Rugged quality,precision imaging
DXR140P-HC | DXR140P-HE
DXR100P-HP | DXR75P-HR detectors

Customer driven
leadership in industrial
radiography
Some things don’t change: like the need in non-destructive
testing (NDT) for reliable, high-quality, fast and easy
imaging. But that doesn’t mean we can’t do better.
With the next-generation DXR wireless portable detectors
Waygate Technologies, has developed a new direct
radiography (DR) platform that combines the latest
technical advances with extensive customer insight.
The result is a range of DR detectors that meet the strictest
ISO standards, designed specifically for the demands ofdifferent industrial applications. Rugged inside and out,
they offer the high quality and fine details an inspection
task requires, from corrosion monitoring to weld inspectionand everything in between. Easy to transport and easy tooperate, they offer a complete solution for safe, high quality
imaging, even in the harshest conditions.

High-quality, portable imaging for
inspection of field installations:• Rugged additional shielding for extended lifetime• Easy to transport and handle• Choice of wireless or wired use• Wi-Fi range up to 100 meters• Two hot-swappable batteries, for longer
uninterrupted operation

A new detector portfolio for high-quality
industrial field operation
The new range of DXR detectors provide the high-quality,
efficient imaging solution required for industrial field testing.
The detectors have a reduced thickness that makes them
ideal in situations with limited detector positioning freedom.
Choose wired or wireless use for simplified operation thatmatches your needs under every circumstance.

Rugged inside and out
Dedicated to industrial use, the customer-driven designof the next generation DXR detectors offers a rugged
reliability that perfectly fits the criteria for mobile inspectionof installations.
Each detector comes in a rugged casing that guards
against drops, knocks and even radiation. Additional layersof shielding in the detectors better protect them fromthe harmful effects of radiation, for a longer lifetime andconsistent performance. The IP67 rating ­(immersion up to
1 meter depth) ensures these detectors are suitable for allindustrial environments.
A tailored flight case, pipe support, 50 m outdoor network
cable and shoulder strap makes the package complete,
and ready for safe, easy transportation.

• Dust tight and waterproof• Compliant with prevailing standards,including ISO/ASTM• Minimum backscatter and noise, maximum
image quality• Rhythm Insight RT acquisition and review
analysis software• Powered by Flash! automatic image processing
technology

Designed for high quality,
efficient imaging inharsh field radiography
conditions

DXR140P-HC

DXR100P-HP

Ideal for corrosion monitoring in the oil & gas and energy
industries, the DXR140P-HC detector offers high contrast.
And due to its increased dose sensitivity allows shorter
exposure times and faster throughput.

The DXR100P-HP gives you the high, 100-micron pixel
resolution, so this versatile detector can cover both worlds:
O&G inspections but also weld inspection, supporting ISO
17636-2 class B for weld inspection.

The DXR140P-HC can be used with X-ray as well
as isotopes, and is suited for general radiography
applications, including (but not limited to):

The DXR100P-HP can be used with X-ray as well as
isotopes, suited for a wide variety of applications,
including (but not limited to):• Oil & gas and energy, in-service inspection:• Weld inspection in oil & gas, energy and aviation:

Large size,
high contrast imager
for all-purpose
radiography

Large size,
high performing
imager, covering
most applications- CUI- transport pipelines- valve positioning- complex structures (spool)- WT measurement- boiler & pressure tubes- pipe support- fuel pipes- boiler tubes- pressure vessels and storage tanks• Casting inspection• Shipyard weld inspection• Aviation MRO• Oil & gas and energy, in-service inspection:• Military and security- CUI & WT measurement• Structure inspection:- valve positioning- concrete, bridges, supports, …

- pipe support• Science, art and archeology• Casting inspection• Power line inspection, GIS• Aviation MRO• Military and security • Structure inspection:- concrete, bridges, supports, …
• Science, art and archeology• Power line inspection, GISNEWDXR75P-HR

DXR140P-HE

Small size,
highest resolution
imager for criticalapplications

Large size,
high contrast imager
for high-energy
radiography

The DXR75P-HR gives you the high, 75-micron pixel resolution
required to distinguish fine details for critical applications.
The detector supports ISO 17636-2 class B for weld inspection,offering precision imaging that meets the most stringent
standards.

The DXR140P-HE is the ideal portable detector designed forhigh-energy applications (Isotopes). Due to its optimized
internal shielding, we prevent low-energy scattered radiation,
having a negative influence on image quality and electronics
lifetime.

The detector’s narrow width makes it ideal for imaging
in situations with limited detector positioning freedom.

The DXR140P-HE can be used with isotopes andhigh-energy X-rays (above 450kV), and is suited forgeneral radiography applications, including (but notlimited to):

The DXR75P-HR is suited for critical applications,including (but not limited to):• Weld inspection in oil & gas, energy and aviation:• Oil & gas and energy, in-service inspection:- CUI- transport pipelines- valve positioning- complex structures (spool)- WT measurement- boiler tubes- pipe support- fuel pipes- boiler tubes- pressure tubes• Casting inspection- pressure vessels and storage tanks• Aviation MRO• Shipyard weld inspection• Military and security• Structure inspection:- concrete, bridges, supports, …
• Science, art and archeology• Power line inspection, GISReady to inspect!
Each DXR is a complete solution, with:• Detector, mounted in a ruggedized case
• Accessories for optional external front shielding• Transport casep

• Wi-Fi and wired accessories

tra• Four batteries with chargerrs• Workstation with Rhythm Insight RT acquisition andanalysis software, powered by Flash!

