iMINI LCD User Guide
S i m p l e a n d S m ar t

S I NG LE U SE TE M PE RA TU RE DA T A LE GG ER

User Guide
Rev2.1

Cryopak Verification Technologies, Inc. reserves the right to change this specification without noticeiMINI LCD User Manual | Document Rev. 2.1 | ©Copyright 2015

P a g e |1

iMINI LCD User Guide
S i m p l e a n d S m ar t

Contents

1.

Introduction ........................................................................................................................................ 3

2.

Safety Instructions ............................................................................................................................. 3

3.

Product Code ..................................................................................................................................... 44.

Size / Dimensions .............................................................................................................................. 55.

Images ............................................................................................................................................... 6

6.

Specifications ..................................................................................................................................... 77.

Display ............................................................................................................................................... 8
7.1.

Hardware Firmware info. ............................................................................................................... 8

7.2.

Start/ Stop ..................................................................................................................................... 8

7.3.

Bookmark ...................................................................................................................................... 9

7.4.

Other LCD functions ...................................................................................................................... 9

7.5.

Information on the iMINI LCD display ........................................................................................... 98.

Software & Software dependencies ................................................................................................ 109.

Pre-program form ............................................................................................................................ 11

10.

Quick Guide..................................................................................................................................... 12

11.

Alarm Settings Options ................................................................................................................... 13

12.

Download, Save and Export Data ................................................................................................... 13

13.

Warranty information ....................................................................................................................... 14

14.

Recycling of electronic devices ....................................................................................................... 15

15.

Contact details................................................................................................................................. 15

Cryopak Verification Technologies, Inc. reserves the right to change this specification without noticeiMINI LCD User Manual | Document Rev. 2.1 | ©Copyright 2015

P a g e |2

iMINI LCD User Guide
S i m p l e a n d S m ar t

1. Introduction
Cryopak Verification Technologies, Inc. offers new and advanced, temperature data loggers.
New iMINI LCD is now available for single use temperature data monitoring solutions. iMINI LCD datalogger offers,Multi-function display



START & STOP button

Waterproof pouch with IP rating IP67



Economical single use data loggerReport output in .pdf, .csv & .txt format (with ConsolePlus software)

Slightly larger than credit card; lays flat (refer to dimensions)

Durable lightweight; weighs less than 23 gramsEasy programming with Console Plus software or pre-programmed

Pre-programmed to customer specifications

High accuracy

Large memory capacity 8K (7808 records)



21 CFR compliance (when used with UAP1 enabled ConsolePlus)

Compact profile

4 alarm thresholds

The new iMINI LCD is now available for storage and transit temperature monitoring solutions. It iscompletely food safe with IP rating 67. The iMINI LCD is an internal temperature sensor data logger withfour alarm conditions, start and stop buttons, and much more…

2. Safety Instructions
Under no circumstances should heavy force be applied to your iMINI LCD logger. Applying
heavy force to any part of your iMINI LCD could result in logger malfunction, LCD damage
and/or injury.
Your iMINI LCD logger should only be operated within the parameters specified in thetechnical data discussed within this user manual. A failure to follow these instructions
could result in you iMINI LCD logger malfunctioning and cause permanent damage to theunit.

1 UserAdmin Package

Cryopak Verification Technologies, Inc. reserves the right to change this specification without noticeiMINI LCD User Manual | Document Rev. 2.1 | ©Copyright 2015

P a g e |3

iMINI LCD User Guide
S i m p l e a n d S m ar t
iMINI LCD loggers are vulnerable to damage when contact with water takes place. Please
do not puncture or tear the plastic sleeve before placing your iMINI LCD logger in suchconditions. Ensure to verify that there is no damage to sleeve before using.
Your iMINI LCD logger is not to be subjected to a naked flame. Exposures to suchconditions may damage the logger or possibly cause an explosion of the battery.

Do not operate your iMINI LCD logger with a low battery.

Do not attempt to repair or modify any part of this logger. Such actions will result in a lossof warranty. All repairs are to be made by an official Cryopak Verification Technology
service center.

The iMINI LCD logger has a one trip warranty. For more information refer page 14.

Battery life depends greatly on the temperature environment and recording frequency of
your iMINI LCD logger.

3. Product CodeThe product code for CVT products is represented in a modular fashion to allow distributors andcustomers the possibility of easily recognizing and ordering a variety of different products.
Note: For ease of communication, the dashes within the product code may be omitted.

