FPH_dat-sw_en_3607-2149-22_v0102_cover.indd 1

Data Sheet | 01.02

Test & Measurement

R&S®Spectrum Rider FPH
Handheld Spectrum
Analyzer
Specifications

02.11.2015 17:15:14

Version 01.02, November 2015

CONTENTS
Definitions ....................................................................................................................................................................... 3
Frequency ......................................................................................................................................................................................... 4
Sweep time........................................................................................................................................................................................ 4
Bandwidths ........................................................................................................................................................................................ 4
Level.................................................................................................................................................................................................. 5
Trigger functions ................................................................................................................................................................................ 6
Inputs and outputs ............................................................................................................................................................................. 6
General data...................................................................................................................................................................................... 7
R&S®FPH-K19 channel power meter ................................................................................................................................................. 9R&S®FPH-K29 pulse measurements with power sensor .................................................................................................................... 9
R&S®HA-Z350 log periodic OEM antenna 700 MHz to 4 GHz ............................................................................................................ 9

Ordering information .................................................................................................................................................... 10
Options ............................................................................................................................................................................................ 10
Accessories ..................................................................................................................................................................................... 10
R&S®NRP-Zxx power sensors supported by the R&S®Spectrum Rider ........................................................................................... 11
Service options ................................................................................................................................................................................ 112

Rohde & Schwarz R&S®Spectrum Rider FPH Handheld Spectrum Analyzer

Version 01.02, November 2015

Definitions
GeneralProduct data applies under the following conditions:Three hours storage at ambient temperature followed by 30 minutes warm-up operation
Specified environmental conditions met
Recommended calibration interval adhered to
All internal automatic adjustments performed, if applicableSpecifications with limits
Represent warranted product performance by means of a range of values for the specified parameter. These specifications aremarked with limiting symbols such as <, ≤, >, ≥, ±, or descriptions such as maximum, limit of, minimum. Compliance is ensured by
testing or is derived from the design. Test limits are narrowed by guard bands to take into account measurement uncertainties, drift
and aging, if applicable.

Specifications without limits
Represent warranted product performance for the specified parameter. These specifications are not specially marked and represent
values with no or negligible deviations from the given value (e.g. dimensions or resolution of a setting parameter). Compliance isensured by design.
Typical data (typ.)Characterizes product performance by means of representative information for the given parameter. When marked with <, > or as arange, it represents the performance met by approximately 80 % of the instruments at production time. Otherwise, it represents themean value.
Nominal values (nom.)
Characterize product performance by means of a representative value for the given parameter (e.g. nominal impedance). In contrast to
typical data, a statistical evaluation does not take place and the parameter is not tested during production.
Measured values (meas.)
Characterize expected product performance by means of measurement results gained from individual samples.
Uncertainties
Represent limits of measurement uncertainty for a given measurand. Uncertainty is defined with a coverage factor of 2 and has been
calculated in line with the rules of the Guide to the Expression of Uncertainty in Measurement (GUM), taking into accountenvironmental conditions, aging, wear and tear.

Device settings and GUI parameters are indicated as follows: “parameter: value”.

Typical data as well as nominal and measured values are not warranted by Rohde & Schwarz.

Rohde & Schwarz R&S®Spectrum Rider FPH Handheld Spectrum Analyzer

3Version 01.02, November 2015

Frequency
Frequency range

R&S®Spectrum Rider
with R&S®FPH-B3 option installedwith R&S®FPH-B3 and R&S®FPH-B4
options installed

Frequency resolution

Reference frequency, internal
Aging per year
Temperature drift
Achievable initial calibration accuracy
Total reference uncertainty

1 Hz

0 °C to +50 °C

Frequency readout
Marker resolution
Uncertainty

Number of sweep (trace) points
Marker tuning frequency step size
Frequency counter resolution
Count uncertainty

SNR > 25 dB

Frequency spanwith R&S®FPH-B3 option installedwith R&S®FPH-B3 and R&S®FPH-B4
options installed
Span uncertainty
Spectral purity SSB phase noise
Carrier offset

