7A10626000A6

Thank you for purchasing the PD-200 Electric Delivery Bed.
This instruction manual describes precautions and how to safely use
and operate the bed.
●Before using this bed, please read this instruction manual
to fully understand the safe and proper operating procedures.
●The correct operating procedures should be explained not
only to the persons using the bed but also to caregivers.
●After reading this instruction manual, store it where it can beeasily accessed for later reference.
●Users and/or carers are requested to report information about any
serious incident related to this bed that occurs within the EU to the
manufacturer and the regulating authority of the member countryof residence. 'Serious incident' means any incident that directly or
indirectly led, might have led or might leadto any the following :a) The death of a user, carer or other person.
b) The temporary or permanent serious deterioration of a user's,
carer's or other person's state of health.
c) A serious public health threat.
●The actual product purchased may differ from detailed
descriptions given in this instruction manual due to
product improvements.
●Values in the specification such as working range, dimensions,angles and mass are subject to normal manufacturing tolerances.
Besides some exemptions, display purpose words suchas“approximately”, “about” etc. are omitted.
●For convenience, the term ''patient'' indicates ''pregnant woman''in this instruction manual.
●For any queries, please feel free to contact your dealer directly.

1
2
3
4

．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．2

INTENDED USE / PART NAMES．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．3-13

SAFETY PRECAUTIONSINSTALLATION．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．14

ASSEMBLING THE BED
1. Attaching / detaching the head board．．．．．．．．．．．．．．．．．．．．．15

2. Attaching / detaching the fluid basin．．．．．．．．．．．．．．．．．．．．．．16

3. Attaching / detaching the hand grips．．．．．．．．．．．．．．．．．．．．．．17

4. Attaching / detaching the calf supports．．．．．．．．．．．．．．．．．．．．18

5. Attaching the battery unit．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．19

5 CONNECTING TO THE POWER SUPPLY
1. Power outlets．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．20

2. Using the cable storage hook．．．．．．．．．．．．．．．．．．．．．．．．．．．20．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．21
6 OPERATION OVERVIEW．．．．．．．．．．．．．．．．．．．．．．．22-23
7 USING THE HANDHELD CONTROLLER
8 OPERATION PROCEDURES
1. Using Emergency CPR (back-lowering) function
2. Retracting the leg section base．．．．．．．．．．．．．．24．．．．．．．．．．．．．．．．．．．．．．．．．．25

3. Locking the casters．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．26

4. Using the bed side-rails．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．27-28

5. Using the battery unit．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．29．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．30
9 MATTRESS
10 SUITABLE PRODUCTS
1. Suitable products table．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．31．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．32
2. Foot rest
3. Foot support bar．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．33

4. Oxygen cylinder holder．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．34
5. IV pole．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．35．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．36
10 11 BEDSIDE TABLE．．．．．．．．．．．．．．．．．．．．．．．．．．37-38
12 PRODUCT SYMBOL DEFINITIONS．．．．．．．．．．．．．．．39-41
13 REGULAR INSPECTION AND DAILY MAINTENANCE．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．42-43
14 TROUBLESHOOTING．．．．．．．．．．．．．．．．．．．．．．．．．．44
15 LONG-TERM STORAGE．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．45-50
16 SPECIFICATIONS．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．51
17 AFTER-SALES SERVICE．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．．52
18 BED DISPOSAL

1

1. Intended Use■This bed is intended to be used as a birthing bed for women who are ready to deliver and notintended for use as a general hospital bed.
※Please refer to the 「SPECIFICATION」 for the patient target groups and the intended users.

2. Part Names

p.27-28

p.17

p.21

p.22-23

p.18

p.15

p.24
p.25

p.20

p.16

p.19, 29

p.17

p.30

p.26

p.18, 32

p.35

【

】
Part of ME EQUIPMENT that in normal
use necessarily comes into physical contact
with the patient for ME EQUIPMENT toperform its function.

p.30

p.20

2

■Be sure to read these safety precautions beforehand to ensure proper usage.
●The safety precautions described here are intended to ensure safe use of the product and prevent
personal injury and property damage. Precautions are classified as "WARNING" and "CAUTION" toindicate the degree of hazard or injury that may result from improper use. These are importantsafety precautions that must be strictly observed.
Indicates actions that may result in death or serious injury (brokenbones, pressure or paralysis) if the indication is ignored.
Indicates actions that may result in light personal injury (bruises,
wounds, or cuts) or property damage if the indication is ignored.
Among items described in SAFETY PRECAUTIONS, be sure to read these notes and use the product safely
and properly. Be especially careful to check patients who might act in unexpected ways or cannot maintaintheir posture independently.
※After reading this instruction manual, store it in a place that can be easily accessed for later reference.

3

[Safety label]●Precautions requiring special attention by persons using the bed are affixed as pictured in the figure.
●If the safety label is removed or defaced, replace it with a new one obtained from your dealer.

4

■Be careful not to allow any part of the body (especially the head or neck) to enter the gap
between the side-rail and board or the mattress base (especially when raised).
●When the head or neck enters the gap, the patient
and/or user might not be able to withdraw, and there
is a risk of injury.

■Be careful not to allow any part of the body to enter the gap in the side-rail and the board.
●If the bed is operated with a body part placed in
the gap, the part may get caught, resulting in injury.
●When the head or neck enters the gap, the patient
and/or user might not be able to withdraw, and there
is a risk of injury.

■Be careful not to allow any part of the body (especially the head or neck) to enter the gap
between the side-rail and the mattress base or the mattress.
●A patient and or/user may fall off the bed through the gap between a board and side-rail and get injured.
●If a patient bends forward over a side rail, they may fall off the bed and get injured.

■When performing the height adjustment or inclination angle adjustment with the side rail
lowered, do not place legs under the side-rail.
●There is a risk that the legs may get caught, resulting in injury.

■Do not place hands (fingers) or legs between the bed frame and the mattress base during
electrical operation.
●The body part may get caught in the gap between the lowered mattress base and the bed frame or theboard, resulting in injury.

■Check the patient’s condition when adjusting the base angle or bed height.
●If the patient on the bed moves when the base angle or bed height is adjusted, the patient may fall off the
bed or get caught in the gap between the bed and the side-rail or the board, resulting in injury.

■Do not crawl or place your head or hands/legs under the bed.
●There is a risk of getting caught between the moving
part on the bed and the frame or the side-rail, resulting ininjury. Check to make sure that there are no people orobjects around the bed before operating the bed.

5

■Be careful not to allow any part of the body to enter the gap between the bed and wall orsurrounding obstacles.
●There is a risk that a part of the body may get caught, resulting in injury.

■Take care to prevent a patient from falling off the bed even when side-rails are raised.
●A patient may fall off the bed through the gap between the side-rails, or a board and side-rail, resulting ininjury.
●If a patient bends forward over a side-rail, they may fall off the bed, resulting in injury.

■Do not adjust the side-rails from inside the bed.
●The side-rails may drop suddenly and there
is a risk of the patient falling off the bed,
resulting in injury.

■When transporting a patient, raise the side-rails.
●There is a risk of the patient falling off the bed, resulting in injury.

■Do not sit on the side-rails, the head board, or the foot board.
●A patient could fall off the bed, or the bed or siderails could be damaged or deformed, resulting ininjury.

