REDEL 2P
HIGH VOLTAGElll®

®

A LEMO Group CompanyIntroducing 3 New High Voltage
configurations (2, 5 & 8 contacts)
Test Voltage higher than 10 kV AC
IEC 60601-1 (3rd Ed.) Compliant
2 MOPP / 2 MOOP (Means Of
Patient / Operator Protection)

REDEL introduces its latest plastic High Voltage connector. These 2, 5 and 8 contacts configurations are the ideal solution for medicaland industrial applications.
Building on the 2P series legacy in medical applications,these new insert configurations are the ideal solutions forlatest high voltage PFA (Pulsed-Field Ablation) and PEF(Pulsed Electrical Field) catheter technologies.
Besides specific applications, new medical standards foruser safety become mandatory (IEC60601-1, 3rd Ed.). Thisrequires extended creepage and clearance distance that
can’t be matched by commodity multipin connectors.
The inherent safety, high voltage specification (up to 10 kVAC), lightweight design, and IP66 rating make it an ideal
candidate for Industrial and Test/Measurement segments
and applications such as automotive testing and electriccars.

Front insert (Fig 1)

Back insert (Fig 2)

Heatshrink
Contact

BENEFITSlTest Voltage higher than 10 kV ACl

3 High Voltage configuration (2, 5 & 8 contacts)l

Water-resistant models available up to IP66
when matedlSeveral keyings / colour codingslMeets electrostatic discharge: IEC 60601-1 (3rd Ed.)
(15 kV)l

RoHS compliant / UL Recognition

FEATURESl

Push-Pull quick and secure lockinglSterilizable by gas or by steamlBlind mating (3 guides)lScoop & Touch prooflLightweight and compact design for space savingslTemperature range -50°C to 170°Cl

UL94 V-0 certified as auto-extinguishablel

For cable range up to ø 9.2 mmInsulator

Air Clearance distance

Creepage distance ** The contribution to the creepage distances of any groove or air gap
less than 1 mm wide shall be limited to its width

PSU and PEI enables
multiple sterilization cycles
Colour front ring codes
Water-resistant
IP66 available

APPLICATIONS

Scoop & Touch proof
3 High Voltage configurations
2, 5 & 8 contacts
3 guides and several colour flanges

MedicalIndustry

Test & Measurement

Automotive

Product safety notice & disclaimers:
Please read and follow all instructions specified on https://www.lemo.com/en/about-lemo/product-safety-notice-and-disclaimers

®

REDEL 2P
HIGH VOLTAGE®

A LEMO Group Company

PART NUMBERING SYSTEM

Plug

C A B

. H05. G L

A . C 92 GVariant
Z = collet nut
for fitting a bend relief
Front ring colour

Model
Keying: B, C, D, H or J
Contact configuration

Collet: 52 = (cable ø 3.2 mm - 5.2 mm)
72 = (cable ø 5.3 mm - 7.2 mm)
92 = (cable ø 7.3 mm - 9.2 mm)

Number of contacts
Outershell+ collet nut:

G = grey PSU
N = black PSU
B = white PSU
E = orange PSU

J = yellow
R = redS = orange
V = green

G = greyN = blackA = blueB = white

Cable fixing type: C = cable collet

S1) = grey PEIContact typeA = male to solder

L = female to solder 2)

Insulator: L = PEEK for solder contacts

Free socket

C R B

. H

05. G L

L . C 92 GVariant
Z = collet nut
for fitting a bend relief
Front ring colour

Model

G = greyN = blackA = blueB = white

Keying: B, C, D, H or J
Contact configuration

Collet: 52 = (cable ø 3.2 mm - 5.2 mm)
72 = (cable ø 5.3 mm - 7.2 mm)
92 = (cable ø 7.3 mm - 9.2 mm)

Number of contacts
Outershell+ collet nut:

G = grey PSU
N = black PSU
B = white PSU
E = orange PSU

J = yellow
R = redS = orange
V = green

Cable fixing type: C = cable collet

S1) = grey PEIContact typeA = male to solder 2)

L = female to solder

Insulator: L = PEEK for solder contacts

Fixed socket

C K B . H 05. G L

L GFront flange colour
G = greyN = blackA = blueB = white

Model
Keying: B, C, D, H or J

J = yellow
R = redS = orange
V = green

Contact typeContact configuration

L = female to solder

A = male to solder 2)

Number of contacts
Outershell: G = grey PSU
N = black PSU
B = white PSU
E = orange PSU

S1) = grey PEI

Insulator: L = PEEK for solder contacts

Note: 1) for extensive steam sterilization we propose polytherimide ULTEM® (PEI).
2) model available only with H and J keying (inverted contacts).

