ABOUT COMPANY

INTRODUCTION
Company name    GASTRON Co., Ltd. President Mr. Choi, Dong-Jin Established 1992. 6. 26 Head office 23, Gunpocheomdansaneop 1-ro, Gunpo-si, Gyeonggi-do, 15881, Rep. of KOREA Tel. +82-31-490-0800   Fax. +82-31-490-0801 Web-site : www.gastron.com E-mail : info@gastron.com Busan branch      55, Gonghangap-gil 85beon-gil, Gangseo-gu, Busan, Korea Industrial Fixed Gas Detector Monitoring System Engineering (HMI & PLC) Colorimetric Gas Detector Open Path Gas Detector Portable Gas Detector Vision Gas Camera & Thermal Imaging Camera Alarm Signalling / Manual Call Point Flame Detector Acid Leak Detection Paint Safety Shield Prodcut 2002   Promotion of Excellent Venture Business (Gyeonggi Provincial Small and Medium Business Administration) Encouragement Prize for 2005 Safety Equipment Test (Korea Occupational Safety & Health Agency) 2006 Selected as a superior technology company (Technology guarantee fund) 2008 Commendation of industrial development of measuring instruments (Ministry of Commerce, Industry and Energy) Designated as a promising export small and medium business by Gyeonggi Small and Medium Business Administration 2013   Sincere Taxpayer Certification (Ansan mayor) 2014 Commendation to fulfill tax payment duty (Minister of Strategy and Finance) Awards Patent details      2007   Gas leakage alarm control system using touch screen Infrared flammable gas detector Gas Leak Detectors Regulate Output Current Constantly 2008 Contacted combustion gas sensor for flammable gas leak detector Receiver for gas leak detector with battery charging circuit 2009 Diffusion Infrared Gas Sensor Module Inhalation Infrared Gas Sensor Module 2014   Hot Line Semiconductor Gas Sensor 2017 Gas Detecting Apparatus Having Explosion-proof Sensor Cartridge using Air Sampling 2018 Infrared sensor to prevent alarm by interference gas Gas sensing apparatus and method
HISTORY
1990 1992 Established Hwaseong Instrument Co., Ltd. (CEO: Chan Young Song)
1997
Awarded by the Minister of Commerce, Industry and Energy Registered as a member of Korea Instrument Research Association 1999 Acquired IS9001 certification
2000
Changed company name to Gastron Co., Ltd. Won the Excellent Explosion Proof Product Selection Contest (Korea Occupational Safety & Health Agency)
2001
Utility Model Registration: Diffusion Gas Leak Detector (No.0258970) Utility model registration: Inhalation type gas leak detector (No. 0258971)
2000
2002
Utility model registration: Gas leakage detection device (No. 0283456) Completion of new building completion and transfer (75-10 Bugok-dong, Sang roku-gu, Ansan-si, Gyeonggi-do)
2004
Selected as a promising small business (Gyeonggi-do)
2005
Established a branch office in Malaysia Establishment of affiliated research institute
2006
Selected as a venture company (Gyeonggi Provincial Small and Medium Business Administration) Selected as a technologically innovative small and medium enterprise Selected as a superior technology company (Technology guarantee fund)
2007
Selected as export promising small and medium enterprise (Gyeonggi Small and Medium Enterprise Export Support Center) Established Ulsan branch
2008
Awarded by Minister of Commerce, Industry and Energy
2009
Acquired European explosion-proof certification (Ex d IIC Gb T5) ATEX, IECEx
2010
2018
! ! ! 2010 Establishment of ERP system
2011
Information support project (product information management system construction)
2012
Design Registration: Portable Gas Detector GPD-100 (12-22935GH) Design Registration: Installable Gas Detector Outdoor Cover (12-30412)
2013
Registered as a member of HART COMMUNICATION PROTOCOL Expansion of sales department / development department office
2014
Obtained SIL 2 certification Established Yeongnam branch
2015
Developed Multi Sensor Detector (GTM-1000, 2000)
2016
Acquired ISO14001 and OHSAS18001 certification
2017
Acquired Russian explosion proof certification Completion and relocation of new building
2018
Selected as a promising export small and medium enterprise (Gyeonggi Provincial Small Venture Enterprise Division) Recognition of Gyeonggi Export Excellence (The director of the Gyeonggi Provincial Small and Medium Venture Enterprise) Acquired Flame Detector fire type approval Acquired FM Certification Selected as a youth-friendly enterprise (Ministry of Employment and Labor) GASTRON PRODUCTS Gas Sampling & Detection System Gas Sampling Monitoring System GTD-5000 / GTD-5100F / GIR-3000 GTD-5000F / GIR-3000A GTD-5100F / GTD-5000F / GTD-3000Ex GTD-2000Ex / GTD-1000Ex / GTD-5000 GTD-6000 GTD-5100F / GTD-5000F / GTD-3000Tx GTD-2000Tx / GTD-1000Tx / GTD-5000 GTD-6000 SafEye Quasar 900 ASC-100 / GTC-200A Series / GTC-200F Series GTC-500 Series / GTC-520F Series GTC-700 Series / GMS-2000 DOD64 FTIR / ChemLogic 96 / ChemLogic 8 CLPX G-Finder Single / GPD-100 / TANGO TX1 GasBadge Pro / VENTIS MX4 / VENTIS PRO MX6 iBrid / RADIUS BZ1 / iNet® Now GTF-1100U / GTF-1100T 975MR / 975HR / 975UF / 975UR Incus Ultrasonic Gas Leak Detector GTL-200 / BExBG Series / MB005 BExS Series / MA112 / BExCP3A / BExCP3B Series Can Gas / Cylinder  / Acid Leak Detection paint Safety Shield Configuration-1 / Configuration-2 Configuration-3 / Configuration-4 Configuration-5
GTM-1000 Fixed Type Freon, Flammable MULTI Gas Detector GTM-2000 Fixed Type Freon, Flammable, Oxygen & Toxic MULTI Gas Detector GASTRON
GTM-1000  Fixed Type Freon, Flammable MULTI Gas Detector
| Detect Freon, Flammable, VOC & Toxic gases simultaneously  | Apply algorithm to prevent malfunction for interference gas (PA, Ethanol, FC3283 and Alcohol types) | Fast response of gas leakage with a high efficient pump inside  | Detection gas from low to high concentration owing to various detection principle  | Stable quality and proved detection principle for service life  | Built in Data & Event Logging  | Parameter setting with Color graphic LCD  | Self diagnosis function  | Flow Sensor to control flow rate automatically  | 2-stage alarm, 1-trouble display and relay contact output  | 4-20mA output signal  | Easy configuration of monitoring by RS-485 Modbus (Isolated) and PoE communication(option)  | Calibration and maintenance by one-man
IMAGE
ACCESSORIES
SPECIFICATION