5 kHz to 2 GHz
5 kHz to 3 GHz
5 kHz to 4 GHz

1 × 10–6
1 × 10–6
5 × 10–7(time since last adjustment × aging rate) +temperature drift + calibration accuracy

1 Hz±(marker frequency × reference
uncertainty + 10 % × resolution bandwidth+ ½ (span/(sweep points – 1) + 1 Hz)
711
span/710
0.1 Hz±(frequency × reference uncertainty +½ (last digit))
0 Hz, 10 Hz to 2 GHz
0 Hz, 10 Hz to 3 GHz
0 Hz, 10 Hz to 4 GHz
1 % (nom.)

30 kHz
100 kHz1 MHz

f = 500 MHz< –88 dBc (1 Hz),–95 dBc (1 Hz) (typ.)< –98 dBc (1 Hz), –105 dBc (1 Hz) (typ.)< –118 dBc (1 Hz), –125 dBc (1 Hz) (typ.)span = 0 Hz
10 Hz ≤ span ≤ 600 MHz
span > 600 MHz
span = 0 Hz
span ≥ 10 Hz

1 ms to 1000 s
20 ms to 1000 s
20 ms × span / 1600 MHz to 1000 s
1 % (nom.)
3 % (nom.)–3 dB bandwidth
1 Hz ≤ RBW ≤ 300 kHz
300 kHz < RBW ≤ 1 MHz

1 Hz to 3 MHz in 1/3 sequence< 5 % (nom.)< 10 % (nom.)< 5 (nom.), (Gaussian type filters)–3 dB bandwidth

1 Hz to 3 MHz in 1/3 sequence

Sweep time
Sweep time

Uncertainty

Bandwidths
Resolution bandwidths
Range
Bandwidth accuracy
Selectivity 60 dB:3 dB
Video filters
Range

4

Rohde & Schwarz R&S®Spectrum Rider FPH Handheld Spectrum Analyzer

Version 01.02, November 2015

Level
Display range
Maximum rated input level
DC voltage
CW RF power
Peak RF power
Max. pulse voltage
Max. pulse energy
Intermodulation
Third-order intercept (TOI)

Second-harmonic intercept (SHI)

Displayed average noise level

displayed noise floor to +30 dBm

duration < 3 spulse width 10 µs

intermodulation-free dynamic range, signal level 2 × –20 dBm, RF attenuation = 0 dB,
RF preamplifier = off
f = 1 GHz
+7 dBm (meas.)f = 2.4 GHz+10 dBm (meas.)
RF attenuation = 0 dB, RF preamplifier = off, signal level = –40 dBm
fin = 20 MHz to 1.5 GHz–60 dBc (nom.)
fin = 1.5 GHz to 2 GHz–80 dBc (nom.)
0 dB RF attenuation, termination 50 Ω, RBW = 1 kHz, VBW = 10 Hz,
sample detector, log scaling, normalized to 1 Hz
frequency
preamplifier = off
1 MHz to 10 MHz< –135 dBm, –142 dBm (typ.)10 MHz to 1 GHz< –142 dBm, –146 dBm (typ.)1 GHz to 4 GHz< –140 dBm, –144 dBm (typ.)frequency
preamplifier = on
1 MHz to 10 MHz< –150 dBm, –160 dBm (typ.)10 MHz to 3 GHz< –158 dBm, –163 dBm (typ.)3 GHz to 4 GHz< –156 dBm, –161 dBm (typ.)Immunity to interference, nominal values
Image frequencies
fin – 2 × 30.15 MHz
fin – 2 × 830.15 MHz
f < 3 GHz , fin – 2 × 830.15 MHz
f < 3 GHz, fin – 2 × 4042.65 MHzf ≥ 3 GHz, fin + 2 × 830.15 MHz
Intermediate frequencies
30.15 MHz, 830.15 MHz, 4042.65 MHz
Other interfering signals,
f ≤ 3 GHz, spurious at fin – 2021.325 MHz
signal level – RF attenuation < –30 dBm
Other interfering signals, related to local∆f ≥ 300 kHz
oscillators
f = receive frequency
Residual spurious response
input matched with 50 Ω,without input signal, RBW ≤ 30 kHz,f ≥ 3 MHz, RF attenuation = 0 dB
Level display
Logarithmic level axis
Linear level axis
Number of traces
Trace detectors
Trace functions
Setting range of reference level
Units of level axis

