2016年度

量子ビームサイエンスフェスタ
Techno AP Co., Ltd.
2976-15 Mawatari, Hitachinaka-city, Ibaraki 312-0012, Japan
Phone: +81-29-350-8011, FAX: +81-29-352-9013
Web: http://www.techno-ap.com
E-mail: order@techno-ap.com

弊社は茨城県ひたちなか市に拠点を置き放射線・物理計測用モジュールや検出器など自社で設計・開発・製造・販売する国内でも数少ない精密測定機器メーカーです。

放射光施設や放射線に関連した研究を行う大学や施設などから数多くの引き合いを頂いており市場のニーズを反映した新製品の開発やさまざまな課題に対し積極的に取り組んでおります。

4element SDD for X-ray
XSDD50-04-25
JAPANMADE

Window-less

For Soft X-ray
Measurable from low energy
region to hundreds of eV

65mm2 collimated to 50mm2

Count Rate1Mcps or more

Energy Resolution

Surface of Detector
Angle degrees 25・244eV@5.9keV Mn KαPeaking Time: 0.25μs
・130eV@5.9keV Mn KαPeaking Time: 1μs

4element SDD system for X-ray
XSDD50-04-25
JAPANMADE

検出器
背面部分

Data measurement device

Function・Histogram
・List-mode
・Wave
・FAST Output
・ROI-SCA Output
・Quick-Scan

APU504GbE

4CH 100Msps 14bit

APU504GbE
4CH 100Msps 14bit

Quick-Scanモードとは、時間分解測定を行うもので外部トリガタイミング（最小10ms）間隔でその間のヒストグラムデータをPC側に送信し、PC側で連続してHDDにデータを保存するモードです。
QXAFS測定に最適です。
Quick-Scanデータ例
Event#1
Event#2

CH1 4096ch

CH2 4096ch

CH3 4096ch CH4 4096ch

CH1 4096ch

CH2 4096ch

CH3 4096ch CH4 4096ch：Event#N

CH1 4096ch

CH2 4096ch

データ収集用アプリケーション画面(185eV@5.9keV ピーキングタイム0.25μs)

CH3 4096ch CH4 4096ch

APN504X GbE for Quick XAFS
● Quick scan● Interface
Preamp out signals
4GATE

4

Event#N(10*Nms)
MultiElements
Detector

10ms(min.) interval
4096ch * 16bit *4CH / Event
Giga bit Ether TCP/IP and UDP
NIM Power Supply
APN504 #1
IP : 192.168.1.11

Giga bit Ether:
APN504 #2
IP : 192.168.1.12

4APN504 #3
IP : 192.168.1.13
4APN504 #4
IP : 192.168.1.14
3APN504 #5
IP : 192.168.1.15

TimingSupply

APN504XGbE

APN504X GbE for Quick XAFS
● Data format

Event#1(10ms)

CH1 4096ch

CH2 4096ch

CH3 4096ch

CH4 4096ch

Event#2(20ms)

CH1 4096ch

CH2 4096ch

CH3 4096ch

CH4 4096ch

CH1 4096ch

CH2 4096ch

CH3 4096ch

CH4 4096ch

Irradiation Test on Synchrotron
Target (Mn), ICR 1.1Mcps, OCR 1.0Mcps,FWHM 194eV@5.9keV MnKα 0.15μsPT

7 element SDD system
for fluorescence X-ray● Element area

455mm² (65ｍｍ² ｘ 7 elements)● Effective area
350mm² (65mm² Collimated to 50mm² x 7 elements)● Function
Histogram, List, Waveform, ROI-SCA● ADC
4CH 100Msps 14bit● Energy Resolution
244eV@5.9keV MnKαPeaking time 0.25μs, 1000kOCR● Power Supply for SDD-200V, ±5V, +3.3V● Interface
Ethernet (TCP/IP)

