AppendixB:SummaryofPMTestReportsReviewedforConsistencyPass/Fail FacilityIDUnitID FuelCategory CombustorDesign TestDate Run#Methodused(26or26A)Sampletime(min.)PercentH2O(Bws) O2MassHCl(μg)Notes ActionsTakeninEPAICRDatabasesPASS CTElectricBoat EMU17 LightLiquid N/A 8/17/2009 1 Method26A 60 9.7 4.63 7.3E+02Didnothavebasicsolutionimpingers.Calculatedcorrectedppm,notinreport.PASS CTElectricBoat EMU17 LightLiquid N/A 8/17/2009 2 Method26A 60 9.7 4.67 4.0E+02Didnothavebasicsolutionimpingers.Calculatedcorrectedppm,notinreport.PASS CTElectricBoat EMU17 LightLiquid N/A 8/17/2009 3 Method26A 60 11 4.68 6.7E+02Didnothavebasicsolutionimpingers.Calculatedcorrectedppm,notinreport.FAIL FLSugarCaneGrowersCoop BoilerNo.8 Biomass Suspension/Grate 1/19/2005 1 Method26A 60 26.2 7.93 2.6E+03 Stainlesssteelnozzlesinsteadofglass?FAIL FLSugarCaneGrowersCoop BoilerNo.8 Biomass Suspension/Grate 1/19/2005 2 Method26A 60 28 7.38 2.7E+03FAIL FLSugarCaneGrowersCoop BoilerNo.8 Biomass Suspension/Grate 1/19/2005 3 Method26A 60 27.1 7.32 2.0E+03PASS FLSugarCaneGrowersCoop BoilerNo.8 Biomass Suspension/Grate 12/16/2002 1 Method26A 60 25.49 8.6 3.0E+02 MetalsandHCltrainscombined???ppmnotincluded,calculatedPASS FLSugarCaneGrowersCoop BoilerNo.8 Biomass Suspension/Grate 12/16/2002 2 Method26A 60 25.63 7.85 2.8E+02PASS FLSugarCaneGrowersCoop BoilerNo.8 Biomass Suspension/Grate 12/16/2002 3 Method26A 60 24.66 8.85 1.9E+02 Leakcheckperformedat14"whenhighestvacuumwas15"FAIL FLUSSugarCorp BoilerNo.7 Biomass Suspension/Grate 12/31/2002 H‐1 Method26A 60 22.55 8.48 9.6E+02MetalsandHCltrainscombined???DGMcalibration‐onereadingisoutoftoleranceandpost‐testnotincluded;ppmnotincluded,calculatedFAIL FLUSSugarCorp BoilerNo.7 Biomass Suspension/Grate 12/31/2002 H‐2 Method26A 60 19.14 ? 7.2E+02Noteonruns2and3:O2andCO2valuesficticiousforuseof30asmolecularwt.Notsurewhatthismeans.FAIL FLUSSugarCorp BoilerNo.7 Biomass Suspension/Grate 12/31/2002 H‐3 Method26A 60 23.33 ? 3.6E+02Filtertemperaturewaslowforfirst10minutesofthetest(220insteadof223)FAIL IDMoyieSpringsLumber420 HFB1 DryBiomass Stoker/SlopedGrate/Other 9/12/2003 1 26A 72 15.06 Notgiven 6.9E+02FAIL IDMoyieSpringsLumber420 HFB1 DryBiomass Stoker/SlopedGrate/Other 9/12/2003 2 26A 72 21.04 Notgiven 1.4E+02 Blankvalueshigherthanrun2result.FAIL IDMoyieSpringsLumber420 HFB1 DryBiomass Stoker/SlopedGrate/Other 9/12/2003 3 26A 72 14.53 Notgiven 2.4E+02 Blankvalueshigherthanrun3result.PASS INAlcoaWarrick Unit#3 Coal PC 8/17/2009 1 26A 120 17.6 5.4 1.0E+01Resultsareallnon‐detect,howeverIcannotfindANYcalculatedresultinanyofthefilesinthefolder.Iambasingmyreviewonfieldsheetsandlabdata.PASS INAlcoaWarrick Unit#3 Coal PC 8/17/2009 2 26A 120 17.6 5.49 1.0E+01PASS INAlcoaWarrick Unit#3 Coal PC 8/17/2009 3 26A 120 17.3 5.55 1.0E+01ReporteddetectionlimitabouthalfofthemethodpublishedDL.Surprisinlylow.PASS MEFPLEnergyWyman Unit#5 HeavyLiquid N/A 10/5/2009 1 26A 2.4E+03 NOTESTMETHODDOCUMENTATIONDOCUMENTATIONPASS MEFPLEnergyWyman Unit#5 HeavyLiquid N/A 10/5/2009 2 26A 1.7E+03PASS MEFPLEnergyWyman Unit#5 HeavyLiquid N/A 10/5/2009 3 26A 2.0E+03FAIL MIConsumerEnergyCo‐Campbell EUAUXBLR12 LightLiquidN/A9/1/2009 1 320 153Therearenoemissionscontrolsforthisboilerasitisanauxiliaryboilertokeepsteamlinesheated.PpmvaluesarewetFAIL MIConsumerEnergyCo‐Campbell EUAUXBLR12 LightLiquidN/A9/1/2009 2 320 149FAIL MIConsumerEnergyCo‐Campbell EUAUXBLR12 LightLiquidN/A9/1/2009 3 320 124FAIL MIConsumerEnergyCo‐Campbell EUAUXBLR12 LightLiquidN/A9/1/2009 4 320 124PASS MNGPDuluth EU33Boiler#3HeavyLiquid N/A8/6/2009 1 26A 120 4.17 8.0E+02PASS MNGPDuluth EU33Boiler#3HeavyLiquid