Synos

GENERAL-PURPOSE SYSTEM SOLUTION / NFP/FFP SIMULTANEOUS MEASUREMENT SYSTEM
NFP/FFP SIMULTANEOUS MEASUREMENT SYSTEM
Simultaneous measurement of NFP & FFP by a single optical unit.

NFP/FFP Simultaneous measurement system realizes simultaneous observation and analysis
of NFP and FFP by a single optical unit. Previously, for measurement and analysis of NFP andFFP by optical method, dedicated two kinds of optics, NFP and FFP measurement optics areneeded each. NFP&FFP simultaneous measurement system with M-Scope type D realizedsimultaneous observation and analysis of NFP and FFP without changing optics.
【Features】○M-Scope type D, NFP/FFP simultaneous measurement optics● Simultaneous observation & analysis of NFP and FFP by a single optical unit.
○Working distance during FFP measurement is approx. 17 mm.
○Possible to measure in 400nm to 1700nm wavelength range by selecting detector.
○Optical beam analysis module AP013, specially designed high-functional image processing
software for optical beam proﬁle analysis● All-in-one package of PC, optical beam analysis software, detector driver, correction data.
● High-performance image processing software for optical beam proﬁle measurement Optimetrics BA Standard is pre-installed.
【Standard component】○Optics● NFP/FFP simultaneous measurement optics M-Scope type D○Measurement wavelength●Please specify the measurement wavelength because AR coating to some optical parts is required.
○Available detectors selection●400~1100nm︓Hi-resolution CMOS detector ISA071/ISA071GL●950~1700nm︓InGaAs high sensitivity NIR detector ISA041H2●400~1700nm : InGaAs high resolution NIR detector ISA041HRA☞Regarding the ﬁeld of view and pixel resolution during NFP measurement and the measurement angle coverage and pixel resolution
during FFP measurement by the detector used, please refer to P50【Detector selection and NFP/FFP simultaneous measurement
speciﬁcations】○Optical beam analysis module AP013● PC for image processing, optical beam analysis software Optometrics BA Standard, detector driver, calibration data, USB key○Accessories● Cables, instruction manuals, etc.
【Option】○Objective lens selection● Objective lens for FFP measurement: 50× (ﬁxed)●Objective lens for NFP measurement: Selectable○Option for optics (for M-Scope type D)● 2× intermediate lens port MS-OP011-RL2 : Intermediate lens unit that doubles the overall magniﬁcation● 1/2× intermediate lens port MS-OP011-RLH : Intermediate lens unit that halves the overall magniﬁcation● Coaxial epi-illumination port MS-OP011-CEP : Coaxial epi-illumination port with removable half mirror○ND ﬁlter● Visible (400~700nm): NDF-5 (5 types per set)
● NIR (700~1100nm): NDF NIR-5 (5 types per set)
● IR (1310~1550nm): NDF IR-5（5 types per set）
○Coaxial epi-illumination light source● Visible~NIR : LED epi-illumination system● IR : 950nm IR-LED epi-illumination system○Optics bench● Optics bench for ﬁber measurement with manual stages● Vertical setting optics bench【Component selection of NFP/FFP simultaneous measurement system】○Stages・optics bench○Optics selection○Optical beam analysis module AP013○Detector selection●for visible~1100nm●Personal computer
・Main unit
・Accessories

Sample stages
Optics stages
Optics bench for ﬁbermeasurement

Vertical settingoptics bench＊Can be combined with various
motorized/manual stages

https://www.synos.jp

NFP/FFP simultaneous
measurement optics
M-Scope type D●Option・2× intermediate lens port
MS-OP011-RL2
・1/2× intermediate lens port
MS-OP011-RLH
・Coaxial epi-illumination port
MS-OP011-CEP

High resolution CMOS
detector ISA071●Optical beam analysis software
Optometrics BA Standard●Detector driver●Calibration data●USB licence key●for 950~1700nm

InGaAs high sensitivity NIR
detector ISA041H2○Accessories●ND ﬁlter●for 400~1700nm●Objective lens
InGaAs high resolution NIRdetector ISA041HRA/HRVA

22●Coaxial epi-illuminationsystem

SYNERGY OPTOSYSTEMS CO., LTD