TempStick Probe

TempStick Probe is a miniaturized temperature recorderwith cable and rigid metal probe of verious lengths, with awide range of monitoring applications especially in thefood, environment and pharmaceutical industry, forcontinuous monitoring in fridges and fridge cells according
to ISO 9001 and 12830 standards, HACCP, FDA, BRC
ecc. certiﬁcation. Its reduced size and no need of external
power supply make the TempStick system extremely
versatile. The data logger can be supplied with calibration
certiﬁcate with Accredia (Italian reference center, NISTequivalent) traceability.

Applications

Food & Beverages

Environment

The battery is User replaceable.
The rigid metal probe is made with stainless steel AISI
316L, food grade, and is available in diﬀerent types:
3 cm length, 4 mm diameter, rounded tip for immersion(ordering code 03)
8 cm length, 4 mm diameter, rounded tip forimmersion (ordering code 04)
3 cm length, 4 mm diameter, pointed tip forpenetration (ordering code 05)
8 cm length, 4 mm diameter, pointed tip forpenetration (ordering code 06)

Cold Chain

Pharmaceutical

Healthcare

Main features
Mission programming and reading using a PC through the TecnoStick Interface or SRI and FLI portableinterfaces
Reduced size for easy use for every need
Quick response timeStand-by function: start the logger and then, using the metallic clip, set it to a stand by status. In this way
you can send many pre-programmed loggers to the sender of the products without providing him with thesoftware and the interface. Before sending the goods he will just have to remove the clip to start theacquisitions
Works with StickLog Pro, TS Manager and FridgeLog Z software

LaboratoriesMedicalWarehouses

Plus
High accuracy and precision
Product's core temperature controlled directly
Easy to use with SRI and FLI portable interfaces
Compliant with HACCP and EN12830 regulations
MKT formula implemented, for quick data analysis, also on SRI and FLI
Provided with calibration certiﬁcate Accredia (NIST equivalent) traceable (on demand)
Long life battery
User replaceable battery

Logistic&Transport

Industrial

The system
The basic system, for transport, is made up by:
TempStick data logger
StickLog Pro software
TecnoStick Interface or SRI
The FridgeLog Z system, for continuous monitoring, is made up by:
TempStick data logger
FridgeLog Z sotware
FLI interface
It may be used for warehouses, environment, fridges validation:
TempStick data logger
TS Manager software
TecnoStick Interface

Accessories
TS Manager
StickLog Pro
FridgeLog Z
TecnoStick Interface
SRIFLI
TempStick metal clip

Technical speciﬁcations
Dimensions

54 X 32 X 11 (mm)Probe dimensions

Cable 300 l X 4 Ø (mm) or longer - Probe 30 or 80 l X 4 Ø (mm) pointed or rounded

Weight14 gTemperature range-30 °C ÷ +60 °C

Sensor temperature range-40 °C ÷ +80 °C

Standard calibration points (temperature)-30/0/20/60°C

Extra calibration points (temperature)

Within the range -40 °C ÷ +80 °C

Temperature resolution

0,03 °C

Temperature accuracy± 1 °C without certiﬁcate / ± 0,25 °C with Accredia traceable certiﬁcate (valid in thecalibration range)

Memory (n. of acquisitions)

2.730

Acquisition step

From 1 every minute to 1 every 255 minutes

Battery type

3.0V CR2032 lithium, user replaceable (2017 IATA DGR: PI970 Section II, < 4 cells)

Battery life

Up to 10 years or 3 millions of acquisitions

Protection degree

IP63

Software&Mobile App

StickLog Pro, FridgeLog Z, TS Manager

Accessories

TecnoStick Interface, SRI, FLI