adsabs/AstroCLIMB huggingface dataset:
Paper: The Simons Observatory: Design, Integration, and Testing of the Small Aperture Telescopes
Authors: Nicholas Galitzki, Tran Tsan, Jake Spisak, Michael Randall, Max Silva-Feaver, Joseph Seibert, Jacob Lashner, Shunsuke Adachi, Sean M. Adkins, Thomas Alford, Kam Arnold, Peter C. Ashton, Jason E. Austermann, Carlo Baccigalupi, Andrew Bazarko, James A. Beall, Sanah Bhimani, Bryce Bixler, Gabriele Coppi, Lance Corbett, Kevin D. Crowley, Kevin T. Crowley, Samuel Day-Weiss, Mark J. Devlin, Simon Dicker, Brooke DiGia, Peter N. Dow, Cody J. Duell, Shannon M. Duff, Remington G. Gerras, John C. Groh, Jon E. Gudmundsson, Kathleen Harrington, Masaya Hasegawa, Erin Healy, Shawn W. Henderson, Johannes Hubmayr, Jeffrey Iuliano, Bradley R. Johnson, Brian Keating, Ben Keller, Kenji Kiuchi, Anna M. Kofman, Brian J. Koopman, Akito Kusaka, Adrian T. Lee, Richard A. Lew, Lawrence T. Lin, Michael J. Link, Tammy J. Lucas, Marius Lungu, Aashrita Mangu, Jeffrey J McMahon, Amber D. Miller, Jenna E. Moore, Magdy Morshed, Hironobu Nakata, Federico Nati, Laura B. Newburgh, David V. Nguyen, Michael D. Niemack, Lyman A. Page, Kana Sakaguri, Yuki Sakurai, Mayuri Sathyanarayana Rao, Lauren J. Saunders, Jordan E. Shroyer, Junna Sugiyama, Osamu Tajima, Atsuto Takeuchi, Refilwe Tanah Bua, Grant Teply, Tomoki Terasaki, Joel N. Ullom, Jeffrey L. Van Lanen, Eve M. Vavagiakis, Michael R Vissers, Liam Walters, Yuhan Wang, Zhilei Xu, Kyohei Yamada, Kaiwen Zheng
Figure caption: Example of a COMSOL simulation to assess thermal gradients across the PTC2 stage with the fiducial cold head value of 4.2 K used as a conservative upper bound. The stage walls use 1100 aluminum, the stage plates use 6061 aluminum, heat straps are composed of a combination of 4N aluminum and OFHC copper, and all surface contacts are assumed to be ideal. Though the actual Al heat straps are 5N8, 4N Al serves as a reasonable approximation and leads to a more conservative estimate of the thermal gradient. In the design phase, the simulations guided materials choices and heat-strap locations. In the testing phase, we used the simulations to assess measured temperatures and gradients, and set upper bounds on realized loading.
