McLeod gauge | Wikipedia audio article
Channel: wikipedia tts
Date: 04/10/2019

This is an audio version of the Wikipedia Article:
https://en.wikipedia.org/wiki/McLeod_gauge


00:02:30 See also



Listening is a more natural way of learning, when compared to reading. Written language only began at around 3200 BC, but spoken language has existed long ago.

Learning by listening is a great way to:
- increases imagination and understanding
- improves your listening skills
- improves your own spoken accent
- learn while on the move
- reduce eye strain

Now learn the vast amount of general knowledge available on Wikipedia through audio (audio article). You could even learn subconsciously by playing the audio while you are sleeping! If you are planning to listen a lot, you could try using a bone conduction headphone, or a standard speaker instead of an earphone.

Listen on Google Assistant through Extra Audio:
https://assistant.google.com/services/invoke/uid/0000001a130b3f91
Other Wikipedia audio articles at:
https://www.youtube.com/results?search_query=wikipedia+tts
Upload your own Wikipedia articles through:
https://github.com/nodef/wikipedia-tts
Speaking Rate: 0.9490742745709871
Voice name: en-US-Wavenet-A


"I cannot teach anybody anything, I can only make them think."
- Socrates


SUMMARY
=======
A McLeod gauge is a scientific instrument used to measure very low pressures, down to 10−6 Torr. It was invented in 1874 by Herbert McLeod (1841–1923). McLeod gauges were once commonly found attached to equipment that operates under vacuum, such as a lyophilizer. Today, however, these gauges have largely been replaced by electronic vacuum gauges.
The design of a McLeod gauge is somewhat similar to that of a mercury-column manometer. Typically it is filled with mercury. If used incorrectly, this mercury can escape and contaminate the vacuum system attached to the gauge.

McLeod gauges operate by taking in a sample volume of gas from a vacuum chamber, then compressing it by tilting and infilling with mercury. The pressure in this smaller volume is then measured by a mercury manometer, and knowing the compression ratio (the ratio of the initial and final volumes), the pressure of the original vacuum can be determined by applying Boyle's law.
This method is fairly accurate for non-condensible gases, such as oxygen and nitrogen. However, condensible gases, such as water vapour, ammonia, carbon dioxide, and pump-oil vapors may be in gaseous form in the low pressure of the vacuum chamber, but will condense when compressed by the McLeod gauge. The result is an erroneous reading, showing a pressure much lower than actually present. A cold trap may be used in conjunction with a McLeod gauge to condense these vapors before they enter the gauge.
The McLeod gauge has the advantage that it is simple to use and that its calibration is nearly the same for all non-condensable gases.
The device can be manually operated and the scale read visually, or the process can be automated in various ways. For example, a small electric motor can periodically rotate the assembly to collect a gas sample. If a fine platinum wire is in the capillary tube, its resistance indicates the height of the mercury column around it.
Modern electronic vacuum gauges are simpler to use, less fragile, and do not present a mercury hazard, but their reading is highly dependent on the chemical nature of the gas being measured, and their calibration is unstable. For this reason, McLeod gauges continue to be used as a calibration standard for electronic gauges.