Small Tool Instruments and DataManagement

Inspection Instrument for Indicators

i-Checker

Catalog No. E12015(2)For efficient in-house calibration

Various types of indicators can beFeatures• Inspection can be performed 2.5 times faster compared tothe previous model.
• This instrument achieves the highest accuracy in its class(Mitutoyo survey, February 2016) and, therefore, guarantees
ultra-reliable inspection results.
• Digital indicators equipped with a data output function arechecked very efficiently due to spindle positioning at theinspection points and recording of measurement results being
under fully automatic control.
• Analog type indicators are inspected in semi-automatic modewith the pointer of the indicator being manually adjusted ateach measuring point with automatic transfer of inspection
results and movement to the next measuring point.

Inspection using analog type indicator2

Inspection using digital type indicator• The latest standard for lever-operated dial test indicators,
ASME and DIN, are applied.
• Inspection standards to a customer's own specification can beimplemented.

inspected with a single tester.

• A laptop PC can be connected using a USB cable, which contributes
to space-saving.

• All functions necessary for inspection are combined in the control box so thatthe operator need not rely on excessive eye movement to adjust the pointer.
1.【DATA】button

Inputs the current position data andmoves spindle to just before the next
inspection position automatically.

STOP button

Automatictransfer

2. Final positioning is by
Jog-dial

RETRY button
DATA button

Automatictransfer

Jog dial/shuttle control
Jog shuttle (continuous up/down movement)
Jog dial (fine incremental movement)

3.【DATA】button

Inputs the current position data andmoves spindle to just before the next
inspection position automatically.
Notes: The distance at which spindle
movement is stopped before the next
graduation is user-deﬁnable.

CANCEL button

3

Optional Accessories and Applications
Applications for Bore Gage Accessory Set

Applications for bore gage measurement accessory set (type C)

Application for mounting bore gage accessory set (type C)
Order No. 02ASU161

i-Checker mountable bore gages
1

List of components for bore gage accessory set(type C) Order No.02ASU161
(1)

(3)

(4)

ab(10)

cabc4

Range (mm)ø6 - ø18
50 or below
120 or below

(5)
(2)

(1) Bore gage accessory(12)(6)(2) Stand B(11)(7)(3) V-shaped attachment B
(8)
(13)
(9)
(4) Bore gage stopper 5(5) Anvil (for calibration, M11 female)(10) V-shaped attachment C(6) Anvil (for calibration, M3.5 x 0.35 male) (11) Bore gage stopper 3(7) Anvil (for calibration, M3.5 x 0.5 male) (12) i-Checker's special(8) Anvil (for calibration, M5 male)contact point(9) Anvil (for calibration, M11 male)(13) Key wrenchOptional Accessories and Applications
Applications for Dial Test Indicator Accessory Set

Applications for dial test indicator accessory set (for horizontal type)

Applications for dial test indicator accessory set (for vertical type)

Dial test Indicator accessory set (for ø6 stem)
Order No.02ASK000

734148.590240

10059.5

Wooden box

2515

2746.55

Optional accessories
Attachments/Stem bush/Others/Connecting cables

Required optional accessories
Dedicated software (i-Pak specification)

Order No.

• Supported OSWindows10（64bit）

Item

Usage and remarks

Attachment
02ASK000 Test indicator accessory set For attachment of ø6 stem test indicator(for ø6mm stem)
Test indicator accessory set For attachment of ø8 stem test indicator
02ASK180 (for ø8mm stem)
02ASK370 Test indicator accessories set Holder to fix ø6 stem to 02ASK180(for ø6 stem)
02ASK380 Test indicator accessories set Holder to fix ø8 stem to 02ASK000(for ø8 stem)
Stem to attach Mu-Checker with knurled902803
ø6 dovetail grooved stem clamp ring (lever head type) to test indicator
accessory set (02ASK000)
Stem to attach Mu-Checker with knurled902804ø8 dovetail grooved stem clamp ring (lever head type) to test indicator
accessory set (02ASK180)
gage accessory set Accessory set for holding grip type bore gage
02ASU161 Bore(type C)
Stem bush
02ASK040 Stem bush ø602ASJ856 Stem bush ø8
02ASK150 Stem bush ø8, short

For attachment of cartridge head MCHP-341

02ASL150 Stem bush ø10For attachment of LGB2 fixing nut type withstem diameter ø9.5
Stem bush (ø15mm) is required separately.

