Public procurement notice: Contract Notice
Published: 2023-05-23
Buyer: JZU UK za drzavna kardiohirurgija/ PHI UC for State Cardiac Surgery
Country: MK
Contract type: Supplies
Procedure: Open procedure
CPV classification: 33000000 | 33140000: Medical consumables
Estimated value: 10000000 MKD
Submission deadline: 2023-06-15
Award criteria: Lowest price

Technical requirements: Vascular graft for thoracic aorta powder (woven), polyester, impregnated with collagen, MRI safe, water permeability < 5ml/cm2/min, dilation resistant with minimum diameter range from 6-38mm and a minimum length span of 15-60cm.
ePTFE straight prosthesis reinforced with a ring along the entire length, with dimensions, minimum length of 80cm and diameter: 6mm, 7mm, 8mm.; made of expanded polytetrafluoroethylene (ePTFE)
Thromboembolectomy cath. 2Fr FOGARTY, with shaft of polyurethane with latex balloon 0.2ml, graduated, color coded, with atraumatic tip, single lumen, X-ray visible, with rigid stylet, length 67.5cm, sterile packed in hard packaging, with a tolerance of +/- 3%
Thromboembolectomy cath. 3Fr FOGARTY, (same as above) 0.6ml, graduated, color coded, with atraumatic tip, single lumen, X-ray visible, with rigid stylet, length 82cm, to be sterile packed in hard packaging, with a tolerance of +/- 3%
Thromboembolectomy catheter 4Fr FOGARTY, (same as above) 1.2ml, graduated, color coded, with atraumatic tip, single lumen, X-ray visible, with rigid stylet, length 82cm, to be sterile packed in hard packaging, with a tolerance of +/- 3%
Thromboembolectomy catheter 5Fr FOGARTY, (same as above) 3ml, graduated, color coded, with atraumatic tip, single lumen, X-ray visible, with rigid stylet, length 82cm, to be sterile packed in hard packaging, with a tolerance of +/- 3%
Thromboembolectomy catheter 6Fr FOGARTY, (same as above) 4.5ml, graduated, color coded, with atraumatic tip, single lumen, X-ray visible, with rigid stylet, length 82cm, to be sterile packed in hard packaging, with a tolerance of +/- 3%
Thromboembolectomy catheter 7Fr FOGARTY, (same as above) 5ml, graduated, color coded, with atraumatic tip, single lumen, X-ray visible, with rigid stylet, length 82cm, to be sterile packed in hard packaging, with a tolerance of +/- 3%
Thromboembolectomy catheter 8Fr FOGARTY, (same as above) 5.5ml, graduated, color coded, with atraumatic tip, single lumen, X-ray visible, with rigid stylet, length 82cm, to be sterile packed in hard packaging, with a tolerance of +/- 3%
Carotid shunts TYPE 3 Carotid shunt with "T" side branch for aspiration and irrigation; conical, with a wider end on one side of 12-15Fr., and a narrower end on the other side of 6-9Fr.; with length of 12-15 cm.
Mitral mechanical valve for mitral valve replacement. Constructed of an annular body and outside coated with polyester and Teflon. On the ring body there are two valves made of pyrolytic carbon or titanium; ring diameter size 25mm, 27mm, 29mm, 31mm, 33mm, 35mm, 37mm.; transvalvular gradient across valves of different sizes to be delivered. Valves to be with low transvalvular gradient. On the ring there should be a marker for the placement of the sutures. The valves should have the possibility to rotate (to correct the position of the discs after implantation) i.e. to be rotatable,in situ”,
Composite Ascending Aorta and Valve Replacement Graft composed of an aortic mechanical valve for intraannular implantation and vascular graft. Annular body with two valves, to be built from pyrolytic carbon. Valve ring diameter size to be 19 cm, 21cm, 23cm, 25cm, 27cm, 29cm, 31cm, the graft should be 12cm +/-10% in length, made of double woven tri-marker collagen coated suede.
Complete mitral annulus. Complete annulus for mitral valve repair; complete ring with flexible anterior part of silico/polyester and semi-rigid posterior part of cobalt/chromium; enables physiological movement of the mitral valve in the ratio 3:4; 4:4; diameter of the ring 24mm, 26mm, 28mm, 30mm, 32mm, 34mm, 36mm, 38mm, 40mm.
Tricuspid 3D Ring. Three-dimensional anatomically shaped incomplete ring for repair of tricuspid valve with a silicon and polyester coated titanium core for easing the insertion of the needle; height profile 3.3mm; diameter 26mm, 28mm, 30mm, 32mm, 34mm, 36mm
Pericardial patch from bovine pericardium, treated with glutaraldehyde, which increases durability and prevents collection etc.

Eligibility requirements: - statement of the economic operator that no effective verdict has been pronounced against the economic operator within the past five years for a committed crime stated in Art. 88, paragraph 1 of the LPP; - certificate stating that no bankruptcy procedure has been initiated by a competent body; - certificate stating that no liquidation procedure has been initiated by a competent body; - certificate of paid taxes, contributions and other public duties issued by a competent body in the country where the economic operator is registered; - certificate from the Register of penalties for committed criminal offenses of legal entities proving that no secondary penalty - ban on participating in open call procedures, awarding public procurement contracts and contracts on public - private partnership has been pronounced; - certificate from the Register of penalties for committed criminal offenses of legal entities proving that no secondary penalty - temporary or permanent ban on performing particular activity has been pronounced; - certificate stating that no misdemeanor sanction - ban on the performance of the profession, activity or duty, i.e. temporary ban on the performance of a particular activity has been pronounced with an effective verdict.

Award value: 10000000