Synos

HIGH POWER LASER BEAM PROFILER SYSTEM SOLUTION / NFP MEASUREMENT SYSTEMSIMPLIFIED OPTICAL BEAM NFP MEASUREMENT SYSTEM FOR HIGH POWER LASER
Simpliﬁed optical beam proﬁle and NFP measurement system especially targeting for high power laser.

Simpliﬁed optical beam NFP measurement system for high power laser is a optical beam
proﬁle measurement system targeting for ~10W class high power laser. NFP measurement
optics for high power laser M-Scope type HL is used. The light ﬂux emitted from high power
laser sample is attenuated by beam sampler unit placed before objective lens, and is further
attenuated by ND ﬁlter at subsequent stage. The NFP image that has been attenuated toappropriate beam power is captured and image processing analysis is performed.
【Features】○M-Scope type HL, simpliﬁed NFP measurement optics for high power laser● Attenuation of incident beam with beam sampler placed before objective lens and ND ﬁlters● Optical magniﬁcation is maximum 20x (option, 10x objective lens and 2x intermediate lens）○Optical beam analysis module AP013, specially designed high-functional image processing software for optical beam proﬁle analysis● All-in-one package of PC, optical beam analysis software, detector driver, calibration data.
● High-performance image processing software for optical beam proﬁle measurement Optimetrics BA Standard is pre-installed.
【Option】【Standard component】○Option for optics (for M-Scope type HL)○Optics● 2× intermediate lens port MS-OP011-RL2● M-Scope type HLIntermediate lens unit that doubles the overall magniﬁcation*Please specify the measurement wavelength from 400~1100● 1/2× intermediate lens port MS-OP011-RLH
range because AR coating to some optical parts is required.
Intermediate lens unit that halves the overall magniﬁcation○Objective lens
● Coaxial epi-illumination port MS-OP011-CEP● M-Plan Apo 10× (ﬁxed)
Coaxial epi-illumination port with removable half mirror○Detector selection● 400-1100nm︓Hi-resolution CMOS detector ISA071/ISA071GL ○ND ﬁlter (for M-Scope type HS, dedicated 30φ）
○Optical beam analysis module AP013● Visible (400~700nm): NDF-5 (5 types per set)
● PC for image processing, optical beam analysis software● NIR (700~1100nm): NDF NIR-5 (5 types per set)
Optometrics BA Standard, detector driver, calibration○Coaxial epi-illumination light source
data, USB key● Visible~NIR: LED epi-illumination system○Accessories○Optics bench● Cables, instruction manuals, etc.
● Optics bench for ﬁber measurement with manual stages● Vertical setting optics bench【Detector selection, ﬁeld of view, pixel resolution (approx. value)】
Detector
Spectral range
Sensor size
Total pixels
Pixels pitch
Magniﬁcation
10×High resolution CMOS detector
ISA071/ISA071GL
400~1100nm
1/1.8 inch
2048×1536
3.45μｍField of view
Pixel resolution(unit:mm)(unit:μm)
0.7×0.52
0.345*Pixel resolution: The measurement length equivalent to 1 pixel of thedetector calculated from ﬁled of view and sensor pitch of the detector.
*The optical magniﬁcation when using MS-OP011-RL2 is 2 times themagniﬁcation in the table on the left, and the actual ﬁeld of view and pixel
resolution are 1/2. The maximum optical magniﬁcation is 20x when usinga 10x objective lens.
＊The optical magniﬁcation when using the MS-OP011-RLH is 1/2 of eachmagniﬁcation in the table on the left, and the actual ﬁeld of view and pixel
resolution are 2 times.

【Component selection of simpliﬁed NFP measurement system for high power laser】○Stages・optics bench○Optics selection○Detector selection

Sample stages
Optics stages
Optics bench for ﬁber
measurement○Optical beam analysis module AP013●Personal computer
・Main unit
・Accessories●for 400~1100nm

Simpliﬁed NFP measurement optics
for high power laser
M-Scope type HL

Hi-resolution CMOS detector
ISA071/ISA071GL

Objective lens 10× (ﬁxed)●Optical beam analysis software
Optometrics BA Standard●Detector driver●Calibration data●USB licence key○Accessories

Vertical setting
optics bench＊Can be combined with various
motorized/manual stages

https://www.synos.jp

ND ﬁlter●Option・2× intermediate lens port
MS-OP011-RL2
・1/2× intermediate lens port
MS-OP011-RLH
・Coaxial epi-illumination port
MS-OP011-CEP

Coaxial epi-illumination system

24

SYNERGY OPTOSYSTEMS CO., LTD