Reagents■ Faecal Immunochemical Test (FIT)■ Faecal Calprotectin Test (FCa)

OC-SENSOR FIT Latex Reagent
OC-SENSOR FIT Buffer

OC-FCa Reagent

Faecal Immunochemical Test (FIT)
Faecal Calprotectin Test (FCa)
OC-FCa Control LV1
OC-FCa Control LV2OC-FCa Control LV3OC-FCa Calibrator

OC-FIT Control LV1
OC-FIT Control LV2
OC-FIT Control LV3OC-FIT Calibrator

All reagents are ready-to-use.
OC-SENSOR Sample Diluent is also available.

Specifications

Dimensions

Product Name

OC-SENSOR PLEDIAPrinciple

Latex agglutination immunoturbidimetry

Throughput

320 tests/hour

Sample Loading Capacity

10-sample rack × 20 *200-sample rack-tray (optional)

STAT Sample

Up to 10 samples simultaneously

Sample Dilution

15 times and 250 times

Reaction Cuvette

Semi-disposable acrylic cuvette (11 serial cells × 5 cuvettes)
Automatic cuvette cleaning with wash solution and purified water

Sampling System

Drawn up by sample nozzle(Liquid surface detection, automatic cleaning withwash solution and purified water)

Reagent Dispense

Drawn up by reagent nozzle(Liquid surface detection, automatic cleaning with purified water)

Mixing System

By spinning mixer (automatic cleaning with wash solution and purified water)

Thermostatic System

Reaction wheel: air bath block heater, Reagent housing: block heater

Light Source

LED (wavelength 660 nm)

Light Detector

Photodiode

Data Input

Color LCD touch panel (10.5 inches)

Memory Capacity

100,000 test results

Data Output

Built-in thermal printer, RS-232C, USB

BarcodesRack barcode, sample barcode, reagent barcode,
two-dimensional barcode (optional)

Dimensions

W630mm × D630mm × H560mm

Weight58kg

Power Required

AC100-240V 50/60Hz, 500VA

Applicable standards and laws

The PMDAct, CE, TGA, TFDA

Please use this product after reading the Operation Manual carefully.

Scan to learn more about
OC-SENSOR PLEDIA

680 mm

630 mmSIMPLEM
CO560 mm

SPEED
Q UAL
630 mm

3074 CIB05
August 2022

IT Y

YPACT

FEATURES & PERFORMANCEHigh Throughput Sampling
Improved processing speed,
increased sampling loading capacity

Random AccessFully Automated System

Throughput

Multiple Items Ordering

Full automation

Capable of analysing 320
tests per hour.

Bidirectional communication between
PLEDIA and the host allows for
multi-item order processing in eachsample.

Equipped with the capability
for automatic startup, start,and one-touch shut down.

Set reagentsHOST

Automatic identification ofmeasurement modes

With the number of reagent
slots increased from two tothree, PLEDIA is more efficientfor high throughput population
screening.

Barcodes on racks enable
the measurement mode to beidentified automatically.

Onboard sample capacity

After completing an analysis

Up to 200 samples can beloaded at a time.
Operation with a tray isalso available. *The tray system is optional.

Retest racks enable simple
re-testing.
With a dilute rack, 15-times
and/or 250-times dilution (s) is
automatically performed.

Retest rack

Sample rack detection sensor

Analysis starts automaticallywhen the sample rack is set.

Dilute retest rack

Advanced Usability
Improved screen operability

Measured value

Prozone recognition

Upper limit ofmeasurement

Measurement
range∆Abs T6-8

User-friendly touch panel
10.5 inch LCD touch panel.
The adjustable display angle
improves visibility.

Lower limit of
PRC

Theoretical value

If the ∆Abs at the T6-8
photometric point ofsamples under the upper
limit of measurement isabove the lower limit ofthe PRC, the sample isrecognized as within
the prozone(PRC region).

Prozone of highly-concentrated samples is checked
through the Primary Rate Check (PRC) method.

Squeeze-up system
Real-time information
Real-time information isavailable, including information
on reagents andsample processing time.

Squeeze - up s ystem
Sample nozzle
Sample cup part
FilterQuality control function
X-R control screen,
CV auto calculation, etc.
Massive QC function isavailable.

As sampling bottle is squeezed,
samples are filtered into the sample cup part,and then drawn by the sample nozzle.