Public procurement notice: Contract Notice
Published: 2018-12-01
Buyer: University Clinic of Nephrology Skopje
Country: MK
Contract type: Supplies
Procedure: Open procedure
CPV classification: 33000000 | 33600000: Pharmaceutical products
Estimated value:  MKD
Submission deadline: 2019-01-03
Award criteria: Lowest price

Technical requirements: Lääketieteelliset laitteet, farmaseuttiset valmisteet ja henkilökohtaiseen hygieniaan liittyvät tuotteet
KUF 7.5 mL/(h•mm Hg) (Ultrafiltration coefficient grater/equal to 7.5; expressed in mL/(h•mm Hg), Urea clearance <FT TYPE="SUP">3</FT> 160 mL/min (Urea clearance grater/equal to 160; expressed in mL/min) Dialyzer to be sterilized with water vapor or radiation (must not be sterilized with ethylene oxide).
KUF ≥ 8 mL/(h•mm Hg); (Ultra-cavitation coefficient grater/equal to 8; expressed in mL/(h•mm Hg)), Urea clearance ≥ 170mL/min (Urea clearance greater/equal to 170; expressed in mL/min) The dialyzer must be sterilized with water vapor or radiation (must not be sterilized with ethylene oxide).
KUF ≥ 30 mL/(h•mm Hg); (Ultra-cavitation coefficient grater/equal to 30; expressed in mL/(h•mm Hg)), Urea clearance ≥ 220 mL/min (Urea clearance greater/equal to 220; expressed in mL/min), КоА ≥ 600 (Dialyzer mass transfer-area coefficient for urea greater/equal to 600) The dialyzer must be sterilized with water vapor or radiation (must not be sterilized with ethylene oxide).
KUF ≥ 30 mL/(h•mm Hg); (Ultra-cavitation coefficient greater/equal to 30; expressed in mL/(h•mm Hg)), Urea clearance ≥ 220 ml/min. (Urea clearance greater/equal to 220 expressed in mL/min.), КоА ≥ 600 (Dialyzer mass transfer-area coefficient for urea grater/equal to 600) The dialyzer must be sterilized with water vapor or radiation (must not be sterilized with ethylene oxide).
KUF ≥ 40 mL/(h•mm Hg); (Ultra-cavitation coefficient greater/equal to 40; expressed in mL/(h•mm Hg)), Urea clearance ≥ 230 ml/min. (Urea clearance greater/equal to 230 expressed in mL/min.), КоА ≥ 700 (Dialyzer mass transfer-area coefficient for urea grater/equal to 700) The dialyzer must be sterilized with water vapor or radiation (must not be sterilized with ethylene oxide).
KUF ≥ 50 mL/(h•mm Hg); (Ultra-cavitation coefficient greater/equal to 50; expressed in mL/(h•mm Hg)), Urea clearance ≥ 240 ml/min. (Urea clearance greater/equal to 240 expressed in mL/min.), КоА ≥ 800 (Dialyzer mass transfer-area coefficient for urea grater/equal to 800) The dialyzer must be sterilized with water vapor radiation (must not be sterilized with ethylene oxide).
The AV system should have an artery and vein chamber and lines for measuring artery and vein pressure compatible (in accordance with the technical characteristics of) with dialyzers in 1. The lines should have a volume of 78 mL/required for children weighing 7.5 to 35 kg. The AV system should be sterilized with water vapor or radiation (must not be sterilized with ethylene oxide).
The AV system should have an artery and vein chamber and lines for measuring artery and vein pressure. The lines should have a volume of 170 mL/required for adult populations. The AV system should be sterilized with water vapor or radiation (must not be sterilized with ethylene oxide).
The hemodialysis needles with opening must contain a sharp metallic, stainless steel piece, non-pyrogenic with a length of ¾“ to 1¼“ (0.75-1.25 cm). The needles must be sterilized with water vapor or radiation (must not be sterilized with ethylene oxide).
The hemodialysis needles without an opening must contain a sharp metallic, stainless steel piece, non-pyrogenic with a length of ¾“ to 1¼“ (0.75-1.25 cm). The needles must be sterilized with water vapor or radiation (must not be sterilized with ethylene oxide).
Catheter introducer splitting in two branches after the catheter is inserted (size according to the ordered catheter - lot no. 21 and lot no. 22); non-pyrogenic plastic, sterile.
Artery and vein extension for permanent catheters for HD (to fit the ordered permanent catheter - lot 21 and lot 22); non-pyrogenic plastic, sterile.

Eligibility requirements: To prove their personal competence, economic operators are to submit the following documents: — statement by the economic operator that it has not been imposed a legally valid court decision for participation in criminal organization, corruption, fraud or money laundering in the last 5 years, — certificate by a competent body that a bankruptcy procedure has not been initiated, — certificate by a competent body that a liquidation procedure has not been initiated, — certificate by a competent body from the country where that economic operator is registered for paid taxes, contributions and other public fees, — certificate by the Register of Sentences for committed crimes of legal entities that a secondary sentence prohibition on participation in an open call procedures, awarding public procurement contracts and contracts for public private partnership, has not been imposed, — certificate by the Register of Sentences for committed crimes of legal entities that a secondary sentence temporary or permanent prohibition on performing a certain activity, has not been imposed, — certificate that a misdemeanour sanction resulting in prohibition for practicing profession, performing activity or duty, i.e. temporary prohibition for performing a particular activity has not been imposed by a legally valid court decision. To prove the ability to perform professional activity, the economic operator should submit: — A certificate of registered activity - CRA form, or proof of registered activity, — Certificate for performing wholesale of medical devices in compliance with the law on medicinal products and medical devices.