Keysight 16195B
Calibration Kit

Operation Note1

Notices
The information contained in this document is subject to change without notice.
This document contains proprietary information that is protected by copyright. All rights
are reserved. No part of this document may be photocopied, reproduced, or translated toanother language without the prior written consent of Keysight Technologies.
Microsoft®,MS-DOS®,Windows®,Visual C++®,Visual Basic®,VBA® and Excel® areregistered
UNIX is a registered trademark in U.S. and other countries, licensed
exclusively through X/Open Company Limited.
Portions ©Copyright 1996, Microsoft Corporation. All rights reserved.
© Copyright 1999, 2001, 2006, 2015 Keysight Technologies

Manual Printing History
The manual’s printing date and part number indicate its current edition. The printing datechanges when a new edition is printed. (Minor corrections and updates that areincorporated at reprint do not cause the date to change.) The manual part number changes
when extensive technical changes are incorporated.
December 1999

Edition 1

March 2001

Edition 2

October 2006

Edition 3

February 2015

Edition 42

Safety Summary
The following general safety precautions must be observed during all phases of operation,service, and repair of this instrument. Failure to comply with these precautions or withspecific WARNINGS elsewhere in this manual may impair the protection provided by theequipment. In addition it violates safety standards of design, manufacture, and intended useof the instrument.
Keysight Technologies Company assumes no liability for the customer’s failure to complywith these requirements.
NOTE

16195B comply with INSTALLATION CATEGORY II and POLLUTION DEGREE 2 in
IEC61010-1. 16195B are INDOOR USE product.
•DO NOT Operate In An Explosive Atmosphere
Do not operate the instrument in the presence of flammable gasses or fumes. Operationof any electrical instrument in such an environment constitutes a definite safety hazard.

•DO NOT Substitute Parts Or Modify Instrument
Because of the danger of introducing additional hazards, do not install substitute partsor perform unauthorized modifications to the instrument. Return the instrument to a
Hewlett-Packard Sales and Service Office for service and repair to ensure that safety
features are maintained.

•Dangerous Procedure Warnings
Warnings, such as the example below, precede potentially dangerous procedures
throughout this manual. Instructions contained in the warnings must be followed.

Safety Symbol
General definitions of safety symbols used on the instrument or in manuals are listed
below.
WARNING

This warning sign denotes a hazard. It calls attention to a procedure, practice,condition or the like, which, if not correctly performed or adhered to, could result ininjury or death to personnel.

CAUTION

This Caution sign denotes a hazard. It calls attention to a procedure, practice, condition or
the like, which, if not correctly performed or adhered to, could result in damage to or
destruction of part or all of the product.

NOTE

Note denotes important information. It calls attention to a procedure, practice, condition orthe like, which is essential to highlight.

3Certification
Keysight Technologies certifies that this product met its published specifications at thetime of shipment from the factory. Keysight Technologies further certifies that its
calibration measurements are traceable to the United States National Institute of Standards
and Technology, to the extent allowed by the Institution’s calibration facility, or to thecalibration facilities of other International Standards Organization members.

Warranty
This Keysight Technologies instrument product is warranted against defects in material
and workmanship for a period corresponding to the individual warranty periods of its
component products. Instruments are warranted for a period of one year. Fixtures andadapters are warranted for a period of 90 days. During the warranty period, KeysightTechnologies will, at its option, either repair or replace products that prove to be defective.
For warranty service or repair, this product must be returned to a service facility designatedby Keysight Technologies. Buyer shall prepay shipping charges to Keysight Technologies
and Keysight Technologies shall pay shipping charges to return the product to Buyer.
However, Buyer shall pay all shipping charges, duties, and taxes for products returned toKeysight Technologies from another country.
Keysight Technologies warrants that its software and firmware designated by KeysightTechnologies for use with an instrument will execute its programming instruction whenproperty installed on that instrument. Keysight Technologies does not warrant that theoperation of the instrument, or software, or firmware will be uninterrupted or error free.

Limitation of Warranty
The foregoing warranty shall not apply to defects resulting from improper or inadequate
maintenance by Buyer, Buyer-supplied software or interfacing, unauthorized modification
or misuse, operation outside the environmental specifications for the product, or impropersite preparation or maintenance.
IMPORTANTNo other warranty is expressed or implied. Keysight Technologies specifically disclaims
the implied warranties of merchantability and fitness for a particular purpose.

4Exclusive Remedies
The remedies provided herein are buyer’s sole and exclusive remedies. KeysightTechnologies shall not be liable for any direct, indirect, special, incidental, orconsequential damages, whether based on contract, tort, or any other legal theory.

