MODEL 1207F ACCELEROMETERSPECIFICATIONS

DIMENSIONSMEMS DC Response
Durable Cable
Low Cost
Reliable Performance

The Model 1207F accelerometer is a small,compact uniaxial device designed for vehicle impact androad testing. Its mechanical overload stops provide highshock protection in rugged applications. Featuring ranges
from 50g to 1000g and frequency response to 3000 Hz, thissensor is easily mounted in hard to get places on vehicles
under test.

FEATURESMEMS Sensing Element±50g to ±1000g Dynamic Range
2-10 VDC Excitation
0 to 50°C Temperature Compensation±40 mV Zero Acceleration Output
Rugged Strain Relief
Mechanical Overload Stops

APPLICATIONSSENSOR SOLUTIONS /// Model 1207F Rev A

Crash Testing
Impact testing
Off-Road Testing
Road Testing

07/2017

Page 1

MODEL 1207F ACCELEROMETERPERFORMANCE SPECIFICATIONS
All values are typical at +24°C, 80Hz and 10Vdc excitation unless otherwise stated. Measurement Specialties reserves the right to update
and change these specifications without notice.

1Parameters
DYNAMIC
Range (g)Sensitivity (mV/g) 1
Frequency Response (Hz)Phase Response
Natural Frequency (Hz)Non-Linearity (%FSO)
Transverse Sensitivity (%)
Damping Ratio
Shock Limit (g)±50
20-8000-300
2000±1<30.65000

ELECTRICAL
Zero Acceleration Output (mV)
Excitation Voltage (Vdc)
Input Resistance (Ω)
Output Resistance (Ω)
Insulation Resistance (MΩ)
Residual Noise (µV RMS)
Ground Isolation±40
2 to 102400-5000
2400-5000>100
10Isolated from Mounting Surface

ENVIRONMENTAL
Thermal Zero Shift (%FSO/°C)
Thermal Sensitivity Shift (%/°C)
Operating Temperature (°C)
Compensated Temperature (°C)
Storage Temperature (°C)
Humidity±0.05-0.20 ±0.05-40 to +100
-20 to +85-40 to +100
Epoxy Sealed, IP65

PHYSICAL
Case Material
Cable
Weight (grams)
Mounting
Mounting TorqueAnodized AluminumPFA Insulated Leads, Braided Shield, Polyurethane Jacket
3.0
2x #4 or M3 Screws
4.0 lb-in (0.5 N-m)

Notes
±100
0.90-1500
0-500
3000±1<30.6
5000±200
0.70-1800
0-500
4000±1<30.6
5000±500
0.40-2700
0-10006000±1<30.3
5000±1000
0.150-40000-20007000±1<30.15000@10Vdc Excitation±1dB<±5

Differential@50VdcMaximum

Output is ratiometric to excitation voltage

Calibration supplied:

CS-FREQ-0100

NIST Traceable Amplitude Calibration from 20Hz to ±1dB Frequency Response Limit

Supplied accessories:

2x #4-40 (1/4” length) Socket Head Cap Screw and Washer

Optional accessories:

121
140A

Three Channel DC Differential Amplifier
Auto-zero Inline Amplifier

The information in this sheet has been carefully reviewed and is believed to be accurate; however, no responsibility is assumed for inaccuracies. Furthermore, thisinformation does not convey to the purchaser of such devices any license under the patent rights to the manufacturer. Measurement Specialties, Inc. reserves theright to make changes without further notice to any product herein. Measurement Specialties, Inc. makes no warranty, representation or guarantee regarding thesuitability of its product for any particular purpose, nor does Measurement Specialties, Inc. assume any liability arising out of the application or use of any product or
circuit and specifically disclaims any and all liability, including without limitation consequential or incidental damages. Typical parameters can and do vary in different
applications. All operating parameters must be validated for each customer application by customer’s technical experts. Measurement Specialties, Inc. does notconvey any license under its patent rights nor the rights of others.

SENSOR SOLUTIONS /// Model 1207F Rev A

07/2017

Page 2

MODEL 1207F ACCELEROMETER

ORDERING INFORMATIONPART NUMBERING

Model Number+Range+Cable Length

1207F-XXXX-ZZZ
ll_____Cable (360 is 360 inches)
l___________Range (0050 is 50g)
Example: 1207F-0050-360
Model 1207F, 50g, 360” (30ft) CableASIANORTH AMERICAEUROPE

Measurement Specialties, Inc.,a TE Connectivity Company
Tel: 800-522-6752
Email: customercare.hmpt@te.com

MEAS France SAS
a TE Connectivity Company
Tel: 800-440-5100
Email: customercare.lcsb@te.com

Measurement Specialties (China), Ltd.,a TE Connectivity Company
Tel: 0400-820-6015
Email: customercare.shzn@te.com

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to hereinmight be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

All Rights Reserved.

SENSOR SOLUTIONS /// Model 1207F Rev A

07/2017

Page 3