International Nuclear
Physics Conference
2016
TechnoAP Co., Ltd.
http://www.techno-ap.com☎ 029-350-8011, 📠 029-352-9013✉ order@techno-ap.com
2976-15 Mawatari, Hitachinaka-city, Ibaraki
Japan, 312-0012

X-ray spectrum meter
APU101●Detector type: SDD, SSD●Throughput: 1Mcps and more
●Power supply: HV(+/-4000V),
PreampFront●Energy resolution[SSD]
139eV@5.9keV PT6us
250eV@5.9keV PT0.5us[SDD]
125eV@5.9keV PT2μs
145eV@5.9keV PT0.5μs●Interface: TCP/IPRear

Stand-alone model*Images is for illustration purpose.
*Please note that contents may change without prior notice.

ALL-IN-ONE
HV Power Supply
Preamplifier Power Supply
Multi Channel Analyzer

X-ray spectrum meter
APU101

Analog input
Analog gain
Sampling
ADC Gain
Digital Processing
HV power supply
Pre-amp power

Unit panel Switch Button
Connector

Interface
Dimension / Weight
Operation system
Electricity consumption*Please note that contents may change without prior notice.

1CH, ±1V range, Input-Impedance1kΩ
Coarse Gain: x2, x4, x10, x20, Fine: x0.5 to x1.5100MSPS, (Resolution: 14Bit)
8K, 4K, 2K, 1K, 512, 256ch
Trapezoidal Filter 0.1 to 16μs, Baseline Restorer,
Pileup Rejecter, Coarse Gain, Fine Gain
0V to ±4000V (Max: 1.0mA), Ripple: 20mVp-p(typ.)*Customizable up to ±5000V (Max: 0.67mA)
Bias shut down input terminal equipped±12V, ±24V (NIM-Standard)【FRONT】
HV power supply monitor LED, Dead-time monitor LED
Emergency Stop button, LAN Connector, POWER button【BACK】
SHV connector for HV power supply
D-sub9 pin-connector for pre-amp power
BNC input connector for pre-amp output signal
LEMO connector for output DAC MONITOR etc.
Ethernet TCP/IP
210mm(W) x 45mm(H) x 275mm (D) (Without connector), Approx.. 1,800g
Windows 7 (32/64bit) or more, Display WXGA or more+12V(Approx. 1.0A)

X-ray spectrum meter
APN101X●Detector type
SDD, SSD●Throughput
over 1Mcps●Power supply
HV(+/-4000V), Preamp●Energy resolution[SSD]
139eV@5.9keV PT6us
250eV@5.9keV PT0.5us[SDD]
125eV@5.9keV PT2μs
145eV@5.9keV PT0.5μs

NIM model*Images is for illustration purpose.
*Please note that contents may change without prior notice.

●InterfaceTCP/IPX-ray spectrum meter
APN101X

NIM model● Detector
Semiconductor detector (Ge, CdTe, Si etc.)
Scintillator (LaBr3(Ce), NaI (Tl), CsI (Tl) etc.)● Resolution
1.7keV@1.33MeV (Ge Semiconductor detector)
2.8 to 3.5%@662keV (LaBr3(Ce) scintillator)● Measurement Mode
Histogram● Multi-function
Spectroscopy amp, Filter waveform output DAC

ALL-IN-ONE
HV Power Supply
Measurement Mode: Histogram

Preamplifier Power Supply
Multi Channel Analyzer

X-ray spectrum meter
XS100[Energy range]
2keV ～ 20keV[Detector cooling]
Cooling by Peltier device[Energy resolution]
125eV@5.9keV 2 us peaking time
150eV@5.9keV 0.15 us peaking time

For X-ray fluorescence(XRF)
The XS100 is X-ray spectrometer with SDD detector (30 mm2),
which is a high-resolution. XS100 was downsizing by integrated
combination of SDD, DSP, high-voltage power supply and Peltierdevice. The pre-amplifier signal from SDD detector was carriedout DSP processing by ADC (100 MHz, 14 bit) and FPGA. That
data is transferred to a PC by USB connection. It was carried outtransistor reset processing in an appropriate way. Therefore, it
has substantially-improve the throughput.
*Images is for illustration purpose.
*Please note that contents may change without prior notice.

FWHM=125eV@5.9keV

Application (spectrum, Mn-Kα)

X-ray spectrum meter
XS100

DAC output of trapezoidal filter shaping(Digital Signal Processing)*Images is for illustration purpose.
*Please note that contents may change without prior notice.

Application (spectrum, Mn-Kα)

X-ray spectrum meter
XS100

Peaking 0.15us
Peaking 0.3us
Peaking 0.5us
Peaking 1us
Peaking 2us

Peaking 0.05us
Peaking 0.3us
Peaking 0.5us
Peaking 1us
Peaking 2us

Input vs Resolution*Images is for illustration purpose.
*Please note that contents may change without prior notice.