Assistance
Product maintenance agreements and other customer assistance agreements are availablefor Keysight Technologies products.
For any assistance, contact your nearest Keysight Technologies Sales and Service Office.
Addresses are provided at the back of this manual.

5

6

1

General Information
This chapter provides Keysight Technologies 16195B description, characteristics andrelated general information.

General Information
Description

Description
Keysight 16195B is a calibration kit for Keysight 4287A RF LCR Meter or E4991A
impedance / material analyzer. The calibration using the Keysight 16195B is performed on
the 7 mm connector plane.

Kit Contents
The calibration kit contents are listed in Table 1-1.

Table 1-1

Contents of the 16195B
Description

Keysight Part No.

Qty.

50 Ω Termination

04287-60021

1

Open Termination

E4991-60022

1

Short Termination

E4991-60021

1

Low-Loss Capacitor

04287-60022

1

Carrying Case

16195-60002

1

Operation Note (This manual)16195-90020

1

Initial Inspection
Inspect the shipping container for damage. If the shipping container or cushioning material
is damaged, it should be kept until the shipping contents have been checked forcompleteness and the performance test kit has been checked mechanically and electrically.
The shipping contents should consist of the items shown in Table 1-1. If the shipment isincomplete, or if there is mechanical damage or other defects, notify your nearest KeysightTechnologies office. If the shipping container is damaged, or the cushioning materialshows signs of stress, notify the carrier as well as the Keysight Technologies office. Keepthe shipping materials for the carrier’s inspection.

8Chapter 1

General Information
Characteristics

Characteristics
This section provides the characteristics of the Keysight 16195B calibration kit.

Electrical Characteristic
Nominal Values of the kit are in Table 1-2.

Table 1-2

NOTE

Nominal Values
OPEN Conductance

0S

OPEN Capacitance

82 fF

SHORT Resistance0Ω

SHORT inductance0H

50 Ω resistance50 Ω50 Ω inductance0H

The low loss capacitor does not have a defined value.

Operating Conditions
The calibration kit must be operated under the ambient environmental conditions in Table
1-3.

Table 1-3

Operating Conditions
Temperature

23 ºC ± 5 ºC

Relative Humidity≤ 70 % at 28 ºCAltitude0 m to 4500 m

Storage Conditions
The calibration kit must be stored under the ambient environmental conditions in Table
1-4.

Table 1-4

Chapter 1

Storage Conditions
Temperature- 40 ºC to + 75 ºC

Relative Humidity≤ 95 % at 40 ºC

Altitude

0 m to 15000 m

9General Information
Characteristics

Carrying Case Dimensions
The carrying case dimensions are;350 mm (W) × 100 mm (H) × 270 mm (D)10Chapter 1

2Service
This chapter provides information of 16195B service.

ServiceServiceService
If your calibration kit requires serivce, contact the Keysight Technologies office nearestyou. When you send the kit to Keysight Technologies, please provide followitng
information.
•Your company name and address•

A technical contact person within your company, and the person’s phone and fax
number•

The model number and serial number of the kit•

The type of the service required•

A detailed description of the problem

12Chapter 2

REGIONAL SALES AND SUPPORT OFFICES
For more information about Keysight Technologies test and measurement products, applications, services, andfor a current sales office listing, visit our web site: http://www.keysight.com/find/tmdir. You can also contact one21/01/2004of the following centers and ask for a test and measurement sales representative.

United States:
Test and Measurement Call Center(tel) 1 800 452-4844(fax) 1 888 900-8921
Canada:Test and Measurement Call Center(tel) 1 877 894-4414(fax) 1 888 900-8921
China:(tel) 800 810-0189(fax) 800 820-2816
Europe:(tel) (31 20) 547-2323(fax) (31 20) 547-2390
Japan:
Call Center(tel) 0120 421-345(tel) (81) 426 56-7832(fax) (81) 426 56-7840
Korea:(tel) (82 2) 2004-5004(fax) (82 2) 2004-5115Latin America:(tel) (305) 269-7500(fax) (305) 269-7599
Taiwan:(tel) 0800 047 866(fax) 0800 286 331

Australia/New Zealand:(tel) (61 3) 9210-5555 (Australia)(fax) (61 3) 9210-5899(tel) (64 4) 939-0636 (New Zealand)(fax) (64 4) 972-5364
Asia Pacific:(tel) (65) 6375-8100(fax) (65) 6836-0252
Email: tm_asia@keysight.com

This information is subject to change without notice.
© Keysight Technologies 1999, 2001, 2006, 2015
Edition 4, February 2015

*16195-90040*16195-90040
www.keysight.com