取扱説明書

Ior 漏電監視ロガー

KEW

5050

共立電気計器株式会社

KEW5050

目次

KEW5050

目次

目次梱包内容の確認 ................................................................................................................................. 5
安全に関するご使用上の注意 ............................................................................................................... 7
1章
1.1
1.2
1.3

製品の概略.................................................................................................................................11
特長 ...............................................................................................................................................11
システム構成図 ........................................................................................................................12測定までの手順 ........................................................................................................................13

2章
2.1
2.2
2.3
2.4

各部の名称.................................................................................................................................14
表示部（LCD）／キー操作部 ...............................................................................................14
コネクタ部 ....................................................................................................................................15
側面部 ..........................................................................................................................................16電圧用測定コードとクランプセンサ ...................................................................................17

3章
3.1
3.2
3.3
3.4

基本操作 .....................................................................................................................................18
操作キーの説明 .......................................................................................................................18
LCD の上部に表示するマーク ............................................................................................19
画面の表示記号 .......................................................................................................................19
画面表示と画面構成 ..............................................................................................................20
全体図 ..........................................................................................................................................20
測定値（ベクトル） .....................................................................................................................21
イベント .........................................................................................................................................22
設定 ...............................................................................................................................................23

4章
4.1
4.2

漏洩電流について ...................................................................................................................24
漏洩電流（Io）測定 ...................................................................................................................24
対地抵抗成分漏洩電流（Ior）測定 ....................................................................................25
単相２線式 ..................................................................................................................................25
単相３線式 ..................................................................................................................................25
三相３線式 ..................................................................................................................................26
三相４線式 ..................................................................................................................................26
異容量結線の対地抵抗成分漏洩電流（Ioｒ）測定に関する注意...........................27

5章
5.1
5.2

測定前の準備 ............................................................................................................................27
識別マーカーを取り付ける ...................................................................................................27
電源について .............................................................................................................................27
電池の使用.................................................................................................................................27
乾電池のセット方法 ................................................................................................................28
画面の表示／電池の残量 ...................................................................................................29
ＡＣアダプタの使用 ..................................................................................................................30
ＡＣアダプタの接続 ..................................................................................................................31
ＳＤカードの挿入／取り出し方法 .......................................................................................32
挿入方法 .....................................................................................................................................32
取り出し方法 ..............................................................................................................................32

5.3

1

−1 −

KEW5050

KEW5050

KEW5050

目次

KEW5050

目次

5.4
5.5

電圧用測定コードとクランプセンサの接続 ....................................................................33
電源の投入.................................................................................................................................34
初期画面表示 ............................................................................................................................34
注意表示 .....................................................................................................................................34
測定対象への接続 ..................................................................................................................35
結線方法（結線方式の選択：結線図） .............................................................................36
記録の手順.................................................................................................................................38
記録の開始.................................................................................................................................38
記録の終了.................................................................................................................................38

5.6
5.7

6章
6.1
6.2

6.3

6.4

6.5

6.6

7章
7.1

7.2

KEW5050

KEW5050

設定 ...............................................................................................................................................39
項目の選択（移動） ..................................................................................................................39
詳細設定項目 ............................................................................................................................40
基本設定 .....................................................................................................................................41
センサ識別..................................................................................................................................41
電流クランプ/ｃｈ.......................................................................................................................42
周波数 ..........................................................................................................................................45
イベント設定 ...............................................................................................................................45
共通 ...............................................................................................................................................45
上限しきい値(Ｈ)/ｃｈ ...............................................................................................................46
下限しきい値(Ｌ) ........................................................................................................................48
ピークしきい値(Ｐｋ)/ｃｈ .........................................................................................................50
記録設定 .....................................................................................................................................53
記録方法 .....................................................................................................................................53
記録可能時間について .........................................................................................................56
保存データ ..................................................................................................................................57
記録データ ..................................................................................................................................57
本体設定 .....................................................................................................................................59
保存データの種類について .................................................................................................61
その他 ...........................................................................................................................................63
環境設定 .....................................................................................................................................63
本体システムに関する設定 .................................................................................................64
画面ごとの表示項目 ...............................................................................................................67
測定値 ..........................................................................................................................................67
系統ごとの測定値を一覧で表示する...............................................................................67
系統ごとのベクトル図を表示する ......................................................................................68
全体の測定値を一覧で表示する.......................................................................................69
イベント .........................................................................................................................................71
イベントの発生状況を表示する ..........................................................................................71

2

−2 −

KEW5050

目次

KEW5050

目次

8章

その他の機能 ............................................................................................................................75

9章
9.1
9.2
9.3

周辺機器との接続 ...................................................................................................................77
ＰＣ：コンピュータへのデータ転送 ......................................................................................77
外部機器との信号制御 .........................................................................................................78
出力端子を接続する...............................................................................................................78
測定ラインから電源を供給する..........................................................................................79

10 章

設定・解析用ＰＣソフトウェア ...............................................................................................80

11 章
仕様 ...............................................................................................................................................81
11.1 安全要求仕様 ............................................................................................................................81
11.2 一般仕様 .....................................................................................................................................81
11.3 測定仕様 .....................................................................................................................................83
測定項目別、解析データ数..................................................................................................83
瞬時測定項目 ............................................................................................................................84
演算項目 .....................................................................................................................................85
イベント項目 ...............................................................................................................................88
12 章
故障かなと思ったら .................................................................................................................89
12.1 トラブルシューティング ...........................................................................................................89
12.2 入力項目と表示項目の制限................................................................................................91
12.3 エラーメッセージの内容とその対処方法 ........................................................................91
13 章

アフターサービス ......................................................................................................................93

3

−3 −

KEW5050

KEW5050

KEW5050

KEW5050

目次−4 −KEW5050

梱包内容の確認

梱包内容の確認
このたびは弊社Ｉｏｒ漏電監視ロガーKEW5050 をご購入していただきありがとうございます。まずは、お手元に届きました本製品の梱包内容を確認してください。

●梱包内容
1

本体

KEW5050：1 台

2電圧用測定コードMODEL7273：1 セット 赤、黒、各 1 本

3AC アダプタ

MODEL8262：1 台

4アースコード

MODEL7278：1 本

5USB ケーブル

MODEL7219：1 本

6

ＳＤカード（2GB）7

CD-ROMPC ソフトウェア

8

電池

単３形アルカリ乾電池（LR6）：6 個

9キャリングバッグ

MODEL9125

10

識別マーカー

4 色×各 2 本（赤/黄/青/緑)

11

取扱説明書

本体用

12

インストールマニュアル

PC ソフトウェア用 ：1 枚

（ワニグチクリップ付き）

：1 枚
：1 枚
：1 個
：1 冊

1. 本体

2. 電圧用測定コード

3. AC アダプタ

4. アースコード

5. USB ケーブル

6. ＳＤカード(2GB)−5 −KEW5050

KEW5050

梱包内容の確認

7. CD-ROM

8. 電池

9. キャリングバッグ

10. 識別マーカー

11. 取扱説明書

12. インストールマニュアル＜収納方法＞●製品の間違い、品不足、破損、印刷不良等がございましたら、お買上店（販売店）
までご連絡をください。
●取扱説明書には保証書が付いています。大切に保管してください。

KEW5050−6 −KEW5050

安全に関するご使用上の注意

KEW5050

安全に関するご使用上の注意●オプション
Ior 用リーク電流検出型

1

クランプセンサ

2

電源供給アダプタ

1. Ior 用リーク電流検出型
クランプセンサ

KEW8177

KEW8178(φ40mm)(φ68mm)

KEW8177（10A/φ40mm）

KEW8178（10A/φ68mm）

MODEL8329 （CATⅢ150V、CATⅡ240V）
2.電源供給アダプタ

安全に関するご使用上の注意
本製品は IEC 61010：電子測定装置に関する安全規格に準拠して、設計・製造の上、検査合格した最良の状態にて出荷されています。
こ の 取 扱 説明書 に は 、 使 用さ れ る 方 の 危 険 を 避 け る た め の 事項 お よ び 本 製品 を 損 傷 さ せ ずに 長 期 間 良 好な 状 態 で 使用して頂くための事柄が書かれています。ご使用前に必ずこの取扱説明書をお読みください。

警告
取扱説明書について●本製品を使用する前に必ずこの取扱説明書をよく読んでご理解ください。
●この取扱説明書は手近な所に保管し、必要な時にいつでも取り出せるようにしてください。
●取扱説明書で指定した製品本来の使用方法を守ってください。
●取扱説明書の安全に関する指示に対しては、指示内容を理解の上、必ず守ってください。
●使用するクランプセンサの取扱いについては、クランプセンサの取扱説明書も必ずよくお読みになって、
ご理解ください。
以上の指示を必ず厳守してください。指示に従わないと、怪我や事故の恐れがあります。危険および警告、注意に反した使用により生じた事故や損傷については、弊社として責任と保証を負いかねます。

7

−7 −

KEW5050

KEW5050

KEW5050

安全に関するご使用上の注意

KEW5050

安全に関するご使用上の注意
マークは、安全に使用するため取扱説明書を読む必要性を表しています。尚、この本製品に表示のマー

クには次の 3 種類がありますので、それぞれの内容に注意してお読みください。

危険 ： この表示を無視して誤った取扱いをすると、人が死亡または重傷を負う危険性が高い内容を示しています。

警告 ： この表示を無視して誤った取扱いをすると、人が死亡または重傷を負う可能性が想定される内容を表示しています。

注意 ： この表示を無視して誤った取扱いをすると、人が傷害を負う可能性が想定される内容および物的損害の発生が想定される内容を示しています。

○測定カテゴリについて
安全規格 IEC61010 では測定器の使用場所についての安全レベルを測定カテゴリという言葉で規定し、以下のように
O～CAT Ⅳの分類をしています。この数値が大きいほど過渡的なインパルスが大きい電気環境であることを意味します。CAT Ⅲで設計された測定器は CAT Ⅱで設計されたものより高いインパルスに耐えることができます。
O(None, Other) ： 主電源に直接接続しないその他の回路です
CAT II ： コンセントに接続する電源コード付機器の１次側の電気回路
CAT III ： 直接配電盤から電気を取り込む機器の１次側および分岐部からのコンセントまでの電路
CAT IV ： 引込み線から電力量計および１次過電流保護装置（配電盤）までの電路危険●指定した操作方法および条件以外で使用した場合、本体の保護機能が正常に動作せず本器を破損したり感電等の重大な事故を引き起こす可能性があります。本器の使用前あるいは指示結果に対する対策を取る前に、既知の電源で正常な動作を確認してください。
●本体の対地間最大定格電圧は、測定カテゴリに準じて CAT Ⅳで AC300V、CAT Ⅲでは AC600V です。これより高い対地間電圧のある回路では絶対に使用しないでください。
●引火性ガスや爆発性のガスおよび、蒸気のある場所で使用すると大変危険ですので、使用しないでください。
●本製品や手が濡れている状態や、湿気などの水滴が付着した状態では、絶対に使用しないでください。
測 定 に つい て●測定の際には、測定範囲を超える入力を加えないでください。
●測定中は絶対に電池蓋を開けないでください。

KEW5050

KEW5050

8−8 −KEW5050

安全に関するご使用上の注意

KEW5050

安全に関するご使用上の注意

危険
電圧用測定コードについて●付属のものをご使用ください。
●測定コードと本体の測定カテゴリが違っている場合は低い方の測定カテゴリが優先されます。測定電圧と定格が合っているか必ず確認してください。
●本体に接続していない状態で測定ラインに接続しないでください。
●測定の際は指先等が、保護用フィンガガードを超えることのないよう充分注意してください。
保護用フィンガガード：操作中の感電事故を防ぐため、最低限必要な沿面および空間距離を確保するための目印です。
●測定中（測定ラインからの通電中）は絶対に本体のコネクタから取りはずさないでください。
●先端の金属部で測定ラインの 2 線間を接触させないでください。
●先端の金属部には絶対に触れないでください。
●測定コードのコード内部から金属部分または外装被覆と異なる色が露出したときは、直ちに使用を中止してください。
クランプセンサについて●本製品専用のものをご使用ください。
●測定電流と定格が合っているか必ず確認し、対地間最大定格電圧以下の電路で使用してください。
●Ior 用クランプセンサ(KEW8177/8178)の測定カテゴリは CAT Ⅲ300V です。本体の基準電圧入力端子の測定カ
テゴリ CAT Ⅳ300V、CATⅢ600V とは測定できる場所が違います。基準電圧入力端子の測定場所と同じ場所
を Ior 用クランプセンサで測定する場合は低い方の測定カテゴリが優先されるため CAT Ⅲ300V までです。
●測定に必要のないものは絶対に接続しないでください。
●本体に接続していない状態で測定ラインに接続しないでください。
●測定の際は指先等が、バリアを超えることのないよう充分注意してください。
バリア：操作中の感電事故を防ぐため、最低限必要な沿面および空間距離を確保するための目印です。
●測定中（測定ラインからの通電中）は絶対に本体のコネクタから取りはずさないでください。
●必ずブレーカーの二次側に接続してください。1 次側は電流容量が大きく危険です。
●コアを開いたとき、金属部で測定ラインの 2 線間を接触させないでください。
電池について●測定中は絶対に電池交換を行わないでください。
AC アダプタについて●電源コードが AC アダプタの奥まで確実に差し込まれていることを確認してください。
●使用する AC アダプタおよび電源コードは、必ず本体に付属している MODEL8262 をご使用ください。
●AC アダプタの定格電源電圧は AC100V～AC240V です。ただし、付属の電源コード MODEL7169 を使用する場
合には、電源コードの最大定格電圧 AC125V 以下の電圧にてご使用ください。
●AC アダプタの定格電源周波数は 50Hz/60Hz です。
●使用する電源電圧と周波数とが合っているか必ず確認し、AC240V より高い電位のある場所には、絶対に接続
しないでください。AC アダプタまたは KEW5050 本体の破損および、電気事故に繋がる可能性があります。
アースコードについて●本体 KEW5050 を接地する場合には、必ず付属のアースコードを使用して既知の接地端子へ接続してください。
アースコードは高い電圧に対して保護されていません。絶対に活線へは接続しないでください。本体の破損および、電気事故に繋がる可能性があります。

9

−9 −

KEW5050

KEW5050

KEW5050

安全に関するご使用上の注意

KEW5050

安全に関するご使用上の注意

警告●本製品を使用する前に必ず既知の電源で正常に測定できることを確認してください。

注意●被測定導線が高温の場合がありますので注意してください。
●各レンジの測定範囲を超える電流や電圧を入力しないでください。
●電源が OFF の状態で、電圧用測定コードやクランプセンサに電圧や電流を入力しないでください。
●埃の多い場所や、水のかかる環境で使用しないでください。
●強力な電磁波が発生したり、帯電したりしているものの近くで使用しないでください。
●振動や衝撃を与えたり、落下させたりしないでください。
●ＳＤカードの向きを必ず確認して正しい方向で本体へ挿入してくだい。無理に挿入しようとするとＳＤカードまたは、本体を破損する恐れがあります。
●ＳＤカードを挿入／取り出すときは必ず、ＳＤカードへアクセス中でないことを確認してください。
（アクセス中はが点滅します。）ＳＤカードへアクセス中に取り出しを行うと、保存されたデータや本体を破損する恐れがあります。
クランプセンサについて●クランプセンサのケーブルを折ったり引っ張ったりしないでください。
電 池 に つい て
●銘柄や種類の違う電池を混ぜて使用しないでください。
使用後について●必ず電源を OFF にし、電源コード、電圧用測定コードおよびクランプセンサを外してください。
●長期間ご使用にならない場合は、電池を取り外した状態で保管してください。
●持ち運ぶときは、ＳＤカードを本体から抜いてください。
●運搬の際には振動や衝撃を与えたり、落下させたりしないでください。
●高温多湿、結露するような場所および直射日光の当たる場所に放置しないでください。
●クリーニングには研磨剤や溶剤を使用しないで、中性洗剤か水に浸した布を使ってください。
●濡れているときは、乾燥後保管してください。
また、各章の危険、警告、注意、注記の内容も必ず守ってください。

本製品に使用している安全記号
取扱説明書を参照する必要があることを示します。
二重絶縁または強化絶縁で保護されている機器を示します。

～

交流(AC)を示します。
（機能）接地端子を示します。
本製品は、WEEE 指令（2002／96／EC）マーキング要求に準拠します。この電気電子製品を一般家庭廃棄物として廃棄してはならないことを示します。