N/A8/6/2009 2 26A 120 4.38 9.0E+02PASS MNGPDuluth EU33Boiler#3HeavyLiquid N/A8/6/2009 3 26A 120 3.96 6.3E+02PASS MSGPNewAugusta AA‐015PowerBoiler DryBiomass Stoker/SlopedGrate/Other7/12/2006 1 26 60 22.3 9.2E+01 NodetailedFieldDatasheets(norealdocumentationPASS MSGPNewAugusta AA‐015PowerBoiler DryBiomass Stoker/SlopedGrate/Other7/12/2006 2 26 60 21.1 8.3E+01PASS MSGPNewAugusta AA‐015PowerBoiler DryBiomass Stoker/SlopedGrate/Other7/12/2006 3 26 60 19.8 2.9E+01PASS MSGPNewAugusta AA‐015PowerBoiler DryBiomass Stoker/SlopedGrate/Other9/22/2010 2 26 240 3.1E+02ThetestdonothavetheQA/QCforthepitotandthemeterbox.Thecontractorsaidtheinformationwasretainedonsite.IfdocumentationisneededforthatIwouldrecommendcontactingthecontractor.Theppmvalueisnotat7%O2itisat3%PASS MSGPNewAugusta AA‐015PowerBoiler DryBiomass Stoker/SlopedGrate/Other9/22/2010 3 26 240 4.4E+02PASS MSGPNewAugusta AA‐015PowerBoiler DryBiomass Stoker/SlopedGrate/Other9/22/2010 4 26 240 2.6E+02PASS MSGPNewAugusta AA‐015PowerBoiler DryBiomass Stoker/SlopedGrate/Other9/28/2010 1 26 240 2.2E+02ThetestdonothavetheQA/QCforthepitotandthemeterbox.Thecontractorsaidtheinformationwasretainedonsite.IfdocumentationisneededforthatIwouldrecommendcontactingthecontractor.Theppmvalueisnotat7%O2itisat3%PASS MSGPNewAugusta AA‐015PowerBoiler DryBiomass Stoker/SlopedGrate/Other9/28/2010 2 26 240 1.0E+02PASS MSGPNewAugusta AA‐015PowerBoiler DryBiomass Stoker/SlopedGrate/Other9/28/2010 3 26 240 2.6E+02PASS NCLPRoaringRiverBoiler#2WetBiomassStoker/SlopedGrate/Other10/13/2004 1 26A 64 24 2.3E+02PASS NCLPRoaringRiverBoiler#2WetBiomassStoker/SlopedGrate/Other10/13/2004 2 26A 64 25 2.3E+02PASS NCLPRoaringRiverBoiler#2WetBiomassStoker/SlopedGrate/Other10/13/2004 3 26A 64 25 2.3E+02PASS PABoeingRidleyPark033 HeavyLiquid N/A9/22/2009 1 26A 120 13.1 1.4E+03MMBtuvaluewascalculatedusingmethod19andanffactorwasusedinthelb/MMBtuvaluePASS PABoeingRidleyPark033 HeavyLiquid N/A9/22/2009 2 26A 120 13.7 1.2E+03PASS PABoeingRidleyPark033 HeavyLiquid N/A9/22/2009 3 26A 120 13.3 9.5E+02PASS SCDAKAmericas P8F HeavyLiquid N/A 7/15/2009 1 26A 120 13.99525514 3.6 1.4E+03PASS SCDAKAmericas P8F HeavyLiquid N/A 40009 2 26A 120 14.0534304 3.5 1.3E+03PASS SCDAKAmericas P8F HeavyLiquid N/A 40010 3 26A 120 14.61601307 3.6 1.4E+03PASS TNBowaterNewsprint BFBNorth Biomass 40092 1 26A 60 18.75146841 5.6 3.1E+02PASS TNBowaterNewsprint North Biomass 40092 2 26A 60 22.16646681 5.4 1.5E+03PASS TNBowaterNewsprint North Biomass 40092 3 26A 60 21.97186437 5.8 3.9E+02FAIL TNBowaterNewsprint BFBSouth Biomass 40092 1 26A 60Failedthermocouplecalibration(differenceof6.9%recorded,shouldbelessthan1.5%);pitotnozzlediameteris0.225,not0.235inchesFAIL TNBowaterNewsprint South Biomass 40092 2 26A 60Failedthermocouplecalibration(differenceof6.9%recorded,shouldbelessthan1.5%);pitotnozzlediameteris0.225,not0.235inchesFAIL TNBowaterNewsprint South Biomass 40092 3 26A 60Failedthermocouplecalibration(differenceof6.9%recorded,shouldbelessthan1.5%)PASS WINewPageBiron B24‐boiler5 Coal Stoker/SlopedGrate/Other 9/22/2009 1 26A 140 11.78219396 7.1 5.3E+03PASS WINewPageBiron B24‐boiler5 Coal Stoker/SlopedGrate/Other 9/22/2009 2 26A 140 11.07332599 7.1 5.7E+03PASS WINewPageBiron B24‐boiler5 Coal Stoker/SlopedGrate/Other 9/22/2009 3 26A 140 11.00051762 9.7 5.0E+03TestignoredfromallanalysessincealltestrunsareflaggedasfailuresTestignoredfromallanalysessincealltestrunsareflaggedasfailuresTestignoredfromallanalysessincealltestrunsareflaggedasfailuresTestignoredfromallanalysessincealltestrunsareflaggedasfailuresTestignoredfromallanalysessincealltestrunsareflaggedasfailures