MS Bench (All Models)Modular LC-MS bench system designed exclusively for SCIEX.
Part Number : 99-9999
See Reverse

Service Kit : See Reverse

Your local gas generation partner

Developed exclusively for SCIEX, the MS Bench SCI range provides a
modular workstation with provision for either integrated gas generation ora noise abating enclosure for roughing pumps. MS Bench SCI is designed
specifically for use with the current and latest mass spectrometers atSCIEX*.
Two variants of the MS SCI Bench are available, both identical in form
factor, aesthetics and work surface. MS Bench (G) SCI features a selfcontained gas generator, providing a reliable and cost-efficient sourceof both nitrogen (Curtain Gas™) and clean, dry oil-free air for source andexhaust gas at flows and pressures configured to meet SCIEX instrument
requirements (exluding IVD medical devices). The other variant, theMS Bench SCI, comes without the gas generator, instead providing
a compartment below the bench, suitable for housing up to two MS
roughing pumps.

Applications)

MS Bench (G) SCI for SCIEX LC-MS instruments (excluding IVD
medical device instruments)

Key Features
MSMS Bench SCI 1, (G) SCI 1 & (G) SCI 2914 mm)

787 mmDesigned exclusively for SCIEX to provide modular
bench solution for all SCIEX LC-MS instruments*
Noise abated compartment suitable for housing
vacuum pump(s) (non generator variant only)

762 mmNoise & vibration dampening, suitable for use as
instrumentation bench* (*MS can only be placed on
MS Bench with NO internal gas generator)

MSMS Bench SCI 2

1040 mm‘Genius Inside’: true plug & play gas generation (MSBench (G) SCI 1 only).
No external compressed air source required.

787 mmCSA / FCC / CE compliant
Chemical-resistant phenolic resin worktop
762 mm

Self-levelling castor wheels for seamless installation
with existing lab bench system*suitable for all current and latest MS from SCIEX including the new 5500+, but excluding IVD
medical device instrument. Floor-standing Triple TOFs can house vac pumps in MS Bench SCI.
Though ZenoTof 7600 can be supplied by the MS Bench (G) 1, it cannot be situated on top of anyMS Bench.

Copyright 2021 Peak Scientific - Rev - MS Bench Data sheet Rev 8 RSID 3060 EN - 18/06/21

Technical Specifications

MS Bench SCI 1

MS Bench (G) SCI 1

MS Bench SCI 2

Minimum Operating Ambient Temperature

5°C (41°F)

Maximum Operating Ambient Temperature

30°C (86°F)

MS Bench (G) SCI 2

Curtain Maximum Flow^

19 L/min @ 4.5 bar(0.67 cfm @ 65 psi)

25 L/min @ 4.48 bar(0.882 cfm @ 65 psi)

Source Maximum Flow^

26 L/min @ 6.9 bar(0.92 cfm @ 100 psi)

26 L/min @ 6.89 bar(0.918 cfm @ 100 psi)

25 L/min @ 4.1 bar(0.88 cfm @ 60 psi)

N/A

Exhaust Maximum Flow^25 L/min @ 4.13 bar(0.882 cfm @ 60 psi)

N/AParticles<0.01μm<0.01μm

Phthalates

None

None

Suspended Liquids

None

None

Fans

8 x Fans
IO: 85-264 VAC 50/60
Hz,1 Amps

220 - 240v ±10% 50/60
Hz,7 Amps

Power Consumption

200VA

1,100VA

Maximum Heat Output

784 BTU/hr

3753 BTU/hr

Noise abating to54dB(A)59dB(A)580 x 650 x 470 mm
22.8 x 25.6 x 18.5”

640 x 290 x 680 mm
25.2 x 11.4 x 26.7”

Electrical Requirements

Noise Level
Usable Compartment Space (H x W x D)

762 x 914 x 787 mm /30 x 36 x 31’’

MS Bench Dimensions (H x W x D)
MS Bench Weight

103kg (227.1 lbs)

160kg (352.8 lbs)

Max Weight Load

272kg (599.7lbs)

215kg (474lbs)

100 - 240v ±10% 50/60
Hz,2 Amps

220 - 240v ±10% 50/60
Hz,9 Amps

762 x 1040 x 787 mm /30 x 42 x 31’’762 x 914 x 787 mm /30 x 36 x 31’’103kg (227.1 lbs)

169kg (372.5 lbs)

^NOTE – These performance characteristics are valid only when paired with an approved Sciex application. Please refer to current application matrix for a list of supported products.

Ordering Information
Part Number3302382

Annual Service3302383
visit: www.peakscientific.com/ordering

Complete Maintenance Plan

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that your investment continues tobe [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you thereassurance of instant access to worldwide technical support and priority on-site response in the untimely event of a breakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America

Product Certifications

Tel: +1 866 647 1649
Fax: +1 978 608 9503
Peak Scientific China
Tel: +86 21 5079 1190
Fax: +86 21 5079 1191
For a full list of our worldwide
office locations, please visit:

Peak Scientific’s Quality
Management System
conforms to:

Web: www.peakscientific.com
Email: discover@peakscientific.com

Copyright 2021 Peak Scientific - Rev - MS Bench Data sheet Rev 8 RSID 3060 EN - 18/06/21