KEW5050

KEW505010− 10 −

1.1 特長

KEW5050

1.1 特長KEW5050

1章 製品の概略
1.1

特長

本製品は多彩な結線方式に対応したクランプ式 Ior 漏電監視ロガーです。従来の高調波を含む漏洩電流実効値
Iom、電圧実効値 Vm に加え、基本波の周波数成分のみを抽出した漏洩電流値実効値 Io、対地抵抗成分漏洩電流
Ior、電圧実効値 V、電圧周波数 f、参考値として電圧電流位相差θおよび絶縁抵抗値 R をすべて同時に測定してＳ
Ｄカードに記録します。また、電流値や電圧値に対してイベントのしきい値を設けることで、記録インターバルの設定に関係なく瞬間の漏電を記録できます。
結線方式
単相 2 線、単相 3 線、三相 3 線、三相 4 線の結線図を表示して、各種測定ラインごとに 4 系統まで選択できます。
なお、電圧と電流のベクトル表示からクランプセンサの向きを参照して結線を間違えていないか確認することもできます。
高調波キャンセル
独自の演算方法により、公称周波数帯域の基本波のみから測定値を求めています。そのため高調波が重畳した漏洩電流であっても測定値に影響しません。※漏洩電流実効値 Iom、基準電圧実効値 Vm は高調波を含んだ値です。
インターバル測定
最短 200ms ごとのすべての測定データをギャップなしで連続して保存するため、間欠リークの検出に最適です。
200ms を超える記録間隔では、その間の最大・最小・平均・瞬時値をインターバルごとに保存します。
イベント測定電流実効値・電圧実効値および、パルス状の瞬時値に対してイベントのしきい値を設けることで、インターバルの測定データに加え瞬間の漏電と発生日時とをすべて同時に記録することができます。
測定データの保存
記録間隔が設定可能なロギング機能を搭載しています。記録開始と終了は手動の他に、日付や時間帯を指定して測定データを自動で保存することもできます。また、プリントスクリーン機能で現在表示している画面をＳＤカードへ保存します。
ベクトル表示基準電圧 V と漏洩電流 Io との位相関係をベクトルで表示することができます。
２つの電源方式
AC アダプタと電池のどちらでも駆動できる 2 電源方式です。電池は、単３形アルカリ乾電池（ＬＲ６）と市販の充電式
単３形ニッケル水素電池(Ni-MH)の使用が可能です。（※充電式単３形ニッケル水素電池(Ni-MH)への充電は、ご使用になる電池メーカの充電器をご使用ください。本体での充電は行えません。）
アプリケーション
ＳＤカードに保存したファイルを USB 経由でパソコンへダウンロードできます。ダウンロードしたファイルは、付属のＰＣソフト（KEW Windows for KEW5050）で簡単に解析することが可能です。またパソコンから本製品の設定を簡単に変更することもできます。
外部信号出力機能
イベントの発生を接点信号としてデジタル出力から警報装置等に送ることができます。
安全設計
安全規格 IEC 61010-1 CAT Ⅳ 300V／CAT Ⅲ 600V に準拠した安全設計です。

11− 11 −

KEW5050

KEW5050

KEW5050

1.2 システム構成図

KEW5050

1.2

1.2 システム構成図

システム構成図

基準電圧入力

電流入力

・・・

接地端子

デジタル出力アースコード

警報装置等

ＵＳＢ

単３形アルカリ乾電池(ＬＲ６)単３形ニッケル水素電池（Ｎi-ＭＨ）

SD カードＰＣ
コンセント

AC アダプタ

KEW5050

KEW505012− 12 −

1.3 測定までの手順

KEW5050

1.3 測定までの手順

1.3

KEW5050測定までの手順
ご使用になる前に必ず「「安全に関するご使用上の注意」(7 頁)をお読みください。
測定準備「5 章 測定前の準備」27 頁

本体へコードとセンサを接続「5.4 電圧用測定コードとクランプセンサの接続」33 頁

電源を入れる「5.5 電源の投入」34 頁

本体設定の読み込み「本体設定」59 頁

各測定共通の項目を設定する「6.2 基本設定」41 頁

測定ラインへ結線「5.6 測定対象への接続」35 頁

結線を確認する「系統ごとのベクトル図を表示する」68 頁

測定ごとの設定と保存方法を設定する「6.3 イベント設定」45 頁／「6.4 記録設定」53 頁

SD カードの空き容量と測定値を確認する「記録データ」57 頁／「7.1 測定値」67 頁

記録を開始する／記録を終了する「5.7 記録の手順」38 頁

イベントの発生状況を確認する「イベントの発生状況を表示する」71 頁

測定ラインからコードとセンサを外し、電源を切る

PC で記録データを解析する「9.1 ＰＣ：コンピュータへのデータ転送」77 頁「10 章 設定・解析用ＰＣソフトウェア」80 頁

13− 13 −

KEW5050

KEW5050

KEW5050

2.1 表示部（LCD）／キー操作部

KEW5050

2.1 表示部（LCD）／キー操作部

2章 各部の名称
2.1

表示部（LCD）／キー操作部

表示部（LCD）

キー操作部

KEW5050

KEW505014− 14 −

2.2 コネクタ部

KEW5050

2.2 コネクタ部

2.2

KEW5050コネクタ部
電流入力端子カバー
基準電圧入力端子（N(T)、L(R)）

電流入力端子（A１、A２、A３、A４）

基準電圧入力端子結線方式※1

電流入力端子(×系統数※2)

単相 2 線

1P2W

N、L

A1(～A4)

単相 3 線

1P3W

N、L1

A1(～A4)

三相 3 線

3P3W

T、R

A1(～A4)

三相 4 線

3P4W

N、R

A1(～A4)電圧・電流ロギング

V・A

N(T)、L(R)

A1(～A4)※1電流測定のみの場合でも必ず基準電圧を接続してください。測定誤差が大きくなり性能を保証できません。

※2

複数の系統を同時に測定する場合、必ず「A1」側から順に測定系統数分のクランプセンサを接続してください。

15− 15 −

KEW5050

KEW5050

KEW5050

2.3 側面部

KEW5050

2.3

2.3 側面部

側面部＜コネクタカバーを閉じた状態＞

USB 部

デジタル出力部
SD カード部＜コネクタカバーを開いた状態＞

USB コネクタ

デジタル出力端子

USB コネクタ

デジタル出力端子
SD カード部

KEW5050

KEW505016− 16 −

2.4 電圧用測定コードとクランプセンサ

KEW5050

2.4 電圧用測定コードとクランプセンサ

KEW5050

AC アダプタ接続コネクタ
2.4

電圧用測定コードとクランプセンサ＜ワニグチクリップ＞※

電圧用測定コード先端部保護用フィンガガード＜クランプセンサ＞

バリア
保護用フィンガガードとバリアは操作中の感電事故を防ぐため、最低限必要な沿面および空間距離を確保するための目印です。
測定の際は指先等が、保護用フィンガガードとバリアを超えることのないよう充分に注意してください。
電圧用測定コードのワニグチクリップは被覆の上からは測定できません。

17− 17 −

KEW5050

KEW5050

KEW5050

3.1 操作キーの説明

3章 基本操作3.1

操作キーの説明ホールド/キーロックキー
表示値をホールドします。
※
ホールド時も測定は継続しています。

画面保存キーLCD に表示中の画面を、BMP(ビットマップ)ファイルとして保存します。

キーの誤操作を防止したい場合には 2 秒以上の長押しで、すべてのキー操作を無効にできます。
再度 2 秒以上長押しすると解除します。
カーソルキー
選択項目の移動、表示の切り換えに使用します。
決定キー
選択を決定します。
戻るキー
カーソルキーで設定選択したものを決定せずに前回の設定に戻します。
ステータス LED

イベントキー
イベントの発生状況を表示します。

緑

点灯：記録測定中
点滅：スタンバイ中
記録/停止キー

設定キー
結線／基本設定／測定設定／記録設
定／保存データ編集／その他の設定を変更します。
電源キー
電源を ON／OFF します。

KEW5050− 18 −

測定を開始／終了します。

3.2 LCD の上部に表示するマーク

3.2

KEW5050

LCD の上部に表示するマーク
マーク

表示時の状態
電池で駆動しています。残量によって 4 種類に変化します。
AC アダプタで駆動しています。
画面の表示更新をホールドしています。
キーがロックされています。
ＳＤカードが使用可能です。
ＳＤカードに記録中です。
ＳＤカードに記録できるだけの容量がありません。
ＳＤカードへのアクセスエラーです。
記録待機状態です。
測定値を記録中です。
記録対象の媒体が、いっぱいになっています。
USB が使用可能です。

3.3

画面の表示記号
画面表示記号

Iom
Vm
θ

R

高調波成分を含む
漏洩電流実効値
高調波成分を含む
基準電圧実効値
位相角

－

遅れ

＋

進み

絶縁抵抗値
(参考値)

Io
V

50/60Hz 基本波のみの漏洩電流実効値
50/60Hz 基本波のみの基準電圧実効値

Iorｆ

対地抵抗成分のみの
漏洩電流実効値
基準電圧の周波数

V:基準電圧実効値(基本波)の位相角を基準(0.0°)として、Io:漏洩電流実効値(基本波)の位相角を表示します。
「V:基準電圧実効値(基本波)÷Ior:漏洩電流実効値(対地抵抗成分)」
を絶縁抵抗値として表示します。注意)絶縁抵抗計とは測定方法が異なるため、同じ抵抗値にならないことがあります。参考値としてご使用ください。

記号のみの表示は全ｃｈの合計値になります。記号の後につく数字がｃｈの番号です。

− 19 −

KEW5050

KEW5050

3.4 画面表示と画面構成

KEW5050

3.4

3.4 画面表示と画面構成

画面表示と画面構成
全体図
測定値

ベクトル

イベント

結線図

詳細設定

結線図を確認しながら、クランプセンサと電圧用測定コードを正しく結線してください。
電圧用測定コードのワニグチクリップは被覆の上からは測定できません。

KEW5050

KEW505020− 20 −

3.4 画面表示と画面構成

KEW5050

測定値（ベクトル）※例は単相 2 線式

− 21 −

KEW5050

KEW5050

3.4 画面表示と画面構成

イベント
表示項目の切り替え+“◀”“▶”に囲まれている範囲で移動できます。

※下図の例は４系統(Ａ１～Ａ４)に対して全イベントを検出した状態です。

KEW5050− 22 −

3.4 画面表示と画面構成

KEW5050

3.4 画面表示と画面構成

KEW5050設定
注記●電流だけを測定したい場合でも必ず電圧用測定コードを結線してください。測定値を演算するタイミングが特定できなくなるので、測定値が安定しません。
●異容量接続の三相4線式V結線、デルタ結線では、対地抵抗成分漏洩電流(Ior)と対地容量成分漏洩電流(Ioc)との位相が重なることがあります。そのため、IoからIocのみを除くことができないため、Iorを正確に測定することができません。また、主に病院等で用いられる非接地配線方式（フローティング電源）では漏洩
電流自体が流れないために、測定できません。

表示項目の切り替えで移動します。

画面に表示した結線図を確認しながら、クランプセンサと電圧用測定コードを正しく結線してください。
電圧用測定コードのワニグチクリップは被覆の上からは測定できません。

23− 23 −

KEW5050

KEW5050

KEW5050

4.1 漏洩電流（Io）測定

KEW5050

4.1 漏洩電流（Io）測定

4章 漏洩電流について
一般的に絶縁監視装置などにて判定基準とされている漏洩電流（Io）の測定値には、火災や感電、電力損失の原因となる対地抵抗成分漏洩電流（Ior）と、危険な状態とならない対地容量成分漏洩電流（Ioc）とが含まれています。
そのため、配線距離が長い設備やインバータ機器が接続されている設備など、Ioc が大きい設備では絶縁状態を正確に判断でない場合があります。

Io

Ioc

単相２線式の漏洩電流（Io）の例
Ioｒ

ベクトル：Io = Ior + Ioc

V＜Ioc が無視できる理由＞
Ior は電圧(V)と同じ位相（位相差 0°）で流れるため、有効電力に換算すると次の式のようになります。
有効電力(P) = V×Ior×cos0°(cos0°= 1） = V×Ior
つまり電力として消費されるため、場合によっては発熱や発火につながる可能性があり、大変危険です。
これに対して対地容量成分漏洩電流（Ioc）は、電圧(V)に対して 90°進んだ位相で流れるため、有効電力に換算す
ると次の式のようになります。
有効電力(P) = V×Ioc×cos90°(cos90°= 0） = 0
つまり電力がゼロとなり、危険な状態になりません。

4.1

漏洩電流（Io）測定
フーリエ変換(FFT)を使用して高調波を除いた漏洩電流の基本波(電源公称周波数 50Hz/60Hz の１次成分)から求めています。
Io = �Io_kr � + Io_ki�

_kr：FFT 変換後の実数成分、_ki：FFT 変換後の虚数成分、k = 1：FFT 解析次数(１次)KEW5050

KEW505024− 24 −

4.2 対地抵抗成分漏洩電流（Ior）測定

KEW5050

4.2 対地抵抗成分漏洩電流（Ior）測定

4.2

KEW5050

対地抵抗成分漏洩電流（Ior）測定
単相２線式
Io

Ioc

Ioｒ

V

Io および基準電圧実効値(V)の虚数と実数とから、有効電力値 P を求め、そこから V を除くことで Ior だけを求めています。
P_k = V_kr×Io_kr＋V_ki×Io_ki
V=
Ior =

V_kr2＋V_ki2_kr：FFT 変換後の実数成分、_ki：FFT 変換後の虚数成分、k = 1：FFT 解析次数(１次)

単相３線式
Ioc_L1

Ioc
L2

Io
Ioｒ_L1

Ioｒ_L2
Ioｒ

L1
V

Ioc_L2
Ior_L1、Ior_L2 は L1 相、L2 相ごとの対地抵抗成分漏洩電流、Ioc_L1、Ioc_L2 は L1 相、L2 相ごとの対地容量成分漏洩電流を示しています。L1 相と L2 相とが同時に絶縁劣化した場合には、L1 相と L2 相の電圧がそれぞれ逆位相のため、Ior が打ち消されることになりますが、L1 相と L2 相とが同じ程度、同時に劣化することは極稀であるため、単相ごとで判断することができます。劣化した相を特定するには、ベクトル図を参考にしてください。
Io のベクトル方向によって劣化の可能性がある相を推測できます。
Io および基準電圧実効値(V)の虚数と実数とから、有効電力値 P を求め、そこから V を除くことで Ior だけを求めています。
P_k = V_kr×Io_kr＋V_ki×Io_ki
V=
Ior =

V_kr2＋V_ki2_kr：FFT 変換後の実数成分、_ki：FFT 変換後の虚数成分、k = 1：FFT 解析次数(１次)
結線図を確認しながら、クランプセンサと電圧用測定コードを正しく結線してください。
電圧用測定コードのワニグチクリップは被覆の上からは測定できません。

25− 25 −

KEW5050

KEW5050

KEW5050

4.2 対地抵抗成分漏洩電流（Ior）測定

KEW5050

4.2 対地抵抗成分漏洩電流（Ior）測定

三相３線式
T

R
Q÷V

Io

Ioｒ = Ior_T＋Ior_R
Ior_R
Ioc_R

Ioc

Ior_T

V

S
Ioc_T

Ior_R、Ior_T は R 相、T 相ごとの対地抵抗成分漏洩電流、Ioc_R、Ioc_T は R 相、T 相ごとの対地容量成分漏洩電流を示しています。Io および基準電圧実効値(V)の虚数と実数とから、無効電力値 Q を求め、そこから V を除いた値を基準とします。Ior は R 相電圧および T 相電圧に対して同位相で流れ、Ioc は Ioc_R と Ioc_T とが平衡状態であるとすると V に対して逆位相で流れるため、上図のようなベクトル関係となり以下の式にて Ior が求まります。
Q_k = V_kr×Io_ki＋V_ki×Io_kr
V=
Ior =

V_kr2＋V_ki2√

×_kr：FFT 変換後の実数成分、_ki：FFT 変換後の虚数成分、k = 1：FFT 解析次数(１次)
劣化した相を特定するには、ベクトル図を参考にしてください。Ｉｏのベクトル方向によって劣化の可能性がある相を推測できます。
注意）Io が R 相電圧と T 相電圧の位相間にある場合には Ior≧Io≧(Q÷V)となるため Io よりも Ior が大きくなります。Ioc_R と Ioc_T とが不平衡状態の場合には測定に誤差が生じます。

三相４線式
Ior_R、Ior_S、Ior_T は R 相、S 相、T 相ごとの対地抵抗成分漏洩電流、Ioc_R、Ioc_S、Ioc_T は R 相、S 相、T 相ごとの対地容量成分漏洩電流を示しています。各相の Ioc が平衡状態であるとすると、その合計はゼロになるため、無視することができます。つまり Io と Ior とは等しくなります。
R
V
Ior_R

Ioｒ = Io

Ioc_S
Ioc_R
Ior_T

N
Ioc_T

T

S

劣化した相を特定するには、ベクトル図を参考にしてください。Ｉｏのベクトル方向によって劣化の可能性がある相を推測できます。
結線図を確認しながら、クランプセンサと電圧用測定コードを正しく結線してください。
電圧用測定コードのワニグチクリップは被覆の上からは測定できません。
KEW5050