XSDD50-07
DSP modules & Power supply

7 element SDD system
for fluorescence X-ray

400k ICR

185eV@5.9keV,Peaking Time 0.25μs

Digital Signal Processing for X-ray● Energy resolution[ SSD with 19 elements]
139eV@5.9keV, 6μs Peaking Time
144eV@5.9keV, 4μs Peaking Time
250eV@5.9keV, 0.25μs Peaking Time
[ SDD ]
125eV@5.9keV, 2μs Peaking Time● Throughput
Over 1Mcps● ModeHistogram, List
● Option
TTL output of ROI-SCA
Measurements boards
for 19 elements of SSD

APV8004X

Digital Signal Processing for X-ray

Trapezoidal filter

Screenshotof Application

APV8016 Digital Signal Processor
Analog Input

16CH, Input Impedance 1kΩ, LEMO00 series connector

ADC

Sampling 100MSPS, Resolution 14bit, Input range ±1V

Coarse Gainx2, x4, x10, x20

Fine Gainx0.5～x1.5 *Setting by PC

ADC GAIN

8K, 4K, 2K, 1K, 512, 256ch

Trapezoidal Filter
Digital SignalProcessing
External terminal*Setting by PC

0.1～12μs （0.01step）
Baseline Restorer, Pileup Rejecter, and CFD etc.
All parameter can set by PC.
Output filter waveform, Input clock, Input GATE, Input VETO,
Input clear,
2 CH for function enhancement

Interface

TCP/IP or UDP, 100BaseT

External
dimensions

VME6U 20(W) x262(H) x187(D)Unit 300(W) x56(H) x335(D)Total weight

VME6U About 470gUnit About 3300g

APV8016

APV8016 Digital Signal Processor

Application
Histogram Mode

Outputted DAC
Upper ：PreampLower: Trapezoidal Filter

APV8016 Digital Signal Processor◆ Resolution[Energy] 1.70 keV @ 1.33 MeV,[Time] Resolution 0.625 ns (minimum unit)◆ Throughput
100kcps and over
◆ ModeHistogram, List, Waveform
Maximum transfer rate of the List mode160Kcps (using 1 CH)◆ Multifunction
Spectroscopy amp, Timing filter amp, CFD,Input and Filter waveform output DAC◆ Option
Coincidence, Rise Wave
◆ Software
Application and Instruction manual of Hard and SoftwareAPV8016 Digital Signal Processor

4element SDD for X-ray
Detector
All element area

SDD detector 50mm2, Window-less or AP3.3260mm2 (65mm2 x 4 element)

Active area

200mm2 (65mm2 collimated to 50mm2 x 4 element)

Function

Histogram, List, Waveform, ROI-SCA

ADC Sampling
Energy Resolution

4CH 100MSPS 14bit
244eV@5.9keV MnKα(Peaking time 0.25μs, 1000kOCR)

SDD Power-200V, ±5V, +3.3V

Throughput

Max 150kcps: 2us Peaking timeMax 1000kcps: 0.15us Peaking time

Interface

Ethernet (TCP/IP)

Power Supply
OptionNIM-BIN Support, VME Power Supply
Z-axis movement mechanism, UHV bulb etc.

Vacuum capable<10⁻3 Pa

Flange type

ICF114

Accessory

Application and ManualXSDD50-04

4element SDD for X-ray
XSDD50-04

Detector selectable
・Active Area
30mm² or 50mm²
・Window
Be, AP3.3, Window-less【Use cases】
Transport with bellows corresponding to the vacuum

4element SDD for X-ray
XSDD50-04
Recommendable Measurement
Module

APN504XGbE

High-Speed ADC（100MHz・14Bit）
Digital signal processing by【Use cases】
Transport with bellows corresponding to the vacuum

Highly-integrated FPGA

Measurement Data・Histogram・List・Quick Scan

Single element SDD for X-ray
Detector

SDD detector 50mm2, Window-less or AP3.3

Element Area

65mm2

Active Area

65mm2 collimated to 50mm2

Function

Histogram, List, Waveform, ROI-SCA

ADC Sampling
Energy Resolution

100MSPS 14bit
125eV@5.9keV 2us Peaking time
150eV@5.9keV 0.15us Peaking timeSDD Power-200V, ±5V, +3.3V

Throughput

Max 150kcps: 2us Peaking timeMax 1000kcps: 0.15us Peaking time

Interface

Ethernet (TCP/IP)

Power Supply
OptionNIM-BIN Support, VME Power Supply
Z-axis movement mechanism, UHV bulb etc.