Format:[XX] - [YY] - [T] - [M] - [Options][XX] - [YY] - [T] - [M] - [Options] - [P / T]

(Example: ML-ST-S-8-L)(Example: ML-ST-S-8-L-P)

The initial letters “XX” indicates product family as detailed below:

ML

New generation CRYOPAK iMINI LCD SINGLE USE Intelligent MINI Temperature Data
Logger with LCD

The field “YY” represents the sensor combination code with following options:ST

Single Trip logger (Internal Temperature Sensor only)

The next field “T” identifies the temperature measurement range:

S

Internal Temperature Sensor with measuring range -40°C to +80°C“M” identifies the Memory capacity of the logger. The following options are available:

Cryopak Verification Technologies, Inc. reserves the right to change this specification without noticeiMINI LCD User Manual | Document Rev. 2.1 | ©Copyright 2015

P a g e |4iMINI LCD User Guide
S i m p l e a n d S m ar t

8

8K memory, iMINI LCD can store 7,808 data points“Options” identifies if the logger has additional options

L

Data logger with an LCD display“P” identifies if the logger has additional options

P
T

Pharma compliant (Title 21 CFR Part 11)
NIST Traceable calibration certificate at 3 set points -20°C, 25°C & 60°C

4. Size / Dimensions

Cryopak Verification Technologies, Inc. reserves the right to change this specification without noticeiMINI LCD User Manual | Document Rev. 2.1 | ©Copyright 2015

P a g e |5iMINI LCD User Guide
S i m p l e a n d S m ar t

5. Images
Front View

Rear View

Product label

Side view

SensorBattery

Perforated cut
To tear the sleeve
START button

Hole toTie the unit
USB Port

LCD
Cryopak Verification Technologies, Inc. reserves the right to change this specification without noticeiMINI LCD User Manual | Document Rev. 2.1 | ©Copyright 2015

STOP button
P a g e |6

iMINI LCD User Guide
S i m p l e a n d S m ar t

6. Specifications
Description

Specifications

Product Code

ML-ST-S-8-L
ML-ST-S-8-L-P (Pharma compliant) 21CFR part 11

Memory Options

8K, 7808 readings

Program Interval

5 seconds to 17 hours

Time Accuracy±1 hour per year

Sensors

Internal

Temperature Range-40 ฀C to +80 ฀C (-40 ฀F to +176 ฀F)

Temperature Accuracy±0.5 ฀C (-40 ฀C to -10 ฀C), ±1 ฀F (-40 ฀F to +14 ฀F);
±0.3 ฀C (-10 ฀C to +80 ฀C), ±0.6 ฀F (+14 ฀F to +176 ฀F)

Resolution

0.1 ฀C (0.1 ฀F)

Sensor Response TimeT90 of 5 minutes in moving air

File Format

PDF, CSV, TXT & CVT

Alarm Thresholds

4 thresholds; 2 Red LEDs, 2 Blue LEDs

Alarm Configuration

HIGH HIGH, HIGH, LOW & LOW LOW

Bookmark

Yes

Preprogram Option

Factory programmed

Start Option

Push button and/or timer

Auto Restart

N/A

Start Delay

Yes, 1 minute to 99 days

Stop Option

Yes, STOP button (can be disabled)

Size

68mmx37mmx5mm (without sleeve)
81mmx52mmx5mm (with sleeve)

Weight23 gCase Material

Plastic sleeve

Battery

3.0V

IP Rating

IP 67

Warranty

1 TripCalibration

A 3 point NIST traceable calibration certificate is available as a pdf
download with the ML-ST-S-8-L-P productAccuracy Certificate

A mass certificate is available upon request for the ML-ST-S-8-L

Other Certification

ISO9001:2008, CE, RoHS

Shelf Life

1 YearInterfaceUSB

SoftwareConsolePlus

Default File Format

CVT (data can be exported as PDF, CSV, TXT with ConsolePlus)

Security

N/A

Cryopak Verification Technologies, Inc. reserves the right to change this specification without noticeiMINI LCD User Manual | Document Rev. 2.1 | ©Copyright 2015

P a g e |7

iMINI LCD User Guide
S i m p l e a n d S m ar t

7. Display
The iMINI LCD data logger has a multifunction LCD display
with a refresh rate per second. Below is the list of displayed
text on the LCD and the functions:

7.1.

iMINI LCD

Hardware Firmware info.

Firmware & information

The iMINI LCD data logger is also capable of displaying
firmware version, memory size and number of activatedsensors, by simply pressing the STOP button for 3-5 seconds,
when it is not logging.