KEW505026− 26 −

5.1 識別マーカーを取り付ける

KEW5050

異容量結線の対地抵抗成分漏洩電流（Ioｒ）測定に関する注意
異容量接続の三相４線式 V 結線、デルタ結線では、Ior と Ioc との位相が重なる場合があります。そのため、Ioから Ioc のみを除くことができないため、Ior を正確に測定することができません。
また、主に病院等で用いられる非接地配線方式（フローティング電源）では漏洩電流自体が流れないために、測定できません。

5章 測定前の準備
識別マーカーを取り付ける
複数のクランプセンサを使用する場合には、電流入力端子の識別色（赤 A1／黄 A2／青 A3／緑 A4）に対応した
識別マーカーをクランプセンサの両端に取り付けます。識別マーカーは４色（赤・青・黄・緑）２本セットの計８本です。
識別マーカー

クランプセンサの両端に取り付けます（４色２本セット、計８本）

電源について本製品は、AC アダプタ／電池駆動の 2 電源方式です。
停電などが原因で AC アダプタから供給が止まった場合でも、電源の供給を電池に切り換えて測定を行います。

電池の使用
単３形アルカリ乾電池（LR6）と単３形ニッケル水素電池(Ni-MH)の使用が可能です。
充電式電池への充電は、ご使用になる電池メーカの充電器をご使用ください。本体での充電は行えません。
※

単３形アルカリ乾電池（ＬＲ６）が付属品となっています。

危険●測定中は絶対に電池交換を行わないでください。
●電池駆動時の AC アダプタ接続コネクタは絶縁されていますが、絶対にさわらないでください。

警告●電池の交換の際には電源コード、電圧用測定コードおよびクランプセンサを本体からはずし、電源を OFF にしてください。

− 27 −

KEW5050

KEW5050

5.2 電源について

KEW50505.2 電源について注意●銘柄や種類の違う電池を混ぜて使用しないでください。
●電池は古いものと混ぜて使用しないでください。
●電池の極性をまちがえないよう、ケース内の彫刻の向きに合わせて入れてください。

注記
本製品のご購入時は電池が内蔵されておりません。必ず付属の電池をセットしてください。
電源 OFF の状態でも電池を消費しますので、長時間使用されない場合は電池を抜き取って保管してください。

乾電池のセット方法
以下の手順で乾電池をセットします。

電池蓋

２

５

ネジ

２
５

単３形アルカリ乾電池：ＬＲ６３１AC アダプタ、アースコード、電圧用測定コードおよびクランプセンサなど、すべてのコードを本体からはずして電源を切ります。

２
３
４
５

KEW5050

KEW5050４

本体裏側のネジ２個を緩めて電池蓋をはずします。
電池をすべて取りはずします。
正しい極性で単３形アルカリ乾電池：ＬＲ６を必ず６本セットします。
電池蓋を取り付けて、ネジ２個を締めます。

28− 28 −

5.2 電源について

KEW5050

画面の表示／電池の残量
画面右上に表示した電源アイコンが、電源の状態によって下記のように変化します。
4 分割

AC アダプタ駆動

単３形アルカリ乾電池の場合は約１１時間測定可能です。
単３形ニッケル水素電池(1900ｍA/h)の場合はフルに充電
電池駆動

された状態から約９時間測定できます。
注記
充電式電池への充電は、ご使用になる電池メーカの充電器をご使用ください。本体での充電は行えません。
通常測定 ※ニッケル水素電池は満充電時の電圧が、アルカリ乾電池に比べて低いため、左の表示以上にならないことがあります。
測定は続行しますが、保存は終了します。
(終了までの測定データは保存されます。)− 29 −

KEW5050

KEW5050

5.2 電源について

ＡＣアダプタの使用
付属の AC アダプタを使用すると長時間のロギングを行えます。AC アダプタをご使用になる場合でも電池を入れた状態でご使用ください。停電などで AC アダプタから電源が供給できない場合には、自動的に電池駆動に切り換えて測定を継続します。AC アダプタおよび、電源コードの定格は下表のとおりです。

電源コード
定格電源電圧125V AC

定格電源電流7A max

ＡＣアダプタ「MODEL8262」
定格電源電圧100～240V AC（±10％）定格電源周波数50/60Hz

最大消費電力20VA max

必ず確認してください。

危険●使用する AC アダプタおよび電源コードは、必ず本体に付属した製品をご使用ください。
●付属の電源コードを AC125V より高い電位のある場所には、絶対に接続しないでください。
●使用する電源電圧と周波数とが合っているか必ず確認し、付属の AC アダプタを AC240V(50Hz/60Hz)より高い電位のある場所には、絶対に接続しないでください。AC アダプタまたは KEW5050 本体の破損および、電気事故に繋がる可能性があります。
●本体 KEW5050 を接地する場合には、必ず付属のアースコードを使用して既知の接地端子へ接続してください。
アースコードは高い電圧に対して保護されていません。絶対に活線へは接続しないでください。本体の破損および、電気事故に繋がる可能性があります。

警告●本体の電源が OFF になっていることを確認してから接続してください。
●接続は必ず先に本体側から行い、奥まで確実に差し込んでください。
●使用しているうちに亀裂が生じたり、金属部分が露出したときは、直ちに使用を中止してください。
●本製品を使用しない場合は、電源コードをコンセントから抜いてください。
●電源コードのプラグをコンセントから抜くときは必ず差し込みプラグを持って抜いてください。

注記●AC アダプタを使用して測定を行う場合には、必ず付属のアースコードを使用して、接地した状態でご使用ください。接地していない状態では、測定値が安定しません。
●AC アダプタから供給がある場合は、電池を消耗しません。
●本体に電池が内蔵されていない状態で AC アダプタからの供給が止まった場合、本体の電源が切れ、記録のデータが失われる可能性があります。充分注意してください。

KEW5050− 30 −

5.2 電源について

KEW5050

ＡＣアダプタの接続
以下の手順でＡＣアダプタを接続します。

１
２
３
４

本体に電源が入っていないことを確認します。
電源コードが AC アダプタに確実に差し込まれていることを確認します。
AC アダプタの出力端子をアースコードの雌口コネクタに接続します。
アースコードの接続端子を本体の AC アダプタ接続コネクタに奥まで確実に差し込みます。

※

ＰＣへ接続してのデータ解析時など、測定を行わない場合には、本体の AC アダプタ接続コネクタに、直接AC アダプタの出力端子を接続してご使用になれます。

５

アースコードのクリップを既知の接地端子に接続します。

６

電源コードをコンセントに接続します。

危険：接続する前に必ず接地端子であることを確認し、絶対に活線へ接続しないでください。

接地端子

５

アースコードクリップ

１ 本体
アースコード雌口コネクタ
３
ＡＣアダプタ出力端子４アースコード接続端子

２

電源コード

コンセント
６
電源コード

ＡＣアダプタ ２

オプションの電源供給アダプタ「MODEL8329」を使用して対地間電圧 240V 以下の測定ラインから、電源を供給することができます。詳細は「9.3 測定ラインから電源を供給する」 79 頁を参照してください。

− 31 −

KEW5050

KEW5050

5.3 ＳＤカードの挿入／取り出し方法

KEW5050

5.3

5.3 ＳＤカードの挿入／取り出し方法

ＳＤカードの挿入／取り出し方法
必ず確認してください。

注意 「挿入方法」に従って正しい向きでＳＤカードを本体に挿入してください。間違った方向で挿入するとＳＤカードまたは、本体が破損する恐れがあります。
 ＳＤカードを挿入／取り出すときは必ず、ＳＤカードへアクセス中でないことを確認してください。
（アクセス中はが点滅します。
）ＳＤカードへアクセス中に取り出しを行うと、保存されたデー
タや本体を破損する恐れがあります。
 記録中（が点滅します。
）はＳＤカードを取り出さないでください。保存されたデータや本体を破損する恐れがあります。取り出す場合には、必ず記録を終了して「記録を停止しました」のメッセージが消えた後に取り出してください。

注記●使用するＳＤカードは、必ず本体に付属した製品または、オプションのＳＤカードを使用してください。
●新しいＳＤカードを使用する場合には、必ず本体にてフォーマットを行ってください。ＰＣでフォーマットを行うと正しく記録できないことがあります。詳細は「フォーマットする」58 頁を参照してください。
●頻繁に使用したＳＤカードや長期間使用したＳＤカードは、フラッシュメモリの寿命からデータを記録できなくなることがあります。記録できないＳＤカードは新しいものと取り換えてください。
●ＳＤカードの記録データは、事故や故障によって、消失または変化してしまうことがあります。記録したデータは定期的にバックアップを取って保存してください。なお、データが消失または変化した場合の損害につきましては、原因および、内容にかかわらず保証できません。あらかじめご了承ください。

挿入方法
１
２
３

コネクタカバーを開きます。
ＳＤカードの向きを確認し表面を上にした状態で図の矢印の方向へコネクタの奥まで押し込みます。
コネクタカバーを閉じます。特に必要のない場合には必ずコネクタカバーを閉めて使用してください。

取り出し方法１２
３
４コネクタカバーを開きます。
ＳＤカードを奥に押すとカードを取り出せる状態になります。
ＳＤカードをつまんで引き抜いてください。
コネクタカバーを閉じます。特に必要のない場合には必ずコネクタカバーを閉めて使用してください。

SD カードコネクタカバー
KEW5050

KEW505032− 32 −

5.4 電圧用測定コードとクランプセンサの接続

KEW5050

5.4 電圧用測定コードとクランプセンサの接続

5.4

KEW5050

電圧用測定コードとクランプセンサの接続必ず確認してください。

危険●電圧用測定コードは付属の専用コードを使用してください。
●クランプセンサは本製品対応のものを使用してください。また、測定電流と定格が合っているか必ず確認してください。
●測定に必要のないクランプセンサは絶対に接続しないでください。
●本体に接続していない状態で測定ラインに接続しないでください。
●測定中（測定ラインからの通電中）は絶対に本体のコネクタから取りはずさないでください。

警告●本体の電源が OFF になっていることを確認してから接続してください。
●接続は必ず先に本体側から行い、奥まで確実に差し込んでください。
●使用しているうちに亀裂が生じたり、金属部分が露出したときは、直ちに使用を中止してください。

以下の手順で電圧用測定コードおよびクランプセンサを接続します。

１

本体に電源が入っていないことを確認します。

２

本体の基準電圧入力端子へ電圧用測定コードを接続します。

３ 本体の電流入力端子へ測定に必要なクランプセンサを接続します。このときクランプセンサの出力端子の矢印と本体の電流入力端子の矢印が向き合うように接続してください。

矢印の向きに合わせる

クランプセンサの使用数は、同時に測定する系統数によって異なります。
「結線図」37 頁を参照してください。
電圧用測定コードのワニグチクリップは被覆の上からは測定できません。

33− 33 −

KEW5050

KEW5050

KEW5050

5.5

5.5 電源の投入

電源の投入
初期画面表示注記●を押しても電源が切れない場合、画面にが設定されています。

を２秒以上押して、を表示していないか確認してください。「キーロック機能」
を消してから

を２秒以上押すと電源を切ることができます。

画面が表示されるまでを押し続けると電源が入り、下記のような画面が表示されます。

電源を切る場合には、２秒以上１を押し続けてください。

本体の電源を入れると、モデル名／バージョン画面を表示します。
正しく起動しない場合には、直ちに使用を中止し「12.1 トラブルシューティング」 89 頁を参照してください。

モデル名

バージョン
シリアル№

２

「測定値」の画面を表示します。

注意表示
前回測定時の電流クランプセンサと異なる電流クランプセンサを自動検出した場合には、今接続している
クランプセンサ種別とシリアル№および、位相補正値を 5 秒間画面に表示し、同時に自動で設定します。
現在の接続状態と異なる場合には、必ず「基本設定」から正しいクランプセンサの種類に変更してください。

注記● 汎用のリーク電流検出型クランプセンサおよび、負荷電流検出型クランプセンサは自動検出できません。接続しているクランプセンサを確認して、手動でKEW5050− 34 −「基本設定」から設定してください。

5.6 測定対象への接続

KEW5050

5.6 測定対象への接続

5.6

KEW5050

測定対象への接続
必ず確認してください

危険●本体の対地間最大定格電圧は、測定カテゴリに準じて CAT Ⅳで AC300V、CAT Ⅲでは AC600V です。これより高い対地間電圧のある回路では絶対に使用しないでください。
●電圧用測定コードとクランプセンサは本製品専用のものをご使用ください。
●Ior 用クランプセンサ(KEW8177/8178)の対地間最大定格電圧は、測定カテゴリに準じて CAT Ⅲ、AC300V です。
本体基準電圧入力端子の測定カテゴリ CAT Ⅳ300V、CAT Ⅲ600V とは測定できる場所が違います。測定電圧と定格が合っているか必ず確認してください。
●クランプセンサ、電圧用測定コード、電源コードは必ず測定物や電源よりも先に本体に接続してください。
●測定コードと本体の測定カテゴリが違っている場合は低い方の測定カテゴリが優先されます。測定電圧と定格が合っているか必ず確認してください。
●測定に必要のないクランプセンサは絶対に接続しないでください。
●クランプセンサは必ずブレーカーの２次側に接続してください。1 次側は電流容量が大きく危険です。
●通電中は CT の２次側が開放しないよう充分注意してください。万一開放状態になりますと、２次側に高電圧が発生して大変危険です。
●結線時に電圧用測定コードの先端の金属部で電源ラインを短絡しないように注意してください。また、先端の金属部には絶対に触れないでください。
●クランプセンサのコア先端部は被測定物を短絡しないような構造になっていますが、絶縁されていない導線を測定する場合はコアで被測定物を短絡しないように注意してください。
●測定の際は指先等が、保護用フィンガガードとバリアを超えることのないよう充分注意してください。
保護用フィンガガードとバリア：操作中の感電事故を防ぐため、最低限必要な沿面および空間距離を確保するための目印です。
●測定中（測定ラインからの通電中）は絶対に本体のコネクタから取りはずさないでください。
●コアを開いたとき、金属部で測定ラインの２線間を接触させないでください。

警告●感電、短絡事故をさけるため、接続をする場合は測定ラインの電源を切ってください。
●電圧用測定コードの先端の金属部には絶対にさわらないでください。

35− 35 −

KEW5050

KEW5050

KEW5050

5.6 測定対象への接続

KEW5050

5.6 測定対象への接続

結線方法（結線方式の選択：結線図）
で現
現在設定している結線図を表示します。

で結線方式の種類を変更し、で測定する系統数を変更します。
表示した結線図を確認しながら、クランプセンサと電圧用測定コードとを正しく結線してください。その後、で「基本設定」画面へ移動することで選択していた結線方式が有効になります。

では、設定を更新しませんので注意してください。

注記●電流だけを測定したい場合でも必
必ず電圧用測定コードを結線してください。測定値を演算するタイミングが特定できなくなるので、測定値が安定しません。
●異容量接続の三相４線式 V 結線、デルタ結線では、対地抵抗成分漏洩電流（Ior）と対地容量成分漏洩電流（Ioc）との位相が重なることがあります。そのため、Io から Ioc のみを除くことができないため、Ior を正確に測定することができません。また、主に病院等で用いられる非接地配線方式（フローティング電源）では漏洩電流自体が流れない
ために、測定できません。
正確に測定するために●測定ラインと本製品の結線方式の設定は正しく行ってください。
●クランプセンサは下記のように矢印を負荷側または、中性線側（接地線測定時）に向けてクランプしてください。
接地線測定時/ＡＡ（負荷電流）×１～４

１Ｐ２Ｗ(単相２線式)×１～４：Ｌ、Ｎ線を一括してクランプ

中性線側/負負荷側

負荷側

矢印：中性線側/負荷側
矢印：負荷側
電源側

B 種接地極側/電
電源側
１Ｐ３Ｗ(単相３線式)×１～４ ：Ｌ１、Ｌ２、Ｎを一括してクランプ
３Ｐ３Ｗ(三相３線式)×１～４ ：Ｒ、Ｓ、Ｔを一括してクランプ

３Ｐ４Ｗ(三相４線式)×１～４：Ｌ１、Ｌ２、Ｌ３、Ｎを一括してクランプ

負荷側

負荷側

矢印：負荷側

矢印：負荷側電源側

KEW5050

KEW5050

電源側

36− 36 −

5.6 測定対象への接続

KEW5050

結線図
設

定

内

容

負荷電流・電圧

単相２線式

単相３線式

三相３線式

ロガー（A）(1P2W)(1P3W)(3P3W)(3P4W)