GTM-2000  Fixed Type Freon, Flammable, Oxygen & Toxic MULTI Gas Detector
|  Detect Freon, Flammable, Oxygen & Toxic gases simultaneously  |  Apply algorithm to prevent malfunction for interference gas (PA, Ethanol, FC3283 and Alcohol types)  | Fast response of gas leakage with a high efficient pump inside  | Detection gas from low to high concentration owing to various detection principle  | Stable quality and proved detection principle for service life   |  Built in Data & Event Logging  |  Parameter setting with Color graphic LCD   |  Self diagnosis function  |  Flow Sensor to control flow rate automatically  | 2-stage alarm, 1-trouble display and relay contact output  | 4-20mA output signal  | Easy configuration of monitoring by RS-485 Modbus (Isolated) and PoE communication(option)  | Calibration and maintenance by one-man IMAGE ACCESSORIES
Fixed Gas Sampling & Detection System LNG Carrier Crude Oil Tanker, Product Carrier, VLCC, Chemical Tanker & etc. CYSTRON
╔ Fixed Gas Sampling & Detection System  LNG Carrier
| Compliance with revised MSC & IMO rules  | Easy to control  | Application of Pre-sampling function  | Loop based system  | 1st & 2nd alarm display  | Sampling time setup | High resolution color display | Both analog and digital output signal (Loop) can be configured between gas detection panel and local gas detectors. · Repeat alarm unit has many functions like dimmer switch, alarm test, Modbus function, volume control and configuration mode. · Separation of electric devices and sampling equipment prevents explosion from sparks caused by electric devices in case of gas leakage inside the panel. · Sampling time (1 Cycle) Within 30 minutes Pump Capacity is 20L / MIN and the highest level in the market. Increased visibility and usability through 15 inch touch Screen and color display – Easy to use and edit the lay-out of the system. · The values for the measured gas name, gas concentration, First / Second alarm, FAULT alarm and FLOW alarm for all sampling points can be grasped at a glance. Gastron is a manufacturer producing from the sensor to the housing of gas detectors. 80% components of the system are manufactured at its facility. In house manufacturing enables the system to be more competitive in price and prompt service to customers. Data logging function enables to record all alarm events and some or all data can be deleted with the delete function.
SPECIFICATION

|=  Fixed Gas Sampling System  Crude Oil Tanker, Product Carrier, VLCC, Chemical Tanker & etc
| Compliance with revised MSC & IMO rules  | Easy to control  | Application of Pre-sampling function  | Loop based system  | 1st & 2nd alarm display  | Sampling time setup | High Resolution color display | Oxygen and hydrogen sulfide sensors | GAS DETECTOR
RECEIVER'S FEATURE

SPECIFICATION

GTD-5100F Explosion Proof Type Sampling VOC Gas Detector GTD-5000F Explosion Proof Type Sampling VOC Gas Detector GTD-2000Tx Explosion Proof Type Diffusion VOC Gas Detector GTD-5000 Intelligent & Sampling Type VOC Gas Detector CASTRON
GTD-5100F Explosion Proof Type Sampling VOC Gas Detector
|  Continuous detection of VDC gas in explosion hazardous area with flame proof structure  | Self diagnosis function with built-in Microprocessor  |  Easy replacement with a cartridge type sensor  | Digital flow sensor to maintain regular flow rate  | Fast response of gas leakage with a high efficient purp inside  | Low concentration VOC gas detection by PID sensor | Stable quality and proved detection principle for service life | Auto calibration and program setting with magnet bar on FND display board  |  4-Digit digital gas concentration display  |  4-20mA (Current source or Sink) output signal  |  Easy configuration of monitoring by RS-485 Modbus  | 3-Wire Type  | 2-stage alarm, 1-trouble display and relay contact output  | Calibration and maintenance by one-man  |
SPECIFICATION

GTD-5000F  Explosion Proof Type Sampling VOC Gas Detector
|  Continuous detection of VOC gas in explosion hazardous area with flame proof structure  |  Self diagnosis function with built-in Microprocessor  |  Easy replacement with a cartridge type sensor  |  Digital flow sensor to maintain regular flow rate  |  Fast response of gas leakage with a high efficient pump inside  | Low concentration VOC gas detection by PID sensor | Stable quality and proved detection principle for service life | Auto calibration and program setting with a magnet bar | 4-digit digital gas concentration display  | 4-20mA (Current source or Sink) output signal  | Easy configuration of monitoring by RS-485 Modbus  | 3-wire connection for signal & power  | 2-stage alarm, 1-trouble display and relay contact output  | Calibration and maintenance by one-man  |

GTD-2000TX  Explosion Proof Type Diffusion VOC Gas Detector
| Continuous detection of VOC gas in explosion hazardous area with flame proof structure  | Self diagnosis function with built-in Microprocessor  | Low concentration VOC gas detection by PID sensor  | Stable quality and proved detection principle for service life  | Multi functions and high quality with low cost of ownership  | Built- in back light LCD or OLED display  | Easy Configuration of monitoring by HART communication  | Auto calibration and program setting with a magnet bar  |  4-20mA (Current source or Sink) output signal   |  With the alarm unit - Normal : Green LED - Alarm : Red LED, Buzzer(95 dB) - Trouble : Yellow LED   |  3-wire connection for signal & power  | Calibration and maintenance by one-man  |

GTD-5000 Intelligent & Sampling Type VOC Gas Detector
|  Fast response of VOC gas leakage with a high efficient pump inside  |  Self diagnosis function with built-in Microprocessor  |  Easy replacement with a cartridge type sensor | Digital flow sensor to maintain regular flow rate | Low concentration VOC gas detection by PID sensor | Stable quality and proved detection principle for service life  | Fine LCD display for showing various information  |  4-digit digital gas concentration display  |  4-20mA (Current source or Sink) output signal  |  Easy configuration of monitoring by RS-485 Modbus or PoE  | 2-stage alarm, 1-trouble display and relay contact output  | 3-wire connection for signal & power  | Calibration and maintenance by one-man  |
CASTRON Fixed type Infrared Gas Detector SIN )) NOUS 7AL2 GTD-5100F(IR) GTD-5000 Non-Explosion Proof Type Interference Gas FtItering Gas Detector GTD-5100F Explosion Proof Type Interference Gas FtItering Gas Detector GIR-3000 Explosion Proof Type Diffusion Infrared Gas Detector GTD-5000F Explosion Proof Type Sampling Infrared Gas Detector GIR-3000A Fixed Type Infrared Gas Detector
GTD-5000 Non Flame proof Type interference gas filtering gas detector
|  Application of anti-malfunction algorithm to interference gas (Ethanol, FC-3283, IPA)  |   Fast response speed by applying new infrared measurement principle   | Built-in high-performance pump to detect gas leakage quickly | Built-in automatic flow control with flow sensor | Built-in self-diagnosis function | Application of proven detection principle with stable quality and service life | Color graphic 1: 1 interactive parameter setting using LCD | 4-20mA (Current source or Sink) output | Easy to install monitoring system by RS-485 Modbus and PoE communication | Two-stage alarm, one fault indication and relay contact output | Built-in Data & Event log function | Calibration and maintenance by one-man |
ACCESSORIES
SPECIFICATION

GTD-5100F Explosion Proof Type interference gas filtering gas detector
|   Built-in explosion-proof structure, installed in dangerous area of gas leakage, continuously detecting gas  |   Application of anti-malfunction algorithm to interference (Ethanol, FC-3283, IPA) |    Fast response speed by applying new infrared measurement principle  |   Built-in high-performance pump to detect gas leakage quickly  |   Built- automatic flow control with flow sensor | Built-in self-diagnosis function | Application of proven detection principle with stable quality and service life | Parameter setti of color graphic LCD and 1: 1 conversation method using magnet bar  |   4-20mA (Current source or Sink) output  |   Easy to install monitoring system by RS-485 Modbus and PoE communication | Two-stage alarm, one fault indication and relay contact output | Built-in Data & Event log function | Calibration and maintenance by one-man | IMAGE

GIR-3000  Explosion Proof Type Diffusion Infrared Gas Detector
|   Continuous detection of flammable, 002, 00 and N20 gases in explosion hazardous area with flame proof structure  |   State-of-the art infrared detection principle  |   L service life time(more than 5 years) | Operation in non oxygen environment | Self diagnosis function with built-in Microprocessor | Fast response time(-5 sec) | Built-in bac light LCD or OLED display | Auto calibration and program setting with a magnet bar | 2-stage alarm, 1-trouble display and relay contact output. | 4-20mA (Current source or Sink) output signal | Easy configuration of monitoring system by RS-485 Modbus or HART communication | With the alarm unit, GTL-100 - Normal : Green LED - Alarm : Red LED, Buzzer(95dB) - Trouble : Yellow LED  |   Calibration and maintenance by one-man  | [ IMAGE

GTD-5000F  Explosion Proof Type Sampling Infrared Gas Detector
|  Continuous detection of flammable, CO2, CO and N20 gas in explosion hazardous area with flame proof structure  |  State-of-the art infrared detection principle  | Operation in non oxygen environment | Digital flow sensor to maintain regular flow rate | Fast response detection of gas leakage with a high efficient pump inside |  Self diagnosis function with built-in Microprocessor  |  Fast response time(<6 sec)  |  Stable quality and proved detection principle for service life  |  Auto calibratio and program setting with a magnet bar | 4-digit digital gas concentration display | 4-20mA (Current source or Sink) output signal | Easy configuration of monitoring system by RS-485 Modbus communication  |  3-Wire Type  |  2-stage alarm, 1-trouble display and relay contact output  |  Calibration and maintenance by one-man  | IMAGE ACCESSORIES
SPECIFICATION
※ Specifications are subject to be changed without prior notice.

