - HI 3604 - SN 76111

METER 60Hz_NON-SINUSOIDAL_RESPONCE DATA

Date Time

Feb-14-91 11327AM

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD

200 mG 146.94 mG

146.80 mG

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD

200 mG 148.44 mG

148.53 mG

SINE WAVE TESTING

RANGE SETTING ACTUAL FIELD

200 mG 145.13 mG

145.11 mG

METER ANGULAR RESPONCE DATA

Date Time
Feb-14-91 11:39AM
Angle ACTUAL FIELD
o° 145.08 mG
10° 146.94 mG
20° 153.63 mG
30° 166.22 mG
40° 186.86 mG
50° 221.96 mG
60° 292.67 mG
70° 403.35 mG
80° 879.26 mG
90° 1101.49 mG

Temperature Meter Background
75° F 0.179
MEASURED FIELD % ERROR
150.05 mG 2.11%
149.95 mG 2.15%
2.13%
MEASURED FIELD % ERROR
149.50 mG 0.71%
150.00 mG 0.98%
0.85%
MEASURED FIELD % ERROR
149.95 mG 3.32%
150.05 mG 3.41%
3.36%
Temperature Meter Background
76° F 0.184

MEASURED FIELD

BRRRRE
SSSSss

149.

Ra’assss
2RRRBRRBB

bel

Appendix  IXA

ANGULAR RESPONCE

RELATIVE RESPONCE

0.18 dB
0.18 dB

0.18 db

RELATIVE RESPONCE

0.06 dB
0.08 dB

0.07 db

RELATIVE RESPONCE

0.28 aB
0.29 dB

0.28 db


08

40

20 +

i)
al

Responce (dB)
1
8
|

-40 4

Frequency Responce - Meter XVIB 1 Square Wave
A Sawtooth Wave

MEDA Model PLM-100 SN 1381 “+ Sine Wave

. — Bezier Fit

TT TTT TT TTT TT TTT TTT TTT]

10 10? 107 10
Frequency (Hz)


DINDIMA - GAUSSMAUS - SN GM0401
MINIMUM DETECTABLE SIGNAL

Frequency Responce Setting - WIDE BAND

Date Time Temperature Meter Background
Mar-28-91 01:24PM 68° F min 0.1 mG
max 0.1 mG

Threshold Current m& = Threshold Field uG

1 6.20 mA 334.49 UG
Trial 2 6.30 mA 339.89 UG
Number 3 6.20 mA 334.49 uG
4 6.20 mA 334.49 UG
5 6,20 mA 334.49 uG
NOTE: Meter readings increase by 0.1 mG 335.57 uG AVG THRESHOLD FOR SENSITIVITY (RMS)
NOTE: This is Peak Measurement
METER RESOLUTION DATA
Frequency Responce Setting - WIDE BAND
Date Time Temperature Meter Background
Mar-28-91 01:29PM 65° F min 0.1 mTM
max 0.1 mG
Meter Range 99MG
Orive Current Start A Threshold Current A Resolution uG Resolution %
1 1.4200 A 1.4360 A 863.19 uG 1.13%
Trial 2 1.4250 A 1.4360 A 593.44 uG 0.77%
Number 3 1.4250 A 1.4350 A «539.49 uG 0.70%
4 1.4250 A 1.4340 A 485.54 uG 0.63%
5 1.4250 A 1.4340 A 485.54 uG 0.63%
NOTE: Meter readings increase by 1.0 mG (RMS) 593.44 uG 0.77%

NOTE: This is Peak Measurement

Appendix XVB


Magnetic Tech - Model ELF 12 SN 2

MINIMUM DETECTABLE SIGNAL

Date Time

Aug-29-91 10:36AM

Threshold Current mA

1 680.00 m
Trial 2 680.00 ma
Number 3 680.00 mA
4 610.00 ma
5 700.00 mA

NOTE: Meter readings increased by 30 mG

METER RESOLUTION DATA

Date. Time
Aug- 29-91 10:44AM
Meter Range 96 mG

Drive Current Start A

1 0.8900 A

Trial 2 0.8900 A
Number 3 0.8900 A
4 0.8900 A

5 0.8900 A

NOTE: Meter readings increased by 16 mG

Temperature
92° F

Threshold Field

36686.00
36686.00
36686.00
32909.50
37765 .00

36146.5

Temperature
93° F

Threshold Current

1.2500
1.2500
1.2500
1.2500
1.2500

Appendix XVIIA

min
max

>

PP r>>

Meter Background

2
2

im
mG

AVG THRESHOLD FOR SENSITIVITY

Meter Background

2
2

Resolution

19421.67
19421.67
19421.67
19421.67
19421.67

19421.67

mG
m

Resolution %

40.45%
40.45%
40.45%
40.45%
40.45%

40.45%


eet

as mMozZovUMD

Susceptibility - Meter XA
WALKER ELF-50D SN71181-204

(ORIEL OFF
FIELD ON

o om Om 3 osm of =

5)

60 Hz 4 HHz = 10 KHz 180 MHz MHz = 18 Miz 100 Miz
E-FIELD 10kVAt ‘UNMOD Gonz MOD


SEI

OB MOzZOTUMD

44

me

mn ee

Susceptibility - Meter VIIA
SYDKRAFT Model 88-05

Om

q

a

60 Hz
E-FIELD 1¢KV Ai

1 MHz

iz 160 MHz

4 titz

10 WHz 100 MHz
Genz MOD

OFELD OFF
FIELD ON


€1

Percent Error

Accuracy and Linearity - Meter Il

WRI Model 670 SN G1257
40 -
20 -

2 -

10 =

-20 } a or

107 1 10 10? 107
Applied Field (mG)


EFM - Metex-3800 - SN 542513
METER 60Hz € FIELD SUSCEPTIBILITY DATA

Date . Time Temperature

May-20-91 08:31AM TRF
E Field OFF Reading E Field ON Reading
EFH Metex 0.20 10.0 kV/M #-5%
E Field OFF Reading(mG) E Field ON Reading(mG)

Meter Range Meter Range
200 200
1 0.00 0.00
Trial 2 0.00 0.00
Number 3 0.00 0.00
4 0.00 0.00
5 0.00 0.00
0.00 AVG 0.00

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA ~ UNMODULATED

Date Time Temperature
May-17-91 12:54PM 77 F
RF OFF ‘Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
200 200
1 MHZ 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 0.00

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MOOULATED

“pate Time Temperature
May-17-91 01:05PM TF
RF OFF Reading(mG) RF ON Reading(ma)
Méter Range Meter Range
200 200
1 Miz 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 0.00

Appendix" XIVA


Let

os MaAZOVHMSD

244

18.

124

Susceptibility - Meter VIIIA
EMD Model ACGM-1 SN01-0116

60 Hz
E-FIELD 10K¥ 1

4 MHz

10 MHz 488 MHz
‘UNHOD

4 MHz

10 MHz
6OHz MOD

100 MHz

OOFIELD OFF
FELD ON


SS

Square W
Frequency Responce - Meter IIIA mn Suitooth vave
EXPAN Model 1A SN 1103 + Sine Wave
40 — ~ — Bezier Fit
20
oO 4
3
a
5 -20-
o
2
-40 4
-60 T ToT rrTyy er T Le | T ay
1 10 10? 107 10

Frequency (Hz)


Project Organization

This effort was executed in three parts. The first part included
creation of required test plans and procedures. Part two provides technical
support for execution of the test phase. The last part addresses the
reporting and summarizing of test results.

Test Planning (QAPP) - Task 1

Recommendations for calibration and survey meter test plans and
procedures were developed. In addition, a run-through of the procedures
against a control survey meter was performed with EPA supervision for the
purpose of testing the developed procedures.

The final product of this task was the Quality Assurance Project Plan,
which incorporated test and calibration procedures with standard EPA quality
assurance procedures.

Test Execution - Task 2
A technician executed the survey meter test plans and procedures under
EPA supervision at the EPA’s Las Vegas Facility.

Test Reporting - Task 3

A final report (this document) was developed providing the results of
this study including, data collation, graphic presentation, and procedural
analysis.


METER CODE

1A2

WRI - Model 660 - SN G1242

MANUFACTURER

Widerange Instruments
110 Shelter Lagoon
Santa Cruz, CA 95060

METER NAME

Magalert

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Date

dan-24-91

RANGE SETTING

100 mG
100 mG
100 mG
100 mG
100 mG

Time

10:17AM

ACTUAL FIELD

SSNzRRSaRe
BRRABRRRBBR

eaoraSslRrge

Temperature

72° F

MEASURED FIELD

100.00
100.00
100. 0
100.0

y
3
2

8

50.0

20.01
10.01
10.01

y
3S
2
Sssssss

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date

Mar-14-91

RANGE SETTING

100m

100mG

100mG

100mG

100mG

Time

01:56PM

ACTUAL FIELD

32.

COCK C Oss RENN

SRESRBRASERREES
BRRRRRRRRRBBABR

Temperature

62°

MEASURED FIELD

Appendix

100.00
100.00
50.00
50.00
20.00
20.00
10.00
10.00
5.00
5.00
2.00
2.00
1.00
1.00
0.50
0.50
0.20
0.20

TA2

2aeRRRBRRR

BRRRAZAARARARRBBRARD

MODEL #
660

SERIAL #
61242

Meter Background- mG

0.5

‘% ERROR

22.97%
27.77%
17.82%
22.07%
28.72%
26.96%
23.57%
19.59%
18.82%
18.82%

22.71%

Meter Background

0.2

‘% ERROR

25.24%

23.21%

mG

AVG % ERROR

AVG % ERROR


EXP - ELF - MODEL 1A - SN 1104

MINIMUM DETECTABLE SIGNAL

Date Time Temperature Meter Background
War-18-91 01:50PM ore F min 0.05 mG
: max 0.05 mG
Threshold Current mA Threshold Field us
1 1.30 ma 70.14 uG
Trial 2 0.90 ma 48.56 UG
Number 3 1.40 ma 75.53. uG
4 0.80 ma 43.16 uG
5 4:00 mA 53.95 us
NOTE: Needle moves slightly 58.27 UG AVG THRESHOLD FOR SENSITIVITY
METER RESOLUTION DATA
Date Time Temperature Meter Background
Mar-18-91 01:56PH 67 F min 0.05 mG
mex 0.05 mG

Meter Range 30 mG

Drive Current Start A Threshold Current A Resolution uG

1 0.3580 A 0.3595 A 80.92 uG

Trial 2 0.3580 A 0.3587 A 37.76 uG
Number 3 0.3580 A 0.3596 A 86.32 uG

4 0.3580 A 0.3605 A 134.87 uG

5 0.3580 A 0.3601 A 113.29 uG

NOTE: Needle moves slightly 90.63 uG

Appendix ITIB

Resolution %

0.41%
0.19%
0.44%
0.69%

0.46%


MINIMUM DETECTABLE SIGNAL

Date Time

Mar-21-91 02:13PM

Threshold Current

1 7.20
Trial 2 7.10
Number 3 7.30
4 7.30
5 7.50

NOTE: Meter readings increased by 0.3mG

METER RESOLUTION DATA

Date Time
Mar-21-91 02:22PM
Meter Range Auto Range - 200mG

Drive Current Start

1 1.7520

Trial 2 1.7520
Number 3 1.7520
4 1.7520

5 1.7520

IRC - MODEL IRC 27

Temperature
56° F

mA Threshold Field

388.44
383.05
393.84
393.84
404.63

22232

392.76

Temperature

A Threshold Current

1.7550
1.7560
1.7560
1.7540
1.7540

>> pr>

NOTE: Meter readings increased by about 0.15mG

Appendix VB

Meter Background

UG AVG THRESHOLD FOR SENSITIVITY

Meter Background

min 70.1
max 0.1
A Resolution
A 161.85
A 215.80
A 215.80
A 107.90
A 107.90
161.85

mG
mG

uG

Resolution %

0.17%
0.22%
0.22%
0.11%
0.11%

0.17%


WRI - Model 670 - SN G1259

METER CODE MANUFACTURER METER NAME

12 Widerange Instruments Magalert 2
110 Shelter Lagoon

Santa Cruz, CA 95060

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Date

Feb-07-91

RANGE SETTING
100*10 me

100*1 mG
100*1 mG

Time

02:19PM

ACTUAL FIELD

Temperature
77° F

MEASURED FIELD
500.00 mG

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date

Mar-14-91

RANGE SETTING

100"1

100*1

100*1

100*1

Time

12:53PM

ACTUAL FIELD

v3

we
a

RPRRSSKSURRASIHaN
33333288332383333

Temperature
59° F

MEASURED FIELD

100.00
100.00
50.00
50,00
20.00
20.00
10.00
10.00
5.00
5.00
2.00
2.00
1.00
1.00
0.50
0.50

BRB RR AAR R ARB RRR

Appendix 12

MODEL #
670

Meter Background
0.5 mG

‘% ERROR

27.83%

32.87%

Meter Background

0.2 mG

‘% ERROR

27.83%

42.50%

SERIAL #
G1259

AVG % ERROR

AVG % ERROR


ek

Responce (dB)

40 -

20 -

-20 5

-40

- (= Square Wave
Frequency Responce Meter X T._ sawtooth Wave
WALKER ELF-50D SN 71181-205 + Sine Wave
: — Bezier Fit
T T TT TtTT TT T T TT TrTy Ty T T T TTrTy 7]
to 10? 10? 10

Frequency (Hz)


Le

Accuracy and Linearity - Meter VIB - Linear Mode
MONITOR Model 42b-1 SN 1287

10 -
5 4
. $A
fF oH
iw
»
8
2
-5 4
-10 Ne ee ee |
107 1 10 10? 10°

Applied Field (mG)


02
Responce (dB) -

- (Square Wave
Frequency Responce - Meter IXB T_ savtooth Wave
HOLADAY HI-3604 ‘SN 76112 + Sine Wave
— Bezier Fit

TTT TT TTT TTT T1117 TT 111111}

10 10? 107 10°
Frequency (Hz)


Susceptibility - Meter XVA - Band Pass
DINDIMA GAUSSMAUS SN GM0400

OOFIELD OFF
IEFIELD ON

9vT

aos mazZzorvama

4

I

60 Hz
E-FIELD 10kV/4

1 Maz

10 MHz 108 HH
‘UNHOD:

+ MHz

10 MHz

6eHz MO!