Firmware

The iMINI LCD will cycle firmware version first. In this case,the firmware version is 1.54.

Important Note:

The firmware version of iMINI LCD data logger is being updated regularly for features; your device may
have a different one.
Memory size & sensors
 Then, memory size & No. of active sensors.
Value before the decimal point denotes memory size of iMINILCD (8K)
Value after decimal point means number of active sensors; inthis case there is one active sensor on the iMINI LCD

7.2.

Start/ Stop

Start/Stop depends on how the loggers has been configured:
Configured?

LCD display
Before startWithidelayi

READE

Withouti
Delayi

READE

WithiTimeri
startii

Withidelayi&i
Timeristarti

READE IN
DELAE

READE IN
DELAE

Action required

After Start LCD display

Push & hold
START button 3-5
sec.
Push & hold
START button 3-5
sec.
iMINI LCD starts
automatically withtimer
Auto-start: iMINILCD starts
automatically withtimer
Manual start: Push
& hold START
button 3-5 sec.

3 broken lines, then IN DELAE text on screen.
When delay time is met, latest temperature on
display & text LATEST flashing
3 broken lines, then latest temperature on
display & text LATEST flashing
When start time is met, latest temperature on
display & text LATEST flashing
When start time is met, latest temperature on
display & text LATEST flashing
3 broken lines, then IN DELAE text on screen.
When delay time is met, latest temperature on
display & text LATEST flashing

Cryopak Verification Technologies, Inc. reserves the right to change this specification without noticeiMINI LCD User Manual | Document Rev. 2.1 | ©Copyright 2015

P a g e |8

iMINI LCD User Guide
S i m p l e a n d S m ar t

Withoutidelayi&iTimeristarti

7.3.

READE IN
DELAE

Auto-start: iMINILCD starts
automatically withtimer
Manual start:
Push & hold
START button 3-5
sec.

When start time is met, latest temperature on
display & text LATEST flashing
3 broken lines, then latest temperature on
display & text LATEST flashing

Bookmark

To place a bookmark in the iMINI LCD logger when recording, press & hold START button for 3-5
seconds. The text MARKED, along with the temperature, will be displayed. Upon download, the marked
readings will have a ‘*’ symbol and on the graph, a star icon will be placed. There is no limit on
bookmarked readings.

7.4.

Other LCD functions

Before Program (when purchased un-programmed units): LCD will display: STOP
Before Program (when purchased pre-programmed units): LCD will display: READE
To view highest, lowest & average temperature:
Press START button quickly,
once LCD will cycle through
HIGHEST, LOWEST &AVERAGE temperature
To view time out of specification:
Press START button quickly, twice LCD will cycle through time above & below
specification, display will show text ABOVE & BELOW
with the information of duration in minutes.

7.5.

Information on the iMINI LCD display

LCD display

Occurrence Description

READE

Constant

iMINI LCD is programmed and READY to start with START button

READE IN
DELAE

Constant

iMINI LCD is programmed with delay and will start after designated delay
when push START button is detected

IN DELAE

Constant

iMINI LCD is programmed with Date/Time, will start recording whendate/time is met

STOPConstant

iMINI LCD is not logging or has finished the last trip

Constant

Battery icon will appear when battery is low

Cryopak Verification Technologies, Inc. reserves the right to change this specification without noticeiMINI LCD User Manual | Document Rev. 2.1 | ©Copyright 2015

P a g e |9iMINI LCD User Guide
S i m p l e a n d S m ar t
HIGH ALARM

Constant

Temperature went above HIGH alarm conditions

HIGH ALARM

Text Flashing Temperature went above HIGH HIGH alarm conditions

LOW ALARM

Constant

LOW ALARM

Text Flashing Temperature went below LOW LOW alarm conditions

LATEST

Text Flashing iMINI LCD is logging

MARKED

Flash Once

A bookmark is placed (by pressing START button when logging for 3-5
seconds)

HIGHESTLOWESTAVERAGE*ConstantHighest, Lowest & Average temperature of the entire trip will cycle on
display when START button is pressed once

Temperature went below LOW alarm conditions

When START button is pressed twice to display time above/below
specifications

HOURS

฀

*Constant

Celsius฀*ConstantFahrenheit* Text or stmbol displat next to occurrence

8. Software & Software dependencies
All new products, including iMINI LCD from Cryopak Verification Technologies, will be supported byConsolePlus. The software has built-in auto update feature; user will be prompted to upgrade when newrelease is available.
Setup file can be downloaded from http://72.55.148.15/CVT/ConsolePlusSetup.zip
For installation and program/download information refer to ConsolePlus user Manual.
Software dependencies
ConsolePlus can be installed on the following platforms:
Windows XPWindows VistaWindows 7

Windows 8 & 8.1



Windows 10

Important Note:
To view or save pdf manually or automatically any pdf viewer application (for example Adobe reader(latest Version) must be installed on the computer.