■When leaving the bed area or temporarily taking your eyes off the patient, checkthe mattress base angle, bed height, and the side-rail status.
●If a physician, nurse, or caregiver leaves the bed area or temporarily takes their eyes off the patient,
flatten the mattress base angle and set the bed height at the lowest position according to the patient’s
condition because there is a slight chance that the patient could fall off the bed.

■Do not allow the bed to be operated by a child aged 12 or under, or someone deemedincapable of comprehending the operation of the bed.
●If a child aged 12 or under, or someone deemed incapable of comprehending the operation (such as
persons with dementia) operates the handheld controller by themselves, there is a risk of unexpected injury
such as the body getting caught in the bed by mistake. If such persons can come into contact with the
handheld controller, unplug the power plug in order to prevent accidents caused by incorrect operation.

■Pay attention to prevent the patient from being compressed during back raising.
●Especially before birthing, if the back section base is raised excessibly, the patient might be compressed,
resulting in injury.

6■Use the bed in the correct orientation.
●Lying down on the bed with the feet at head end
and the head at the foot end induces an unnatural
posture during bed operation (base angle) which
may result in injury.

■Do not perform any angle adjustment when the patient is lying face down or on theirside (in any posture other than lying face up).
●There is a risk of injury caused by extending the joints backward.

■Use the bed according to the patient’s medical condition.
●The bed operations may adversely affect the medical conditions of patients with specific symptoms.
●Before the patient or a family member operates the bed, make sure that the physicians or nurses explain
details given in the instruction manual and proper usage to suit the patient’s medical condition.

■Do not attach a restraint belt.
●Otherwise it may cause an injury or bed damage or deformation.

■Do not stand on the bed.
●There is a risk of injury caused by the person
falling off the bed or falling down, or the bed could
be damaged or deformed. Pay close attention tochildren.

■Do not sit on the leg section base.
●The leg section base may get damaged or the bed
may tip over, causing an injury.

■Be careful not to allow fingers to enter the gap between the hip section base and the leg
section base when operating the leg section base.
●There is a risk that fingers may get caught,
resulting in injury.

■Do not spill liquid on the bed.
●Do not spill liquid on the bed’s electrical components, such as the motors and handheld controller. This
could cause an electric shock due to a short circuit or product failure. If liquid is spilt accidentally on thebed, disconnect the power plug from the power outlet and contact your dealer.

7

■Make sure that there is nothing in the IV Pole attachment holes, the hand grip attachment
holes or the accessories attachment holes when mounting the calf supports, hand grips or
optional items.
●If there are obstacles in the attachment holes, you cannot mount the hand grips, the calf supports oraccessories items, and the patient may fall off the bed or get caught in the gap, resulting in injury.

■Do not use a mobile phone or other wireless device in the vicinity of the bed.
●This could cause the bed to malfunction or affect other medical equipment.

■Do not combine the bed with products other than the suitable products specified byParamount Bed.
●For suitable products directly attached to the bed such as mattresses, use suitable products specified byParamount Bed. Using other products may create unintended gaps, cause interference between products, or
lower the product safety, which may result in injury or operational failure.

■Keep the bed away from fire.
●Do not use thermal appliances such as heaters near the bed. Otherwise, the bed may be damaged ordeformed, or even catch fire.

■When moving the bed, watch your step.
●Your leg may bump into the caster operation pedal or your foot may get caught by a caster, resulting ininjury.

■Lock the casters at all times except when moving the bed.
●The bed may move when a patient gets into or out of the bed, resulting in an unexpected injury.
●When moving the bed, operate the caster operation pedal and unlock the casters.

■Do not step on the caster operation pedal.
●There is a risk of injury due to falling or damage to the bed.

■Do not trip over the power cable or cables of the handheld controller.
●The power plug or cables may get damaged and short circuit, resulting in electric shock or fire.
●There is a risk of injury caused by the patient falling down.

■Do not run cables (such as the power cable) of the bed or other electrical appliances
under the bed.
●The cables may get caught in casters or moving parts and become damaged and short circuit, resulting inelectric shock or fire.

■Do not apply excessive force to the power plug.
●There is a risk of electric shock or a fire due to a short circuit.

8

■Do not allow dust to accumulate on the power plug pins.
●Dust on the power plug can absorb moisture andconduct electricity, causing a short circuit andpossibly resulting in electric shock or fire.
●Wipe away any dust accumulating on the surfaceof the power plug using a dry cloth.

■Always unplug the power plug before cleaning.
●There is a risk of injury due to accidental operation or electric shock due to a short circuit.

■Do not insert or remove the power plug with wet hands.
●There is a risk of electric shock or product failure
due to a short circuit.

■Do not damage cables (such as the power cable).
●Damage to the cables may cause electric shock or fire due to a short circuit.
○Be careful not to allow cables to get caught in the moving parts of the bed.
○Be careful not to place heavy objects on the cables or apply excessive force to them.
○Do not run over cables with the bed.
●Request repair (or replacement) of damaged cables.

■Do not bump or damage electrical parts such as the handheld controller.
●The bed may malfunction, resulting in injury, and there is a risk of electric shock or fire due to a shortcircuit.
●This may cause operational failure.
●Request repair (or replacement) of a damaged handheld controller.

■When disconnecting the power plug, hold the power plug body to disconnect.
●When holding onto only the power cable and pulling it to disconnect the plug from the outlet, the cable
could be damaged and there is a risk of electric shock or fire due to a short circuit.

9

■Always disconnect the power plug for the bed from the power outlet when using
electronic medical equipment.
●Using the bed in conjunction with electronic medical equipment (e.g. microwave or ultrashortwave
medical equipment) may cause operational failure or malfunction.
Check for safety considerations before using the bed in conjunction with other medical equipment.

■Do not repair or modify the product or its components.
●This may create unintended gaps or may cause abnormal operation, resulting in injury.

■Inspect the bed regularly.
●The product exhibits wear and deterioration according to the frequency and conditions of use. Inspect
for loose parts, the operation of moving parts on the bed, and the presence of damage regularly. There is arisk of unexpected injury.

■Beds damaged in accidents must be inspected and repaired before being reused.
●Beds damaged in earthquakes, fires, or floods or other incidents must be inspected and repaired by yourdealer. Otherwise, there is a risk of electric shock or fire due to short circuit or current leakage fromelectrical parts, or injury due to malfunctioning caused by deformation of the bed.

■The bed should not be used by two or more persons.
●This bed is designed for use by one person. When used by two or more persons, the bed could bedamaged, resulting injury. User's maximum weight is 180kg. If a physician, nurse, or caregiver needs to
get on the bed for a short time, confirm the following conditions before hand.
○The total weight of the person to get on the bed must not exceed 220kg.
○The mattress base has been lowered all the way.

■Do not apply more than the safe working load on the bed.
●The bed could be damaged, resulting in injury. The safe working load of this bed is 2150N (220kgf).
The safe working load is the maximum load with which the bed can be operated and is the total load of theuser's weight and incidental items including the mattress and suitable products.

■Do not use as a general hospital bed.
●This bed is not designed as a general hospital bed but designed for women who are ready to deliver.
Using as a general hospital bed may cause unexpected injury.

■Do not place the bed on the inclination which angle is 5 degree or more.
●The bed may tip over, resulting in injury.

10■Do not use the mattress when the fastener is open.
●If liquid gets inside the mattress, the urethane foam may deteriorate.

