Task: Intent Identification
Input (English): After the introduction of the Biotronik Linox S/SD high-voltage lead, several cases of early failure have been observed.', 'The purpose of this article was to assess the performance of the Linox S/SD lead in comparison to 2 other contemporary leads.', 'We used the prospective Erasmus MC ICD registry to identify all implanted Linox S/SD (n = 408), Durata (St. Jude Medical, model 7122) (n = 340), and Endotak Reliance (Boston Scientific, models 0155, 0138, and 0158) (n = 343) leads. Lead failure was defined by low- or high-voltage impedance, failure to capture, sense or defibrillate, or the presence of nonphysiological signals not due to external interference.', 'During a median follow-up of 5.1 years, 24 Linox (5.9%), 5 Endotak (1.5%), and 5 Durata (1.5%) leads failed. At 5-year follow-up, the cumulative failure rate of Linox leads (6.4%) was higher than that of Endotak (0.4%; P < .0001) and Durata (2.0%; P = .003) leads. The incidence rate was higher in Linox leads (1.3 per 100 patient-years) than in Endotak and Durata leads (0.2 and 0.3 per 100 patient-years, respectively; P < .001). A log-log analysis of the cumulative hazard for Linox leads functioning at 3-year follow-up revealed a stable failure rate of 3% per year. The majority of failures consisted of noise (62.5%) and abnormal impedance (33.3%).
Output (English): 1