Paper: Overview of the Instrumentation for the Dark Energy Spectroscopic Instrument
Authors: DESI Collaboration, B. Abareshi, J. Aguilar, S. Ahlen, Shadab Alam, David M. Alexander, R. Alfarsy, L. Allen, C. Allende Prieto, O. Alves, J. Ameel, E. Armengaud, J. Asorey, Alejandro Aviles, S. Bailey, A. Balaguera-Antolínez, O. Ballester, C. Baltay, A. Bault, S. F. Beltran, B. Benavides, S. BenZvi, A. Berti, R. Besuner, Florian Beutler, D. Bianchi, C. Blake, P. Blanc, R. Blum, A. Bolton, S. Bose, D. Bramall, S. Brieden, A. Brodzeller, D. Brooks, C. Brownewell, E. Buckley-Geer, R. N. Cahn, Z. Cai, R. Canning, R. Capasso, A. Carnero Rosell, P. Carton, R. Casas, F. J. Castander, J. L. Cervantes-Cota, S. Chabanier, E. Chaussidon, C. Chuang, C. Circosta, S. Cole, A. P. Cooper, L. da Costa, M.-C. Cousinou, A. Cuceu, T. M. Davis, K. Dawson, R. de la Cruz-Noriega, A. de la Macorra, A. de Mattia, J. Della Costa, P. Demmer, M. Derwent, A. Dey, B. Dey, G. Dhungana, Z. Ding, C. Dobson, P. Doel, J. Donald-McCann, J. Donaldson, K. Douglass, Y. Duan, P. Dunlop, J. Edelstein, S. Eftekharzadeh, D. J. Eisenstein, M. Enriquez-Vargas, S. Escoffier, M. Evatt, P. Fagrelius, X. Fan, K. Fanning, V. A. Fawcett, S. Ferraro, J. Ereza, B. Flaugher, A. Font-Ribera, J. E. Forero-Romero, C. S. Frenk, S. Fromenteau, B. T. Gänsicke, C. Garcia-Quintero, L. Garrison, E. Gaztañaga, F. Gerardi, H. Gil-Marín, S. Gontcho A Gontcho, Alma X. Gonzalez-Morales, G. Gonzalez-de-Rivera, V. Gonzalez-Perez, C. Gordon, O. Graur, D. Green, C. Grove, D. Gruen, G. Gutierrez, J. Guy, C. Hahn, S. Harris, D. Herrera, Hiram K. Herrera-Alcantar, K. Honscheid, C. Howlett, D. Huterer, V. Iršič, M. Ishak, P. Jelinsky, L. Jiang, J. Jimenez, Y. P. Jing, R. Joyce, E. Jullo, S. Juneau, N. G. Karaçaylı, M. Karamanis, A. Karcher, T. Karim, R. Kehoe, S. Kent, D. Kirkby, T. Kisner, F. Kitaura, S. E. Koposov, A. Kovács, A. Kremin, Alex Krolewski, B. L'Huillier, O. Lahav, A. Lambert, C. Lamman, Ting-Wen Lan, M. Landriau, S. Lane, D. Lang, J. U. Lange, J. Lasker, L. Le Guillou, A. Leauthaud, A. Le Van Suu, Michael E. Levi, T. S. Li, C. Magneville, M. Manera, Christopher J. Manser, B. Marshall, Paul Martini, W. McCollam, P. McDonald, Aaron M. Meisner, J. Mena-Fernández, J. Meneses-Rizo, M. Mezcua, T. Miller, R. Miquel, P. Montero-Camacho, J. Moon, J. Moustakas, E. Mueller, Andrea Muñoz-Gutiérrez, Adam D. Myers, S. Nadathur, J. Najita, L. Napolitano, E. Neilsen, Jeffrey A. Newman, J. D. Nie, Y. Ning, G. Niz, P. Norberg, Hernán E. Noriega, T. O'Brien, A. Obuljen, N. Palanque-Delabrouille, A. Palmese, P. Zhiwei, D. Pappalardo, X. PENG, W. J. Percival, S. Perruchot, R. Pogge, C. Poppett, A. Porredon, F. Prada, J. Prochaska, R. Pucha, A. Pérez-Fernández, I. Pérez-Ràfols, D. Rabinowitz, A. Raichoor, S. Ramirez-Solano, César Ramírez-Pérez, C. Ravoux, K. Reil, M. Rezaie, A. Rocher, C. Rockosi, N. A. Roe, A. Roodman, A. J. Ross, G. Rossi, R. Ruggeri, V. Ruhlmann-Kleider, C. G. Sabiu, S. Gaines, K. Said, A. Saintonge, Javier Salas Catonga, L. Samushia, E. Sanchez, C. Saulder, E. Schaan, E. Schlafly, D. Schlegel, J. Schmoll, D. Scholte, M. Schubnell, A. Secroun, H. Seo, S. Serrano, Ray M. Sharples, Michael J. Sholl, Joseph Harry Silber, D. R. Silva, M. Sirk, M. Siudek, A. Smith, D. Sprayberry, R. Staten, B. Stupak, T. Tan, Gregory Tarlé, Suk Sien Tie, R. Tojeiro, L. A. Ureña-López, F. Valdes, O. Valenzuela, M. Valluri, M. Vargas-Magaña, L. Verde, M. Walther, B. Wang, M. S. Wang, B. A. Weaver, C. Weaverdyck, R. Wechsler, Michael J. Wilson, J. Yang, Y. Yu, S. Yuan, Christophe Yèche, H. Zhang, K. Zhang, Cheng Zhao, Rongpu Zhou, Zhimin Zhou, H. Zou, J. Zou, S. Zou, Y. Zu
Figure caption: Average measured total throughput compared to the engineering model. The average throughput is measured with standard stars on photometric nights. All measurements have been corrected for losses due to the fiber acceptance fraction and therefore do not include the contribution of seeing. The darker lines (blue, red, and black) show the throughput of the three spectrograph channels with a correction that removed atmospheric extinction, while the lighter lines show the throughput at the zenith. The points are the predicted throughput of the engineering model for both cases. The agreement between the measurement and the model is excellent for the blue channel and very good for the red and NIR channels. The minor differences between the model and the measurement are likely due to many small differences between estimates and the final performance. The dip at 440 nm is due to an issue with the spectrograph collimator coating. The average throughput without atmospheric extinction has been smoothed to reduce the impact of individual telluric absorption features on the measured throughput.