Package: pysnr
PySNRPySNR is a Python library which provides a suite of tools for performing various types of noise analysis on  signals.
PySNR aims to provide five main functionalities:SNR (Signal to Noise Ratio)THD (Total Harmonic Distortion)SINAD (Signal to Noise and Distortion Ratio)TOI (Third Order Intercept)SFDR (Spurious Free Dynamic Range)The following sections elaborate on each of these utilities further.Signal to Noise Ratio (SNR)This calculates the signal-to-noise ratio of an input signal. Signal-to-Noise ratio is defined as the ratio
between the power of the desired signal to the power of the background noise.If the noise magnitude is provided along with the signal, the SNR is calculated using the following formula:
$$20 \log_{10}\left({\frac{\sqrt{\sum signal^2}}{\sqrt{\sum noise^2}}}\right)$$If the noise magnitude is not provided, a modified periodogram is computed using a Kaiser window with $\beta = 38$
and the SNR is computed using the power of fundamental frequency and the power of the signal after removing the top 6
harmonics. The formula used in this case is:
$$10 \log_{10}\left({\frac{\sqrt{\sum signal^2}}{\sqrt{\sum noise^2}}}\right)$$The SNR value can also be computed by providing the periodograms ofpower spectral density$(V^{2}/Hz)$ orpower spectrum$(V^{2})$. In case of power spectrum periodograms,the resolution bandwidth needs to be provided
as well. Utilities provide theenbw()function which computes the estimated noise bandwidth for assessing
the resolution bandwidth.Total Harmonic Distortion (THD)This calculates the total harmonic distortion of a signal. Total harmonic distortion is defined as the ratio
of the power of the harmonics to the power of the fundamental frequency.A modified periodogram is computed using a Kaiser window with $\beta = 38$ and this information is then used to
determine the fundamental frequency and its harmonics. The formula used for computing the THD is:
$$\frac{\sqrt{V^2_{H_1} + V^2_{H_2} + V^2_{H_3} + ...}}{V_{H_0}}$$The THD value can also be computed by providing the periodograms ofpower spectral density$(V^{2}/Hz)$ orpower spectrum$(V^{2})$. In case of power spectrum periodograms, the resolution bandwidth needs to be provided
as well. Utilities provide theenbw()function which computes the estimated noise bandwidth for assessing
the resolution bandwidth.Signal to Noise and Distortion Ratio (SINAD)This calculates the signal-to-noise-and-distortion ratio of a signal. SINAD is defined as the ratio between the
power of the signal's fundamental frequency to the power of the background noise and harmonics.A modified periodogram is computed using a Kaiser window with $\beta = 38$ and this information is then used to
determine the fundamental frequency and its harmonics. The formula used for computing the SINAD is:
$$\frac{P_{fundamental}}{P_{noise} + P_{harmonics}}$$The SINAD value can also be computed by providing the periodograms ofpower spectral density$(V^{2}/Hz)$ orpower spectrum$(V^{2})$. In case of power spectrum periodograms, the resolution bandwidth needs to be provided
as well. Utilities provide theenbw()function which computes the estimated noise bandwidth for assessing
the resolution bandwidth.Third Order Intercept (TOI)This calculates the third order intercept point for a signal. There are six third-order intermodulation points from the
top two dominant frequencies $F_1$ and $F_2$ (fundamental signals) — $3F_1$, $3F_2$, $2F_1 + F_2$, $2F_2 + F_1$,
$2F_1 - F_2$ and $2F_2 - F_1$. Amongst these, the hardest to handle are $2F_1 - F_2$ and $2F_2 - F_1$ because of how c
lose they are to the fundamental signals. TOI helps us compute the point at which the power of the third order products
intercepts the power of the fundamental signals. In real world devices, this does not happen because the output power is
limited; hence, the TOI is a theoretical value. TOI helps evaluate the linearity of the signal source. The higher the
TOI, the better the linearity with lower levels of intermodulation distortion.A modified periodogram is computed using a Kaiser window with $\beta = 38$ and this information is then used to identify
the two dominant frequencies $F_1$ and $F_2$ which is considered the fundamental signal. The formula used for computing
the TOI is:
$$\overline{P_{fundamental}} + \frac{\overline{P_{fundamental}} - \overline{P_{intermodulation}}}{2}$$The TOI value can also be computed by providing the periodograms ofpower spectral density$(V^{2}/Hz)$ orpower spectrum$(V^{2})$. In case of power spectrum periodograms, the resolution bandwidth needs to be provided
as well. Utilities provide theenbw()function which computes the estimated noise bandwidth for assessing
the resolution bandwidth.Spurious Free Dynamic Range (SFDR)This calculates the spurious free dynamic range of a signal. SFDR is defined as the strength ratio of the fundamental
signal to the strongest spurious signal.A modified periodogram is computed using a Kaiser window with $\beta = 38$ and this information is then used to
determine the fundamental frequency and its harmonics. The formula used for computing the SFDR is:
$$10 \log_{10}\left({\frac{P_{fundamental}}{P_{spurious}}}\right)$$The SFDR value can also be computed by providing the periodograms ofpower spectral density$(V^{2}/Hz)$ orpower spectrum$(V^{2})$. In case of power spectrum periodograms, the resolution bandwidth needs to be provided
as well. Utilities provide theenbw()function which computes the estimated noise bandwidth for assessing
the resolution bandwidth.