50 V
33 dBm (= 2 W)
36 dBm (= 4 W)
150 V
10 mWs< –70 dBc (nom.)< –70 dBc (nom.)< –70 dBc (nom.)–60 dBc (nom.)–60 dBc (nom.)< –60 dBc (nom.)< –60 dBc (nom.)< –60 dBc (nom.)< –90 dBm (nom.)

1/2/3/5/10/20/30/50/100/120/150 dB,
10 divisions
0 % to 100 %, 10 divisions
2
max. peak, min. peak, auto peak, sample,
RMS
clear/write, max. hold, min. hold, average,view–130 dBm to +30 dBm
dBm, dBmV, dBµV, V, W

Rohde & Schwarz R&S®Spectrum Rider FPH Handheld Spectrum Analyzer

5Version 01.02, November 2015

Level measurement uncertainty
Absolute level uncertainty at 100 MHz
Frequency response (+20 °C to +30 °C)
Attenuator uncertainty
Uncertainty of reference level setting
Display nonlinearity
Bandwidth switching uncertainty
Total measurement uncertainty+20 °C to +30 °C
5 kHz ≤ f < 10 MHz
10 MHz ≤ f ≤ 4 GHz< 0.3 dB
< 1.5 dB (nom.)< 1 dB< 0.3 dB
< 0.1 dB (nom.)< 0.3 dB

SNR > 16 dB, 0 dB to –50 dB,
logarithmic level display
reference: RBW = 10 kHz< 0.1 dB (nom.)
95 % confidence level, +20 °C to +30 °C,
SNR > 16 dB, 0 dB to –50 dB below reference level, RF attenuation auto
10 MHz ≤ f ≤ 4 GHz< 1.25 dB, typ. 0.5 dB

Trigger functions
Trigger
Trigger source
External trigger level thresholdlow → high transitionhigh → low transition
maximum

free run, video, external
2.4 V
0.7 V
3.0 V

100 kHz ≤ f ≤ 1 GHz
1 GHz < f ≤ 4 GHz
RF input only

50 Ω (nom.)
N female< 1.5 (nom.)< 2 (nom.)
0 dB to 40 dB in 5 dB steps

Inputs and outputs
RF input
Impedance
Connector
VSWR
Input attenuator
AF output
AF demodulation types
Connector
Output impedance
Voltage (open circuit)
External reference, external trigger
Connector
Mode
External reference
External trigger threshold6

AM and FM
3.5 mm mini jack32 Ω (nom.)
VRMS adjustable from 0 V to > 100 mV

required level
frequency
low → high transitionhigh → low transition

Rohde & Schwarz R&S®Spectrum Rider FPH Handheld Spectrum Analyzer

BNC, 50 Ω
ext. reference, ext. trigger
0 dBm
10 MHz
2.4 V
0.7 V

Version 01.02, November 2015

General data
Manual operation
Languages

Chinese, Chinese Traditional, English,
French, German, Italian, Hungarian,Japanese, Korean, Portuguese, Russian,
SpanishRemote control
Command set
LAN interface
USB
Display
Resolution
Audio
Speaker
USB interface

SCPI 1997.0
10/100BASE-T, RJ-45
mini B plug, version 2.0
WVGA, 800 × 480 pixel

number of interfaces
Mass memory
Mass memory
Data storage
Temperature

Climatic loadinginternal
on USB stick or micro SD card, ≥ 1 Gbyte
operating temperature range
storage temperature range
battery charging mode
relative humidity
IP class of protection