Cryopak Verification Technologies, Inc. reserves the right to change this specification without noticeiMINI LCD User Manual | Document Rev. 2.1 | ©Copyright 2015

P a g e | 10

iMINI LCD User Guide
S i m p l e a n d S m ar t

9. Pre-program form

Cryopak Verification Technologies, Inc. reserves the right to change this specification without noticeiMINI LCD User Manual | Document Rev. 2.1 | ©Copyright 2015

P a g e | 11

iMINI LCD User Guide
S i m p l e a n d S m ar t

10.

Quick GuideiMINI LCD
Pre-programmed

LCD display
showing:“READY”

Three
Broken lines
Appear

Press
&
Hold
START
ButtonFor 3 to 5
seconds

LATEST
Temp
Displayed
On LCDiMINI LCD
starting

When logging

Press
&
Hold
STOP
ButtonBased onConfigurationOr
If Manually
StoppedFor 3 to 5 seconds
To manually stop
(if stop button enabled during programming)

Tear off
The sleeve
AcrossThe dotted
Lines

STOP text on LCD
Alarm status is constantly displayed on LCD

Using thumbPush
Left side,
Sliding theBoard along
Revealing theUSB port

Connect USB port of iMINI LCD temp data logger
to computer running ConsolePlus software
To download, view and analyze the data&
To export in various formats (PDF, CSV and TXT)

Cryopak Verification Technologies, Inc. reserves the right to change this specification without noticeiMINI LCD User Manual | Document Rev. 2.1 | ©Copyright 2015

P a g e | 12

iMINI LCD User Guide
S i m p l e a n d S m ar t

11.

Alarm Settings Options

HH (HIGH HIGH) enable
Triggers an alarm when the temperature is greater than the HIGH HIGH specification that has beenrecorded. The text ‘HIGH ALARM’ will start blinking on LCD.
H (HIGH) enable
Triggers an alarm when the temperature is greater than the HIGH specification that has been recorded.
The text ‘HIGH ALARM’ will be displayed on LCD (not blinking)
L (LOW) enable
Triggers an alarm when the temperature is lower than the LOW specification that has been recorded. The
text ‘LOW ALARM’ will be displayed on LCD (not blinking)
LL (LOW LOW) enable
Triggers raise an alarm when the temperature is lower than the LOW LOW specification that has beenrecorded. The text ‘LOW ALARM’ will start blinking on LCD.
Consecutive delay before alarm (1 to 254)
A preset alarm will trigger when conditions goes out of specification: ‘consecutive alarm delay’ has been
met. For example if the value is set ‘5’ in this field, as soon as 5 consecutive readings are out ofspecification, logger will trigger an alarm, if the consecutive readings are 4 or less, there will be no alarm.
Total delay before alarm (1 to 254)
The preset alarm will trigger when conditions goes out of specification: when ‘total alarm delay’ has been
met. For example if the value is set ‘5’ in this field, as soon as 5 readings (in any order) are out ofspecification, the logger will trigger an alarm. For example, if 2 readings were above or below spec andthen readings come within the specifications followed by another 3 readings being out of specification
then, once the total of 5 readings are above or below spec, there will be an alarm.

12.

Download, Save and Export Data

ConsolePlus must be installed on computer. Under the options, select auto download, auto save anddesired file format to save the data automatically. When the iMINI LCD will be connected to the computerthe data will be downloaded and saved automatically as selected in options.
For more information download ConsolePlus user manual.
Data from multiple loggers can be multi-linked by first reading or by date and time for evaluation. Multilinked files, data can be exported in CSV and TXT format, by manually selecting the option ‘save as’,when the files are multi-linked.

Cryopak Verification Technologies, Inc. reserves the right to change this specification without noticeiMINI LCD User Manual | Document Rev. 2.1 | ©Copyright 2015

P a g e | 13

iMINI LCD User Guide
S i m p l e a n d S m ar t

13.