Paper: The Simons Observatory: Design, Integration, and Testing of the Small Aperture Telescopes
Authors: Nicholas Galitzki, Tran Tsan, Jake Spisak, Michael Randall, Max Silva-Feaver, Joseph Seibert, Jacob Lashner, Shunsuke Adachi, Sean M. Adkins, Thomas Alford, Kam Arnold, Peter C. Ashton, Jason E. Austermann, Carlo Baccigalupi, Andrew Bazarko, James A. Beall, Sanah Bhimani, Bryce Bixler, Gabriele Coppi, Lance Corbett, Kevin D. Crowley, Kevin T. Crowley, Samuel Day-Weiss, Mark J. Devlin, Simon Dicker, Brooke DiGia, Peter N. Dow, Cody J. Duell, Shannon M. Duff, Remington G. Gerras, John C. Groh, Jon E. Gudmundsson, Kathleen Harrington, Masaya Hasegawa, Erin Healy, Shawn W. Henderson, Johannes Hubmayr, Jeffrey Iuliano, Bradley R. Johnson, Brian Keating, Ben Keller, Kenji Kiuchi, Anna M. Kofman, Brian J. Koopman, Akito Kusaka, Adrian T. Lee, Richard A. Lew, Lawrence T. Lin, Michael J. Link, Tammy J. Lucas, Marius Lungu, Aashrita Mangu, Jeffrey J McMahon, Amber D. Miller, Jenna E. Moore, Magdy Morshed, Hironobu Nakata, Federico Nati, Laura B. Newburgh, David V. Nguyen, Michael D. Niemack, Lyman A. Page, Kana Sakaguri, Yuki Sakurai, Mayuri Sathyanarayana Rao, Lauren J. Saunders, Jordan E. Shroyer, Junna Sugiyama, Osamu Tajima, Atsuto Takeuchi, Refilwe Tanah Bua, Grant Teply, Tomoki Terasaki, Joel N. Ullom, Jeffrey L. Van Lanen, Eve M. Vavagiakis, Michael R Vissers, Liam Walters, Yuhan Wang, Zhilei Xu, Kyohei Yamada, Kaiwen Zheng
Figure caption: Schematic diagram of the helium and vacuum systems on the SAT including all hoses between the PTC compressors, turbo rack, GHS, and the SAT itself with manual valves called out. The location and length of the SATP cable wraps for azimuth (AZ), elevation (EL), and boresight (BS) as well as the conduit run are identified. The vacuum lines are installed temporarily for service as they are not used in operation and thus do not route through the cable wrap. The KF-40 line pulls vacuum from the SAT, while the KF-16 line simplifies servicing and pulling vacuum on he DR circuit. Red dashed lines denote distinct zones of the telescope where equipment is mounted.
Paper: The Simons Observatory: Design, Integration, and Testing of the Small Aperture Telescopes
Authors: Nicholas Galitzki, Tran Tsan, Jake Spisak, Michael Randall, Max Silva-Feaver, Joseph Seibert, Jacob Lashner, Shunsuke Adachi, Sean M. Adkins, Thomas Alford, Kam Arnold, Peter C. Ashton, Jason E. Austermann, Carlo Baccigalupi, Andrew Bazarko, James A. Beall, Sanah Bhimani, Bryce Bixler, Gabriele Coppi, Lance Corbett, Kevin D. Crowley, Kevin T. Crowley, Samuel Day-Weiss, Mark J. Devlin, Simon Dicker, Brooke DiGia, Peter N. Dow, Cody J. Duell, Shannon M. Duff, Remington G. Gerras, John C. Groh, Jon E. Gudmundsson, Kathleen Harrington, Masaya Hasegawa, Erin Healy, Shawn W. Henderson, Johannes Hubmayr, Jeffrey Iuliano, Bradley R. Johnson, Brian Keating, Ben Keller, Kenji Kiuchi, Anna M. Kofman, Brian J. Koopman, Akito Kusaka, Adrian T. Lee, Richard A. Lew, Lawrence T. Lin, Michael J. Link, Tammy J. Lucas, Marius Lungu, Aashrita Mangu, Jeffrey J McMahon, Amber D. Miller, Jenna E. Moore, Magdy Morshed, Hironobu Nakata, Federico Nati, Laura B. Newburgh, David V. Nguyen, Michael D. Niemack, Lyman A. Page, Kana Sakaguri, Yuki Sakurai, Mayuri Sathyanarayana Rao, Lauren J. Saunders, Jordan E. Shroyer, Junna Sugiyama, Osamu Tajima, Atsuto Takeuchi, Refilwe Tanah Bua, Grant Teply, Tomoki Terasaki, Joel N. Ullom, Jeffrey L. Van Lanen, Eve M. Vavagiakis, Michael R Vissers, Liam Walters, Yuhan Wang, Zhilei Xu, Kyohei Yamada, Kaiwen Zheng
Figure caption: Top: the SAT-MF1 G10 vacuum/PTC1/PTC2 mechanical truss (inverted). Middle: the assembled truss after MLI has been attached to individual tabs and the RF shields have been installed but before the MLI skirts have been added. There are 40 layers of MLI on the PTC1 stage and 10 layers on the PTC2 stage. See Figure 2 for more details of the truss design. Left: CAD image of a single vacuum to PTC1 G10 monolithic truss tab. Bottom right: the testing setup of a vacuum to PTC1 tab on the Instron universal testing machine. A series of tabs were tested to destruction as part of our development program and subsequent validation of the SAT G10 truss.
