Public procurement notice: Contract Notice
Published: 2019-04-04
Buyer: Ss. Cyril and Methodius University Faculty of Medicine Skopje
Country: MK
Contract type: Supplies
Procedure: Open procedure
CPV classification: 33100000 | 39711120 | 42931100
Submission deadline: 2019-05-13
Award criteria: Lowest price

Technical requirements: Digestor for working with strong acids and alkalis, air flow 0,3- 0,8 m/s, microprocessor control with LCD display, noise at work less than or equal to 68db, body made of white porcelain resistant to strong acids and alkalies with thickness not less than 8 mm, cabinets with compartments and minimum 3 protective covers in the lower part of the digester mounted on a stand, a built-in motor with speed regulation, a front window made of reinforced glass with a thickness of at least 5 mm acid-resistant and alkalis that moves in the direction of the top- down, with a working chamber in the foot has a ceramic plate with internal dimensions: width 1 280-1 380 mm, depth 570-670 mm and height 700-900 mm, with external dimensions: width 1540-1590 mm, depth 800-840 mm, height 2 000-2 200 mm (with hose connection) interior gloss lighting with total power of all lamps of at least 60w, safe anti-corrosive water tap, remote control for gas and water drain control, minimum 2 integrated waterproof voltage outlets, operating at 220 V AC mains with memory function in accidental cessation of power supply including mounting and drainage hose connection or channel in the existing evaporative drainage system, with materials resistant to strong acids and chemical evaporation.
Set of transport saw the autopsy of batteries placed in easily transportable shipping container, which operates exclusively on battery with a capacity of nickel metal hybrid battery of 2600mAh and a minimum voltage of 10,8Vdc with speed oscilacijaod 16000cpm, include accompanying charger of batteries that are connected to a 220V / 50Hz mains voltage, contain 2 spare batteries and 3 spare blades with dimensions of 64mm, spanner, warranty period of 2 years.
Laboratory biological microscope for observation of native preparations prepared with nafrax medium, infinitive optical system with 60 mm parfomal distance, high luminoscent white LED lighting with life span of minimum 60 000 working hours, Sidentof binocular with an incline of minimum 30? and a minimum interupillary distance of 47-74 mm, two eyepieces with an increase of 10 times with dioptric compensation and a 20 mm field of view, a refocusratraal plate for a preparation with a non-sterile glass stand, a polarization set with a phase condenser with a numerical aperture of 1.25, Infinitive CFI E Plan lenses with 4.10 and 40 magnification, 5-year warranty.
Microscope for practical instruction, with built-in keyless entry controls, adjustable motion resistance with a minimum setting of 2.5 micrometers, 4-lens mounting lens, 30 cm binocular tube with 30 degree bending, adjustable interpupillary distance from 48-75mm, one-way dioptre adjustment, wide angle eyepieces (2 pcs.) 10x / FN 20, mechanical table for right-hand preparations for moving the chair and adjustment of abbey abbe condenser with NA 1,25 or equivalent, built-in LED transmission light, volume Design: Akhromatic Plan 4x, NA 0.10, working distance 27mm, Plan Akhromatic 10x, AT 0.25, working distance 8mm, Plan Akhromatic 40x, NA 0.65 working distance 0.6mm, correction of glass window 0.17, protection spray preparation from breaking. All lenses and eyepieces have protection against mold
Laboratory freezer with temperature range from -18 to -22 degrees Celsius; with a gross volume of minimum 600 liters to a maximum of 700 liters; built-in electronic regulator with automatic defrosting and display (display) of temperature, including 4 shelves made by Inox with a minimum one year warranty
A mini incubator with a temperature range of + 5 C to a minimum of + 60 C and a volume of 9 to 10 liters, with a minimum warranty of 1 year
Equipment for seeding with the suspension of microorganisms on the surface of media spread in petri plates,
Equipment for seeding with the suspension of microorganisms on the surface of media spread in petri plates, with rotation of the Petri plate; semi-automated inoculation of petri plates with a diameter of 90mm (+/- 1 %); minimum three adapters for the size of a petri dish; with the option of selecting the rotary speed with an external potentiometer of minimum 0 to 500rpm; with a temperature range of 15 to 35