Warranty information(Clause from CVT Terms & Conditions)

9.1 CVT’s warranties in respect of the iroducts are:(a) The warranties and conditions implied bt the Uniform Commercial Code
Article 2, iart 3 312(Warrantt of Title and Against Infringement), 313-2(b) (sale bt description) and 314-2, 315 (merchantable
qualitt); or(b) A specific warrantt or iroduct specification included in the Order Confirmation or on the CVT website www.crtopak.com.
9.2 The warrantt in subclause 9.1 is to the exclusion of all other warranties, conditions and liabilities wherever expressed or
implied and whether arising in contract, court or bt statute or otherwise.
9.3 No representation in relation to the iroducts shall be binding on CVT unless in writing and signed bt CVT or one of its
Directors, or included in either of the warranties detailed in clause 9.1.
9.4 CVT will not be liable for ant personal injurt, propertt damage, consequential or contingent loss or damage caused throughthe negligence or otherwise of CVT, its servants or agents or rising out of the goods being defective or otherwise not in accordancewith ant warrantt given.
10. Specific Warrantt irovisions
10.1 Warrantt is understood as “return to base”. You will be responsible for freight/taxes and duties back to CVT; CVT will pat
the return freight back to You.
10.2 iroduct Warrantt is twentt four months on all products, except for the Single Trip loggers where the warrantt term is limited
to a single trip to be performed within the 24 month period. This does not include batteries.
10.3 The warrantt does not cover:(a) RH calibration later than 1 tear after production, or if the unit has been subject
to environmental conditions outside those specified.
(b) Willful damage, mistreatment, misuse or abuse of the goods.
(c) Loss or damage caused bt ingress of moisture unless ordered with immersion rating;(d) Batteries(e) Circumstances where the unit has been modified from CVT specifications.
(f) Exposure of the logger to temperatures outside the specified storage temperature, or operating temperature.
(g) Exposure to environmental conditions outside the specified conditions.
(h) Circumstances where the printed circuit board has been removed or tampered with.
10.4 In the case of a warrantt claim CVT will repair the iroduct or, at its option supplt an equivalent replacement.
10.5 In some circumstances, where a specific return is authorized, CVT, mat permit the use of its Federal Express account number
for returning iroduct. Such permission is valid ONLY for that authorized shipment. CVT will not accept ant freight charges forgoods that have been returned without its express permission.
10.6 You mat return iroduct to CVT for accuract testing, if there are reasonable doubts as to the iroducts overall accuract.
10.7 If a iroduct is returned within the warrantt period, a Traceabilitt Certification will be performed. Should the logger read
within the specifications, a Traceabilitt Certificate will be issued and You will be charged with the cost and the cost of return
freight. If the logger does not pass the certification, ie: the logger is at fault, the logger will be replaced or repaired. The new or
repaired logger will be issued with a Traceabilitt
Certificate free of charge.
10.8 If the iroduct is returned outside the warrantt period, a Traceabilitt Certificate will be issued and charged regardless of theoutcome of the test.
10.9 CVT will not pat for ant performance tests undertaken bt ant outside organization, without prior approval

Cryopak Verification Technologies, Inc. reserves the right to change this specification without noticeiMINI LCD User Manual | Document Rev. 2.1 | ©Copyright 2015

P a g e | 14

iMINI LCD User Guide
S i m p l e a n d S m ar t

14.

Recycling of electronic devices

Some parts in products from Cryopak Verification Technologies consist of recyclable materials, but others
should not be disposed of in household waste. To avoid pollution, we kindly ask you to adhere to national
policies and regulations concerning waste disposal and recycling. iMINI LCD data loggers must be
returned to your distributor for disposal (European Standard EN 50419:2005).

15.

Contact details

If you require further information regarding CRYOPAK Verification Technologies, Inc. products
please contact us at:USA
Mailing Address:
Office:Phone:
Fax:
Email:PO Box 309, Buchanan, VA 24066
120 Parkway Drive, Buchanan, VA 24066+1-540-254-1433+1-732-346-9200 Ext. 150+1-540-254-2433
techsupport@cryopak.com

CANADA
Phone:
Fax:
Email:+1-514-324-4720+1-514-773-5966+1-514-324-9623
support@cryopak.com

FRANCE
Address:

Phone:
Email:TCP Cryopak France
1 Rue Berthelot
Z.I. La Maine
76150 Maromme+33 2 32 82 59 65
techsupport@cryopak.com

You can find additional information regarding all of our products on our website:www.cryopak.com

http://www.cryopak.com/en/verification-products/

Cryopak Verification Technologies, Inc. reserves the right to change this specification without noticeiMINI LCD User Manual | Document Rev. 2.1 | ©Copyright 2015

P a g e | 15