NIST Technical Note 1668

A Green Laser Pointer Hazard

Jemellie Galang

Alessandro Restelli

Edward W. Hagley

Charles W. Clark
NIST Technical Note 1668

A Green Laser Pointer Hazard

Jemellie Galang

Alessandro Restelli

Edward W. Hagley

Charles W. Clark

Electron and Optical Physics Division

Physics Laboratory

July, 2010

U.S. Department of Commerce 

Gary Locke, Secretary 

National Institute of Standards and Technology 

Patrick D. Gallagher, Director 
Certain commercial entities, equipment, or materials may be identified in this

document in order to describe an experimental procedure or concept adequately.

Such identification is not intended to imply recommendation or endorsement by the

National Institute of Standards and Technology, nor is it intended to imply that the

entities, materials, or equipment are necessarily the best available for the purpose. 

National Institute of Standards and Technology Technical NOTE 1668

Natl. Inst. Stand. Technol. Tech. Note 1668, 9 pages (July, 2010) 

CODEN: NTNOEF
1

A Green Laser Pointer Hazard

Jemellie Galang, Alessandro Restelli, Edward W. Hagley* and Charles W. Clark

Joint Quantum Institute, National Institute of Standards and Technology and University of 

Maryland, Gaithersburg, MD 20899-8410

*also Acadia Optronics, Rockville, MD 20850

Abstract: An inexpensive green laser pointer was found to emit 20 mW of infrared radiation 

during normal use. This is potentially a serious hazard that would not be noticed by most users 

of such pointers. We find that this infrared emission derives from the design of the pointer, and 

describe a simple method of testing for infrared emissions using common household items. 

1. Introduction

In December, 2009, we made a retail purchase of three inexpensive green laser pointers for

US$15 each. These devices were advertised to have output power of 10 mW and had common 

packaging. One of these lasers had a much weaker green light intensity compared to the others, 

as judged by normal eyesight. We measured the output power of that laser and found that it 

emitted more than ten times more invisible infrared light than visible green light.

This is a serious hazard, since humans or animals may incur significant eye damage by exposure 

to invisible light before they become aware of it. 

We have found that this problem is common in low-cost green laser pointers, though its 

seriousness varies widely. The hazards would not be immediately apparent to ordinary users. In 

this paper, we describe the origin of this problem, and we suggest a simple method for 

diagnosing it that can be used by those who have access to a digital or cell phone camera and an 

inexpensive web camera.

2. Principles of inexpensive green laser pointer operation

The common green laser pointer is a technological marvel that has recently become a commodity 

item. Twenty years ago, 10 mW of coherent green light could be obtained only from research 

laboratory-grade lasers that cost over $100,000 and occupied much of an experimental laser 

table. Today's green laser pointer delivers much the same performance for less than $20, in a 

package the size of a slim cigar, powered by two AAA batteries.

The atomic energy level diagram of the green laser pointer is shown in Fig. 1. Hereafter, we use 

the acronym GLP (Green Laser Pointer) to refer only to devices employing the diode-pumped, 

frequency-doubling architecture shown there. Figure 1 exhibits three essential elements of 

operation, each of which was at its time a highlight of laser technology:

a semiconductor pump diode laser operating at an infrared wavelength of 808 nm

a neodymium ion oscillator at a wavelength of 1064 nm

a frequency-doubling crystal that generates light of half of that wavelength, the familiar 

GLP color at 532 nm
2

Today, all three of these functions are integrated in an inexpensive package that is produced in 

high-volume manufacturing. A schematic of a typical GLP is shown in Fig. 2. 

Figure 1. Atomic energy levels and light involved in the operation of a GLP. Triply-charged ions of the 

neodymium atoms, Nd3+, are present as dopants in a crystal of yttrium orthovanadate (Nd:YVO4). The Nd3+ ion 

contains three 4f electrons outside closed electron shells, so its electronic configuration is thus designated 4f 3. The 

horizontal lines indicate some of the energy levels of the 4f 3configuration, labeled by conventional spectroscopic 

notation. A semiconductor diode laser with an infrared wavelength of 808 nm excites the lowest 4I state to an 

electronically excited 4F state. The Nd3+ ion emits infrared radiation, at a wavelength of 1064 nm, as it drops from 

the excited state into a different 4I state. This radiation is directed into a “frequency doubling” crystal of potassium 

titanyl phosphate, which emits light of half the wavelength: 532 nm. Figure credit: Dr. Joseph Reader, National 

Institute of Standards and Technology.

Figure 2. Schematic of the operation of a GLP 

based on a Multiple Crystal Assembly (MCA). 

