Honeywell BW Solo Ammonia (NH3) Calibration and Bump
Channel: Ideal Calibrations
Date: 2021-02-16

Hey everyone, today we're looking at the Honeywell BW Solo Ammonia (NH3) unit and going through calibration and bump tests. 

If you have a problem during your calibration, give us a call or send it in for repair here:
https://idealcalibrations.com/gas-detector-service/

Calibration Steps:
1. Power unit on by holding button
2. When in standard reading mode, double tap button to get into menu
3. Press button once to advance menus. Press button enough times to get to Calibrate
4. Press and hold to get into calibration mode.
5. Confirm gas blend is 50ppm Ammonia (NH3). Press and hold.
6. Wait for zero to complete.
7. Screen will ask APPLY GAS? Press and hold button to advance
8. Quickly attach calibration adapter and turn gas on
9. When calibration completes, unit will ask to remove gas. Remove calibration adapter and turn gas off.
10. Your unit is calibrated!

Calibration Gas:
50ppm Ammonia (NH3)
Nitrogen balance

PN: 58DAL-0021-24
https://idealcalibrations.com/ammonia-nh3-calibration-gas/?SKU=58RAL-0021-24

Regulator:
1.0lpm Stainless Steel Fixed Flow

PN: SS-100-C10
https://www.idealcalibrations.com/regulators/fixed-flow-c10-regulators-for-34l-58l-116l-aluminum-cylinders-and-103l-steel-cylinders

If you have any gas monitor or calibration gas questions, please feel free to call us at 1-734-956-0539 or to email me support@idealcalibrations.com.  We'd be happy to answer any questions you have.

Please see our websites at:
http://www.idealcalibrations.com