ESCORT iMINI USB Pdf
CRYOPAK iMINI USB Pdf

Simple and Smart

User Guide
Ver. 3.0

iMINI USB pdf User Guide

Simple and SmartContents
1.

Introduction................................................................................................................................................... 32.

Size and Dimensions...................................................................................................................................... 33.

Safety Instructions......................................................................................................................................... 4

4.

Product Code ................................................................................................................................................. 55.

Software ........................................................................................................................................................ 5

6.

Specifications ................................................................................................................................................ 6

7.

iMINI USB pdf Features ................................................................................................................................. 7

8.

iMINI USB pdf minimum system requirements .............................................................................................. 8

9.

Accessories / Connection ............................................................................................................................... 8

10.

Files format ............................................................................................................................................... 9

11.

LEDs.......................................................................................................................................................... 9

12.

How did logger stop? ............................................................................................................................... 11

13.

Sample PDF............................................................................................................................................. 11

14.

Alarm Settings Options ........................................................................................................................... 15

15.

Important Notes ...................................................................................................................................... 15

16.

iMINI USB pdf not detected ..................................................................................................................... 16

17.

Quick User Guide......................................................................................................................................17

18.

Warranty Information .............................................................................................................................. 19

19.

Recycling of electronic devices ................................................................................................................ 20

20.

Contact Details ........................................................................................................................................ 20

iMINI USB pdf User Manual version 3.0

Page 2

iMINI USB pdf User Guide

Simple and Smart

1. Introduction
Cryopak Verification Technologies, Inc. (CVT) offers a new advanced plug and play USB temperature data logger.
The device is a true plug and play: it does not require any software to download the data. Data can be viewed anddownloaded as PDF, CSV & TXT on PC, MAC, and ANDROID.


Economical single use data loggerReport output in .pdf, .csv & .txt format, no custom software required

Smaller than a credit card; lays flatDurable lightweight; weighs less than 13 gramsEasy programming with Console Plus software or pre-programmed

LED indicators provide status alert of logger

The new iMINI USB pdf is now available for storage and transit temperature monitoring solutions. It is completely
food safe with IP rating 65. The iMINI USB pdf is an internal temperature sensor data logger with four alarm
conditions, start and stop buttons, and much more…

2. Size and Dimensions

67.15 mm2.64”

36.9 mm
1.45”

12.15 mm
0.48”

35 mm
1.38”

Weight: 13 grams (with protective sleeve)

iMINI USB pdf User Manual version 3.0

Page 3

iMINI USB pdf User Guide

Simple and Smart

3. Safety Instructions
In no circumstance should heavy force be applied to your iMINI USB pdf logger. Applying heavy
force to any part of your iMINI USB pdf could result in logger malfunction and/or injury.

iMINI USB pdf logger should only be operated within the parameters specified in the technicaldata discussed within this user manual. A failure to follow these instructions could result in you
iMINI USB pdf logger malfunctioning and cause permanent damage to the unit.

iMINI USB pdf loggers are vulnerable to damage when contact with water takes place. Pleasecheck your iMINI USB pdf loggers IP rating before placing your iMINI USB pdf logger in suchconditions.

Your iMINI USB pdf logger is not to be subjected to a naked flame. Exposures to such conditions
may damage the logger or possibly cause an explosion of the battery.

Never operate your iMINI USB pdf logger with a low battery.

Do not attempt to repair or modify any part of this logger. Such actions will result in a loss of
warranty. All repairs are to be made by an official Cryopak Verification Technology service center.

The iMINI USB pdf logger has a one trip warranty. For more information, refer page 16.

Battery life depends greatly on the temperature environment and recording frequency of youriMINI USB pdf logger.

iMINI USB pdf User Manual version 3.0

Page 4

iMINI USB pdf User Guide

Simple and Smart

4. Product CodeThe product code for CVT products is represented in a modular fashion to allow distributors and customers thepossibility of easily recognizing and ordering a variety of different products.
For ease of communication, the dashes within the product code may be omitted.