■Do not place something cold (ex. ice pack or refrigerant) on the mattress directly.
●The urethane foam may deteriorate due to condensation.

■Do not sleep face down.
●If the patient sleeps face down for a prolonged time, there is a risk of asphyxiation because the mattresscover is not breathable.

■Do not spray a spray-type insecticide directly on the bed.
●There is a risk that the solvents contained in the insecticide may damage, discolor, or dissolve casters orother resin parts of the bed. Check for damage caused by the insecticide because damaged or dissolved
parts may cause unexpected injury.

11

■When performing height or inclination angle adjustments, pay attention not to hit thewall, beam, or power outlet on the wall.
●The bed or wall may be damaged or deformed.
Make sure that the bed does not hit the wall, beam,
or power plug during height or inclination angle
adjustments.

■When setting up the bed, make sure that there is at least 10 cm clearance from the wall.
●The bed operations may damage objects locatedaround the bed, or may cause the bed to be damaged
or deformed.

■Do not get on the mattress base when it is raised.
●Excessive force on the mattress base can cause
damage or deformation.

■Check the surroundings before operating the bed.
●The bed operations may damage objects located around the bed, or may cause the bed to be damaged ordeformed.

■Make sure the stoppers of the head board are in the lock position.
●If the board is not mounted properly, the board
may suddenly disengage when a staff pushes (pulls)
the board to move the bed or a patient grabs theboard to support their body, resulting in injury
should the patient lose balance and falls.

12

■Do not move the bed by holding on to the side-rails.
●Excessive force applied to the side-rails can cause damage or deformation.

■Avoid transporting over uneven floors.
●This may cause damage or deformation to the bed.
If the bed must be moved on uneven floors, do it asslowly as possible.

■The handheld controller must be attached to the appropriate bed.
●If it is attached to an inappropriate bed, the bed could operate unexpectedly, resulting in injury or bed
failure.
●Even if the bed is of the same series, the handheld controllers may be different due to specification
changes. Before attaching the handheld controller to the bed, check that the part number and shape of thehandheld controller are the same as those of the bed.

13

■To avoid possible malfunctions, do not install the bed in the following locations.
●Locations exposed to direct sunlight●Locations near heat sources such as a stove or heater●Locations which are heavily electrically charged such as near an MRI.
●Locations that have a high oxygen levels●Locations exposed to excessive amounts of water vapor or oil vapor●Locations that tend to be exposed to high temperatures, high humidity, low temperatures or
dry conditions※The optimum ambient conditions for the bed are as follows.
Temperature 5℃ to 40℃ / Humidity 20%RH to 90%RH / Atmospheric pressure 700hPa to1060hPa.
●Locations with large amount of dust, smoke, salt, sulfur, or corrosive substances●Locations with poor ventilation●Locations exposed to vibrations or shocks (including during transportation)●Locations where the floor is not strong enough●Locations where the floor is not level●When performing height or inclination angle adjustments, pay attention not to hit the wall, beam,
or power outlet on the wall. The bed or wall may be damaged or deformed.

●When setting up the bed, make sure that there is at least 10 cm clearance from the wall. The bed
operations may damage objects located around the bed, or may cause the bed to be damaged ordeformed.

14

1.Attaching / detaching the head board●When attaching the head board, make sure that the stopper is in the lock position. Otherwise, theboard may disengage when the bed is moved, causing injury due to toppling.
●Do not attach the board to a mounting lugs other than the board attachment mounting lugs. Attachment
could become insufficient and the board could be rattled or come off, resulting in injury.
●When removing the board, do not forcefully lift it while the board is tilted. There is the risk of deformationof the board insertion parts.

■The head board can be attached or detatched.
[Attaching]①Hold the grip of the head board, lift and mount it sothat the board brackets engage with the mounting lugs
on the bed.(The board stopper is automaticallylocked.)②Lightly lift the board and make sure that it is locked.

[Detaching]①Move the board stopper to the unlock position.
②Hold the grip of the board and detach it.

15

2.Attaching / detaching the fluid basin●Open the Fluid basin receiver before storing the Fluid basin. If it is stored incorrectly, the fluid receiver may
be dropped or pinched, resulting in damage to the equipment or injury.

■The fluid basin can be attached or detached.
[Attaching]①Pull out both fluid basin receivers as shown in the
picture.
②Mount the fluid basin on the fluid basin receivers.

fluid basin receiver

fluid basin[Detaching]①Lift and detach the fluid basin.
②Push in the fluid basin receivers as shown in the
picture.

fluid basin receiver

16

3.Attaching / detaching the hand grips●Be careful not to allow any part of the body to enter the gap between the hand grip and the mattressbase, the side-rails or the optional items. When any part of the body enters the gap, the patient might
not be able to withdraw, and there is a risk of injury. Detach the hand grips except when giving birth.
■The hand grips can be used during delivery.
Mounting section[Attaching]①Direct outward the planar section of the mounting
section of the hand grip, and insert the mounting
section into the hand grip attachment hole as shown in
the picture.
②Turn the knob to fasten the hand grip.
③Lightly pull or rotate the hand grip and make surethat it is fastened.
●The right hand grip and the left hand grip are
different.
Be careful not to attach wrongly.

Accessories attachment
hole

Main frame side label

Hand gripside label[Detaching]①Turn the knob to loosen.
②Lift the hand grip as shown in the picture.

17

4.A�aching / detaching the calf supports●Fasten the lever when the calf supports are mounted in the accessories attachment holes. Otherwise,
the calf supports may drop, causing an injury.
■The calf supports can be used when examining the patient or giving birth.
Mounting section[Attaching]①Insert the mounting section into the accessories
attachment hole as shown in the picture.
②Turn the lever to fasten the calf supports.
③Lightly lift the calf supports and make sure that it
is fastened.

Main frame side label
Accessories attachment hole

Mounting section label
Accessories attachment hole

[Detaching]①Turn the lever to loosen.
②Lift the calf supports as shown in the picture.

18

5. Attaching the battery unit

●When connecting the battery cable, unplug the power plug. There is a risk of injury due toaccidental operation or electric shock due to a short circuit.
■The battery cable must be connected to the control box in order to use the battery.

①Set minus driver in the hole at the side of the control
box and slide the control box cover to open it.
②Connect the battery cable to the battery adapter ofthe control box.
③Attach the control box cover.

<After opening the control box cover>Control Box

19

1. Power outlets●Never connect more than one cable to a power outlet.

●Each Bed should be pluged into a separate,
appropriately wired, single-phase 230VAC, 10A or
greater grounded power outlet using the power plug.

2. Using the cable storage hook
●Wind the power cable onto the cable storage hook when moving the bed.

※View after detaching the head board

20

●Do not allow the patient to get on or off the bed while the bed is tilted. There is a risk of injury due
to the patient falling down. Tilting the bed must only be performed under the observation of a doctor
or nurse.

●The angle of the back section base can be
adjusted electrically from an angle of 0°to 65°.

00 to 650

●The height from the floor to the mattress basesurface can be adjusted electrically from 50cm to83cm.

50 cm to 83 cm●The head end of the bed can be lowered. The
angle can be adjusted electrically up to 12°.

