SAXSess Alignment
Channel: drheaddamage
Date: 2016-06-07

Alignment procedure for the SAXSess II instrument. 

Note 0: You can insert the 0.01 mm Ni filter during the mirror alignment procedure to remove most of the Cu Kb radiation, avoiding the pitfall of aligning to the wrong wavelength. 

Note 1: After tilting the mirror to select the Ka reflection, you can adjust the height of the entrance slit once more to remove the remaining unreflected radiation. This will simplify the adjustment of the collimation. 

Note 2: After aligning the sample capillary, remove the capillary for the alignment of the beamstop. Remember to record the beam profile without sample holder and without beam stop in place. 

This video has not been sponsored or endorsed by Anton Paar, and should also not be seen as a sponsoring or endorsement of Anton Paar. It is what it is.