The product codes of iMINI USB pdf Data Loggers have the following format:[XX] - [YY] - [T] - [M] - [Options] (e.g. MS-ST-S-8-P)


The initial letters “XX” indicates product family as detailed below:
MS





Single trip logger (usually with internal temperature sensor only)

The next field “T” identifies the temperature measurement range:
Code

Sensor location

Temperature range

S

Internal-40°C to +80°C“M” identifies the Memory capacity of the logger. The following options are available:8New generation Intelligent MINI plug and play

The field “YY” represents the sensor combination code with following options:ST

iMINI USB pdf ESCORT / CRYOPAK

8K memory

Logger can store 7,928 data points“Options” identifies if the logger has built-in traceable calibration certificate

For pharma compliant units “P” is added with the product code i.e. MS-ST-S-8-P (21 CFR part 11 compliant)
The specification sheets detail in more depth which logger models are available with which memory options. All
iMINI USB pdf data loggers are fitted with five LED’s.

5. Software
All new products from CVT will be supported by ConsolePlus. Software has built-in auto update feature. When a
new release is available it will prompt user to upgrade to newest version.
Setup file can be downloaded from www.cryopak.com
For installation and program/download information refer to the ConsolePlus user Manual
No special software is required to view and download the data from iMINI USB pdf logger. It is a true plug and playlogger, hence no device driver needed. Once plugged in, a computer recognizes the logger as a mass storage
device named “iMINI USB”. It contains four files:(Serial no.).cvt, (serial no.).csv, (serial no.).pdf and (serial no.).txt. More information on the file formats is available
in section 10

iMINI USB pdf User Manual version 3.0

Page 5

iMINI USB pdf User Guide

Simple and Smart

6. Specifications
Description

Specifications

Hardware iMINI USB pdf Temperature Data Logger (Single Use)
MS-ST-S-8
Product code
MS-ST-S-8-P (pharma compliant)File format *.PDF, *.CVT, *.CSV and *.TXT
Start option Push button and/or Timer
Stop button Yes, Option to enable or disable STOP button
Stop Option Push STOP button and/or Timer
Temperature Range -40 °C to +80 °C
Resolution 0.1 °C
0.5 °C from -40 °C to -10 °C
Temperature accuracy
0.3 °C from -10 °C to +80 °C
Bookmark Yes, Brief press of START button when logging (maximum 8)Interface USB, Built inMemory 8K, 7928 readings
Sensor 1, Internal NTC
Alarm 4 thresholds: HIGHHIGH; HIGH; LOW; LOWLOWRed LEDs flash - HIGH / HIGHHIGH
Alarm thresholds with LED visualBlue LEDs flash - LOW / LOWLOW
Smaller than a credit card 68mm x 37mm x 5mm (without plastic
Size (Dimensions)sleeve) 81mm x 52mm x 5mm (with plastic sleeve)
Weight Less than 14g
IP Rating IP 67, Completely Food Safe
Accuracy Certificate
Other certification
Shelf LifeTime Accuracy
Calibration (points)
Program Option
Alarm Configuration
Time above/below specification
Temperature Units °F and °C
Start delay
Program interval
Warranty

iMINI USB pdf User Manual version 3.0

In the memory (for MS-ST-S-8-P)
CE, ISO9001:2008, RoHS
1 Year
±1 hour per year
3-point Calibration (for MS-ST-S-8-P) or on customer request
Pre-programmed andUser Programmable (special conditions apply)Consecutive and Non-Consecutive
By LED blink counts (refer to user manual)
Yes
Yes, 0 minute to 30 days 23 hours 59 minutes
1 Sec - 17 hours
1 Trip / 1 Year

Page 6

iMINI USB pdf User Guide

Simple and Smart

7. iMINI USB pdf Features
Internal temperature sensor, single use plug and play PDF data loggerProduct label specifying product code and serial number plus bar coded serial number according to EAN
128.