02ASK050 Bush ø9.5 - ø1502ASK060 Stem bush ø1202ASK070 Stem bush ø15
02ASK080 Stem bush ø20
02ASK710 Stem bush ø28

For attachment of indicator with stem diameterø9.525mm
Storage of 7 stem bushes and 1 bush(ø9.5 - ø15mm)

02ASK090 Stem bush ø3/8 inWooden box for No.
02ASK130 02ASK130 stem bushes• Supported Industrial Standards
ISO, JIS, JMAS, ANSI/ASME, DIN, VDI/VDE/DGQ, BS• Basic inspection functions· Creation of inspection standard conformed to the industrial standard· Creation and edition of original inspection standard, by customizing theindustrial standard· Inspection of indicator using the inspection standard(inspection for accuracy and repeatability)· Graphical display of measurement result· Creation, edition, and printing of simplified inspection certificate• Function· Retry "measurement positioning"· Re-measurement· Data cancellation· Change of speed for jog shuttle• Input/Output specifications· Digimatic data connector (rear side of i-Checker main unit)*3· PC serial-data input port (RS-232C)· rinting of simplified inspection certificates*3 Automatic measurement requires the indicator's connecting cable.
Additionally some form of indicator, along with the normally connected accessory(the optional accessory for the indicator such as a Digimatic power-supply unit inan EF counter) will be required.

Others937179T

Used instead of [DATA] button in the operationbox.

Foot switch

02ASK730 Observation mirror
Connecting cables
905338905409936937965014937387

1Connecting cable (1m)* Used for connecting with Digimatic Indicators
Connecting cable (2m)*1 (ID-C, ID-S, ID-U, and IDU)
Connecting cable (1m)*1 (ID-F, IDF, Digimatic power supply unit andConnecting cable (2m)*1 between the testers)

965013Connecting cable (1m)*1 Used for connecting with Digimatic Indicators
Connecting cable (2m)*1 (IDC and IDA)

965275

Digimatic power supply Used for the Digimatic Indicators that require
external power supplyunit*2*1 Use the cable described in the user's manual.
*2 Confirm the specification in the user's manual.

6

Supported indicators• Dial Indicator• Dial Test Indicator• Mitutoyo Hicator• Bore Gage
• Digimatic Indicator• Linear Gage
• Electronic micrometer (Mu-Checker)* It cannot inspect the indicator with accuracy that is higher than the main unit
accuracy of the tester.
* Resolution of electronic micrometer (Mu-checker) is greater than 1μm (e.g.
1μm/5μm/10μm/50μm).

Supported Industrial Standards
Dial Indicator463-2006
R463-1965
B7503-2011
B7503-1997
B7503-1992
2001-1998
2001-1994
2003-1994
B89.1.10M-2001
B89.1.10M-1987
878-2006
878-1983

ISOJISJMAS
ANSI/ASMEDIN
VDI/VDE/DGQ

2618
Blatt11-1991

BS
Mitutoyo Standard

907-1965—

Dial Test Indicator

HicatorBore Gage

Digimatic Indicator

Linear Gage

Electronic
micrometer

9493-2010

—

—

———

B7533-2015
B7533-1990

B7519-1994

B7515-1982——

B7536-1982——

5009-1988——

5003-1962

B89.1.10M-2001
B89.1.10M-1987——

B89.1.10M-2001——

2270-1985

879-1983

—

—

—

—

2618
Blatt21-1991

—

—

—

2618
Blatt26-1991

—
—

—
—

—
✔

—
✔

—
—

2618
Blatt11.3-2002
2618
Blatt20-1991
2795-1981—

Specifications
Order No.
Measuring Range
Resolution
Accuracy vertical position(20°C) lateral position
Feed speedø8mm / ø9.525mm (3/8 in),Standard accessory
73