The familiar external package contains two AAA 

batteries that power the unit, a printed circuit 

board with the pump Laser Diode (LD) driving 

circuitry and the Diode Pumped Solid State Laser 

(DPSS) Module. The 808 nm pump LD is 

optically coupled to the Nd:YVO4 conversion 

crystal (violet section of the MCA), which emits 

1064 nm light into the KTP frequency-doubling 

crystal (light blue section). The 532 nm light 

from the KTP crystal is sent through an 

expanding and collimating lens assembly to 

produce a collimated output beam. In this 

configuration, an IR Filter prevents the 808 nm 

and 1064 nm light from exiting the laser. In the 

GLP discussed in this paper, no such infrared 

filter was present. HR: High Reflectivity; HT: 

High Transmissivity; AR: Anti-Reflective; OC: 

Output Coupler; IR: InfraRed. This figure is the 

copyrighted work of Samuel M. Goldwasser [1], 

and is reprinted here with his permission.
3

3. Standard hazards of GLP operation

A 10 mW GLP is rated as a Class IIIb device [2], which is capable of causing serious eye 

damage. However, there is little documented data on eye injuries actually caused by exposure to 

GLPs. The light from a GLP is at a wavelength to which the human eye is most sensitive (Fig. 

3). This is why a GLP appears far brighter than a red laser pointer that emits the same optical 

power. Thus, GLPs are superior to red laser pointers for demonstration purposes. Direct 

exposure of the eye to GLPs seems to be relatively rare, because of the blink reflex that protects 

the eye from light of strong visual intensity.

Figure 3. Diagram of human visual response showing the wavelength of the GLP line at 532 nm, and the 650 nm

wavelength of a typical red laser pointer. The blue curve shows the perceived brightness of light from a source of 

uniform intensity across the wavelength range. Thus a 5 mW GLP would appear as bright as a 41 mW red laser.

Data from the 1931 report of the International Commission on Illumination (CIE), as presented in Ref. [3]

4. Less-appreciated hazards of the GLP

The green light of a GLP is produced by much more intense sources of invisible infrared light: 

the diode pump laser and the neodymium oscillator. In some green laser pointer designs, this 

invisible radiation is blocked from the output beam by an infrared-blocking filter, as shown in 

Fig. 2. In the case that we report here, the infrared radiation was not blocked, due to the absence 

of this filter. Intense infrared radiation poses a grave danger to humans and animals because, 

being invisible, it does not activate the blink reflex. Thus, a considerable dose of infrared 

radiation can enter the eye and cause retinal damage, imperceptible to the victim before serious 

damage occurs. Invisible lasers, in the infrared and ultraviolet, are among the most frequently 

reported sources of laser-induced eye injuries, with many reported instances of permanent 

damage to vision [4]. A survey of the 100 accidental, non-medical laser-induced eye injuries 

that had been reported in the scientific literature up to 1999 found that Nd-based lasers with 

wavelengths of 1060-1064 nm were implicated in 49 % of all cases, while 532 nm lasers 

accounted for only 7 %. [5]
4

The main danger associated with GLPs occurs in cases of low conversion efficiency from 

infrared to visible radiation. This can result from the manufacturing process. Normally, if the 

conversion efficiency is high, the inter-cavity infrared power will be low because conversion to 

green light draws down the power of the infrared. However if the conversion efficiency is low, 

then the inter-cavity infrared power can build up to high levels, resulting in strong infrared

emission. In the extreme case of zero conversion efficiency, it would be possible for the GLP to 

emit intense infrared light but no visible green light. Inclusion of an infrared-blocking filter in 

GLP design can prevent such infrared emission. As shown in Fig. 4, for the laser that we discuss

here, about 10 times more light was emitted in the infrared than in the green. This was due to 

both low infrared/green conversion efficiency and the absence of an infrared-blocking filter. The 

total emitted power as measured with a thermal detector was 20 mW. It is dangerous for such 

intensities of infrared laser light to be present in an uncontrolled environment. For example, 

many windows in modern commercial and industrial buildings have infrared reflective coatings 

to reduce solar heating of the interior. If a green laser is directed on such a window, the green 

light will pass through it, but any infrared emitted by the GLP will be reflected. This is a 

substantial hazard. Any differential reflection of infrared vs. green light poses such hazards, 

since the green light that travels with the infrared light may not be sufficiently intense to activate 

the blink reflex.

Figure 4. Power spectrum of the laser under test, showing the ratios of the measured power of each laser line to the 

measured power of the green line at 532 nm.

5. A simple test for green laser pointers

Inexpensive green laser pointers are produced in vast quantities. Few that we have seen carry 

traceable manufacturers' marks that are visible to the consumer. Widespread production of this 

commodity item is likely accompanied by inconsistent adherence to practices of safe design 
5

and/or quality control. The drive to lower costs on such commodity items may encourage

manufacturers to eliminate the infrared-blocking filter, resulting in the type of hazard described 

here, especially when the conversion efficiency from infrared to green is low. The consequence 

has been the apparent release into the U.S. retail market of GLPs that emit potentially hazardous

and unappreciated levels of infrared radiation. 