Five LED’s displays temperature above or below specification, depending on the set alarm conditions.
LOWLOW, LOW, OK, HIGH and HIGHHIGHStart the logger via START button immediately or with delay and pre-program to start at a specific date
and time.



Logging interval as low as 5 second per reading and as high as 17 hours

LED Window

USB Port

START Button

Battery

STOP Button
Top LabelSensorProduct Label

Rear LabeliMINI USB pdf User Manual version 3.0

Page 7

iMINI USB pdf User Guide

Simple and Smart

8. iMINI USB pdf minimum system requirements
Operating System
Windows 2000
- Service Pack 4
Windows XP- Service Pack 2
Windows 7
Windows 8, 8.1
Windows 10
Android- Ver 4.1.1
MAC OSX-Ver 10.7.5
Linux-Ubuntu Ver 12.04

PDF Software
Adobe Reader
Ver 9.0 or above

Foxit Reader
Ver 5.4.3 or above

Preview
N/A

Ver 9.0 or above

Ver 3.0 or above

N/A

Ver 11.0 or above
Ver 11.0 or above
Ver 11.0 or above
Ver 10.6.1 or above

Ver 5.4.3 or above
Ver 5.4.3 or above
Ver 5.4.3 or above
Ver 1.7.0.0809 or above

N/A
N/A
N/A
N/A

N/A

N/A

5.5 or above

Ver 9.5.5 or above*

N/A

N/A

*with Linux OS, error messages as below appears which does not affect the data or pdf

9. Accessories / Connection
No accessories are required. The logger is direct plug and play.

iMINI USB pdf User Manual version 3.0

Page 8

iMINI USB pdf User Guide

10.

Simple and Smart

Files format

iMINI USB pdf logger when connected to computer or tablet, appears as mass storage device. A drive letter is
automatically assigned and drive name is by default “iMiniUSB”. Double click to open the contents of the folder.
There are four files in the folder “iMiniUSB” and all files are named with serial number (prefix is removed). For
example if the connected logger has serial number MS-CD-123-4567, the files will be as follows:
D1234567.CSV
D1234567.CVT
D1234567.PDF
D1234567.TXT
Drive nameFiles

iMiniUSB
Drive letter:(automaticallyassigned)

D1234567.CSV
D1234567.CVT
D1234567.PDF
D1234567.TXT

11.

File size(full memory)
249KB
17KB
14KB
249KB

File format
CSV
CVT
PDF
TXT

Application
Excel
ConsolePlus
PDF reader
Notepad / textedit

Pages
159-170
N/A
2
159-170

LEDs

The iMINI USB pdf has five LEDs: two RED LEDs, two BLUE LEDs and one GREEN LED.
These LEDs will function during programming, downloading and when any of four alarms are triggered.

1
2
3
4
5

11.1. During connection with PC, MAC, ANDROID
All 5 LEDs will start in running flash mode quickly in the direction of LL  L  OK  H  HH  H  OK  L  LL
every time it is connected to USB.
11.2. During programming
All 5 LEDs will flash quickly (once), when the iMINI USB pdf is programmed. Two BLUE LEDs sometimes are notviewable in all lighted environments.

iMINI USB pdf User Manual version 3.0

Page 9

iMINI USB pdf User Guide

Simple and Smart

11.3. After program (not started)
When the logger is programmed but not yet started with START button and has no delay in start, GREEN LED will
blink (once) every 8 seconds.
11.4. During start (when configured - no delay)
By pressing and holding START button for 3-5 seconds, HH (RED LED) will flash once and then all LEDs will flash
in running light mode, quickly from LL  L  OK  H  HH.
11.5. During start (when configured - with delay)
By pressing and holding START button for 3-5 seconds, HH (RED LED) will flash once and then all LEDs will flash
in running light mode, quickly from LL  L  OK  H  HH.
When there is delayed start, GREEN LED will blink each second. When delay time is met, all LEDs will flash inrunning light mode once again, quickly from LL  L  OK  H  HH. This time logger will actual starts
recording.
11.6. During recording / logging
Immediately after start, GREEN LED will start blinking twice every 4 sec intervals. When any alarm is activated,
respective LED(s) will blink twice every 4 sec intervals. If there is no alarm activated during entire logging period
GREEN LED will blink until the end of the logging trip,
11.7. Alarm
The iMINI USB pdf supports four alarm conditions, if temperature goes out of specification thencorresponding LEDs will start flashing during logging or when stopped.
Alarm

