MODEL 4810A ACCELEROMETERSPECIFICATIONSHigh Performance DC Response
Ultra-Stable MEMS Accelerometer
Advanced Temp Compensation
Hermetically Sealed

The Model 4810A is an ultra-stable MEMS
accelerometer in a welded stainless steel package. The
accelerometer offers an amplified signal conditioned
output in ranges from ±2 to ±200g. The model 4810A
incorporates a gas damped silicon MEMS sensing
element that incorporates mechanical overload stops forshock protection to 5,000g and a wide bandwidth fromDC to 2000Hz.

DIMENSIONSFEATURES

±2g to ±200g Dynamic Range
8-36Vdc Excitation Voltage
Hermetically Sealed, Detachable Cable
Integral Cable Option, Model 4810B
Gas Damped MEMS Element
Temperature Compensated

APPLICATIONS




SENSOR SOLUTIONS /// Model 4810A Rev B

Low Frequency Monitoring
Transportation
Flight Testing
Machine Control
Road Vehicle Testing
Trains9/2015

Page 1

MODEL 4810A ACCELEROMETERPERFORMANCE SPECIFICATIONS
All values are typical at +24°C, 80Hz and 12Vdc excitation unless otherwise stated. Measurement Specialties reserves the right to update
and change these specifications without notice.

1ParametersDYNAMIC
Range (g)Sensitivity (mV/g)
Frequency Response (Hz)Natural Frequency (Hz)Non-Linearity (%FSO)
Transverse Sensitivity (%)
Damping Ratio
Shock Limit (g)Residual Noise (µV RMS)
Spectral Noise (µg/Hz)

±2
1000
0-250
700
±1.0
<30.7
5000
600
35

ELECTRICAL
Zero Acceleration Output (mV)
Excitation Voltage (Vdc)
Excitation Current (mA)
Bias Voltage (Vdc)
Full Scale Output Voltage (V)Output Resistance (Ω)
Insulation Resistance (MΩ)
Turn On Time (msec)
Ground Isolation±508 to 36<52.5±2<100>100<100
Isolated from Mounting Surface

Differential

ENVIRONMENTAL
Thermal Zero Shift (%FSO/°C)
Thermal Sensitivity Shift (%/°C)
Operating Temperature (°C)
Storage Temperature (°C)
Humidity±0.004±0.010-55 to 125-55 to 125
Hermetically Sealed, IP67 1

TypicalTypicalPHYSICAL
Case Material
Weight (grams)
Mounting
Mounting Torque

Stainless Steel16
2x #4 or M3 Screws
6 lb-in (0.7 N-m)

Notes
±5400
0-700
800±1.0<30.7
5000
750
38±102000-1000
1000±1.0<30.7
5000
800
75±20
1000-1000
1500±1.0<30.75000
800
132±50400-1000
4000
±1.0
<30.7
5000
800
316±100200-1500
6000
±1.0
<30.7
5000
400
516±200100-1500
8000
±1.0
<30.6
5000
400
1033±5%<1 Typical
Passband
Passband@100Vdc

Mating cable needs to also have minimum IP67 rating and be properly sealed to accel connector in accordance with IEC 60529.

Calibration supplied:

CS-FREQ-0100

NIST Traceable Amplitude Calibration from 20Hz to ±5% Frequency Response Limit

Supplied accessories:

AC-A02285

2x #4-40 (7/16 length) Socket Head Cap Screw and Washer

Optional accessories:

AC-D02669
340A-XXX
343-XXX
121

Triaxial Mounting Block
Cable Assembly, #28 AWG, -54 to +121°C (XXX designates length in inches, 5ft standard)
Cable Assembly, #28 AWG, -40 to +85°C (XXX designates length in inches, 5ft standard)
3-Channel Precision Low Noise DC Amplifier

The information in this sheet has been carefully reviewed and is believed to be accurate; however, no responsibility is assumed for inaccuracies. Furthermore, thisinformation does not convey to the purchaser of such devices any license under the patent rights to the manufacturer. Measurement Specialties, Inc. reserves theright to make changes without further notice to any product herein. Measurement Specialties, Inc. makes no warranty, representation or guarantee regarding thesuitability of its product for any particular purpose, nor does Measurement Specialties, Inc. assume any liability arising out of the application or use of any product or
circuit and specifically disclaims any and all liability, including without limitation consequential or incidental damages. Typical parameters can and do vary in different
applications. All operating parameters must be validated for each customer application by customer’s technical experts. Measurement Specialties, Inc. does notconvey any license under its patent rights nor the rights of others.

SENSOR SOLUTIONS /// Model 4810A Rev B

07/2017

Page 2

MODEL 4810A ACCELEROMETER

2g Range
5g Range
10g Range
20g Range
50g Range
100g-500g Range

Typical Frequency Response
40353025Sensitivity Deviation (%)

2015
1050-5-10-15-20-25-30-35-40
101001000

10000

Frequency (Hz)2g Range
5g Range
10g Range
20g Range
50g Range
100g Range200g Range

Typical Phase Response
4035302520Phase (degrees)

151050-5-10-15-20-25-30-35-40
10100

100010000

Frequency (Hz)SENSOR SOLUTIONS /// Model 4810A Rev B

07/2017

Page 3

MODEL 4810A ACCELEROMETER

ORDERING INFORMATIONPART NUMBERING

Model Number+Range

4810A-GGGG-Cll______Range (0010 is 10g)
l__________Electrical Interface (A; Connector, B; Integral Cable)
Example: 4810A-0010-C
Model 4810A, 10g, ConnectorASIA

NORTH AMERICAEUROPE

Measurement Specialties, Inc.,a TE Connectivity Company
Tel: 800-522-6752
Email: customercare.hmpt@te.com

MEAS France SAS
a TE Connectivity Company
Tel: 800-440-5100
Email: customercare.lcsb@te.com

Measurement Specialties (China), Ltd.,a TE Connectivity Company
Tel: 0400-820-6015
Email: customercare.shzn@te.com

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

All Rights Reserved.

SENSOR SOLUTIONS /// Model 4810A Rev B

07/2017

Page 4