00 to 120

21

●Do not allow the bed to be operated by a child aged 12 or under, or someone deemed incapable ofcomprehending the operation of the bed. If a child aged 12 or under, or someone deemed incapable ofcomprehending the operation (such as persons with dementia) operates the handheld controller bythemselves, there is a risk of unexpected injury such as the body getting caught in the bed by mistake.
If such person can come into contact with the handheld controller, unplug the power plug in order toprevent accidents caused by incorrect operation.
●The angle of the back section base, the bed height and the bed inclination angle can be smoothly
adjusted with no increments by using the handheld controller.
Pressing the button adjusts the bed, and releasing the button stops adjustment, allowing you to adjust
the bed to the desired position. Press the [↑] to raise the desired section and press [↓] button to lower
the desired section.
Power indicator lamp
●Lamp will turn on when in normal electricity condition.
●Lamp will turn off if power plug or handheld controller plugloosed or if black out and disconnection

UP and DOWN buttons●For adjustment of the angle of the back section base.

UP and DOWN buttons●For adjustment of the bed height.

UP and DOWN buttons●For adjustment of the bed inclination angle.
●This function only available in handheld controller for
caregiver. Doesn't install in handheld controller for user.

For caregiver

For user●Pressing two or more buttons simultaneously stops adjustment. If you stop adjustment bypressing two or more buttons, remove your hands from all buttons and then perform
adjustment again.

22

●Do not drop the handheld controller or pull the cable. The handheld controller may get damaged ordeformed, causing it to operate incorrectly. Request repair (or replacement) of a damaged handheld
controller.

●The handheld controller must be properly set in the storage position or hung on the side-rail withthe buttons facing outwards using a hook. Otherwise, when operating the bed or side-rails, buttons on
the handheld controller could be pushed and the bed could be moved accidentally, or the handheld
controller could be damaged or deformed.

<Handheld controller storage position>●Store the handheld controller in the handheld
controller storage position inside the bed side-rail.

23

1. Using Emergency CPR (back-lowering) function●During the emergency CPR operation, make sure that the patient's head, arms, feet or other parts ofthe body have not entered the gap between the back section base and the frame. There is a risk ofthem getting caught, resulting in injury.
■In case of an emergency or a power failure, the raised back section base can be quickly returned toflat status.

<Emergency CPR procedure>①Place one hand on the side of the back section
base and pull the emergency CPR lever (orange).
Levers are located at both sides of the bed under theback section base and either lever can be used.
②Keep pulling the lever until the back section basereturns to flat status. (If you release the emergency
CPR lever halfway through, the back-lowering
operation is stopped.)●Depending on the patient's weight, theback section base may not be lowered. Insuch a case, lightly push down the backsection base.

Back section base

<Operation after emergency CPR>
Back-raising and back-lowering can be operatedusing the buttons on the handheld controller in power
on status.
Emergency CPR lever (orange)24

2. Retracting the leg section base●Do not sit on the leg section base. Otherwise, the leg section base
may get damaged or the bed may tip over, causing injury.
●Be careful not to allow fingers to enter the gap between the hip
section base and the leg section base when operating the leg section
base. There is a risk that fingers may get caught, resulting in injury.
●When operating the leg section base, make sure that it is locked.
Otherwise, the leg section base may move unexpectedly, causing
injury.

■The leg section base can be retracted during delivery.
[Retracting the leg section base]①Remove the foot mattress from the leg section
base.
②Lift the leg section retraction lever to unlock, andpush the leg section base inward.
③Lightly pull the leg section base and make surethat it is locked.

Leg section baseLift
Leg section retraction lever

Push to retract[Extracting the leg section base]①Lift the leg section retraction lever to unlock, andpull the leg section base outward.
②Lightly push the leg section base and make surethat it is locked.
③Place the foot mattress on the leg section base.

Pull to extract

Bed condition

Extracted position

25

Retracted position3. Locking the casters●Lock the casters at all times except when moving the bed. The bed may move when a patient gets
into or out of the bed, resulting in an unexpected injury.

●Do not attempt to forcibly move the bed when the casters are locked (fixed), as doing so will cause
malfunctions.
<Total locking system>●The casters can be set to either of the positions below by operating the caster operation pedal on thefoot end.

①Locking the four wheels(Lock position)
Depress the red pedal to lockswiveling and rotation of thefour wheels.

②Unlocking the four wheels(Free position)
Returning the pedal to thehorizontal position unlocks thefour wheels, allowing themswivel and rotate.

③Straight-line movement(Steering position)
Depress the green pedal to lockswiveling of the caster (A) (onewheel only) in a straightforward direction. This makes
it easier for the bed to be
moved forward in a straightline.

●Inspect the casters regularly. Depending on the frequency of use and the environment, casters
will exhibit wear and deterioration. (High-temperature and high-humidity conditions may
accelerate the deterioration.)26

4. Using the bed side-rails●There are a total of two side-rails, one side-rail on the left and another on the right.
●Side-rails prevent the patient from falling off the bed. The patient can also hold onto the side-rail
when giving birth.

Side-rail position when it is raised.

①②

Side-rail position when it is lowered.
[Raising the side-rails]①Lift the side-rail and push in until it clicks into
place.
②Pull the side-rail toward you to make sure thatthe side-rail is locked.

[Lowering the side-rails]①Use one hand to hold the grip of the side-rail.
②Use the other hand to pull the operation levertoward you and, while supporting the side-rail,
lower the side-rail until it stops.
①②

Operation lever27

●Be careful not to allow any part of the body (especially the head or neck) to enter the gap between
the side-rail and the mattress base or the mattress. When the head or neck enters the gap, the patient
might not be able to withdraw, and there is a risk of injury.
●Be especially careful to check patients who might act in unexpected ways or cannot maintain their
posture independently.
●Do not adjust the side-rails from inside the bed. The side-rails may drop suddenly and there is arisk of injury caused by a patient falling off the bed.
●Do not let the patient lean against the side-rail while it is raised. The patient may lose balance andfall off the bed, resulting in injury.
●Do not place legs on the side-rail while it is lowered. The bed could turn over, resulting injury.
●When raising side-rails, make sure that the side-rails are locked. If the side-rails are not completely
locked, they may move unexpectedly, resulting in injury.
●Take care to prevent the intravenous tubes or the drain tubes from getting caught. When operating
the side-rails, tubes could get caught between the side-rail and mattress and be damaged, resulting ininjury.
●When performing the height adjustment or inclination angle adjustment with the side-rail lowered,
do not place legs under the side-rail. There is a risk of getting caught, resulting in injury.
●When operating the side-rails, make sure that the patient’s hands or feet have not entered the gap
between the side-rail and mattress base. When the hands or feet enter the gap, the patient could getcaught in the gap, resulting in injury.
●When operating the side-rail with the back section base raised, operate the side-rail with caution.
Depending on the back section base angle, the side-rails may move unexpectedly and there is a riskof injury.

28

5. Using the battery unit

●Incorrect operation in disregard of the details given below will result in personal injury or damage
due to leakage, fire, explosion, electric shock, or rupturing.
○Do not disassemble the battery case or built-in battery.
○Do not incinerate the battery unit.
○Do not recharge the battery other than with the suitable control box of the bed.
○Do not use the battery unit with equipment other than the suitable bed.
○Do not subject the battery unit to impact.
○Do not recharge the battery unit or operate the bed with the bed turned upside down.
■The battery is an auxiliary power supply for operating an electrically operated bed when notconnected to a power outlet.
[Operating the bed on battery power]●The bed is powered by the battery when the power plug is notconnected to the power outlet or during a power failure.
[Recharging]●Recharging starts as soon as the bed power plug in plugged
into the power outlet.
●The battery should be charged (for more than 24 hours) immediately after purchase.