We now describe a simple test that can be used to detect infrared leakage which can be 

performed by anyone with access to a common digital or cell phone camera, a compact disc 

(CD), a web camera and a TV remote control. This test is an elementary exercise in laser 

spectroscopy, which uses a diffraction grating to separate light of different wavelengths onto a 

viewing screen [6]. Despite the simple nature of the test, we cannot over-emphasize the 

importance of following proper laser safety procedures in carrying it out [4].

A simple layout of this experiment is shown in Fig. 5. The CD serves as the diffraction grating, 

which deflects light of different wavelengths in different directions. The digital or cell phone 

camera is sensitive only to visible light, and is used to photograph the green diffraction pattern, 

as shown in Fig. 6. On the other hand, inexpensive web cameras are, or may easily be modified 

to be, sensitive to both visible and infrared radiation. An infrared TV remote control unit may be

used to determine whether the web camera is infrared sensitive; if it is not, it can be modified by 

the procedure described in Appendix B. The infrared-sensitive web camera is then used to 

photograph the diffraction pattern. Comparison of images acquired by digital and web cameras 

determines the presence of infrared light, as shown in Fig. 6 below. 

CAUTION

DIRECT EXPOSURE ON THE EYE BY A BEAM OF LASER LIGHT SHOULD ALWAYS BE AVOIDED WITH 

ANY LASER, NO MATTER HOW LOW THE POWER

NEVER LOOK INTO A DIRECT, REFLECTED OR DIFFRACTED LASER SOURCE

TEST PROCEDURE MAY PRODUCE VISIBLE AND INVISIBLE LASER RADIATION 

KEEP ALL LASER BEAMS AT TABLE LEVEL; KEEP EYES WELL ABOVE TABLE LEVEL

WEAR SAFETY GLASSES APPROVED FOR LASER AND INFRARED RADIATION WHEN CONDUCTING 

TEST PROCEDURE

CONDUCT TEST PROCEDURE ONLY UNDER PROPER SUPERVISION

IN A PROPERLY CONTROLLED ENVIRONMENT
6

6. Details of test procedure

We now describe an actual implementation of this experiment in more detail.

Figure 5. Experimental setup for determining whether infrared radiation is emitted by a GLP. The GLP light 

passes through a hole in the paper screen and is diffracted by the black-masked CD on the left. Five green 

diffraction spots are visible on the paper screen.

Ordinary disposable drink cups (or stacks of books) are used as optical mounts for the GLP, the 

CD, and the cameras. 

Pressing a knife or scissors blade across the diameter of a cup cuts slots that provide an 

adjustable vertical mount for the CD. 

Another cup provides a level mount for the laser, whose power button can be held on by 

duct tape, a binder clip, or a cable tie. 

A small transmission hole is made in a piece of ordinary ink-jet or copier paper, which is 

positioned between the laser and the CD. The transmission hole allows the laser light to 

pass through the paper and hit the CD. Black tape is used to cover most of the CD 

surface, leaving a small bare patch exposed to the laser spot. This suppresses distracting 

reflections of the light from other parts of the CD surface. 

The light reflected from the bare patch on the CD returns to the paper, which forms a 

back-lit screen that can be viewed by the eye or photographed from behind the laser. The 

pattern of visible light in this configuration is shown in the top frame of Fig. 6. The 

center spot is the direct reflection of the laser by the CD. The spots on either side are the 

diffraction orders described in Appendix A below.

After viewing and understanding this pattern, and photographing it with a digital camera, 

we place an infrared-sensitive web camera so as to view the same scene. Such infraredenabled web cameras will show the presence of infrared light, as shown in the bottom 

frame of Fig. 6. 
7

7. Origin of the problem

After studying the visible and infrared light emitted by this GLP, we disassembled it. No 

infrared-blocking filter of the type shown in the schematic of Fig. 2 was found in this particular 

device. Inspection of the assembly revealed that the manufacturer had not incorporated a holder

for such a filter. We thus believe that the absence of the filter in this case was due to a design 

decision.

8. Summary

We describe a green laser pointer purchased in ordinary commerce which could potentially 

expose the eye to dangerous levels of infrared radiation. We devised a procedure by which 

infrared radiation from a green laser pointer may be detected with common household apparatus.