Shouldeoor cable

Rhythm Insight RT advanced digital tools• Intuitive user interface for out-of-the-box
scanning and image processing50• Touch user interface for optimal operation
under all conditions

outd

Built on 15 years of experience and proven knowledge, Rhythm
Insight RT offers a powerful, scalable, yet easy-to-use NDT
acquisition and analysis platform, dedicated to industrial
requirements, and powered by Flash!

m(164 feet)• Application-guided automation• Powerful image quality validation tools• DICONDE compliant

Technical specificationsNEW

DXR140P-HC

DXR100P-HP

Pixel pitch (micron)

140100Detector size(incl. armor case & battery pack)460 x 568 x 27mm (18,1 x 23 x 1,1”)

460 x 568 x 27mm (18,1 x 23 x 1,1”)

Detector active area

353 x 424 mm (14 x 17”)348 x 425 mm (14 x 17”)

Detector weight(incl. armor case & battery pack)9,2 kgs (20 lbs)

9,2 kgs (20 lbs)

Scintillator

Gadolinium oxysulfide (GOS)

Gadolinium oxysulfide (GOS)

Environmental protection detector

IP67

IP67

Operating temperature-20 °C to +50 °C-20 °C to +50 °C

Drop test: 1 m

PassedPassedGrayscaleCommunication

16 bitWired or wireless (802.11n/ac)

Power supply

DC +24 V (max 0.8 A)• Wired: power by SCU with tether interface cable• Wireless: powered by battery pack (2 x 3.100 mAh)

Operating time

Up to 7 hours for imaging, 9 hours on standby

Radiation hardness
Compliance> 1 kGy
ISO, AST, ASME, JIS and KS• DICONDE compliant (ASTM E3147)• Spatial Resolution Measurement conforms to ASTM E2597

Useful Assescoires

pp

ort

Pipe support
Curved support on the armour case that allows smooth positioningof the detector on pipes and ensures that the dda is always in thecentre.
Shoulder strap.
A simple but neat solution that allows easy transport of the detector.
Equipped with quick-lock for easy use.

PipeseTransport case (flight case).
Customised transport case offering full protection during transportwith protection class IP67.

surt

Ca

Outdoor communication cable
A cable solution consisting of a 50 metre (164 feet) UV protected andshielded flexible network cable that allows direct communication
between the detector and the computer. This provides the user withthe most stable and fastest data communication.

Tran

spoTripod
Allows completely flexible and easy positioning around the object.
Two versions available: for large and small detectors.

od

Ethercon cabel (7 m - 22,9 feet)
Allows the supplied tether cable to be extended to 14 metres forcable communication between SCU (WiFi access point) and detector.

Tr

DXR75P-HR

DXR140P-HE

Pixel pitch (micron)

75

140

Detector size(incl. armor case & battery pack)268 x 358 x 38mm (10,6 x 14,1 x 1,5”)

460 x 568 x 27mm (18,1 x 23 x 1,1”)

Detector active area

168 x 225 mm (7 x 9”)

353 x 424 mm (14 x 17”)

Detector weight(incl. armor case & battery pack)4,5 kgs (9,9 lbs)8 kgs (17,6 lbs)

Scintillator

Gadolinium oxysulfide (GOS Fine)

Gadolinium oxysulfide (GOS)

Environmental protection detector

IP67

IP67

Operating temperature-20 °C to +50 °C-20 °C to +50 °C

Drop test: 1 m

PassedPassedGrayscale

16 bitCommunication

Power supply

Wired or wireless (802.11n/ac)
DC +24 V (max 0.8 A)• Wired: power by SCU
with tether interface cable• Wireless: powered by battery pack(2 x 3.100 mAh)

DC +24 V (max 1.0 A)• Wired: power by SCU
with tether interface cable• Wireless: powered by battery pack(2 x 3.100 mAh)

Operating time

Up to 7 hours for imaging, 9 hours on standby

Radiation hardness> 1 kGy

Compliance

ipISO, AST, ASME, JIS and KS• DICONDE compliant (ASTM E3147)• Spatial Resolution Measurement conforms to ASTM E2597

Powered by Flash!
intelligent image processing
technology
One Click

When it comes to revealing details and reliably make good/fail decisions,Waygate Technologies Flash! digital radiography image processing software
sets new standards for defect detection and inspection efficiency.

Combining 25+ years of experience and patents with
next-generation technology, Flash! automatically, quickly
and consistently optimizes your digital radiographs.
You get exquisite image quality and comfortable reading, with a faster,smoother workflow that enhances your productivity, maximizes your resourcesand gives your customers peace of mind.

Next-level imaging
Selecting the right imaging hardware has a major impact on image quality –but the software you choose is just as important!

SHARP

Flash! uses innovative, proven and leading image processing technologythat offers high image quality. Images are made consistent, regardless of theoperator, for easier reading and more confident reviewing.

A layered perspective
With Flash! you see more, with a clear vision of both high- and low-density
areas, in a single image. Intelligent image processing automatically minimizes
noise while maximizing details.

SMART

You can read images quickly, viewing all the layers in a glance without manualadjustments. No unsharp masking or edge enhancement creating artefacts.
No over- or under-shooting around higher-density areas. No window levelingrequired. Flash! provides optimal image quality even whit minimal contrast. Theinvisible becomes visible, and the material no longer hides radiographic secrets
from you.

Good to go!

SIMPLE

Flash! is easy to learn and use, and doesn’t require specialized training. Operator
independent, it automatically adjusts to variations in density, materials,geometry, radiation quality, etc. The software saves time and effort for operator
and inspector, so you get more out of your human and material resources.

Copyright 2022 Baker Hughes Company. This material contains one or more registeredtrademarks of Baker Hughes Company and its subsidiaries in one or more countries. All
third-party product and company names are trademarks of their respective holders.
BHFF-DXR 140-75P-brochure-EN (03/2022)

waygate-tech.com