£9

. W
Frequency Responce - Meter VIB - Linear Mode a sueetn weve
MONITOR Model 42b-1 SN 1287 + Sine Wave
40 — — Bezier Fit
20
ia]
A
o 4
= +
qa
2g -20-
a
on
fe
-40 4
-60 T TT ITINy T TTT T TOTTI T TTTTINy] YT TTT}
1 10 10? 107 104 107

Frequency (Hz)


02

Accuracy and Linearity - Meter IIIB
EXPAN Model 1A SN 1104

10 -
5 4
i.
B94
»
8
ct
“os
-10
i

T rr T TT TT TITY] T TT TT TTT]

10 10? 107
Applied Field (mG)


Le

Percent Error
Sg
\ |
T

Accuracy and Linearity - Meter XIVA

EFM - Metex-3800 - SN 542513
10 4

10 - T TT TTT T Toy T TT rt

1 — 10 9g? 107
Applied Field (mG)


METER CODE
"4

WRI - Model 670 - SN G1257

MANUFACTURER

Widerange Instruments
110 Shelter Lagoon
Santa Cruz, CA 95060

METER NAME

Magalert 2

METER ACCURACY AND LINEARITY AT 60H2 - TEST LOCATION LAB

Date

Feb-08-91

RANGE SETTING

100*10 mG
100*10
100*10 mG
100*10
100*10 mG

100*10

100*10 mG

Time

08:15AM

ACTUAL FIELD

384.51
176.6!
103.41
93.4;
36.6
17.2
8.8

BaS

assn

BABB RAB

Temperature
74°

MEASURED FIELD

500.01
200.01
100.0
100.01
50.01
20.01
10.01

2422883

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date

Mar-14-94

RANGE SETTING

100*1

100*1

100*1

100*1

Time

14:15AM

ACTUAL FIELD

ueeee
2

a BWR Oe
Seccconannuuv ot T8588

S2eskessuasseagkese
2B2222223828282328323

Temperature
58° F

MEASURED FIELD

Appendix 11

MODEL #
670

Meter Background
0.5 mG

‘% ERROR

30.04%
13.22%
+3.29%

7.05%
36.29%
15.85%
13.02%

16.03%

Meter Background

0.1 mG

% ERROR

16.21%
20.36%
10.33%
10.99%
+2.03%
0.02%
+7.55%
+14.78%
+B.24%
78.24%
6.76%
4.26%
1.01%
1.84%
°10.02%
9.58%

443%

SERIAL #
G1257

AVG % ERROR

AVG % ERROR


SYDKRAFT - MODEL 88.05 - VIIB

FREQUENCY RESPONCE PRIMARY DATA

Date Time Temperature Meter Background
Feb- 13-91 10:30AM T3° F 0.06 mm
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
1 Hz 2 mG 989.37 mG 0.11 mG +79.08 dB
2 Hz 2 mG 989.37 mG 0.12 mG 78.32 dB
5 Hz 2 mG 989.37 mG 0.15 mG +76.10 cB
10 Hz 2% 989.37 mG 0.24 mG +72.30 cB
20 Hz 206 986.09 mG 1.85 mG 754.53 0B
50 Hz 200 mG 753.28 mG 150.00 mG 714.02 oB
60 Hz 200 m6 146.37 m6 150.00 mG 0.21 dB
100 Hz 200 m6 1009.86 mG 55.00 mG +25.28 dB
200 Hz 20 mG 995.11 mG 13.30 mG +37.48 dB
500 Hz 20 mG 246.87 mG 1.00 mG 47.85 dB
1 kHz 20 mG 218.38 mG 0.40 mG 54.74 cB
FREQUENCY RESPONCE AUXILIARY DATA
Date Time Temperature Meter Background
Feb-13-91 12:44PM 76° F 0.03 mm
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
30 Hz 20 mG 1011.75 mG 8.30 mG “41.72 GB
40 Hz 200 mG 994.21 mG 34.00 mG +29.32 dB
80 Hz 200 mG 1004.37 mG 141.00 mG 17.05 dB
55 Wz 200 mG 261.43 mG 150.00 mG
65 Hz 200 mG 199.75 mG 150.00 mG

Appendix VIIB


4g

Responce (dB)

40 5

20

- (1 Square Wave
Frequency Responce - Meter IVA A Savtooth wave
MSI Model MSI-20 SN 79011626 -++ Sine Wave
— Bezier Fit

TTT TTT TT TTT TTI TTT TTT TTT TTT]

10 10? 107 104
Frequency (Hz)


SYOKRAFT - MODEL 88_05

METER COOE MANUFACTURER METER NAME
VIIB Sydkraft H Flux
AVD TEK 4

GAMLA SEGEVAGEN
232 91 ARLOV Sweden

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

> VIB
MODEL #

88-05

Date Time Temperature Meter Background
Feb- 13-91 09:51AM 75° F 0.03 mG

RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR

2000 mG 491.98 mG 500.00 mG 1.63%

200 146.37 mG 150.00 mG 2.48%

200 mG . 73.38 mG 75.00 mG 2.21%

200 26.28 mG 25.00 mG 2.98%

20 mG 9.74 mG 10.00 mG 2.72%

2.40%

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION RI

Date Time Temperature Meter Background
Mar-21-91 12:33PM 58° F Uy) mG

RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR

200 mG 97.65 mG 100.00 mG 2.41%

97.65 mG 100.00 mG 2.41%

20 me 9.76 mG 10.00 mG 2.41%

9.76 mG 10.00 mG 2.61%

2 mG 0.97 mG 1.00 mG 2.41%

0.97 mG 1.00 mG 2.41%

2 mG 0.20 me 0.20 mG 0.19%

0.20 mG 0.20 mG 0.19%

1.85%

Appendix VIIB

SERIAL #

AVG % ERROR

AVG % ERROR


£2

Percent Error

Accuracy and Linearity - Meter VA

IRC Model IRC-109
10 -

-10 T TTT TTI TY T TTT TTT T TTT TTT

I 10 10? 10°
Applied Field (mG)


bol

as moazovuma

1.8

(|

Susceptibility - Meter XIVB
EFM Metex-3800 SN 542832

ot

ee ee

68 Hz
E-FIELD 18kV 4

a ,
4 MHz = 10 MHz «180 MHz
UNHOD

4MHz | 10 MHz 468 Miz
Genz MOD

OOFIELW OFF
WFIELD ON


06

Responce

La

a8

06-

a4—-

a2-

a0a-

10

Angular Responce - Meter IIIA

EXPAN
+

20

30

Model 1A SN 1103
+
40 50 60

Rotation Angle

70

80

90

+

Meter Responce

Cosine Curve


Magnetic Tech - Model ELF 12 SN 4

MINIMUM DETECTAB! IGNAL

Date Time Temperature
‘Aug-29-91 10:52AM 94° F
Threshold Current mA Threshold Field
1 124.00 ma $689.80
Trial 2 100.00 ma 5395.00
Number 3 202.00 ma 10897.90
4 200.00 mA 10790.00
5 180.00 ma 9711.00
NOTE: Meter readings increase by 4 mG 8696.74
METER RESOLUTION DATA
Date Time Temperature
Aug- 29-91 11:00AM 94° F
Meter Range 96 mG

Drive Current Start

1 0.9700

Trial 2 0.9700
Number 3 0.9700
4 0.9700

5 0.9700

NOTE: Meter readings increase by 16 mG

A Threshold Current
A 1.3700
A 1.3700
A 1.3700
A 1.3700
A 1.3700

Appendix XVIIB

min

5

5555S

min
max

A

> Pe>>

Meter Background

2
2

mG
mm

AVG THRESHOLD FOR SENSITIVITY

Meter Background

2
2

Resolution

21579.63
21579.63
21579.63,
21579.63,
21579.63,

21579.63,

mG
mG

uG

65555

5

Resolution %

41.24%
41.24%
41.24%
41.24%
41.24%

41.24%


EFM - Metex-3800 - SN 542832

METER CODE MANUFACTURER METER NAME MODEL # SERIAL #
XIVB Electric Field Measurements Metex Digital Metex-3800 542832
Multimeter
METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB
Date Time Temperature Meter Background
May- 15-91 01:06PM 77° F 0.1 mG
RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR
2000 mG 499.80 mG 500.00 mG 0.03%
200 mG 149,88 mG 149.95 mG 0.04%
0.04% AVG % ERROR
METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE
Date Time Temperature Meter Background
Mar-28-91 02:18PM 72° F 0.05 mG
RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR
100.35 mG 99.95 mG 70.39%
200 mG 100.35 mG 99.90 mG 70.44%
50.17 mG 50.05 mG 70.24%
50.17 mG 50.05 mG 70.24%
10.03 mG 10.00 mG 0.34%
1.03 me 1.00 me 72.44%
0.33% AVG % ERROR

3

Appendix XIVB


WRI - Model 660 - SN G1241

METER 60Hz_E FIELD SUSCEPTIBILITY DATA

Date Time Temperature
Feb-27-91 01:00PM 78° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10 kV/M +-5%_
€ Field OFF Reading(mG) E Field ON Reading(m
Meter Range Meter Range
Full Full
1 0.30 1.10
Triat 2 0.30 115
Number 3 0.30 115
4 0.30 1.10
5 0.30 4.10

0.30 AVG 1.12

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Date Time Temperature
Feb-20-91 01:22PM “80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Full Full
1 MHz 0.35 0.35
10 MHz 0.35 0.35
100 MHz 0.35 0.36

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date Time Temperature

Feb-20-91 01:49PM 80° F

RF OFF Reading(mG) RF ON Reading(mG)

Meter Range Meter Range

Full Full

1 MHz 0.35 0.35

10 MHZ 0.35 0.35

100 MHz 0.35 5.00

Appendix IA1


bp

Percent Error

10 -

Accuracy and Linearity - Meter XVIB
MEDA Model PLM-100 “SN 1381

TT TT TTT7 T | T T—T-T1TT

10 10?
Applied Field (mG)

107


s8

Responce

1o-

ae-

a6 -

a4

a2

Angular Responce - Meter 12
WRI Model 670 SN G1259

-++ Meter Responce
— Cosine Curve

a0-
a

CUUREERU SS AUREUS SRL PULLEY ER

10 20 30 40 50 60 70 80 90

Rotation Angle


EMD - MODEL ACGM-1 - SN 01-0117

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date

Feb-11-91

SQUARE

RANGE SETTING
FULL

‘SAWTOOT!

RANGE SETTING

FULL

SINE

RANGE SETTING

FULL

METER ANGULAR RESPONCE DATA

Date

Feb-11-91

Angle

0°
10°
20°
30°
40°
50°
60°
70°
80°
90°

Time

10:15AM

WAVE TESTING

ACTUAL FIELD

289.05 mG
288.90 mG

'H WAVE TESTING

ACTUAL FIELD

293.14 mG
293.86 mG

WAVE TESTING

ACTUAL FIELD

263.11 mG
263.26 mG

Time

10:30AM

ACTUAL FIELD

262.99 mG
270.99 mG
289.77 mG
320.79 mG
374.76 1G
469.50 mG
584.25 mG
829.86 mG
972.55 mG
1052.68 mG

Temperature
74° F

MEASURED FIELD

250.00
250.00

MEASURED FIELD

250.60
251.20

MEASURED FIELD

250.20
250.20

Temperature
74° F

MEASURED FIELD

250.00
250.30
250.30
250.20
250.20
250.20
250.30
250.50
145.20

7.60

Appendix VIII8

Meter Background

0.1 mG
% LKROR
mG 13.51%
mG 13.46%
12.49%
% ERROR
mG 14.51%
mG +14.52%
714.52%
% ERROR
mG 4.91%
mG 74.96%
4.94%

Meter Background

0.1 mG

ANGULAR RESPONCE

mG 1.00
mG 0.97
mG 0.90
mG 0.82
mG 0.70
mG 0.56
mG 0.45
mG 0.31
mG 0.15
mG 0.00

RELATIVE RESPONCE

71.26 dB
71.26 GB

71.26 db

RELATIVE RESPONCE

ote
ae
71.36 db

RELATIVE RESPONCE

0.43 dB
70.44 dB

-0.44 db


Magnetic Tech - Model ELF 12 SN 4

METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date

Hay-20-91

EFM Metex

Trial
Number

wnune

Time

09:59AM

E Field OFF Reading

0.00

E Field OFF Reading(ms)
Meter Range
‘Auto
2.00
2.00
2.00

2.00
2.00

2.00

AVG

Temperature
75° F

E Field ON Reading
10.00 KV/M +-5%

E Field ON Reading(mG)
Meter Range
Auto
2.00
2.00
2.00

2.00
2.00

2.00

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA > UNMODULATED

Date

May-17-91

1 MHz
10 MHz
100 MHz

Time
02:06PM
RF OFF Reading(mG)
Meter Range
‘Auto
2.00

2.00
2.00

Temperature
Tre
RF ON Reading(mG)
Meter Range
Auto
2.00

0.00
2.00

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date

May-17-91

1 MHZ
10 MHz
100 MHZ

Time
02:06PM
RF OFF Reading(ms)
Meter Range
Auto
2.00

2.00
2.00

Temperature
7m e

RF ON Reading(mG)

Meter Range

Auto
16.00
96.00
2.00

Appendix XVIIB


EFM - Metex-3800 - SN 542832

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date ~ Time Temperature
May- 15-91 01:54PM TF

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD
200 me 187.80 mG 149.95
187.66 mG 1469.95

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD
200 mG 147.67 mG 149.95
147.84 mG 150.05

SINE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD
200 mG 149.75 mG 149.95
149.80 mG 149.95

METER ANGULAR RESPONCE DATA

Date Time Temperature
May- 15-91 02:16PM 77 F

Angle ACTUAL FIELD MEASURED FIELD
149.75 mG 149.95

151.71 mG 150.05

158.74 mG 149.95

170.42 mG 149.95

192.00 me 169.95

232.12 mG 149.95

296.72 mG 149.95

426.97 mG 149.95

782.02 mG 150.05

1195.13 mG 26.95

Appendix XIVB

Meter Background

0.1 mG
% ERROR
mG +20.16%
mG + 20.09%
720.12%
‘% ERROR
mG “1.55%
mG 1.50%
1.52%
% ERROR
mG 0.13%
mG 0.09%
0.11%

Meter Background

0.1 mG

METER RESPONCE

eeocoosso>
Sau
RSUsKasess

BBRRRRRRBR

RELATIVE RESPONCE

71.96 dB
“1.95 dB

+1.95 db

RELATIVE RESPONCE

0.13 dB
0.12 dB

0.13 db

RELATIVE RESPONCE

0.01 dB
0.00 dB

0.01 db


WRI - Model 660 - SN G1242

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date Time Temperature Meter Background

Jan-24-91 12:08PM 72° F 0.4 im

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR RELATIVE RESPONCE
100 mG 77.32 mG 100.00 mG 29.33% 2.23 dB

75.31 mG 100.00 mG 32.78% 2.46 dB
31.06% 2.35 db

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR RELATIVE RESPONCE
100 mG 108.25 mG 100.00 me +7.62% -0.68 dB

100.77 mG 100.00 mG 0.76% -0.06 oB
74.19% +0.37 db

SINE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR RELATIVE RESPONCE
100 mG "82.79 mG 100.00 mG 20.79% 1.64 dB

82.12 mG 100.00 mG 21.78% 1.71 dB
21.29% 1.68 db

METER ANGULAR RESPONCE DATA

Date Time Temperature Meter Background
Jan-24-91 01:10PM T° F 0.3 mG
Angle ACTUAL FIELD MEASURED FIELD ANGULAR RESPONCE

o° 80.63 mG 100.00 mG 1.00

10° 76.06 mG 100.00 mG 1.06

20° 85.59 mG 100.00 mG 0.9%

30° 87.88 mG 100.00 mG 0.91

40° 106.46 mG 100.00 mG 0.75

50° 122.75 mG 100.00 mG 0.65

60° 166.32 mG 100.00 mG 0.48

70° 269.35 mG 100.00 mG 0.29

80° 604.82 mG 100.00 mG 0.13

90° 601.66 mG 20.00 mG 0.02

Appendix 1A2


IRC - MODEL IRC 109

FREQUENCY RESPONCE PRIMARY DATA

Date Time Temperature Meter Background
Feb-11-91 02:50PM 76° F 0.13 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
1 uz 20 mG 954.99 mG 0.14 mG +76.37 cB
2 Hz 20 mG 954.99 mG 0.30 mG +70.06 dB
5 hz 20 mG 954.99 mG 1.62 mG 755.41 dB
10 Hz 20 mG 954.99 mG 6.54 mG +43.29 dB
20 Hz 200 mG 1015.70 mG 28.80 mG +30.95 dB
50 Hz 200 mG 442.70 mG 150.00 mG +9.40 dB
60 Hz 200 mG 151.99 mG 150.00 mG -0.11 GB
100 Hz 200 mG 548.98 mG 150.00 mG -11.27 GB
200 Hz 200 mG 1009.96 mG 109.30 mG 19.31 dB
500 Hz 200 mG 1007.18 mG 38.70 mG 728.31 GB
1 kHz 200 mG 1002.36 mG 14.70 mG +36.67 dB
2 khz 200 mG 1009.51 mG 4.40 mG 47.21 GB
5 ktz 20 mG 794.91 mG 0.67 mG +61.48 dB
10 kHz . 20 mG 553.75 mG 0.13 mG -72.59 GB
FREQUENCY RESPONCE AUXILIARY DATA
Date Time Temperature Meter Background
Feb-11-91 01:33PM 76° F 2 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
70 Hz 2000 mG 278.02 mG 250.00 mG -0.92 cB
80 Hz 2000 mG 514.63 mG 250.00 mG -6.27 cB
90 Hz 2000 mG 725.41 mG 250.00 mG 79.25 dB
40 Hz 2000 mG 1015.80 mG 155.00 mG +16.33 cB
30 Hz 2000 mG 1010.56 mG 72.00 mG 22.94 dB
65 Hz 2000 mG 194.14 mG 250.00 mG 2.20 dB
55 Hz 2000 mG 448.98 mG 250.00 mG °5.09 GB
150 Hz 2000 mG 1008.79 mG 153.00 mG +16.38 GB
- FREQUENCY RESPONCE AUXILIARY DATA
Date Time Temperature Meter Background
Feb-12-91 08:40AM 76° F 0.13 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
65 Hz 200 mG 116.49 mG 150.00 mG 2.20 oB
S5 Hz 200 mG 270.52 mG 150.00 mG -5.12 0B

