Synos

IMAGING DETECTOR / CCD DETECTOR, CMOS DETECTOR FOR VISIBLE~1100nm SPECTRAL RANGEHIGH RESOLUTION CMOS DETECTOR
ISA071 / ISA071GL
High resolution & high performance, USB 3.0 interface CMOS detector for 400nm~1100nm spectral range
ISA071/ISA071GL is high resolution & high performance CMOS detectors for 400nm~1100nm
spectral range. It is suitable for optical beam proﬁle analysis of various optical device and module
such as laser diodes, optical ﬁber, etc. in 400~1100nm wavelength range. In combination withSynosʼ optical measurement optics M-Scope series and optical beam analysis module AD013,
it has wide application ﬁeld in optical beam observation and beam proﬁle analysis.
【Summary of speciﬁcation】【Spectral sensitivity】
Response
RelativeRelative
Response[%] [%]High resolution CMOS detector High resolution CMOS detector for laser
ISA071GL
ISA071
Speciﬁcation outline of image sensor
Imaging device
1/1.8” CMOS (with cover glass) 1/1.8” CMOS (cover glassless)Spectral range
400nm~1100nm
Total pixels
2048×1536 pixels (approx. 3.2 megapixels)
Pixels pitch
3.45×3.45 (μｍ)
Speciﬁcation outline of detector hardware
InterfaceUSB 3.0
Frame rate
60fps (at full resolution)
Shutter
Global shutter
Exporsure control 25μsec~4sec
Gradation
12bit ADC
Camera mount
C/CS mountPower supply
USB bus power (4.5~5.5 VDC supply voltage)
Operation environment +5℃~+45℃

1.0-

0.9-

0.8-

0.7-

0.6-

0.5-

0.4-

0.3-

0.2-

0.1-0 400(CAUTION) High Resolution CMOS Detector For Laser ISA071GL is using cover glassless
type CMOS image sensor. For this reason, free warranty after delivery does not apply.
When selecting ISA071GL, please be aware of this in advance and handle with care.

500600700

8009001000

Wavelength [nm]Wavelength [nm]【Standard component】○Detector head: 11” HIGH RESOLUTION CMOS DETECTOR
ISA0611 inch format high resolution & high performance digital CMOS detector for 400nm~1100nm spectral range
ISA061 is a CMOS detector using 2048×2048 pixels, NIR enhanced 1” progressive scan CMOS
imaging device. ISA061 is the dedicated detector of wide area type FFP measurement optics
M-Scope type FW, and high power laser FFP measurement optics M-Scope type HF.
【Summary of speciﬁcation】
Summary of speciﬁcation / Imaging device
Imaging device
1 inch progressive scan CMOS (NIR enhanced)
Spectral range
400nm - 1100nm
Pixels number
2048 × 2048 pixels / 4M pixels
Sensor pitch
5.5 x 5.5μm
Summary of speciﬁcation / Detector
Detector control I/F
Gigabit Ethernet (GigE Vision)
Frame rate
25fps
Shutter
GlobalExposure control
Programable mode
Gradation
8/12bit
Camera mountC mount
Power supply
12VDC【Spectral sensitivity】
Quantum Eﬃciency(%)

70【Standard component】○Detector head: 1○AC adapter: 16050403020100400

500600700800

Wavelength(nm)

900

1000

USB CMOS DETECTOR
ISA003-01
General purpose USB CMOS detector, mainly used for image observation and image processing
USB CMOS detector ISA003-01 is a monochrome USB CMOS detector, mainly used for image
observation and image processing in general purpose.
【Summary of speciﬁcation】○Image sensor:1/1.8” progressive scan monochrome CMOS
○Shutter:Global/rolling shutter○Eﬀective total pixels: 1280×1024 pixels (approx. 1.3M pixels）○Pixels pitch:
5.3×5.3μｍ○Frame rate:
60fps○Camera interface:USB3.0 Super speed (USB3.0 microB)○Gradation:
8bit /10bit○Minimum subject illuminance: 0.26Lux at F1.2
○Exposure time:
15.72μsec~10.3sec○Power supply/power consumption: +5V (typ.)(USB standard compliant)/2.0W or lesshttps://www.synos.jp

39〇測定制御PC

SYNERGY OPTOSYSTEMS CO., LTD