Logger windowLED#

LED color

HIGH HIGH

HH

LED 1

RED LED

HIGH

H

LED 2

RED LED

LED 3

GREEN LED

OK
LOW

L

LED 4

BLUE LED

LOW LOWLL

LED 5BLUE LED

Alarms LEDs will stop flashing after 7 days after the logger is connected to USB port to download or view the data.
In ideal mode if alarm LEDs need to be viewed again, a brief press of START button will enable ‘alarm mode’ fornext 7 days. After 7 days’ alarm LED’s will go in sleep mode again. This feature helps preserve battery life.
11.8. Stopped mode
When the logger is stopped after finishing the trip, and no alarm was activated, the GREEN LED will blink (once)
every 4 seconds.
When the logger is stopped after finishing the trip, and an alarm was activated, then the respective LED(s) will
blink (once) every 4 seconds.
11.9. During stopiMINI USB pdf User Manual version 3.0

Page 10

iMINI USB pdf User Guide

Simple and Smart

If STOP button is enabled during configuration /programming, by pressing and holding STOP button for 3-5
seconds or when the trip is completed then all LEDs will flash in running light mode, quickly from HH  H  OK 
L  LL (reverse direction of start LEDs)
11.10. During download
When PDF file is opened, LL (BLUE) LED will constantly lit, till the file is open
When CSV file is opened, LL & L (both BLUE) LEDs will constantly lit, till the file is open
When TXT file is opened, LL (BLUE) LED will constantly lit, till the file is open
When CVT file is opened, no LED have lit or blink

12.

How did logger stop?

Easily represented by the logger when a PDF is download, a letter is added at the end of stop time, which signifies
how did the logger actually stopped recording.
This information is available in generated PDF on 1st page under device specification

Text
[B]
[R]
[T]
[F]
[C]
[X]

Description
Logger is stopped by STOP button (if enabled)
Logger is stopped by readings (less than maximum readings)
Logger is stopped by timer (to be implemented)
Logger is stopped after recording full memory
Logger is plugged in with computer/phone/tablet & stopped recording
Logger was RESET by unknown reason – flat battery / moisture / damage

13. Sample PDF
Page 1iMINI USB pdf User Manual version 3.0

Page 11

iMINI USB pdf User GuideiMINI USB pdf User Manual version 3.0

Simple and Smart

Page 12

iMINI USB pdf User GuidePage 2

1.

Product code MS-ST-S-8 (non-pharma compliant)Without 21 CFR part11 compliance & NIST traceable certificate

2.

Product code MS-ST-S-8-P (pharma compliant)With 21 CFR part11 compliance & NIST traceable certificate

iMINI USB pdf User Manual version 3.0

Simple and Smart

Page 13

iMINI USB pdf User GuideiMINI USB pdf User Manual version 3.0

Simple and Smart

Page 14

iMINI USB pdf User Guide

14.