Appendix VA


MSI - MODEL MSI 20 - SN 79011626
METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date Time Temperature
Feb-25-91 12:23PM 78° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.0 kV/M +-5%
& Field OFF Reading(mé) E Field ON Reading(mG
Meter Range Meter Range
200 200
1 0.00 1.05
Trial 2 0.00 1.05
Number 3 0.00 1.05
4 0.00 1.05
5 0.00 1.05
0.00 AVG 1.05
E Field OFF Reading(mG) E Field ON Reading(mG
Meter Range Meter Range
2000 2000
1 0.00 0.03
Trial 2 0.00 0.03
Number 3 0.00 0.03
4 0.00 0.03
5 0.00 0.02
0.00 AVG 0.03
E Field OFF Reading(mG) E Field ON Reading(mG
Meter Range Meter Range
20000 20000
1 0.00 . 0.02
Trial 2 0.00 0.02
Number: 3 0.00 0.02
4 0.00 0.02
5 0.00 0.02
0.00 ANG 0.02
E Field OFF Reading(mG) E Field ON Reading(mG
Meter Range Meter Range
2006 & 7506 2006 & 7506
1 WO RESPONCE NO RESPONCE
Trial 2 WO RESPONCE NO RESPONCE
Number 3 WO RESPONCE NO RESPONCE
4 WO RESPONCE NO RESPONCE
5 NO RESPONCE NO RESPONCE

Appendix IVA


29

Frequency R

_ _ (1 Square Wave

esponce - Meter VIA - Flat Mode A Sautooth Wave

MONITOR Model 42b-1 SN 1286 + Sine Wave
— Bezier Fit

40 —

20 5

o 4
g
8

5 20-
o
fe

-40 4

-60
1

TTT TTT

10

TT TTT TTT TTT TT T1117]

10? 10? 10°
Frequency (Hz)


WALKER - ELF-SOD - SN 71181-204

FREQUENCY RESPONCE PRIMARY DATA

Date

Feb-07-91

FREQUENCY

5
10
20
50
60

100
200
500

1

2

5
10

Hz
Hz
hz
Hz
Hz
Hz
Hz
Hz
kHz
kkz
kHz
kHz

Time

11:26AM

RANGE SETTING

Lo
to
Lo
Lo
Lo
Lo
Lo
Lo
Lo
Lo
Lo
Lo

FREQUENCY RESPONCE AUXILIARY DATA

NO AUX TESTING REQUIRED

Temperature
77° F

ACTUAL FIELD

aes
reee sss

FARRGISISARG

was
Naa

114.

ARRRRRARBBRS

8

Appendix XA

Meter Background

0

mm

MEASURED FIELD

140.00
151.00
152.00
150.00
150.00
150.00
151.00
150.00
150.00
150.00
150.00
150.00.

BaRARBZBR BBS

RELATIVE RESPONCE

715.66
+15.00

9.62
72
0.13

4.34
10.56
19.46
20.69
13.38

2.32
“8.11

B
oB
oB
oB
dB
dB
dB
oB
dB
dB
dB
Bo


98

Responce

1.04

ae
06 -
a4—-
42 -

40-

Angular Responce.~ Meter IAl
Model 660

10

WRI

20

30

SN G1241

40 50

Rotation Ang] e

60

70

+

Meter Responce
Cosine Curve


DINDIMA - GAUSSMAUS - SN GMO400

METER 60Hz E FIELD SUSCEPTIBILITY DATA

Frequency Responce Setting - WIDE BAND

Date

May-20-91

EFM Metex

Trial
Number

wnune

Time

09:01AM

E Field OFF Reading
0.00

E Field OFF Reading(mG)
Meter Range
9.9
0.10
0.10
0.10

0.10
0.10

0.10

Frequency Responce Setting - BAND PASS

Date

May-20-91

EFM Metex

Trial
Number

wnuna

Time
09:07AM

€ Field OFF Reading

0.00

E Field OFF Reading(mG)
Meter Range
9.9
0.30
0.20
0.20

0.20
0.20

0.22

AVG

AVG

Temperature
75° F

£ Field ON Reading

10 KV/M +-5%

E Field ON Reading(mG)

Meter Range

Temperature
74° F

E Field ON Reading

10 KV/M +-5%

E Field ON Reading(mG)
Meter Range

9.9

Appendix XVA


[21

aB Mazorvuoma

Susceptibility - Meter |A1-
WRI Model 660 SN G1241

CORED OFF
FELD ON

‘| ttt | el

6e Hz 1 MHz 10 MHz §=6188 Hz en
E-FIELD 1eKvt UNHOD Genz HOD


EXP - ELF - MODEL 1A - SN 1104

FREQUENCY RESPONCE PRIMARY DATA

Date Time Temperature Meter Background
Feb- 05-91 09:24AM 74° F 0.1 mG

FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

1 hz 3 mG 973.17 mG 0.10 mG +79.76 dB

2 Hz 3G 973.17 mG 0.20 mG +73.74 dB

5 Hz 3 mG 973.17 mG 0.40 mG -67.72 dB

10 Hz 3 0G 973.17 mG 1,50 mG +5624 dB

20 Hz 100 mG 982.39 me 30.00 ms -30.30 dB

50 Hz 1000 mG 167.96 mG 150.00 mG 0.11 de

60 Hz 1000 me 168.44 mG 150.00 mG 0.09 dB

100 Hz 1000 mG 137.13 mG 150.00 mG 0.77 dB

200 Hz 1000 mG 142.87 mG 150.00 mG 0.42 de

500 Hz 1000 mG 146.60 mG 150.00 mG 0.19 dB

1 kHz 1000 mG 153.20 mG 150.00 mG -0.18 dB

2 kHz 1000 mG 178.32 mG 150.00 mG =1.50 dB

5 kHz 1000 mG 289.29 mG 150.00 mG +5.70 dB

10 kHz 1000 mG 499.65 mG 125.00 mG 712.04 dB -

FREQUENCY RESPONCE AUXILIARY DATA

Date Time Temperature Meter Background
Feb-05-91 10:40AM 74° F 0.1 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
80 Hz 1000 mG 141.98 mG 150.00 mG 0.47 dB
150 Hz 1000 mG 141.28 mG 150.00 mG 0.52 dB
40 Hz 1000 mc 150.12 mG 150.00 mG
30 Hz 1000 mG 591.80 mG 150.00 mG
60 Hz 1000 mG 148.16 mG 150.00. mG
300 Hz 1000 mG 143.79 mG 150.00 mG
25 Hz 1000 mG 1116.32 mG 100.00 mG
35 Hz 1000 mG 241.90 mG 150.00 mG

Appendix 1118


$21

os MOZO THM

e.a4

|

Susceptibility - Meter 112

_ SAFE PROMETER SN 1264

Pat al

l

I

60 Hz
E-FIELD 10kV 1

1 Maz 10 hue 900 Hz

10 MHz
6eHz MOD

CORED OFF
FELD ON


Te

10 -

Percent Error

Accuracy and Linearity - Meter VITIA
EMD Model ACGN-1 SN 01-0116

T | T TT TTT TTT T TT T1177]

10 10? 107
Applied Field (mG)


APPENDIX

156


SAFE - SAFE PRO METER - SN 1263

METER CODE MANUFACTURER METER NAME MODEL # SERIAL #
1 Safe Computing Company Safe Meter 1263
368 Hillside Ave.
Needham, MA 02194
METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB
Date Time Temperature . Meter Background- mG
Feb-05-91 12:51PM 76° F 0.2
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
200 mG 139.07 mG 150.10 mG 7.93%
23.30 mG 25.30 mG 8.59%
200 mG 69.61 mG 75.10 mG 7.89%
23.15 mG 25.00 mG 7.97%
8.10% AVG % ERROR
METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE
Date Time Temperature Meter Background
Mar- 18-91 11:48AM 64° F 0.15 mG
RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR
92.79 mG 100.10 mG 7.87%
200 mG 92.79 mG 100.10 mG 7.87%
18.61 mG 20.00 mG 7.45%
18.61 mG 20.05 mG TA72h
9.28 mG 10.00 mG 7.77%
9.28 mG 10.00 mG 7.7%
1.89 mG 2.00 mG 5.92%
1.89 mG 2.00 mG 5.92%
0.94 mG 1.00 mG 5.92%
0.94 mG 1.05 mG 11.21%
7.54% AVG % ERROR

Appendix 111


HP 33144
FUNCTION
GENERATOR

HELMHOLTZ COIL SYSTEM
HIGH LEVEL TESTING

2 OHM, 800 WATT (20 AMPS) LOAD

RED 30 FEET
CROWN M600 : 10 awG <=>
AMPLIFIER Ap Last

BLACK

DC - 10 kHZ HELMHOLTZ COIL
UP 70 20 AMP 0.5 m RADIUS
INTO 2 OHMS 3 TURNS
FLUKE A40A-20A ——B=1 (S3.95 milligauss /amp)
HP 3456A Loe 0.021 OHM
DIGITAL VOLTMETER CURRENT SHUNT
HI® 20 AMPS MAXIMUM

K(amp)=V/0.021 ohm


MSI - MODEL MSI 20 - SN 79010854

FREQUENCY RESPONCE PRIMARY DATA

Date Time Temperature Meter Background
Jan-28-91 09:22AM Th? F 0 mG

FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

1 kz 200 mG 1004.60 mG 2.50 mG +52.08 dB

2 Hz 200 mG 1004.60 mG 25.70 mG 31.84 dB

5 Hz 200 mG 1004.60 mG 61.60 mG 24.25 dB

10 Hz 200 mG 1004.60 mG 101.70 mG +19.89 dB

20 Hz 200 mG 600.55 mG 148.80 mG 12.12 cB

50 Hz 200 mG 184.80 mG 149.90 mG 1,82 dB

60 Hz 200 mG 150.94 mG 150.00 mG -0.05 dB

100 Hz 200 mG 102.48 mG 149.90 mG dB

200 Hz 200 mG 172.31 mG 149.90 mG oB

500 Hz 200 m6 468.31 mG 149.90 mG aB

1 kHz 200 mG 957.55 mG 149.90 mG B

2 kHz 200 mG 1202.48 mG 92.90 mG dB

5S kHz 200 mG 901.14 mG 26.70 mG -30.57 dB

10 kHz 200 mG 553.92 mG 7.00 mG +37.97 dB

FREQUENCY RESPONCE AUXILIARY DATA

Date Time Temperature Meter Background
dan-28-91 10:56AM 74° F 0 mG

FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

80 Hz 200 mG 116.41 mG 149.90 mG 2.35 dB

150 Hz 200 mG 126.48 mG 150.00 mG 1.48 cB

30 Hz 200 mG 318.63 mG 149.70 mG +6.56 dB

300 Hz 200 mG 271.76 mG 149.90 mG *5.17 oB

1.5 kHz 200 mG 1231.27 mG 127.60 mG +19.69 dB

3 khz 200 mG 1076.70 mG 54.70 mG -25.88 dB

4 kkz 200 mG 993.91 mG 37.40 mG *28.49 dB

Appendix  1VB


6eI

OB MazZOrvTUMD

Susceptibility - Meter IXA
HOLADAY HI-3604 -SN 76111

68 Hz
E-FIELD 10kV 1

4 MHz 10 MHz «6108 MHz
UNMOD

4 MHz

19 MHz 190 MHz
Genz HOD

OFEDOFF —

FELD ON


DINDIMA - GAUSSMAUS - SN GMO400

METER CODE MANUFACTURER
XVA The Dindima Group
Pry LTD

METER NAME MODEL #

GAUSSMAUS

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Frequency Responce Setting - WIDE BAND

Date Time
Mar-28-91 12:50PM
RANGE SETTING ACTUAL FIELD
9.9 mG 10.99 mG
mG 10.99 mG
mG 5.65 mG
99.0 mG 56.46 mG
mG 109.87 mG
Frequency Responce Setting - BAND PASS
Date Time
Mar-28-91 12:54PM
RANGE SETTING ACTUAL FIELD
9.9 mG 11.14 mG
mG 11.14 mG
mG 5.91 mG
99.0 mG 57.99 mG
mG 109.87 mG

Temperature Meter Background

64° F 0.1

MEASURED FIELD ‘% ERROR

80 mG -10.80%
9.80 mG + 10.80%
5.00 mG 71146%
0.00 mG 71166%
8.00 mG +10.80%

11.06%

Temperature Meter Background

68° F 0.1

MEASURED FIELD % ERROR

12.02%
12.02%
715.33%
“13.77%
+ 10.80%

ssgse
RaBaR

ou

12.79%

Appendix  XVA

mG

mG

SERIAL #
M0400

AVG % ERROR

AVG % ERROR


Let

as Mazo Tuma

4.24

0.44

Susceptibility - Meter XVB - Wide Band

DINDIMA GAUSSMAUS SN GM0401

(COFIELD OFF
FELD ON

60 Hz
E-FIELD 10kV/1

41 MHz

4 MHz

10 MHz 100 Hz
6eHz HOD


SYDKRAFT - MODEL 88.05 - VIIB

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date

Feb-13-91

RANGE SETTING

200 mG

RANGE SETTING
200

RANGE SETTING
200

METER ANGULAR

Date

Feb-13-91

Angle

0°
10°
20°
30°
40°
50°

Time
12:05PM

SQUARE WAVE TESTING

ACTUAL FIELD

161.45 mG
161.87 mG

SAWTOOTH WAVE TESTING

ACTUAL FIELD

mG 176.68 mG

177.62 mG

SINE WAVE TESTING

ACTUAL FIELD

145.93 mG
146.08 mG

RESPONCE DATA

Time

12:18PM

ACTUAL FIELD

Temperature
76° F

MEASURED FIELD

150.00 mG
150.00 mG

MEASURED FIELD

150.00
150.00

MEASURED FIELD

150.00
150.00

Temperature
76° F

MEASURED FIELD

150.00
150.00
150.00
150.00
150.00
150.00
150.00
150.00
150.00

3.40

Appendix  VIIB

ms
mG

mG
mG

BRaaBB RRS

Meter Background

0.03 mG

‘% ERROR

+7.09%
+7.33%

*7.21%

‘% ERROR

15.10%
+15.55%

+15.32%

‘% ERROR

2.79%
2.6%

2.76%

Meter Background

0.03 mG

ANGULAR RESPONCE

ass

Baxereeass

e209090000-~

RELATIVE RESPONCE

-0.63 dB
70.66 cB

+0.65 db

RELATIVE RESPONCE

-1.42 GB
“1.47 cB

71.44 db

RELATIVE RESPONCE

0.23 dB
0.23 dB

0.23 db


METER CODE
XVITA

MANUFACTURER

Magnetic Technology

Magnetic Tech - Model ELF 12 SN 2
MODEL #

ELF 12

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date

Aug-29-91

RANGE SETTING

BELEZEEB
BaBBRBBS

FREQUENCY RESPONCE PRIMARY DATA

Date

May- 14-91

FREQUENCY

wane
=
*

10 Kz
50 Hz
100 Hz

200 Hz
500 Hz

10 kHz

Time Temperature Meter Background
11:21AM 97° F 0.01 mG
ACTUAL FIELD MEASURED FIELD % ERROR