Vacuum capable<10⁻3 Pa

Flange type

ICF70

Accessory

Application and ManualXSDD50-01WL

Single element SDD for X-ray
XSDD50-01WL

Detector selectable

Spectrum of Soft X-ray

・Active Area
30mm² or 50mm²
・Window
Be, AP3.3, Window-less

C (277eV)
O (525eV)【Use cases】
Gate valve and Angle valve Transportwith bellows corresponding to the vacuum

Single element SDD for X-ray
XSDD50-01WL

Recommendable
Measurement Module

APU101X
DIGITAL SPECTROSCOPY【Use cases】
Gate valve and Angle valve Transportwith bellows corresponding to the vacuum

within・1CH HV Power Supply・Pre-amp Power・MCA

APU101 X-ray spectra meter● Detector type

SDD, SSD● Throughput
Over 1Mcps● Power supply
HV(+/-4000V), Preamp● Energy resolution[SSD]
139eV@5.9keV PT6us
250eV@5.9keV PT0.5us[SDD]
125eV@5.9keV PT2μs
145eV@5.9keV PT0.5μs● Interface
TCP/IP

Front

Rear
APU101

APN101X X-ray spectra meter● Detector type

SDD, SSD● Throughput
Over 1Mcps● Power supply
HV(+/-4000V), Preamp● Energy resolution[SSD]
139eV@5.9keV PT6us
250eV@5.9keV PT0.5us[SDD]
125eV@5.9keV PT2μs
145eV@5.9keV PT0.5μs● Interface
TCP/IP

APN101X

APV8102-14 MWPSAGb◆ ADC

2CH, 14bit, 1GHz Synchronism Sampling◆ Memory

1GB (512MB+512MB)×2◆ Time Resolution

Coarse: 2ns | Fine: 7.8ps◆ Throughput

1Mcps or more / CH
◆ Analyze Mode

Waveform acquisition, List (TDC+QDC etc.)◆ Analyze Functions(Digital) CFD, TDC, DC, PSA etc.
◆ Interface

1000BASE-T (over 1M List event)

APV8102-14

APV8102-14 MWPSAGb
Discrimination of γ and neutron(Vertical axis: Waveform fall time integral calculus level
Horizontal axis: Fall time integral calculus level/ All waveform integral calculus level)*Capable number of channel change may occur
due to the signal processing custom and selected options.

Application Waveform acquisition

APV8102-14 MWPSAGb
Analog signal input
typeTrigger

CFD

QDC

PSA

DET

TDC

Time

EnergyFast, Slow

APV8102-14MW
PSAGb

Controller(CAMC, VME, etc)

DET

Wave-Dump, List-Data(Time, Energy)
EternetAnalog signal
input range±3V (Zin:50Ω, GAIN x 1) (LEMO connector x 2)
*Customizable maximum ±4V capable

Analog offsetadjustment±2V (12bit)*Customizable ±20mV to±4V capable

Analog gain switch
Analog signal rise time
Outside in/output
signal terminal(TTL level)
Interface

Unified functions of each modules

PMT anode signal, Fast-NIM signal etc.

