Public procurement notice: Contract Notice
Published: 2018-03-24
Buyer: Public Health Institute University Urology Clinic, Skopje
Country: MK
Contract type: Supplies
Procedure: Open procedure
CPV classification: 33141000: Disposable medical supplies
Submission deadline: 2018-04-05
Award criteria: Lowest price

Technical requirements: Instruments and materials for open and laparascopic surgery and titanium joints for the annual requirements of the Clinics. Detailed description provided in the technical specification
electrode with a length of 19,5mm, with knife 17,8mm and possibility for foot and hand activation, blunt tip and two sided jaw opening
electrode with a length of 20mm, with knife 18mm and possibility for foot and hand activation, jaw opening 13mm, possibility for foot and hand activation and Maryland jaw type or equivalently
for open surgery with a sine-shaped, curved jaw with a length of 16.5mm and a knife length of 15mm, possibility of foot and hand automatic activation
, for open surgery with curved jaw with a length of 36mm and a knife length of 34mm, possibility of foot and hand activation
for laparoscopic surgery with electrode with length of 22mm with knife 20mm, possibility of foot and hand activation
for laparoscopic surgery with electrode with length of 18mm with knife 12mm, possibility of foot and hand activation and dolphin tip
for laparoscopic surgery with electrode with length of 19,5mm with knife 17,8mm, possibility of foot and hand activation and blunt tip and two sided jaw opening
for open surgery with electrode with length of 20mm with knife 18mm, possibility of foot and hand activation, jaw opening 13mm, hand activation and Maryland jaw type or equivalently
for laparoscopic surgery with bend Maryland tip electrode, with length of 20mm, with knife 18,5mm, with a retractable L monopolar hook which, if necessary, replaces the electrode. To have the possibility of rotating the body of the instrument 315º, as well as a handle for replacing the electrode with the monopole hook and its retraction
Endoscopic automatic clipper with 360º rotating body with diameter of 5mm and working length 33cm, pre-filled with 16 locking titanium clips and an LCD display showing the number of clips. The clips should have a mechanism for leveling the jaws to prevent their crossing. Closing the clips should be from the knives to the head. The jaw should be empty, with the possibility of removing the clip in the jaw before closing, the forehead and the sidewall of the jaw. It should have a safety mechanism for locking the handle after the blowing of all the clips.
360º rotating body with diameter of 10mm and working length 29cm, pre-filled with 20 locking titanium clips. Clip locking should be from the knives to the head. The jaw should be empty, with the possibility of removing the clip in the jaw before closing, forehead and side jaw opening. It should have a safety mechanism for locking the handle after emptying all the clips and a visible indicator.
Endoscope automatic clipper, with 360º rotating body with diameter of 10mm and working length 33cm, pre-filled with 15 locking titanium clips. Clip locking should be from the knives to the head. The jaw should empty, with the possibility of removing the clip in the jaw before closing, forehead and side jaw opening. It should ave a safety mechanism for locking the handle after emptying all the clips and a visible indicator.
with 360º rotating body with diameter of 10mm and working length 28cm pre-filled with 20 locking titanium clips. The jaw opening should be lateral, at an angle of 90º with the body of the instrument.
Automatic opening calliper for open surgery with jaw opening and jaw angle of 30º and with automatic clip closing mechanism, with length of 11.5 inches, pre-filled with 30 locking titanium clips, with rectangular cross-sectional shape 0,47mm x 0.76mm and span of the jaw of 3.5-4mm.

Eligibility requirements: DRD form of form for registered activity, decision on wholesales of medical devices in accordance to the Law on Medicines and medical devices Decision on registering a medical device in the Register of medical devices available in RM issued by an authorised body in RM Approval for putting a medical device into circulation (except for item 50 and 51)
Decision on registering a medical device in the Register of medical devices available in RM issued by an authorised body in RM Approval for putting a medical device into circulation (except for item 50 and 51)
Decision on registering a medical device in the Register of medical devices available in RM issued by an authorised body in RM
Approval for putting a medical device into circulation (except for item 50 and 51)