Mechanical resistanceVibrationShocksinusoidal
randominternal, external headphone supported
type A plug, version 2.0
2
USB stick/micro SD card (not supplied)size ≤ 32 Gbyte, USB version 1.1 or 2.0> 160 instrument settings and traces> 10000 instrument settings and traces–10 °C to +55 °C–40 °C to +70 °C
0 °C to +40 °C+25/+55 °C at 95 % relative humidity(EN 60068-2-30)
51
EN 60068-2-6, MIL-PRF-28800F class 2
EN 60068-2-64, MIL-PRF-28800F class 2
40 g shock spectrum,in line with MIL-STD-810E, method 516.4
procedure 1, MIL-PRF-28800F

Rohde & Schwarz R&S®Spectrum Rider FPH Handheld Spectrum Analyzer

7Version 01.02, November 2015

Power supply
R&S®HA-Z301 AC power supply

input specifications
output specifications
operating temperature range
storage temperature range
test markExternal DC voltage
Battery
Capacity
Voltage
Operating time with new,fully charged battery
Charging time

Life time

R&S®HA-Z306
R&S®HA-Z306
instrument switched off or charge with
R&S®HA-Z303 battery charger
instrument switched on
charging cycles

Power consumption
SafetyTest mark
EMC

Dimensions

W×H×D

WeightRecommended calibration interval

8

Rohde & Schwarz R&S®Spectrum Rider FPH Handheld Spectrum Analyzer

100 V to 240 V AC, 50 Hz / 60 Hz,
1.0 A to 0.5A
15 V, 2.67 A, max. 40 W
–30 °C to +60 °C–40 °C to +85 °C
CE, UL, PSE, TUV
14.65 V to 15.45 V
Lithium-ion battery
72 Wh
11.25 V (nom.)
8h
3.5 h
4h
> 80 % or more of its initial capacity after
300 charge/discharge
8 W (meas.)
IEC 61010-1, EN 61010-1, UL 61010-1(Third Edition),
CAN/CSA-C22.2 No. 61010-1-12
VDE, CSA, CSA-NRTL
in line with European EMC Directive
2004/108/EC including
EN 61326-1 class B (emission)
CISPR 11/EN 55011/group 1class B (emission)
EN 61326-1 table 2(immunity, industrial)
202 mm × 294 mm × 76 mm(8.0 in × 11.6 in × 3 in)
2.5 kg (5.5 lb)
1 yearVersion 01.02, November 2015

R&S®FPH-K19 channel power meter
Frequency range

Channel bandwidth
Amplitude
Unit
LimitsMeasurement range
Level measurement uncertainty
Absolute level uncertainty at 100 MHz
Frequency response (+20 °C to +30 °C)

R&S®Spectrum Rider
with R&S®FPH-B3 option installedwith R&S®FPH-B3 and R&S®FPH-B4
options installed

5 kHz to 2 GHz
5 kHz to 3 GHz
5 kHz to 4 GHz
100 kHz to 1 GHz
offset, dB relative, zeroingdBm, W
on/off, upper limit, lower limit, beep on fail–120 dBm to +30 dBm+20 °C to +30 °C
100 kHz ≤ f < 10 MHz
10 MHz ≤ f ≤ 4 GHz< 0.3 dB
< 1.5 dB (nom.)< 1.25 dB

R&S®FPH-K29 pulse measurements with power sensor
In combination with one of the power sensors R&S®NRP-Z81, -Z85 or -Z86, the R&S®Spectrum Rider supports measurements onpulsed signals. The achievable RF performance is documented in the data sheet specifications of the R&S®NRP-Z81/-Z85/-Z86 power
sensors. The list below shows which measurements are supported by the R&S®FPH-K29.
Measurements
Pulse power parameters
Peak power
Pulse top power
Average power
Base power
Minimum power
Positive overshoot
Negative overshoot
Pulse timing parameters
Pulse duration
Pulse period
Pulse start/stop time
Rise/fall time
Duty cycle

R&S®FPH-K29
●
●
●
●
●
●
●
●
●
●
●
●
●
●

R&S®HA-Z350 log periodic OEM antenna 700 MHz to 4 GHz
Frequency range
Gain
Impedance
VSWR
Connector type
Dimensions
WeightAccessories supplied

