Precision Nitrogen Headspace
Nitrogen Generator
Part Number : 65-3000

Service Kit : See reverse

Your local gas generation partner

Description
The Precision Nitrogen Standard Headspace 250 further
compliments our Precision range of products for GC. Speciallydesigned based on pressure swing technology, the Precision
Nitrogen Headspace offers 250cc/min of UHP Nitrogen ata pressure of up to 100psi to supply the purest gas to yourHeadspace Autosampler.
The ground- breaking stackable design as part of the Precision
Series also means that this generator is a true space saver.
As with all Precision Series generators, Nitrogen models benefit
from a compact and modular, stackable design, minimizing the total
footprint required for GC gas supply, and providing flexibility to add
or remove modules as your laboratory requirements evolve over time.

Applications) Headspace Autosamplers) Ultra High Purity nitrogenmake up gasKey Features) Suitable as make up gas for any GC as well as
Headspace Autosamplers
540 mm) Nitrogen generated on demand380 mm) Constant and consistent supply, avoid running out ofgas during analysis256 mm) Increased output pressure to meet the requirementsof Headspace Autosamplers) Avoid contaminants entering system as result ofchanging over empty cylinders) Dedicated Precision compressor module optional
where house supply not suitable or not available) Integrates seamlessly with other Precision units) Minimum maintenance with an annual filter change) Compact, space-saving modular design

Precision Nitrogen Headspacein a typical Precision stack.

Copyright © Peak Scientific Instruments Ltd 2024 — Precision Headspace Rev 3 RSID 414 EN —10/09/2024

Technical Specifications

Precision Headspace

Max Flow Rate

250cc/min

Max Pressure

100 psi / 6.9 bar

Purity> 99.9995%

Gas Outlets

1 x 1/4 " BSPP

Min/Max Air Inlet Pressure

120 psi / 8.3 bar

Min Air Inlet Flow25 lpm

Min Inlet Air Quality

ISO8573 - 1:2010 Class 1.4.1

Phthalates

None

Suspended Liquids

None

Operating Temperature

5°C - 30°C / 41°F - 86 °F

Electrical Requirements

110-230 VAC / 50/60 Hz / 0.03A

Power Consumption

6.9 Watts

Heat Output

25 BTU/Hr

Generator Dimensions (WxDxH)

38.0 x 54 x 25.6 cm / 14.9 x 21.2 x 10”

Generator Weight

22.5 kg / 49.6 lbs

Ordering Information
Part Number

65-3000Annual Service
Fixed Price Preventative Maintenance

Visit: www.peakscientific.com/ordering

Complete/Premium Maintenance PlanPrecision Air Compressor

110v

230v

Part Number

65-1555

65-0555

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that you’re investment continues
to be [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you thereassurance of instant access to worldwide technical support and priority on-site response in the untimely event of a breakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management System
conforms to: ISO:9001:2008

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com

Copyright © Peak Scientific Instruments Ltd 2024 — Precision Headspace Rev 3 RSID 414 EN —10/09/2024