# 

Skip to Main Page ContentSkip to SearchSkip to Site Map NavigationSkip to Footer Links

        - Home

        - FAQ
        - Glossary
        - Facility Locator
        - What's New
        - Site Help
        - Index A-Z
        - Contact Us
        - Browse Aloud
        - Email Updates

Report a Safety Concern

      - Nuclear Reactors  - Power Reactors
- Research & Test Reactors
- Operating Reactors
- Operator Licensing
- New Reactors
- Advanced Reactors
- Operator Licensing for New Reactors
- Nuclear Reactor Quick Links
 
      - Nuclear Materials  - Special Nuclear Material
- Source Material
- Byproduct Material
- Med, Ind, & Academic Uses
- Source Materials Facilities
- Uranium Recovery
- Fuel Cycle Facilities
- Materials Transportation
- Nuclear Materials Quick Links
 
      - Radioactive Waste  - Low-Level Waste
- Waste Incidental to Reprocessing
- High-Level Waste
- Uranium Mill Tailings
- Low-Level Waste Disposal
- High-Level Waste Disposal
- Storage of Spent Nuclear Fuel
- Transporation of Spent Nuclear Fuel
- Radioactive Waste Quick Links
 
      - Nuclear Security  - Domestic Safeguards
- Information Security
- Radioactive Material Security
- Contact Us
 
      - Public Meetings & Involvement  - The NRC Approach to Open Government
- About Meetings Open to the Public
- Conferences & Symposia
- Documents for Comment
- Facilitating Stakeholder Involvement
- NRC Information Quality Guidelines
- Subscribe to E-mail Updates
- Commission Schedule
- Public Meeting Schedule
 
      - NRC Library  - Basic References
- Document Collections
- ADAMS Public Documents
- Public Document Room
- Get Copies of Documents
- FOIA & Privacy Act Requests
- Photos & Video
- Records Management 
- Withholding of Sensitive Information
- FAQ Index
- Adjudications (Hearings)
 
      - About NRC  - The Commission
- Governing Legislation
- Plans, Budget, & Performance
- Organization & Functions
- Locations
- History
- Values
- Direction-Setting & Policymaking
- Radiation Protection
- Fire Protection
- How We Regulate
- Emergency Preparedness & Response
- Public Affairs
- Congressional Affairs
- International Programs
- State & Tribal Programs
- Alternative Dispute Resolution Programs
- Employment Opportunities
- Contracting Opportunities
- Grant Opportunities
- Civil Rights
 

Print

HomeNRC LibraryDocument CollectionsGeneral CommunicationsInformation Notices1993

# 

```



                                 UNITED STATES
                         NUCLEAR REGULATORY COMMISSION
                     OFFICE OF NUCLEAR REACTOR REGULATION
                            WASHINGTON, D.C.  20555

                                March 31, 1993


Information Notice No. 93-23:  WESCHLER INSTRUMENTS MODEL 252
                                   SWITCHBOARD METERS


Addressees

All holders of operating licenses or construction permits for nuclear power
reactors.

Purpose

The U.S. Nuclear Regulatory Commission (NRC) is issuing this information
notice to alert addressees to possible problems with Weschler Instruments
(Weschler) Model 252 commercial grade switchboard meters.  It is expected that
recipients will review the information for applicability to their facilities
and consider actions, as appropriate, to avoid similar problems.  However,
suggestions contained in this information notice are not NRC requirements;
therefore, no specific action or written response is required.  

Description of Circumstances

On November 9, 1992, Weschler informed the NRC of sticking meter movements in
some Model 252 meters manufactured and supplied as commercial grade items. 
Although Weschler reported having no evidence that meters manufactured
specifically for safety-related service are affected, Weschler was concerned
that affected commercial grade meters could have been dedicated for use in
nuclear safety-related applications.

Discussion

Weschler determined that some of the commercial grade meters returned for
repair or calibration were inaccurate and erratic.  The vendor engineering
report attributed this problem to sticking of the meter movement because of an
off-center torsion spring.  Weschler determined that this condition resulted
from a deviation in the dimensions associated with the torsion head screws
combined with several coincidental factors that included (1) degradation of
clip-to-torsion screw lubrication, (2) low initial force from the spring-
retaining clip or fatigue of the spring-retaining clip, (3) variations in the
alignment of the meter movement during initial assembly, and (4) the effect of
the summation of the dimensional tolerances in the assembly of the
subcomponents.





9303240274.

                                                            IN 93-23
                                                            March 31, 1993
                                                            Page 2 of 2


To prevent this deficiency in future products, Weschler revised its production
inspection procedures to specifically require checking the dimensions in
question.  Weschler also revised its nuclear quality assurance procedures to
require specific checks of torsion head screw dimensions, although the vendor
has not detected a problem with meters manufactured specifically for safety-
related applications.

Weschler could not determine whether or not any Model 252 switchboard meters
sold as commercial grade items through the various distributors had been
dedicated for use in nuclear safety-related applications.  Therefore, Weschler
has not directly notified any licensees about this problem.   

Weschler stated that the principal corrective action it has taken for affected
meters returned for repair and recalibration is to replace the faulty torsion
head screws because they can't be repaired.  Upon request, Weschler will give
licensees information about the problem and the recommended corrective
measures.

This information notice requires no specific action or written response.  If
you have any questions about the information in this notice, please contact
the technical contact listed below or the appropriate Office of Nuclear
Reactor Regulation (NRR) project manager.


                                      ORIGINAL SIGNED BY

                                   Brian K. Grimes, Director
                                   Division of Operating Reactor Support
                                   Office of Nuclear Reactor Regulation

Technical contact:  S. V. Athavale, NRR
                    (301) 504-2974

Attachment:  
List of Recently Issued NRC Information Notices
..
```

*Page Last Reviewed/Updated Wednesday, February 16, 2011*

#### Home

            - News Releases
            - Event Reports
            - ADAMS
            - Open Gov
            - Students & Teachers
            - Photos & Video

#### About Us

            - Strategic Plan
            - Budget & Performance
            - Perf & Accountability Rept
            - History of the NRC
            - Employment
            - NRC Ethics
            - Agency Status
            - Contact Us

#### Popular Documents

            - Info Digest
            - Factsheets & Brochures
            - Forms
            - Electronic Submittals
            - NRC Reports – NUREG
            - NRC Regulations – 10-CFR
            - Inspection Reports
            - Plain Writing
            - Enforcement Actions

#### Stay Connected

            - Blog
            - Flickr
            - Twitter
            - YouTube
            - RSS

Regulations.govUSA.govRecoveryFOIANo Fear EEOInspector GeneralSite MapAccessibilityPrivacy PolicySite DisclaimerFor Employees