×1～4 系統

×1～4 系統

×1～4 系統

×1～4 系統

×1～4 系統※

三相４線式

は初期値です。

結線図を確認しながら、クランプセンサと電圧用測定コードを正しく結線してください。
電圧用測定コードのワニグチクリップは被覆の上からは測定できません。

− 37 −

KEW5050

KEW5050

5.7 記録の手順

KEW5050

5.7

5.7 記録の手順

記録の手順
記録の開始
を押す。

記録を開始する前に現在の記録設定を表示します。変更する必要がない場合には「開始する」を選択してください
現在の設定に準じて記録を開始します。記録設定を変更するには、一旦「取り消す」を選択して記録をキャンセル
してから、「記録設定」から変更してください。また、測定前には必ず『安全の確認』、『測定準備』を行ってください。
「取り消す／開始する」の項目へ移動

決定

キャンセル

記録の終了
記録データナンバーを押す。

記録開始方法
記録イベント項目

記録に関する情報の表示と、記録を終了できます。
画面表示項目
記録データナンバー
経過時間

記録データの識別番号を表示します。測定データを保存するフォルダ名としても使用します。
記録開始からの経過時間を表示します。
手動記録開始方法

連続記録「記録開始日時」を表示します。
「記録開始日時」／「記録終了日時」を表示します。

時間帯指定 「記録開始日時」／「記録期間」／「記録時間帯」を表示します。
記録イベント項目

判定イベントとして設定している項目を表示します。

「取り消す／停止する」の項目へ移動

KEW5050

KEW5050

決定38− 38 −

キャンセル

6 章 設定

KEW5050

6章 設定
測定を始める前にあらかじめ測定条件やデータの保存について設定をする必要があります。
設定を行う場合にはを押して、先ず「結線図」から結線方式を選択してください。

で「詳細設定」モードへ移行します。再度

または、で前画面へ戻ります。

項目の選択（移動）
本製品は、す。

で選択(移動)し、で決定、で決定せずに元の設定に戻る操作を基本としていま「基本設定」－「電流クランプ」の入力操作を例に、詳細を説明します。他の表示画面での入力

操作も、ほぼ同様の操作方法です。
黒い背景に白抜き文字の項目が選択中の項目になります。「電流クランプ」では、で各 ch の電流クランプを選択し、戻りたい場合には、で決定します。選択に関係なく前画面にで戻ります。

選択項目が縦横に表示してある画面ではを選択する左の画面では、で項目の移動が可能です。「クランプ」

で接続したいクランプ種別を選択し、します。選択に関係なく前画面に戻るにはを押してください。

シリアル№(日付/時刻なども同じ)のように数値を入力するには桁へ移動し、
る左の画面では、で決定で変更したいで数値のカウントアップ／ダウンを行います。「シリアル№」を選択すでシリアル№の百の桁を選択しています。この状態でを操作すると、百の桁を１ずつカウントアップ、またはカウントダウンできます。変更を決定するにはを押してください。数値に関係なく前画面に戻るにはを押してください。

− 39 −

KEW5050

KEW5050

6.1 詳細設定項目

KEW5050

6.1

6.1

詳細設定項目「詳細設定」は、下記の５つの項目に分かれています。それぞれの項目へはで移動します。

注記
変更した設定はで、他の画面へ移動することで有効になります。画面を移動せずに電源を切ると、設定は更新されません。注意してください。

KEW5050

KEW505040− 40 −

6.2 基本設定

KEW5050

6.2 基本設定

6.2

KEW5050

基本設定「基本設定」タブへ移動

センサ識別
Ior 用クランプセンサのみ自動設定できます。
本体にＩｏｒ用クランプセンサを接続してから「センサ識別」を行ってください。クランプセンサの種類とシリアル№、位相補正量とを自動で設定します。前回測定時の電流クランプセンサと異なる電流クランプセンサを検出した場合には、検出したｃｈの先頭に「！
！」を表示します。

汎用のリーク電流検出型クランプセンサおよび、負荷電流検出型クランプセンサは自動設定できません。手動で設定してください。

「センサ識別」の項目へ移動

決定／

41

− 41 −

キャンセル

KEW5050

KEW5050

KEW5050

6.2 基本設定

KEW5050

6.2 基本設定

センサ識別が NG となる原因
確認事項

原因

電流クランプセンサの種類

・汎用の電流クランプセンサを接続していませんか？ＫＥＷ５０５０で自動で識別できるのはＩｏｒ用クランプセンサのみです。汎用のリーク電流検出型クランプセンサおよび、
負荷電流検出型クランプセンサを接続する場合には「電流クランプ」から手動で設定
してください。

？？？（識別不能）・電流クランプセンサが本体に確実に接続されていますか？
・故障かなと思ったら？
NG と識別されたｃｈに接続しているＩｏｒ用クランプセンサを、他の正しく識別できているｃｈに入れ換えて再度「センサ識別」を行ってください。この時、前回と同じｃｈが NGと識別された場合には、本体が故障している可能性があります。前回 NG と識別されていたＩｏｒ用クランプセンサを接続しているｃｈが、NG と識別された場合にはＩｏｒ用クランプセンサが故障している可能性があります。故障が疑われる場合は、直ちに使用を中止してください。

電流クランプ/ｃｈ
電流クランプセンサの詳細情報を設定します。

設定ｃｈの項目へ移動

KEW5050

KEW505042− 42 −

6.2 基本設定

KEW5050

クランプ：
使用するクランプセンサを選択します。クランプの一覧を表示している状態で、各クランプセンサへ移動すると、定格電流と被測定導体径を画面下へ表示します。
設

定

内

容

Ｉｏｒ用リーク電流検出型クランプセンサ
8177/8178：10.000mA／100.00mA／1000.0mA／10.000A／AUTO

汎用リーク電流検出型クランプセンサ
8146/8147/8148：10.000mA／100.00mA／1000.0mA／10.000A／AUTO

8141/8142/8143：5.000mA／50.00mA／500.0mA／1.000A／AUTO

負荷電流検出型クランプセンサ
8128：500.0mA／5.000A／50.00A／AUTO

8121/8127：1000mA／10.00A／100.0A／AUTO

8126：2.000A／20.00A／200.0A／AUTO

8122/8125：5.000A／50.00A／500.0A／AUTO

8123/8124/8130：10.00A／100.0A／1000A／AUTO

8133：30.00A／300.0A／3000A／AUTO

8129：300.0A／1000A／3000A※「クランプ：」の項目へ移動

決定／は初期値です。

クランプ一覧表示

クランプセンサを選択

キャンセル

レンジ：
使用する電流レンジを選択します。

注記●「ＡＵＴＯ」を設定すると、設定したｃｈのイベント検出が自動的に「ＯＦＦ」になります。イベントを検出する場合には、必ずレンジを固定レンジに設定してください。なお、イベント設定は「ＡＵＴＯ」を再度、固定レンジに設定しなおすことで、「ＡＵＴＯ」を設定する前の状態に戻ります。
「レンジ：」の項目へ移動
決定/

プルダウンメニュー表示

レンジを選択キャンセル− 43 −

KEW5050

KEW5050

6.2 基本設定

KEW5050

6.2 基本設定

シリアル№：
リーク電流検出型クランプセンサのシリアル№を登録して、測定時に使用したクランプセンサのシリアル№を、
アプリケーションから参照できます。
「Ｉｏｒ用クランプセンサ」
接続し「センサ識別」､「電源オン」、「記録開始」のいずれかを行うと自動的に登録します。または、表示した一覧から登録済みのシリアル№を選択することもできます。登録済みのシリアル№は削除できません。
「汎用リーク電流検出型クランプセンサ」「新規」からクランプセンサ本体に貼付しているナンバーを登録してください。登録後は一覧から登録済みのシリアル№を選択できます。また、登録後は下段にある「シリアル№削除」にて、現在選択しているシリアル№（現在選択している汎用リーク電流検出型クランプセンサ）を削除できます。

「負荷電流検出型クランプセンサ」
リーク電流検出型クランプセンサのみが本機能の対象です。入力項目を表示しません。
「シリアル№：」の項目へ移動

新規／シリアル№一覧表示

シリアル№を選択「新規」＋

シリアル№入力

設

定

内0001～9999※

KEW5050

KEW5050

決定／は初期値です。

44− 44 −

容キャンセル6.3 イベント設定

KEW5050

6.3 イベント設定

KEW5050

周波数
測定対象の公称周波数を設定します。停電時のように電圧の周波数が特定できない状態では、設定した公称周波数を基準にして測定を行います。
設

定

内

容

50Hz／60Hz※「公称周波数」の項目へ移動
決定／

6.3

は初期値です。

プルダウンメニュー表示

周波数を選択キャンセル

イベント設定「イベント」タブへ移動

共通
ヒステリシス：
イベント判定をしない測定領域を、しきい値に対する％で設定します。適切なヒステリシスを設定することで、しきい値付近の電圧変動／電流変動によって起こる必要以上のイベント判定を防ぐことができます。
設

定

内

容

しきい値に対して 1～10%（5%）※

「ヒステリシス」の項目へ移動

は初期値です。
数値入力ウインドウを表示※※

入力できる範囲を同時にポップアップに表示します。

ヒステリシス[%]を入力

決定／

45

− 45 −

キャンセル

KEW5050

KEW5050

KEW5050

6.3 イベント設定

上限しきい値(Ｈ)/ｃｈ
上限しきい値の詳細情報を設定します。

イベント項目へ移動

設

定

項

目

漏洩電流実効値 ：Iom
基準電圧実効値 ：Vm

上限しきい値

漏洩電流 ：IoＨ：

対地抵抗成分漏洩電流 ：Ior
基準電圧 ：V

トリガー：「ON／OFF」で、それぞれのイベント「しきい値」を入力できる状態にします。ON したイベント判定はすべてのｃｈに対して有効になります。特定のｃｈを判定したくない場合には、そのｃｈの電流レンジ設定を「ＡＵＴＯ」に設定するか、判定対象外になるように、しきい値を調整してください。

注記●クランプセンサの電流レンジを「AUTO」設定にしていると電流に関するイベント判定をONにできません。イベン
トしきい値を含んだ固定電流レンジに変更してから ON にしてください。

設

定

内

容

ＯＮ／ＯＦＦ※「トリガー：」へ移動
決定／

KEW5050は初期値です。

入力ウインドウを表示
キャンセル− 46 −

ON・OFF を選択6.3 イベント設定

KEW5050

6.3 イベント設定

KEW5050ｃｈ：
連続して 200ms ごとに求める実効値の上限しきい値を、レンジの最大値に対する％で設定します。このしきい値にはヒステリシスが有効になります。

注記●漏洩電流実効値の上限しきい値は電流レンジに対してパーセンテージで設定するため、電流レンジの設定を変更すると、しきい値の電流値が変わります。注意してください。基準電圧レンジは 1000V 固定です。
設

定

内

容

電圧レンジに対して 0～100%（100%）
電流レンジに対して 0～110%（100%）※は初期値です。

設定ｃｈの項目へ移動

数値入力ウインドウを表示※※

入力できる範囲と実効値とを同時にポップアップに表示します。

レンジの最大値に対する％を入力

上限イベントの検出例

電流・電圧波形

決定／

キャンセル

※

連続した 200ms(50/60Hz：10/12 周期)ごとの実効値

実効値①

実効値②

実効値③

～

～

～

しきい値

実効値ｎ

～

～

ヒステリシス
①

②

③

実効値波形

47

− 47 −

・・・ｎ

KEW5050

KEW5050

KEW5050

6.3 イベント設定

KEW5050

6.3 イベント設定

下限しきい値(Ｌ)
下限しきい値の詳細情報を設定します。

イベント項目へ移動

設

定

項

目

基準電圧実効値 ：Vm

下限しきい値

基準電圧 ：VＬ：

トリガー：「ON／OFF」で、それぞれのイベント「しきい値」を入力できる状態にします。
設

定

内

容

ＯＮ／ＯＦＦ※「トリガー：」へ移動
決定／

KEW5050

KEW5050は初期値です。

入力ウインドウを表示
キャンセル

48

− 48 −

ON・OFF を選択6.3 イベント設定

KEW5050

6.3 イベント設定

KEW5050

Ｖｍ：/Ｖ：
連続して 200ms ごとに求める基準電圧値の下限しきい値を、レンジの最大値（1000V）に対する％で設定します。このしきい値にはヒステリシスが有効になります。
設

定

内

容

レンジに対して 0～100%（5%）※

は初期値です。

数値入力ウインドウを表示※「しきい値」へ移動※入力できる範囲と実効値とを同時にポップアップに表示します。

レンジの最大値に対する％を入力

下限イベントの検出例

電圧波形

実効値波形

決定／

キャンセル

※

連続した 200ms(50/60Hz：10/12 周期)ごとの実効値

実効値①

実効値②

実効値③

～

～

～

①

②

実効値ｎ

～

③

～

・・・ｎ

ヒステリシス

しきい値

49− 49 −

KEW5050

KEW5050

KEW5050

6.3 イベント設定

ピークしきい値(Ｐｋ)/ｃｈ
ピークしきい値の詳細情報を設定します。

イベント項目へ移動

設

定

項

目

ピークしきい値

漏洩電流実効値 ：IomＰｋ：

基準電圧実効値 ：Vm

トリガー：「ON／OFF」で、それぞれのイベント「しきい値」を入力できる状態にします。ON したイベント判定はすべてのｃｈに対して有効になります。特定のｃｈを判定したくない場合には、そのｃｈの電流レンジ設定を「ＡＵＴＯ」に設定するか、判定対象外になるように、しきい値を調整してください。

注記●クランプセンサの電流レンジを「AUTO」設定にしていると 電流に関するイベント判定をONにできません。イベン
トしきい値を含んだ固定電流レンジに変更してから ON にしてください。
設

定

内

容

ＯＮ／ＯＦＦ※「トリガー：」へ移動
決定／

KEW5050は初期値です。

入力ウインドウを表示
キャンセル− 50 −

ON・OFF を選択6.3 イベント設定

KEW5050

6.3 イベント設定

KEW5050

Ｉｏｍ、ｃｈ： / Ｖｍ：
漏洩電流値と基準電圧値のしきい値を瞬間の電流ピーク値・電圧ピーク値で設定します。

注記
電流レンジの設定を変更すると、しきい値の電流値がレンジの 200%Apeak に変わります。注意してください。基準
電圧レンジは 1000V 固定です。
設

定

内

容

Iom：

電流レンジに対して 1.5%Apeak（1mA≦）～200%Apeak（200%Apeak）

Vm：

50V～2000Vpeak(2000Vpeak）※

は初期値です。

設定ｃｈの項目へ移動

数値入力ウインドウを表示※※

入力できる範囲を同時にポップアップに表示します。

電圧ピーク値を入力

決定／

キャンセル

数値入力ウインドウを表示※「しきい値」へ移動※入力できる範囲を同時にポップアップに表示します。

電圧ピーク値を入力

決定／51− 51 −

キャンセル

KEW5050

KEW5050

KEW5050

6.3 イベント設定

KEW5050

6.3 イベント設定「ピークしきい値」
漏洩電流実効値(Iom)・基準電圧実効値(Vm)
漏洩電流値と基準電圧値のしきい値を瞬間の電流ピーク値・電圧ピーク値で設定します。
設

定

内

容

Iom：

電流レンジに対して 1.5%Apeak（1mA≦）～200%Apeak（200%Apeak）

Vm：

50V～2000Vpeak(2000Vpeak）※

は初期値です。
数値入力ウインドウを表示※

イベントを検出する項目の「しきい値」へ移動※入力できる範囲を同時にポップアップに表示します。

電圧値を入力決定／

キャンセル

ピーク検出例
しきい値（電流値・電圧値）
電流・電圧波形

KEW5050

KEW5050

52− 52 −

6.4 記録設定

KEW5050

6.4 記録設定

6.4

KEW5050

記録設定「記録設定」タブへ移動

記録方法

インターバル
測定データをＳＤカードに記録する時間の間隔を選択します。設定可能なインターバルは 12 種類あります。
設

定

内

容

200 ミリ秒／400 ミリ秒／1 秒／5 秒／15 秒／30 秒／
1 分／5 分／15 分／30 分／1 時間／2 時間※「インターバル」の項目へ移動

決定／は初期値です。
選択ウインドウを表示

記録間隔を選択

キャンセル

開始方法
記録開始の方法を選択します。
設

定

内

容

手動／連続記録／時間帯指定※「開始方法」の項目へ移動
決定／は初期値です。

プルダウンメニュー表示

記録開始方法を選択

キャンセル

53

− 53 −

KEW5050

KEW5050

KEW5050

6.4 記録設定

KEW5050

6.4 記録設定「手動」
を操作して記録開始／記録終了を行った間のみ記録します。

「連続記録」
記録を開始する日付と時間、記録を終了する日付と時間を設定します。設定した開始日時から終了日時までの間、インターバルごとに記録を行います。
設

定

項

目

設

定

内

容

開始日時

年／月／日

時：分 （0000／00／00

00：00）

終了日時

年／月／日

時：分 （0000／00／00

00：00）「開始：」／「終了：」の項目へ移動

決定／

日時入力ウインドウを表示

キャンセル『例』 下図のように設定した場合には、次の期間記録を行います。
２０１７年５月３０日の６時１０分～２０１７年６月１０日の１０時２０分

KEW5050

KEW505054− 54 −

日時を入力

6.4 記録設定

KEW5050

6.4 記録設定

KEW5050「時間帯指定」記録する時間帯を開始時刻と終了時刻で設定し、記録する期間を開始日付と終了日付とで設定します。設定した時間帯の間のみインターバルごとに記録し、これを設定した期間繰り返します。
設

