i-FlowLab 6XX1

Scalable, high-flow, nitrogen generation solution for laboratories
Part Number : See back page

Service Kit : Contact us for details

Description
i-FlowLab nitrogen generators have been engineered by Peak
Scientific gas generation specialists to meet the supply needs oflaboratories which require a large capacity, on-site nitrogen supply.
Harnessing the latest gas purification technologies i-FlowLab 6XX1
removes oxygen from air, delivering nitrogen at flow rates from 155 l/min
to 3402 l/min. Available in over 100 pre-configured flow and purity
specifications, i-FlowLab is modular with the option to add columns
to increase the flow rate as your nitrogen requirements grow.
The 6XX1 range measures generator purity output as oxygen
content in percentage, as opposed to its 6XX2 counterpart whichmeasures oxygen content in ppm. Integrated with the compressed
air pre-filtration system, PureAir, ensures the highest quality nitrogen
output by cleaning air before it enters the i-FlowLab generator.

Applications)

Multiple LC-MS
instruments)

Sample Concentrators)

Whole lab solution

500.00 mm (19.68 inch))

FT-IR/Glove Boxes/TOC)

Larger scale laboratories

760 mm (29.92 inch)

160 mm(6.29 inch)

1738 mm (68.42 Inch)i-FlowLab 6XX1 Dimensions

Copyright 2020 Peak Scientific - Rev - i-FlowLab 6XX1 datasheet Rev 4 RSID 1268 EN - 07/10/20

Key Features)

Oxygen Anaylzer provides continuous real-timemonitoring of purity produced in %)Modular design allowing for columns to be added as
your requirements grow)Sustainable on-site gas generation, no need forstorage of high volumes of bulk gas supply)

Consistent gas purity, so no fluctuations as can beexperienced with bulk gas storage)

Minimal maintenance, only scheduled preventative
maintenance required to ensure long life andperformance of the system)

A safer solution with no storing of highly pressurizedgas on-site)

Remote monitoring, Live product status information
at your fingertips, wherever you are)

Reduce carbon footprint by removing the requirement
for bulk gas supply delivery)

Comprehensive installation including compressed
air supply, pipework, pre-filtration system, nitrogengenerator and storage

Technical Specifications

LPM *Product Name

6011

6021

6031

6041

6051

6061

6071

6081

6091

6101

Oxygen Content - 0.50%

155

305

450

585

730

874

1015

1160

1300

1450

Oxygen Content - 1%

190

380

530

725

885

1062

1238

1415

1574

1748

Oxygen Content - 2%

245

490

665

845

1065

1280

1480

1685

1885

2090

Oxygen Content - 3%

305

578

810

1025

1278

1574

1815

2065

2330

2580

Oxygen Content - 4%

362

651

915

1115

1420

1704

1988

2272

2556

2840

Oxygen Content - 5%

402

750

1100

1385

1649

2045

2478

2778

3045

3402

Voltage100-240 VAC +/- 10%

Frequency

50/60 Hz

Current

2.0 - 1.0 Amp

Input Connection

C20 Plug

Power Cord (Supplied)

C19 Socket to local connection

Power Consumption

250 Watts

Operating Temperature

5°C - 50°C / 41°F - 122°F

Pollution Degree2Installation Category

II

Width mm (inch)

500 (19.68)

Height mm (inch)

1738 (68.42)

Depth mm (inch)
Weight kg (lbs)
Shipping weight kg (lbs)

760

920

1080

1240

1400

1560

1720

1880

2040

2200

(29.92)(36.22)(42.52)(42.52)(55.12)(61.42)(67.72)(74.02)(80.31)(86.61)

197

282

367

452

537

622

707

792

877

962

(433)

(620)

(807)

(994)

(1181)(1368)(1555)(1742)(1929)(2116)

277

364

452

538

625

712

799

886

973

1060

(609)

(801)

(992)

(1184)(1375)(1566)(1758)(1949)(2141)(2333)

20-6071

20-6081

20-6091

20-6101

Noise Level

59dBa @ 1m

*Performance data is based on 7 bar (G) inlet air pressure and 20 - 25 deg C ambient temperature.
(Flow reference conditions, 20 deg C, 1013 millibar (a), 0% Relative Humidity)

Ordering Information
Part Number

20-601120-602120-603120-6041

20-6051

20-6061

Service Kit

08-0318

08-0319

Complete Plan

09-3705

09-3706

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that you’re investment continues
to be [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you thereassurance of instant access to worldwide technical support and priority on-site response in the untimely event of a breakdown.

Peak HQ
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management System
conforms to: ISO:9001:2008

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com

Copyright 2020 Peak Scientific - Rev - i-FlowLab 6XX1 datasheet Rev 4 RSID 1268 EN - 07/10/20