Public procurement notice: Contract Notice
Published: 2023-01-04
Buyer: University Clinic of Cardiology
Country: MK
Contract type: Supplies
Procedure: Open procedure
CPV classification: 33000000
Estimated value: 99894700 MKD
Submission deadline: 2023-01-30
Award criteria: Lowest price

Technical requirements: Lääketieteelliset laitteet, farmaseuttiset valmisteet ja henkilökohtaiseen hygieniaan liittyvät tuotteet
Cobalt Alloy Coronary Stent, continuous sinusoidal wire like stent design, for tortuous coronary arteries, coated with permanent polymer and cytostatic zotarolimus (DES), minimum diameter span of 2.25 to 4 mm and minimum span length of 8 to at least 38mm compatible with 5F guiding catheter for all dimensions.
Cobalt-planitum iridium alloy coronary stent, continuous sinusoidal wire like stent design, for large diameter coronary arteries and for tortuous lesions, coated with permanent polymer and cytostatic zotarolimus (DES), minimum diameter span of 2, up to 5 mm and minimum length span from at least 12 to 30mm, with the possibility of expansion to a larger diameter without significant shortening, for large coronary arteries, for LM, compatible with 5F guiding catheter for all dimensions.
Distal protection kit consisting of: a catheter containing a wire with a filter mesh with radio-visible markers at the distal, proximal end and the entire proximal mouth of the filter. The system should show independent movement of the wire and the possibility of longitudinal and rotational movement of the filter. Filter sizes 3-7mm, wire length 320 cm with the option of breaking the wire at 190 cm. The system is compatible with 0.014"/ 01018" guide wire and 5F introducer and is indicated for carotid, coronary and peripheral interventional procedures.
Cobalt-Chromium alloy coronary stent, coated with drug "umirolimus" (semi synthetic sirolimus derivative), with biodegradable "poly lactic acid" polymer, with stent design with 6 crowns (for small arteries from 2.25 mm to 3.0 mm) and with 9 crowns (for medium arteries from 3.5 mm to 4.0 mm), minimum range of diameter 2.25 to 4.00 mm and minimum range of length from 9-29 mm.
Cobalt-Chromium alloy coronary stent, abluminally coated with umirolimus (semi synthetic sirolimus derivative), without polymer and without carrier, with stent design with 6 crowns (for small arteries from 2.25 mm to 3.0 mm ) and with 9 crowns (for medium arteries from 3.5 mm to 4.0 mm), a minimum diameter range of 2.25 to 4.00 mm and a minimum length range of 9-29 mm
Self-expanding nitinol stent intended for the treatment of femoral artery, with a minimum diameter of 5-8 mm and a minimum length of up to 150 mm, with a three-axial stent placement system, with "close-cell" design at the ends and "open-cell" " center design for flexibility and break resistance, compatible with 0.035" guide wire
Platinum Chromium Alloy coronary stent, non-linear link open cell design, coated with permanent polymer and cytostatic everolimus (DES), span diameter from 2.25 to 4mm and length from 8 to at least 38mm compatible with 5F guiding catheter for all dimensions.
Open Cell Nonlinear Link Design, Abluminally Coated with Bioresorbable Polymer and Cytostatic Everolimus (DES), Diameter Range 2.25 to 5mm and Length 8 to 48mm Compatible with 5F Guiding Catheter for all dimensions.
Cobalt - chromium alloy coronary stent, Open cell design with biodegradable polymer, coated with immunosuppressive drug sirolimus, Thickness of "stent strut" 60 µm with PROBIO coating (amprphous silicon-carbid), Diameter 2.25 - 4mm and length 9 - 40mm. Should have a CE certificate for patients with a high risk of bleeding.
Coronary Stent Graft System, made of cobalt Chromium Alloy Balloon Expandable, designed to deal with complications during interventions, with a maximum profile of 1.25 mm, compatible with 5 French catheter for 2.5 to 4.0 mm diameter and 6 French catheter for 4 ,5 and 5.0 diameter, and lengths from 15 to 26mm.
Nitinol alloy stent for leg arteries below and above the knee, as well as for the popliteal artery, OTW system, Tri-axial precision placement system, compatible with 0.018" guidewire, and with 4F introducer introducer, Stent to be dimensioned for diameter range from 4-7 mm and for a length of 20-200 mm, carrying catheter with a length of 90 and 135 cm.
Peripheral balloon expandable stent, made of cobalt-chromium, OTW system, compatible with 6 Fr introducer, thread thickness 110 micrometers (for stents 5-7 mm diameter), with PROBIO coating, with 2 markers, on catheter carrier 90, 135 and 170 cm, and stent length 18-78 mm, diameter 5-10 mm.
Cobalt Chromium Alloy Coronary Stent, (DES) with open cells as a stent design intended for coronary arteries with bifurcations and complex lesions, Specific and selectively placed coating with bioresorbable polymer and cytostatic sirolimus on the abluminal side, The stent is with diameter span of 2, 25, up to 4 mm and length from 9 to 38 mm, compatible with 5F guiding catheter for all sizes.
Self-expanding carotid stent made of nitinol alloy, with a flexible two-layer micromesh closed cell design, for protection against embolization, and possibility of repositioning during expansion up to 50%. compatible with 5F introducer sheath Rx system. (diameter 5-10mm, length from 16 to 40 mm of the stent)


Award value: 99894700