Infinity XE 60 Series (All Models)Installation Guide

Copyright © Peak Scientific Instruments Ltd 2021 — Infinity XE 60 Series Installation Guide Rev 1 PN IG-0031 RSID 4127 EN —09/11/2021

Change History
RevCommentNameDate

How to use this Installation Guide
This document is intended to be used only as a Quick Start Guide, to facilitate safeand correct installation and initial configuration of your Peak Scientific gas generator.
For more extensive user operation directions, we recommend you download the User
Manual from the website.

Please visit www.peakscientific.com/downloads todownload the full User Manual for your gas generator.
Thank you for selecting Peak Scientific to meet your laboratory gas generation needs,
and should you require any further assistance or support please do not hesitate tocontact Peak Scientific or Peak Partner from which you purchased your generator.

Page 2

Safety Notices
Peak Scientific Instruments cannot anticipate every possible circumstance which may
represent a potential hazard. The warnings detailed within this document detail themost known potential hazards, but by definition cannot be all inclusive. If the user
employs an operating procedure, item of equipment or a method of working which
is not specifically recommended by Peak Scientific, the user must ensure that theequipment will not be damaged or become hazardous to persons or property.

Symbols
This document uses the following symbols to highlight specific areas important to thesafe and proper use of the Generator

WARNINGA WARNING notice denotes a hazard. It calls attention to an operating procedure,
process or similar, which if not correctly performed or adhered to, could cause
personal injury or in the worst case death. Do not proceed beyond a WARNING
notice until the indicated conditions are fully understood or met.

CAUTIONA CAUTION notice denotes a hazard. It calls attention to an operating procedure,
process or similar, which if not correctly performed or adhered to, could cause
damage to the Generator or the Application. Do not proceed beyond a CAUTION
notice until the indicated conditions are fully understood or met.
Caution, risk of electric shock. Ensure power to the Generator has been removed
before proceeding.

Safety Notice to UsersWARNINGWARNINGWARNING

Page 3

These instructions must be read thoroughly and understood before
installation and operation of your Peak Generator. Use of thegenerator in a manner not specified by Peak Scientific MAY impair theSAFETY provided by the equipment.
When handling, operating or carrying out any maintenance, personnel
must employ safe engineering practices and observe all relevant local
health and safety requirements and regulations. The attention of UK
users is drawn to the Health and Safety at Work Act 1974, and theInstitute of Electrical Engineers regulations.
If the equipment is used in a manner not specified by themanufacturer, the protection provided by the equipment may beimpaired.

Fittings Kit

x1x11/4” Tee Fitting 02-1048

US Mains Cable 230V C19 6-15 04-1024

x18mm Hex Key 00-0007

x1

UK Mains Cable C19 04-1025x3m*
12mm OD x 10mm ID PE 00-1267

x1

Shut-Off Valve 02-4471

x1

EU Mains Cable C19 04-1026

x3mx1*x16mm OD x 4mm ID PE 00-1282

1/4” - 12mm Push Fit Fitting02-5801

US Mains Cable 110V C19 5-15 04-1035

x1x1x1JP Mains Cable 100V C19 04-1004
Page 416mm Push Fit Fitting 3301060

1/4” - 6mm Push Fit Fitting02-4600* x2 with Dry Air option fitted

1.
WARNING0°0°
Unpack the generator from the shipping crate and position on a flat surface, indesired area.
An Infinity XE 6010 has been used for the purposes of this Installation Guide.
WARNING Ensure all inlets and outlets are connected to correct sources and applications.
All Connections should only be carried out by trained personnel. Generator must beswitched off and unplugged prior to any connection , cleaning or maintenance operations.

2.

3301060

02-4600
Select the 16mm Push Fit Fitting 3301060 from the fittings kit and attach to the Air
Inlet port at the rear of the generator.
Then fit the 6mm Push-Fit Fitting 02-4600 to the Drain port.
Page 5

3.
02-4471

02-5801

02-1048

Select the Tee Fitting 02-1048 and attach this to the Nitrogen port.
Next, attach the Shut-Off Valve 02-4471 to the vertical female port on the Tee Fitting,ensuring PTFE tape is used for tight fit.
Finally attach the 12mm Push-Fit Fitting 02-5801 from the fittings kit to the remaining
female port on the Tee Fitting, again ensuring PTFE tape is used.
For units without Dry Air option fitted please proceed to step 5.

4. (Dry Air option only)

02-5801

If the generator is fitted with the optional Dry Air output, select the 12mm Push-Fit
Fitting 02-5801 from the fittings kit and attach it to the ‘Air’ port on the rear of thegenerator.

Page 65.

Fit
appropriate
tubing00-1267

00-1282
Connect the 6mm PE Tubing 00-1282 to the ¼”- 6mm Push Fit Fitting 02-4600.
Fit the other end of the tubing to a suitable drain connection or container. The container
MUST NOT have an airtight seal as water and air will be expelled periodically under a slightpressure.
Attach 3m of 12mm PE Tubing 00-1267 to the fitting attached to the fittings on theNitrogen port and connect to the gas inlet of your instrument.
Lastly connect appropriate 16mm tubing (not supplied) to the fitting on the Air Inlet port
and connect the other end to your air supply (ISO 8573-1:2010 Class [1.4.1]).
For units without Dry Air option fitted please proceed to step 7.