(©) LEMO All right reserved

REDEL 2P
HIGH VOLTAGE®

®

A LEMO Group Company

PRODUCT RANGE
Standard models
Fixed sockets

Free sockets

CAB

CKB

CRB

CAB

CLB

CRB

Fixed socket

Free socket

CNB

CSB

Straight plugs

IP66 Water-resistant models
Straight plug

CFB

KEYING
00000

Keying (plug front view)

ReferenceContact type for plugContact type for socket
Availability

BC

D

H

J

male
femalemale
femalemale
femalefemalemale

female
male

•

••

•

•

Material

Colour

G GreyN Black
B WhiteE Orange

PSU

Temperature

Availability-50° / +150°C

•­•
•
•

Material
PEI

ColourTemperature

Availability

SGrey-50° / +170°C•­Note: for extensive steam sterilization we propose polytherimide ULTEM® (PEI).

FRONT FLANGES / RING COLOUR		GNABJRSV		 greyblackbluewhiteyellow

redorangegreen

•

•••••

•

•

Ref.

Ref.

OUTERSHELL

Note: •­ Standard / • On demand(©) LEMO All right reserved

®

REDEL 2P
HIGH VOLTAGE®

A LEMO Group Company

INSERT CONFIGURATION
Female solder contacts

Print (elbow)

Test voltage (kV rms)1)
Contact-contact

Air clearance min 2) (mm)Creepage distance min (mm)Rated current (A)3Print (straight)

2Crimp2Solder3

Crimp bucket ø (mm)4Solder bucket ø (mm)1Contact ø (mm)1

Number of contacts

4Contact type

Reference

Male solder contacts

H02

2

0.70.8–

•

–––

12.7

8.90

11.0

H05

50.70.8–

•

–––

11.6

7.96

8.0

H08

80.7

0.8–

•

–––

10.3

7.42

6.5

Note: 1) The use of potting type Epoxy* (insert config. H02, H05 and H08) or adhesive-lined heatshrink tubes over each termination (only H02 and H05) is necessary to guarantee
the indicated Test voltage.
2) The use of potting type Epoxy* (insert config H02, H05 and H08) or heatshrink tubes over each termination (only H02 and H05) is necessary to guarantee the indicated
Air Clearance and Creepage distance values.		 Adhesive-lined heatshrink tube can improve these values. IEC 60601-1 Compliant, 2 MOOP / 2 MOPP (Means Of Operator/Patient Protection).
* Minimum characteristics: Dielectric constant (εr, 50Hz, 25°C ) >4 and Volume resistivity (ρ, 25°C) < 1014 Ωcm

EXPLODED VIEW (PLUG AND FREE SOCKET)

321

1. Bend relief

2. Collet nut

3. Cable collet

45

4. Insulator + Contacts5. Outershell +colour front ring

A bend relief absorbs the force that may be exerted on cables.
These are designed for plugs and free sockets with cable collet and nut.

Part NumberøA

LDimensions (mm)Bend reliefCable øAL		max. min.

GMA.2B.040.DG
GMA.2B.045.DG
GMA.2B.050.DG
GMA.2B.060.DG
GMA.2B.070.DG
GMA.2B.080.DG

4.0
4.5
5.0
6.0
7.0
7.8

3636
36
36
36
36

4.5
5.0
5.5
6.5
7.7
8.8

4.0
4.5
5.0
6.0
7.0
7.8

GMA.2B.040.RG
GMA.2B.045.RG
GMA.2B.051.RG
GMA.2B.057.RG
GMA.2B.063.RG
GMA.2B.071.RG
GMA.2B.080.RG

4.0
4.5
5.1
5.7
6.3
7.1
8.0

41
41
41
41
41
41
41

4.4
5.0
5.6
6.2
7.0
7.9
9.0

4.0
4.5
5.1
5.7
6.3
7.1
8.0

Material

Temperature range
in dry atmospherein water steam

TPU(ThermoplasticPolyurethane)-40°C, +80°C–Siliconeelastomer
VMQ-60°C, +200°C+140°C

Note: the last letter «G» of the part number indicates a grey colour, see the adjacent table and replace letter «G» by the letter of the colour required.
(©) LEMO All right reservedRef.

ColoursABGJM
NRSV

bluewhite
greyyellow
brown
black
red
orangegreen

CAT.2P.REN.P1121, Data subject to change, printed in Switzerland, updated July 2023

BEND RELIEF