700 MHz to 4000 MHz
4 dBi (typ.)50 Ω< 1:2 (nom.)
SMA (f)
WxHxD
340 mm × 200 mm × 25 mm(13.3 in × 7.9 in × 1 in)
270 g (0.6 lb)
hardcase with foam, typical calibration data in 10 MHz steps, pistol grip with mini-tripod
function, one set of SMA toolset

Rohde & Schwarz R&S®Spectrum Rider FPH Handheld Spectrum Analyzer

9Version 01.02, November 2015

Ordering information
Designation
Type
Order No.
Handheld Spectrum Analyzer, 5 kHz to 2 GHz
R&S®Spectrum Rider FPH 1321.1111.02
Accessories supplied
Lithium-ion battery pack, USB cable, AC power supply with country specific adapters for EU, GB, US, AUS, CH, CD-ROM with
R&S®Instrument View software and documentation, quick start guide, side strap

Options
Designation
Spectrum Analyzer Frequency Upgrade 2 GHz to 3 GHz
Spectrum Analyzer Frequency Upgrade 3 GHz to 4 GHz(requires R&S®FPH-B3)
Spectrum Analyzer Preamplifier
Power Sensor Support
Channel Power Meter
Pulse Measurements with Power Sensor

Type
R&S®FPH-B3
R&S®FPH-B4

Order No.
1321.0667.02
1321.0673.02

R&S®FPH-B22
R&S®FPH-K9
R&S®FPH-K19
R&S®FPH-K29

1321.0680.02
1321.0709.02
1321.0721.02
1321.0738.02

Type
R&S®HA-Z220
R&S®FSH-Z320

Order No.
1309.6175.00
1309.6600.00

R&S®FSH-Z321

1309.6617.00

R&S®RAM
R&S®RAZ
R&S®FSH-Z38

R&S®RDL50
R&S®RBU100
R&S®RBU100
R&S®HA-Z303
R&S®HA-Z306
R&S®FSH-Z36
R&S®TS95A16

0358.5414.02
0358.5714.02
1300.7740.02
0118.2812.00
0092.6581.00
4012.5837.00
3530.6646.00
3530.6630.00
4048.9790.00
0017.6742.00
1035.1700.52
1073.8495.20
1073.8495.30
1321.1328.02
1321.1334.02
1145.5838.02
1118.6943.16

R&S®HA-Z350
R&S®HZ-15

1321.1405.02
1147.2736.02

R&S®HZ-16
R&S®HA-Z211
R&S®HA-Z210
R&S®HA-Z301

1147.2720.02
1309.6169.00
1309.6152.00
1321.1386.02

Accessories
Designation
Soft Carrying Bag
RF Cable (length: 1 m), DC to 8 GHz, armored,
N male/N female connectors
RF Cable (length: 3 m), DC to 8 GHz, armored,
N male/N female connectors
Matching Pad, 50/75 Ω, L section
Matching Pad, 50/75 Ω, series resistor 25 Ω
Matching Pad, 50/75 Ω, L section, N to BNC
Adapter N (m) – BNC (f)Adapter N (m) – N (m)Adapter N (m) – SMA (f)Adapter N (m) – 7/16 (f)Adapter N (m) – 7/16 (m)Adapter N (m) – FME (f)Adapter BNC (m) – Banana (f)Attenuator, 50 W, 20 dB, 50 Ω, DC to 6 GHz, N (f) – N (m)Attenuator, 100 W, 20 dB, 50 Ω, DC to 2 GHz, N (f) – N (m)Attenuator, 100 W, 30 dB, 50 Ω, DC to 2 GHz, N (f) – N (m)Battery Charger for R&S®HA-Z306 1
Lithium-Ion Battery Pack, 6.4 Ah
Headphones
GSM/UMTS/CDMA Antenna Magnetic Mount 850/900/1800/1900/2100
band, N connector
Logarithmic Periodic OEM Antenna, 700 MHz to 4 GHz
Compact Probe Set for E and H Near-Field Measurements,
30 MHz to 3 GHz
Preamplifier 3 GHz, 20 dB, Power Adapter 100 V to 230 V, for R&S®HZ-15
Spare USB Cable
Spare Ethernet Cable
Spare Power Supply, incl. mains plug for EU, GB,US, AUS, CH

1

The battery charger is dedicated for charging an additional battery outside the instrument.
The battery can be charged via the instrument as well.