Paper: The Simons Observatory: Design, Integration, and Testing of the Small Aperture Telescopes
Authors: Nicholas Galitzki, Tran Tsan, Jake Spisak, Michael Randall, Max Silva-Feaver, Joseph Seibert, Jacob Lashner, Shunsuke Adachi, Sean M. Adkins, Thomas Alford, Kam Arnold, Peter C. Ashton, Jason E. Austermann, Carlo Baccigalupi, Andrew Bazarko, James A. Beall, Sanah Bhimani, Bryce Bixler, Gabriele Coppi, Lance Corbett, Kevin D. Crowley, Kevin T. Crowley, Samuel Day-Weiss, Mark J. Devlin, Simon Dicker, Brooke DiGia, Peter N. Dow, Cody J. Duell, Shannon M. Duff, Remington G. Gerras, John C. Groh, Jon E. Gudmundsson, Kathleen Harrington, Masaya Hasegawa, Erin Healy, Shawn W. Henderson, Johannes Hubmayr, Jeffrey Iuliano, Bradley R. Johnson, Brian Keating, Ben Keller, Kenji Kiuchi, Anna M. Kofman, Brian J. Koopman, Akito Kusaka, Adrian T. Lee, Richard A. Lew, Lawrence T. Lin, Michael J. Link, Tammy J. Lucas, Marius Lungu, Aashrita Mangu, Jeffrey J McMahon, Amber D. Miller, Jenna E. Moore, Magdy Morshed, Hironobu Nakata, Federico Nati, Laura B. Newburgh, David V. Nguyen, Michael D. Niemack, Lyman A. Page, Kana Sakaguri, Yuki Sakurai, Mayuri Sathyanarayana Rao, Lauren J. Saunders, Jordan E. Shroyer, Junna Sugiyama, Osamu Tajima, Atsuto Takeuchi, Refilwe Tanah Bua, Grant Teply, Tomoki Terasaki, Joel N. Ullom, Jeffrey L. Van Lanen, Eve M. Vavagiakis, Michael R Vissers, Liam Walters, Yuhan Wang, Zhilei Xu, Kyohei Yamada, Kaiwen Zheng
Figure caption: Monolithic G10 tabs were tested to destruction to ensure the design met our specifications (see Table 3). Top: the results from testing two vacuum to PTC1 type tabs. The blue line is from a tab as manufactured. The orange line shows a tab that, prior to being tested, has been flexed 100 times with the two ends of the tab displaced horizontally by 2 mm to simulate the maximum expected flex from the differential contraction of the truss rings. Bottom: fatigue testing of a tab that was repeatedly stressed to different loading set points 10 times at each set point before its ultimate failure. All three tabs exceeded our requirements with typical ultimate strengths >10.5 kN.