Simple and Smart

Alarm Settings Options

HH (HIGH HIGH) enable
Triggers an alarm when the temperature is greater than the HIGH HIGH specification that has been recorded. The
HH LED will start blinking.
H (HIGH) enable
Triggers an alarm when the temperature is greater than the HIGH specification that has been recorded. The H
LED will start blinking
L (LOW) enable
Triggers an alarm when the temperature is lower than the LOW specification that has been recorded. The L LED
will start blinking
LL (LOW LOW) enable
Triggers raise an alarm when the temperature is lower than the LOW LOW specification that has been recorded.
The LL LED will start blinking
Consecutive delay before alarm (1 to 254)
A preset alarm will trigger when conditions goes out of specification: ‘consecutive alarm delay’ has been met. For
example, if the value is set ‘5’ in this field, as soon as 5 consecutive readings are out of specification, logger will
trigger an alarm, if the consecutive readings are 4 or less, there will be no alarm.
Total delay before alarm (1 to 254)
The preset alarm will trigger when conditions goes out of specification: when ‘total alarm delay’ has been met. For
example, if the value is set ‘5’ in this field, as soon as 5 readings (in any order) are out of specification, the logger
will trigger an alarm. For example, if 2 readings were above or below spec and then readings come within thespecifications followed by another 3 readings being out of specification then, once the total of 5 readings areabove or below spec, there will be an alarm.

15.

Important Notes



The protective sleeve must not be removed until
the download, if removed during the trip logger
may encounter issues.



The iMINI USB pdf logger will finish the logging
trip when connected to USB. The device will notrecord furtherIf ‘corrupt pdf’ message appears, re-connect
the device to USB and re-open the PDF

The maximum number of bookmarks is 8If during connection, the logger LEDs keep
continuously flashing in running light mode, then disconnect the logger and reconnect. If the problem
persists, short the battery terminal for 1 second.

iMINI USB pdf User Manual version 3.0

Page 15

iMINI USB pdf User Guide

16.

Simple and SmartiMINI USB pdf not detected

Problem Description:
On some computers with WINDOWS 7 (64 bit) operating systemwith i7 core processor, when iMINI USB pdf logger is connected,
error appears“USB Device Not Recognized”

In the event of computer is not able to detect the iMINI USB pdf data logger

Solution 1:
The USB3 ports work perfectly with Cryopak monitoring devices on Windows 8, 8.1 and 10 – We recommend
upgrading your OS to one of the aforementioned operating systems. Windows are offering free upgrades toWindows 10 (available till 29th July 2016)

Solution 2:
Try to locate USB 2.0 port on thecomputer and plug iMINI USB pdf
logger on USB 2.0 port
Identification USB 3.0 vs USB 2.0
The color: Blue means USB 3.0. Yes
but this standard is quickly
disappearing as USB 3.0 becomesmainstream.
The # of pins: A USB 3.0 port has 9 total pins. 5 in the front and thenormal 4 in the back. You can shine a light into the port and if you see
the 5 gold pins near the opening and 4 gold pins that are morerecessed, you've got a USB 3.0 port. (see picture below: Left is USB 3.0
and right is USB 2.0)
The symbol: SS stands for "Super Speed" and this also means you havea USB 3.0 port.

Solution 3:
Restart the computer and boot into BIOS, pay close attention to the first screen that appears. Look for a
message that indicates which key you need to press in order to access the BIOS. That key is likely to be:

iMINI USB pdf User Manual version 3.0

Page 16

iMINI USB pdf User Guide

Simple and Smart A Function key(such as F1, F2, or F3)
 The Esc key
 The Delete key
On some PCs, entering the BIOS
may require pressing another key
or a combination of keys. Keepyour eyes open for an on-screen
message that says somethinglike: 'Press F1 to enter setup'
 'BIOS settings: Esc'
 'Setup = Del'
 'System configuration: F2'
If you don't catch it the first time,
simply try again. Your computer's
documentation may also tell you
which key to press.

 Once in BIOS, scroll / browse to ‘advanced’
settings and locate ‘Legacy USB Support:’
If it is set to ‘Auto’ or ‘Disabled’, set it to
ENABLE If the option to disable USB 3.0 port isavailable – disable it Press F10 key or save the settings.
 Reboot the computer
Now iMINI USB pdf logger should be detected by computer after the reboot.

17.