10

Rohde & Schwarz R&S®Spectrum Rider FPH Handheld Spectrum Analyzer

Version 01.02, November 2015

R&S®NRP-Zxx power sensors supported by the R&S®Spectrum Rider 2
Designation
Type
Order No.
Universal Power Sensor, 10 MHz to 8 GHz, 100 mW, 2-path
R&S®NRP-Z211
1417.0409.02
Universal Power Sensor, 10 MHz to 18 GHz, 100 mW, 2-path
R&S®NRP-Z221
1417.0309.02
Universal Power Sensor, 10 MHz to 18 GHz, 2 W
R&S®NRP-Z22
1137.7506.02
Universal Power Sensor, 10 MHz to 18 GHz, 15 W
R&S®NRP-Z23
1137.8002.02
Universal Power Sensor, 10 MHz to 18 GHz, 30 W
R&S®NRP-Z24
1137.8502.02
Universal Power Sensor, 10 MHz to 40 GHz, 100 mW
R&S®NRP-Z41
1171.8801.02
Thermal Power Sensor, 0 Hz to 18 GHz, 100 mW
R&S®NRP-Z51
1138.0005.02
Thermal Power Sensor, 0 Hz to 33 GHz, 100 mW
R&S®NRP-Z52
1138.0505.02
Thermal Power Sensor, 0 Hz to 40 GHz, 100 mW
R&S®NRP-Z55
1138.2008.02
Thermal Power Sensor, 0 Hz to 50 GHz, 100 mW
R&S®NRP-Z56
1171.8201.02
Thermal Power Sensor, 0 Hz to 67 GHz, 100 mW
R&S®NRP-Z57
1171.8401.02
Thermal Power Sensor, 0 Hz to 110 GHz, 100 mW
R&S®NRP-Z58
1173.7031.02
Universal Power Sensor, 10 MHz to 40 GHz, 100 mW
R&S®NRP-Z61
1171.7505.02
Wideband Power Sensor, 50 MHz to 18 GHz, 100 mW
R&S®NRP-Z81
1137.9009.02
Wideband Power Sensor, 50 MHz to 40 GHz, 100 mW (2.92 mm)
R&S®NRP-Z85
1411.7501.02
Wideband Power Sensor, 50 MHz to 40 GHz, 100 mW (2.40 mm)
R&S®NRP-Z86
1417.0109.40
Wideband Power Sensor, 50 MHz to 44 GHz, 100 mW (2.40 mm)
R&S®NRP-Z86
1417.0109.44
Average Power Sensor, 9 kHz to 6 GHz, 200 mW
R&S®NRP-Z91
1168.8004.02
Average Power Sensor, 9 kHz to 6 GHz, 2 W
R&S®NRP-Z92
1171.7005.02
Three-Path Diode Power Sensors, 100 pW to 200 mW, 10 MHz to 8 GHz
R&S®NRP8S
1419.0006.02
Three-Path Diode Power Sensors, 100 pW to 200 mW, 10 MHz to 8 GHz, LAN version
R&S®NRP8SN
1419.0012.02
Three-Path Diode Power Sensors, 100 pW to 200 mW, 10 MHz to 18 GHz
R&S®NRP18S
1419.0029.02
Three-Path Diode Power Sensors, 100 pW to 200 mW, 10 MHz to 18 GHz, LAN version
R&S®NRP18SN
1419.0035.02
Three-Path Diode Power Sensors, 100 pW to 200 mW, 10 MHz to 33 GHz
R&S®NRP33S
1419.0064.02
Three-Path Diode Power Sensors, 100 pW to 200 mW, 10 MHz to 33 GHz, LAN version
R&S®NRP33SN
1419.0070.02
R&S®NRP-Zxx power sensors require the following adapter cable for operation on the R&S®Spectrum Rider
USB Adapter Cable (passive), length: 2 m (78.7 in), to connect R&S®NRP-Zxx S/SN power R&S®NRP-Z4
1146.8001.02
sensors to the R&S®Spectrum Rider
R&S®NRP power sensors require the following adapter cable for operation on the R&S®Spectrum Rider
USB Interface Cable, length: 1.5 m (59 in), to connect R&S®NRP sensors to the
R&S®NRP-ZKU
1419.0658.03
R&S®Spectrum Rider