Paper: The Simons Observatory: Design, Integration, and Testing of the Small Aperture Telescopes
Authors: Nicholas Galitzki, Tran Tsan, Jake Spisak, Michael Randall, Max Silva-Feaver, Joseph Seibert, Jacob Lashner, Shunsuke Adachi, Sean M. Adkins, Thomas Alford, Kam Arnold, Peter C. Ashton, Jason E. Austermann, Carlo Baccigalupi, Andrew Bazarko, James A. Beall, Sanah Bhimani, Bryce Bixler, Gabriele Coppi, Lance Corbett, Kevin D. Crowley, Kevin T. Crowley, Samuel Day-Weiss, Mark J. Devlin, Simon Dicker, Brooke DiGia, Peter N. Dow, Cody J. Duell, Shannon M. Duff, Remington G. Gerras, John C. Groh, Jon E. Gudmundsson, Kathleen Harrington, Masaya Hasegawa, Erin Healy, Shawn W. Henderson, Johannes Hubmayr, Jeffrey Iuliano, Bradley R. Johnson, Brian Keating, Ben Keller, Kenji Kiuchi, Anna M. Kofman, Brian J. Koopman, Akito Kusaka, Adrian T. Lee, Richard A. Lew, Lawrence T. Lin, Michael J. Link, Tammy J. Lucas, Marius Lungu, Aashrita Mangu, Jeffrey J McMahon, Amber D. Miller, Jenna E. Moore, Magdy Morshed, Hironobu Nakata, Federico Nati, Laura B. Newburgh, David V. Nguyen, Michael D. Niemack, Lyman A. Page, Kana Sakaguri, Yuki Sakurai, Mayuri Sathyanarayana Rao, Lauren J. Saunders, Jordan E. Shroyer, Junna Sugiyama, Osamu Tajima, Atsuto Takeuchi, Refilwe Tanah Bua, Grant Teply, Tomoki Terasaki, Joel N. Ullom, Jeffrey L. Van Lanen, Eve M. Vavagiakis, Michael R Vissers, Liam Walters, Yuhan Wang, Zhilei Xu, Kyohei Yamada, Kaiwen Zheng
Figure caption: Stress tests of the CFRP PTC2/OT truss in shear and the CFRP OT/FPA truss in all three configurations. After some initial settling, all tests show the trusses remain in the elastic regime as predicted, exceeding the loading requirements set in Table 5 (horizontal lines). The FPA compression and tension tests start from an offset of +150 N and −150 N, respectively, due to the weight of the testing jig.
Paper: The Simons Observatory: Design, Integration, and Testing of the Small Aperture Telescopes
Authors: Nicholas Galitzki, Tran Tsan, Jake Spisak, Michael Randall, Max Silva-Feaver, Joseph Seibert, Jacob Lashner, Shunsuke Adachi, Sean M. Adkins, Thomas Alford, Kam Arnold, Peter C. Ashton, Jason E. Austermann, Carlo Baccigalupi, Andrew Bazarko, James A. Beall, Sanah Bhimani, Bryce Bixler, Gabriele Coppi, Lance Corbett, Kevin D. Crowley, Kevin T. Crowley, Samuel Day-Weiss, Mark J. Devlin, Simon Dicker, Brooke DiGia, Peter N. Dow, Cody J. Duell, Shannon M. Duff, Remington G. Gerras, John C. Groh, Jon E. Gudmundsson, Kathleen Harrington, Masaya Hasegawa, Erin Healy, Shawn W. Henderson, Johannes Hubmayr, Jeffrey Iuliano, Bradley R. Johnson, Brian Keating, Ben Keller, Kenji Kiuchi, Anna M. Kofman, Brian J. Koopman, Akito Kusaka, Adrian T. Lee, Richard A. Lew, Lawrence T. Lin, Michael J. Link, Tammy J. Lucas, Marius Lungu, Aashrita Mangu, Jeffrey J McMahon, Amber D. Miller, Jenna E. Moore, Magdy Morshed, Hironobu Nakata, Federico Nati, Laura B. Newburgh, David V. Nguyen, Michael D. Niemack, Lyman A. Page, Kana Sakaguri, Yuki Sakurai, Mayuri Sathyanarayana Rao, Lauren J. Saunders, Jordan E. Shroyer, Junna Sugiyama, Osamu Tajima, Atsuto Takeuchi, Refilwe Tanah Bua, Grant Teply, Tomoki Terasaki, Joel N. Ullom, Jeffrey L. Van Lanen, Eve M. Vavagiakis, Michael R Vissers, Liam Walters, Yuhan Wang, Zhilei Xu, Kyohei Yamada, Kaiwen Zheng
Figure caption: Top left: the CFRP PTC2/OT truss with the RF shield installed. Bottom left: the CFRP OT/FPA truss with the RF shield installed. The carbon fiber supports and copper ring for the CRA components at the MC stage is also integrated into the pictured assembly. Right: validation of the CFRP PTC2/OT truss in the Instron universal testing machine for the shear force test.
