Synos

OPTICAL LIGHT SOURCE / LD LIGHT SOURCEHIGH STABILITY FC CONNECTOR OUTPUT LD LIGHT SOURCEHigh stability DC-LD light source unit with FC connector output type.

LSL002 SERIESLSL002 series is FC connector output type, high stability DC-LD light source unit. As is FC
connector output type lightsource, various optical ﬁbers on the market are available.
Furthermore, as optional version, it is available to prepare polarization mainraining optical ﬁberoutput. As various types of laser diodes on the market are set in, it is possible to choose fromvarious types such as wavelength, output power, etc.
【Features】○Easy to use because of all-in-one unit including light source, driver circuit, power supply, etc.
○Because of using laser diode on the market, it is possible to choose various type of lightsource in power, wavelength, etc.
○High stability in output power and wavelength with APC and temperature control.
○As standard version, it is singlemode ﬁber and FC connector output. As optional version, it isavailable to prepare polarization mainraining optical ﬁber output.
【Summary of speciﬁcations】○Optical output:approx. 1mW (standard)【Standard component】○Main unit: 1○Power supply cable: 1○Key for key switch: 1*Special speciﬁcations such as high output are available as an option.

○Optical output connector: SMF+FC connector (standard)
PMF+FC connector (option)○LD driver:
APC+temperature control bypeltier device○Output stability:
typ.±1%○Power supply:
AC100V±10％、50/60Hz【LD selection】
Peak wavelength (nm)Wavelength tolerance (nm)Max. output power (mW)Power stability (％)
Mode ﬁeld diameter (μm)
N.A.

375±51±12.20.13405±51±130.13488±51±13.50.12

520±101±13.50.12635±101±14.50.12658±101±14.50.12785±101±15.00.13

850±151±15.00.13980±101±15.90.141064±101±15.90.141310±201±19.20.141550±301±110.40.14

COMPACT TYPE SMF OUTPUT LD LIGHT SOURCE LSL011 SERIES
Compact and SMF output type LD light source unit with SM ﬁber output type. Low cost and easy to use.
LSL011 series is compact and SMF output type LD light source unit. As is SM ﬁber output typelightsource with LD, driver circuit, TEC, power supply, it is very easy to use. As various types oflaser diodes on the market are set in, it is possible to choose from various types such as
wavelength, output power, etc. Pulse emmision is also available with external TTL trigger input.
【Features】○Easy to use as is all-in-one unit with SM ﬁber, driver circuit, TEC, power supply.
○Because of using laser diode on the market, it is possible to choose various type of lightsource in power, wavelength, etc.
○High stability in output power and wavelength with ACC and temperature control.
○Pulse emmision with external TTL trigger input.
【Standard component】○Main unit: 1【Summary of speciﬁcation】○Light source module: SMF coupling FP-LD○Fiber length:
50cm or more
○Output connector:
FC/PC○Drive control:
ACC and temperature control○Pulse emission:
TTL input (high : on, low : oﬀ)(except 532nm)
maximum repetition ＞1kHz○Power supply:
AC100-240V 50/60Hz 0.3A
○dimension, weight:60(W) x 50(H) x 120(D)mm/400g(except prodection component)
AC100V±10％、50/60Hz【LD selection】
Peak wavelength (nm) 375
Wavelength tolerance (nm) ±5
Max. output power (mW) 1
Power stability (％)±1Mode ﬁeld diameter (μm) 2.9
N.A.
0.12

https://www.synos.jp

405±51±12.90.12445±510±12.90.12473±510±13.50.13488±510±13.50.13520±1010±13.50.13635±105±14.00.13

41660±1010±14.00.13785±1010±15.00.13

850±1010±15.00.13980±1010±14.20.141064±1010±14.20.141310±1010±19.00.13

1550±1010±19.00.131620±154±19.00.13

SYNERGY OPTOSYSTEMS CO., LTD