GIR-3000A   Fixed Type Infrared Gas Detector
|  Continuous detection of flammable, CO2, CO and N2O gases in explosion hazardous area with flame proof structure  |  State-of-the art infrared detection principle  | Long service life time(more than 5 years) | Operation in non oxygen environment | Self diagnosis function with built-in Microprocessor | Fast response time(<6 sec) |  Built-in back light LCD or OLED display(Only GIR-3000A) |  Auto calibration and program setting with a magnet bar  |  2-stage alarm, 1-trouble display and relay contact output  |  4-20mA (Current source or Sink) output signal  |  Easy configuration of monitoring system by RS-485 Modbus or HART communication(Only GIR- 3000A) | Calibration and maintenance by one-man |
IMAGE
［ SPECIFICATION

B & Fixed type Flammable Gas Detector eastron GTD-5100F Explosion Proof Type Sampling Flammable Gas Detector GTD-5000F Explosion Proof Type Sampling Flammable Gas Detector GTD-3000Ex Intelligent Diffusion Flammable Gas Detector GTD-2000Ex Smart Diffusion Flammable Gas Detector GTD-1000Ex Transmitter Diffusion Flammable Gas Detector GTD-5000 Sampling Flammable Gas Detector GTD-6000 Gas Receiver Combination Flammable Gas Detector
GTD-5100F  Explosion Proof Type Sampling Flammable Gas Detector
|  Continuous detection of Flammable gas in explosion hazardous area with flame proof structure   |   Self diagnosis function with built-in Microprocessor   |  Easy replacement with a cartridge type sensor  | Digital flow sensor to maintain regular flow rate  | Fast response of gas leakage with a high efficient pump inside  | Low concentration VDC gas detection by PID sensor | Stable quality and proved detection principle for service life | Auto calibration and program setting with magnet bar on FND display board  | 4-Digit digital gas concentration display  | 4-20mA (Current source or Sink) output signal  | Easy configuration of monitoring by RS-485 Modbus communication  | 3-Wire Type  | 2-stage alarm, 1-trouble display and relay contact output  | Calibration and maintenance by one-man  |

GTD-5000F  Explosion Proof Type Sampling Flammable Gas Detector
|  Continuous detection of flammable gases in explosion hazardous area with flame proof structure   |  Self diagnosis function with built-in Microprocessor  |  Easy replacement with a cartridge type sensor  |  Digital flow sensor to maintain regular flow rate  |  Fast response of VOC gas leakage with a high efficient pump inside  | Detection gas from low to high concentration owing to various detection principle   |  Stable quality and proved detection principle for service life   |  Auto calibration and program setting with a magnet bar | 4-digit digital gas concentration display | 4-20mA (Current source or Sink) output signal | Easy configuration of monitoring system by RS-485 Modbus communication  |  3-wire connection for signal & power  |  2-stage alarm, 1-trouble display and relay contact output  |  Calibration and maintenance by one-man  |
SPECIFICATION

GTD-3000EX  Intelligent Flammable Gas Detector
| Continuous detection of flammable gases in explosion hazardous area with flame proof structure  | Self diagnosis function with built-in Microprocessor  |  Detection gas from low to high concentration owing to various detection principle   |  Stable quality and proved detection principle for service life   |  Built-in back light LCD or OLED display  |  Auto calibration and program setting with a magnet bar  |  2-stage alarm, 1-trouble display and relay contact output  |  4-20mA (Current source or Sink) output signal  |  Easy configuration of monitoring by RS-485 Modbus communication  |  With the alarm unit - Normal : Green LED - Alarm : Red LED, Buzzer(95 dB) - Trouble : Yellow LED  | Calibration and maintenance by one-man  | [ IMAGE

GTD-2000EX   Smart Diffusion Flammable Gas Detector
|  Continuous detection of flammable gases in explosion hazardous area with flame proof structure  |  Self diagnosis function with built-in Microprocessor  |  Detection gas from low to high concentration owing to various detection principle   |  Stable quality and proved detection principle for service life   |  Built-in back light LCD or OLED display  |  Easy Configuration of monitoring by HART communication  |  Auto calibration and program setting with a magnet bar  |  4-20mA (Current source or Sink) output signal  |  With the alarm unit - Normal : Green LED - Alarm : Red LED, Buzzer(95 dB) - Trouble : Yellow LED  |  Calibration and maintenance by one-man  | IMAGE

GTD-1000EX  Transmitter Diffusion Flammable Gas Detector
| Continuous detection of flammable gases in explosion hazardous area with flame proof structure  | Detection gases from low to high concentration owing to various detection principle  |  Stable quality and proved detection principle for service life  |  High quality and low cost of ownership  |  Simple structure and compact design  | 4-20mA (Current source or Sink) output signal  | Easy calibration & maintenance with inside FND display  | Calibration and maintenance by one-man  | IMAGE

GTD-5000  Sampling Flammable Gas Detector
|  Fast response of flammable gases leakage with a high efficient pump inside  |  Self diagnosis function with built-in Microprocessor  |  Easy replacement with a cartridge type sensor  |  Digital flow sensor to maintain regular flow rate  |  Detection gas from low to high concentration owing to various detection principle  | |  Stable quality and proved detection principle for service life   |  Fine LCD display for showing various information   |  4-digit digital gas concentration display   | |  4-20mA (Current source or Sink) output signal   |  Easy configuration of monitoring by RS-485 Modbus or PoE  |  2-stage alarm, 1-trouble display and relay contact output | 3-wire connection for signal & power | Calibration and maintenance by one-man | ACCESSORIES
SPECIFICATION

GTD-6000  Gas Receiver Combination Flammable Gas Detector
| Self diagnosis function with built-in Microprocessor  | Detection gas from low to high concentration owing to various detection principle   | Easy installation with a wall mount type  |  Check it from a distance by large display and flashing light  |  4-digit gas concentration display  |  3-stage alarm level, 1-trouble LED display and relay contact output  | Easy configuration of monitoring by RS-48S Modbus communication  | Programmable alarm setting (Level, Time delay)  | Self diagnosis  | | 4-20mA(Current source or Sink) output signal  | [ IMAGE ACCESSORIES
SPECIFICATION

3 GASTRON Fixed type Oxygen & Toxic Gas Detector GTD-5100F Explosion Proof Type Sampling Oxygen & Toxic Gas Detector GTD-5000F Explosion Proof Type Sampling Oxygen & Toxic Gas Detector GTD-3000Tx Intelligent Diffusion Oxygen & Toxic Gas Detector GTD-2000Tx Smart Diffusion Oxygen & Toxic Gas Detector GTD-1000Tx Transmitter Diffusion Oxygen & Toxic Gas Detector GTD-5000 Sampling Oxygen & Toxic Gas Detector GTD-6000 Gas Receiver Combination Oxygen & Toxic Gas Detector
GTD-5100F  Explosion Proof Type Sampling Oxygen & Toxic Gas Detector
|  Continuous detection of toxic gases and oxygen gas in explosion hazardous area with flame proof structure  |  Self diagnosis function with built-in Microprocessor  | |  Easy replacement with a cartridge type sensor  |   Digital flow sensor to maintain regular flow rate  |   Fast response of gas leakage with a high efficient pump inside | Superior performance for detecting gases below TLV  | Stable quality and proved detection principle for service life   |  Auto calibration and program setting with magnet bar on FND display board  |  4-Digit digital gas concentration display  |  4-20mA (Current source or Sink) output signal  |  Easy configuration of monitoring by RS-485 Modbus (Isolated) Serial data communication (Option)  |   3-Wire Type  |   2-stage alarm, 1-trouble display and relay contact output  |   Calibration and maintenance by one-man ■ IMAGE
SPECIFICATION

