Public procurement notice: Contract Notice
Published: 2016-08-27
Buyer: PHI Clinical Hospital Shtip
Country: MK
Region: Istočen
Contract type: Supplies
Procedure: Open procedure
CPV classification: 33000000
Estimated value: 8500000 MKD
Submission deadline: 2016-10-07
Award criteria: Lowest price

Technical requirements: Service of the machine for computer tomography GE Light Speed VCT with installation of an X-ray tube and enabling the machine to function.
— Original new (unrepaired) x-ray tube type Maxiray or Performix for computed tomography General Electric Light Speed VCT 64 from 2011 or equivalent, which supports minimum 800 mA of power by helical and axial scans without interruptions for cooling the samem;
— the insertion of the tube has two focuses: small 0,9 x 0,7 and big 1,2 x 1,1 (according to the standard IE60336/2005) with tolerance ± 2 % or equivalent;
— measure value of power with small or big focus with tolerance of 2±% with the following voltage: 80 kV: 300 mA & 675 mA, 100 kV: 310 mA & 770 mA, 120 kV: 335 Ma & 800 mA, 140 kV: 335 mA & 715 mA;
— effective cooling system through external system of cooling which eliminates the interruptions at the time of encumbered helical scans;
— the whole construction of the x-ray tube can provide stability of the above mentioned dimensions of the small and big focus of ±2 % for a high resolution picture of 1 024 x 1 024 with tolerance of ±1 %, at rotations of 0,35 seconds;
— to support the following single helical protocols of scanning at 120 kV, time of scanning and maximum power according to the tolerance of ±2 %: 5 sec. and 800 mA, 10 sec. and 790 mA, 20 sec. and 645 mA, 30 sec. and 560 mA, 40 sec. and 510 mA, 50 sec. and 485 mA, 60 sec. and 46 0Ma, 80 sec. and 290-430 mA, 100 sec. and 280-400 mA, 120 sec. and 270-280 mA;
— possibility for continuous scanning of minimum 5 sec. at maximum power, without interruption for cooling of the same;
— the x-ray tube should have identification module through which the computed tomography could recognize the type of tube and determine and guarantee the dose of the x-ray radiation to the patient;
— cooling pump of the x-ray tube with transformational oil and built-in diode identification of its functional condition compatible with the real system of cooling of the computed tomography;
— high voltage interface module for interconnection of x-ray tube with generator of high voltage of the computed tomography piece 1;
— memory module of 500 ± 5 % with built-in case in 6 Gbps memory line for recording data directly of the computed tomography with appropriately built-in software for automatic recognition of the same, work power of 5 to 12 volts with tolerance of ±5 % and is controlled by chip with clock generator with maximum speed of 1 333 MHz;
— 2 paired minimum 19 inch monitors with resolution of minimum 1 mp for medical-radiological use with relation 5:4, lightning of minimum 300 cd/m², contrast of min. 2000:1, with minimum 2 entries: DVI-D and D-Sub mini 15 pin and possibility for real adjustment of the same of 40 to 50 cm with tolerance of ±5 % calculated from the top side of the monitors.
— module for recording of CD and DVD media with dimension of 120 mm±1 % and capacity of 700 MB and 4,7 GB accordingly and dimensions of 145 x 175 x 40 mm ±1 % compatible for built-in external case for extents of the console of the computed tomography. set 1.
— maximum manual labor for realization of items 1 and 2, with all the needed adjustments and calibrations for putting the computed tomography into function hour 30.

Eligibility requirements: — Statement of the economical operator that in the past 5 years there has not been adjudged an effective verdict for participating in a criminal organization, corruption, fraud or money laundering; — Confirmation that there is not an open procedure for bankruptcy by a competent authority; — Confirmation that there is not an open procedure for liquidation by a competent authority; — Confirmation for paid taxes, contributions and other public expenditures by a competent authority from the state where the economical operator is registered; — Confirmation from the Register for penalties for criminal acts of the public entities that there aren't imposed minor fines, prohibition for participating in procedures for public call, granting contracts for public procurement and contracts for public-private partnership; — Confirmation from the Register for penalties for criminal acts of the public entities that there aren't imposed minor fines, temporary or permanent prohibition for performing certain activity; — Confirmation that there isn't adjudged a penalty for violation with a legally-binding verdict — prohibition for performing the professional activities and duties, respectively a temporary prohibition for performing certain activity.