moisture.IQ

Panametrics Moisture
Analyzer
Applications

FeaturesThis multichannel analyzer measures moisture in gases
and non-aqueous liquids and oxygen (optional) in gases.
Designed for permanent installations, it is used in conjunction
with Panametrics Moisture Image Series and M Seriesmoisture probes for industries and applications including:• Up to 6 channels of moisture measurement• Petrochemical• Natural gas• Industrial gas• Semiconductor• Furnace gas/heat treating• Power generation• Air dryer• Pharmaceutical• Aerospace• Each channel can include optional pressure, temperature,oxygen, and other analog inputs• Calibrations traceable to national standards• Rack, bench, and panel, weatherproof and explosion-proof
configurations available• Computer-enhanced response software for abrupt
dry-down applications• Real time touch screen display of six or twelve parameterssimultaneously• Displays trace measurements in the most commonmoisture measurement units• Built-in data logging• USB port for data handling and instrument program updates• Digital capabilities including Ethernet, Modbus, and RS232/485• Analog outputs include recorder outputs andmeasurement relays• Fault condition alarm relay• Built in Henry’s Law constants database for liquid applications

Multifunction
The moisture.IQ is the multichannel, multifunction flagshipmodel in the IQ Series of Panametrics aluminum oxide-based
moisture analyzers. The moisture.IQ measures trace moisture,pressure and temperature in non-aqueous liquids and gases.
It accepts inputs from electrochemical sensors for measuringoxygen concentrations in gases. The auxiliary inputs canaccept analog inputs from any transmitter with 0/4 to 20
mA or -1 to +4 V output, including a variety of Panametrics
process control instruments, this feature makes the moisture.
IQ a true multifunction analyzer, providing cost savings
through system integration.

Bench mount version

Multichannel
For additional cost savings, the moisture.IQ is available withtwo module bays. Each can accept a 1-channel or 3-channel
module. With up to six sets of sensors coming to one set ofelectronics, customers can significantly reduce their cost permeasurement point.

Touch screen display
Designed with an industrial touch screen display, the graphicaluser interface is intuitive enough to negate the need for a user’s
manual (although one is provided)! Six or twelve measurements
can be displayed at one time. Multiple display pages canbe set up for easy scrolling. Color coded status gives you
measurement alarm information at a glance.

Moisture image series probe
The moisture.IQ can accept inputs from the Panametrics MIS
probes and the M Series probes that were available for theMoisture Image Series 1 and Moisture Monitor Series 3 analyzers.
The MIS probes (MISP and MISP2) provide integral moisture,temperature, and pressure inputs. The calibration data for thesesensors is stored digitally. Upon connection to the analyzer, thecalibration data is downloaded to the moisture.IQ.

Weatherproof version

Explosion-proof version

Electronics

Pressure Measurement

Intrinsic Safety

All non-auxiliary inputs are intrinsicallysafe through internal isolation andenergy-limiting circuitry.

InputsTwo module bays are available. Each
module bay can accommodate a
1-channel or 3-channel module. Eachchannel can accept:• 1 Moisture input (MIS Probe or M Probe)• 1 Temperature input(MIS Probe or M Probe)• 1 Pressure input (MIS Probe)• 1 Oxygen input (electrochemical sensor)• 2 Auxiliary inputs

Analog Outputs

2 per available channel

Measurement
Alarm Relays

2 per available channel

Fault Condition
Alarm Relay

1 per meter

Type

Optional transducer built into Moisture
Image Series moisture probes orstandard external pressure transmitter

Full-Scale Ranges
Available• 300 psig (21 bar)
• 500 psig (35 bar)• 1000 psig (69 bar)• 3000 psig (207 bar)
• 5000 psig (345 bar)

Accuracy±1% of full scale

Pressure Rating

Three times the span of the available
range to a maximum of 7500 psig (518 bar)

Oxygen Measurement
TypeElectrochemical sensor

Available Sensor
Ranges / Accuracy

ppb O2 Range• DFOX-1, 0 to 500 ppb/5 ppm/50 ppm,
1/4 VCR± 3% of reading or 25 ppb
ppm O2 Range• DFOX-9, 0 to 1/10/100 ppm, 1/4 VCR• DFOX-2, 0 to 1/10/100 ppm,1/8 compression± 3% of reading or 50 ppb• DFOX-3, 0 to 10/100/1000 ppm,
1/8 compression± 3% of reading or 200 ppb• DFOX-4, 0 to 100/1000/10,000 ppm,1/8 compression± 3% of reading or 2 ppm• DFOX-5, 0 to 50/500/5,000 ppm,1/8 compression± 3% of reading or 1 ppm% O2 Range• DFOX-6, 0 to 5%, 1/8 compression± 3% of reading or 10 ppm• DFOX-7, 0 to 10%, 1/8 compression± 3% of reading or 20 ppm• DFOX-8, 0 to 25%, 1/8 compression± 3% of reading or 50 ppm

Pressure Rating

0.2 to 1 psig (0.07 bar)

Rack mount: 482 x 133 x 357 mm(19.0 x 5.2 x 14.1 in.); 11.2 kg [24.7 lb.]
Bench mount: 440 x 133 x 357 mm(17.4 x 5.2 x 14.1 in.); 10.4 kg [22.9 lb.]
Dimensions (w x h x d)and Weight

Panel mount: 542 x 201 x 357 mm(21.4 x 8.0 x 14.1 in.); 11.3 kg [25.0 lb.]
Weatherproof: 508 x 508 x 229.8 mm(20.0 x 20.0 x 9.05 in.); 24.9 kg (55 lb.)
Explosion-Proof: 590.6 x 590.6 x 304.8 mm(23.25 x 23.25 x 12.0 in.); 113.6 kg (250 lb.)

Moisture Measurement
Type

Panametrics Moisture Image Series andM Series thin-film aluminum oxide probes

Calibration Ranges(Dew/Frost Point)• Standard: 50 to -112°F (+10 to –80°C) withdata from +68 to -166°F (+20 to -110°C)• Ultralow: -58 to -148°F (–50 to –100°C)
with data to -166°F (–110°C)

Accuracy(Dew/Frost Point)• ±3.6°F (±2°C) above -148°F (-100°C)• ±5.4°F (±3°C) below -148°F (-100°C)

Repeatability(Dew/Frost Point)• ±0.4°F (±0.2°C) above -148°F (-100°C)• ±0.9°F (±0.5°C) below -148°F (-100°C)

Operating Pressure

5 μ of Hg to 5000 psig (345 bar) limitedby optional pressure sensor: seepressure sensor ranges

Temperature Measurement
Type

Optional thermistor built intomoisture probe

Calibration Ranges(Dew/Frost Point)–22 to 158°F (–30 to 70°C)

Accuracy±0.9°F (±0.5°C) at –22°F (–30°C)

Copyright 2020 Baker Hughes Company. All rights reserved.
BHCS38681		

(12/2020)

panametrics.com