GTD-5000F  Explosion Proof Type Sampling Oxygen & Toxic Gas Detector
|  Continuous detection of toxic gases and oxygen in explosion hazardous area with flame proof structure   |  Self diagnosis function with built-in Microprocessor  | Easy replacement with a cartridge type sensor  | Digital flow sensor to maintain regular flow rate  | Fast response of gas leakage with a high efficient pump inside  | Stable quality and proved detection principle for service life   |   Excellent selectivity by adapting detection principle only for toxic gases & oxygen  |   Superior performance for detecting gases below TLV  |  Auto calibration and program setting with a magnet bar  |  4-digit digital gas concentration display  |  4-20mA (Current source or Sink) output signal  | Easy configuration of monitoring system by RS-485 Modbus communication  |  3-Wire Type  |  2-stage alarm, 1-trouble display and relay contact output  | Calibration and maintenance by one-man  | [ IMAGE DIMENSION (Unit : mm)

GTD-3000TX  Intelligent Diffusion Oxygen & Toxic Gas Detector
|  Continuous detection of toxic gases and oxygen in explosion hazardous area with flame proof structure  |  Self diagnosis function with built-in Microprocessor  | Excellent selectivity by adapting detection principle only for toxic gases & oxygen  |  Stable quality and proved detection principle for service life  |  Built-in back ligh LCD or OLED display  |  Auto calibration and program setting with a magnet bar  |  2-stage alarm, 1-trouble display and relay contact output  |  4-20mA (Current source or Sink) output signal   |  Easy configuration of monitoring system by RS-485 Modbus communication   |  With the alarm unit - Normal : Green LED -Alarm : Red LED, Buzzer(95 dB) -Trouble : Yellow LED  | Calibration and maintenance by one-man  | Superior performance for detecting gases below TLV  | [ IMAGE ACCESSORIES

GTD-2000TX   Smart Diffusion Oxygen & Toxic Gas Detector
|  Continuous detection of toxic gases and oxygen in explosion hazardous area with flame proof structure   |  Self diagnosis function with built-in Microprocessor  | Excellent selectivity by adapting detection principle only for toxic gases & oxygen  | Superior performance for detecting toxic gases below TLV  | Multi functions and high quality with low cost of ownership   |  Built-in back light LCD or OLED display   |   Easy configuration of monitoring system by HART communication   |   Auto calibration and program setting with a magnet bar  | 4-20mA (Current source or Sink) output signal  | With the alarm unit - Normal - Green LED - Alarm - Red LED, Buzzer(95 dB) - Trouble - Yellow LED ·Calibration and maintenance by one-man  | IMAGE
SPECIFICATION

GTD-1000TX  Transmitter Diffusion Oxygen & Toxic Gas Detector
|  Continuous detection of toxic gases and oxygen in explosion hazardous area with flame proof structure  |  Excellent selectivity by adapting detection principle only for toxic gases & oxygen | Superior performance for detecting toxic gases below TLV | High quality and low cost of ownership | 4-20mA (Current source or Sink) output signal  | Simple structure and compact design  | Easy calibration & maintenance with inside FND display  | Calibration and maintenance by one-man  | ACCESSORIES

GTD-5000  Sampling Oxygen & Toxic Gas Detector
|  Fast response of toxic gases & Oxygen leakage with a high efficient pump inside  |  Self diagnosis function with built-in Microprocessor  |  Easy replacement with a cartridge type sensor  |  Digital flow sensor to maintain regular flow rate  |  Stable quality and proved detection principle for service life  |  Superior performance for detecting toxic gases below TLV  | Fine LCD display for showing various information  | 4-digit digital gas concentration display  | 4-20mA (Current source or Sink) output signal  |  Easy configuration of monitoring system by RS-485 Modbus communication or PoE  |  2-stage alarm, 1-trouble display and relay contact output  | 3-Wire Type  | Calibration and maintenance by one-man  |
SPECIFICATION

GTD-6000  Gas Receiver Combination Oxygen & Toxic Gas Detector
|  Self diagnosis function with built-in Microprocessor  |  Superior performance for detecting toxic gases below TLV  |  Easy installation with a wall mount type  | 4-digit gas concentration display   |  3-stage alarm level, 1-trouble LED display and relay contact output   |  Easy configuration of monitoring by RS-485 Modbus communication  |  Programmable alarm setting (Level, Time delay)  |  Self diagnosis  |  4-20mA(Current source or Sink) output signal  | [ IMAGE _ DIMENSION (Unit : mm) _ACCESSORIES
Open Path Gas Detector SafEye Quasar 900
Infrared Open Path Gas Detector CASTRON
SafEye Quasar 900  Infrared Open Path Gas Detector
[ SPECIFICATION

|  Quasar 900 is totally immune to interference from sunlight or any other sources of radiation such as flare stacks, arc welding or lightning.  |  The high power xenon lamp will compensate for changing weather conditions, including rain, fog, mist, snow and makes it immune to influences from solar radiation, arc-welding, stack flares or vibration from machinery.  | Quasar 900 is approved to SIL2 (IEC61508), equipped with heated optics and tolerates a very wide temperature range to provide reliable detection. | No unrevealed failures. In normal operation, the output signal is 4 to 20 mA, depending on the measured gas concentration. Sub-4mA signals includes indications for beam blockage (2mA), a fault (1mA) In addition ■ IMAGE Mount(U-Bolt, 5 inch) GMS-2000 Touchscreen Monitoring Panel GTC-700 Series Analogue Mux Controller (8 Slot GTC-520F Series Single Channel Explosion Proof Type Gas Dete GTC-500 Series Single Channel Gas Detector Re GTC-200F Series Multi Channel Explosion Proof Type Gas Detector Receiver GTC-200A Series Multi Channel Gas Detector Received ASC-100W Safety Controller GASTRON Safety Controller Receiver

SC-100 Safety Controllor
|   Up to 10 charnels of gas detector monitoring (4 – 20mA detection and 24/ supply)  |   Expansion of adtitional 10 charnels when using extension unit (monitoring up to 20 |   Relevy 4 – 20mA output function individual module configuration  |   10.1" touch screen with wide viewing angle and mults-language support  |   Intuitive user interface configuration | Detector position and alarm status can be displayed using map monitoring mode | Detection channel log function and various event storage function | Material sheet (MSDS) viewer for seriesing gas  |   High output buzer and user voice alarm function according to alarm occurrence  |   Various external interfaces such as TDP / IP a Provides earthquake alarm detection function (intensity 3 – 7) | Robust case construction and IP65 waterproof / dustproof  |
[ IMAGE
ACCESSORIES
SPECIFICATION

|   Easy expansion with using a global standard DIN type case  |   Various receiver types available(Wall Mount, Panel Mount and 19* Rack Type)  |   Gas concentration display with 4-digit & Bar-graph (32 segment, 3-color LED)  |   3-color display according to alarm stages (Green, Yellow, Red)  |   3-stage alarm, 1-trouble LED display and relay contact output  |   Easy configuration of monitoring system with RS-485 Modbus isolated serial data communication(Option)  |   Programmable alarm setting (Level, Time delay) | Peak hold | Self diagnosis | HVAC & DP Card Option | IMAGE
DIMENSION (Unit : mm)
ACCESSORIES
SPECIFICATION