Paper: The Simons Observatory: Design, Integration, and Testing of the Small Aperture Telescopes
Authors: Nicholas Galitzki, Tran Tsan, Jake Spisak, Michael Randall, Max Silva-Feaver, Joseph Seibert, Jacob Lashner, Shunsuke Adachi, Sean M. Adkins, Thomas Alford, Kam Arnold, Peter C. Ashton, Jason E. Austermann, Carlo Baccigalupi, Andrew Bazarko, James A. Beall, Sanah Bhimani, Bryce Bixler, Gabriele Coppi, Lance Corbett, Kevin D. Crowley, Kevin T. Crowley, Samuel Day-Weiss, Mark J. Devlin, Simon Dicker, Brooke DiGia, Peter N. Dow, Cody J. Duell, Shannon M. Duff, Remington G. Gerras, John C. Groh, Jon E. Gudmundsson, Kathleen Harrington, Masaya Hasegawa, Erin Healy, Shawn W. Henderson, Johannes Hubmayr, Jeffrey Iuliano, Bradley R. Johnson, Brian Keating, Ben Keller, Kenji Kiuchi, Anna M. Kofman, Brian J. Koopman, Akito Kusaka, Adrian T. Lee, Richard A. Lew, Lawrence T. Lin, Michael J. Link, Tammy J. Lucas, Marius Lungu, Aashrita Mangu, Jeffrey J McMahon, Amber D. Miller, Jenna E. Moore, Magdy Morshed, Hironobu Nakata, Federico Nati, Laura B. Newburgh, David V. Nguyen, Michael D. Niemack, Lyman A. Page, Kana Sakaguri, Yuki Sakurai, Mayuri Sathyanarayana Rao, Lauren J. Saunders, Jordan E. Shroyer, Junna Sugiyama, Osamu Tajima, Atsuto Takeuchi, Refilwe Tanah Bua, Grant Teply, Tomoki Terasaki, Joel N. Ullom, Jeffrey L. Van Lanen, Eve M. Vavagiakis, Michael R Vissers, Liam Walters, Yuhan Wang, Zhilei Xu, Kyohei Yamada, Kaiwen Zheng
Figure caption: Cross section of the optical design of the SAT with ray tracing for central and edge pixels on the focal plane. See Section 3.8 for more details.
Paper: The Simons Observatory: Design, Integration, and Testing of the Small Aperture Telescopes
Authors: Nicholas Galitzki, Tran Tsan, Jake Spisak, Michael Randall, Max Silva-Feaver, Joseph Seibert, Jacob Lashner, Shunsuke Adachi, Sean M. Adkins, Thomas Alford, Kam Arnold, Peter C. Ashton, Jason E. Austermann, Carlo Baccigalupi, Andrew Bazarko, James A. Beall, Sanah Bhimani, Bryce Bixler, Gabriele Coppi, Lance Corbett, Kevin D. Crowley, Kevin T. Crowley, Samuel Day-Weiss, Mark J. Devlin, Simon Dicker, Brooke DiGia, Peter N. Dow, Cody J. Duell, Shannon M. Duff, Remington G. Gerras, John C. Groh, Jon E. Gudmundsson, Kathleen Harrington, Masaya Hasegawa, Erin Healy, Shawn W. Henderson, Johannes Hubmayr, Jeffrey Iuliano, Bradley R. Johnson, Brian Keating, Ben Keller, Kenji Kiuchi, Anna M. Kofman, Brian J. Koopman, Akito Kusaka, Adrian T. Lee, Richard A. Lew, Lawrence T. Lin, Michael J. Link, Tammy J. Lucas, Marius Lungu, Aashrita Mangu, Jeffrey J McMahon, Amber D. Miller, Jenna E. Moore, Magdy Morshed, Hironobu Nakata, Federico Nati, Laura B. Newburgh, David V. Nguyen, Michael D. Niemack, Lyman A. Page, Kana Sakaguri, Yuki Sakurai, Mayuri Sathyanarayana Rao, Lauren J. Saunders, Jordan E. Shroyer, Junna Sugiyama, Osamu Tajima, Atsuto Takeuchi, Refilwe Tanah Bua, Grant Teply, Tomoki Terasaki, Joel N. Ullom, Jeffrey L. Van Lanen, Eve M. Vavagiakis, Michael R Vissers, Liam Walters, Yuhan Wang, Zhilei Xu, Kyohei Yamada, Kaiwen Zheng
Figure caption: Diagram of the SAT housekeeping system and related components. Principal locations of components are identified with different color schemes. Lines denote cable connections between locations. Internal connections, chiefly ethernet and power, are omitted. Also omitted are vendor-supplied systems for SATP monitoring, such as motors and encoders. More details on the computing structure of the housekeeping system can be found in Koopman et al. (2020).
