Solaris XE for Waters™
Dedicated nitrogen solution for ACQUITY™ QDa™
Part Number : 99-9999

Service Kit : 99-9999

Your local gas generation partner

Description
Peak Scientific’s Solaris XE nitrogen generator offers a reliable,
compressor-free gas solution that can comfortably meet thedemands of most mass spectrometers within Waters’ portfolio ofinstruments. Using our tried and tested membrane technology,
Solaris XE can operate with varying flow rates (up to 35L/min ofhigh purity nitrogen), purity (95-99.5%) and outlet pressure can be
adjusted down from 116psi.
Solaris XE for Waters has been designed in collaboration withWaters and is the only compressor-free nitrogen generator available
which is tested and approved by Waters for use with the ACQUITY
QDa Detector. Solaris XE is fitted with alignment blocks specificallydesigned to enable the generator to be stacked underneath theACQUITY QDa in most modular stacking configurations*, providing
a compact and discreet space-saving solution within the lab.
Alternatively this unit can be wall mounted or free standing on a labbench.
The Solaris XE has been engineered to provide nitrogen tolaboratories that utilize an external source of compressed air, whichmeets a minimum quality grade of ISO 8573-1:2010 Class 1.4.1.
*Please ask your local sales representative for advice on supported stacking configurations.

Applications)

Waters ACQUITY QDa Detector))

A list of instruments that are compatible with Solaris XE can be foundon our Waters application list.

Nitrogen up to 35L/min, up to 116 psi @95-99.5% purity)

Tested and approved by Waters)

Compact, wall-mountable and stackable*with Waters ACQUITY QDa, maximizing theuse of valuable laboratory floor or bench space)

Variable outlet flow, pressure and purity)

Colour indicative LED lighting for easy systemstatus awareness)

Manufactured and tested to highest spec inUK, CE/FCC/CSA certified)

Safer and more convenient than pressurized
cylinders, dewars or bulk storage)

12 month on-site comprehensive warranty)

Backed by [Peak Protected] global on premise
servicing and technical support

343 mm / 13.5”

650 mm / 25.6”

156 mm / 6.2”
Copyright © Peak Scientific Instruments Ltd 2022 — Solaris XE Waters Data sheet Rev 5 RSID 2950 EN —01/02/2022

Technical Specifications

Solaris XE for Waters

Max Flow Rateup to 35 L/min

Max Pressure

up to 116 psi / 8 bar(user adjustable)

Purity95% - 99.5%*

Gas Outlets

1 x 1/4 “ BSPP

Min/Max Air Inlet Pressure
Min Inlet Air Quality

80 psi/5.5 bar - 145 psi/10 bar (116 psi/8 bar required for QDa)
ISO8573-1:2010 Class 1.4.1**

Contaminants

Phthalate & BHT Free

Suspended Liquids

None

Operating Temperature

5°C - 30°C / 41°F - °F

Electrical Requirements

Universal 100 / 230 VAC / 50 / 60 Hz / 0.1 A

Power Consumption

6.9 Watts

Heat Output

25 BTU/Hr

Generator Dimensions (HxWxD)

156 x 343 x 650 mm
6.2 x 13.5 x 25.6’’

Generator Weight

15.1 kg / 33.3 lbs

Max Weight Bearing Capability

100 kg / 220.5 lbs

Solaris Compressor compatible

NO*depending on inlet air pressure and outlet flow** This unit should be connected to an air supply that, as a minimum, meets ISO8573-1:2010 Class 1.4.1

Ordering Information
Part Number

3300670

Annual Service

www.peakscientific.com/ordering

FPPM Maintenance Plan

3300671-FPPM

Complete Maintenance Plan

3300671-CSP

Supported Maintenance Plan

3300671-SMP

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that your investment continues tobe [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you
the reassurance of instant access to worldwide technical support and priority on-site response in the untimely event of abreakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management Systemconforms to: ISO:9001:2008

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com

Copyright 2022 Peak Scientific - Rev - Solaris XE Waters Data sheet Rev 5 RSID 2950 EN - 01/02/22