GTC-200A Series  Multi Channel Gas Detector Receiver
GTC-200F Series  Multi Channel Explosion Proof Type Gas Detector Receiver
|   Usable in explosion hazardous area with flame proof structure  |    Integrated type receiver with built-in Microprocessor  |   Gas concentration display with 4-digit & Bar-graph (32 segment, 3-color LED)  |    3-color display according to alarm stages (Green, Yellow, Red)  |    3-stage alarm, 1-trouble LED display and relay contact output  |    Easy configuration of monitoring system with RS-485 Modbus isolated serial data communication(Option)  |   Programmable alarm setting (Level, Time delay) | Peak hold | Self diagnosis | HVAC & DP Card Option | ACCESSORIES

DIMENSION (Unit : mm)
GTC-500 Series  Single Channel Gas Detector Receiver
|   Connectable with various types of gas detectors  |  Large-scale digital concentration display and warning light can be applied for remote viewing   |   Gas concentration display with 3-digit, 4-digit,or Bar-graph (32 segment, 3-color LED) | 3-color display according to alarm stages (Green, Yellow, Red) | 3-stage alarm, 1-trouble LED display and relay contact output  |    Easy configuration of monitoring system with RS-485 Modbus isolated serial data communication(Option) | Programmable alarm setting (Level, Time delay) | Peak hold | Self diagnosis |

GTC-520F Series  Single Channel Explosion Proof Type Gas Detector Receiver
|   Usable in explosion hazardous area with flame proof structure  |    Easy installation with a wall mount type  |    Gas concentration display with 4-digit & Bar-graph (32 segment, 3-color LED) | 3-color display according to alarm stages (Green, Yellow, Red) | 3-stage alarm, 1-trouble LED display and relay contact output | Easy configuration of monitoring system with RS-485 Modbus isolated serial data communication(Option) | Programmable alarm setting (Level, Time delay) | Peak hold | Self diagnosis  | ■ IMAGE
ACCESSORIES

DIMENSION (Unit : mm)
GTC-700 Series  Analogue Mux Controller (8 Slot)
|   Microprocessor applied for central system  |   Easy monitoring system configuration with Isolated RS-485 & Ethernet communication  |   Remote interface, channel calibration, data check, device setting is possible using the common unit  |   1 set of controller can configure 64 detectors (8 channel / slot)  |   Controller can be expanded (up to 32 sets) |    Detected value can be displayed in 10uA unit  |    Compensation possible on the difference made due to wiring path  |   High-definition 16-BIT A/D Converter applied  |   Fast scan of input value (0.32 sec / 64 channels)  |   Isolated Analog Input Port  |
IMAGE

GMS-2000  Touch Screen Monitoring System
|  Integrated management of all gas detectors with one graphic   |  Easy configuration with low cost by serial data communication (RS-485 Modbus)  |  Easy output configuration on alarm (by group or single) | Display switch to a trouble point screen when alarm is occurred | Possible to check logged alarm history | Real time logging data  | Easy expansion by serial port  | Ethernet applicable  | Easy to analyze data by providing real-time history trend  | Easily search for data and ranges  |
[ IMAGE
[ DIMENSION (Unit : mm)
Gas detectors location
GAS AL ARM MONTTOR DOD64 FTIR Fourier Transform InfraRed Gas Detection Monitoring ChemLogic 96 ChemLogic 96 Point Continuous Gas Monitor ChemLogic 8 ChemLogic 8 Point Continuous Gas Monitor CLPX ChemLogic Portable Gas Detector CASTRON Colorimetric (ChemLogic) Gas Monitor
DOD64 FTIR  Fourier Transform InfraRed Gas Detection Monitoring

|  Advanced Interferometer : Increased reliability / Lower Maintenance  |  16 - 64 Points of Detection : Flexible & Expandable System  |  20 gas selection per point : Cost effective monitoring per point  | 50 Initial Gases to select from : Flexible to adapt to new requirements  | MCT Stirling cooled detector : Detects gases at a much lower level (PPB)  |  Ceramic Globar IR Source : Increased reliability / Lower Maintenance  |  VCSEL reference laser : Increased reliability / Lower Maintenance  | Sample Cell volume 400ml : Quick sample fill time & analysis (min 10 sec) | USB Port with Memory Stick : Easily Retrievable Data and Spectra | Real Time Trend Display : Quick Real Time Information and Spectra  | Complete Front Access : Easy to Service  | Touch Screen Control : Easy to Startup & Operate  |  Compact Size : Simple to Install/ saves you space  |  Advanced HMI & Flash control : More reliable and easy to learn  | ACCESSORIES IMAGE
ChemLogic 96  chemLogic 96 Point Continuous Gas Monitor

| Intelligent Optics : Reduced Maintenance and Risk of False Alarms  | 16 to 96 Points of Detection : Flexible & Expandable System  |  Up to 3 Gas Families : Reduced Installation & Operating Costs  | Optimized Flow System : Faster Response  | 4 Month Cassette Life : Reduced Operating Cost  |  ChemLogic Technology : Field Proven Reliability | Remote Mountable UO : Reduced Cost of Installation | USB Port with Memory Stick : Easily Retrievable Data | Real Time Trend Display : Quick Real Time Information  | Complete Front Access : Easy to Service  | Touch Screen Control : Easy to Startup & Operate  | Compact Size : Simple to Install  | Energy Efficient (Green) : Uses –90% Less Energy Per Point than the Competition  | * Additional ranges may be available and are subject to change. Please see specific product brochure or contact DO Technologies. ** Roquires pyrolyzer option for detection ^  ^^ Compounds which hydrolyze to HUI ・・・ Compounds which hydrolyze to HBr
ChemLogic 8  ChemLogic 8 Point Continuous Gas Monitor

| New intelligent Optics : Reduced Maintenance and risk of False Alarms  | Optimized flow system : Quicker response  | 2 & 4 Month Cassettes : Reduced operation cost  |  ChemLogic Technology : Field proven reliability  |  Remote mountable I/O : Reduced cost of installation  |  Flash Memory Storage : Easily retrievable data  | Real time trend display : Quick real time information  | Complete front access : Easy to service  | Touch screen control : Easy to startup & operate  |  Compact size : Simple to install  | [ IMAGE * Additional ranges may be available and are subject to change. Please see specific product brochure or contact DOD Technologies. ** Requires pyrolyzer option for detection A  AA    Compounds which hydrolyze to HCl a a a  Compounds which hydrolyze to HBr Alarm 1 Default LOL 1.9 0.4 0.2 0.4 0.9 0.9 0.3 0.2 90.1 1 0.4 39.5 6.7 0.4 14.9 0.2 86.5 0.4 98.3 0.4 0.2 1.5 26 1.2 a 02 0.3 0.4 0.4 18.2 16.7 0.3 3 1 12.5 0.7 30 02 49 5.3 0.2 0.2 0.4 0.3 0.2 0.4 0.2 0.2 233 0.4 0.4 0.2 1.1 0.2 24.9 1 0.4 0.2 4 0.2 14.6 19.3 0.2 97.1 0.1 0.2 0.2 0.2 02 39 * Additional ranges may be available and are subject to change. Please see specific product brochure or contact DOO Technologies ** Requires pyrolyzer option for detection ^ Compounds which hydrolyze to HF ^^ Compounds which hydrolyze to HCl ^^^ Compounds which hydrolyze to HBr
|  Improved Optics Design : Provides low level gas detection in seconds   |  Immediate Alert Indicator Bar : Immediate Indication that gas is present   |  Automated Calibration : Eliminates the need for optical calibration | Built in Data Logging : Micro SD card can record up to 5 years of data | Real Time Clock w/Data Stamp : Accurate recording of data  | SW Selectable Calibration Curves : All gas calibrations are user selectable(no gase yks needed) system is flexible to meet your gas detectione gruirements  | Extended Battery Life : Up to 12 hours of continuous monitoring  | Built in STEL : Unit constantly calculates for a rolling 15 minute STEL  | 2 Alarm Levels : User selectable alarm levels  |  Purged Cassette : Insures colorimetric media will not be contaminated by ambient gas  |
ChemLogic Portable Gas Detector
Portable Gas Detector CASTRON
Cartridge Sensor & Sampling Portable Detector TANGO TX1 MX6 iBrid Single Gas Portable Detector (CO, H2S, NO2, SO2)    6-Gas Portable Detector (LEL, Toxic , IR, VOC) RADIUS BZ1 GasBadge Pro Single Gas Portable Detector (02, NH3, CL2, PH3, 7-Gas Area Monitor (LEL, Toxic, VOC) HCN, H2, CLO2) VENTIS MX4 4-Gas Portable Detector (LEL, Toxic) VENTIS PRO 5-Gas Portable Detector (LEL, Toxic , IR, VOC)
┌ G-Finder Single  single Gas Portable Detector
| 2 years warranty without replacement of battery and sensor  | Easy to operate with one button  | Real-time gas detection and display  | Regular calibration period notification function | Periodic Bump Test Cycle Notification Function |
■ IMAGE
ACCESSORIES