86.86 mG 96.00 mG 10.52%

86.86 mG 96.00 mG 10.52%

34.53 mG 32.00 mG -7.32%

34.53 mG 32.00 mG 7.32%

11.33 mG 12.00 mG 5.92%

11.38 mG 12.00 mG 5.42%

5.18 mG 4.00 mG 22.77%

5.18 mG 4.00 m6 22.77%

*3.47%
Time Temperature Meter Background
09:48AM 77° F 2 mG

RANGE SETTING ACTUAL FIELD

FREQUENCY RESPONCE AUXILIARY DATA

Date

May-14-91

FREQUENCY

30 Hz
80 Hz
300 Hz
15 Hz

Auto mG 427.67 mG 2.00 mG
Auto me 427.67 mG 64.00 mG
‘Auto me 427.67 mG 96.00 mG
Auto mG 427.67 mG 96.00 mG
Auto mG 136.51 mG $6.00 mG
Auto mG 90.44 mG 96.00 mG
Auto mG 91.90 me 96.00 mG
‘Auto mG 92.18 mG 96.00 mG
Auto mG 120.29 mG 96.00 mG
Auto mG 232.07 mG 96.00 mG
Auto mG 438.92 mG 96.00 mG
Auto mG 873.70 mG 96.00 mG
Auto mG 477.27 mG 16.00 mG
Auto mG 219.18 mG 8.00 mm
Time Temperature Meter Background

10:52AM 77° F 2

RANGE SETTING ACTUAL FIELD

Auto mG 107.58 mG 96.00
Auto mG 88.57 mG 96.00
Auto mG 157.33 mG 96.00
Auto mG 198.66 mG 96.00

Appendix XVIIA

MEASURED FIELD

mG

MEASURED FIELD

mG

mG
mG
ms

SERIAL #
2

AVG % ERROR

RELATIVE RESPONCE

-46.60 dB
+16.50 dB
“12.98 dB
-12.98 dB
+3.06 dB

0.51 dB

0.37 dB

0.35 dB
1.96 GB
7.67 cB
+13.20 dB
-19.18 dB
29.49 dB
28.75 dB

RELATIVE RESPONCE

-0.98 dB
0.69 dB
74.29 cB
6.32 dB


Minimum Detectable Signal and Resolution Measurements

Sensitivity was established by testing the magnetic field meters for
minimum discernable signal at 60 Hz. A remote location in the desert west
of Las Vegas was identified where the ambient magnetic field was
approximately 2uG. Testing took place at this site.

Since minimum discernable signal can be a somewhat subjective reading,
it was repeated a number of times for each meter. To perform the
measurement the Helmholtz coil was configured for 60 Hz operation and the
current reduced to zero. The current was slowly increased until the minimum
signal is discerned. This process was repeated five times. The formula to
calculate the minimum detectable signal was as follows:

MIN_DETECTABLE_SIGNAL = THRESHOLD_CURRENT * 53.95

Resolution measurements were taken with the field typically near 100mG
and the appropriate range setting for the test meter. Next the field was
slowly increased until a discernable change was noted on the meter reading.
The change in drive current (field) was recorded on the data sheet. Meter
resolution was calculated with the following formula.

% RESOLUTION = ABS(START_CURRENT - THRESHHOLD_CURRENT) / START_CURRENT
Results of the resolution and -minimum detectable signal tests are
included in Table 3 on the next page. Minimum detectable signal is given in

microGausse units. Meter resolution is given as a percent at the field
level which the test was performed.

153


9b

2 -

10 4

10 4

Percent Error

-20 4

Accuracy and Linearity - Meter XVIIB
MAG TEC Model ELF 12 SN 4

T T1177 T TTT 17]

10 10?
Applied Field (mG)


SYDKRAFT - MODEL 88.05 - VIIB

MINIMUM DETECTABLE SIGNAL

Date Time

Mar-21-91 12:41PM

Temperature
56° F

Threshold Current mA Threshold Field

1 0.17
Trial a 0.17
Number 3 0.18
4 0.18
5 0.18

METER RESOLUTION DATA

Date Time
Mar-21-91 12:45PM
Meter Range 20uT

Drive Current Start

1 1.8100

Trial 2 1.8100
Number 3 1.8100
4 1.8100

5 1.8100

NOTE: Meter reading changed by 1mG

322323

9.17
9.17
9.71
9.71
9.71

9.5

Temperature
58° F

A Threshold Current

1.8190
1.8180
1.8180
1.8160
1.8170

>> >>>

Appendix VIB

Meter Background

min 0
max 0
uG
uG
uG
uG
uG
uG

mm
mG

UG AVG THRESHOLD FOR SENSITIVITY

Meter Background

Resolution

A

A 485.54
A 431.59
A 431.59
A 323.69
A 377.64

410.01

mG
mG

uG

uG

uG

uG

Resolution %

0.49%
0.44%
0.46%
0.33%
0.38%

0.42%


MSI - MODEL MSI 20 - SN 79010854

METER 60Hz2 E FIELD SUSCEPTIBILITY DATA

Date Time Temperature
Feb-27-91 12:05PM 78° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.0 kV/M #-5%
E Field OFF Reading(ma) E Field ON Reading(mG)
Meter Range Meter ange
200 200
1 0,00 1.10
Trial 2 0.00 1.10
Number 3 0,00 1.20
4 0.00 4.10
5 0.00 1.10
0.00 AVG 1.12
E Field OFF Reading(mG) E Field ON Reading(mG)
Meter Range Meter Range
2000 2000
1 0.00 0.03
Trial 2 0.00 0.03
Number 3 0.00 0.03
4 0.00 0.03
5 0.00 0.03
0.00 AVG 0.03
E Field OFF Reading(mé) E Field ON Reading(mG)
Meter Range Meter Range
20000 20000
1 0.00 0.02
Trial 2 0.00 0.02
Number 3 0.00 0.02
4 0.00 0.02
5 0.00 0.02
0.00 AVG 0.02
E Field OFF Reading(mG) E Field ON Reading(mG)
Meter Range . Meter Range
2006 & 750G 2006 & 750G
1 0.00 0.00
Trial 2 0.00 0.00
Number 3 0.00 0.00
4 0.00 0.00
5 0.00 0.00

Appendix VB


MONITOR - MODEL 42b-1 - SN 1286

METER ANGULAR RESPONCE DATA

Frequency Responce Setting - LINEAR MODE

Date Time Temperature Meter Background
Feb-28-91 12:13PM 74° F 0.25 mG
Angle ACTUAL FIELD MEASURED FIELD ANGULAR RESPONCE

o° 148.76 mG 150.00 mG 1.00

10° 151.84 mG 150.00 mG 0.98

20° 158.37 mG 150.00 mG 0.93

30° 174.04 mG 150.00 mG 0.85

40° 194,83 mG 150.00 mG 0.76

50° 231.69 mG 150.00 mG 0.64

60° 289.82 mG 150.00 mG 0.51

70° 459.91 mG 150.00 mG 0.32

80° 926.73 mG 150.00 mG 0.16

90° 1007.63 mG 10.00 mG 0.00

Frequency Responce Setting - FLAT MODE

Date Time Temperature Meter Background
Feb-28-91 12:29PM The F 0.12 mG
Angle ACTUAL FIELD MEASURED FIELD ANGULAR RESPONCE

0° 149.30 mG 150.00 mG 1.00

10° 150.12 mG 150.00 mG 0.99

20° 157.15 mG 150.00 mG 0.95

30° 169.10 mG 150.00 mG 0.88

40° 191.73 mG 150.00 mG 0.77

50° 224.96 mG 150.00 mG 0.66

60° 297.91 mG 150.00 mG 0.50

70° 412.64 mG 150.00 mG 0.36

g0° 829.26 mG 150.00 mG 0.18

90° 1013.51 mG 7.50 mG 0.00

Appendix VIA


Test Procedures

Test procedures were developed to obtain meter performance data
relative to the following catagories. Every meter was subjected to all
procedures shown below in Table 2.

Test Procedures - Major Categories

1] 60Hz Accuracy and Linearity - High Level Lab Measurements
2] Frequency Response Measurements

3] 60Hz Non-Sinusoidal Response Measurements

4] Angular Response Measurements

5] 60Hz E-Field Susceptibility Measurements

6] Radio Frequency Susceptibility Measurements

7] 60Hz Accuracy & Linearity - Low Level Field Measurements
8] Sensitivity and Resolution Measurements

Table 2

Schematic diagrams of the four test systéms used are shown on the
following pages. These figures can be referenced when reviewing the test
results.


SYDKRAFT - MODEL 88_05 - VIIA

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date

Feb-13-91

RANGE SETTING
200 mG

RANGE SETTING
200 mG

RANGE SETTING
200 mG

Time
03:35PM
SQUARE WAVE TESTING
ACTUAL FIELD

160.73 mG
160.83 mG

SAWTOOTH WAVE TESTING

ACTUAL FIELD

176.80 mG
176.28 mG

SINE WAVE TESTING

ACTUAL FIELD

146,22 m6
165.93 mG

METER ANGULAR RESPONCE DATA

Date

Feb-13-91

Angle

o°
10°
20°
30°
40°
50°
60°
70°
0°
90°

Time

03:52PM

ACTUAL FIELD

166.10
167.78
176.65
191.06
219.80
260.21
340.44
484.92
1059.76
1116.67

BRaAARRARRS

Temperature
76° F

MEASURED FIELD

150.00 mG
150.00 mG

MEASURED FIELD

150.00 mG
150.00 mG

MEASURED FIELD

150.00 mG
150.00 mG

Temperature
76° F

MEASURED FIELD

150.00 mG
150.00.
150.00
150.01

sss

Baaaa
3-5-8 -3-)
22000

BRABRAARB

27.0

Appendix VIIA

Meter Background

0.03

% ERROR

6.68%
-6.73%

+6.70%

‘% ERROR

+15.16%
“14.91%

15.03%

% ERROR

2.58%
2.79%

2.69%

Meter Background
0.05

ANGULAR RESPONCE

RaFSGARRSS

ecole Dep CoCo

RELATIVE R.SPONCE

0.60 dB
0.60 dB

+0.60 db

RELATIVE RESPONCE

+1.42 db

RELATIVE RESPONCE

0.23 db


Ov

J

-10 5

Percent Error

SH

Accuracy and Linearity - Meter XVA - Band Pass
DINDIMA - GAUSSMAUS - SN GMO400

T TT TT TTT] T TTT TTT T TT T1111)

10 10? 107
Applied Field (mG)


02r

OB MOZOVHMD

Susceptibility - Meter 12
WRI Model 670 SN G1259

COFIELD OFF
FELON

(| Mt nl

60 Hz 1 MHz 8610 MHz §=6108 MHz 1 hiiz 18 nix tee tne
E-FIELD 10kV/4 ‘UNHOD


IRC - MODEL IRC 109

METER 60Hz £ FIELD SUSCEPTIBILITY DATA

Date Time Temperature
Feb-27-91 11:20AM 74° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.0 KV/M +-5%
E Field OFF Reading(mG) E Field ON Reading(mG)
Meter Range Meter Range
2mG 2mG
1 0.06 0.15
Trial 2 0.06 0.14
Number 3 0.06 0.13
4 0.06 0.15
5 0.06 0.13
0.06 AVG 0.14
E Field OFF Reading(mG) E Field ON Reading(mG)
Meter Range Meter Range
20mG 20mG
1 0.08 0.15
Trial 2 0.08 0.16
Number 3 0.08 0.16
4 0.08 0.18
5 0.08 0.14
0.08 AVG 0.16
E Field OFF Reading(mG) E Field ON Reading(mG)
Meter Range Meter Range
200mG 200mG
1 0.20 0.30
Trial 2 0.20 0.30
Number 3 0.20 0.30
4 0.20 0.30
5 0.20 0.30
0.20 ANG 0.30
— Field OFF Reading(mG) € Field ON Reading(mG)
Meter Range. Meter Range
2000mG 2000mG
1 0.20 0.20
Trial 2 0.20 0.20
Number 3 0.20 0.20
4 0.20 0.20
5 0.20 0.20
0.20 AVG 0.20

Appendix VA


61

Percent Error

Accuracy and Linearity - Meter ITIA
EXPAN Model 1A SN 1103

10>

5 4

7 ee

5

-10 ay oy ry
1 10 10? 10?

Applied Field (m6)


oI

Percent Error

90

100 —

60 +

70 4

60 +

30 4

40 +

3

20

Accuracy and Linearity ~- Meter I2
Model 670

WRI

SN 61259

197

TT TTT

1

TTT TITTY] TTT TTY

10
Applied Field (mG)

10?

T TTTTTT)
107


Quality Assurance

The construction of a calibrated magnetic field meter is relatively
easy. A small coil is used to transduce the magnetic field to a voltage
proportional to field strength (or time rate of the change of field) and the
voltage can be read using a commercially availab.e voltmeter. The instrument
so built can be calibrated using a large coil as a known field source. The
current measured in the source coil with a common ammeter and the geometry of
the source coil can be used to calculate a standard calibrating field. The
accuracy of the meters is typically quoted in the range of 0.5 to 3%. An
IEEE standard states that meters more than 5% in error should be considered
inaccurate. The data quality objective of this evaluation is that: the EPA
system error should be better than + 2%.

Also of interest was documenting meter susceptibility to fields to
which the instruments should not respond. When an instrument responds to a
factor to which it is not designed to be sensitive, the response may be
unstable. This implies that no special effort should be made to establish
susceptibility to fields with greater than routine accuracy. Two fields used
for testing were radiofrequency fields and 60 hertz electric fields.
Accuracies for the radio frequency fields were + 2 decibels for field
strength and + 5% for amplitude modulation percentages. For 60 hertz
electric fields the accuracy was + 5%.


ze

om MazZoQVvNMD

0.44

I

Susceptibility - Meter VIA - Flat Mode
MONITOR Model 42b-1 SN 1286

!