Quick User GuideiMINI USB pdf User Manual version 3.0

Page 17

Simple and SmartiMINI USB pdf User Guide
Quick User Guide –Guía de Inicio Rápido-

Mini Guide de l’Utilisateur –Thank you for purchasing Cryopak’s
iMINI USB pdf Temperature Data Logger

Gracias por adquirir el Registardor de TemperaturaiMINI USB pdf fabricado por la firma Cryopak

Merci d’avoir acheté l’acquisiteur detempérature iMINI USB pdf de Cryopak A GREEN LED blinks once every 8 sec.,
indicating Logger is in sleep mode El indicador verde se apagará y prenderá cada 8
segundos indicando que el registrador esta en estado
de espera.
Para Programar el Registrador (Registradores que noestán programados se pueden ordenar) Launch Consoleplus 1.15/1.16
 Conecte el Registrador a un Puerto USB de lacomputadora.
 Vaya al Tabulador de programación
 Ingrese la configuración requerida
 Clic el botón de programar
 Espere que todas las 5 luces indicadoras se prendan y
se apaguen
 Desconecté el Registrador
Para comenzar los Registradores que estan
programados Oprima y mantenga el botón de comenzar de 3 a 5
segundos HH el indicador de color rojo se prenderá y apagará una
sola vez, a continuación todas los indicadores de luces
se encenderán y se apagarán de acuerdo al siguiente
order: LL → L → OK → H → HH
 El indicador Verde se apagará dos veces cada 4
segundos indicando que el Registrador a comenzado agrabar.
Para señalar un MARCADOR (maximo 8)
 Oprima y mantenga el botón de comenzar de 3 a 5
segundos.
 Todas las luces indicadoras se prenderán y apagaránuna sola vez y al mismo tiempo, indicando que elmarcador a sido colocado.
Para Parar el Registrador (si el botón de apagar esta
prendido)
 Oprima y mantenga el botón de entre 3 a 5 segundos LL el indicador (azul) se prenderá y apagará una solavez, inmediatamente despues todas las luces
indicadoras se prenden en serie rapidamente en elsiguiente order: HH → H → OK → L → LL
 El Registrador dejará de grabar
Para Bajar la información grabada (para leer el archivo
se require el programa de lectura PDF)
 Retire las manjas de plastico
 Conecté el Registrador a un puerto USB
 La unidad de almacenamiento massivo de datos de‘iMiniUSB’ aparecerá Todos los archivos con información estarán disponibles
 Copie el PDF a lugar deseado Doble clic para abrirlo

 Une lumière LED VERTE clignote à chaque 8 secondes
pour indiquer que l’acquisiteur est en mode sommeil

To PROGRAM Logger (un-programmed
loggers available upon request)
 Launch ConsolePlus 1.15 /1.16
 Plug the logger into USB port
 Go to ‘Program’ tab Enter the required settings
 Click ‘program’ button
 Wait for all 5 LEDs to flash
 Remove the logger
To START programmed loggers
 Press & hold START button for 3 to 5 sec.
 HH LED (RED) will blink once & thenall LEDs will quickly flash in running line
LL  L  OK  H  HH
 GREEN LED blinks twice every 4 sec.
Indicating logger has started recording
To Place a BOOKMARK (max 8)
 Press & hold START button for 3 to 5 sec.
 All LEDs will flash together once,
indicating a bookmark has been placed
To STOP logger (if STOP button enabled) Press & hold STOP button for 3 to 5 sec.
 LL LED (BLUE) will blink once & thenall LEDs will quickly flash in running line
HH  H  OK  L  LL
 Logger stopped logging
To DOWNLOAD data
(Requires PDF reader software) Remove plastic sleeve Plug the logger into USB port
 Mass storage drive ‘iMiniUSB’ appears All data files are available
 Copy PDF to desired location
 Double click to open

