Guiding the Design of Radiation Imagers with Experimentally Benchmarked Geant4 Simulations for Electron-Tracking Compton Imaging

Radiation imagers are import tools in the modern world for a wide range of applications. They span the use-cases of fundamental sciences, astrophysics, medical imaging, all the way to national security, nuclear safeguards, and non-proliferation verification. The type of radiation imagers studied in this thesis were gamma-ray imagers that detect emissions from radioactive materials. Gamma-ray imagers goal is to localize and map the distribution of radiation within their specific field-of-view despite the fact of complicating background radiation that can be terrestrial, astronomical, and temporal. Compton imaging systems are one type of gamma-ray imager that can map the radiation around the system without the use of collimation. Lack of collimation enables the imaging system to be able to detect radiation from all-directions, while at the same time, enables increased detection efficiency by not absorbing incident radiation in non-sensing materials. Each Compton-scatter events within an imaging system generated a possible cone-surface in space that the radiation could have originated from. Compton imaging is limited in its reconstructed image signal-to-background due to these source Compton-cones overlapping with background radiation Compton-cones. These overlapping cones limit Compton imaging's detection-sensitivity in image space. Electron-tracking Compton imaging (ETCI) can improve the detection-sensitivity by measuring the Compton-scattered electron's initial trajectory. With an estimate of the scattered electron's trajectory, one can reduce the Compton-back-projected cone to a cone-arc, thus enabling faster radiation source detection and localization./nHowever, the ability to measure the Compton-scattered electron-trajectories adds another layer of complexity to an already complex methodology. For a real-world imaging applications, improvements are needed in electron-track detection efficiency and in electron-track reconstruction. One way of measuring Compton-scattered electron-trajectories is with high-resolution Charged-Coupled Devices (CCDs). The proof-of-principle CCD-based ETCI experiment demonstrated the CCDs' ability to measure the Compton-scattered electron-tracks as a 2-dimensional image. Electron-track-imaging algorithms using the electron-track-image are able to determine the 3-dimensional electron-track trajectory within +/- 20 degrees. The work presented here is the physics simulations developed along side the experimental proof-of-principle experiment. The development of accurate physics modeling for multiple-layer CCDs based ETCI systems allow for the accurate prediction of future ETCI system performance. The simulations also enable quick development insights for system design, and they guide the development of electron-track reconstruction methods./nThe physics simulation efforts for this project looked closely at the accuracy of the Geant4 Monte Carlo methods for medium energy electron transport. In older version of Geant4 there were some discrepancies between the electron-tracking experimental measurements and the simulation results. It was determined that when comparing the electron dynamics of electrons at very high resolutions, Geant4 simulations must be fine tuned with careful choices for physics production cuts and electron physics stepping sizes. One result of this work is a CCDs Monte Carlo model that has been benchmarked to experimental findings and fully characterized for both photon and electron transport. The CCDs physics model now match to within 1 percent error of experimental results for scattered-electron energies below 500 keV./nFollowing the improvements of the CCDs simulations, the performance of a realistic two-layer CCD-stack system was characterized. The realistic CCD-stack system looked at the effect of thin passive-layers on the CCDs' front face and back-contact. The photon interaction efficiency was calculated for the two-layer CCD-stack, and we found that there is a 90 percent probability of scattered-electrons from a 662 keV source to stay within a single active layer. This demonstrates the improved detection efficiency, which is one of the strengths of the CCDs' implementation as a ETCI system./nThe CCD-stack simulations also established that electron-tracks scattering from one CCDs layer to another could be reconstructed. The passive-regions on the CCD-stack mean that these inter-layer scattered-electron-tracks will always loose both angular information and energy information. Looking at the angular changes of these electrons scattering between the CCDs layers showed us there is not a strong energy dependence on the angular changes due to the passive-regions of the CCDs. The angular changes of the electron track are, for the most part, a function of the thickness of the thin back-layer of the CCDs. Lastly, an approach using CCD-stack simulations was developed to reconstruct the energy transport across dead-layers and its feasibility was demonstrated. Adding back this lost energy will limit the loss of energy resolution of the scatter-interactions. Energy resolution losses would negatively impacted the achievable image resolution from image reconstruction algorithms. Returning some of the energy back to the reconstructed electron-track will help retain the expected performance of the electron-track trajectory determination algorithm.

Advisor: Vetter, Kai