Remote Operation from Your PC! Control and Communication Functions in a Slimline Unit! APERATURE CONTROL
Specifications

When adding options, enter them by punctuating with a comma.
Name and functions of the sections
Small and space saving
Width: 17.5mm, Height: 75mm, Depth: 85mm (excluding the plug) Its compactness ensures easy mounting when only limited space permits. 17.5mm Ultra Thin
Multi-input
18 type of inputs such as thermocouple (10 types), RTD (2 types), DC current (2 types), DC voltage (4 types) are available. 0.25 second input sampling period allows applications for various processes
Multi-drop communication
Various operations and settings can be carried out from programmable PCs interfaces and through communication function (Up to 31 units of the NCL-33A) can be connected.) For the communication protocol, Shinko protocol,
Modbus protocol are available. (For Modbus protocol,
RTU mode and ASCII mode can be selected by the DIP switch.) Connection to the open network Modbus is possible.
Standard 4 alarm points
Four points of alarm are standard feature. However, for Alarm 2, Alarm 3 and Alarm 4, the available.
Easy mounting, removal and maintenance
Hook the NCL-13A into the DIN rail, then fit the unit to the DIN rail. Since the unit is fixed tightly to the DIN rail, it is resistant to vibration, whereby ensuring ease of maintenance.
Easy connectivity
Using a screw type plug, the power and communication connection of the NCL-13A units The spring type plug for input and output can also be connected or removed.
separately) between input terminals. ①Power indicator (POW) A green LED lights up when the instrument is switched on A green LED flashes when an alarm (temperature alarm, Heater burnout alarm/Actuator short circuit alarm, Loop break alarm) is activated. ②Communication indicator (T/R) A yellow LED lights up during Serial communication TX output. ③Control output indicator (OUT) When Control output is ON, green LED lights up. For DC current output, a green LED flashes corresponding to the output manipulated variable. ④Rotary switch for instrument setting Sets the instrument number of the NCL 1 3A ©Spring type plug(Removable) Plug for the input, control output and Event output. ©CT1, CT2 input connectors Connectors for CT input of Heater burnout alarm/Actuator short circuit alarm (Option: W, W3) ⑦DIP switch for Communication speed and Communication protocol Dip switch for selecting Communication speed and Communication protocol ®Bus plug (Removable) Plug for lines (Power bus, Communication bus) When connecting multiple units, the bus plug can connect to another bus plug.

Power supply terminals
■CT dimensions
■ DIN rail mounting (Applicable DIN rail specifications)
[POWER SUPPLY] [EVENT OUTPUT] Event output terminals [Alarm 1 (shares output terminals with Heater burnout alarm/Actuator short circuit alarm and Loop break alarm) or Cooling control output.] Main control output terminals [MAIN OUTPUT] Serial communication terminals CT (current transformer) input terminals These terminals are attached only when the option Heater burnout alarm is applied. Thermocouple input terminals RTD input terminals DC current, voltage input terminals ■External dimensions CTL-6-S (for 20A) CTL-12-S36-10L1U (for 100A) Width : 35mm Height : 7.5mm or greater Groove width : 23mm or greater DIN rail mounting screw height: 3.4mm or less (At a DIN rail height of 7.5mm) [RS-485] [CT1, CT2] [TC] [RTD] [DC]
When connecting to a PC (Basic configuration)
If a personal computer is connected to the NCL-13A units via a Communication converter(IF-400), up to 31 temperature control points can be monitored.
When connecting to a touch panel unit
Bý connecting to a touch panel unit (programmable interface), up to 31 points of temperature control and monitoring can be performed. The NCL-13A complies with the following touch panel units. Digital Electronics Corp.: GLC series, GP series
When connecting to the PLC
A maximum of 31 units of the NCL-13A can be connected with the PLC via PLC interface unit SIF-400. Please contact us to determine which PLCs correspond to the SIF-400. To ensure safe and correct use, thoroughly read and understand this manual before using this instrument. This instrument is intended to be used for industrial machinery, machine tools and measuring equipment. Verify correct usage after consulting purpose of use with our agency or main office. (Never use this instrument for medical purposes with which human lives are involved.) External protection devices such as protection equipment against excessive temperature rise, etc. must be installed. as melfunction of this product could result in serious damage to the system or intery to personnel. Also proper periodic maintenance is required. This instrument must be used under the canditions and environment described in this manual. Shinko Technas Co, Ltd, does not accept lebility for any injury, loss of life or damage occurring due to the instrument being used under conditions nat atherwise stated in this manual. • This catalog is as of September 2006, and specifications are subject to change without notice. • If you have any inquiries, please consult our agency or with us directly.
SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION
Reg. Office : 2-5-1, Senbahigashi, Minoo, Osaka, 562-0035, Japan : 81 - 72 - 727 - 6100 Tel
Caution with respect to Export Trade Control Ordinance To avoid this instrument from being used as a oomsonent in areas bong used in the status of the status of the status of the status.
One status as mostlation the end used in the end used in the status.
In the cooo oreas