A probe for intraoperative radioactive measurement with a curved tip and a body thickness of 16 mm maximum, with a radial sensitivity distribution at a distance of 30 cm maximum 400 (FWHM-400) with a spatial resolution (distribution of lateral sensitivity) maximum 25 mm with a sensitivity of probe minimum 5000 cps / MBq, with probe protection minimum 0.1 % of the system's maximum sensitivity, with good energy resolution with the possibility of complete / photocopier discrimination and possibility of checking the selected energy, with the possibility to work with low, medium and high energy isotopes Tc-99m, In-111 and I-131, with a clear correlation between the acoustic signal and the measurement signal, with a digital and / or analogue display that can be clearly read at a minimum of 2 meter, with continuous measurement, as well as adjustment of the measurement time, with power supply via mains and / or rechargeable batteries and a cable for connection to the electric network, minimum 2 meters long, with a minimum warranty period of one year, with power supply from 210V to 240V and frequency from 50Hz to 60Hz
Observation device - a sagittal section of the head and neck with a preserved parotid gland, an infematemporal pit and a pit content, made of silicone rubber by means of a plate of a pre-screened sagi
Observation device - a sagittal section of the head and neck with a preserved parotid gland, an infematemporal pit and a pit content, made of silicone rubber by means of a plate of a pre-screened sagittal cross-section of the head and neck of human cadaver
A device for disinfecting air by generating an extremely low ozone (<0.05 ppm) and ions reacting as catalytic substances in cartridges
A device for disinfecting air by generating an extremely low ozone (<0.05 ppm) and ions reacting as catalytic substances in cartridges, in the reaction to produce hydroxyl radicals, a reactive oxygen that neutralizes bacteria, viruses and fungi in the air and on surfaces of the non-living environment of minimum 300 m2, not to be harmful to the people in the room, daily included, with a daily consumption of hydrogen peroxide from 1.0 g to 0.2 g, to include a multicolor LED indicator to have a bluetooth communication, to be positioned and the wall of the tank capacity of at least 90ml, include cartrides 12 months warranty period of 1 year minimum
Centrifuge with a minimum speed of 5000 rpm, Minimum centrifugal force of 2800 xg, Included rotor of minimum 12x5ml / 6ml tubes and diameter of 17 mm, Digital display for speed and centrifugal force and time, Cover indicators and Start, Programmable Microprocessor control system, Setting the speed range minimum in the range from 1000 to 5000 rpm, Minimum speed setting of only 10 rpm, Time range from 1 to 99 minutes as well as HOLD function, Speed ​​setting the slowdown and acceleration, Induction brushless motor and maintenance free, Lid lock, Casing made of stainless steel with durable Epoxy polyester paint, Minimum radius of the rotor of 101 mm, Operating temperature 5 to 35 degrees, Noise level less than 60dB, Wide range of adapters for different types of test tubes, Mass not more than 18 kg

Eligibility requirements: — statement by the economic operator that it has not been sentenced, in the last 5 years, with effective court decision for participation in criminal organisation, corruption, fraud or money laundering, — certificate that it is not subject to bankruptcy procedure issued by competent authority, — certificate that it is not subject to liquidation procedure issued by competent authority, — certificate for paid taxes, contributions and other public dues by the competent authority in the country where that economic operator is established, — certificate from the legal entities penal registry that it is not being imposed a side sanction — prohibition to participate in public competitions, public procurement contract award and public — private partnership contract award procedures, — certificate from the legal entities penal registry that it is not being imposed a side sanction — temporary or permanent prohibition to perform certain professional activity, — certificate that it is not being sanctioned for misdemeanor with an effective court decision, resulting in prohibition for pursuing professional activity or duty, i.e. temporary prohibition for performing certain business activity. Ability of the economic operator to perform business activity: — document for registered business activity, — certificate for wholesale of medical devices in accordance with the Law on medicines and medical devices for items 7763; 8846; 8978; 9231; 9232;9233; 9259; 9304; 9338 (see tender documentation)