Service options
Service options
Extended Warranty, one year
Extended Warranty, two years
Extended Warranty with Calibration Coverage, one year
Extended Warranty with Calibration Coverage, two years

R&S®WE1
R&S®WE2
R&S®CW1
R&S®CW2

Please contact
your local
Rohde & Schwarz
sales office.

Extended warranty with a term of one to four years (WE1 to WE4)
Repairs carried out during the contract term are free of charge 3. Necessary calibration and adjustments carried out during repairs arealso covered. Simply contact the forwarding agent we name; your product will be picked up free of charge and returned to you in top
condition a couple of days later.
Extended warranty with calibration (CW1 to CW4)
Enhance your extended warranty by adding calibration coverage at a package price. This package ensures that yourRohde & Schwarz product is regularly calibrated, inspected and maintained during the term of the contract. It includes all repairs 6 andcalibration at the recommended intervals as well as any calibration carried out during repairs or option upgrades.
CDMA2000® is a registered trademark of the Telecommunications Industry Association (TIA-USA).
For product brochure, see PD 5214.0482.12 and www.rohde-schwarz.com.

2

For average power measurements only.

3

Excluding defects caused by incorrect operation or handling and force majeure. Wear-and-tear parts are not included.

Rohde & Schwarz R&S®Spectrum Rider FPH Handheld Spectrum Analyzer

11Service that adds value
Worldwide
Local and personalized❙ Customized and ﬂexible❙ Uncompromising quality❙ Long-term dependability❙❙

About Rohde & Schwarz
The Rohde & Schwarz electronics group offers innovative
solutions in the following business fields: test and measurement, broadcast and media, secure communications,cybersecurity, radiomonitoring and radiolocation. Founded
more than 80 years ago, this independent company has
an extensive sales and service network and is present inmore than 70 countries. The electronics group is among
the world market leaders in its established business fields.
The company is headquartered in Munich, Germany. It
also has regional headquarters in Singapore and Columbia,
Maryland, USA, to manage its operations in these regions.
Sustainable product design❙❙ Environmental compatibility and eco-footprint❙❙ Energy efficiency and low emissions❙❙ Longevity and optimized total cost of ownership
Certified Quality Management

ISO 9001

Certified Environmental Management

ISO 14001

Rohde & Schwarz GmbH & Co. KG
www.rohde-schwarz.com
Rohde & Schwarz Training
www.training.rohde-schwarz.com

R&S® is a registered trademark of Rohde & Schwarz GmbH & Co. KG
Trade names are trademarks of the owners
PD 3607.2149.22 | Version 01.02 | November 2015 (as)R&S®Spectrum Rider FPH Handheld Spectrum Analyzer
Data without tolerance limits is not binding | Subject to change© 2015 Rohde & Schwarz GmbH & Co. KG | 81671 Munich, Germany

3607.2149.22 01.02 PDP 1 en

Regional contact❙❙ Europe, Africa, Middle East | +49 89 4129 12345
customersupport@rohde-schwarz.com❙❙ North America | 1 888 TEST RSA (1 888 837 87 72)
customer.support@rsa.rohde-schwarz.com❙❙ Latin America | +1 410 910 79 88
customersupport.la@rohde-schwarz.com❙❙ Asia Pacific | +65 65 13 04 88
customersupport.asia@rohde-schwarz.com❙❙ China | +86 800 810 82 28 | +86 400 650 58 96
customersupport.china@rohde-schwarz.com

3607214922

FPH_dat-sw_en_3607-2149-22_v0102_cover.indd 2

02.11.2015 17:15:14