GPD-100 Cartridge Sensor & Sampling Portable Detector
|  Fast response of gas leakage with a high efficient pump inside  |  Easy replacement with a cartridge type sensor (same as GTD-5000 cartridge)  |  Easy calibration and get information from the face displayed with backlit color LCD  |  10 hours continuously operation with a rechargeable Li-ion battery and charge power with a power adapter  |  Data logging function with Micro SD (T-flash) card (Text File)  |  Zero calibration button Convenient usage structure through front placament  | IMAGE

GasBadge Pro  single Gas Portable Detector
TANGO TX1   Single Gas Portable Detector
SPECIFICATION
■ CERTIFICATIONS

|  First-ever single gas monitor that uses two sensors to detect the same gas  | Two years of continuous battery runtime  | Easy Battery Replacement  | Louder Alarm Sound than other brands (100dB @ 10cm) | Selectable Detection Gas(CO, CO / H2 low, H2S, NO2, SO2) | |  Full-featured, single gas detector  |  Lifetime warranty  |  Interchangeable sensors  |  iNet™ Ready and Docking Station™ compatible  |  Easy Battery Replacement  | | Selectable Detection Gas(CO, H2S, 02, NO2, SO2, NH3, CL2, CLO2, PH3, HCN, H2, CO/H2 low) | IMAGE ■ IMAGE
VENTIS PRO  5-Gas Portable Detector
VENTIS MX4 4-Gas Portable Detector

| Monitors 1-4 gases | Sampling Pump | Configures with or without a pump | Available with a black or safety orange overmold | iNet™ Ready and DS2 compatible  | Selectable Detection Gas(LEL, CH4, 02, CO, CO/H2 low, H2S, NO2, SO2)  | |  Monitors 1-4 gases  |  Sampling pump  |  Choice of black or orange  |  Man alarm and panic button  |  Track users and sites with iAssign ™  |  Lens Wireless(Real- time alarm and measurement check) function | i-Net now available | Communication between Radius BZ1 and monitor using LENS ™ Wireless | [ IMAGE [ IMAGE _ACCESSORIES ACCESSORIES
RADIUS BZ1  7-Gas Area Monitor
– MX6 iBrid  6-Gas Portable Detector
SPECIFICATIUN

| Monitors 1-6 gases  |  24"Plug-and-Play" fie Id-replaceable sensors including PID and Infrared options  |  Full-color LCD display  |  Sampling Pump  |  Compatible with iNet ™ and docking station  |  Up to 6 gases monitored simultaneously  | |  Monitors 1-7  |  Sampling pump  |  Longest monitoring with up to 7 days (168 hours) of runtime  |  Communication between Ventis Pro and monitor using LENS ™ Wireless  |  108dB powerful sound at 1m distance  |  DualSenseß Configurable  |  Peer to Peer for real-time alarm and measurement  |  Long distance communication up to 300M | IMAGE ACCESSORIES ACCESSORIES [ IMAGE
– iNet® NOW  Constant monitoring configuration for sealed operation
■ CONSTANT MONITORING CONFIGURATION FOR SEALED OPERATION
[ FEATURE
| Real-time warning about panic and manhole in gas leak area  | | Track instrument status via real-time text messages, email alerts, and real-time maps from any device connected to the web, even outside the office  | | When the detector is connected to iNet, it can automatically grasp the status and position of operator  | | Connectable up to 9M Bluetooth distance  | | Update operator's status and location every 15 seconds  | · With LENS Wireless, you can be aware of the alarm and respond appropriately in a place that is close to the place of alarm. · Operate with the detector itself without setup or IT support • No other equipment required - Provide alarms and data via peer reading · Up to 25 devices can be connected in one group Alarms from one device can be checked on all connected devices · Real-time warning about panic and manhole in gas leak area Track instrument status via real-time text messages, email alerts, and real-time maps from any device connected to the web, even outside the office · When the detector is connected to iNet, it can automatically grasp the status and position of operator · Connectable up to 9M Bluetooth distance Update operator's status and location every 15 seconds iNet® Now Control iNet® Now Sync Devices ▶ If iNet is disconnected, the detector can be audible and visual alarm ▶ Smart device is compatible with iNet Now Sync app ▶ iOS 10.0+ or Android 5.1+ ▶ Bluetooth Low Energy (BLE) v4.1 or higher ▶ Running GPS-based applications With Lens Wireless, you can gain visibility into the risks that occur up to 100 meters away so you can take the appropriate safety precautions when the gas reading in your area reaches a critical level Radius BZ1 (LEL,02) Lens Wireless LENS Wireless Sampling tube LENS Wireless
GTF-1100U UV/IR Flame Detector 975UF UV/IR Flame Detector GTF-1100T Triple IR Flame Detector 975UR UV/IR Flame Detector 975MR Triple IR Flame Detector 975HR Multi IR Flame Detector GASTRON
GTF-1100U  UV/IR Flame Detector
| Detection distance: 30m  | Viewing angle: Vertical 100 °, Horizontal 100 °  | Automatic & manual self-diagnosis function  |  interface - Relay contact output (fire & breakdown) - RS-485 communication - 0 ~ 20mA output  |  Indicators - Status LED (Bi-Color: RED & GREEN), Alarm LED (RED)  |
ACCESSORIES

GTF-1100T Triple IR Flame Detector
|  Detection distance: 50m  |  Viewing angle: Vertical 100 °, Horizontal 100 °  |  Automatic & manual self-diagnosis function  |  interface - Relay contact output (fire & breakdown) - RS-485 communication - 0 ~ 20mA output  |  Indicators  |  Status LED (Bi-Color: RED & GREEN), Alarm LED (RED)  | ACCESSORIES

