FN2114
Pedal Load Cell
SPECIFICATIONS•Ranges 200 to 2500 N (40 to 500 Lbf)

•Aluminum body•

Cable Gland Output•

Accuracy 1% regardless force application point

The FN2114 load cell has been developed for pedal load
testing. Its size allows the sensor to be easily installed on
the pedals of any vehicle or a fatigue/endurance testbench.
Since the forces to be investigated on the brake,
clutch or accelerometer pedals are different, and vary froma small car to a heavy goods vehicle, TE CONNECTIVITY
offers different measurement ranges as standard.

FEATURES•••

Compression onlyIntegrated amplifier optional
Rugged design

APPLICATIONS•••

On-board vehicle equipment
Test bench equipment
Laboratory and researchWith an internal mechanical decoupling system,
accuracy is 1% F.S., regardless of the point of force
application. Integrated amplifier for high level output is
optionally available for all ranges.
A similar pedal load cells with thicker dimension
exists as model FN2570.
With many years of experience as a designer andmanufacturer of sensors, TE CONNECTIVITY often workswith customers to design or customize sensors for specificuses and testing environments.
On request, Instruction documents can beprovided to ease the selection and use of our sensors andprovide helpful tips.

SENSOR SOLUTIONS /// FN2114_en03/2016

Page 1

FN2114Pedal Load Cell

STANDARD RANGES
Ranges in N (FS)

200500

1000

2000

2500

Ranges in lbf (FS)

40100

200

400500PERFORMANCE SPECIFICATIONS (typical values at temperature 23±3°C)
Parameters
Operating Temperature Range (OTR)-20 to 80 °C [-4 to 176° F]

Compensated Temperature Range (CTR)

0 to 60 ºC [32 to 140° F]

Zero Shift in CTR<0.5% FS /50º C [/100° F]

Sensitivity Shift in CTR<1% of reading / 50º C [/100° F]

Over-Range
Without Damage

1.5 x F.S.

Without Destruction

3 x F.S.

Accuracy
Combined non-linearity & hysteresis≤±1% F.S.

Electrical Characteristics
Model
Supply Outage
Sensitivity “FSO”
Zero Offset

44

Input Impedance/Consumption

FN2114

FN2114-A1

FN2114-A2

10Vdc

10 – 30Vdc±15Vdc (±12 to ±18Vdc)±2mV/V4V ±0.2V

5V ±0.2V±5% F.S.

0.5V ±0.2V

0V ±0.2V

700Ω

<50mA50mA

5Output Impedance

700Ω

1 kΩ

Insulation under 50Vdc≥100MΩ

≥100MΩ

1 kΩ

5≥100MΩ

Notes
1. Standard electrical termination: cable gland with Φ3mm shielded cable, 2 meters length
2. Material: aluminum alloy.
3. Protection Index: IP50
4. Other signal output on request
5. Output impedance < 100Ω on request
6. CE conformance according to EN 61010-1, EN 50081-1, EN 50082-1

SENSOR SOLUTIONS /// FN2114_en03/2016

Page 2FN2114
Pedal Load CellDIMENSIONS & WIRING SCHEMATIC (IN METRIC)
Wiring Schematic

Dimension A :28 mm for ranges 200N, 500N
29 mm for ranges 1kN, 1.5kN, 2kN, 2.5kN

SENSOR SOLUTIONS /// FN2114_en

03/2016

Page 3

FN2114Pedal Load CellOPTIONS
L00M : special cable length, replace “00” with total length in meters

ORDERING INFORMATIONFN2114 - A1 - 2KN- /L5M
Options (L05M, ...)
Range in Newton
Amplified version (none, A1 or A2)
Model

NORTH AMERICAEUROPE

ASIA

Measurement Specialties, Inc.,a TE Connectivity Company
Vibration Design Center
32 Journey - Suite 150
Aliso Viejo, CA 92656
United States USA
Tel: 1-949-716-0877
Fax: 1-949-916-5677
CustomerCare.lcsb@te.com

Measurement Specialties (Europe), Ltd.
a TE Connectivity Company
26 Rue des Dames
78340 Les Clayes-Sous-Bois, France
Tel: +33 (0) 130 79 33 00
Fax: +33 (0) 134 81 03 59
CustomerCare.lcsb@te.com

Measurement Specialties (China), Ltd.,a TE Connectivity Company
No. 26 Langshan Road
Shenzhen High-Tech Park (North)Nanshan District, Shenzhen 518057 China
Tel: +86 755 3330 5088
Fax: +86 755 3330 5099
CustomerCare.shzn@te.com

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred toherein might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s StandardTerms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or
misuse of the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

SENSOR SOLUTIONS /// FN2114_en

All Rights Reserved.

03/2016

Page 4