Connectable peristaltic pump-cassettes without tools Micro tube-pump systems Micro actuator x Fluid control Powered by Micro gear tecnology Provide flexibility for your needs ↑ Antibacterial resistance Minimal contamination risk: Autoclavable and disposable Pump cassette ♂ Laboratory scale Available in CO2 incubator due to its small size and light weight Precision fluid control Variable flow rate pump. Flow range: 0.1 to 2,000 μL/min
Icomes lab Co., Ltd is one of leading suppliers of life-Science and medical pump. Micro Tube-pump systems provide flexibility base on your needs and requirements. In addition, it allow to customize for varius purpose with abundant option parts. Customizable depending on your needs The pump systems can be customized as your needs, Just switch variable parts. Tray for For microfluidics option
Various pump lineup
Variable flow-rate is Flow-range 0.1 to 2,000 µL/min. Connectable up to 6 pump cassettes with nub-Bricks Feature. Note that the number of cassette varies depending on flow rate.
Sterilizable pump-cassette
Pump cassette is autoclavable and disposable. Minimize contamination risks. Note that no tools required for installation and removal Effects on cells: Can transport cell suspensions Experimental conditons : NIH/3T3 (passage 3) Cell line Medium : DMEM (10% FBS, 1%Penicillin / Streptomycin) : 2mL/min Flow rate Infusion volume : 1mL Cells/mL x10 5 Viscosity: Can transport high viscosity fluid Stress: Reduced stress on cells during medium exchange · Human fetal-derived hepatocytes (293T) • Culture for 3 days Reference : R.Yasuda et al., "Cell stress reduction by a novel perfusion - culture system using commercial culture dish" ,Appl Sci, 10, 95, 2020. • Measured value of flow rate when fluide is sent at a flow rate of 280 µ L / min. • A fluide with a viscosity 300 times higher than that of blood (about 4 cP) can be sent. Perfusion citure for Regenerative medical research Connected with Perfusion attachments (iP-TEC Perfusion attachment, Sanplatech Co., Ltd.) iP-TEC Perfusion attachment can perfuse individual culture wells by single driving unit with 6 pump cassettes and observe through a microscope. Time-lapse imaging for drug susceptibility assay Setup on the stage of a time-lapse imaging system ( BioStudio-T, Nikon Engineering Co., Ltd. ) installation in various incubators. Objective: CFI Plan Fluor DL 4x Camera  : 131 mega-pixels, monochromatic Expose   : 9 msec Enhance/gain  : auto Micro physiological sytem (microfluidic cell culture) Connected with Microfluidic tip (μ-slide, ibidi Co., Ltd.) µ-slide, ibidi Co., Ltd. Example of connection with Kyodo International "Chip Holder"
Extracellular potential recording system -
Connected with extracellular potential recording system(MED64 system, Alpha MED Scientific Inc.) (iP-TEC Perfusion attachment, Sanplatech Co., Ltd.) Basic specification
Connectable up to 6 pump-cassettes
Disposable casset specification

Adaptable tube Inner Φ0.8 to Φ2.5 mm Pump cassette Note that the design specifications of our products are subject to change without prior notice, For inquiries, please contact the company. ■ Manufacturer Icomes Lab Co., Ltd. E-mail:globalinfo@icomes.co.jp URL:https://www.icomes.co.jp/en/ ■ Contact information TOLIMS Co E-mail:global@tolims.co.jp URL:https://www.tolims.co.jp/en/