Genius 1025
Nitrogen & air generator for PerkinElmer Instruments
Part Number : 99-9999

Service Kit : 99-9999

Your local gas generation partner

Description
Producing a single output of LC-MS grade nitrogen (up to 15 L/min) and air (up to 35 L/min) specifically to meet the requirementsof PerkinElmer’s QSight Triple Quad 110 and 210 LC-MS/MS
systems, the Genius 1025 has been designed using membrane
technology to produce LC-MS grade purity and an internal air
dryer to produce a clean supply of dry air.
The Genius 10 series models are compressor-based systems
Peak manufacture utilizing membrane technology to ensure highperformance and supply of analytical instrument grade nitrogen.

Applications)

1 x PerkinElmer QSight Triple Quad 110)

1 x PerkinElmer QSight Triple Quad 210

Key Features
600 mm / 23.7”)

Self-contained solution with integrated compressors,no need for an external air supply)

Highly economical source of nitrogen/dry air with low
lifetime running costs)

Easy to use, gas at the push of a button)

Wheels make it easy to manoeuvre generator aroundthe lab)

12 month comprehensive on-site warranty600.00 mm

713 mm / 28.1”

713.00 mm

750 mm / 29.6”

750.00 mm

713.00 mm

713 mm / 28.1”

Copyright © Peak Scientific Instruments Ltd 2022 — Genius 1025 Data Sheet Rev 6 RSID 2789 EN —01/02/2022

Technical Specifications
Max Flow

Genius 1025
Up to 15 L/min Nitrogen & up to 35L/min Dry Air

Min/ Max Pressure

Up to 80 psi Nitrogen & up to 110 psi Dry Air

Gas outlets

2 x 1/4” BSPP

Max Relative Humidity

80% Non-Condensing

Max Altitude

2000m

Particles<0.01µm

Phthalates

NONE

Suspended liquids

NONE

Operating Temperature

5°C (41°F) to 30°C (86°F)

Electrical Requirements

230V±10% 50 / 60 Hz 9A

Power Consumption<1,265VA

Heat Output

3,925 BTU /Hr

Generator Dimensions (H x W x D)

713 x 600 x 750 mm
28.1 x 23.7 x 29.6’’

Generator Weight108.5 kg / 239.3 lbs

Shipping Weight

137 kg / 302.1 lbs

Ordering Information
Part Number

3301714Annual Service
Complete Maintenance Plan†

visit: www.peakscientific.com/ordering

Noise level expressed as SPL (Sound Pressure Level) measured at 1m from source in a reverberant chamber

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that your investment continues tobe [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you
the reassurance of instant access to worldwide technical support and priority on-site response in the untimely event of abreakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North AmericaTel: +1 866 647 1649
Fax: +1 978 608 9503

Peak Scientific’s Quality
Management System
conforms to:ISO 9001:2015

Product Certifications

For a full list of our worldwide
office locations, please visit:www.peakscientific.com/offices
Web: www.peakscientific.com
Email: discover@peakscientific.com

Copyright 2022 Peak Scientific - Genius 1025 Data Sheet Rev 6 RSID 2789 EN - 01/02/22