iMINI USB pdf User Manual version 3.0

Pour PROGRAMMER l’acquisiteur (acquisiteur non
programmé disponibles sur demande) Démarrer ConsolePlus 1.15 /1.16
 Insérer l’acquisiteur dans un port USB
 Aller à l’onglet « Programmer » Entrer les paramètres désirés Cliquer sur le bouton « Programmer » Attendre que les 5 lumières LED clignotent
 Retirer l’acquisiteur
Pour DÉMARRER les acquisiteurs préprogrammés Appuyer et maintenir le bouton START de 3 à 5secondes La lumière LED HH (rouge) clignotera une fois puis
toutes les lumières LED clignoteront rapidement une àla suite de l’autre LL  L  OK  H  HH
 La lumière LED VERTE clignotera deux fois à toutes les
4 secondes, indiquant ainsi que l’acquisiteur acommencé à enregistrer
Pour insérer un MARQUE-PAGE (max 8)
 Appuyer et maintenir le bouton START de 3 à 5secondes Toutes les lumières LED clignoteront en même temps
une fois, indiquant ainsi qu’un marque-page a été inséré
Pour ARRÊTER l’acquisiteur (seulement si le bouton
STOP est activé)
 Appuyer et maintenir le bouton STOP de 3 à 5 secondes
 La lumière LED LL (bleu) clignotera une fois puis toutes
les lumières LED clignoteront rapidement une à la suitede l’autre HH  H  OK  L  LL
 L’acquisiteur cessera d’enregistrer
Pour TÉLÉCHARGER les données (requière un logiciel delecture PDF)
 Retirer de l’enveloppe de plastique Insérer l’acquisiteur dans un port USB
 Le périphérique de stockage « iMiniUSB » apparaîtra
 Tous les fichiers de données sont disponibles
 Copier le fichier PDF dans le fichier désiré
 Double cliquer pour ouvrir le fichier

Guida rapida –Grazie per aver acquistato Cryopak vi
iMINI USB pdf Temperatura Logger un LED verde lampeggia una volta ogni 8 sec.,
indicando Logger è in modalità sleep
Per programmare Logger (non programmato
logger disponibili su richiesta ) Lancio ConsolePlus 1.15 / 1.16
 Collegare lo strumento alla porta USB
 Vai alla scheda ' Programma ' Immettere le impostazioni necessarie
 Fare clic sul pulsante ' programma ' Attendere che tutti e 5 i LED a lampeggiare Rimuovere il logger
A START logger programmati
 Premere e tenere premuto il tasto START per 3 e 5sec.
 LED HH (rosso) lampeggia una volta e poi tutti i
LED lampeggiano rapidamente in linea che
LL  L  OK  H  HH
 LED verde lampeggia due volte ogni 4 sec.
Indicando logger ha iniziato la registrazione
Collocare una PREFERITI (max 8)
 Premere e tenere premuto il tasto START per 3 e 5sec.
 Tutti i LED lampeggiano insieme una volta ,indicando un segnalibro è stato posto
A ARRESTO logger (se tasto STOP abilitato) Premere e tenere premuto il pulsante di STOP per 3e 5 sec.
 LED LL (blu) lampeggia una volta e poi tutti i LED
lampeggiano rapidamente in linea che
HH  H  OK  L  LL
Logger arrestato registrazione
Per scaricare i dati (richiede software PDF reader) Rimuovere il manicotto di plastica Collegare lo strumento alla porta USB
 Apparirà Memoria di massa di auto 'iMiniUSB'
 Tutti i file di dati sono disponibili
 Copia PDF nella posizione desiderata
 Fare doppio clic per aprire

Page 18

iMINI USB pdf User Guide

18.