定

項

目

設

定

内

容

記録期間

開始－終了

年／月／日（YYYY／MM／DD）－年／月／日（YYYY／MM／DD）

記録時間帯

開始－終了

時：分（hh：mm）－時：分（hh：mm）「期間：」の項目へ移動

決定／

日付入力ウインドウを表示

キャンセル

日付を入力

「時間帯：」の項目へ移動

時間入力ウインドウを表示

時間を入力

決定／

キャンセル

『例』 下図のように設定した場合には、次の期間と時間帯で記録を行います。
午前６時１０分～８時３０分の間のみ記録し、その他の時間帯は記録開始待ちになります。
これを、２０１７年５月２０日～６月１０日までの期間繰り返します。

55− 55 −

KEW5050

KEW5050

KEW5050

6.4 記録設定

記録可能時間について注記●使用するＳＤカードは、必ず本体に付属した製品または、オプションのＳＤカードを使用してください。
●新しいＳＤカードを使用する場合には、必ずＫＥＷ５０５０本体でフォーマットを行ってください。ＰＣや他の機器にてフォーマットを行うと、正しく記録できないまたは、極端に記録可能な時間が短くなる可能性があります。詳細は「フォーマットする」58 頁を参照してください。
●イベントの記録を設定していた場合には、その発生量によって記録可能な残り時間が減少します。１回の記録開始／終了で保存できるイベントデータは最大 1GB です。
２GB のＳＤカードで、インターバルデータのみを記録した場合の計算値です。記録可能な時間を保証するものではありません。条件および測定環境によって記録できる時間は変わるため、参考値としてのみご使用ください。
インターバル４００ミリ秒は、瞬時値に加えて平均値、最大値、最小値を記録するので、瞬時値のみを記録するイ
ンターバル２００ミリ秒よりも、記録できる時間が短くなります。
インターバル

記録可能時間２００ミリ秒７日

４００ミリ秒

３日

１秒

９日

５秒

６．７ヶ月

１５秒

２０ヶ月

３０秒

４０ヶ月

１分

６．７年以上

５分

３３年以上

１５分

１００年以上３０分
１時間

２００年以上２時間

KEW5050− 56 −

6.5 保存データ

6.5

KEW5050

保存データ「保存データ」タブへ移動

記録データ

操作項目へ移動決定

空き容量を確認する
本体に挿入しているＳＤカードのデータ容量を表示します。

で「保存データ」の画面に戻ります。

項目
容量

表示内容
合計サイズ

使用領域+空き領域の容量

空き容量

空き領域のみの容量

記録可能時間

現在のインターバル設定で記録できる目安の時間− 57 −

KEW5050

KEW5050

6.5 保存データ

KEW5050

6.5 保存データ

データを削除する
ＳＤカードに記録している「S0001～9999：測定データ」、「PS-SD000～999：プリントスクリーン」、「SUPS0000～
9999：本体設定データ」を、一覧から選択して削除します。ファイルを表示する順番は不定のため、選択時にはファイル名の右側に表示している日付を参考にしてください。全体の記録ファイル数は画面右側に表示しているスクロールバーが目安になります。

チェックボックス
削除したいデータへ移動

データ選択

－

／ 選択解除「データ削除」確認メッセージ表示

決定／

削除したいデータを選択

スクロールバー－

はい／いいえを選択

で入力を終了して「保存データ」の画面に戻ります。

するとチェックボックスがに変わり、選択されていることを表示します。一度に複数のデータを削除したい場合には続けて選択してください。

フォーマットする
ＳＤカードをフォーマットします。フォーマットを行うとＳＤカードに記録しているすべてのデータを削除します。確認し十分に注意して行ってください。全体の記録ファイル数は画面右側に表示しているスクロールバーを目安にしてください。
スクロールバー「フォーマット」確認メッセージ表示

決定／

KEW5050

KEW5050

はい／いいえを選択

で入力を終了して「保存データ」の画面に戻ります。

58− 58 −

6.5 保存データ

KEW5050

6.5 保存データ

KEW5050

本体設定

操作項目へ移動決定

設定を保存する
ＳＤカードに「SUPS0000～9999：本体設定データ」を保存します。ファイルを表示する順番は不定です。全体の記録
ファイル数は画面右側に表示しているスクロールバーが目安になります。
スクロールバー「保存する」確認メッセージ表示

決定／

はい／いいえを選択

で入力を終了して「保存データ」の画面に戻ります。

59− 59 −

KEW5050

KEW5050

KEW5050

6.5 保存データ

以下の本体設定を保存します
基 本 設 定設

そ の 他 設 定
定

項

目

設

結線

定

項

環境設定

クランプ／シリアル№／電流レンジ

日付形式
ID 番号

本体設定

周波数

目

ブザー音

イ ベ ン ト 設 定
設

定

項

目

ヒステリシス：

上限しきい値Ｈ：

漏洩電流実効値：Iom

ON／OFF

１ｃｈ～４ｃｈしきい値

基準電圧実効値：Vm

ON／OFF

しきい値

漏洩電流：Io

ON／OFF

１ｃｈ～４ｃｈしきい値

対地抵抗成分漏洩電流：Ior

ON／OFF

１ｃｈ～４ｃｈしきい値

基準電圧：V

ON／OFF

しきい値

下限しきい値

基準電圧実効値：Vm

ON／OFF

しきい値

基準電圧：V

ON／OFF

しきい値

Ｌ：
ピークしきい値

漏洩電流実効値：Iom

ON／OFF

１ｃｈ～４ｃｈしきい値

Ｐｋ：

基準電圧実効値：Vm

ON／OFF

しきい値

記 録 設 定
設
記録方法

定

項

インターバル
開始方法
開始日時

連続記録
時間帯指定

KEW5050

目

終了日時
記録期間

開始－終了

記録時間帯

開始－終了

− 60 −

6.5 保存データ

KEW5050

6.5 保存データ

KEW5050「設定を読み込む」
ＳＤカードから「SUPS0000～9999：本体設定データ」を読み込んで設定を変更します。ファイルを表示する順番は不定のため、選択時にはファイル名の右側に表示している日付を参考にしてください。全体の記録ファイル数は
画面右側に表示しているスクロールバーが目安になります。

スクロールバー

転送したいデータへ移動

決定／「読み込む」確認メッセージ表示

はい／いいえを選択

で入力を終了して「保存データ」の画面に戻ります。

保存データの種類について
データの取り扱いに関してファイル名は自動的にファイル№をカウントアップしてつけられます。カウントアップしたファイル№は本体で記録しているため電源をオフしても初期値になりません。システムリセットを行うか、最大のカウント数を越えるまでカウントアップし続けます。

注記●保存するファイル№のファイル名と同じ名前のファイルをＳＤカードに保存していると、測定データは自動的に次のファイル№にカウントアップしてフォルダを作成しデータを保存しますが、「プリントスクリーン」と「本体設定」のファイルは同じファイル名で上書きします。システムリセットを行ってファイル№がゼロから始まる場合と、複数の異なるＫＥＷ５０５０で１枚のＳＤカードを転用する場合には、ファイル名が重複しないように注意してください。ただし、すべてのファイル No.(S0000～S9999) の全種類をＳＤカードに保存していると測定データも上書きします。
注意してください。
●ＰＣ：コンピュータ等でファイルを削除したり保存したフォルダまたは、ファイルの名前を変更すると、本体でのデータ操作と解析用ＰＣソフトウェアでの解析ができなくなります。フォルダ名および、ファイル名は絶対に変更しな
いでください。

61− 61 −

KEW5050

KEW5050

KEW5050

6.5 保存データ

KEW5050

6.5 保存データ「プリントスクリーン」のＢＭＰデータを記録します。
ファイル名： ＰＳ－ＳＤ

０００

．ＢＭＰ

｜

｜

ファイル№

拡張子

（000～999）（ＢＭＰファイル）「本体設定データ」の「保存データ」-本体設定から本体設定データを記録します。
ファイル名：ＳＵＰＳ

００００

．ＫＥＷ

｜
ファイル№
（0000～9999）「データフォルダ」
１回の測定ごとに「測定データ」を記録するためのフォルダを順次作成します。
フォルダ名：／ＫＥＷ／

Ｓ００００
｜
ファイル№
（0000～9999）「測定データ」
本体設定データ（記録開始時） ファイル名：

ＩＮＩＳ

インターバルデータ

ＤＡＴＳ

イベントデータ

ＥＶＴＳ

００００

｜
ファイル№
（0000～9999）

KEW5050

KEW505062− 62 −．ＫＥＷ

6.6 その他

KEW5050

6.6 その他

6.6

KEW5050

その他「その他」タブへ移動

環境設定

言語：
表示言語を選択します。
設

定

内

容

日本語／英語※は初期値です。ただしシステムリセットを行っても変更した設定は初期化されません。

「言語」の項目へ移動
決定／

プルダウンメニュー表示

言語を選択

キャンセル

63

− 63 −

KEW5050

KEW5050

KEW5050

6.6 その他

KEW5050

6.6 その他

日付形式：
日付の表示形式を選択します。画面の右上に表示している現在の日付、記録開始終了に関する日付表示と設定入力形式などがすべて変更できます。
設

定

内

容

YYYY/MM/DD ／ MM/DD/YYYY ／ DD/MM/YYYY※

は初期値です。ただしシステムリセットを行っても変更した設定は初期化されません。

「日付形式」の項目へ移動
決定／

プルダウンメニュー表示

形式を選択

キャンセル

本体システムに関する設定

現在日時：
システムクロックの時間を設定します。現在の日時に合わせてください。
設

定内yyyy/mm/dd

容

hh：mm

※

「日付形式」に同期して入力形式も変わります。

「現在日時」の項目へ移動

決定／

KEW5050

KEW5050

日時入力ウインドウ表示

キャンセル

64

− 64 −

日時を入力

6.6 その他

KEW5050

6.6 その他

KEW5050

ＩＤ番号：
本体の認識番号を設定します。複数台使用する場合や１台で複数の場所を定期的に測定する場合に番号を使い分けると、記録データを解析する時に便利です。
設

定

内

容

00-001～99-999※「ＩＤ番号」の項目へ移動

決定／は初期値です。

数値入力ウインドウ表示

認識番号を入力

キャンセル

ブザー音：
キーを操作した時のブザー音のあり、なしを選択します。乾電池の警告ブザーには関係しません。
設

定

内

容

あり／なし
※

「ブザー音」の項目へ移動
決定／は初期値です。

プルダウンメニュー表示

ブザー音あり／なしを選択

キャンセル

電源：
記録中を除き一定の時間以上、無操作状態が続いた場合に、自動で電源をＯＦＦするか、ＯＦＦしないかを選択します。乾電池で駆動している場合は、乾電池の容量を節約するために「ＯＦＦしない」を選択できません。
設

定

対

象

設

AC アダプタ定容

５分後に OFF ／OFF しない

乾電池

２分後に OFF（固定）※「電源」の項目へ移動

内は初期値です。

プルダウンメニュー表示

自動で電源をＯＦＦ／ＯＦＦしないを選択

決定／

65

− 65 −

キャンセル

KEW5050

KEW5050

KEW5050

6.6 その他

KEW5050

6.6 その他

バックライト：
一定の時間以上、無操作状態が続いた場合に、自動でバックライトをＯＦＦするか、ＯＦＦしないかを選択します。
乾電池で駆動している場合は、乾電池の容量を節約するために「ＯＦＦしない」を選択できません。
設

定

対

象

設

AC アダプタ定内容

５分後に OFF ／OFF しない

乾電池

２分後に OFF（固定）※「バックライト」の項目へ移動

は初期値です。

プルダウンメニュー表示

自動でバックライトをＯＦＦ／ＯＦＦしないを選択

決定／

キャンセル

システムリセット
環境設定「言語」､「日付形式」および､「現在日時」を除いた設定を出荷時の状態に初期化します。
「システムリセット」の項目へ移動
決定／

KEW5050

KEW5050

確認メッセージ表示

キャンセル

66

− 66 −

はい／いいえを選択7.1 測定値

KEW5050

7章 画面ごとの表示項目
7.1

測定値
系統ごとの測定値を一覧で表示する
系統ごとの測定値を１画面に表示します。系統数が複数の場合には「被測定系全体」の画面をはじめに表示します。

一覧表示（例）３Ｐ３Ｗ（三相３線式）１系統を表示しています。

※

3P3W の結線では Io が R 相電圧と T 相電圧の位相間にある場合は Io よりも Ior が大きくなります。

※

記号の後につく数字が系統の番号です。
画面表示記号

Io

50/60Hz 基本波のみの漏洩電流実効値

Iom

高調波成分を含む漏洩電流実効値

Ior

対地抵抗成分のみの漏洩電流実効値

絶縁抵抗値「 V:基準電圧実効値(基本波)÷Ior:漏洩電流実効値(対地抵抗成分) 」
R

注記 絶縁抵抗計とは測定方法が異なるため、同じ抵抗値にはなりません。参考値としてご使用ください。

V

50/60Hz 基本波のみの基準電圧実効値ｆ

基準電圧の周波数

表示する系統を変更する
複数の系統を同時に測定している場合には、を操作することで表示する系統を変更できます。

系統ごとのベクトル図を表示する「ベクトル表示」に移動します。

− 67 −

KEW5050

KEW5050

7.1 測定値

系統ごとのベクトル図を表示する
系統ごとのベクトル図を１画面に表示します。

ベクトル表示（例）３Ｐ３Ｗ（三相３線式）を表示しています。

実線 V の矢印（）

実線Ｉｏの矢印（）

漏洩電流実効値(基本波)のベクトル線

点線の補助線（）

基準電圧基本波を除く、その他の電圧

円弧θ （

）

基準電圧基本波のベクトル線

位相角※

記号の後につく数字が系統の番号です。

画面表示記号
Iomθ

高調波成分を含む漏洩電流実効値※

ベクトル線は Io の位相角です。

位相角

Vm

高調波成分を含む基準電圧実効値

0～＋180 進み

基準電圧基本波の位相角を 0°として、漏洩電流基本波の位相0～－180 遅れ

角を表示します。

対地容量成分がなく、電圧と電流が平衡な状態では下図のようなベクトル図を表示します。Ｉｏ：漏洩電流実効値(基本波)の位相が下記の範囲内にない場合には、クランプセンサの取り付け向きか、ワニグチクリップの取り付け
極性を間違えている可能性があります。
※

基準電圧(V)を０°として表示します。

KEW5050− 68 −

7.1 測定値

KEW5050

表示する系統を変更する
複数の系統を同時に測定している場合には、を操作することで表示する系統を変更できます。

系統ごとの測定値を表示する「測定値」に移動します。

全体の測定値を一覧で表示する
被測定系統全体の測定値を１画面に表示します。測定対象が１系統のみの場合には本画面を表示しません。系統
1 の表示画面のみとなります。記号のみの表示は全系統の合計値になります。記号の後につく数字が系統の番号
です。

一覧表示
（例）３Ｐ３Ｗ（三相３線式）４系統を表示しています。

※

3P3W の結線では Io が R 相電圧と T 相電圧の位相間にある場合は Io よりも Ior が大きくなります。

※

記号の後につく数字が系統の番号です。
画面表示記号

Io

50/60Hz 基本波のみの漏洩電流実効値

Iom

高調波成分を含む漏洩電流実効値

Ior

対地抵抗成分のみの漏洩電流実効値

絶縁抵抗値「 V:基準電圧実効値(基本波)÷Ior:漏洩電流実効値(対地抵抗成分) 」
R

注記 絶縁抵抗計とは測定方法が異なるため、同じ抵抗値にはなりません。参考値としてご使用ください。

V

50/60Hz 基本波のみの基準電圧実効値ｆ− 69 −

基準電圧の周波数

KEW5050

KEW5050

7.1 測定値

KEW5050

7.1 測定値

系統ごとの測定値を一覧で表示するを操作することで系統ごとの測定値を表示します。

測定項目ごとに一覧で表示するにて以下のように表示を切り替えます。
「被測定系全体」：

各測定項目の合計値を表示します。

「漏洩電流」：

測定しているすべての系統のＩｏを一覧で表示します。

「対地抵抗成分漏洩電流」：

測定しているすべての系統のＩｏｒを一覧で表示します。

「漏洩電流実効値」：

測定しているすべての系統のＩｏｍを一覧で表示します。

「絶縁抵抗」に移動します。 ：KEW5050

KEW5050

測定しているすべての系統のＲを一覧で表示します。

70− 70 −

7.2 イベント

7.2

KEW5050

イベント
を押す
再度

を押すことで前画面へ戻ります。

イベントの発生状況を表示する
ｃｈ (系統) の切り替え

イベント種別の切り替え

ｃｈ (系統) ごとの発生回数
イベント発生ｃｈ (系統)