Dimension and Weight

Environmental
conditions
Electricity consumption
Accessory

X1/x3*Customizable under input range limit x 10 capable
Under 1ns (@Gain x 1)
CLK input, CLK output, GATE input, VETO input
CLR input, OR output (LEMO connector x 6)
*Customizable switch in/output signal

Ethernet (TCP/IP) 1000BASE-T
VME1width 20mm(W) x 262mm(H) x 187mm(D), 540g
Temperature 5 to 25℃+5V(2.5A), +12V(0.6A), -12V(0.3A)
Application and Manual

USB-MCA4
USB bus power, List of 4CH corresponding MCA● Channel

4CH● Static Dead Time
1.5μs
● Throughput
50kcps and more
● ADC Gain
16384, 8192, 4096, 2048, 1024, 512ch● ModeSpectrum, List
● PowerUSB bus power● BodyLightweight, Compact aluminum case
● Interface
USB2.0● Data transfer
List data 100kcps/sec and more
● Software
Driver & Application

APG7400A

USB-MCA4General configuration example
DETECTOR#1AMP#1DETECTOR#2AMP#2DETECTOR#3AMP#3DETECTOR#4AMP#4USB Data （Spectrum, List）
USB bus power

LIST Data
Event #1 Empty(4bit)
Event #2 Empty(4bit)Time(44bit)Time(44bit)

CH(2bit)
CH(2bit)

PHA(14bit)
PHA(14bit)

CH(2bit)

PHA(14bit)：

Event #N Empty(4bit)Time(44bit)* Data Size 8Byte/Event, Time 40ns/bit

KD-3052 PSD System
GATENET

3He-PSD System*Multi PSD detector use

Ethernet
3He-PSD

8 tubes / 1 Set

AMP97

NEUNET08
PowerSupply＋5V

PSD board

T0 Signal

PC
Another Power Supply Acceptable*Setting: ±12v, ±5V, 3.3V, 1.8V, 1.2V

KD-3052 PSD System
High Energy Accelerator Research Organization developed this board.
This is based on technical specification [KENS-DAQ-012]
which Bee Beans Technologies (BBT) Co., Ltd. shows.

We have a regular license from BBT and produce it.

The neutron scientific facilities (KENS) of High Energy Accelerator Research
Organization develop an experiment to realize structure analysis and this of thematerial as one of the neutron dispersion studies and a measuring system.
This measuring system is called NEUNET system(Neutron Position Sensitive Model Detector System).
This data handling board is used for this NEUNET system.

NUENET
Analog Digital Convertor Processing and Data Processing board
for Output signal of 3He Position Sensitive Proportional Counter

NEUNET08

NUENET
ITEM
Product Name / Model
Analog Input Signal

T0 Input Signal

Input Connector

External Dimensions

Power Supply

Environmental Condition

SPECIFICATION
Date processing board for Neutron / Model: KD-3052
Differential Signal, Negative Unipolar Pulse, 0 ～ -1.0v*Input impedance: 100ΩTTL / Positive Logic Pulse, Input Impedance: 1kΩ
Pulse Width: > 0ns, Rise Time: > 20ns
Connector for Power Supply: DC Plug φ5.5 x φ2.1Connector for T0 signal: LEMO 00 typeConnector for Analog signal: HIF3BA-26PA-2.54DS*MIL Standard150(W) x 93(D) x38(H)*Storable to the PSD detector box of ½ inch x 600 x 8 tubes
Operating by power supply of the board: 5V / 2.2A
*AMP97 and PSD detector board are used by In-houseproduct
Another Power Supply Setting: ±12v, ±5V, 3.3V, 1.8V, 1.2V
Temperature: 0～50°C
Maximum humidity: 80% (No dew condensation)

OUTLINE[Business lineup](1) Sales of radiation measuring instrument and radiation counter.
(2) Development of radiation measuring device and radiation counter.
(3) Development of research and development device, measurement
controlling system, and inspection apparatus.

[Contact]🏠 2976-15 Mawatari, Hitachinaka-city, Ibaraki, JAPAN, 312-0012☎ +81-29-350-8011
📠 +81-29-352-9013
🌏 http://www.techno-ap.com/index_e.html
📩 order@techno-ap.com

ご訪問によるご提案やデモ等随時受け付けております。
是非お気軽にご相談ください。
皆様のご期待に沿えるよう一意専心努力してまいりますのでこれからも末永くご支援賜りますようお願い申し上げます。

何かお役に立てることがございましたらご遠慮なくご連絡ください。