68 Hz
E-FIELD 18kVM"4

4 MHz 10 MHz 86180 MHz
UNMOD

41 MHz

10 MHz 100 MHz
6eHz MOD

(COFIELD OFF
FIELD ON


DINDIMA -

MINIMUM DETECTABLE SIGNAL
Frequency Responce Setting - WIDE BAND

Date Time

Mar-28-91 01:00PH
Threshold Current mA
1 5.40 mA
Trial 2 5.30 mA
Number 3 5.20 ma
4 5.30 mA
5 5.20 ma

NOTE: Meter reading increased by 0.1mG

This is Peak Value Meter

METER RESOLUTION DATA

Frequency Responce Setting - WIDE BAND

Date Time
Mar-28-91 01:05PM
Meter Range 100m6

Drive Current Start

4 1.4350

Trial 2 1.4350
Number 3 1.4350
4 ~.4.4350

5 1.4350

NOTE: Metér readings increased by 1m

This is Peak Value Meter

GAUSSMAUS - SN GM0400

Temperature
66° F

Threshold Field
291.33
285.96
280.54

285.94
280.54

284.86

Temperature
66° F

A Threshold Current

> rere

1.4490
1.4500
1.4480
1.4470
1.4480

Appendix XVA

Meter Background

0.1
0.1

mm
mG

min
max

uG

uG AVG THRESHOLD FOR SENSITIVITY (RMS)

Meter Background

min 0.1 mG
max 0.4 mG
A Resolution uG Resolution %
A 755.29 uG - 0.97%
A 809.24 uG 1.05%
A 701.34 uG 0.90%
A 647.39 uG 0.83%
A 701.34 uG 0.90%
(RMS) 722.92 uG 0.93%


Olt

Responce

LOA

08

ab

a4

a2 -

a0

Angular Responce - Meter XVA - Wide Band
DINDIMA’- GAUSSMAUS - SN GM0400

10 20 30 40 50 60 70 &0

Rotation Angle

90

+ Meter Responce
— Cosine Curve


69
Responce (dB) - .

_ (1 Square Wave

Frequency Responce - Meter IXA A Sastooth wave
HOLADAY HI-3604 SN 76111 + Sine Wave

— Bezier Fit

TTT Tir TTT TTT TTT TT TTT}

10 10? 10? 104
Frequency (Hz) :


MSI - MODEL MSI 20 - SN 79010854

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date

Jan-28-91

RANGE SETTING
200 mG

RANGE SETTING
200 mG

RANGE SETTING
200 mc

Time
11:20AM

SQUARE WAVE TESTING

ACTUAL FIELD

182.07 mG
182.09 mG

SAWTOOTH WAVE TESTING

ACTUAL FIELD

158.32 mG
158.27 mG

SINE WAVE TESTING

ACTUAL FIELD

151.44 mG
151.52 mG

METER ANGULAR RESPONCE DATA

Date

Jan: 28-91

Angle

0°
10°
20°
30°
40°
50°
60°
70°
80°
90°

Time

11:34AM

ACTUAL FIELD

852.11
1233.95

Temperature
74° F

MEASURED FIELD

150.00 me
150.00 mG

MEASURED FIELD

150.00 mG
150.00 mG

MEASURED FIELD

150.00 mG
150.00 mc

Temperature
74° F

MEASURED FIELD

Appendix VB

Meter Background

0 mG

% ERROR

17.61%
17.62%

“17.62%

‘% ERROR

+5.26%
5.23%

5.24%

‘% ERROR

0.95%
-1.00%

0.97%

Meter Background

0 mG

ANGULAR RESPONCE

SUFSRSRLIS

ec09000000—

RELATIVE RESPONCE

1.68 dB
1.68 dB

-1.68 db

RELATIVE RESPONCE

70.46 dB
-0.46 dB

-0.46 db

RELATIVE RESPONCE

70.08 dB
+0.08 dB

-0.08 db


v2

Accuracy and Linearity - Meter VB
IRC Model IRC-27

10 —
5 4
5
fal o4
WwW
»
5
2
a
5 4
-10 T TT T1111] T TTT 1717 T |
1 10 10? 10?

Applied Field (mG)


eel

OB mMazovuma

Susceptibility - Meter XIVA
EFM Metex-3800 SN 542513

COFIELD OFF
FIELD ON

— ee ee
6e Hz 4 MHz 46 MHz =6168 MHz (MHz 10 MHz 100 MHz
E-FIELD 10K UNMOD 6eHz MOD


SYDKRAFT - MODEL 88 05 - VIIA
FREQUENCY RESPONCE PRIMARY DATA

Date Time Temperature Meter Background
Feb-13-91 02:07PM 76° F 0.03 mG

FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

1 Hz 2m 1018.13 mG 0.07 mG 783.25 dB

2 Hz 2 mG 1018.13 mG 0.12 mG +78.57 dB

5 Hz 2 mG 1018.13 mG 0.16 mG +76.07 cB

10 Hz 2 mG 1018.13 mG 0.32 mG +70.05 dB

20 Hz 2 mG 970.72 mG 1.96 mG -53.90 dB

50 Hz 200 mG 641,33 mG 150.00 mG -12.62 dB

60 Hz 200 mG 146.20 mG 150.00 mG 0.22 dB

100 Hz 200 mG 1012.20 mG 67.00 mG +23.58 dB

200 Hz 20 mG 1002.41 mG 16.20 mG 35.83 dB

500 Hz 20 mG 251.09 mG 1.20 mG 46.41 dB

1 kHz 20 mG 206.51 mG 0.40 mG 54.26 dB

FREQUENCY RESPONCE AUXILIARY DATA

Date Time Temperature Meter Background
Feb-13-91 02:26PM 76° F 0.03 mG

FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

30 Hz 20 mG 1029.14 mG 10.00 mG +40.25 dB

40 Hz 200 mG 1040.71 mG 43.00 mG 27.68 dB

80 Hz 200 mG . 850.10 mG 150.00 mG -15.07 dB

55 Wz 200 mc 228.54 mG 150.00 mG +3.66 GB

65 Hz 200 me 168.43 mG 150.00 mG 71.01 dB

Appendix VITA


RF SUSCEPTIBILITY TEST SETUP
SCHEMATIC DIAGRAM

HP S9305A
RELAY ACTUATOR

HP 3325A
SYNTHESIZER
ouT
9
HP 8660B RF
SYNTHESIZER ied OUT
9

HP 436A
POWER METER
HP 8484A
POWER SENSOR
HP 8566A a
X HP 8761A
SPECTRUM ANALYZER Y RF SWITCH
rd

ATTENUATOR

YELLOW SYSTEM

AMPLIFIER RESEARCH
10W1000 © N
AMPLIFIER our

BOONTON 91-12

BOONTON 9200

ati RF PROBE

VOLTMETER

DUT tT
FOAM 0.4m
110 MHz BLOCK v
LOW PASS
FILTER
IFI CC 101.5
TEM CELL

10



eae

On mMozZOVvTUMsD

Susceptibility - Meter X
WALKER ELF-50D SN71181-205

OOFIELD OFF
(FELD ON

o of of 3 ome Om 2

60 Hz 4 HHz = 10 MHz = 108 MHz 4 MHz = 10 MHz 100 Hz
E-FIELD 10k¥At ‘UNHOD 6eHz HOD


SAFE - SAFE PRO METER - SN 1264
METER CODE MANUFACTURER METER NAME MODEL # SERIAL #
M2 Safe Computing Company Safe Meter 1264

368 Hillside Avenue
Needham, MA 02194

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Date Time Temperature Meter Background: mG
Feb-06-91 09:39AM 76° F 0.2

RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
200 me 142.45 mG 151.20 mG 6.14%

23.40 mG 25.00 mG 6.85%

200 mG 70.68 mG 75.30 mG 6.53%

23.52 mG 25.10 mG 6.71%

6.56% AVG % ERROR

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date Time Temperature Meter Background
Har-18-91 11:13AM oF 0.1 7G
RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR
94.61 mG 100.00 mG «5.92%
200 mG 96.41 mG 100.00 me 5.92%
18.83 mG 20.00 mG 6.22%
18.83 mG 20.00 mG 6.22%
9.44 mG 10.00 m6 5.92%
9.44 mG 10.00 mG 5.92%
1.89 mG 2.00 mG 5.92%
1.89 mG 2.00 me 5.92%
0.94 mG 1.00 mG 5.92%
0.96 mG 1.00 mG 5.92%
5.98% AVG % ERROR

Appendix 112


62

Accuracy and Linearity - Meter VIIA
" SYDKRAFT Model 88-05

10 -
54
c o-
uj
Ps)
G
oO
&
-5 4
LO i a oo oy
1g? 1 10 10? 10?

Applied Field (mG)


WRI - Model 670 - SN G1259
METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date Time Temperature

Feb-27-91 01:37PM 78° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.00 KV/M 4-5%
E Field OFF Reading(ms) E Field ON Reading(ma)
Meter Range Meter Range
100*1 100*1
1 0.30 1.50
Trial 2 0.30 1.50
Number 3 0.30 1.50
4 0.30 1.50
5 0.30 1.50
0.30 AVG 1.50
E Field OFF Reading(mG) E Field ON Reading(ms)
Meter Range Meter Range

100*10 100*10
1 1.90 : 3.00
Trial 2 1.90 3.00
Number 3 1.90 3.00
4 1.90 3.00
5 1.90 3.00
1.90 AVG 3.00

Appendix 12


WALKER - ELF-50D - SN 71181-204

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date Time

Feb-07-91 12:03PM

SQUARE WAVE TESTING

RANGE SETT.NG ACTUAL FIELD
LO mG 145.36 mG
145.18 mG

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD
LO mG 134.75 mG
134.62 mG

SINE WAVE TESTING

RANGE SETTING ACTUAL FIELD
LO mG 151.79 mG
152.06 mG

METE! IGULAR RESPONCE DATA

Date Time
Feb-07-91 12:20PM
Angle ACTUAL FIELD
o° 152.33 mG
10° 156.38 mG
20° 164.11 mG
30° 182.89 mG
40° 210.11 mG
50° 254.84 mG
60° 344.36 mG
70° 546.87 mG
go° 1024.69 mG
90° 1109.56 mG

Temperature
77° F

MEASURED FIELD

150.50 mG
150.50 mG

MEASURED FIELD

150.50 mG
150.50 mG

MEASURED FIELD

150.00 mG
150.00 mG

Temperature
TF

MEASURED FIELD

a
2
2
8
a

a
o
2
Ss

SSSsssss

Appendix XA.

s8éssssss

BaRBRBRAR

Meter Background
one

‘% ERROR

3.54%
3.66%

3.60%

% ERROR

11.69%
11.79%

11.74%

‘% ERROR

18%
71.36%

71.27%

Meter Background
0 7G

ANGULAR RESPONCE

ereessecee=
SSVESAaSLS

RELATIVE RESPONCE

0.30 dB
0.31 dB

0.30 db

RELATIVE RESPONCE

0.96 dB
0.96 dB

0.96 db

RELATIVE RESPONCE

-0.10 dB
70.11 dB

70.11 db


SAFE - SAFE PRO METER - SN 1264

METER 60Hz NON-SINUSOIDAL RESP\ ATA
Date Time Temperature Meter Background
Feb-06-91 11:00PM 76° F 0.2 mG

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR RELATIVE RESPONCE
200 mG 921.26 mG 92.20 mG 89.99% 19.99 dB

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELO MEASURED FIELD % ERROR RELATIVE RESPONCE
200 mG 213.76 mG 150.30 mG + 29.69% -3.06 cB

SINE WAVE TESTING

RANGE SETTING ACTUAL FIELD . MEASURED FIELD ‘% ERROR RELATIVE RESPONCE
200 mG 148.58 mG 150.00 mG 0.95% 0.08 dB

IETER_ANI RESPONCE DATA
Date Time: Temperature Meter Background
Feb-06-91 11:28AM 76° F 0.2 mG
Angle ACTUAL FIELD MEASURED FIELD ANGULAR RESPONCE
o° . 148.88 mG 150.10 mG 1.00 .

10° . 150.07. mG 150.20 mG 0.99

20° . 155.86 mG 150.50 mG 0.95

30° 167.21 mG - 150.60 mG 0.89

40°. 184.97 mG 150.10 mG 0.80

50° 217.51 mG 150.30 mG 0.68

60° 275.78 mG 149.70 mG 0.53

70° 387.93 mG 149.70 mG 0.38

B0° ~ 696.62 mG 150.10 mG 0.21

90° 942.00 mG 45.40 mG 0.04

Appendix 112


WALKER - ELF-SOD - SN 71181-205
METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date Time Temperature

Feb-25-91 02:57PM 78° F
& Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.0 kV/M +-5%
E Field OFF Reading(mG) E Field ON Reading(mG)
Meter Range Meter Range
Lo Lo

1 0.00 6.00
Trial 2 0.00 6.00
Number 3 0.00 6.00
4 0.00 6.00
5 0.00 6.00
0.00 AVG 6.00

Meter Range Meter Range

HI HI

1 0.00 20.00
Trial 2 0.00 20.00
Number 3 0.00 20.00
4 0.00 20.00
5 0.00 20.00
0.00 AVG 20.00

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Date Time Temperature
Feb-22-91- 09:58AM 79° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Lo LO
1 MHZ 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 +17.00
Meter Range Meter Range
HI HI
1 MHz 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 80.00

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date Time Temperature
Feb-22-91 10:20AM 79° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Lo Lo
1 MHz 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 +11.00
Meter Range Meter Range
HI HI
1 MHz 0.00 0.00
10 MHz 0.00 0.00
100 MHz 0.00 -80.00

Appendix x


HOLADAY - HI 3604 - SN 76112
METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date Time Temperature

Feb-25-91 "03:28PM 7B° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.0 kV/M +-5%
E Field OFF Reading(mG) —_E Field ON Reading(mc)
Meter Range Meter Range

Auto Auto
1 0.16 0.16
Triat 2 0.16 0.16
Number 3 0.16 0.16
4 0.16 0.16
5 0.16 0.16
0.16 ANG 0.16

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Date Time Temperature
Feb-22-91 12:39PM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Auto Auto
1 MHz 0.16 0.16
10 MHz 0.16 0.16
100 MHz 0.16 0.16

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date Time Temperature
Feb-22-91 12:45PM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Auto Auto
1 MHZ 0.16 0.16
10 MHz 0.16 0.16

100 MHz 0.16 0.16

Appendix  IXB


86

Responce

10-
2.8 -
06 -
a4-

422-5

a0 -

Angular Responce - Meter VIB - Linear Mode
MONITOR

10

20

30

Model 42b-1

40 50

Rotation Angle

SN 1287

60

70

60

90

+

Meter Responce

Cosine Curve


Ort

an MOzZOVUMD

0.4

I

Susceptibility ~ Meter IXB
HOLADAY HI-3604 SN 76112

Va

WL

60 Hz
E-FIELD 10k¥4

4 MHz 10 MHz «6100 Hitz
UNTOD

1 Hz 1@ MHz 100 MHz
6eHz MoD

OFIELD OFF
FELDON


DINDIMA - GAUSSMAUS - SN GMO400
METER ANGULAR RESPONCE DATA

Frequency Responce Setting - WIDE BAND

Dete Time Temperature Meter Background

May-17-91 08:39AM The F 1 mG
Angle ACTUAL FIELD MEASURED FIELD METER RESPONCE

o° 113.76 mG 99.00 mG 1.00

10° 114.27 mG 99.00 mG 0.99

20° 118.80 mG 99.00 mG 0.95

30° 132.32 mG 99.00 mG 0.86

40° 146.80 mG 99.00 mG 0.77

50° 176.58 mG 99.00 mG 0.64

60° 245.68 mG 99.00 mG 0.46

70° 345.44 mG 99.00 mG 0.32

60° 718.00 mG 99.00 mG 0.15

90° 1441.51 mG 30.00 mG 0.02

Frequency Responce Setting - BAND PASS

Date Time Temperature Meter Background

May-16-91 12:31PM . 76° F 1 mG
Angle ACTUAL FIELD MEASURED FIELD METER RESPONCE

o° 108.61 mG 99.00 mG 1.00

10° 111.18 mG 99.00 mG 0.97

20° 119.36 mG 99.00 mG 0.91

30° 123.82 mG 99.00 mG 0.87

40° 140.33 mG 99.00 mG 0.77

50° 167.10 mG 99.00 mG 0.65

60° 221.79 mG 99.00 mG 0.49

70° 318.50 mG 99.00 mG 0.34

80° 663.96 mG 99:00 mG 0.16

90° 1501.08 me 10.00 mé 0.00

‘Appendix XVA


16

Responce

10-

ae

ab 4

a4-

42H

ao-
o

10

Angular Responce - Meter IIIB

EXPA!:

20

30

Model 1A

4050

Rotation Angle

SN 1104

60

70

80

+

Meter Responce
Cosine Curve


Magnetic Tech - Model ELF 12 SN 2

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

METER 60Hz NON-SINUSOTDAL RESPONCE DATA

Date Time

May- 14-91 10:55AM

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD
Auto mG 98.76 mG
98.78 mG

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD
Auto mG 81.30 mG
82.49 mG

SINE WAVE TESTING

RANGE SETTING ACTUAL FIELD
Auto mG 89.54 mG
92.67 mG

METER ANGULAR RESPONCE DATA

Date Time
May- 14-91 12:08PM
Angle ACTUAL FIELD
o° 89.77 mG
10° 93.27 mG
20° 102.91 mG
30° 104.52 mG
40° 417.64 mG
50° 140.51 mG
60° 185.42 mG
70° 280.53 mG
80° 583.46 mG
90° 1122.11 mG

Temperature
TT? F

MEASURED FIELD

96.00
96.00

MEASURED FIELD

96.00
96.00

MEASURED FIELD

00
96.00

Temperature
77° F

MEASURED FIELD

a

wSSSRRERRE
288888888
BRRBBRBBA

n
Appendix XVIIA

mG
mG

mTM
mG

mG
mG

Meter Background

2 mG

‘% ERROR

*2.79%
-2.82%

72.80%

‘% ERROR

18.09%
16.38%

17.23%

% ERROR

7.21%
3.59%

5.40%

Meter Background

2 mG

METER RESPONCE

1.00
0.96
0.87
0.85
0.76
0.63
0.48
0.32
0.15
0.00

RELATIVE RESPONCE

70.24 GB
70.24 cB

0.24 db

RELATIVE RESPONCE

1.44 cB
1.32 dB

1.38 db

RELATIVE RESPONCE

0.60 dB
0.30 dB

0.45 db


IRC > MODEL IRC 27

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date Time

Feb-12-91 03:11PM

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD
200 mG 158.35 mG
158.15 mG

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD
200 mG 172.83 mG
172.98 mG

SINE WAVE TESTING

RANGE SETTING ACTUAL FIELD

200 mG 142.92 mG
142.87 mG

METER ANGULAR RESPONCE DATA

Date Time
Feb: 12-91 03:23PM
Angle ACTUAL FIELD
o 142.92 mG
10° 145.13 mG
20° 151,02 mG
30° 164.56 mG
40° 186.05 mG
50° 219.25 mG
60° 275.61 mG
70° 401.32 mG
80° 796.60 mG
90° 997.02 me

Temperature Meter Background
76° F 0.1 mG
MEASURED FIELD ‘% ERROR
150.20 mG 75.14%
149.95 mG 75.18%
+5.16%
MEASURED FIELD ‘% ERROR
149.85 mG 13.29%
149.90 mG 13.34%
+13.32%
MEASURED FIELD ‘% ERROR
150.00 mG 4.95%
149.90 mG 4.92%
4.96%
Temperature Meter Background
76° F 0 mG

MEASURED FIELD ANGULAR RESPONCE

150.10 mG 1.00
149.95 mG 0.98
150.00 mG 0.94
149.85 mG 0.86
149.80 mG 0.77
149.90 mG 0.65
150.00 mG 0.51
150.10 mG 0.35
149.95 mG 0.17

9.50 mG 0.00

Appendix VB

RELATIVE RESPONCE

-0.45 dB
0.46 cB

70.46 db

RELATIVE RESPONCE

1.26 aB
+1.26 GB

1.24 db

RELATIVE RESPONCE

0.42 dB
0.41 dB

0.41 db


EXP - ELF - MODEL 1A - SN 1103
MINIMUM DETECTABLE SIGNAL

Resolution %

0.97%
0.78%
0.76%
1.13%
0.86%

Date Time Temperature Meter Background
Mar-18-91 12:50PM 68° F min 0.1 mG
max 0.1 mG

Threshold Current m\ Threshold Field uG

1 1.40 mA 75.53 uG

Trial 2 1.60 ma 86.32 uG

Number 3 1.50 ma 80.93 uG

4 1.80 mA 97.11 UG

5 1.80 ma 97.11 UG

87.4 UG AVG THRESHOLD FOR SENSITIVITY
METER RESOLUTION DATA
Date Time Temperature Meter Background
Mar- 18-91 12:56PM 68° F min 0.1 mG
max 0.1 mG
Meter Range 30 mG

Drive Current Start A Threshold Current A Resolution uG
1 0.3800 A 0.3837 A 199.61 uG
Trial 2 0.3800 A 0.3830 A 161.85 uG
Number 3 0.3800 A 0.3829 A 156.45 uG
4 0.3800 A 0.3843 A 231.98 uG
5 0.3800 A 0.3833 A 178.03 uG
185.58 uG

Appendix I1IA

0.90%


Internal Quality Control Checks

All the procedures except 5 & 6 from table 2 on page 6 were performed
in a Helmholtz coil system designed for generating known magnetic fields.
The quality control check on the Helmholtz coil system took the form of
perindic characterization of the circuits used to operate the coil. This
characterization is described in the test plan (QAPP) standard operating
procedures as the ’Current Shunt DC Calibration’. This periodic calibration
procedure ensured that the Helmholtz coil system was functioning properly and
accurately.

Additional verification was provided by a ’Standard Meter Check’. In
this check one meter was chosen as a control against which the Helmholtz
coil’s performance over time was monitored. The control meter need not be
especially accurate; the absolute value of the field in the coil is not to
be gauged by the meter reading. The field value measured over time however
should not deviate. This procedure provided a backup check against unforseen
failure modes.

Two other instrumentation systems were used in this test effort.
Together they provided for the susceptibility testing portions of this test
sequence (categories 5 & 6 from Table 2). The E field cell is a parallel
plate structure which was energized with a l0kV/m 60Hz E field. The field
was verified using both a low level voltage measurement from a voltage
divider circuit, and a measurement of the actual voltage on the E field cell
plates themselves using an electrostatic voltmeter. A 60Hz electric field
meter was also used to check the electric field.

The RF TEM cell is the last piece of instrumentation. The value of the

field in the TEM cell is established by measuring the power output of a known
attenuator system connected to the output port of the cell.

11


16

Responce

LOA

a8 -

a6 -

a4

a2-

00-

Angular Responce - Meter VIA - Flat Mode
MONITOR Model 42b-1 SN 1286

10

20

30

40 50

Rotation Angle

60

70

80

90

+

Meter Responce

Cosine Curve


SAFE - SAFE PRO METER - SN 1264
METER 60H2 E FIELD SUSCEPTIBILITY DATA

Date Time Temperature

Feb-27-91 12:40PM 78° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.0 kV/M +-5%
E Field OFF Reading(mG) E Field ON Reading(m

Meter Range Meter Range
Auto Auto
1 0.20 . 0.75
Trial 2 0.20 0.55
Number 3 0.20 0.55
4 0.20 0.75
5 0.20 0.55
0.20 AVG 0.63

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Date Time Temperature
Feb-21-91 10:56AM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Auto Auto
1 MHz 0.15 0.15)
10 MHz 0.15 0.15
100 MHz 0.15 1.50

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date Time Temperature
Feb-21-91 11:36AM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Auto Auto
4 MHz 0.15 0.15
10 Miz 0.15 0.30
100 Miz 0.15 1.80

Appendix 112


201

Responce

10

Angular Responce - Meter VIIIA
EMD Model 'ACGM-1 SN 01-0116

20 30 40 50 60 70

Rotation Angle

80

90

-++ Meter Responce
— Cosine Curve


MONITOR - MODEL 42b-1 - SN 1286

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Frequency Responce Setting - LINEAR MODE

Date

Mar-26-91

RANGE SETTING

100.0 mG
50.0 mG
25.0 mG
10.0 mG
5.0 mG
2.5 mG
1.0 mG
0.5 mG

Time

41:07AM.

ACTUAL FIELD

conus S83

BySeeue.
BRRRRBBB

Frequency Responce Setting - FLAT MODE

Date

Mar-26-91

RANGE SETTING

100.0 mG
50.0

Time

41:19AM

ACTUAL FIELD

S3

Sorunes

easeesne
BRRBRRBR

Temperature
59° F

MEASURED FIELD

Temperature
56° F

MEASURED FIELD

Appendix VIA

Meter Background

0.5

% ERROR

0.73%
0.32%
1.29%
0.87%
1.57%
1.29%
1.22%
1.29%

1.07%

Meter Background

0.5

‘% ERROR

0.19%
0.60%
1.01%
1.57%
1.57%
1.29%
2.12%
2.61%

1.34%

mG

mG

AVG % ERROR

AVG % ERROR


MEDA - Model PLM-100 - SN 1381

FREQUENCY RESPONCE PRIMARY Di

Date Time Temperature Meter Background
May-15-91 08: 164M 76° F 0.1 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
5 Hz 200 mG 998.91 mG 1.50 mG +56.47 dB
10 Hz 200 mG 998.91 mG 10.10 mG -39.90 dB
20 Hz 200 mG 983.18 mG 77.40 mG -22.08 dB
50 Hz 200 mG 696.56 mG 150.00 mG =13.31 cB
60 Hz 200 m6 150.60 mG 150.00 mG -0.03 dB
100 Hz 200 m6 998.63 m6 58.70 mG +26.62 dB
200 Hz 200 mc 1018.03 mG 20.70 mG -33.84 dB
500 Hz 200 mG 1029.78 mG 7.50 mG 242.75 dB
1 kHz 200 mG 1029.06 mG 3.60 mG -49.12 dB
2 kHz 200 mG 925.26 mG 1.70 mG -56.72 dB
FREQUENCY RESPONCE AUXILIARY DATA
Date Time Temperature Meter Background
May-14-91 08:43AM 76° F 0.1 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
70 Hz 200 mG 508.72 mG 150.00 mG -10.61 dB
80 Hz 200 m6 959.59 mG 105.30 mG 719.19 dB
90 Hz 200 mG 1014.78 mG 74.70 mG +22.66 dB
110 Hz 200 mG 1013.11 mG 48.60 mG +26.38 dB
120 Hz 200 mG 1013.61 mG 41.40 mG +27.78 dB
140 Hz 200 m6 1022.45 mG 33.00 mG -29.82 dB
180 Kz 200 mG 1011.23 mG 23.50 mG -32.68 dB
190 Hz 200 mG 1008.59 mG 21.90 mG +33.27 dB
40 Hz 200 mG 999.85 mG 72.10 mG +22.84 dB
30 Hz 200 mG 1003.78 mG 61.80 mG +24.21 dB
51 Hz 200 mG 564.75 mG 150.00 m6 -11.52 dB
52 Hz 200 mG 492.18 mG 150.00 mG
53 Hz 200 mG 422.45 mG 150.00 mG
54 Hz : 200 mG 358.17 mG 150.00 mG
55 Hz 200 m6 300.75 mG 150.00 mG
56 Hz 200 mG 255.32 mG 150.00 mG
57 Hz 200 mG 216.95 mG 150.00 mG
58 Hz 200 m6 181.96 mG 150.00 m6
59 Hz 200 mG 159.76 me 150.00 mG
61 Hz 200 mG 149.79 mG 150.00 mG
62 Hz 200 m6 176.90 mG 150.00 mG
63 Hz 200 mG 205.27 mG 150.00 mG
64 Hz 200 mG 239.57 mG 150.00 mG
65 Hz 200 mG 276.01 mG 150.00 mG
66 Hz 200 mG 316.67 mG 150.00 mG
67 Hz 200 mG 366.03 mG 150.00 mG
68 Hz 200 mG 413.09 mG 150.00 mG
69 Hz 200 m6 462.59 mG 150.00 mG -9.78 dB

Appendix XVIB


DINDIMA - GAUSSMAUS - SN GM0401
METER ANGULAR RESPONCE DATA

Frequency Responce Setting - WIDE BAND

Date Time Temperature Meter Background
May-17-91 09:49AM T° F 1 mG
Angle ACTUAL FIELD MEASURED FIELD METER RESPONCE
112.27 mG 99.00 mG 1.00
113.07 mG 99.00 mG 0.99
116.62 mG 99.00 mG 0.96
127.55 mG 99.00 mG 0.88
143.74 mG 99.00 mG 0.78
171.03 mG 99.00 mG 0.65
220.04 mG 99.00 mG 0.51
344.84 mG 99.00 mG 0.32
666.43 mG 99.00 m6 0.16
1533.61 mG 31.00 mG 0.02
Frequency Responce Setting - BAND PASS
Date Time Temperature Meter Background
May- 16-91 10:37AM 76° F 1 mG
Angle ACTUAL FIELD MEASURED FIELD METER RESPONCE
o° 110.26 mG 99.00 mG 1.00
10° 106.75 mG 99.00 mG 1.03
20° 119.61 mG 99.00 mG 0.92
30° 124.28 mG 99.00 mG 0.88
40° 138.47 mG 99.00 mG 0.79
so° 166.61 mG 99.00 mG 0.66
60° 215.12 mG 99.00 mG 0.51
70° 306.31 mG 99.00 mG 0.36
80° 676.41 mG 99.00 mG 0.16
16.00 mG 0.01

90°" . 1475.83 mG

Appendix XVB



MSI - MODEL MSI-20 - SN 79011626

METER 60Hz NON-SINUSOIDAL RESPONCE DATA

Date Time Temperature Meter Background
Jan-25-91 01:00PM 74° F 0 mG

SQUARE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR RELATIVE RESPONCE
200 mG 184.25 mG 150.00 mG +18.59% 1
184.43 mG 150.00 mG + 18.67% *1.79 GB
18.63% *1.79 db

SAWTOOTH WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR RELATIVE RESPONCE
200 mG 160.11 mG 150.00 mG -6.31% 70.56 dB

160.13 mG 150.00 mG +6.33% 70.56 oB

-6.32% +0.56 db

SINE WAVE TESTING

RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR RELATIVE RESPONCE
200 mG 153.25 mG 150.00 mG 72.12% 70.18 dB

153.38 mG 150.00 mG 2.20% 70.19 dB

72.16% -0.19 db

METER ANGULAR RESPONCE DATA

Date Time Temperature Meter Background
dan-25-91 01:30PM The F 0 mG
Angle ACTUAL FIELD MEASURED FIELD ANGULAR RESPONCE

o° 153.30 mG 150.00 mG 1.00

10° 157.00 mG 150.00 mG 0.97

20° 165.82 mG 150.00 mG 0.92

30° ~ 181.52 mG 150.00 mG 0.84

40° 207.38. mG 150.00 mG 0.73

50° 250.50 mG ~ 150.00 mG 0.61

60° 330.43 mG 150.00 mG 0.46

70° 538.00 mG 150.00 mG 0.28

80° 1169.92 mG 150.00 mG 0.13

90° 1246.20 mG 59.70 mG 0.04

Appendix IVA


EXP - ELF - MODEL 1A - SN 1103

METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date Time Temperature

Feb-27-91 12:34PH 78° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.0 kV/M +-5%
E Field OFF Reading(mG) E Field ON Reading(mG)

Meter Range Meter Range
3m 3G
1 0.10 0.10
Trial 2 0.10 0.10
Number 3 0.10 0.10
4 0.10 0.10
5 0.10 0.10
0.10 AVG 0.10

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA ~ UNHODULATED

Date Time Temperature
Feb-20-91 10:15AM . 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
3mG 3mG
1 MHZ 0.05 0.05
40 MHz 0.05 0.05

100 MHz 0.05 0.05

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date Time Temperature
Feb-20-91 10:15AM 80° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
‘3mG 3G
1 MHZ 0.05 0.05
10 MHZ 0.05 0.05
100 MHz 0.05 0.15

Appendix 111A


Meter 60Hz E-Field Susceptibility Measurements

The meters were tested for susceptibility to external E fields. Each
meter was placed inside a parallel plate test fixture and subjected to a 60
Hz electric field of 10.0kV/m. Any deviations in the meter display were
recorded. The most sensitive range on each meter was selected to begin this
test. Each range on the meter was tested in like manner. The recorded data
is meter readings with and without the E field applied.