イベント発生日付
発生日時と測定値との切り替え
イベント種別 (＋アイコン)
スクロールバーｃｈ（系統）番号

イベント発生時間
時：分：秒．ミリ秒

イベント発生時の測定値− 71 −

KEW5050

KEW5050

7.2 イベント

KEW5050

7.2 イベント

画面表示記号
開始

終了

漏洩電流実効値

Ｈ：Ｉｏｍ

Ｈ：Ｉｏｍ

基準電圧実効値

Ｈ：Ｖｍ

Ｈ：Ｖｍ

漏洩電流

Ｈ：Ｉｏ

Ｈ：Ｉｏ

対地抵抗成分漏洩電流

Ｈ：Ｉｏｒ

Ｈ：Ｉｏｒ

基準電圧

Ｈ：Ｖ

Ｈ：Ｖ

下限
Ｌ：

基準電圧実効値

Ｌ：Ｖｍ

Ｌ：Ｖｍ

Ｌ：Ｖ

Ｌ：Ｖ

ピーク
Ｐｋ：

漏洩電流実効値

Ｐｋ：Ｉｏｍ

Ｐｋ：Ｉｏｍ

基準電圧実効値

Ｐｋ：Ｖｍ

Ｐｋ：Ｖｍ

上限
Ｈ：
イベント記号

基準電圧

イベント開始と終了の回数を表示します。カウントアップは発生した１イベントの開始で１カ
ウント、終了で１カウント、合計２カウントアップします。

回数ｃｈ（系統）番号

イベントを判定したｃｈ(系統１～４）を番号で表示します。

発生日付

イベントの開始と終了を判定した日付を表示します。

発生時間

イベントの開始と終了を判定した時間を表示します。
イベントの開始と終了を判定した時点の瞬時値を表示します。長期間継続しているイベン

測定値

ト中の測定値はインターバル測定データから確認できます。イベントを記録する場合には、できるだけ短いインターバル（最短 200ms）の設定が解析には有効です。

表示範囲を変更するを操作することで画面を縦にスクロールして表示する範囲を変更します。

イベント抽出

◀

▶

選択項目

(点滅)

◀▶

点滅内※を順番に[選択項目] 確定

◀
◀
◀
◀
◀
◀
◀
◀
◀
◀
◀
KEW5050

KEW5050

回数
すべて
Ｈ：Ｉｏｍ
Ｈ：Ｖｍ
Ｈ：Ｉｏ
Ｈ：Ｉｏｒ
Ｈ：Ｖ
Ｌ：Ｖｍ
Ｌ：Ｖ
Ｐｋ：ＩｏｍＰｋ：Ｖｍ

で変更または移動
※

解除

▶ ◀◀
▶
▶
▶
▶
▶ ◀
▶
▶
▶
▶
▶

ＣＨ
ＥＶＴ．

全ｃｈ

決定

発生していないイベントは除きます。

▶
▶
◀
◀
◀
◀
◀
◀

▶

72

− 72 −

Ｖ
Ａ
Ａ１
Ａ２
Ａ３
Ａ４

▶
▶ ◀
▶
▶ ◀
▶
▶

測定値
発生日時

▶
▶

7.2 イベント

KEW5050

7.2 イベント

KEW5050

上限／下限イベントの測定方法

ギャップなしで測定した約 200ms ごとの実効値から各イベントを検出します。イベントを検出した場合、その 200msの先頭をイベント開始時刻とし、その後の 200ms ごとの実効値にてイベントが検出されなかった場合に、その200ms の先頭をイベント終了時刻とします。なお、開始から終了までの間はイベントが継続していると見なします。

上限イベントの検出例※

連続した 200ms(50/60Hz：10/12 周期)ごとの実効値
実効値①

実効値②

実効値③

～

～

～

電流・電圧波形

しきい値

実効値ｎ

～

～

ヒステリシス
①

③

②

・・・ｎ

実効値波形
イベント開始

イベント終了
発生期間

下限イベントの検出例※

連続した 200ms(50/60Hz：10/12 周期)ごとの実効値

電圧波形

実効値①

実効値②

実効値③

～

～

～

①

実効値ｎ

～

・・・ｎ

③

②

～

ヒステリシス

しきい値
実効値波形
イベント終了

イベント開始発生期間

73− 73 −

KEW5050

KEW5050

KEW5050

7.2 イベント

KEW5050

7.2 イベント

ピークイベントの判定方法

漏洩電流波形と基準電圧波形をギャップなしで約４０ｋｓｐｓでモニタし、約２００ｍｓ区間ごとにピーク値を集計しながら判定します。そのため、初めてピークイベントを検出した約２００ｍｓ区間の最初がイベント開始時刻となり、その後の２００ｍｓ区間で、ピークイベントが検出されなかった場合に、その区間の最初をイベント終了時刻とします。な
お、開始から終了までの間はイベントが継続していると見なします。
ピークイベント検出例

しきい値

測定波形

イベント開始保存データ

約２００ｍｓ間

イベント終了

発生期間

イベントが発生した場合には、そのイベントの種別、ｃｈ（系統）、開始時間、終了時間、測定値を記録します。

KEW5050

KEW505074− 74 −

8 章 その他の機能

KEW5050

8 章 その他の機能

KEW5050

8章 その他の機能
データホールド機能
を押すと測定状態に関わらず表示更新をストップして、画面上にを押すとのマークを表示します。再びのﾏｰｸは消え、表示更新をスタートします。データホールド中でも表示を切り替えて、各画面の測定値を確認できます。記録中に表示更新をストップしても、測定値とイベントの情報を常に記録しています。

キーロック機能を２秒以上押すと、画面上にを２秒以上押し続けるとを表示し、その時点からすべてのキー入力をキャンセルします。再びが消え、キーロックを解除します。

注記● 「キーロック機能」を設定するとし、を２秒以上押して、を押しても電源が切れません。画面にを表示していないか確認

を消してから電源をオフしてください。

自動バックライト OFF
AC アダプタ接続時
何も操作しない状態が 5 分間経過すると自動的にバックライトを消灯します。再点灯する場合には、電源キーを除くキーを操作することで、再度 5 分間点灯します。

「その他」［本体設定］「バックライト：」から「OFF しない」を選択することで、バックライトを常に点灯することも可能です。

電池駆動時何も操作しない状態が 2 分間経過すると､自動的にバックライトを消灯します。再点灯する場合には、電源キーを除くキーを操作することで再度 2 分間点灯します。電池駆動では常時点灯できません。

自動電源 OFFAC アダプタ接続時
記録中を除き、何も操作しない状態が 5 分間経過すると自動的に電源を OFF します。再度電源を ON したい場合には電源キーを操作してください。

「その他」［本体設定］「電源：」から「OFF しない」を選択することで、常に電源 ON することも可能です。

電池駆動時
記録中を除き、何も操作しない状態が 2 分間経過すると自動的に電源を OFF します。再度電源を ON したい場合には電源キーを操作してください。電池駆動の場合は常時電源 ON し続けることはできません。

75− 75 −

KEW5050

KEW5050

KEW5050

8 章 その他の機能

KEW5050

8 章 その他の機能

電流オートレンジ
測定した電流実効値から各センサの電流レンジを自動で切り替えます。イベントを記録する場合には選択できません。切り替えは下側レンジの 300%peak または 130%rms を上回るとレンジアップ、下側レンジの 100%rms を下回るとレンジダウンです。

センサ識別機能「基本設定」［電流クランプ］からセンサ識別を操作することにより、現在本体に接続しているクランプセン
サを自動で識別します。
電源起動時には、その時点で接続しているクランプセンサを自動で識別し、前回電源をオフした時のクランプ設定と差がある場合にのみ識別結果を表示します。

プリントスクリーン機能を操作することで、現在表示している画面を BMP ファイルで保存します。1 ファイルのサイズは約 77KB です。

注意
保存されたBMPファイルは、カードリーダを使用してSDカードから直接読み込んでください。
KEW Windows for KEW 5050からの読み込みはできません。

設定記録機能電源を OFF しても前回測定時のすべての設定を記録しているため、次に電源を ON した時には前回測定時と同じ状態で起動します。※出荷後初回の起動時は初期設定値です。

ステータス LED
記録スタンバイ中は緑色で点滅し、記録中は緑色で常に点灯します。

KEW5050

KEW505076− 76 −

9.1 ＰＣ：コンピュータへのデータ転送

KEW5050

9章 周辺機器との接続
9.1

ＰＣ：コンピュータへのデータ転送
ＳＤカードに保存したデータは、USB 接続又はＳＤカードリーダーを使用することにより PC に転送することが可能です。ＵＳＢマスストレージに対応しています。ＰＣに接続すると、ＫＥＷ５０５０本体をリムーバブルディスクとして認識します。

ＵＳＢ接続によるデータ転送に関する注記● 記録中は測定データを保護するために、ＰＣではＳＤカードが抜かれた状態で認識されます。
● すべての機器との動作を保証するものではありません。USB ハブなどを介して接続すると、正しく動作しないことがあります。
● ＵＳＢマスストレージのみを使用する場合でも付属の USB ドライバをインストールしてください。
※

ＰＣへのデータ移動にはＳＤカードの使用をおすすめします。（本製品の転送時間 約 320MB／時）保存容量の大きいデータを直接 USB 接続で PC に転送すると、ＳＤカードを使用した場合に比べて転送する時間が長くなります。
また、確実にデータを保存するために、本製品のファイル以外はＳＤカードに記録しないでください。必要のないファ
イルは事前にすべて削除してください。

USB 接続SD カード− 77 −

KEW5050

KEW5050

9.2

9.2 外部機器との信号制御

外部機器との信号制御
出力端子を接続する危険●アースコードで本体を接地している場合、デジタル出力端子 L はアースコードを介して接地されています。
そのため、デジタル出力端子Ｌへ接地と異なる電圧を絶対に接続しないでください。本器を破損したり感電等の重大な事故を引き起こす可能性があります。デジタル出力端子Ｈに接続できる対地間最大定格電圧は30V、50mA、200mW以下です。
接続可能な信号線の寸法は下記のとおりです。
適合電線：単線φ1.2 ㎜(AWG16)、撚線 1.25 ㎜２(AWG16)素線径φ0.18 ㎜以上使用可能電線 ：単線φ0.4～1.2(AWG26～16)、撚線 0.2～1.25 ㎜２(AWG24～16)素線径φ0.18 ㎜以上
標準むき線長 ：11 ㎜
１コネクタカバーを開きます。

２

端子の下にある四角の部分をマイナスドライバーなどで押しながら、信号線を挿入します。

３ドライバーを離すと信号線が固定されます。
デジタル出力端子
出力方式：オープンコレクタ出力
端子 H ：本体内部プルアップ抵抗 10ｋΩ、5V
対地間最大定格 30V、50mA、200mW まで接続できます。
端子 L ：アースコードを介して接地しています。

信号線

マイナスドライバーなど「デジタル出力端子」+5V

端子 H、L の基本回路は左図のようにオープンコレクタ出力方式です。端子 L はアースコードを介して接地されています。端子 H はプルアップ電圧を持たない外部接
続機器に備え、10ｋΩのプルアップ抵抗を介して本体内部で 5V にプルアップしていHます。
通常 端子 H は 5V です。イベントが発生している間、端子 H は端子 L に接続され、

OUTPUT

端子間電圧が 0V になります。イベントの継続している時間が１秒未満であった場合には 1 秒間、端子間電圧を 0V にします。ただし、イベントが複数重なった場合にはL

各イベントの論理和で 0V になります。出力するイベントを限定したい場合には、「6.3 イベント設定」45 頁を参考にして、その他のイベントをすべて「ＯＦＦ」に設定してください。

KEW5050− 78 −

9.3 測定ラインから電源を供給する

KEW5050

9.3 測定ラインから電源を供給する

9.3

KEW5050

測定ラインから電源を供給する
コンセントから AC アダプタへ電源供給できない場合、ＭＯＤＥＬ８３２９（電源供給アダプタ)を用いることで、電圧用測定コードから電源供給を行うことができます。

危険●電源供給アダプタ、測定コード、本体は、それぞれ違った測定カテゴリです。電源供給アダプタが最も低い測定カテゴリのため、対地間最大定格電圧 CAT Ⅲで AC150V、CAT Ⅱで AC240V より、高い対地間電圧のある回路では絶対に使用しないでください。
●電源供給アダプタの定格電源周波数は 50Hz/60Hz です。
●本体に接続していない状態で測定ラインに接続しないでください。
●測定中（測定ラインからの通電中）は絶対に本体のコネクタから取りはずさないでください。
●必ずブレーカーの二次側に接続してください。1 次側は電流容量が大きく危険です。

警告●使用しているうちに亀裂が生じたり、金属部分が露出したときは、直ちに使用を中止してください。
●本体の電源が OFF になっていることを確認してから接続してください。
●接続は必ず先に本体側から行い、奥まで確実に差し込んでください。
以下の手順で本製品を接続します。

注意●安全に使用するため、以下の手順通り接続してください。
●手順通り接続しないとヒューズが溶断することがあります。
７

１
２
３

ＫＥＷ５０５０本体とＭＯＤＥＬ８３２９の電源スイッチがＯＦＦになっていることを確認します。
電圧用測定コードをＭＯＤＥＬ８３２９の電源入力端子(N/L)に差し込みます。
次にＭＯＤＥＬ８３２９のプラグをＫＥＷ５０５０の基準電圧入力端子に差し込みます。
ＭＯＤＥＬ８３２９の電源プラグをＡＣアダプタの奥まで確実に差し込みます。

４

AC アダプタの出力端子をアースコードの雌口コネクタに接続します。

５

アースコードの接続端子を本体の AC アダプタ接続コネクタに奥まで確実に差し込
みます。

６

２４０Ｖ
以下

アースコードのクリップを既知の接地端子に接続します。

危険：接続する前に必ず接地端子であることを確認し、絶対４に活線へ接続しないでください。
７

電圧用測定コードのワニグチを被測定回路に接続します。
電圧用測定コードのワニグチクリップは被覆の上からは測定できません。

８

ＫＥＷ５０５０の電源スイッチをＯＮにします。

９

ＭＯＤＥＬ８３２９の電源スイッチをＯＮにします。

２

１ / ９

２

本製品をはずす場合は、逆に ９ から １ へ向かっての手順になります。使用方法の詳細はＭＯＤＥＬ８３２９の取扱説明書
を参照ください。

３「ＭＯＤＥＬ８３２９」１ / ８

測定カテゴリ ：CATⅢ150V CATⅡ240V （50/60Hz）

６

ヒューズ定格 ：AC500mA／600V 速断タイプφ6.3×32mm
79− 79 −

５

KEW5050

KEW5050

KEW5050

10 章 設定・解析用ＰＣソフトウェア

10章 設定・解析用ＰＣソフトウェア
ＰＣから「ＫＥＷ Ｗｉｎｄｏｗｓ ｆｏｒ ＫＥＷ５０５０」を使用することで、本体で記録したデータの解析※と、本体の設定を行うことができます。
※

記録データからグラフとリストを 1 クリック自動生成、複数台の設定データ、記録データを一元管理など。

「ＫＥＷ Ｗｉｎｄｏｗｓ ｆｏｒ ＫＥＷ５０５０」を使用するには、別紙の「ＫＥＷ Ｗｉｎｄｏｗｓ ｆｏｒ ＫＥＷ５０５０」用のインス
トールマニュアルを参考にして、アプリケーションとＵＳＢドライバをＰＣ：コンピュータへインストールしてください。
●インターフェース
通信方式：USB Ver2.0 準拠

USB 通信で以下のことが行えます。
・SD メモリ内のファイルをパソコンへダウンロードする。
・パソコンから本体を設定する。
・本体で記録したデータの測定値とグラフをパソコン上に表示する。
●パソコンの推奨動作環境
・OS（オペレーションシステム）
対応 OS は CD ケースのバージョンラベルまたは、弊社ホームページを確認してください。
・画面表示
解像度 1024×768 ドット、65536 色以上
・HDD（ハードディスク）空き容量 1Gbyte 以上(Framework を含む)
・.NET Framework 3.5
・.NET Framework 4.6●商標について・Windows®は米国マイクロソフト社の商標です。
最新のソフトのダウンロードは、弊社ホームページから行うことができます。
www.kew-ltd.co.jp

