MODEL 4835A ACCELEROMETERSPECIFICATIONSDIMENSIONSMEMS Triaxial Accelerometer
Ultra-Stable, DC Response
Light Weight Welded Titanium<2% Total Error Band

The Model 4835A is an ultra-stable MEMS
triaxial accelerometer in a rugged, welded Titanium
package.
The
variable
capacitance
MEMS
accelerometer incorporates integral temperaturecompensation that provides a stable output over a wide
temperature range from -55°C to +125°C. Offered inranges from ±2 to ±200g, the model 4835A
accelerometer also provides a wide bandwidth from DC
to 1000Hz for both static and dynamic measurements.

FEATURES

±2g to ±200g Dynamic Range
Amplified Output
8-36Vdc Excitation Voltage
Hermetically Sealed, Detachable Cable
Gas Damped MEMS Element
Temperature Compensated
Low Transverse Sensitivity

APPLICATIONSSENSOR SOLUTIONS /// Model 4835A Rev A

Low Frequency Monitoring
Transportation
Flight Testing
Machine Control
Road Vehicle Testing
Trains

07/2017

Page 1

MODEL 4835A ACCELEROMETERPERFORMANCE SPECIFICATIONS
All values are typical at +24°C, 80Hz and 12Vdc excitation unless otherwise stated. Measurement Specialties reserves the right to update
and change these specifications without notice.

1Parameters
DYNAMIC
Range (g)Sensitivity (mV/g)
Frequency Response (Hz)Non-Linearity (%FSO)
Transverse Sensitivity (%)
Damping Ratio
Shock Limit (g)Residual Noise (µV RMS)

±2
1000
0-200
±1.0
<20.7
2000
550

ELECTRICAL
Zero Acceleration Output (mV)
Excitation Voltage (Vdc)
Excitation Current (mA)
Full Scale Output Voltage (V)Output Impedance (Ω)
Insulation Resistance (MΩ)
Turn On Time (msec)
Ground Isolation±508 to 36<15±2<100>100<100
Isolated from Mounting Surface

Differential

ENVIRONMENTAL
Thermal Zero Shift (%FSO/°C)
Thermal Sensitivity Shift (%/°C)
Operating Temperature (°C)
Storage Temperature (°C)
Humidity
Total Error Band (%)±0.004±0.010-55 to 121-55 to 121
Hermetically Sealed, IP67 1<2 (RSS of Non-Linearity, Thermal Zero Shift, and Thermal Sensitivity Shift

TypicalTypicalPHYSICAL
Case Material
Weight (grams)
Mounting
Mounting Torque

Titanium22
2x #4 or M3 Screws
6 lb-in (0.7 N-m)

Notes
±54000-600±1.0<20.75000700±102000-800
±1.0
<20.7
5000
750±30670-800
±1.0
<20.7
5000
750±50400-800±1.0<20.75000
750±100200-1000
±1.0
<20.7
5000
800

±200
100-1000
±1.0
<20.6
5000
800

±5%

Passband@100Vdc

Mating cable needs to also have minimum IP67 rating and be properly sealed to accelerometer connector in accordance with IEC 60529.

Calibration supplied:

CS-FREQ-0100

NIST Traceable Amplitude Calibration from 20Hz to ±5% Frequency Response Limit

Supplied accessories:

AC-D02995

2x #4-40 (7/8” length) Socket Head Cap Screw and Washer

Optional accessories:

371-120
121

Mating Cable Assembly, (XXX designates length in inches, 10ft standard)
3-Channel Precision Low Noise DC Amplifier

The information in this sheet has been carefully reviewed and is believed to be accurate; however, no responsibility is assumed for inaccuracies. Furthermore, thisinformation does not convey to the purchaser of such devices any license under the patent rights to the manufacturer. Measurement Specialties, Inc. reserves theright to make changes without further notice to any product herein. Measurement Specialties, Inc. makes no warranty, representation or guarantee regarding thesuitability of its product for any particular purpose, nor does Measurement Specialties, Inc. assume any liability arising out of the application or use of any product or
circuit and specifically disclaims any and all liability, including without limitation consequential or incidental damages. Typical parameters can and do vary in different
applications. All operating parameters must be validated for each customer application by customer’s technical experts. Measurement Specialties, Inc. does notconvey any license under its patent rights nor the rights of others.

SENSOR SOLUTIONS /// Model 4835A Rev A07/2017

Page 2

MODEL 4835A ACCELEROMETER

2g Range
5g Range
10g Range
20g Range
50g Range
100g-500g Range

Typical Frequency Response
40353025Sensitivity Deviation (%)

20151050-5-10-15-20-25-30-35-40
10100

1000

10000Frequency (Hz)2g Range
5g Range
10g Range
20g Range
50g Range
100g Range
200g Range

Typical Phase Response
4035302520Phase (degrees)

151050-5-10-15-20-25-30-35-40101001000

10000

Frequency (Hz)SENSOR SOLUTIONS /// Model 4835A Rev A

07/2017

Page 3

MODEL 4835A ACCELEROMETER

ORDERING INFORMATIONPART NUMBERING

Model Number+Range

4835A-GGGG-C
l______Range (0010 is 10g)
Example: 4835A-0010-C
Model 4835A, 10g Range

ASIA

NORTH AMERICAEUROPE

Measurement Specialties, Inc.,a TE Connectivity Company
Tel: 800-522-6752
Email: customercare.hmpt@te.com

MEAS France SAS
a TE Connectivity Company
Tel: 800-440-5100
Email: customercare.lcsb@te.com

Measurement Specialties (China), Ltd.,a TE Connectivity Company
Tel: 0400-820-6015
Email: customercare.shzn@te.com

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015TE Connectivity Ltd. family of companies

All Rights Reserved.

SENSOR SOLUTIONS /// Model 4835A Rev A

07/2017

Page 4