6. (Dry Air option only)

00-1267

Attach 3m of 12mm PE tubing 00-1267 from the fittings kit to the fitting on the Dry Air
output of the generator. Connect the other end to the Air inlet of your instrument.

Page 7

7.

04-1004/04-1024/04-1025/04-1026/04-1035

Select the appropriate mains cable from the fittings kit and plug the IEC 60320/C19
cable into the mains input at the rear of the generator.

8.
UKEUUS110V
JP100VUS230VCAUTION

Plug the mains cable into the corresponding 100/110/230VAC 50/60Hz single phasepower supply.
CAUTION ensure the correct cable has been selected for your electrical supply.
Consult the Electrical Requirements section if you are unsure of the requirementsof your generator.
Page 8

9.

Switch the generator power ON using the switch on the rear panel.

10.

The inlet pressure graph on the external HMI of the generator will gradually increase
from 0Psi to the delivered inlet pressure as the system pressurises

Page 911.

Remove the side panel, and set the desired Nitrogen output pressure using
the internal pressure regulator located near the front of the generator. View
the external outlet pressure reading on the HMI when setting the pressure.
For units without Dry Air option fitted please replace the side panel after theregulator has been set and proceed to step 13.

12. (Dry Air option only)

Set the desired Air output pressure using the internal pressure regulator locatednear the rear of the generator. View the external outlet pressure reading on the HMI
when setting the pressure. Replace the side panel after the regulator has been set.
Page 10

13.

/CONGRATULATIONSYour Peak Scientific gas generator is now fully installed, operational and ready tosupply gas on demand to your instrument.

Page 11

General Notes
Electrical requirements for 230V generators are 195V - 253V 50/60Hz. For 195V- 219V a 06-3200 Dual Tap Transformer is required.

1.

Electrical requirement is 100-230V for all XE60 models.

2.

Electrical Requirements
Generator
Infinity XE 60*0230

Voltage
110100Frequency
5060











Plug TypeUK, EU, US 230V

Please familiarise yourself with the full User Manual available at www.peakscientific.
com/downloads. This includes all other technical specifications, operating
instructions, service requirements, contact details and trouble shooting.

Page 12Go Online or Complete and Return
We know that registering any of your recently purchased products is not
the first thing on your mind- but it is very important to both of us. Not all
warranties are alike and Peak Scientific stand out against other gas suppliers as
we offer a comprehensive, quick response, on-site warranty. This means thatin the very unlikely case that your gas generator develops a fault we have rapid
support teams on-hand around the world who are able to come to your lab andget you back up and running in no time.
Register for your comprehensive 12 month on-site warranty with ease online atwww.peakscientific.com/register.
Alternatively, you can send the completed form to Peak Scientific by post or
email at warranty@peakscientific.com.

Go Online or Complete and Return
You can register for your FREE 12 month Warranty with ease online atwww.peakscientific/protected.
Alternatively, you can send the completed form to Peak Scientific by post or
email at warranty@peakscientific.com.

Product Warranty Registration
Contact name
Email addressCompanyAddress
City/town
PostcodeCountryTelephoneGenerator serial #Model typeInstallation date
Do you still use analternative gas solution i.e.
cylinders or bulk liquid?

Yes

What gas requirements do
you have in your lab?

Hydrogen

NoNitrogen

Zero Air

Extend your cover with
Peak Scientific offer comprehensive gas generator after sales support
packages. Peak [Protected] aftercare support can guarantee an on-site
response within 72 hours*, genuine parts from our ISO9001 approved factory
and a 95% first-time fix rate. See our enclosed Peak [Protected] leaflet for
further information.
Important!
You have 1 month to register your Peak Scientific product from the date of installation. Onceregistered the warranty will be honoured for a period of 12 months. If you wish to defer the installationof your generator, you must notify Peak Scientific immediately by emailing warranty@peakscientific.
com. For generators that remain unregistered after 1 month from the shipment date, the warranty will
be considered active from the date of factory dispatch.
* Complete Plan only

Important!
You have 1 month to register your Peak Scientific product from the date ofinstallation. Once registered the warranty will be honoured for a period of 12months. If you wish to defer the installation of your generator, you must notify
Peak Scientific immediately by emailing warranty@peakscientific.com. Forgenerators that remain unregistered after 1 month from the shipment date, thewarranty will be considered active from the date of factory dispatch.
Page 13

Page 14

Page 15

Please visit www.peakscientific.com/downloads todownload the full User Manual for your gas generator.

Peak Scientific has highly trained, fully certified Field Service Engineers located in over 20 countries
across every continent around the world. This allows us to provide an industry-leading rapid
response service to our customers. With [Peak Protected], your laboratory’s productivity becomes
our top priority.
To discuss Peak Protected generator cover and payment options speak to your local Peak
Representative or for further information contact: protected@peakscientific.com.

Peak Scientific
Fountain Crescent
Inchinnan Business Park
Inchinnan
PA4 9RE
Scotland, UK
Tel: +44 141 812 8100
Fax: +44 141 812 8200
For further information on any of ourgenerator products please contact
marketing@peakscientific.comCopyright © Peak Scientific Instruments Ltd 2021 — Infinity XE 60 Series Installation Guide Rev 1 PN IG-0031 RSID 4127 EN —09/11/2021