Paper: The Simons Observatory: Design, Integration, and Testing of the Small Aperture Telescopes
Authors: Nicholas Galitzki, Tran Tsan, Jake Spisak, Michael Randall, Max Silva-Feaver, Joseph Seibert, Jacob Lashner, Shunsuke Adachi, Sean M. Adkins, Thomas Alford, Kam Arnold, Peter C. Ashton, Jason E. Austermann, Carlo Baccigalupi, Andrew Bazarko, James A. Beall, Sanah Bhimani, Bryce Bixler, Gabriele Coppi, Lance Corbett, Kevin D. Crowley, Kevin T. Crowley, Samuel Day-Weiss, Mark J. Devlin, Simon Dicker, Brooke DiGia, Peter N. Dow, Cody J. Duell, Shannon M. Duff, Remington G. Gerras, John C. Groh, Jon E. Gudmundsson, Kathleen Harrington, Masaya Hasegawa, Erin Healy, Shawn W. Henderson, Johannes Hubmayr, Jeffrey Iuliano, Bradley R. Johnson, Brian Keating, Ben Keller, Kenji Kiuchi, Anna M. Kofman, Brian J. Koopman, Akito Kusaka, Adrian T. Lee, Richard A. Lew, Lawrence T. Lin, Michael J. Link, Tammy J. Lucas, Marius Lungu, Aashrita Mangu, Jeffrey J McMahon, Amber D. Miller, Jenna E. Moore, Magdy Morshed, Hironobu Nakata, Federico Nati, Laura B. Newburgh, David V. Nguyen, Michael D. Niemack, Lyman A. Page, Kana Sakaguri, Yuki Sakurai, Mayuri Sathyanarayana Rao, Lauren J. Saunders, Jordan E. Shroyer, Junna Sugiyama, Osamu Tajima, Atsuto Takeuchi, Refilwe Tanah Bua, Grant Teply, Tomoki Terasaki, Joel N. Ullom, Jeffrey L. Van Lanen, Eve M. Vavagiakis, Michael R Vissers, Liam Walters, Yuhan Wang, Zhilei Xu, Kyohei Yamada, Kaiwen Zheng
Figure caption: The fully assembled SAT-MF1 in the UC San Diego laboratory space. The rack mounted to the side houses the turbo-molecular pumps for the DR circuit and is attached to the receiver with a mounting jig for testing.
