Infinity XE 5010-5040
Single nitrogen source for LC-MS
Part Number : See Reverse

Service Kit : See Reverse

Your local gas generation partner

Description
Producing between 10 and 520 L/min of high purity nitrogengas, Infinity XE 50 series gas generators can comfortably supply
multiple LC-MS instruments with LC-MS grade nitrogen. With theInfinity XE 5010-5040 you can now effectively meet the demands ofnumerous LC-MS instruments simultaneously. With no moving parts
or mechanical sound, our Infinity range is easy to service and requires
minimal maintenance.
Peak Scientific's Infinity XE series generators have been engineered
to provide nitrogen to laboratories that utilize an external sourceof compressed air, which meets a minimum quality grade of ISO
8573-1:2010 Class 1.4.1. Where required, Peak can provide assistanceon external compressors & pre/post filtration as part of delivering acomplete solution for your needs.

Applications)

LC-MS)Infinity XE 5010-5040 models have the capability tosupport other applications. Please speak to your local Peakrepresentative for more information.

500.00mm

Key Features

500.00mm

995.00mm

INFINITY XE 5010-5040 Dimensions

Copyright 2020 Peak Scientific - Rev - Infinity XE 5010-5040 Data sheet | Rev 5 | RSID 2369 | EN - 10/07/20)

Flow rates ranging from 10-520 L/min*)

LC-MS grade N2)

Built upon tried & tested membrane technology)

Compact, space saving design)

Engineered and built in the UK to accredited standards, CE/FCC certified)

30% smaller than predecessor units (Infinity 50X0))

User friendly, easy to operate & maintain)

Low noise and minimal vibration)

No moving parts, highly robust solution* Dependent on inlet pressure, inlet/outlet flow & inlet air quality

Technical Specifications

501050205030

5040

Flow Rate Min - Max (L/min)*10-13020-26030-39040-520Purity N2 Min - Max *LC-MS grade Nitrogen

Max Outlet Pressure

100psi default (user adjustable)

Inlet Air Quality Requirement

ISO 8573-1:2010 (1:4:1)

Inlet Air Pressure Requirement

125psi (8.6 bar) minimum

Voltage100-230 VAC +/- 10%

Frequency

50/60 Hz

Current

0.4A @ 100V / 0.174A @ 230V

Input Connection

C19 (13A)

Power Cord (Supplied)

120V NEMA 5-15 (US), 230V BS 1363 (UK), 230V NEMA 6-15 (US). 230V CEE 7 (EU), 100V JIS C 8303 (Japan)

Power Consumption

40 Watts

Operating Temperature

5°C - 30°C / 41°F - 86°F

Pollution Degree2Installation Category

II

Dimensions mm (inches) W x D x H

500 (19.7) x 500 (19.7) x 995 (39.2)

Weight kg (lbs)

50kg (111lbs)

54kg (120lbs)

59kg (131lbs)

63kg (139lbs)

Shipping weight kg (lbs)

66kg (146lbs)

71kg (157lbs)

77kg (170lbs)

82kg (181lbs)

3300573

3300574

3300575*Dependant on inlet pressure, inlet/outlet flow and Inlet Air QualityOrdering Information
Part Number

3300572

Annual Service Kit
Fixed Price Preventative Maintenance

Visit: www.peakscientific.com/ordering

Complete/Standard Maintenance Plan

Peak Scientific gas generators define the benchmark in reliability, convenience and performance in laboratories around theworld, and come backed by a 12 month warranty. Beyond this period however you can ensure that you’re investment continues
to be [Protected] by our comprehensive generator care cover.
Our world-class aftercare support packages deliver a program of scheduled preventative maintenance whilst giving you thereassurance of instant access to worldwide technical support and priority on-site response in the untimely event of a breakdown.

Peak Scientific UK
Tel: +44 (0)141 812 8100
Fax: +44 (0)141 812 8200
Peak Scientific North America
Tel: +1 866 647 1649
Fax: +1 978 608 9503

Product Certifications
Peak Scientific’s Quality
Management System
conforms to: ISO:9001:2008

For a full list of our worldwide
office locations, please visit:
Web: www.peakscientific.com
Email: marketing@peakscientific.com

Copyright 2020 Peak Scientific - Rev - Infinity XE 5010-5040 Data sheet | Rev 5 | RSID 2369 | EN - 10/07/20