[Battery life]●The battery is not designed to be recharged after being totally discharged. Repeatedly recharging thebattery from zero charge will reduce its operating life. Always recharge the battery after operating the
bed on battery power.
●To ensure a long battery life, recharge the battery promptly.

●The battery life will vary depending on the usage frequency, usage conditions and storage
conditions. Regardless of whether the battery is used, its life will start to reduce after a few
years.

29

●Do not use the bed without the mattress. Body parts (fingers, hands or legs) may get caught in the
gap between the mattress bases, resulting in injury.
●Be sure to use suitable mattresses as specified by Paramount Bed. Using other mattresses may
create unintended gaps, cause interference between products, or lower the product safety, which may
result in injury or operational failure. The codes of suitable mattresses are as follow.
Head Mattress:PE201. Foot mattress: PE202.
●When attaching the mattresses, make sure the Velcro tapes on the mattresses align with the Velcro
tapes on the base.

●The mattress is not used for preventing bedsores. Using for prolonged periods may cause bedsores.
●Do not raise the back section base to an angle that exceeds 30°as far as possible. If an angle of
30°is exceeded, pressure, friction, and shear forces will be applied to the patient's hips and buttocks,
which may cause pressure sores. If circumstances require that the back section base be raised to anangle of greater than 30°, restrict the time of use at that angle, and monitor the patient's conditions.
■Attach and remove the mattress as explained below.
<How to attach>
1.As shown in left figure, set head side mattress while
adjusting hip bottom and fastener surface.
2.Put foot side mattress on the foot bottom, sothat the convex part of mattress stopper insert into
in-out foot bottom.
After placing foot side mattress, make sure that mattressdoes not slip out of position.
3.There are two side in new born receiver mattress. Set graycolor concave part of hip bottom and concave part of newborn receiver mattress is match.
<How to remove>
Remove by reverse the procedure.
When doing cleaning, remove hook foot side mattress if
remove mattress stopper.

30

1. Suitable products table●Use suitable products after confirming that the combination with the bed is correct. Using the bed
with the wrong combination of suitable products could result in injury or could damage the products.
The part number of each suitable product is written on the safety label and on the carton box.
●Do not use suitable products other than those specified in this instruction manual. This may create
unintended gaps or may damage or deform the product, resulting in injury.
●Make sure that there is nothing in the accessories attachment holes, when mounting the suitableproducts. If there are obstacles in the accessories attachment holes, you cannot mount the suitableproducts, and the patient may fall off the stretcher or get caught in the gap, resulting in injury.

●When suitable products may be changed due to specification changes or combinations. In
addition, some suitable products may not be listed in the table due to new release or
discontinuation of sales. For any queries on suitable products, contact Paramount Bed (seep.52 ) or your distributor.
Suitable product*1)*2)

Model No.*1)

Foot rest
Foot support bar
Oxygen cylinder holder
IV polePC-5051, PC-5054, PC-5064
PC-5052
PC-5038
KC-56A, KC-56B, EC-562

Reference
page
32
33
34
35

*1) These products may not be available in some countries or regions. For details, contact
Paramount bed (see p.52) or your distributor.
*2) In some countries or regions, some of the medical device or its accessories fall under the
category of medical devices. For details, contact Paramount Bed (see p.52) or your
distributor.

31

2. Foot rest (PC-5051, PC-5054, PC-5064)
This foot rest is designed to be attached to the bed for for supporting body of patient during delivery.

●Be careful not to allow any part of the body to enter the gap between the foot rest and the mattressbase, or hand grip. When any part of the body enters the gap, the patient might not be able towithdraw, and there is a risk of injury. Detach the foot rests except when giving birth.
●Make sure that the lever is tightened firmly. Otherwise, the foot rest may move, causing an injury.
■The foot rest can be used during delivery.
①Direct outward the planar section of the mounting section of the foot rest, and insert the mounting
section into the accessory attachment hole as shown in the figure.
②Turn the lever to fasten the foot rest.
③Lightly pull the foot rest and make sure that it is fastened.

Main frame side label

Mounting section(PC-5051)(PC-5054)(PC-5064)(PC-5051)

Mattress side label
LoosenLever

TightenAccessory attachment hole

●The right foot rest and the left foot rest are different. Be careful not to attach wrongly.
●The description in this instruction manual may differ from the actual product due tospecification changes. For any queries, please feel free to contact your dealer.

32

3. Foot support bar (PC-5052)
This foot support bar is designed to support a doctor or nurse posture.

●When using the foot support bar, pay attention not to touch the fluid basin. Otherwise, the fluidbasin may be tipped over accidentally.
■The doctor can use the foot support bar.
①Insert the attached bolts into the holes of the attachment plates of the bed and the foot support bar.
Tighten the bolts and nuts.
②Lightly pull the foot support bar and make sure that it is fastened.
Attachment plate (bed)Attachment plate (foot support bar)

●The description in this instruction manual may differ from the actual product due tospecification changes. For any queries, please feel free to contact your dealer.

33

4. Oxygen cylinder holder (PC-5038)
This oxygen cylinder holder is made for attaching to the medical device to use the oxygen cylinder.

●Make sure that the knob bolt is tightened firmly or there may be a risk of insufficient oxygen supply.
●When using the oxygen cylinder, take care to prevent it from coming into contact with any objects inthe vicinity of the bed. There is a risk of injury due to damage to a flow meter or the oxygen cylinder if
the cylinder comes into contact with another object.
■The oxygen cylinder holder can be used by hanging it on the head board.
①Hang the oxygen cylinder holder on the head board.
②Loosen the knob bolt on the oxygen cylinder holder, and then slowly insert the oxygen
cylinder into the holder from the top.
③Firmly tighten the knob bolt to fasten the oxygen cylinder.

Tighten

LoosenKnob bolt●The description in this instruction manual may differ from the actual product due tospecification changes. For any queries, please feel free to contact your dealer.

34

5. IV pole (KC-56A, KC-56B, EC-562)
This IV pole is designed to be attached to the medical device when giving an intravenous drip to a patient.

●Do not move the bed using the attached IV pole. The IV pole may break, resulting in injury.
●Do not use the IV pole as a support to stand up. The IV pole may come off or break, causing falls
which may result in injury.
●When using the IV pole, do not put a hand or arm between the mattress base or the head board andthe IV pole. The hand or arm may get caught, resulting in injury.
●Each hook's load capacity is 1 kg. Do not hang an article exceeding the load capacity because it
may damage the hook, resulting in injury.
●Length adjustment of the slide pipe should be kept within 70cm. If the slide pipe is over-extended,it may come off, resulting in injury or damage.
■The IV pole is used for hanging intravenous drip bags and similar items.
①Insert the IV pole into one of the IV pole attachment holes.
②Loosen the knob bolt on the IV pole, and then move the slide pipe up and down to adjust the length.
When the pipe has been adjusted to the desired length, tighten the knob bolt to fasten the pipe.
③Hang the intravenous bags or similar items on the IV pole.

