KRAUTKRÄMERDMS Go+

The ultra-portable, ultra-powerful
thickness gauge.

Featuring intuitive, easy-to-use arrow-keypad control, powerful data management
and the latest industrial electronics to provide accurate, reliable and comprehensive
thickness inspection data.
The Krautkrämer DMS Go+ thickness gauge that can be easily converted into acomprehensive flaw detector with a simple software upgrade.

Krautkrämer DMS Go+
Designed and developedwith the user in mind
Operational excellence
The Krautkrämer DMS Go+ offers a comprehensive, handheld solution to thickness measurement, data recording and data management ina wide range of applications and environments.

High performance thickness measurement•	Zero cross measurement technique for highmeasurement stability and reliability•	Automatic gain control for excellent repeatability
and corrosion monitoring.
•	Built-in temperature compensation for accurate
measurement up to 540°C (1000°F)•	Multiple calibration and zeroing modes for repeatableaccuracy.
•	Multiple measurement modes, including thickness,A-scan, B-scan, Min/max and differential.

•	On-board data recorder, with capacity of hundreds ofthousands of thickness readings, with storage of A-scan,
B-scan and MicroGrid attachments

The Krautkrämer DMS Go+ is suitable for thickness
measurement in a wide variety of applications
throughout the industrial and process spectrum.
It is especially applicable for corrosion measurement
and monitoring, even on coated components andstructures and at high temperatures.

•	Data can be organized in pre-set, custom or advanced
file structures.

Typical applications include:

High capacity data recorder and compatibility
with powerful data management systems•	Data transfer is by SD card or via USB port to PC.
Data can be transferred in various file formats to allow
easy integration with user data management systems.
•	Compatible with UltraMate and UltraMate Lite data
management programs to allow for comprehensive data
analysis and documentation.

Ease of use•	Intuitive arrow-keypad for positive digital control ofparameters•	One hand operation and one-hand, menu-directed
calibration process•	A “Flip” function allows use by both left-handed andright-handed operatives.
•	A large, 800x480 pixel, display screen, ergonomically
sized to reduce eye-strain, which can be adjusted toprovide optimum visibility in various ambient lightconditions.
•	Small size, lightweight (870g, 1.9lb), robust construction
to IP67 for operation in harsh environments.

2A wide range of applications•	Battery allows up to 10 hours operation and can be
re-charged on- or off-board.

Oil & Gas
• Inspection and monitoring of corrosion in tubes,vessels and tanks• Measurement of remaining wall thickness
through paint coatings
Power Generation• Inspection of complex geometry tubes• Monitoring of boiler efficiency by measuring oxide
scale in boiler tubes with special probe OSS-10Aerospace• Maintenance checks
Metals Industry• Thickness measurement of austenitic materials
Optional applications software such as TopCOAT
technology, also allows measurement of coating
thickness as well as metal thickness, while Auto-V
measurement enables thickness to be measured
on components with unknown sound velocities,
without the need for a calibration block.

A simple software upgrade adds
a comprehensive and versatile
flaw detector to the Krautkrämer
DMS Go+The DMS Go+ uses the same operating platform andhardware as the state-of-the-art USM Go+ portable flaw
detector. This offers high Near Surface Resolution todetect flaws near to the surface, as well as a wide Pulse
Repetition Frequency range, allowing it to be used forinspecting forged parts as well as welds.
An up-graded DMS Go+ means that personnel now needto carry only one instrument to perform accurate andreliable thickness measurement and flaw detection.

Technical Specifications of Krautkrämer DMS Go+Display

5 inch,800 × 480 pixels,
108 × 65 mm (W x H),> 200 cd/m2

Size (W x H x D)

175 × 111 × 50 mm (W × H × D)

Weight

850 g with battery

Protection classIP 67

Operating temperature0 – 55 °C

Battery

Li-Ion, rechargeable, > 8 hours operation timePower adapter / charger

100 – 240 V AC, 50/60 Hz

Probe connector

Dual Lemo-00 (T/R)

PC interface

Micro USB

Memory card

SD-Card 16 GB max

Datarecorder

100.000 readings per file. Multiple files can be stored on SD card
8 file formats, Attachment of A-Scan, B-Scan and micro grid

Pulser120 – 250 V, Spike wave, Automatically matched to probe

Puls Repetition Frequency

4, 8 or 16 Hz selectable

Receiver110 dB dynamic, automatic gain control, Manual –high, -low, -auto

Measurement range

0,4 – 14.000 m/s (0.01 – 551 “)

Unitsmm, inch, µs

Digital Display resolution

0,01 mm or 0,1 mm (0.001” or 0.01”) selectable

Measurement techniques

Zero crossing, IP to 1st echo, multi echo, TopCoat, Auto-V

Calibration

One-point, Two-point
Auto or Manual On-block and Off-block Zero
Automatic V-Path correction

Display mode

Thickness and A-Scan, Temperature corrected thickness, B-Scan,
Min/Max capture, Differential

ComplianceEN 61010,EN 61326-1, EN 12668, ASTM E 1324, E317, ANSI/NCSL Z 540-1-1994
MIL-STD 45662A, MIL-STD 2154, EN 15317

Copyright 2020 Baker Hughes Company. This material contains one or more registeredtrademarks of Baker Hughes Company and its subsidiaries in one or more countries. All
third-party product and company names are trademarks of their respective holders.
BHFF-20219EN (07/20)

waygate-tech.com