Paper: The Simons Observatory: Design, Integration, and Testing of the Small Aperture Telescopes
Authors: Nicholas Galitzki, Tran Tsan, Jake Spisak, Michael Randall, Max Silva-Feaver, Joseph Seibert, Jacob Lashner, Shunsuke Adachi, Sean M. Adkins, Thomas Alford, Kam Arnold, Peter C. Ashton, Jason E. Austermann, Carlo Baccigalupi, Andrew Bazarko, James A. Beall, Sanah Bhimani, Bryce Bixler, Gabriele Coppi, Lance Corbett, Kevin D. Crowley, Kevin T. Crowley, Samuel Day-Weiss, Mark J. Devlin, Simon Dicker, Brooke DiGia, Peter N. Dow, Cody J. Duell, Shannon M. Duff, Remington G. Gerras, John C. Groh, Jon E. Gudmundsson, Kathleen Harrington, Masaya Hasegawa, Erin Healy, Shawn W. Henderson, Johannes Hubmayr, Jeffrey Iuliano, Bradley R. Johnson, Brian Keating, Ben Keller, Kenji Kiuchi, Anna M. Kofman, Brian J. Koopman, Akito Kusaka, Adrian T. Lee, Richard A. Lew, Lawrence T. Lin, Michael J. Link, Tammy J. Lucas, Marius Lungu, Aashrita Mangu, Jeffrey J McMahon, Amber D. Miller, Jenna E. Moore, Magdy Morshed, Hironobu Nakata, Federico Nati, Laura B. Newburgh, David V. Nguyen, Michael D. Niemack, Lyman A. Page, Kana Sakaguri, Yuki Sakurai, Mayuri Sathyanarayana Rao, Lauren J. Saunders, Jordan E. Shroyer, Junna Sugiyama, Osamu Tajima, Atsuto Takeuchi, Refilwe Tanah Bua, Grant Teply, Tomoki Terasaki, Joel N. Ullom, Jeffrey L. Van Lanen, Eve M. Vavagiakis, Michael R Vissers, Liam Walters, Yuhan Wang, Zhilei Xu, Kyohei Yamada, Kaiwen Zheng
Figure caption: The RF chain consists of the coaxial cables and components that bring signals from the digital signal processing electronics at room temperature down to the cryogenic multiplexer chips and back up to enable readout of our ﹩{ \mathcal O }﹩(10,000) TES focal plane. The schematic details the components for a single readout chain, with two chains needed for each UFM. The left side of the image is the room-temperature SMuRF electronics and on the right side is the connection to the detector arrays at the 100 mK focal plane. The colored cylinders denote the isothermal and intertemperature coaxial cables construction material and cross section with 0.86 mm CuN (light blue), 2.19 mm CuN (dark blue), 0.″141 hand-formable copper (gray), and 1.19 mm NbTi (pink). All use solid PTFE dielectrics. The isothermal cables are hand flexed to shape during assembly and are sourced from Mini-Circuits for the SMA to SMA sections and from Centric RF for the 100 mK SMA to SMP section. The intertemperature stage coax comes assembled and bent to shape from Coax Co. More details on all of the components in RF chain are given in Rao et al. (2020). Note that in addition to the fixed attenuation, losses in the microwave cabling at temperatures below 4.2 K along the “RF In” line reduce the thermal noise within the coax such that it is subdominant to other noise terms.
Paper: The Simons Observatory: Design, Integration, and Testing of the Small Aperture Telescopes
Authors: Nicholas Galitzki, Tran Tsan, Jake Spisak, Michael Randall, Max Silva-Feaver, Joseph Seibert, Jacob Lashner, Shunsuke Adachi, Sean M. Adkins, Thomas Alford, Kam Arnold, Peter C. Ashton, Jason E. Austermann, Carlo Baccigalupi, Andrew Bazarko, James A. Beall, Sanah Bhimani, Bryce Bixler, Gabriele Coppi, Lance Corbett, Kevin D. Crowley, Kevin T. Crowley, Samuel Day-Weiss, Mark J. Devlin, Simon Dicker, Brooke DiGia, Peter N. Dow, Cody J. Duell, Shannon M. Duff, Remington G. Gerras, John C. Groh, Jon E. Gudmundsson, Kathleen Harrington, Masaya Hasegawa, Erin Healy, Shawn W. Henderson, Johannes Hubmayr, Jeffrey Iuliano, Bradley R. Johnson, Brian Keating, Ben Keller, Kenji Kiuchi, Anna M. Kofman, Brian J. Koopman, Akito Kusaka, Adrian T. Lee, Richard A. Lew, Lawrence T. Lin, Michael J. Link, Tammy J. Lucas, Marius Lungu, Aashrita Mangu, Jeffrey J McMahon, Amber D. Miller, Jenna E. Moore, Magdy Morshed, Hironobu Nakata, Federico Nati, Laura B. Newburgh, David V. Nguyen, Michael D. Niemack, Lyman A. Page, Kana Sakaguri, Yuki Sakurai, Mayuri Sathyanarayana Rao, Lauren J. Saunders, Jordan E. Shroyer, Junna Sugiyama, Osamu Tajima, Atsuto Takeuchi, Refilwe Tanah Bua, Grant Teply, Tomoki Terasaki, Joel N. Ullom, Jeffrey L. Van Lanen, Eve M. Vavagiakis, Michael R Vissers, Liam Walters, Yuhan Wang, Zhilei Xu, Kyohei Yamada, Kaiwen Zheng
Figure caption: Transmission spectra above 10 cm−1 of the IR blocking filters used in the testing phase of SAT-MF1. The two double-sided metal-mesh filters had 15 μm grid spacing and 4 μm substrate thickness. The alumina sample filter has no antireflective coating (ARC) but is of a similar type and thickness to the filters used for testing in SAT-MF1. Measurements were made with a Fourier transform spectrometer (FTS) at room temperature at Cardiff.