The recorded data shown on the graphs on the following pages show
readings at the lowest meter range only. These graphs are combined with the
radio frequency susceptibility data to create one susceptibility chart.

Meter Radio Frequency Susceptibility Measurements

The magnetic field meters were tested for susceptibility to radio
frequency field. The goal of this portion of the test was to determine the
potential for meter response when operated in the vicinity of radio frequency
sources (AM,FM, & TV broadcast towers). The test apparatus for this was a
TEM cell. The field strength for this testing was 10V/m. Testing was
executed at the discreet frequencies 1MHz, 10MHz, 100MHz. The test objects
were oriented in the TEM cell such that the TEM magetic field would not
couple to the pickup coil. At E = 10V/m the radio frequency magnetic field
in the TEM cell is 0.33 mG. Testing was performed on all available range
settings on each test meter. Graphs of this data on the following pages show
results only at the lowest range.

118


s9

Responce (dB)

40 -

20

-20 +

-40 5

- (1 Square wave

Frequency Responce Meter VIIA Sawtooth Wave
SYDKRAFT Model 88-05 ++ Sine Wave

— Bezier Fit

T TT TTT 177 T TTT T TTT TTT]

10 10? 107
Frequency (Hz)


SYDKRAFT - MODEL 88.05 - VIIA

Date Time

Mar-21-91 12:07PM
Threshold Current mA

Trial
Number

wens
eeese
USeTs
232222

WOTE: Meter reading increased by 0.01mG

METER RESOLUTION DATA

Date Time
Mar-21-91 12:13PM
Meter Range 20uT
Drive Current Start A
1 1.8100 A
Trial 2 1.8000 A
Number 3 1,8000 A
4 1.8000 A
5 1.8000 A
6 1.8000 A

NOTE: Meter reading increased by about 1mG

Temperature

64° F

Threshold Field

10.79
11.87
11.33
10.79
12.41

11.44

Temperature
5e° F

Threshold Current

1.8190
1.8140
1.8160
1.8180
1.8150
1.8160

Appendix VITA

Meter Background

min 0 mG
max 0 mG
uG
uG
uG
uG
UG
uG

uG AVG THRESHOLD FOR SENSITIVITY

Meter Background

min 0 mG
max oO mG
A Resolution uG Resolution %
A 485.54 uG 0.49%
A 735.29 uG 0.77%
A 863.19 uG 0.88%
A 971.08 uG 1.00%
A 809.24 uG 0.83%
A 863.19 UG 0.88%
791.25 uG 0.81%


v4

- (Square Wave
Frequency Responce - Meter XA T_ Saitooth Wave
WALKER ELF-50D SN 71181-204 + Sine Wave
40 5 — Bezier Fit
20 4
o-4
g
3 20
2 .29-
2
oO
rd
-40 4
-60 a TT]
1 10 10? 107 104

Frequency (Hz)


MSI - MODEL MSI 20 : SN 79011626

FREQUENCY RESPONCE PRIMARY DATA

Date

Jan-25-91

FREQUENCY

Hz
Hz
Hz
Wz
Hz
Hz
Hz
Hz
Wz
Hz
kiz
kuz
kRz
kHz

Time

10:31AM

RANGE SETTING

200 mG
200 mG
200 mG

200 mG ~

* 200 mG
200 mG
200 mG
200 mG
200 mG

, 200 mG
200 mG

200 m6 ©

200 mG
200 mG

FREQUENCY RESPONCE AUXILIARY DATA

Date

Jan-25-91

FREQUENCY

80
150
30
300
4.5
3

4

Time

11:05AM

RANGE SETTING

200 mG
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG

Temperature
The F

ACTUAL FIELD

1014.70
1014.70
1014.70
1014.70
602.58
187.21
153.45
104.17
176.44
472.75
966.79
1173.57
938.97
549.25

Temperature
“me F

ACTUAL FIELD

115.97
128.09
323.40
274.32
1134.45
1112.22
1010.01

Appendix IVA

ABRBRABRBARBR BBS

BaBBRaB

Meter Background

0

mG

MEASURED FIELD

2.60
26.80
68.50
401.30
149.80
150.00
150.00
150.00
150.00
150.00
150.00
89.00
27.20
6.50

Meter Background

0

BASRRRARRARARB

mG

MEASURED FIELD

150.00
150.00
150.30
150.10
116.50

55.80

37.40

BaBRRBS

RELATIVE

RELATIVE

RESPONCE

-51.83 dB
-31.56 dB
723.41 GB
720.01 dB
712.09 dB
“1.92 cB
70.99 cB

3.17 dB
+1.31 GB
°9.97 OB
+16.18 dB
22.40 B
30.76 dB
+38.54 dB

RESPONCE


MEDA - Model PLM-100 - SN 1380

FREQUENCY RESPONCE PRIMARY DATA

Date

May- 14-91

FREQUENCY

Time

01:18PM

RANGE SETTING

200 me
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG

FREQUENCY RESPONCE AUXILIARY DATA

Date

May-14-91

FREQUENCY

Time

02:18PM

RANGE SETTING

200 mG
200 m6
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG
200 me
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG
200 mG
200 m6
200 mG

Temperature
77 F

ACTUAL FIELD

Temperature
77 F

ACTUAL FIELD

g

&
S
SANSNLUNKESSUASGRIaAa

Nees
SS8E5

185.

wuunpeece
SESSB8 RRS

445.

BB2ARRRRARBRARRRBARBR

Appendix XVIA

Meter Background

0.1 mG

MEASURED FIELD

1.70
12.45
79.70

137.50
150.05
50.50
17.80
6.50
3.10

BaRBRBBRR

Meter Background

0.1 mG

MEASURED FIELD

150.25
90.80
64.30
41.90
36.00
28.40
20.30
19.00
63.00
44.90

150.00

150.00

150.00

150.00

150.00

150.00

150.00

150.00

10

BARR RAARRRRARRRARRARRRRRRARR

RELATIVE

RELATIVE

RESPONCE

-55.26 dB
+37.96 cB
721.63 dB
+17.99 cB

0.00 dB
25.91 oB
+35.14 dB
743.04 dB
47.53 dB

RESPONCE

-10.70 dB
-19.80 dB
-23.31 GB
427.59 dB
-28.96 dB
+34.23 GB
-34.31 dB
+35.47 GB
+24.52 cB
+27.44 GB
*15.40 oB
-10.42 GB
+9.04 dB
7.68 dB
-6.21 cB
+4.79 oB
73.29 cB
+1,83 dB
0.59 GB
70.28 dB
*1.25 oB

9.46 OB


91

Percent Error

60 -

30 +

40 -

30 4

20 +

10 +

Accuracy and Linearity - Meter IA2
WRI Model 660 SN G1242

T T TTT TTT T TT TT TITT T TT TTT]

1 10 10?
Applied Field (mG)


WRI - Model 670 - SN G1257

METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date Time Temperature

Feb-27-91 01:53PM 78° F
E Field OFF Reading E Field ON Reading
EFM Metex 0.00 10.00 KV/M +-5%
E Field OFF Reading(mG) ° & Field ON Reading(m)

Meter Range Meter Range
100"1 100*1
1 0.10 1.00
Trial 2 0.10 1,00
Number 3 0.10 1.00
4 0.10 1.00
5 0.10 1.00
0.10 AVG 1.00

E Field OFF Reading(ma) E Field ON Reading(mG)

Meter Range Meter Range
100*10 100*10
1 0.00 1.00
Trial 2 0.00 1,00
Number 3 0.00 1.00
4 0.00 1.00
5 0.00 1.00
0.00 AVG 1.00

Appendix 11


MONITOR - MODEL 42b-1°- SN 1287

FREQUENCY RESPONCE PRIMARY DATA

Frequency Responce Setting - LINEAR MODE

Date Time | Temperature Meter Background
dan-29-91 11:25AM The F 0.2 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
THz 1 mG 935.64 mG 0.52 mG -65.10 dB
2 Hz 1 mG 935.64 mG 0.56 mG 64.46 cB
5S Hz 1 mG 935.64 mG 0.82 mG 761.15 dB
10 Hz 25 mG 935.64 mG 7.50 mG -41.92 GB
20 Hz 250 mG 489.07 mG 150.00 mG -10.27 cB
50 Hz 250 mG 177.32 mG 150.00 mG +1.45 cB
60 Hz 250 mG 167.24 mG 150.00 mG 0.16 dB
100 Hz * 250 mG 88.47 mG 150.00 mG 4.59 dB
200 Hz 250 mG 44.04 mG 150.00 mG 10.65 dB
500 Hz . 250 mG 17.81 mG 150.00 mG 18.51 cB
1 kHz 250 mG 9.36 mG 150.00 mG 24.09 dB
2 kHz 250 mG 5.66 mG 150.00 mG 28.46 dB
5 kHz 250 mG 4.87 mG 150.00 mG 29.77 dB
10 kHz 250 mG 8.94 mG 150.00 mG 24.49 dB
Frequency Responce Setting - FLAT MODE
Date Time Temperature Meter Background
Jan-29-91 12:36PH Th? F 0.11 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
2 Hz 1m 997.27 mG 0.28 mG -71.03 4B
5 Kz 1 mG 997.27 mG 0.65 mG +63.72 cB
10 Hz 10 mG 997.27 mG 4.60 mG 746.72 dB
20 Hz 250 mo 269.75 mG 150.00 mG +5.10 dB
50 Hz 250 mG 147.69 mG 150.00 mG 0.13 dB
60 Hz 250 mG 148.81 mG 150.00 m6 0.06 dB
100 Hz . 250 mG 146.99 mG 150.00 mG 0.17 dB
200 Hz 250 mG 146.45. mG 150.00 mG 0.20 dB
500 Hz 250 mG 148.31 mG 150.00 mG 0.09 dB
1 kHz 250 mG 153.40 mG 150.00 mG 70.19 dB
2 kHz 250 mG 183.18 mG 150.00 mG 71.74 cB
5 kHz 250 mG 575.93 mG 75.00 mG 17.71 dB
10 kHz 250 mG 578.96 mG 37.00 mG +23.89 dB
FREQUENCY RESPONCE AUXILIARY DATA
Frequency Responce Setting - LINEAR MODE
Date Time Temperature Meter Background
dan: 29-91 12:35PM Th? F 0.2 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
15 Hz 250 mG 1241.08 mG 150.00 mG 718.35 cB
15 Hz 250 mG 995.43 mG 120.00 mG +18.38 cB
20 kHz 250 mG 28.76 mG 150.00 mG 14.35 dB
30 kHz 250 mG 68.48 mG 150.00 mG .81 cB
40 kHz 250 mG 138.45 mG 150.00 mG 0.69 dB
Frequency Responce Setting - FLAT MODE
Date Time Temperature Meter Background
dan-29-91 01:30PM Th? F 0.11 mG
FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE
15 Hz 250 mG 1021.06 mG 150.00 mG +16.66 cB

Appendix VIB


66

Responce

Angular Responce - Meter VIB - Flat Mode
MONITOR Model 42b-1 SN 1287

10 2 30 4050 60 70 =—80

Rotation Angle

90

++ Meter Responce
— Cosine Curve


12

Percent Error

ch
L

-10 4

AS

Accuracy and Linearity - Meter IVA
MSI Model MSI-20 SN 79011626

TT TT TTT] TT TTT 1177 T TTT TTT]

10 10? 107
Applied Field (mG)


0S

Responce (dB)

40 -

20 4

8
!

-20 4

-40 4

Frequency Responce - Meter [2
WRI Model 670 SN G1259

TIT TTY] | TT T1117]

10 10? 107 104

Frequency (Hz)

[+b0O

Square Wave
Sawtooth Wave
Sine Wave
Bezier Fit


S6

Responce

LOA

08 -

a6

a4 -

a2 -

a0

Angular Responce - Meter VB
IRC Model IRC-27

oO

MAAS A A

10

20

30

40 90

Rotation Angle

60

70

80

90

+

Meter Responce
Cosine Curve


SYDKRAFT - MODEL 88.05 - VIIA

METER CODE MANUFACTURER METER NAME MODEL # SERIAL #
VITA Sydkraft H Flux 88-05
AVD TEK 4

GAMLA SEGEVAGEN
232 91 ARLOV Sweden

METER ACCURACY AND LINEARITY AT 60H2 - TEST LOCATION LAB

Date Time Temperature Meter Background

Feb-13-91 01:52PM 76° F 0.03 mG
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
2000 mG 491.31 mG 500.00 mG : 1.77%
200 145.88 mG 150.00 mG 2.83%
200 mG 72.83 mG 75.00 mG 2.98%
200 24.38 mG 25.00 mG 2.52%
20 mG 9.76 mG 10.00 mG 2.41%

2.52% AVG % ERROR

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date Time Temperature Meter Background

Mar-21-91 11:56AM . 57° F 0 mG
RANGE SETTING ACTUAL FIELD MEASURED FIELD ‘% ERROR
200 mG 97.65 mG 100.00 mG 2.41%
97.65 mG 100.00 mG 2.41%
20 mG 9.76 mG 10.00 mG 2.41%
. 9.76 mG 10.00 mG 2.41%
2 mG 0.97 mG 1.00 mG 2.98%
0.97 mG 1.00 mG 2.98%
2 mG 0.19 mG 0.20 mG 2.98%
0.20 mG 0.20 mG 0.19%
2.41% AVG % ERROR

Appendix VIIA


ee

Accuracy and Linearity - Meter IXA

HOLADAY HI-3604 SN 76111
10 —

Percent Error

LO or a oo a,

107 1 10 10? 107
App] ied Field (mG)


WRI - Model 670 - SN G1259

MINIMUM DETECTABLE SIGNAL

Date

Mar- 14-91

Trial
Number

wane

METER RESOLUTION DATA

Date

Mar-14-91

Meter Range

Trial
Number

nuns

Time Temperature
12:53PM 60° F
Threshold Current mA Threshold Field
1.00 mA 53.95
1.00 mA 53.95
0.70 m 37.77
0.70 ma 37.77
0.70 ma 37.77
44.26
Time Temperature
12:59PM 61° F
100*1
Drive Current Start A Threshold Current
0.1380 A 0.1720
0.1280 A 0.1580
0.1310 A 0.1550
0.1310 A 0.1620
0.1340 A 0.1530

Appendix 12

max

i
&

65855

&

min
max

>

Prrr,

Meter Background

0.1 mG
0.3 mG

AVG THRESHOLD FOR SENSITIVITY

Meter Background

0.1 mG
0.3 mG
Resolution uG Resolution %

1834.27 uG 26.64%
1618.47 uG 23.44%
1294.78 uG 18.32%
1672.42 uG 23.66%
1025.03 uG 14.18%
1488.99 UG 20.85%


eel

as mazQqvuma

wal

Susceptibility - Meter VIB - Linear Mode

MONITOR Model 42b-1 SN 1287

JIE III

68 Hz
E-FIELD 16k¥ 1

4 MHz

16 qe 180 MHz

4 MHz

2 ot ao

pie? Hz

OFIELD OFF
FELD ON


rai

Accuracy and Linearity - Meter II1
SAFE PRO METER SN 1263

10 -
7 Veet

5

§ 9S

»

cc

2
-10 To]
1g? 1 10 10? 10?

