GENERAL-PURPOSE SYSTEM SOLUTION / DATA PROCESSING SYSTEM FOR ADVANCED OPTICAL BEAM ANALYSISOPTICAL BEAM ANALYSIS MODULE

Synos

AP013

Image processing hardware and software package for optical beam proﬁle measurement. EF/EAF analysis function is supported as standard function.

Optical beam analysis module AP013 is the image prosessing and data analysis system
focusing on optical beam proﬁle measurement and analysis, widely applicable to optical beam
proﬁle measurement application such as NFP, FFP, collimated beam and various applicationin combination with Synosʼ optical system M-Scope series and imaging detectors ISA series.
AP013 is composed of personal computer for image processing and data analysis, optical beam
analysis software Optometrics BA Standard, calibration data set, and detector driver interface.
Additionally, Optometrics BA Standard is equipped with power distribution analysis function
such as ﬂux analysis function (EF/EAF) that is the standard analysis parameters for MMF, D86
analysis function, D4σ analysis function, etc.
OPTICAL BEAM ANALYSIS SOFTWARE

Optometrics BA Standard（Ver.3.0.0）

Optical beam analysis software Optometrics BA Standard is image processing/analysis software equipped with our unique andabundant optical beam measurement and analysis functions, including general-purpose beam proﬁle measurement, beam widthmeasurement, optical power distribution amalysis such as EF/EAF analysis, D86 analysis, D4σ analysis.
【features of optical beam analysis software Optometrics BA Standard】○Image acquisition and processing
Live image display, Image averaging and accumulation, Dark current correction, Storage and read out of image data
○Center setting function
Peak intensity position of the image, Center of gravity position of the beam, Proﬁle peak position, Center of beam proﬁle width, anyspeciﬁed position○Conversion value (unit conversion) setting function
Conversion value setting, Analysis by converted value（unit: pixel/μm/degree）○Indicator display
Peak intensity position & value (maximum intensity value in the image and its postion), Beam center of gravity position○Measurement and Analysis function●Optical beam proﬁle analysis
Realtime beam proﬁle display(vertical/horizontal, by automatic search or manual set of center position), Beam proﬁle displayof arbitrary direction, 3D display of intensity proﬁle, Peak hold display, Gaussian ﬁtting beam proﬁle●Width measurement and analysis
Beam width analysis (FWHM, 1/e2, arbitrary% down width), Unit setting & conversion●Optical power distribution analysis (*These functions are included only in “Optometrics BA standard” )EF(Encircled Flux)/EAF(Encircled Angular Flux) analysis, D86 analysis(based on EF/EAF analysis), D4σ analysis●Other analysis function
Total intensity analysis in any area, etc.
○Image display
Gray scale, binarized image, pseudo color, 3D intensity display, equal intensity line display○Image and data storage
Image storage (TIFF/BMP/CSV), intensity proﬁle data (CSV), EF/EAF data (CSV), hard copy of measurement screen (JPEG/BMP)【Analysis function of “Optometrics BA Standard (ver.3.0.0.)】○Main operation window of “Optometrics BA Standard”○EF/EAF analysis & D86 analysis○D4σ analysis【Main component】●Personal computer system for image processing and data analysis … 1set
・OS: Microsoft Windows 10 professional 64bit
・Prosessor: Intel core i5 (or corresponding speciﬁcation)
・Main memory: 16GB (or corresponding speciﬁcation)●Optical beam analysis software Optometrics BA Standard (ver. 3.0.0) … 1 license
・Software install media: 1 (including Optometrics BA Standard, Driver and interface software for detector, Calibration data),
・Software license key (USB key): 1【AP013 System selection】
MODEL
AP013
AP013BSOFTWARE
Optometrics BA Standard
Optometrics BA BasicOption

Socket communication external control software
AP013-ExC

https://www.synos.jp

OUTLINE
Supports all the above features
Among the functions of Optometrics BA Standard, it supports functions
excluding the optical power distribution analysis function
Software for external control of Optometrics BA Standard ・Optometrics BA
Basic.

27

SYNERGY OPTOSYSTEMS CO., LTD