KEW5050− 80 −

11.1 安全要求仕様

KEW5050

11.1 安全要求仕様

KEW5050

11章 仕様11.1

安全要求仕様使用環境： 屋内使用高度 2000m 以下

確度保証温湿度範囲： 23℃±5℃

相対湿度 85%以下（結露しないこと）

使用温湿度範囲：-10℃～50℃

相対湿度 85%以下（結露しないこと）

保存温湿度範囲：-20℃～60℃

相対湿度 85%以下（結露しないこと）

耐電圧：

AC5160V／5 秒間
AC3310V／5 秒間(基準電圧入力端子)－(外装)間(基準電圧入力端子)－(電流入力端子、AC アダプタ接続コネクタ、通信(USB)コネクタ)間

絶縁抵抗：50MΩ 以上／1000V

適合規格：IEC 61010-1、-2-030(電圧／電流入力端子、AC アダプタ接続コネクタ)－(外装)間測定カテゴリ 「本体」

CAT Ⅳ300V CAT Ⅲ600V 汚染度 2「電圧用測定コード」
EMC規格
防塵/防水性：IEC 60529 IP40

RoHS

：IEC 63000

11.2

CAT Ⅳ600V CAT Ⅲ1kV 汚染度 2: IEC 61010-031、IEC61326 Class A

一般仕様

表示部

：160×160 ドット FSTN モノクログラフィック液晶

表示更新周期：500 ミリ秒※※

測定演算処理の関係から、画面上に実際の測定値が反映されるまでに、最大で 400ms の遅延があります。ただし記録データとタイムスタンプに遅延はありません。
バックライト

消灯：操作しない状態が乾電池駆動時２分、ＡＣアダプタ接続時で５分以上経過すると自動的に消灯します。
点灯：消灯状態から電源ｷｰを除くｷｰを操作すると点灯します。

外形寸法：165（Ｌ）×115（Ｗ）×57（Ｄ）mm

質量：約 680g（電池含む）

実時間確度：±5 秒／日以内

電源：AC アダプタ：MODEL8262

電圧範囲

AC100V～AC240V

周波数50Hz/60Hz

消費電力

7.5VAmax(使用可能範囲：47Hz～63Hz)：DC 電源
乾電池

充電式電池

電圧

DC4.5V (＝1.5V×3 直列×2 並列)

DC3.6V (＝1.2V×3 直列×2 並列)

使用電池

単３形ｱﾙｶﾘ電池(LR6)単3形ﾆｯｹﾙ水素電池(1900mA/h)

消費電流0.21A typ.(@4.5V)

0.26A typ.(@3.6V)

連続使用時間 ※23℃参考値

11 時間

9 時間 ※ﾌﾙ充電時

81− 81 −

KEW5050

KEW5050

KEW5050
付属品

11.2 一般仕様：電圧用測定ｺｰﾄﾞ MODEL7273(（CAT Ⅲ1kV、CAT Ⅳ600V、赤／黒､ﾜﾆｸﾞﾁｸﾘｯﾌﾟ付)··········· 1 ｾｯﾄ
識別ﾏｰｶｰ·························································································· 各 2 本×4 色（赤/青/黄/緑) 合計８本
AC ｱﾀﾞﾌﾟﾀ MODEL8262······································································································································ 1 個
ｱｰｽｺｰﾄﾞ MODEL7278 ········································································································································ 1 本
USB ｹｰﾌﾞﾙ MODEL7219··································································································································· 1 本
取扱説明書 ······························································································································································ 1 冊
ｲﾝｽﾄｰﾙﾏﾆｭｱﾙ······················································································································································· 1 冊
CD-ROM···································································································································································· 1 枚
設定・解析用 PC ｿﾌﾄｳｴｱ（KEW Windows for KEW5050）
取扱説明書ﾃﾞｰﾀ（PDF ﾌｧｲﾙ）
単３形ｱﾙｶﾘ乾電池（LR6） ·································································································································· 6 個
ＳＤｶｰﾄﾞ (2GB)························································································································································ 1 枚
ｷｬﾘﾝｸﾞﾊﾞｯｸﾞ MODEL9125 ································································································································· 1 個

オプション：クランプセンサ
KEW8177(

Ior ﾘｰｸｾﾝｻ

10A ﾀｲﾌﾟ

φ40mm)

KEW8178(

Ior ﾘｰｸｾﾝｻ

10A ﾀｲﾌﾟ

φ68mm)

KEW8146(

ﾘｰｸｾﾝｻ

10A ﾀｲﾌﾟ

φ24mm)

KEW8147(

ﾘｰｸｾﾝｻ

10A ﾀｲﾌﾟ

φ40mm)

KEW8148(

ﾘｰｸｾﾝｻ

10A ﾀｲﾌﾟ

φ68mm)

MODEL8141 (ﾘｰｸｾﾝｻ

1A ﾀｲﾌﾟ

φ24mm)

MODEL8142 (ﾘｰｸｾﾝｻ

1A ﾀｲﾌﾟ

φ40mm)

MODEL8143 (ﾘｰｸｾﾝｻ

1A ﾀｲﾌﾟ

φ68mm)

MODEL8128 (ｸﾗﾝﾌﾟｾﾝｻ

50A ﾀｲﾌﾟ

φ24mm)

MODEL8127 (ｸﾗﾝﾌﾟｾﾝｻ

100A ﾀｲﾌﾟ

φ24mm)(

ｸﾗﾝﾌﾟｾﾝｻ

100A ﾀｲﾌﾟ

φ24mm)

MODEL8126 (KEW8121

ｸﾗﾝﾌﾟｾﾝｻ

200A ﾀｲﾌﾟ

φ40mm)

MODEL8125 (ｸﾗﾝﾌﾟｾﾝｻ

500A ﾀｲﾌﾟ

φ40mm)

KEW8122(

ｸﾗﾝﾌﾟｾﾝｻ

500A ﾀｲﾌﾟ

φ40mm)

KEW8123(

ｸﾗﾝﾌﾟｾﾝｻ

1000A ﾀｲﾌﾟ

φ55mm)

MODEL8124 (ｸﾗﾝﾌﾟｾﾝｻ

1000A ﾀｲﾌﾟ

φ68mm)
φ110mm)

KEW8130( ﾌﾚｷｼﾌﾞﾙｾﾝｻ

1000A ﾀｲﾌﾟ

KEW8133※( ﾌﾚｷｼﾌﾞﾙｾﾝｻ

3000A ﾀｲﾌﾟ

φ170mm)

KEW8129( ﾌﾚｷｼﾌﾞﾙｾﾝｻ

3000A ﾀｲﾌﾟ

φ150mm)

電源供給ｱﾀﾞﾌﾟﾀ MODEL8329（CAT Ⅲ150V、CAT Ⅱ240V）※

KEW8133 については、本体ファームウェアが下表以降のバージョンでないと、使用することができません。

ご使用されている本体ファームウェアのバージョンを確認してください。
なお、最新の本体ファームウェアのダウンロードは弊社ホームページから行うことができます。
本体ファームウェアのバージョン
V1.10 以降

KEW5050− 82 −

11.3 測定仕様

KEW5050

11.3 測定仕様

KEW5050

リアルタイム OS：

本製品は、T-Engine フォーラム(www.t-engine.org)の T-License に基づき T-Kernel ソースコードを利用しています。
本製品のソフトウェアの著作権の一部は、(c) 2010 The FreeType Project (www.freetype.org)のものです。すべての権利はその所有者に帰属します。
外部通信：USB ※接続 USB ケーブル長は、2m 以下であること

ｺﾈｸﾀ形状mini-B

通信方式

USB Ver2.0 準拠

USB 認識番号

ﾍﾞﾝﾀﾞｰ ID:12EC (Hex)、 ﾌﾟﾛﾀﾞｸﾄ ID:5050 (Hex)、 ｼﾘｱﾙ番号:0+7 桁機体番号

通信速度

12Mbps (ﾌﾙｽﾋﾟｰﾄﾞ)

デジタル出力端子：

端子 H、L の基本回路はオープンコレクタ出力方式です。端子 L はアースコードを介して接地されています。端子 H
はプルアップ電圧を持たない外部接続機器に備え、10ｋΩのプルアップ抵抗を介して本体内部で 5V にプルアップしています。イベントが発生している間、端子 H は端子 L に接続され、端子間電圧が 0V になります。イベントの継続している時間が１秒未満であった場合には 1 秒間、端子間電圧を 0V にします。ただし、イベントが複数重なった場合には各イベントの論理和で 0V になります。
ｺﾈｸﾀ形状貫通型ｽｸﾘｭｰﾚｽ端子台 2 極 (ML800-S1H-2P)

出力方式

ｵｰﾌﾟﾝｺﾚｸﾀ出力、端子 H - L 間 0V ｱｸﾃｨﾌﾞ

入力電圧0～30V、50mAmax、200mWmax
ｲﾍﾞﾝﾄ検出中 ：0V～1V

端子間電圧
入力定格

待機状態

：4V～5V (本体内部ﾌﾟﾙｱｯﾌﾟ抵抗 10ｋΩ、5V)

端子 H

対地間最大定格 30V、50mA、200mW

端子 L

アースコードを介して接地

11.3 測定仕様
周波数 f 〔Hz〕
測定方式

：レシプロカル方式、50Hz は 10 周期／60Hz は 12 周期の累積カウントの逆数から演算する。

表示桁数

3桁

確度

正弦波 40～70Hz、10Vrms 以上の基準電圧 V において ±2dgt

表示範囲

10.0～99.9Hz (範囲外は”----”表示)

信号ｿｰｽ

基準電圧

測定項目別、解析データ数
200ms(50Hz：10 周期、60Hz：12 周期)を 1 測定範囲として、8192 ﾎﾟｲﾝﾄのﾃﾞｰﾀで演算する項目
漏洩電流実効値(Iom)、基準電圧実効値(Vm)200ms(50Hz：10 周期、60Hz：12 周期)を 1 測定範囲として、4096 ﾎﾟｲﾝﾄのﾃﾞｰﾀで演算する項目
漏洩電流(Io)、基準電圧(V)、対地抵抗成分漏洩電流(Ior)、位相角(θ)、絶縁抵抗値(R)約 40.96ksps (24.4μs 間隔)で測定する項目
漏洩電流瞬間のピーク値(IomP)、基準電圧瞬間のピーク値(VmP)

83− 83 −

KEW5050

KEW5050

KEW5050

11.3 測定仕様

瞬時測定項目測定方式：40.96ksps(24.4μs 間隔)、ｷﾞｬｯﾌﾟなしで約 200ms ごとに実効値を算出する。

有効周波数範囲：40～70Hz

漏洩電流実効値 (リーク電流検出型クランプセンサ)
負荷電流実効値 (負荷電流検出型クランプセンサ)
Iom〔A rms〕
リーク電流検出型クランプセンサ
8177/8178(10A タイプ)： 10.000m／100.00m／1000.0m／10.000A／AUTO

8146/8147/8148(10A タイプ)： 10.000m／100.00m／1000.0m／10.000A／AUTO

8141/8142/8143(1A タイプ)：

5.000m／50.00m／500.0m／1.000A／AUTO

負荷電流検出型クランプセンサ
ﾚﾝｼﾞ

表示桁数
有効入力範囲表示範囲
ｸﾚｽﾄﾌｧｸﾀ
確度
入力ｲﾝﾋﾟｰﾀﾞﾝｽ

8128(50A タイプ)：

500.0m／5.000A／50.00A／AUTO

8121/8127(100A タイプ)：

1000m／10.00／100.0A／AUTO

8126(200A タイプ)：

2.000／20.00／200.0A／AUTO

8122/8125(500A タイプ)：

5.000／50.00／500.0A／AUTO

8123/8124/8130(1000A タイプ) ：10.00／100.0／1000A／AUTO

8133(3000A タイプ) ：30.00／300.0／3000A／AUTO

8129(3000A タイプ) ：300.0／1000／3000A
リーク電流検出型クランプセンサ ：5 桁
負荷電流検出型クランプセンサ：4 桁
各ﾚﾝｼﾞの 1%～110% (rms)および、ﾚﾝｼﾞの 200% (peak)
各ﾚﾝｼﾞの 0.15%～130% (0.15%未満は 0 表示、超えた場合は OL 表示)
3 以下
正弦波 40～70Hz の測定波形において ±0.2%rdg±0.2%f.s.＋クランプセンサ振幅確度
約 1MΩ

演算式※１

基準電圧実効値 Vm〔V rms〕
ﾚﾝｼﾞ1000.0V

表示桁数

5桁

有効入力範囲10～1000Vrms および、2000Vpeak

表示範囲

0.9V～1100.0Vrms (0.9V 未満は 0 表示、超えた場合は OL 表示)

ｸﾚｽﾄﾌｧｸﾀ

2 以下

確度

測定波形 正弦波 40～70Hz において

入力ｲﾝﾋﾟｰﾀﾞﾝｽ

約 4MΩ

演算式※１

※１

V: 基準電圧、Io: 漏洩電流、i ：ｻﾝﾌﾟﾘﾝｸﾞﾎﾟｲﾝﾄ№、n：約 8192 ﾎﾟｲﾝﾄ

KEW5050− 84 −±0.2%rdg±0.2%f.s.

11.3 測定仕様

KEW5050

11.3 測定仕様

KEW5050

演算項目
測定方式：ﾃﾞｼﾞﾀﾙ PLL 同期方式

測定方法

：高調波解析後の基本波にて演算する。

有効周波数範囲：40～70Hz

ｳｲﾝﾄﾞｳ幅：50Hz は 10 周期、60Hz は 12 周期

ｳｲﾝﾄﾞｳの種類：ﾚｸﾀﾝｷﾞｭﾗ

解析ﾃﾞｰﾀ数：4096 ﾎﾟｲﾝﾄ

解析ﾚｰﾄ：50Hz/60Hz にて約 1 回/200ms、ｷﾞｬｯﾌﾟなし。

漏洩電流実効値 基本波 (リーク電流検出型クランプセンサ)
負荷電流実効値 基本波 (負荷電流検出型クランプセンサ)
Io〔A rms〕
ﾚﾝｼﾞ

漏洩電流実効値・負荷電流実効値に同じ

表示桁数

漏洩電流実効値・負荷電流実効値に同じ

有効入力範囲漏洩電流実効値・負荷電流実効値に同じ

表示範囲

漏洩電流実効値・負荷電流実効値に同じ測定方式

解析ｳｲﾝﾄﾞｳ幅は 50/60Hz に対して 10/12 ｻｲｸﾙ、測定値は基本波のみで演算

確度

正弦波 40～70Hz、90Vrms 以上の基準電圧 V において±0.2%rdg±0.2%f.s.＋クランプセンサ振幅確度

演算式※２、３

Io c 

Io 10 k r 2  Io 10 k i 2

基準電圧 V〔V rms〕
ﾚﾝｼﾞ

基準電圧実効値に同じ

表示桁数

基準電圧実効値に同じ

有効入力範囲

基準電圧実効値に同じ

表示範囲

基準電圧実効値に同じ測定方式

解析ｳｲﾝﾄﾞｳ幅は 50/60Hz に対して 10/12 ｻｲｸﾙ、測定値は基本波のみで演算

確度
演算式※２、３基準電圧実効値に同じ

V V 10 k r 2  V 10 k i 2

基準電圧電流位相差 θ〔deg〕
表示桁数
表示範囲
測定方式
確度

演算式※３

4桁
0.0°～±180.0°(基準電圧 V の位相を基準値 0.0°とする)
進み：0 ～ ＋180°/ 遅れ：0 ～ －180°
解析ｳｲﾝﾄﾞｳ幅は 50/60Hz に対して 10/12 ｻｲｸﾙ、測定値は基本波のみで演算
正弦波 40～70Hz、90Vrms 以上の基準電圧 V、漏洩電流 Io レンジの 10％以上の入力において：±0.5°以内
Ior 用リーク電流検出型クランププセンサとの組み合わせ：±1.0°以内
その他、汎用クランププセンサとの組み合わせ：±0.5°以内+クランプセンサ確度  Io  V

 Vr 
  Vi  V  tan 1 85

− 85 −

 Io r   Io i  Io  tan 1 

KEW5050

KEW5050

KEW5050

11.3 測定仕様

KEW5050

11.3 測定仕様

対地抵抗成分漏洩電流 Ior〔A rms〕
レンジ

漏洩電流実効値・負荷電流実効値に同じ

表示桁数

漏洩電流実効値・負荷電流実効値に同じ

有効入力範囲漏洩電流実効値・負荷電流実効値に同じ
リーク電流検出型クランプセンサとの組み合わせでのみ表示する

表示範囲

漏洩電流実効値・負荷電流実効値に同じ測定方式

解析ｳｲﾝﾄﾞｳ幅は 50/60Hz に対して 10/12 ｻｲｸﾙ、測定値は基本波のみで演算
正弦波 40～70Hz、90Vrms 以上の基準電圧 V において±0.2%rdg±0.2%f.s.＋クランプセンサ振幅確度+位相確度による誤差※(位相誤差)※