975MR  Triple IR Flame Detector
SPECIFICATION
[ IMAGE
ACCESSORIES

|  Multi spectrum design-for long distance detection and high false alarm immunity  |  Sensitivity selection-to ensure no zone crossover detection  |  Automatic and Manual Built-In-Test (BIT)-to assure continued reliable operation  |  Heated window-for operation in harsh weather conditions (snow, ice, condesation)  |  Multiple output options for maximum flexibility and compatibility - Relays (3) for Alarm, Fault and Auxiliary - 0-20mA (stepped) - Hart Protocol for maintenance and asset management - RS-485, Modbus Compatible | High Reliability-MTBF-minimum 150,000 hours | Approved to Safety Integrity Level 2 (SIL2-TUV) | 5-Year Warranty | | User Programmable via HART or RS-485 |
975HR  Multi IR Flame Detector
SPECIFICATION
ACCESSORIES

|  Multi spectrum design-for ling distance detection of hydrocarbons and hydrogen flames  |  High false alarm immunity  |  Sensitivity selection-to ensure no zone crossober detection  |  Automatic and Manual Built-In-Test (BIT)-to assure continued reliable operation  |  Heated window-for operation in harsh weather conditions (snow, ice, condesation) | Multiple output options for maximum flexibility and compatibility - Relays (3) for Alarm, Fault and Auxiliary - 0-20mA (stepped) - HART Protocol for maintenance and asset management - RS-485, Modbus Compatible | High Reliability-MTBF-minimum 150,000 hours | Approved to Safety Integrity Level 2 (SIL2-TUV )  |  5-Year Warranty  |  User Programmable via HART or RS-48  | [ IMAGE
975UF  UV/IR Flame Detector
SPECIFICATION
ACCESSORIES

|  UV/IR Dual-Sensor  |  Fast Response-200 msec to flash fire  |  Solar blind  |  Automatic Built-In-Test (BIT)* and Manual-to assure continued reliable operation  | |  Heated window-for operation in harsh weather conditions(snow, ice, condesation)  |  Multiple output options for maximum flexibility and compatibility - Relays (3) for Alarm, Fault and Auxiliary - 0-20mA (stepped) - HART Protocol for maintenance and assetmanagement - RS-485, Modbus Compatible | High Reliability-MTBF- minimum 150,000 hours | Approved to Safety Integrity Level 2 (SIL2-TUV ) | model 40/40LB only | 5-Year Warranty User Programmable via HART or RS-485 | ■ IMAGE
975UR  UV/IR Flame Detector
SPECIFICATION
ACCESSORIES

| UV/IR Dual-Sensor  |  High Speed Response-20 msec to flash fire  |  Solar bind  |  Automatic Built-In-Test(BIT) and manual-to assure continued reliable operation  | | Heated window-for operation in harsh weather conditions(snow, ice, condensation) | Multiple output options for maximum flexibility and compatibility - relays (2) for Alarm and Fault - Analogue output for fast detection - 0-20mA (stepped) - HART Protocol for maintenance and asset management - Designed to Safety Integrity Level 2 (SIL2-TUV) - 5-year warranty - User programmable via HART or RS-485  | ■ IMAGE
┌ Incus Ultrasonic Gas Leak Detector  Ultrasonic Gas Detector
|  Instantaneous response to all gas leaks(toxic, combustible, or inert) |   Operates in extreme temperatures  |   Automated electronic self-test offers failsafe operation |  Widest area of coverage through four independent sensors  |  4–20 mA analog or stepped, "multi-drop" RS-485 digital interface, and HART® communication protoco  |  Certified worldwide for hazardous locations  |  Innovative mapping tool helps optimize coverage for a target risk  |  Programmable time delays screen intermittent unwanted alarm sources  | [ IMAGE
[ DIMENSION (Unit : mm)
SPECIFICATION

GTL-200 Flame Proof Type Sounder & Beacon Combination BExBG Series Flame Proof Type Xenon MB005 Non-Flame Proof Type Xenon Strobe Beacon BExS Series Flame Proof Type Sounder MA112 Non-Flame Proof Type Sounder - Effective range 100M BExCP3A / BExCP3B Series Manual Call Poin Alarm Signalling / Manual Call Point
GTL-200B  Explosion Proof Sounder
[ SPECIFICATION

GTL-200L  Explosion Proof Warning Light
［ SPECIFICATION

L-200  Explosion Proof Combination
［ SPECIFICATION

GTL-200  Explosion Proof Type Sounder & Beacon Combination
|  Combined Explosion-proof Sounder &Beacon using LED Light sources   | Excellence of high effect of signal light radiation using our unique lens type luminescence system   | Available of sound types & volume control with internal switch   |  Available in controlling the rotation speed of reflector   | Available in various colors(Red, Yellow, Green, Blue)  |  Max Output : 120 dB  | Available in setting options depend on field condition  |  A total of 64 sound files  |  AC or DC available  |
SPECIFICATION

┌ BExBG Series  Explosion Proof Type Xenon
| Automatic synchronisation on multi-beacon system.  | Beacons can be set to 'flip-flop' alternating mode with other units on multi-beacon systems.  | Xenon tubes mechanically secured against shock & vibration      Glass lens & Stainless Steel guard    | Very large termination area.    | Ratchet adjustable stainless steel 'U' bracket. | IN & OUT terminals (Ex de version only)  |

MB005  Non-Explosion Proof Type Xenon Strobe Beacon
|  Automatic synchronised flash, or Fip-Flop alternating mode.    |   Xenon tube mechanically secured against vibration.   |  Continuously rated.   |   Large termination are |  Stainless steel fixings.  | Ratchet adjustable stainless steel 'U' bracket for 360'positioning.  | Duplicate cable terminations (in & out for daisy-chain installations). | Tropicalised as standard.  | IMAGE

┌ BExS Series  Explosion Proof Type Sounder
| Automatic synchronisation on multi-sounder system.  | Very large termination area.  | Ratchet adjustable stainless steel 'U' bracket.  | IN & OUT terminals (Ex de version only).   |   'Programmable' version available: - 45 alarm tones - 4 remotely selectable stages - Any tone can be assigned to Any stage - User configurable continuous frequency tone  |

MA112  Non-Explosion Proof Type Sounder - Effective range 100M
| Automatic synchronisation on multi-sounder system.  | Continuously rated.  | Large termination area.  | Stainless steel fixings. | Ratchet adjustable stainless steel 'U' bracket for 360°positioning.  | Duplicate cable terminations (in & out for daisy-chain installations). | Tropicalisation available on request.  | Available with custom tone configurations and frequencies. |  'Programmable'version available: - 45 alarm tones - 4 remotely selectable stages - Any tone can be assigned to Any stage - User configurable continuous frequency tone |

┌ BExCP3A / BExCP3B Series  Manual Call Point
|  Alternative housing colours are available to meet specific requirements.   |  DIN rail mounted terminal blocks: 8 x 2.5mm²  |  Stainless Steel lift flap  |  Metalised Polyester "Duty" label.  | Series and/or End of Line resistors.  | [ IMAGE BG(Break glass type) [ DIMENSION (Unit : mm) Smn MOUNING HOLES 4 OFF
Calibration Gas & Industrial Safety Product Can Gas Cylinder Acid Leak Detection paint Safety Shield
Can Gas  Calibration