Appendix A. Image details

Figure 6. Top frame: ordinary camera photograph of diffraction of green laser light by the masked CD, from the 

vantage point of the web camera shown in Fig. 5. The GLP is visible in the foreground. The yellow lines are aligned 

with the green diffraction spots corresponding to zero-, first- and second-order diffraction. Bottom frame: the same 

vignette, photographed by a web camera with no infrared-blocking filter. Here we see intense diffraction spots that 

are not visible to the eye, between the first- and second-order diffraction spots of the green light. These are due to 

808 nm infrared light. Note that the infrared spots exhibit greater divergence than the green. The infrared light 

spreads out beyond the green, which could be injurious, for example, to a cat closely chasing a spot of green light.

The CD acts as a diffraction grating with a grating spacing defined by the CD tracks that are 

separated by d =1.6 μm. Knowing the grating separation d, it is possible to calculate the angle of 

diffraction, , as a function of wavelength λ. For normal incidence it is possible to use the simple 

equation [7] 

sin = mλ/d
8

where m = 0, ±1, ±2, etc. is an integer. The formula indicates that each wavelength will be 

scattered back in a discrete series of angles that match the integral values of m. The case m = 0 

corresponds to simple back reflection.

The two frames of Fig. 6 show images taken with digital (visible) and web (visible + infrared) 

cameras. The top frame was acquired with an ordinary digital camera, which has a high-quality 

infrared-blocking filter. It corresponds closely to what would be seen by the naked eye. Only the 

green light is seen, distributed left to right among diffraction orders corresponding to m = -2, -1, 

0, 1 and 2, as labeled.

The bottom frame was acquired by an inexpensive web camera, with its infrared-blocking filter 

removed, as described in Appendix B. Here we see the 808 nm infrared light, with a central 

(m = 0) spot much more intense than the green, and with bright m = ±1 spots outside the fainter 

m = ±1 spots of the green, as predicted by the diffraction equation above.

The webcam sensitivity cuts off near wavelengths of 1064 nm, so we cannot see the diffraction 

spots (m = ±1) corresponding to this wavelength. If detectable, such spots would overlap the 

m = ± 2 spots of the 532 nm green light. 

Appendix B. Using an inexpensive web camera as an infrared detector

As mentioned in Section 6 above, not all web cameras are sensitive to infrared light. Web 

cameras that are sensitive to infrared are usually labeled as night vision cameras. Not capable of 

imaging in complete darkness, such cameras use an infrared light-emitting diode to illuminate 

the subject. It is possible to modify most other web cameras to extend their sensitivity into the 

infrared range. There is a vast internet-based literature on such modifications, which can easily 

be found by submitting “IR webcam” to a search engine. For completeness, we describe the 

actual procedure that we used ourselves to modify such a camera. However, we do not claim it 

to be optimal or even recommended. It is similar to the procedures that are described online.

After disconnecting the web camera from all data communication and power lines, we 

disassemble the housing by unfastening screws. Figure 7 shows the interior components of the 

camera. The infrared-blocking filter is mounted close to the camera’s sensor, and while 

appearing transparent it reflects red light if observed laterally. 
9

Figure 7. Image of a web camera that was disassembled in order to remove the infrared-blocking filter that is

usually located in proximity to the camera’s sensor. In this particular model the filter is glued inside the lens holder. 

The holder is connected to the main circuit board using two screws through mounting holes indicated by the dashed 

lines. Lens and holder are threaded for both mechanical assembly and adjustment of focus.

Since the procedure of filter removal involves access to electrical and mechanical components, 

we recommend this operation only to those familiar with delicate disassembly and assembly 

work. Power and data cables must be disconnected during the modification, and the camera 

package should be fully reassembled before subsequent operation. This procedure is likely to 

void all manufacturers’ warranties.

References

[1] http://www.repairfaq.org/F_email.html .

[2] Code of Federal Regulations, Title 21, Volume 8, Chapter I, Subchapter J, Part 1040, Sec. 

1040.10: Laser products (revised as of April 1, 2009).

[3] “Machine vision toolbox,” P. Corke, IEEE Robotics and Automation Magazine 12(4), 16

(2005).

[4] Laser Safety Management, Kenneth Barat (CRC Press, Boca Raton, FL, 2006), sec. 9.20; 

“Ocular Radiation Hazards,” David H. Sliney, in Handbook of Optics, Third Edition, Vol. III: 

Vision and Vision Optics, ed. Michael Bass, et al. (McGraw Hill, New York, 2010).

[5] “Laser injuries of the eye,” A. B. Thatch, International Ophthalmology Clinics 39(2), 13

(1999)

[6] “The compact disc as a diffraction grating”, H. Kruglyak, Physics Education 26(4), 255 

(1991); “Teaching optics in North Uganda”, L. Rossi, et al. Optics & Photonics News 21(6), 

14 (2010)

[7] Diffraction Grating Handbook, Sixth Edition, Christopher Palmer (Newport Corporation, 

Rochester, New York, 2005)