Ior 用リーク電流検出型クランプセンサとの組み合わせで、漏洩電流 Io の測定値

確度

±2.0%rdg を追加する (θ：基準電圧電流位相差の確度±1.0°以内)。
「測定確度算出例」

注記：
汎用リーク電流検出型クランプセンサでは、確度を規定しないKEW8178 使用時、10mA レンジにて、Ior= 1mA、Io= 5mA を測定した場合の Ior 測定確度
±0.2%rdg±0.2%f.s.±１.0%rdg (KEW8178 振幅確度)+ Io×±2.0%rdg (クランプセンサ位相誤差：±1.0°)= 1mA (Ior)×±0.2%+10mA (Ior_f.s.)×±0.2%+1mA (Ior)×±1.0%+5mA (Io)×±2.0%= ±0.002mA±0.02mA±0.01mA±0.1mA= ±0.132mA±0.132mA÷1mA(Ior) = ±0.132

従って 1mA に対する Ior 確度は±13.2%rdg となる演算式※２、３
1P2W
1P3W

3P3W

Ior 

Ior

V

  Io10 k r  V10 k i  Io10 k i

10 k rV
2 3 V10 k r  Io10 k i  V10 k i  Io10 k r
3V

平衡状態の静電容量成分漏洩電流(Ioc)の和はゼロになる
3P4W

KEW5050

KEW5050

Ioc  Ioc _ L1  Ioc _ L 2  Ioc _ L 3  0
 Ior  Io

86

− 86 −

11.3 測定仕様

KEW5050

11.3 測定仕様

KEW5050

絶縁抵抗値 R〔ohm〕
レンジ

20.00MΩ

表示桁数

4桁
リーク電流検出型クランプセンサとの組み合わせでのみ表示する

表示範囲

レンジの 0.15%～130%(0.15%未満は 0 表示、超えた場合は OL 表示)
基準電圧 V/漏洩電流 Io が 0 表示または、OL 表示の場合“----”を表示する

演算式※２

R

V
Ior

※２

V: 基準電圧、Io: 漏洩電流※３k=1: 高調波 1 次(基本波)、r: FFT 変換後の実数成分、i: FFT 変換後の虚数成分
演算式の測定周期は 10 周期であり、12 周期ならば式中の"10k"を"12k"に置き換える。

87− 87 −

KEW5050

KEW5050

KEW5050

11.3 測定仕様

KEW5050

11.3 測定仕様

イベント項目
実効値の上限値 H:Iom / H:Io / H:Ior〔A rms〕 / H:Vm / H:V〔V rms〕
実効値の下限値 L:VmL / L:V〔V rms〕
測定方式

各測定項目に同じ

ﾚﾝｼﾞ

各測定項目に同じ

表示桁数

各測定項目に同じ

有効入力範囲各測定項目に同じ

表示範囲

各測定項目に同じ

ｸﾚｽﾄﾌｧｸﾀ

各測定項目に同じ

確度

各測定項目に同じ

入力ｲﾝﾋﾟｰﾀﾞﾝｽ

各測定項目に同じ

漏洩電流瞬間のピーク値Ｐｋ：Iom〔A peak〕
測定方式

約 40.96ksps (24.4μs 間隔)でｷﾞｬｯﾌﾟなしに、ｲﾍﾞﾝﾄの有無を判定する

ﾚﾝｼﾞ

漏洩電流実効値・負荷電流実効値に同じ

表示桁数

漏洩電流実効値・負荷電流実効値に同じ

有効入力範囲各ﾚﾝｼﾞの 0.15% (1mA≦)～200% (peak)

表示範囲

各ﾚﾝｼﾞの 0.15% (1mA≦)～200% (peak)
各レンジの 100% (DC)にて規程
1/10/100 倍レンジ：±0.5%f.s.＋クランプセンサ振幅確度1000 倍レンジ：±5.0%f.s.＋クランプセンサ振幅確度※
Ior 用リーク電流検出型クランプセンサでは、
1 倍：10A / 10 倍：1000mA /100 倍：100mA / 1000 倍：10mA となる

確度

入力ｲﾝﾋﾟｰﾀﾞﾝｽ

約 1MΩ

しきい値

絶対値にてﾋﾟｰｸ電流を指定する

基準電圧瞬間のピーク値Ｐｋ：Vm〔V peak〕
測定方式

約 40.96ksps (24.4μs 間隔)でｷﾞｬｯﾌﾟなしに、ｲﾍﾞﾝﾄの有無を判定する

ﾚﾝｼﾞ

基準電圧実効値に同じ

表示桁数

基準電圧実効値に同じ

有効入力範囲

50V～2000V (peak)

表示範囲

50V～2000V (peak)

確度1000V (DC)にて規程 ±0.5%f.s.

入力ｲﾝﾋﾟｰﾀﾞﾝｽ

約 4MΩ

しきい値

絶対値にてﾋﾟｰｸ電圧を指定する

KEW5050

KEW505088− 88 −

12.1 トラブルシューティング

KEW5050

12.1 トラブルシューティング

KEW5050

12章 故障かなと思ったら
12.1

トラブルシューティング

本製品を使用しているときに故障かなと思われる内容が発生した場合、下記の事項を確認してください。下記以外の不具合が認められる場合は、弊社または販売店までご連絡をください。
症状

確認事項

電源キーを操作しても電源が入 ＜AC アダプタを接続している場合＞らない。(何も表示しない。)

・電源コードがコンセントに正しく接続されているか確認してください。
・ＡＣアダプタ出力端子とアースコードとが本体に正しく接続されているか確認してください。
・電源コード、ＡＣアダプタ出力ケーブル、アースコードが断線していないか
確認してください。
・電源電圧が許容範囲内か確認してください。
＜電池駆動の場合＞
・電池の極性が正しくセットされているか確認してください。
・単３形ニッケル水素電池（Ni－MH）の場合、充分充電されているか確認してください。
・単３形アルカリ乾電池の場合、電池が消耗していないか確認してください。
＜上記が問題ない場合＞
・一旦ＡＣアダプタと、すべての電池を取り外し、再度ＡＣアダプタまたは、
すべての電池をセットし直して電源を入れてください。この状態でも電源が入らない場合には本体が故障している可能性があります。

電源が切れない。

・キーロック機能が動作していないか確認してください。
・画面表示が更新していない場合には、一旦ＡＣアダプタと、すべての電池を取り外し、再度ＡＣアダプタまたは、すべての電池をセットし直して電源を入れてください。この状態でも正常に動作しない場合には本体が故障している可能性があります。

キー操作ができない。

・キーロック機能が動作していないか確認してください。
・本書にて各測定レンジの有効キーを確認してください。

無入力でゼロ「0」を表示しない。 ・基準電圧入力端子間をオープン状態または、電流入力端子にクランプを接続していない場合でも数値が表示されますが、測定には問題ありません。
・電流入力端子にクランプを接続し、被測定導体をクランプしていない場
合、表示がゼロにならないことがありますが、測定には問題ありません。

89− 89 −

KEW5050

KEW5050

KEW5050

12.1 トラブルシューティング

KEW5050

12.1 トラブルシューティング

症状

確認事項

測定値を表示しない。

・電圧用測定コードが正しく接続されているか確認してください。電流だけを測定表示値が不安定、または測定したい場合でも、電圧用測定コードを接続していないと測定値が安定

おかしい。

しません。
・クランプセンサの向きが正しく接続されているか確認してください。
・ＡＣアダプタを接続している場合には、本体を接地する必要があります。
付属のアースコードを使用し、正しく接地されているか確認してください。
・入力している基準電圧の周波数が４０～７０Ｈｚの範囲内か確認してください。４０～７０Ｈｚが測定できる範囲です。
・測定ラインに対して本製品の設定および結線が正しいか確認してください。
・使用しているクランプセンサとクランプの設定が正しいか確認してください。汎用クランプセンサは手動で設定する必要があります。
・電圧用測定コードの断線または、クランプセンサが故障していないか確認
してください。
・入力信号にノイズがのっている可能性がないか確認してください。
・近くに強い電磁波がないか確認してください。
・使用環境が本製品の仕様の範囲内か確認してください。

ＳＤカードに保存ができない。

・ＳＤカードが正しく挿入されているか確認してください。

または、保存したデータが読み・ＳＤカードが本体付属のＳＤカードまたは、オプションのＳＤカードか確認し込めない。

てください。その他のＳＤカードでは正常動作を保証できません。
・ＳＤカードが KEW５０５０本体でフォーマットされたものか確認してください。その他の機器でフォーマットすると記録できる容量が小さくなったり、または記録したデータが読めない可能性があります。
・ＳＤカードの空き容量を確認して記録可能時間が残っているか確認してください。
･ＳＤカードの故障が疑われる場合、既知のハードウエアで正常に動作することを確認してください。

USB 通信でダウンロードおよ・付属の USB コードで正しくＰＣに接続しているか確認してください。

び設定ができない・通信アプリケーションソフト(KEW Ｗｉｎｄｏｗｓ for KEW５０５０)において接続デバイスが表示されていることを確認ください。表示していない場合、
USB ドライバが正常にインストールできていない可能性があります。別紙の「ＫＥＷ Ｗｉｎｄｏｗｓ ｆｏｒ ＫＥＷ５０５０」用のインストールマニュアルを参考にしてＵＳＢドライバをＰＣへ再インストールしてください。

USB 接続で SD カードを認識しない

・記録中は測定データを保護するために、ＰＣではＳＤカードが抜かれた状
態で認識されます。
・USB ドライバが正常にインストールできていない可能性があります。別紙の「ＫＥＷ Ｗｉｎｄｏｗｓ ｆｏｒ ＫＥＷ５０５０」用のインストールマニュアルを参考にしてＵＳＢドライバをＰＣへ再インストールしてください。

KEW5050

KEW505090− 90 −

12.2 入力項目と表示項目の制限

KEW5050

入力項目と表示項目の制限
設定によって入力できる項目と、表示しない項目があります。ここでは、その内容と対処方法を説明します。
症状

確認事項

「シリアル№」を入力または削除・Ior 用リーク電流検出型クランプセンサまたは、負荷電流検出型クランプできない。

センサを設定していないか確認してください。汎用リーク電流検出型クランプセンサのみが本機能の対象になります。詳しくは「シリアル№：」44頁を参照してください。

イベントの検出を「ON」にできな・クランプセンサの電流レンジに「ＡＵＴＯ」を設定していないか確認してくだい。

さい。「ＡＵＴＯ」を設定したｃｈのイベント検出は自動的に「ＯＦＦ」になります。イベントを検出する場合には、イベントしきい値を含んだ固定レンジに設定してからイベントの検出を「ON」にしてください。詳しくは「上限しきい値(Ｈ)/ｃｈ」46 頁と「ピークしきい値(Ｐｋ)/ｃｈ」50 頁とを参照してください。

エラーメッセージの内容とその対処方法
本製品使用中に、画面にメッセージが表示されることがあります。
ここでは、その内容と対処方法を説明します。
メッセージ
記録を開始できません。
ＳＤカードを確認してください。
保存できません。
ＳＤカードを確認してください。
ＳＤカードが挿入されていません。
ＳＤカードの空き領域が足りません。
記録を中止しました。

内容／対処方法
・ＳＤカードが正しく挿入されているか確認してください。
・ＳＤカードに問題があると思われ場合には、「12.1 トラブルシューテ
ィング」 「ＳＤカードに保存ができない。または、保存したデータが読み込めない。」 90頁の欄を参照してください。
・必ず記録をストップし「記録を停止しました」のメッセージが消えた
後に保存しているファイルをＰＣ等にバックアップして、ファイルを削除、またはフォーマットするか、本体でフォーマットした別のＳＤ
カードを使用して記録を再開してください。詳しくは「記録データ」57
頁を参照してください。

ＳＤカードに空き領域がありませんフォーマットまたはデータ削除を行ってくだ
さい。

・ＳＤカードの空き容量を確認してください。容量が少ない場合には本体でフォーマットした別のＳＤカードを使用するか、保存しているファイルをＰＣなどにバックアップして、ファイルを削除するか、またはＳＤカードをフォーマットして、記録を再開してください。詳しくは「記録データ」57 頁を参照してください。

− 91 −

KEW5050

KEW5050

12.3 エラーメッセージの内容とその対処方法
メッセージ

内容／対処方法

設定と異なるクランプが接続されています。接続を確認してください。

・前回の測定時と異なるクランプセンサを接続している場合に表示します。
・ＫＥＷ５０５０ではＩｏｒ用クランプセンサのみ自動認識できます。汎用
クランプセンサを設定するには前回と異なるセンサーを検出しました。
設定を確認してください。

の「基本設定」-「電流クランプ」から手動で行ってください。
・電流センサが本体に確実に接続されているか確認してください。
・故障が疑われる場合には下記の手順で確認してください。

センサーの接続が正しくありません。
接続を確認してください。

NG と識別された電流センサの接続ｃｈを正しく識別されているｃｈと変更して再テストを行ってみてください。この時、前回と同じｃｈがNG と識別された場合には、本体が故障している可能性があります。前回 NG と識別された電流センサを接続しているｃｈが、NG と識別された場合には、電流センサが故障している可能性があります。故障していた場合には直ちに使用を中止してください。

過去の日時が指定されています記録

・記録「開始方法」に連続記録／時間帯指定を選択し、記録終了の開始方法を確認してください。

日時に過去の日時を設定していた場合に表示します。それぞれの記録方法にて設定した日付を確認してください。詳しくは「記録方法」53 頁を参照してください。

記録中および記録待機中は本体設定・記録中には設定の確認しかできません。設定を変更する場合にを変更できません。

は、必ず記録をストップし「記録を停止しました。」のメッセージがAUTO レンジはイベント検出が無効に・クランプセンサの電流レンジを「AUTO」設定にしていると電流に関消えた後に変更してください。
なります。

するイベント判定を ON にできません。イベントしきい値を含んだ固定電流レンジに変更してから ON にしてください。

KEW5050− 92 −

13 章 アフターサービス

KEW5050

13章 アフターサービス●修理・校正を依頼されるにはお買い上げいただいた販売店または弊社サービスセンターにお送りください。
●製品のご使用に関するお問い合わせは弊社お客様相談室にご連絡ください。
●校正周期について
本製品を正しくご使用いただくため、定期的（推奨校正周期１年）に校正することをおすすめいたします。
●補修用部品の保有期間
本製品の機能・性能を維持するために必要な補修部品を製造打ち切り後、５年間を目安に保有しています。
この取扱説明書に記載されている事項を断りなく変更することがありますのでご了承ください。

■ホームページのご案内
www.kew-ltd.co.jp●新製品情報●取扱説明書／ソフトウェア／単品カタログのダウンロード●販売終了製品情報ご使用に関するお問い合わせは共立電気計器
電話受付時間

修理・校正に関するお問い合わせは

お客様相談室

共立電気計器 サービスセンター

9:00 ～ 12:00、13:00 ～ 17:00〒797-0045 愛媛県西予市宇和町坂戸４８０（土・日・祝日・年末年始・夏季休暇を除く）

0120-62-1172※折り返しお電話させていただくことがございますので発信者番号の通知にご協力いただきますようお願いいたします。
※フリーコールをご利用いただけない場合は、最寄りの弊社営業所へおかけください。

0894-62-1172
修理・校正を依頼される場合は事前に電池の消耗、ヒューズや測定コードの断線を確認してから輸送中に損傷しないように十分梱包した上で
弊社サービスセンターまでお送りください。

− 93 −

KEW5050

保ＫＥＷ５０５０
保証期間

証

書

年

月

製造番号

ご購入日（

日）より１年間共立製品をお買い上げいただきありがとうございます。保証期間内に正常なご使用状
態で万一故障が生じた場合は、保証規定により無償修理をさせていただきます。本書を添付の上ご依頼ください。
お名前
ご住所
ＴＥＬ
◎本保証書に製造番号、ご購入日、およびお名前、ご連絡先をご記入の上、大切に保管してください。
◎本保証書の再発行はいたしません。
◎本保証書は日本国内でのみ有効です。This warranty is valid only in Japan.
保証規定
保証期間内に生じました故障は無償で修理いたします。
ただし、下記事項に該当する場合は対象から除外させていただきます。
1． 取扱説明書と異なる不適切な取扱い、または使用方法が原因で発生した故障。
2． お買い上げ後の持ち運びや輸送の間に、落下させるなど異常な衝撃が加わって生じた故障。
3． 弊社サービス担当者以外による改造、修理が原因で生じた故障。
4． 火災、地震、水害、公害およびその他の天変地異が原因で生じた故障。
5． 傷など外観上の変化。
6． その他弊社の責任と見なされない故障。
7． 電池など消耗品の交換、補充。
8． 保証書のご提出がない場合。

9-19
92-2298B
3-2392-2298C