Input vs Throughput

Soft X-ray spectrum meter
XS100V●DetectorSDD (Windows less)●Energy range
C(277eV) ～For Soft X-ray measurement●Detector cooling
Peltier device

C (277eV)
O軟X線測定(525eV)(放射光施設にて)
CUSTOM MADE
for your needs!
eV*Images is for illustration purpose.
*Please note that contents may change without prior notice.

Measurement of Soft x-ray(C & O)

Soft X-ray spectrum meterXS100VDetector
Detector cooling

Cooling by Peltier device

ADC sampling

100MHz, 14bit

ADC gain
Energy resolution

InhibitTransistor Reset Processing

125eV@5.9keV MnKα 10kcps ： 2us Peaking time※1
150eV@5.9keV MnKα 300kcps ： 0.05us Peaking time※1
Max. 150kcps ： 2us Peaking timeMax. 1250kcps ： 0.05us Peaking time

Measurable
Element

C (Carbon) ～10-5 PaInterfaceUSB2.0 or Ethernet

Back panel

Monitor output terminal, TTL（SCA） output terminal,VETO input terminal, DC power socket, miniUSB connector,
Power LED monitor

Externaldimensions

80(W) × 400(D) × 40(H) *Unit: mm

WeightAccessory*Images is for illustration purpose.
*Please note that contents may change without prior notice.

4096, 2048, 1024, 512, 256ch

Throughput

Vacuum Degree

10us

SDD 65mm2 collimated to 50mm2, Windows-lessAbout 1100g
AC adapter, USB cable, Application

Digital Signal Processor for X-ray Spectroscopy
APN504 GbE
Analog input

4CH, LEMO 00 series connector, Input impedance 1kΩ

Coarse Gainx 2, x 4, x 10, x 20

Fine Gain
ADC:ADC Gain
Trapezoidal Filter
Digital Signal
ProcessingInput signal ±1V, Sampling 100MSPS, Resolution 14bit
4096, 2048, 1024, 512, 256ch
0.05～12 μs
Baseline Restorer, Pileup Rejecter, CFD*All parameters setting by PC.

Quick scan mode

Minimum time distance 10ms
Data Size: 32768byte (= 2byte × 4CH × 4096ch)

External terminal

Filter waveform output, Clock input, GATE (Trigger)
input, VETO input, Clear inputInterface
External Dimensions(Unit: mm)
Weight*Images is for illustration purpose.
*Please note that contents may change without prior notice.

X 0.5 ～ x 1.5

Gigabit Ethernet (TCP/IP)
NIM1U 34(W) x 221(H) x 249(T) (Without connector)
About 900g

Digital Signal Processor for X-ray Spectroscopy
APN504 GbE●Channel: 4CH simultaneous sampling

Quick XAFS

●Throughput: 1Mcps and more
●Measurement Mode: Histogram, List
●External: NIM1U●Resolution[In the case of 19 elements SSD]
@5.9keV 139 eV 6 μs Peaking time
@250eV 0.5μs Peaking time*Comparable Analog 0.25μs

Target: Ｍn, ICR 1.1Mcps, OCR 1.0Mcps
FWHM 194eV@5.9keV MnKα 0.15μsPT[In the case of SDD] *High-resolution type125eV2μs Peaking time
145eV 0.5μs Peaking time*Comparable Analog 0.25μs●Multi function
Spectroscopy amp, Timing filter amp, CFD,Input and filter waveform output DAC

Emitted light exposure examination*Images is for illustration purpose.
*Please note that contents may change without prior notice.

●Interface: TCP/IP, Gigabit Ethernet
Data transfer 20MByte/sec and more
*List mode

Digital Signal Processor for X-ray Spectroscopy
APN504 GbE
Preamp out signals
4
GATE

NIM Power Supply
APN504 #1
IP : 192.168.1.11

4
APN504 #2
IP : 192.168.1.12
MultiElements
Detector

4

Quick scan data example
Giga bit Ether

Event#1(10ms)

CH1 4096ch CH2 4096ch CH3 4096ch CH4 4096ch

Event#2(20ms)

CH1 4096ch CH2 4096ch CH3 4096ch CH4 4096ch

Event#N(10*Nms) CH1 4096ch CH2 4096ch CH3 4096ch CH4 4096ch
APN504 #3
IP : 192.168.1.13●Quick scan

4APN504 #4
IP : 192.168.1.14
3APN504 #5
IP : 192.168.1.15
TimingSupply*Images is for illustration purpose.
*Please note that contents may change without prior notice.

