Gas Clip Technologies Multi Gas Clip - IR Calibration Guide
Channel: Ideal Calibrations
Date: 2021-02-01

Welcome to the Ideal Calibrations how to calibrate your gas detector series. Today we're looking at the Gas Clip Technologies Multi Gas Clip - IR unit. A cool unit with an IR LEL sensor which behaves a bit differently than a standard pellistor or catalytic bead sensor. I got over the differences, but keep in mind this LEL sensor will not pick up Acetylene or Hydrogen, which you'll need to use the CO sensor to monitor for.

How to Calibrate:
1. Turn the unit on and wait for standard reading mode.
2. Check your gas and screw it your regulator with the valve open (shut the valve on the regulator once gas begins to flow). Attach cal adapter to the regulator.
3. Press and hold the power button, waiting for the countdown to end, the monitor to go silent, and then for the screen to count down again.
4. When the monitor says AUTO ZERO release the button.
5. When screen says APPLY GAS clip on your calibration adapter with the single port facing down
6. Turn the gas valve to on on the regulator
7. Wait for the calibration to finish. When it complete, turn off the gas and disconnect the calibration adapter.
8. You're all set!

Leave a comment or call me if you have any questions, happy to help. Please remember to Like the video and Subscribe if this helped you out!

Buy Calibration Gas:
https://idealcalibrations.com/calibration-gas-blend-0063/

Buy a Regulator:
https://idealcalibrations.com/0-5-lpm-fixed-flow-c10-regulator/

Buy a replacement Oxygen Sensor here:
https://idealcalibrations.com/oxygen-sensor/

If you have any gas monitor or calibration gas questions, please feel free to call us at 1-734-956-0539 or to email me support@idealcalibrations.com.  We'd be happy to answer any questions you have.

Please see our websites at:
http://www.idealcalibrations.com