Aperio GT 450 1.0.1.8000 Patch Notice
For Research Use Only. Not for use in diagnostic procedures.
Leica Biosystems is always looking for ways to further improve our customers’ experience using ourproducts. Our software patches often include new features, but also include improvements based on
customer input and our own internal testing. Please make sure to read below for the details of this patch
so you can get the best out of your product. Thank you for being a customer of Leica Biosystems.

New Features Included in This PatchComponent

Case #Description

Controller27500

DICOM images are now updated with barcode ID information andcorrect image types (labels) for the thumbnail and macro images toensure compatibility with DICOM standards.

Controller

27501

Aperio GT 450 scan status as shown in Aperio eSlide Manager now
shows SUCCESS instead of SCANSTATUS_SCANNED orSCANSTATUS_IQ.

Console27502

The SAM IP address can now be updated without logging in with aPIN.

Controller

27503

For consistency, units of scan width, height and scan time are now
identical to those of the Aperio AT2 scanner:
Scan width and height => mm, scan time => hh.mm.ss.sss

Controller/SAM

27506

Two new barcode user configuration settings BARCODE VALUEMODIFIER and BARCODE VALUE SUBSTITUTION FORMAT wereadded to SAM. The scanner Barcode decoder is updated to takethese settings as inputs to exclude and modify non-printable/illegal
characters from the decoded barcode value. See Aperio GT 450 IT
Manager and Lab Administrator Guide for information on using thisfeature.

Aperio GT 450 1.0.1.8000 Patch Notice, Revision A© Leica Biosystems Imaging, Inc. 2021

Aperio GT 450 1.0.1.8000 Patch Notice

Improvements Included in This PatchThe following improvements were added in this patch.

Component

Case #Description

Controller

27504

The slide on stage check is now part of the scanner homing sequence. Ifa slide is detected during homing, the user is notified through an error
message displayed on the console requesting the user to remove theslide from the stage and restart the homing sequence.

SAM

27505

Java was replaced with OpenJDK on the SAM server. SAM is now
deployed with open source implementation of Java runtime OpenJDK.

Console27471

An issue related to Purple Screen on Scanner Console was resolved.

Aperio GT 450 1.0.1.8000 Patch Notice, Revision A© Leica Biosystems Imaging, Inc. 2021

Aperio GT 450 1.0.1.8000 Patch NoticePart Number/Revision: PN-0024, Revision A
Date: 2 February 2021

Copyright NoticeCopyright © 2021 Leica Biosystems Imaging, Inc. All Rights Reserved. LEICA and the Leica logo are
registered trademarks of Leica Microsystems IR GmbH. Aperio is a trademark of the Leica Biosystems
group of companies in the USA and optionally in other countries. Other logos, product and/or company
names might be trademarks of their respective owners.

This product is protected by registered patents. For a list of patents, contact Leica Biosystems.

User Resources

For the latest information on Leica Biosystems Aperio products and services, please visit
www.LeicaBiosystems.com/Aperio.

DisclaimersUse normal care in maintaining and using the Aperio eSlide Manager servers. Interrupting network
connections or turning off the Aperio eSlide Manager and DSR servers while they are processing data (suchas when they are analyzing digital slides or generating an audit report) can result in data loss.

This manual is not a substitute for the detailed operator training provided by Leica Biosystems Imaging orfor other advanced instruction. Leica Biosystems Imaging Field Representatives should be contacted
immediately for assistance in the event of any instrument malfunction. Installation of hardware should only
be performed by a certified Leica Biosystems Imaging Service Engineer.

Contact Information – Leica Biosystems Imaging, Inc.

Headquarters
Leica Biosystems Imaging, Inc.
1360 Park Center Drive
Vista, CA 92081
USA
Tel: +1 (866)-478-4111 (toll free)Direct International Tel: +1 (760)539-1100

Customer SupportGeneral Information

Contact your local support representativewith any query and service request.

US/Canada Tel: +1 (866) 478-4111 (tollfree)Direct International Tel: +1 (760) 53991100

https://www.leicabiosystems.com/servicesupport/technical-support/

Email: ePathology@LeicaBiosystems.com

Aperio GT 450 1.0.1.8000 Patch Notice, Revision A© Leica Biosystems Imaging, Inc. 2021