●Interface

10ms(min.) interval
4096ch *16bit *4CH / Event
Giga bit Ether TCP/IP and UDP

Digital Signal Processing for X-ray
APV8004X●Energy resolution[ SSD with 19 elements]
139eV@5.9keV: 6μs Peaking Time
144eV@5.9keV: 4μs Peaking Time
250eV@5.9keV: 0.25μs Peaking Time

[ SDD ]
125eV@5.9keV: 2μs Peaking Time●Throughput: 1Mcps and more
●Measurement Mode: Histogram, List
●Option: TTL output of ROI-SCA*Images is for illustration purpose.
*Please note that contents may change without prior notice.

For SSD with 19 elements

Digital Signal Processing for X-ray
APV8004X

Application*Images is for illustration purpose.
*Please note that contents may change without prior notice.

Monitor of Trapezoidal filter

7 element SDD system for fluorescence X-ray
XSDD50-07● Element area
455mm² (65ｍｍ² ｘ 7 elements)● Effective area
350mm² (65mm² Collimated to 50mm² x 7 elements)● Function
Histogram, List, Waveform, ROI-SCA● ADC
4CH 100Msps 14bit● Energy Resolution
244eV@5.9keV MnKαPeaking time 0.25μs, 1000kOCR● Power Supply for SDD-200V, ±5V, +3.3V

CUSTOM MADE
for your needs!
*Images is for illustration purpose.
*Please note that contents may change without prior notice.

● Interface
Ethernet (TCP/IP)

DSP & Power supply

7 element SDD system for fluorescence X-ray
XSDD50-07

400k ICR 185eV@5.9keV 0.25μs Peaking Time
*Images is for illustration purpose.
*Please note that contents may change without prior notice.

7 element SDD system for fluorescence X-ray
XSDD50-07

ICR vs Energy Resolution*Images is for illustration purpose.
*Please note that contents may change without prior notice.

ICR vs OCR

4 element SDD for X-ray
XSDD50-04

Selectable
Active Area: 30mm2 / 50mm2
Window: Be / AP3.3 / Window-less

CUSTOM MADE
for your needs!

10-5 Pa
Vacuum Degree

Customizable*Images is for illustration purpose.
*Please note that contents may change without prior notice.

Flange type / Tube size

4 element SDD for X-ray
Detector

XSDD50-04

SDD 50mm2, Window-less / AP3.3 / BeElement area

260mm2 (65mm2 × 4 element）

Active area

200mm2 (65mm2 collimated to 50mm2 × 4 element）

Measurement mode
ADC sampling

Histogram / List / Waveform / ROI-SCA
4CH 100Msps 14bit

Energy resolution(typ.)

244eV@5.9keV MnKα*Peaking time: 0.25μs, 1000kOCR

SDD power supply-200Ｖ, ±5V, ＋3.3V

Throughput

Max. 150kcps ： 2us
Max. 1000kcps ： 0.15us

Interface

Ethernet (TCP/IP)

OptionZ-axis movement mechanism, UHV valve

Vacuum capable<10⁻5 Pa

Flange type

ICF114（Standard）

Accessory

Software, Instruction Manual*Please note that contents may change without prior notice.

4 element SDD for X-ray
10-5 PaXSDD50-04

Vacuum Degree

Transport with bellows*Images is for illustration purpose.
*Please note that contents may change without prior notice.

CUSTOM MADE
for your needs!

1 element SDD for X-rayXSDD50-01

Selectable
Active Area: 30mm2 / 50mm2
Window: Be / AP3.3 / Window-less10-5 Pa
Vacuum Degree*Images is for illustration purpose.
*Please note that contents may change without prior notice.

Customizable
Flange type / Tube size

1 element SDD for X-ray
XSDD50-01
Detector

SDD 50mm2, Window-less / AP3.3 / BeElement area

65mm2

Active area

65mm2 collimated to 50mm2

Measurement mode

ADC sampling

Histogram / List / Waveform / ROI-SCA

100Msps 14bit

Energy resolution(typ.)125eV@5.9keV 2us
150eV@5.9keV 0.15us

SDD power supply-200Ｖ, ±5V, ＋3.3V

Throughput

Max. 150kcps: 2us
Max. 1000kcps: 0.15us

Interface

Ethernet (TCP/IP)

OptionZ-axis movement mechanism, UHV valve

Vacuum capable
Flange type
*Images is for illustration purpose.
*Please note that contents may change without prior notice.

<10⁻5 Pa
ICF70（Standard）

1 element SDD for X-ray
XSDD50-01
Gate valve and Angle valve transportwith bellows*Images is for illustration purpose.
*Please note that contents may change without prior notice.

CUSTOM MADE
for your needs!