Synos

ACCESSORIES, PERIPHERALS / OPTICAL WORK BENCH, COAXIAL EPI-ILLUMINATION SYSTEMOPTICS BENCH FOR SAMPLE MEASUREMENTVERTICAL TYPE OPTICS BENCHThis is vertical setting optics bench for optical measurement optics M-Scope series. Rough position
adjustment function in Z axis is equipped so that it is easy to adjust focusing for various unevensample. As sample holders and stage system can be designed and selected based on device shape,
the measurement system is easily conﬁgured.
【Summary of speciﬁcation】○Loading capacity of optics installation part: approx. ５kg○Motion range of Z axis rough position adjustment: approx. 80mm
OPTICS BENCH FOR FIBER MEASUREMENT OP002-F3/OP002-F5
This is optics bench for optical measurement optics M-Scope series, especially for optical ﬁber sample
measurement. As manual 3-axis (or 5-axis) stage for ﬁber position adjustmet is equipped, it is easy toadjust mesurement position of ﬁber sample in NFP/FFP and various beam proﬁle measurement.
【Optics bench for optical ﬁber measurement with manual 3axis stage OP002-F3】○Stage system for ﬁber position adjustment: Manual 3-axis (XYZ axis)○Motion range of XYZ-axis stage: ±6.5mm○Minimum readout: 10μｍ/scale○Fiber adaptor: FC connector(standard), SC connector and other type ﬁber connector (option)【Optics bench for optical ﬁber measurement with manual 5axis stage OP002-F5】○Stage system for ﬁber position adjustment: Manual 5-axis (XYZ+θxθy axis)○Motion range of XY-axis stage: ±6.5mm○Motion range of Z-axis stage: ±3mm
○Minimum readout of XYZ-axis: 10μｍ / scale○Motion range of θxθy axis stage: ±15°
○Fiber adaptor: FC connector (standard), SC connector and other type ﬁber connector (option)【Compatible optics for OP002・OP002-F3・OP002-F5 (common)】○Synosʼ optical measurement optics M-Scope series●Sophisticated optical measurement optics M-Scope type I seires●Simpliﬁed optical measurement optics M-Scope type J series●Optical beam NFP measurement optics M-Scope type S, M-Scope type L●FFP measurement optics M-Scope type F, M-Scope type FW●Collimated beam measurement optics M-Scope type C●NFP/FFP simultaneous measurement optics M-Scope type D●Underﬁlled launch optical system M-Scope type G●Mode selective launch optical system M-Scope type ML●NFP measurement optics for high power laser M-Scope type HS, M-Scope type HL●FFP measurement optics for high power laser M-Scope type HF●Polarization measurement module PMD002/POL

Vertical type optics bench OP002

Optics bench for ﬁber measurementwith 3-axis manual stage OP002-F3

Optics bench for ﬁber measurementwith 5-axis manual stage OP002-F5

COAXIAL EPI-ILLUMINATION SYSTEM
LED Coaxial epi-illumination system
This is LED type coaxial epi-illumination unit. In combination with Synos' CCD detector andoptical measurement optics M-Scope series with coaxial epi-illumination port, it is possible toobserve microscopic image of object, such as edge image of optical ﬁber, optical waveguide
module, and so on. There are monochromatic light type (465nm, 520nm, 645nm, 850nm, etc.)
and white light type.
【Summary of speciﬁcation】
Model
LS-EL001/R
LS-EL001/W
LS-EL001/BL
LS-EL001/GR
LS-EL002L/860
LS-EL002L/950

Wavelength
Red（630nm)
White（CCT5600K）
Blue（465nm）Green（520nm）
860nm
950nm

https://www.synos.jp

Head shape
Cylindrical (or L-shape)
Cylindrical (or L-shape)
Cylindrical (or L-shape)
Cylindrical (or L-shape)
L-shapeL-shape

Tip diameter
φ8mm
φ8mm
φ8mm
φ8mm
φ8mm
φ8mm

44

Controller type
CCS PJ-1505-2CA
CCS PJ-1505-2CA
CCS PJ-1505-2CA
CCS PJ-1505-2CA
CCS PJ2-1505-2CA-PE
CCS PJ2-1505-2CA-PE

LED coaxial epi-illumination light source
LS-EL001

SYNERGY OPTOSYSTEMS CO., LTD