LoosenTighten

Hook
Slide pipe

Knob bolt

IV pole attachment hole

●The description in this instruction manual may differ from the actual product due tospecification changes. For any queries, please feel free to contact your dealer.

35

●Remove the table from the bed when performing positioning operations of the bed. Otherwise, thepatient may get caught between the table and the bed, and may get injured.
■The bedside table can be used as a table for meals or other activities when on the bed.
Suitable bedside table
KF-282
PF-3100A, PF3110A
PF-3120A, PF-3130A●The description in this instruction manual may differ from the actual product due tospecification changes. For any queries, please feel free to contact your dealer.

36

■The symbols below are used for the product.
SymbolIPx4

Description

SymbolDescription●CE mark●4 casters in locked status (Lockposition)●Indoor use●4 casters in unlocked status (Freeposition)●Electrical waste●Entrapment foot warning label●See the instruction manual●Be careful not to allow fingers toenter the gap between the hip section
base and the leg section base.

●Type B equipment●Bed transportation status (steering
position)●AC input●Take care to prevent body parts fromgetting caught in the moving parts on
the bed.

●Protection code

●Take care to prevent the patient's
head from getting caught in the gap
between the side-rails and the bed.

●Model number●Do not allow any part of the body toenter the gap in the side-rail and theboard.

●Authorized representative in theEuropean community●Safe working load

●Serial number●Maximum patient weight●Product weight●Total weight including load capacity

37

■The symbols below are used for the product.

●Do not stand or sit on the side-rail.

●Take care to prevent fingers fromgetting caught by the moving parts on
the bed.

●The side-rails must be operated by acaregiver.

●Check that the side-rail is locked.

●Use side-rails after checking that they
are locked.

●Do not sit on the leg section base.

●Manufacturer*It may be called "Legal Manufacturer"or "Brand Owner" etc.
*This symbol is not applicable in the
country origin.
●Country of manufacture (made inIndonesia)●Manufacturing site
*In the country of origin, the site acts as
the manufacturer.
●Date of manufacture(YYYY-MM-DD)●This a 2D code (GS1 Data Matrix)
that encodes the UDI (Unique Device
Identifier) required by the Unique
Device Identification System designed
to adequately identify device throughdistribution and use.

●Lot number●Medical device(This product is a medical device or anaccessory for medical device in Europe,and may be such a device in somecountries and regions depending on
their regulations. An accessory for
medical device is required to be labeled
as "Medical Device" under medical
device regulations in Europe).

●Indicates the storage humidity●Indicates the storage atmosphericpressure●Indicates the storage temperature●The medical bed for adults

38

■For safety during usage, inspect the product regularly, at least once a year. The product exhibits wear
and deterioration according to the frequency and conditions of use. Inspect for loose parts, the operationof moving parts on the bed, and the presence of damage regularly. When an irregularity has been
noticed or a detailed inspection is desired, contact your dealer.
Operation details
Back raise operation (p.21
to p.23)Height and inclination
operations (p.21 to p.23)

Bed Unit

Leg section base operation(p.25)

Miscellaneous

Board

Board mounting condition(p.15)
Traveling

Casters
Caster locking operation(p.26)

Bed side-rail

Bed side-rail operation(p.27 to p.28)

Electrical part

Visual check of electricalparts

Inspection items
Proper operation can be achieved.
There must not be any strange noises.
Proper operation can be achieved.
There must not be any strange noises.
The stopper must be locked properly.
The leg section base must move smoothly when thelock is released.
There must not be any scratches and deformation at
the mattress base, frame, or any other parts.
There must not be any deformation, damage, or detachmentof bumpers, resin parts or elastic parts.

There must not be any peeling of labels, and the text
must be legible.
The board stopper must be locked properly.
There must not be any remarkable rattling.
There must not be any strange noises when the bed istraveling.
There must not be any strange vibrations when the bed
is traveling.
Proper operation can be achieved.
The operation must not be unusually difficult.
The bed must not move when the casters are locked.
The bed must move smoothly when the lock isreleased.
When in steering position, swiveling of the steering
caster must be locked.
Proper operation can be achieved.
There must not be any strange noises or remarkablerattling.
The operation must not be unusually difficult.
Cable must not get entangled in the bed.
Plugs and cables must have no scratches or damage.
There must not be any scratches and damage on thehandheld controller.

39

〈Lubrication (if there is abnormal noise)〉●Lubrication is needed if there is abnormal noise. Please feel free to contact your dealer to be
lubricated.
●Water adhesion to the bed or the mattress, or moisture may cause corrosion, abnormal noiseor contamination. If water adheres to the bed or the mattress, wipe off immediately.
〈Bed unit〉●To clean the bed, use a cloth that has been soaked in diluted detergent and wrung out well, then wipe
off any residual detergent with a cloth soaked in water and wrung out well. Finally, use a dry cloth towipe the bed to ensure it is dry.
●To wash the bed using water, pour water on the bed to wash out stains. Finally, use a dry cloth towipe the bed to ensure it is dry.
●To disinfect the bed, always dilute detergents used as specified below and follow the instructions
described in the instruction manual.
76.9% - 81.4% ethanol for disinfection
0.05% - 0.2% benzalkonium chloride (e.g. Osvan)
0.05% - 0.2% benzethonium chloride (e.g. Hiamine)
0.05% chlorhexidine gluconate (e.g. Hibitane)
0.02% - 1.0% sodium hypochlorite (e.g. Milton)●The bed should be disinfected before a different patient uses the bed.

●Do not use volatile cleaning fluids (such as paint thinner,benzene or gasoline) and cresol. Discoloration or
deterioration could result.
●Do not use disinfectant other than those specified. Thereis a risk of damage or deterioration.
●After cleaning the bed with detergent, be sure to wipe any residual detergent with a cloth soaked inwater and wrung out well. Otherwise, there is a risk of damage to the plastic parts.
●When washing the stretcher with water, do not pour water from under the base cover. There is a riskof malfunction.
●Do not wash the stretcher by a bed washer or high-pressure water. There is a risk of malfunciton.
●The product should not be cleaned with an ozone sterilizer or autocrave apparatus.

40

●Do not wash the padding. There is a risk of padding degradation.
●Do not use a mattress washer, an ozone sterilizer or autocrave apparatus. There is a risk of damage,deterioration or degradation.
●Do not leave any blood, urine, sudor or sebum on the mattress. Clean dirt frequently with neutral
detergent or soap. If left untreated, the surface material may degrade, smell or stain may result.
○The smell that is left during use may be neutralized with mouthwash.
○If the mattress is stained, wipe off thoroughly with oxydol (3% hydrogen peroxide).
○Use thrombolytic agent, if it is difficult to clean blood. In that case, try using it on
the end of the mattress firstly, because there is a risk of discoloration or deterioration.
●Do not use volatile cleaning fluids (such as paint thinner, benzene or gasoline) and cresol.
Discoloration or deterioration could result.
●Do not use disinfectant other than those specified. There is a risk of damage or deterioration.
〈Mattress unit〉■When the mattress gets dirty or the persons who use the mattress change, be sure to clean anddisinfect both surfaces of the mattress in accordance with the procedures specified in steps ① to ④
below. At this time, take care not to bend the mattress.
①Remove any soiling (blood, excreta, etc).
②Wipe the mattress using a cloth that has been soaked in commercially available soap or neutral
detergent diluted with water, and wrung out well.
③ Disinfect the mattress by using disinfectant.
0.05% - 0.2% benzalkonium chloride (e.g. Osvan)
0.05% - 0.2% benzethonium chloride (e.g. Hiamine)
0.05% chlorhexidine gluconate (e.g. Hibitane)
0.1%-1.0% sodium hypochlorite (e.g. Purelox)④Leave the mattress to dry out naturally on the stretcher or in another location. However, do not dry inthe sun.
●Clean and disinfect the mattress regularly.
●Do not wash the mattress.
●Take care not to soak the padding.

