FN2317
HandBreak Load Cell

SPECIFICATIONS•Ranges 500 N and 1000N (100 and 200 lbf)

•Compatible with most handbrakes•“Easy to mount” through clamping collars•

High accuracy ±0.5 % FS

The FN2317 is designed with the purpose of allowing
quick and easy installation when measuring forces applied
to the handle of the hand brake in automobiles. Clamping
collars facilitate the prompt and uncomplicated installationof the load cell under the hand brake. As such, the FN
2317 is compatible with most models of hand brakes.

FEATURES•••

Compact and ergonomic design
Tension measurements
High accuracy regardless force
application point

APPLICATIONS•••On-board equipment testing
Production quality control
Laboratory and Research

SENSOR SOLUTIONS /// FN2317_en

Through careful placement of metallic strain gages
inside, the sensors provides accurate measurements
regardless of the point of application of force.
With many years of experience as a designer andmanufacturer, TE CONNECTIVITY offers solutions toautomotive industry and can supply standard or custom
sensors for specific uses and testing environments.
On request, Instruction documents can beprovided to ease the selection and use of our sensors andprovide helpful tips.

03/2016

Page 1

FN2317
HandBreak Load Cell

STANDARD RANGES
Ranges in N (FS)

500

1000

Ranges in lbf (FS)

100200PERFORMANCE SPECIFICATIONS (typical values at temperature 23±3°C)
Parameters
Operating Temperature Range (OTR)-20 to 80° C [-4 to 176° F]

Compensated Temperature Range (CTR)

0 to 60º C [32 to 140° F]

Zero Shift in CTR<0.5% F.S. / 50º C [/100° F]

Sensitivity Shift in CTR<1% of reading / 50º C [/100° F]

Over-Range
Without Damage

1.5 x F.S.

Without Destruction

3 x F.S.

Accuracy
Combined non-linearity & hysteresis≤±0.5% F.S.

Electrical Characteristics
Model
Supply Outage
Sensitivity “FSO”
Zero Offset

44

Input Impedance/Consumption

FN2317

FN2317-A1

FN2317-A2

10Vdc

10 – 30Vdc±15Vdc (±12 to ±18Vdc)±1.5mV/V

4V ±0.2V

5V ±0.2V±5% F.S.

0.5V ±0.2V

0V ±0.2V

350 to 700Ω

<50mA50mA

5Output Impedance

350 to 700Ω

1 kΩInsulation under 50Vdc≥100MΩ

≥100MΩ1 kΩ

5≥100MΩNotes
1. standard electrical termination: cable gland with Φ3mm shielded cable, 2meters length
2. Material: stainless steel.
3. Protection Index: IP50
4. Other signal output on request
5. Output impedance < 100Ω on request
6. CE conformance according to EN 61010-1, EN 50081-1, EN 50082-1

SENSOR SOLUTIONS /// FN2317_en03/2016

Page 2

FN2317
HandBreak Load CellDIMENSIONS & WIRING SCHEMATIC (IN METRIC)
Wiring Schematic

SENSOR SOLUTIONS /// FN2317_en03/2016

Page 3

FN2317
HandBreak Load CellOPTIONSA1 : Amplified Tension output with unipolar power supply
A2 : Amplified Tension output with bipolar power supply
SC : LEMO connector output instead of standard cable gland
L00M : special cable length, replace “00” with total length in meters

ORDERING INFORMATIONFN2317 - A1 - 1KN - /L5M
Options (L05M, SC, ...)
Range in Newton
Amplified version (none, A1 or A2)
Model

NORTH AMERICAEUROPE

ASIA

Measurement Specialties, Inc.,a TE Connectivity Company
Vibration Design Center
32 Journey - Suite 150
Aliso Viejo, CA 92656
United States USATel: 1-949-716-0877
Fax: 1-949-916-5677
CustomerCare.lcsb@te.com

Measurement Specialties (Europe), Ltd.
a TE Connectivity Company
26 Rue des Dames
78340 Les Clayes-Sous-Bois, France
Tel: +33 (0) 130 79 33 00
Fax: +33 (0) 134 81 03 59
CustomerCare.lcsb@te.com

Measurement Specialties (China), Ltd.,a TE Connectivity Company
No. 26 Langshan Road
Shenzhen High-Tech Park (North)Nanshan District, Shenzhen 518057 China
Tel: +86 755 3330 5088
Fax: +86 755 3330 5099
CustomerCare.shzn@te.com

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred toherein might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or
misuse of the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies All Rights Reserved.

SENSOR SOLUTIONS /// FN2317_en

03/2016

Page 4