Task: Question Answering
Input (English): Context: Little is known about causes of intimal hyperplasia (IH) after sirolimus-eluting stent (SES) implantation.', 'Intravascular ultrasound was performed in 24 lesions with intra-SES restenosis and a comparison group of 25 nonrestenotic SESs. To assess stent strut distribution, the maximum interstrut angle was measured with a protractor centered on the stent, and the visible struts were counted and normalized for the number of stent cells. In SES restenosis patients, minimum lumen site was compared with image slices 2.5, 5.0, 7.5, and 10.0 mm proximal and distal to this site. The minimum lumen site had a smaller IVUS lumen area at follow-up (2.7+/-0.9 versus 6.2+/-1.9 mm2; P<0.01), larger maximum interstrut angle (135+/-39 degrees versus 72+/-23 degrees; P<0.01), larger IH area (3.4+/-1.5 versus 0.6+/-1.1 mm2; P<0.01) and thickness (0.7+/-0.3 versus 0.1+/-0.2 mm; P<0.01) at maximum interstrut angle, and fewer stent struts (4.9+/-1.0 versus 6.0+/-0.5; P<0.01) even when normalized for the number of stent cells (0.78+/-0.15 versus 0.97+/-0.07; P<0.01). Compared with nonrestenotic SES, the restenosis lesions also had a smaller minimal lumen area, larger IH area, thicker IH at maximum interstrut angle, fewer stent struts, and larger maximum interstrut angle. Multivariate analysis identified the number of visualized stent struts normalized for the number of stent cells and maximum interstrut angle as the only independent IVUS predictor of IH cross-sectional area (P<0.01 and P<0.01), minimum lumen area (P<0.01 and P<0.01), and IH thickness (P<0.01 and P<0.01).\Question: Does nonuniform strut distribution correlate with more neointimal hyperplasia after sirolimus-eluting stent implantation?
Output (English): The number and distribution of stent struts affect the amount of neointima after SES implantation.