●Do not unzip the fastener and do not remove the cover when cleaning and disinfecting the mattress.
Do not unzip the fastener while the mattress is in use. The fastener is provided so that the cover canbe repaired or replaced if it is damaged.
●The mattress should not be cleaned by an ozone sterilizer or autoclave apparatus.

41

■Problems may not always indicate faults. Check the following once again before requesting repair.
■If the problem persists after taking corrective actions, stop using the bed immediately, disconnect thepower plug from the outlet, and contact your dealer for repair.
■Bed damaged in earthquakes, fires, floods or other incidents must be inspected and repaired by yourdealer. Otherwise, there is a risk of electric shock or fire due to short circuit or current leakage fromelectrical parts, or injury due to malfunctioning caused by deformation of the bed.

Corrective action

Reference
page●Plug the power plug into theoutlet.

20●Unplug the power plug from
●The power plug is plugged into the outlet and plug it in again.
the outlet but the bed still does not ●Plug another electricalmove.
device into the outlet to checkthat power is available.

20●Check if the power plug isplugged into the outlet.

●Plug the power plug into theoutlet.

20●The POWER Lamp on
the handheld controller is
OFF.

●Unplug the power plug from
●The power plug is plugged into the outlet and plug it in again.
the outlet but the POWER lamp on ●Plug another electricalthe handheld controller is OFF.
device into the outlet to checkthat power is available.

20●The bed cannot be
positioned correctly.

●Check that there is nothing in
the moving parts on the bed.

-

●The bed cannot be
moved.

●Check if the casters are locked. ●Unlock the casters.

Problem

Check●Check if the power plug isplugged into the outlet.

●The bed does not moveeven when the handheld
controller buttons arepressed.

●Remove any obstacles.

26●The bed does not move in
the intended direction when ●Check if the caster is in thethe patient is being
steering position.
transported.

●Set the caster in the steering
position.

26●The board cannot be
removed.

●Unlock the board stoppers.

15●Check if the board is locked.

●When requesting repairs or contacting your dealer, please provide the number on the productID label.

42

■Product ID label locations

Product ID labels are affixed to the head end of the main frame, inside of the head board, hand grip,
head mattress, foot mattress, fluid basin and calf support.
<Main frame><Head board>※View after detaching the head board

Product ID Label<Hand grip>

Product ID Label<Foot Mattress><Head Mattress>
Product ID label

Product ID label

Product ID label<Calf support><Fluid basin>

Product ID label
Product ID label■Product lifetime

The lifetime of this equipment is typically eight (8) years. Lifetime is defined as the period during
which the product will maintain the specified performance and safety, provided it has been maintained
and operated in conditions of normal use in accordance with the requirements in the instruction manual.

43

■The following precautions should be noted when the bed is not in use for extended periods of time.
1. Lay the back section base flat.
2. Lower the bed to the lowest position.
3. Do not place objects other than mattress on the bed.
4. Do not place any objects on the mattress.
5. Do not place in a location that tends to be exposed to high temperatures, high humidity, lowtemperatures, dry or dusty conditions.
Store at a temperature between -10℃ and 50℃, humidity between 20%RH and 90%RH, andatmospheric pressure between 700hPa and 1060hPa.
6. Store the bed horizontally; do not stand it upright against the wall.
7. For electrically-operated beds, be sure to disconnect the power plug from the power outlet andwind the power cable onto the cable storage hook. (See p.20）●Inspect the product regularly at least once a year, even during storage.

44

Family Name
Product Name
Model No. (*1)
Quantity of motors
Back raising
Operation
Height adjustment
Procedure
Bed inclination

PD-200 Series
Electric Delivery Bed
PD-2003 Motors
Motorized
Motorized
Motorized
AC200-240V±10% ／50/60Hz
2,072mmPower supply voltage ／ frequency

Dimensions

Total length
Total width

1,056mm (When the bed side rails are raised.)
1,140mm (When the bed side rails are lowered.)

Total height883[+10] to 1,213[-20]mm

Mattress base length

1,330mm (In bed length retracted)
1,910mm (In bed length normal)

Mattress base width
Mattress base height (*2)
Casters
Power cable length
Product weight
Mattress baseMain frame
Material
Side rail
Head board
Safe working load (*3)
Inclination angle
BackraisingTime required
Height range
Heightadjustment
Time required
Inclination angle
Bedinclination
Time required
Actuator Type
Back raisingPower
consumption
Height adjustment

910mm
500[+10] to 830[-20]mm
125mm dia. single-wheel casters (Total locking system)
Effective length: 2,500mm
113kg
Made of steel, powder coating (part is resin mold item) / Particle board

Made of steel, powder coating (part is resin mold item)
PE resin mold items
220kgf (2,150N)
0°to 65°(+6.5)028sec. (*4)
330mm
23sec. (*4)
0°to 12°(+2)0
13sec. (*4)
Linear actuator (DC motor)
84VA or less
106VA or less2 min. (stop time: 18min. (*5))

Duty cycle

Dimensions
Mattress

Surface materialCore material
Service life

Head Mattress: Width 910mm × Length 1,330mm × Thickness 130mm
Foot Mattress: Width 910mm × Length 705mm × Thickness 175mm

Vinyl leather
Urethane foam
8 years (based on Paramount Bed data), excluding consumable items
such as casters*1) Put an alphabet after "Model No.", the alphabet represent the product color.
*2) The mattress base height indicates the height from the floor to the mattress base surface.
*3) Safe working load: Maximum load with which the bed can be operated. (The total load of user's weight and incidental
items including accessories.)*4) With no loads*5) Time to protect electrical parts from overheating when the bed is used continuously for duty cycle (2min).

●Put the plug code "A～H" after the product code.

45

Environmental conditions
Temperature
Humidity
Air pressure

Patient
targetgroupsWhen transported or stored-10 to 50℃
20 to 90%RH
700 to 1,060hPa

User weight (min-max)
Minimum user height
Minimum user bodymass index (BMI)
Supported age range
Intended users
Clinical benefit

Range

When in use
5 to 40℃
20 to 90%RH
700 to 1,060hPa

40kg to 180kg
146cm
17

13 years and older*Do not use the bed for children (aged 12 or under) or patient
weighing more than 180kg.
Health care workers, pregnant women ready to delivery,
maintenance staff
Patient comfort
Quality patient care and efficient care provision●Essential performance
Function to support the body of patient, Emergency CPR (back lower) function.
●Frequently used functions
Back-raising, height adjustment, trendelenburg, side rail raising and lowering operation.
●Classification and Standards
The electric delivery bed is designed and manufactured according to the following equipment classificationand standards.
・Classification as medical device in Europe: Class 1

・Safety standards applied: IEC 60601-1：2005 + CORR. 1:2006 + CORR.
2:2007+AM1:2012: IEC60601-1-2：2014: EN 60601-1：2006 +A12:2014: EN 60601-1-2：2015

・Electric shock protection: Class 1 , Type B※Equipment not suitable for use in the presence of a flammable anaesthetic mixture.

