Public procurement notice: Contract Notice
Published: 2019-02-25
Buyer: Oslo Universitetssykehus HF — Strategi og anskaffelsesavdelingen MTV
Country: NO
Contract type: Supplies
Procedure: Open procedure
CPV classification: 33191110
Submission deadline: 2019-03-27
Award criteria: The most economic tender

Technical requirements: 75082 RH OPK Orthopaedic Department Autoclave — 75167 RH HLK KIT Thorax Surgery Department of Transplant Medicine Autoclave
OUS will procure 2 and have the option to procure 1 (similar to operation 2) single door steam sterilisers (approx. 2 sterilisation devices) for sterilisation of invasive medical equipment, i.a. surgical instruments, thermostable endoscopic equipment, anaesthesia equipment and sterilisation of breast implants. Steam sterilisers must have steam generators and a water purification system for steam production. Factory testing documentation must be in place and be approved by OUS before delivery. The number of cycles is anticipated to be approx. 3-4 per day per autoclave/240 working days per year.
(L58 x B28 x H21), (L60 x B28 x H13), (L54 mm x B26 mm x H15 mm),(L60 mm x B 40 mm x H20 mm),(L70 mm x B28 mm x H20 mm) but the chamber must be dimension to and be able to run containing equipment with a width of 34,5 cm.
OUS will procure 2 and have the option to procure 1 (similar to operation 2) single door steam sterilisers (approx. 2 sterilisation devices) for sterilisation of invasive medical equipment, i.a. surgical instruments, thermostable endoscopic equipment, anaesthesia equipment and sterilisation of breast implants. Steam sterilisers must have steam generators and a water purification system for steam production. Factory testing documentation must be in place and be approved by OUS before delivery. The number of cycles is anticipated to be approx. 3-4 per day per autoclave/240 working days per year.
(L58 x B28 x H21), (L60 x B28 x H13), (L54 mm x B26 mm x H15 mm), (L60 mm x B40 mm x H20 mm), (L70 mm x B28 mm x H20 mm) but the chamber must be dimension to and be able to run containing equipment with a width of 34,5 cm.
External measurement are preferably the same for both of the surgical departments but is a challenge due to the container size.
Depth: enter the outer measurement depth (must be able to take container L70 mm x B28 mm x H20 mm, L60 mm x B28 mm x H13 mm), (L54B x B26 mm x 15 mm)
Depth: 1 020 mm (must be able to sterilise the sterilisation device (must be able to hold the sterilisation device: (L58 mm x B28 mm x H21 mm), (L62 mm x B42 mm x H20 mm)/(L60 mm x B20 mm x H20 mm)
The steam steriliser delivered must be able to be transported into the named sterilisation room without building intervention (structural works). OUS will not perform any alteration of doors/transport routes in the corridors to get the sterilisers into the sterilisation room, due to patient activity.
One extra steam steriliser at Rikshospitalet [the National Hospital] needs to be replaced and we would like a price for 1 device as an option.
1.1 Steam steriliser (single door) with integrated steam generator and RO system (for OUS) similar to operating theatre 2.
The certificates shall be issued on a certified form from the tax collector/tax office (www.skatteetaten.no). The certificates must be no more than six months from the deadline for receipt of tenders. Any arrears must be explained/justified.
Foreign tenderers must submit corresponding confirmation that government taxes and fees have been paid in the country where the tenderer has its registered place of business, in accordance with the respective country´s rules for such certificates.
Credit evaluation/rating, not older than 3 months, based on the last known accounting figures. The rating must be carried out by a credit rating company with a licence to conduct this service.
That the Contracting Authority has requested, he can document his economic and financial capacity by presenting other documents that the contracting authority deem to be appropriate. (PPR § 16-4 (2)).

Eligibility requirements: Tax and VAT certificate: The certificates shall be issued on a certified form from the tax collector/tax office (www.skatteetaten.no). The certificates must be no more than six months from the deadline for receipt of tenders. Any arrears must be explained/justified. Foreign tenderers must submit corresponding confirmation that government taxes and fees have been paid in the country where the tenderer has its registered place of business, in accordance with the respective country´s rules for such certificates.