Warranty Information(Clause from CVT Terms and Conditions)

9.1 CVT warranties in respect of the Product are:(a) The warranties and conditions implied by the Uniform Commercial Code Article 2, Part 3 312(Warranty of Title and Against
Infringement), 313-2(b) (sale by description) and 314-2, 315 (merchantable quality); or(b) A specific warranty or Product specification included in the Order Confirmation or on the CVT website www.cryopak.com.
9.2 The warranty in sub clause 9.1 is to the exclusion of all other warranties, conditions and liabilities wherever expressed or
implied and whether arising in contract, court or by statute or otherwise.
9.3 No representation in relation to the Products shall be binding on CVT unless in writing and signed by CVT or one of its
Directors, or included in either of the warranties detailed in clause 9.1.
9.4 CVT will not be liable for any personal injury, property damage, consequential or contingent loss or damage caused
through the negligence or otherwise of CVT, its servants or agents or rising out of the goods being defective or otherwise notin accordance with any warranty given.
10. Specific Warranty Provisions
10.1 Warranty is understood as “return to base”. You will be responsible for freight/taxes and duties back to CVT; CVT will pay
the return freight back to You.
10.2 Product Warranty is twenty-four months’ ex works on all products, except for the Single Trip loggers where the warranty
term is limited to a single trip to be performed within the 12-month period. This does not include batteries.
10.3 The warranty does not cover:(a) RH calibration later than 1 year after production, or if the unit has been subject to environmental conditions outside thosespecified.
(b) Willful damage, mistreatment, misuse or abuse of the goods.
(c) Loss or damage caused by ingress of moisture unless ordered with immersion rating;(d) Batteries(e) Circumstances where the unit has been modified from EVT specifications.
(f) Exposure of the logger to temperatures outside the specified storage temperature or operating temperature.
(g) Exposure to environmental conditions outside the specified conditions.
(h) Circumstances where the printed circuit board has been removed or tampered with.
10.4 In the case of a warranty claim CVT will repair the Product or, at its option supply an equivalent replacement.
10.5 In some circumstances, where a specific return is authorized, CVT, may permit the use of its Federal Express accountnumber for returning Product. Such permission is valid ONLY for that authorized shipment. CVT will not accept any freightcharges forgoods that have been returned without its express permission.
10.6 You may return Product to CVT for accuracy testing, if there are reasonable doubts as to the Products overall accuracy.
10.7 If a Product is returned within the warranty period, a Traceability Certification will be performed. Should the logger read
within the specifications, a Traceability Certificate will be issued and You will be charged with the cost and the cost of return
freight. If the logger does not pass the certification, i.e.: the logger is at fault, the logger will be replaced or repaired. The newor repaired logger will be issued with a Traceability Certificate free of charge.
10.8 If the Product is returned outside the warranty period, a Traceability Certificate will be issued and charged regardless ofthe outcome of the test.
10.9 CVT will not pay for any performance tests undertaken by any outside organization, without prior approval.
iMINI USB pdf User Manual version 3.0

Page 19

iMINI USB pdf User Guide

19.

Recycling of electronic devices

Some parts in products from Cryopak Verification Technologies consist of recyclable materials, but others should
not be disposed of in household waste. To avoid pollution, we kindly ask you to adhere to national policies andregulations concerning waste disposal and recycling. iMINI USB pdf data loggers must be returned to your
distributor for disposal (European Standard EN 50419:2005).

20.

Contact Details

If you require further information regarding Cryopak Verification Technologies products, please contact usat:USA

Mailing Address:
Office:Phone:
OrdersFax:
Email:PO Box 309, Buchanan, VA 24066
120 Parkway Drive, Buchanan, VA 24066+1-540-254-1433+1-732-346-9200 Ext.150+1-540-254-2433
techsupport@cryopak.com

CANADA

Mailing Address:

11000 Parkway Blvd.
Anjou, QC, Canada, H1J 1R6, Canada+1-514-324-4720+1-514-207-1691+1-514-324-9623
support@cryopak.com
d.delia@cryopak.com

Phone:
Cell:
Fax:
Email:FRANCE

Mailing Address:
Phone:
Email:

1 Rue Berthelot, Z.I. La Maine
76150 Maromme, France+33 (0)2 32 82 59 65
ordersEU@cryopak.com

You can find additional information regarding all of our products on our website:www.cryopak.com

iMINI USB pdf User Manual version 3.0

Page 20