Applied Field (mG)


DINDIMA - GAUSSMAUS - SN GM0400
RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Frequency Responce Setting - WIDE BAND

Date Time Temperature
May- 17-91 12:31PM 77° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
» »
1 MHZ 1.00 1.00
10 MHz 1.00 1.00
100 MKz 1.00 1.00

Frequency Responce Setting - BAND PASS

Date Time Temperature
May-17-91 12:27PM 77° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
” ”
1 MHZ 1.00 1.00
10 MHz 1.00 1,00
100 MHz 1.00 1.00
RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 H; TED

Frequency Responce Setting - WIDE BAND

Date Time Temperature
May-17-91 12:25PM T° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
#9 9
1 MHz 1.00 1.00
10 MHz 1.00 1.00
100 MHz 1,00 1.00

Frequency Responce Setting - BAND PASS

Date Time Temperature
May-17-91 12:31PM 77° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
” ”
1 MHZ 1.00 1.00
10 MHz 1.00 1.00
100 MHz 1.00 2.00

‘Appendix  XVA


19

Frequenc et: - (i (1 Square Wave
q y Responce ater VIA - Linear Mode A Saitooth Wave
MONITOR Mc. ~ 42b-1 SN 1286 + Sine Wave
40 — Bezier Fit
20
o
ra
oO -
g +
8
5 ~O0-
o
&
-40 4
760 mr tr a a a
1 10 10? 10? 10¢ 10°

Frequency (Hz)


IRC - MODEL IRC 27
METER CODE MANUFACTURER METER NAME MODEL # SERIAL #
VB Integrity Research Corp. 60 Hz H Field IRC-27

266 Elmiood Ave., ste.298
Buffalo, NY 14222

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB

Date Time .. Temperature Meter Background: mG
Feb- 12-91 12:35PM 76° F 0
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
200 mG 143.09 mG 150.00 m6 4.83%
71.55 mG 75.20 mG 5.11%
200 me 23.85 mG 25.10 m6 5.26%
9.93 mG 10.00 me 0.73%
3.98% AVG % ERROR

METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE

Date Time Temperature Meter Background
Mar-21-91 02:00PM 56° F 0 mG
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
: 96.52 mG 100.05 mG 5.85%
200 m6 96.41 mG 99.95 mG 5.87%
70.84 mG 74.95 mG 5.81%
70.89 mG 74.95 mG 5.73%
23.76 mG 24.95 me 5.11%
23.76 mG 26.95 mG 5.11%
9.71 mG 10.05. mG 3.49%
2.10 mG 2.00 mG 4.95%
4.00% AVG % ERROR

Appendix VB


oe

On MOzZOVTYUMD

6.44

il

Susceptibility - Meter VB
IRC Model IRC-27

66 Hz
E-FIELD 10kVAt

4 MHz 10 MHz «6108 MHz
UNMOD

4 MHz

10H
GOH:

—
Hz 108 MHz
iz MOD

OFIELD OFF
{FIELD ON


Il

Responce

LOA

08 =

06 -

04-5

025

00 -

10

Angular Responce ~ Meter XVIA
MEDA Model PLM-100 SN 1380

20 30 40 50 60 70

Rotation Angle

&0

90

++ Meter Responce

— Cosine Curve


HOLADAY - HI 3604 - SN 76111

METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date Time Temperature

Feb-25-91 03:15PM 78° F
E Field OFF Reading E Fietd ON Reading
EFM Metex 0.00 10.0 kV/H +-5%
E Field OFF Reading(mG) E Field ON Reading(ma)

Meter Range Meter Range
Avto Auto
1 0.17 0.17
Trial 2 0.17 0.17
Number 3 0.17 0.17
4 0.17 0.17
5 0.17 0.17
0.17 AVG 0.17

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMOOULATED

Date Time Temperature
Feb-22-91 12:44PM 79° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Auto Auto
1 MHz 0.15 0.15
10 MHz 0.15 0.15
100 MHz 0.15 0.15

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date Time Temperature
Feb-22-91 01:00PM 79° F
RF OFF Reading(mG) RF ON Reading(mG)
Meter Range Meter Range
Auto Auto
1 MHz 0.15 0.15
10 MHz 0.15 0.15
100 MHz 0.15 0.15

Appendix 1X8


EFM - Metex-3800 - SN 542513

METER 60Kz_ NON-SINUSOIDAL RESPONCE DATA

Date

May-15-91

RANGE SETTING
200 mG

RANGE SETTING

200 mG

RANGE SETTING

200 mG

Time
12:23PM

SQUARE WAVE TESTING

ACTUAL FIELD

177.87 mG
177.92 mG

SAWTOOTH WAVE TESTING

ACTUAL FIELD

145.98 mG
146.00 mG

SINE WAVE TESTING

ACTUAL FIELD

149.53 mG
149.48 mG

METER ANGULAR RESPONCE DATA

Date

May-15-91

Angle

0°
10°
20°
30°
40°
50°
60°
70°
80°
90°

Time

12:37PM

ACTUAL FIELD

149.48 mG
152.21 mG
159.49 mG
175.88 mG
204.15 mG
239.15 mG
304.30 mG
466.05 mG
947.47 mG
1108.54 mG

Temperature
77° F

MEASURED FIELD

150.05
150.00

MEASURED FIELD

150.00
150.00

MEASURED FIELD

149.95
149.95

Temperature
7 F

MEASURED FIELD

150.00
150.05
150.00
150.05
150.05
150.05
150.05
150.05
149.95

17.10

hi

"Appendix XIVA

mi
mm

BBAaR ABBAS

Meter Background
0.4 mG

‘% ERROR

715.64%
15.69%

15.67%

‘% ERROR

2.76%
2.76%

2.75%

% ERROR

0.28%
0.31%

0.29%

Meter Background

0.1 mG

METER RESPONCE

RELATIVE RESPONCE

°1.48 cB
1.48 cB

*1.48 db

RELATIVE RESPONCE

0.23 dB
0.23 dB

0.23 db

RELATIVE RESPONCE

0.02 dB
0.02 dB

0.02 db


SIT

Responce

Loo

ae

a6 A

a4 -

22-5

Angular Responce - Meter XVIIA
MAG TEC Model ELF 12 SN 2

+ Meter Responce
— Cosine Curve

40
Oo

MARA CA

10 20 30 40 50 60 70 80 90

Rotation Angle


EFM - Metex-3800 - SN 542513

METER CODE MANUFACTURER METER NAME MODEL # SERIAL #
XIVA Electric Field Measurements Metex Digital Metex-3800 542513
Multimeter
METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION LAB
Date Time Temperature Meter Background- mG
May: 15-91 11:34AM 77° F 0.1
RANGE SETTING ACTUAL FIELD MEASURED FIELD X ERROR
2000 mG 498.48 mG 500.00 mc 0.30%
200 mG 149.53 mG 149.50 mG 0.01%
0.14% AVG X ERROR
METER ACCURACY AND LINEARITY AT 60Hz - TEST LOCATION REMOTE
Date Time Temperature Meter Background
Mar-28-91 01:49PM 7h? F 70.05 mG
RANGE SETTING ACTUAL FIELD MEASURED FIELD % ERROR
99.59 mG 100.00 m6 0.41%
200 mG 99.59 mG 100.05 mG 0.46%
49.85 mG 50.05 mG 0.40%
49.74 mG 49.95 mG 0.41%
9.98 mG 10.00 mG 0.19%
1.03 mG 1.00 mG $2.64%
-0.09% AVG % ERROR

Appendix  XIVA


EMD - MODEL ACGM-1 - SN 01-0116

FREQUENCY RESPONCE PRIMARY DATA

Date

Feb-07-91

FREQUENCY

1 az

2 Hz

5 Hz
10 Hz
20 Hz
50 Hz
60 Hz
100 Hz
200 Hz
500 Hz

1 kHz
2 kHz

Time

11:42AM

RANGE SETTING

FULL mG

FREQUENCY RESPONCE AUXILIARY DATA

Date

Feb-08-91

FREQUENCY

80 Hz
300 Hz
150 Hz
35 Hz

Time

12:20PK

RANGE SETTING

FULL mG
FULL mG
FULL mG
FULL mG

Temperature
76° F

ACTUAL FIELD

9.71 mG

Temperature
76° F

ACTUAL FIELD

260.46 mG
994.11 mG
324.22 mG
317.41 mG

Appendix VIIIA

Meter Background

0

mG

MEASURED FIELD

0.30
2.60
27.40
150.10
248.50
250.60
250.00
250.40
250.60
65.80
8.50
0.80

Meter Background

0

3322832282328

mG

MEASURED FIELD

250.30
250.30
250.00
250.30

mTM
mms
mG
mG

RELATIVE

RELATIVE

RESPONCE

+70.31 dB
51.56 GB

RESPONCE

70.34 gB
711.98 cB
72.26 dB
72.06 GB


MEDA - Model PLM-100 - SN 1380

METER 60Hz E FIELD SUSCEPTIBILITY DATA

Date

May-20-91

EFM Metex

Trial
Number

wane

Time

09:35AM

E Field OFF Reading

0.00

E Field OFF Reading(mG)
Meter Range

20

AVG

Temperature
74° F

E Field ON Reading

10.00 KV/M +-5%

£ Field ON Reading(mG)
Meter Range
20
0.26
0.09
0.22

0.17
0.17

0.18

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - UNMODULATED

Date

May-17-91

1 MHz
10 MHz
100 MHz

Time
02:25PM
RF OFF Reading(mG)
Meter Range
20
0.
0.
0.

SSs

Temperature
7 F
RF ON Reading(mG)
Meter Range
20
0.09

0.09
0.37

RADIO FREQUENCY SUSCEPTIBILITY MEASUREMENT DATA - 60 Hz MODULATED

Date

May-17-91

1 MHZ
10 MHz
100 MHz

Time
02:24PM
RF OFF Reading(mG)
Meter Range
20

Temperature
7 F
RF ON Reading(mG)
Meter Range
20

222
888

Appendix XVIA


Sampling Procedures

The devices tested were limited to coil type single axis meters having
either analog or digital readout. Excluded from this test were computer
driven devices, multi-axis devices, and devices utilizing non-coil- type
sensing probes. All manufacturers known to the EPA-at the time of the test
were solicited to submit their meters for testing. Each manufacturer was
requested to submit two items of each’ type. In addition several instruments
were purchased from manufacturers. 7

Sample Custody

The vendor samples (2 each type) were received at the EPA testing
facility in Las Vegas NV. Upon receipt all samples were masked by a check-in
technician not directly connected with the testing procedure. Masking the
devices involved covering of’ manufacturer, model number, and serial number
designations. A-test item code was ‘recorded‘on each test object and used
throughout testing to identify each test object. A log was maintained by
the check-in technician which -associates each test object code with the
manufacturer, model number ‘and device serial ‘number.

The coding scheme used a Roman numeral assigned to each test meter
type (manufacturer and model) in the order of their arrival at the EPA test
facility. Following the test object numeral was a letter A or B for the two
items of the same type (each manufacturer was requested to deliver two of
each meter type for testing) .

A list of the meters included in this document and their corresponding
meter codes is provided in Table 1 on the next page. These codes have been
Placed on all the meter performance charts and graphs throughout this
document.


£6

Responce

Angular Responce - Meter IVB
MSI Model MSI-20 SN 79010854

-++ Meter Responce

La - — Cosine Curve

ae
aE A
a4
ae

ao0-
0 0 2 0 0 5% 6 0 8 9%

Rotation Angle


8€

Accuracy and Linearity - Meter XIVB
EFM - Metex-3800 - SN 542832

10
54
5
o .
£ oO-
2
: 4
eo.
-10 Tr oy
1 . 10 10? 107

Applied Field (mG)


WRI - Model 670 - SN G1257

FREQUENCY RESPONCE PRIMARY DATA

Date Time Temperature Meter Background
Feb-08-91 08:40AM 74° F 0.5 mG

FREQUENCY RANGE SETTING ACTUAL FIELD MEASURED FIELD RELATIVE RESPONCE

1 Hz 100*1 mG 868.13 mG 4.00 mG -58.77 dB

2 Hz 100*1 m6 868.13 mG 2.00 mG -52.75 dB

5 Hz 400*1 mG 868.13 mG 21.00 mG -32.33 4B

10 Hz 100*1 mG 868.13 mG 100.00 mG “18.77 cB

20 Hz 100*10 mG 679.36 mG 200.00 m6 +10.62 dB

50 Hz 100*10 me 211.08 me 200.00 mG +0.46 dB

60 kz 100*10 mG 176.92 mG 200.00 mG 1.06 dB

100 Hz 100*10 mG 104.82 mG 200.00 mG 5.61 dB

200 Hz 100*10 mG 58.45 mG 200.00 mG 10.69 dB

500 Hz 100*10 mG 35.05 mG 200.00 mG 15.13 dB

1 kHz 100*10 mG 33.48 mG 200.00 mG 15.52 aB

2 kHz 100*10 mG 36.02 mG 200.00 mG 14.89 dB

5 kHz 100*10 mG 71.46 mG 200.00 mG 8.9% dB

10 kHz 100*10 mG 139.82 mG 200.00 m6 3.11 aB

FREQUENCY RESPONCE AUXILIARY DATA

NO AUX TESTS

Appendix 11


S2l

OB MnzZOvUmMD

ll

Susceptibility - Meter IIIA
EXPAN Model 1A SN 1103

a nh

6@ Hz
E-FIELD 16k¥TM

1 MHz 18 MHz «6108 MHz
‘UNTO

10 MHz $00 MHz
GeHz MOD

COFIELD OFF
(FELD ON


EFM - Metex-3800 - SN 542513

MINIMUM DETECTABLE SIGNAL

Date Time

Mar-28-91 01:57PM

Threshold Current m

1 3.10

Trial 2 2.90
Number 3 3.30
4 3.40

5 3.40

NOTE: Meter reads about 0.1mG higher

METER RESOLUTION DATA

Date Time
Mar-28-91 02:02PM
Meter Range Auto Range - 100mG
Drive Current Start
1 1.8460
Trial 2 1.8460
Number 3 1.8460
4 1.8460
5 1.8560

NOTE: Meter readings increase by 0.1mG

A

>> >>>

Temperature
74° F

Threshold Field

167.25
156.46
178.04
183.43
183.43

173.72

Temperature
70° F

Threshold Current

1.8540
1.8490
1.8510
1.8550
1.8650

Appendix XIVA

min
max

uG

uG
uG
uG
uG
uG

uG

min
max

> Pree

Meter Background

70.1
0

ms
mG

AVG THRESHOLD FOR SENSITIVITY

Meter Background

“0.1
0

Resolution

431.59
161.85
269.75
485.54
485.54

366.85

ms
ms

65565 &

6

Resolution %

0.43%
0.16%
0.27%
0.48%
0.48%

0.36%


00

Responce

105

a8 4

06 -

a4

a2 -

a0-

10

Angular Responce - Meter VIIA

20

SYDKRAFT

30

Model 88-05

40 50
Rotation Angle

60

70

80

90

+

Meter Responce

Cosine Curve


07

Percent Error

Accuracy and Linearity - Meter XVB - Wide Band
DINDIMA - GAUSSMAUS - SN GMO401

po |

1317 T1177 TTT 1177

10 10? 10?
Applied Field (mG)