| Easy to use - Combined pressure regulator and flowmeter in the container allows to understand gas flow at a glance  | Affordable and economical - you can buy whatever you need when you need it compared to traditional high volume products  | High portability - small size of 300 x 65mm and light weight (150g)   | Safe without container explosion - Inner charge pressure is designed to 8Kg / cm², low concentration gas that can not be burned is charged  | [ IMAGE
Safety Shield
┌ Acid Leak Detection paint  Acid Leak Detection Paint
PRODUCTS
PPL safety shield
TEFLON safety shield
FEATURES
VUE safety shield
Metal safety shield
Applicable Acid & Gas
|  Acid Leak Detection Paint that is a visual detection system to help quickly identify leaks in piping connections where acids or caustics are present.   |  It provides a physical color change so that operators, maintenance or other personnel can quickly identify a leak saving valuable time and money in locating the source of a leak versus using electronic detection devices.   |  Another unique property of the On Guard Detection Paint System is that after an "activation" or color change and the leaking component is fixed, the Acid Leak Detection paint can be neutralized and it will return to its Neutral color ready to activate again. This saves money by not having to re-coat the connection.  | Recommend no more then 3-4 exposures before repainting.  | |  Protect personnel from injury  |  Prevent damage to equipment  |  Detect leaks through a pH indication patch or visual  |  Reduce the clean up area after a material release | - Suitability for temperatures up to 60°C  and in the presence of certain chemicals, such as concentrated acids, only at lower temperatures. - Wide applications, particularly in conditions with ambient temperatures and pressures of 21 bar or lower. - Added benefit of a "See-Thru" feature, permitting easy visual inspection of pipe joint-flanges, valves and threaded connections - Used to protect against hydrofluoric acid, a highly virulent chemical that is used to etch glass and is also employed in petroleum cracking operations. - Thermoplastic cloth demonstrates temperature and pressure resistance up to 225 °C and 76 bar. -  at sulfuric acid concentrations of 30% or less, are recommended up to line temperatures of $3 °C - Demonstrate resistance to temperatures up to 232°C pressures up to 114 bar and are inert to a wide spectrum of commercial chemicals. - Outstanding protection against corrosion. - Recommended in the chemical, pharmaceutical, electronic, pulp/paper, food processing industries, as well as other vital manufacturing applications in which corrosive chemicals at high temperatures are frequently used. - For lines carrying sulfunc acid, this is most frequently specified. - It can resist thisacid at any concentration, including fuming oil. - The Acid Detection Paint is YELLOW in a neutral state and changes to RED when it comes in contact with liquid 2.7 pH or below. - STORAGE TEMPERATURE : Minimum 7°C Maximum43 °C - COVERAGE RATE : 250 to 300 square feet per gallon - APPLICATION PROCEDURE : Apply ON GUARD™ to achieve 4 mil dry thickness (2-3 brush coats) - SURFACE TEMPERATURE : Minimum 40 °F (4 °C) / Maximum 120 °F (49 °C) - CIEAN UP INSTRUCTIONS : Clean all equipment immediately after use with Xylene. Flush and clean spray equipment before material dries - APPLICATION EQUIPMENT : Air or airless spray, brush or roller - CAPACITY : 3.79 LITTER / 0.9 LITTER - "See-Thru" shield that permits easy identification and inspection of all pipe joints, including flanges, valves, threaded connections and expansion joints. - Withstands temperatures up to 150°C and pressures to 69 bar. - It demonstrates a broad spectrum of chemical resistance, especially to most corrosive chemicals and organic solvents. - Use as a top coat over primer. Make sure the surface is clean and free of any grease, dirt, or, foreign material. Previously painted surfaces should be free of any loose o scaling paint. Souff-sand overly smooth surfaces. Avoid spraying in windy or dusty conditions. Cover adjacent surfaces to protect them from overspray mist. - Vigorously shake can for at least 1 minute after hearing the ball rattle to ensure uniform color and to prevent clogging. Point spray button opening toward object holding can 12 to 14 inches from the surface to be sprayed. Press button firmly and use steady, even strokes during application. For best results, apply only when container temperature is 18°C – 32°C. Use when humidity is below 85%. Allow more dry time at cooler temperatures - STORAGE TEMPERATURE : Store at room temperature 20°C - COVERAGE RATE : one 12 oz can should cover 20-25 feet 2 with 2-3 thin, even coats. - Note : Multiple thin coats are better than one heavy coat. - CAPACITY : 12 OZ (340 grams) - Galvanized steel, "304" stainless steel, "316" stainless steel - It issufficiently heavy to withstand high pressures and also has flexibility to permit shaping - Galvanized- Installed widely for water, oil andsteam applications - It is vulnerable to corrosive attack and should not be used on lines carrying hazardous chemicals - Withstands temperatures up to 427°C  and pressures to 207 bar. - SUS304 - Applications requiring mild corrosion resistance Withstands temperatures up to 1454°C and pressures to 207 bar. - SUS3016 - To give superior resistance to pitting and to most types of corrosion, making "316" suitable for applications involving severe corrosive conditions. Withstands temperatures up to 1454°C and rpessures to 207 bar. - Any bare or previously painted metal should be primed with a quality primer such as White Primer #315. It may be used over existing paint if the user is assured that the sub coating is stable and the surface is clean, dry, rust, dust, oil and contamination free. - CAPACITY : 3.79 LITTER - RTY Neutralizer is a specially formulated liquid contained in a spray bottle that aids in the neutralization of material pH 2.7 and below so that it can be safely cleaned as well as helping return the Acid Detection paint back to its neutral color of YELLOW. - CAPACITY : 8 02(226 grams) Hydrochloric Acid, Chlorine, Hydrofluoric Acid, Oxalic Acid, Nitric Acid, Nox, TFA(Trifluoroacetic Acid), Sulfuric Acid, Phosphoric Acid, Glacial Acetic Acid, Perchloric Acid, Citric Acid, Other acids and acid gases below pH 3 PVC safety shield al fi Gas Monitoring System Configuration CASTRON Configuration-1 Gas Detector ⇒ Receiver Configuration-2 Gas Detector ⇒ Receiver ⇒ PC Monitoring System Configuration-3 Gas Detector ⇔ PLC & Signal Converter ⇔ PC Monitoring System Configuration-4 Gas Detector Serial Data Communication ⇒ PC Monitoring System Configuration-5 Gas Detector (PoE) ⇔ DB & I/O Server ⇔ PC Monitoring System)
┌ Configuration-1 Gas Detector ⇒ Receiver
− Configuration-2  Gas Detector ↔ Receiver ↔ PC Monitoring System
[ FEATURE
[ FEATURE
|  Real time display (gas detector operation condition)  |  Easy check of gas concentration with 4-digit FND and Bar-graph  |  Easy expansion by using din-case type  | | Easy maintenance due to simple configuration  | Visual and audible indication for each alarm  | Relay contact output for each alarm  | 4-20mA DC output for every single gas detector  | | Real time monitoring and control of gas detectors | Integrated management of whole gas detectors with one graphic | Dual structure system (PC monitoring system ÷ Gas detector receiver) | Display switch to a trouble point screen | Possible to check logged alarm history | Logging real time data | Easy data analyzation with a real time trend  | Easy reporting (day / week / year / user defined date)  | This configuration is the most widely used type in industrial plants and each gas detector is connected with a control unit. The control unit receives 4-20mA DC signal from a gas detector and shows operating condition and gas leakage. An user can easily check them with 4-digit FND, Bar-graph and LED to take immediate action when alarm is triggered. This configuration is PC monitoring system which possible to be checked operating condition of gas detectors and gas leakage on PC screen at a look. An user can take immediate action when alarm is triggered. DC 4-20mA
Configuration-4  Gas Detector Serial Data Communication ⇒ PC Monitoring System
┌ Configuration-3  Gas Detector ⇒ PLC & Signal Converter ⇒ PC Monitoring System
[ FEATURE
[ FEATURE
| Easy expansion with PLC  | Easy configuration of digital output and relay output  | Real time monitoring and control of gas detector with realistic graphic  | |  Integrated management of whole gas detectors with one graphic  |  Display switch to a trouble point screen when alarm is occurred  |  Possible to check logged alarm history | Logging real time data | Easy reporting (day/week/year/user defined date) | |  Integrated management of all gas detectors with one graphic   |  Easy configuration with low cost by serial data communication (RS-485 Modbus)  |  Easy output configuration on alarm (by group or single) | Display switch to a trouble point screen when alarm is occurred | Possible to check logged alarm history | Logging real data | Easy expansion by serial port | Ethernet applicable | This configuration is PC monitoring system and each gas detector is connected to PLC. PLC converts analog signal to digital signal and sends them for PC to show operating condition of gas detectors and gas leakage at a look. An user can take immediate action when alarm is triggered. This is touch screen & PC monitoring type system. It communicates with each gas detector by serial data communication.
┌ Configuration-5  Gas Detector Serial Data Communication ⇒ PC Monitoring System
_ FEATURE

| Real time monitoring and control of gas detectors with the realistic graphic  | Integrated management of all gas detectors with the one graphic  | Display switch to the trouble site on occurring alarm situation | Available of saving, searching and printing a detailed alarm history data | Real time data collecting and logging | |  Easy data analysis with an real time history information   |  Easy reporting (day/week/year/user defined data)   |  Manageable detectors on the Web   |  Easy controlling data in a DB & I/O server  | Bissa Pet Bower Gen Electronic owercovers Steer fathyas Bissa Associations B 2000151800136013601312499