ø160

79.5

1.5×10 -6 / K
189

Semi-automatic measurement
Fully automatic measurement(only when using an indicator equipped with data outputfunction) *1*2
Mass20kg
Operating temperature range
20°C±0.5°C

Stroke 100

Measuring Unit
Thermal expansion
coefficient ofmeasurement mode

Unit: mm559.5

Measurement method

370.5*1 Automatic measurement requires the indicator’s connecting cable. Additionallysome form of indicator, along with the normally connected accessory (the optional
accessory for the indicator such as a Digimatic power-supply unit in an EF counter)
will be required.
*2 The indicator measured via RS-232C has the capability to receive data from themain unit and output the counter value.

384.8

Drive methodDimensions
170-402/170-403
100mm
0.01μm(0.1+0.4L/100) μm L = Arbitrary length (mm)(0.15+0.6L/100) μm L = Arbitrary length (mm)Maximum 10mm/s
Motor drive, semi-automatic, fully automatic(Fully automatic only for Indicator with SPC data output)
Separate type Linear Encoder

Standard accessories supplied Order No.170-402/170-403
Order No.
02AST250
02AST780
02AST830
02ASU001
02ASU002
02ASU003A
02ASJ856
02ASK091
601614-2
99MBD063A
—Item
QTY.
Tester main unit
1
Controller ICMC-2
1
Control box
1EXT. Signal Cable
1EXT. Motor Cable
1EXT. USB2.0 Cable
1
Stem bush Φ8mm, only for No.170-402
1
Stem bush Φ9.525mm (3/8 in), only for No.170-403
1
Dust cover
1
User's Manual (English)
1
Inspection certificate, Certification of calibration and1
Traceability chart (English)

164

ø2007

66 1701 (2) A-(CH)HS, Printed in Japan

Coordinate Measuring Machines

Vision Measuring Systems

Form Measurement

Optical Measuring

Whatever your challenges are, Mitutoyo
supports you from start to finish.
Mitutoyo is not only a manufacturer of top quality
measuring products but one that also offers qualified
support for the lifetime of the equipment, backed up
by comprehensive services that ensure your staff canmake the very best use of the investment.
Sensor Systems

Test Equipment
and Seismometers

Digital Scale and DRO Systems

Small Tool Instruments
and Data Management

Apart from the basics of calibration and repair,
Mitutoyo offers product and metrology training,as well as IT support for the sophisticated software
used in modern measuring technology. We can also
design, build, test and deliver bespoke measuringsolutions and even, if deemed cost-effective, takeyour critical measurement challenges in-house on a
sub-contract basis.

Find additional product literatureand our product catalogue
http://www.mitutoyo.co.jp/global.html
Mitutoyo Corporation20-1, Sakado 1-Chome,
Export permission by the Japanese government may be required for exporting our products according to the Foreign Exchange and Foreign
Trade Law. Please consult our sales office near you before you export our products or you offer technical information to a nonresident.
Note: Product illustrations are without obligation. Product descriptions, in particular any and all technicalspecifications, are only binding when explicitly agreed upon.
MITUTOYO and MiCAT are either registered trademarks or trademarks of Mitutoyo Corp. in Japan and/or other countries/regions.
Other product, company and brand names mentioned herein are for identification purposes only and may be thetrademarks of their respective holders.

Takatsu-ku, Kawasaki-shi,
Kanagawa 213-8533, Japan
T +81 (0) 44 813-8230
F +81 (0) 44 813-8231
http://www.mitutoyo.co.jp