Paper: The Simons Observatory: Design, Integration, and Testing of the Small Aperture Telescopes
Authors: Nicholas Galitzki, Tran Tsan, Jake Spisak, Michael Randall, Max Silva-Feaver, Joseph Seibert, Jacob Lashner, Shunsuke Adachi, Sean M. Adkins, Thomas Alford, Kam Arnold, Peter C. Ashton, Jason E. Austermann, Carlo Baccigalupi, Andrew Bazarko, James A. Beall, Sanah Bhimani, Bryce Bixler, Gabriele Coppi, Lance Corbett, Kevin D. Crowley, Kevin T. Crowley, Samuel Day-Weiss, Mark J. Devlin, Simon Dicker, Brooke DiGia, Peter N. Dow, Cody J. Duell, Shannon M. Duff, Remington G. Gerras, John C. Groh, Jon E. Gudmundsson, Kathleen Harrington, Masaya Hasegawa, Erin Healy, Shawn W. Henderson, Johannes Hubmayr, Jeffrey Iuliano, Bradley R. Johnson, Brian Keating, Ben Keller, Kenji Kiuchi, Anna M. Kofman, Brian J. Koopman, Akito Kusaka, Adrian T. Lee, Richard A. Lew, Lawrence T. Lin, Michael J. Link, Tammy J. Lucas, Marius Lungu, Aashrita Mangu, Jeffrey J McMahon, Amber D. Miller, Jenna E. Moore, Magdy Morshed, Hironobu Nakata, Federico Nati, Laura B. Newburgh, David V. Nguyen, Michael D. Niemack, Lyman A. Page, Kana Sakaguri, Yuki Sakurai, Mayuri Sathyanarayana Rao, Lauren J. Saunders, Jordan E. Shroyer, Junna Sugiyama, Osamu Tajima, Atsuto Takeuchi, Refilwe Tanah Bua, Grant Teply, Tomoki Terasaki, Joel N. Ullom, Jeffrey L. Van Lanen, Eve M. Vavagiakis, Michael R Vissers, Liam Walters, Yuhan Wang, Zhilei Xu, Kyohei Yamada, Kaiwen Zheng
Figure caption: Temperature response of the PTC integrated into the DR (black and green) and of the DR cold heads (red and blue) over six tilt angles. The tilt angle shown is that of the PTC and DR cold heads relative to vertical, not the SAT optical axis. The Still and MC stages had a constant power applied during testing of 15 mW and 500 μW, respectively. Data from rotation around the short (long) axis of the receiver perpendicular to the optical axis are shown with dashed (solid) lines (see inset). The data were obtained from two separate cooldowns. The MC temperature at 51.°2 is a lower bound as the temperature had not stabilized due to the increased temperature on the PTC2 stage effectively exceeding the temperature at which the DR circuit can operate stably.