46

For the PD-200 Series, special attention is required to EMC-related safety.
Install, operate and use the bed in accordance with the following information.
Bed is intended for use in the electromagnetic environment specified by "electromagnetic disturbance" of bed.
The following abnormality may occur if using bed in unintended environment.
*The function for supporting patient's weight does not maintain.
*The function for back lower does not maintain.

●If using the other specified accessories, transducers and cable as a replacement part for internal
components, except for transducers and cables sold by the manufacturer of bed, there is a risk ofincreasing the emissions or reducing the immunity of bed.
●Use portable RF transciever more than 30 cm away from any part of bed (including peripheral
equipment such as antenna cable and external antenna). If it is used within 30 cm, the bed
performance may degrade.
●Bed cannot be used with adjacent to or stacked with other equipment. If it must be used adjacent
to or stacked with other equipment, make sure to monitor that it can operate properly in the
Bed is intended for use in clinic, restricted medical facilities, hospital (except near high-frequency surgicalinstrument, RF shielded room for magnetic resonance). Users or customers must use bed in this environment.
Bed is device that suitable to used in hospital due to emission characteristics. If it used in housing environment,
the equipment may not provide adequate protection againts wireless radio frequency transciever. The user may
need to take mitigation measures such as relocating or redirecting the device.
No.
1
2
3

Cable
AC power cable
Handheld controller cable
Nurse control panel cable

Maximum length [m]
4.6
1.0
0.2

Remarks
Without shield
Without shield
Without shield

Guidance and Manufacturer's Declaration - Electromagnetic Emissions
The PD-200 Series is intended for use in the following electromagnetic environment. The customer or user of the
PD-200 Series should make sure it is used in such an environment.
Emission test

Compliance

RF emissions
CISPR 11/EN 55011

Group 1

RF emissions
CISPR 11/EN 55011

Class A

Harmonic emissions
IEC 61000-3-2/EN 61000-3- Class A
2
Voltage fluctuations/flicker
emissions
Complies
IEC 61000-3-3/EN 61000-33

Electromagnetic Environment - Guidance
The PD-200 Series uses RF energy only for its internalfunctions. Therefore, its RF emissions are low and are not
likely to cause any interference with nearby electronicequipment.

The PD-200 Series is suitable for use in all establishments
other than domestic and those directly connected to the
public low-voltage power supply network that supplies
buildings used for domestic purposes.

47

Guidance and Manufacturer's Declaration - Electromagnetic Immunity
The PD-200 Series is intended for use in the following electromagnetic environment. Users or customers must use
bed in this environment.
Immunity test

Static electricity
discharge (ESD)
IEC 61000-4-2/EN 61000-4-2

Electrical fasttransient/burst
IEC 61000-4-4/EN 61000-4-4

IEC 60601-1-2/EN 60601-1-2
Test level

Compliance level

Contact discharge: ±8kV
Contact discharge: ±8kV
Air discharge : ±2kV, ±4kV, ±8kV, ±15kV Air discharge : ±2kV, ±4kV, ±8kV, ±15kV±2kV power supply line

±2kV power supply lineSurge
IEC 61000-4-5/EN 61000-4-5

Line - line: ±0.5kV, ±1.0kVLine - ground: ±0.5kV, ±1.0kV, ±2.0kVLine - line: ±0.5kV, ±1.0kVLine - ground: ±0.5kV, ±1.0kV, ±2.0kV

Power frequency
magnetic fields
IEC 61000-4-8/EN 61000-4-8

30 A/m30 A/m

0%UT: 0.5 cycles
Phase: 00, 450, 900, 1350, 1800, 2250,
2700, 3150

0%UT: 0.5 cycles
Phase: 00, 450, 900, 1350, 1800, 2250,
2700, 3150

0%UT: 1 cycles
70%UT: 25/30 cycles
Single phase: 00

0%UT: 1 cycles
70%UT: 25/30 cycles
Single phase: 00

0%UT: 250/300 cycles

0%UT: 250/300 cycles

Voltage Dips
IEC 61000-4-11/EN 61000-4-11

Short interuptions
IEC 61000-4-11/EN 61000-4-11

Note: UT is the AC power voltage before applying the test level.

48

Guidance and Manufacturer's Declaration - Electromagnetic Immunity
The PD-200 Series is intended for use in the following electromagnetic environment that used wireless RFtransciever. Electromagnetic interference can be prevented by maintaining the minimum distance between RF
transciever and PD-200 Series depend on the maximum output and frequency of the communication device asrecommended below.
Immunity test

IEC 60601-1-2/EN 60601-1-2
Test level

Compliance
level
(V/m)

Band
(MHz)

Service27V/m
385MHz27

380-390TETRA 400

28V/m
450MHz28

430-470

GMRS 460,
FRS 4609704-787

Maximum
power (W)Distance (m)

1.8

0.3

2

0.3

Pulse
LTE Band 13,
modulation
17
217kHz

0.2

0.3

28

800-960

GSM
800/900,
TETRA 800,
iDEN 820,CDMA 850
LTE Band 5

Pulse
modulation
18kHz

2

0.3

28

GSM 1800,
CDMA 1900,
GSM 1900,
1700-1990
DECT, LTE
Band 1, 3,
4.25, UMTS

Pulse
modulation
217kHz2

0.3

28

Bluetooth,
WLAN,
2400-2570 802.11 b/g/n
RFID 2450,
LTE band 7

Pulse
modulation
18kHz

2

0.3

9

5100-5800

WLAN
802.11 a.n

Pulse
modulation
217kHz

0.2

0.3

9V/m
710MHz9V/m
745MHz28V/m
810MHz28V/m
870MHzEnclosure port
to RF wirelesscommunications
equipment
IEC 61000-4-3/EN 61000-4-3

28V/m
930MHz28V/m1720MHz
28V/m
1845MHz
28V/m
1970MHz

28V/m
2450MHz

Modulation
Pulse
modulation
18kHz
FM±5kHzdeviation
1kHz sine

9V/m
5240MHz
9V/m
5500MHz
9V/m
5785MHz

49

Guidance and Manufacturer's Declaration - Electromagnetic Immunity
The PD-200 Series is intended for use in the following electromagnetic environment. The customer or user of the
PD-200 Series should make sure it is used in such an environment.
Immunity test

RF Conductivity
IEC 61000-4-6/EN 61000-4-6

Radioactive RF
electromagnetic
field
IEC 61000-4-3/EN 61000-4-3

IEC 60601-1-2 /EN
60601-1-2
Test level

Compliance level

3V
0.15MHz-80MHz

3V
0.15MHz-80MHz

6V in ISM bands
between 0.15MHz80MHz
80%AM at 1kHz

6V in ISM bands
between 0.15MHz80MHz
80%AM at 1kHz

3 V/m
80 MHz -2.7 GHz
80%AM at 1kHz

3 V/m
80 MHz -2.7 GHz
80%AM at 1kHz

Electromagnetic Environment - Guidance

Field intensity from the fixed RF transmitter
determined by a field investigation of the magnetic
field must be lower than the compliance level of therespective frequency band. Near devices with thefollowing symbol, interference could occur.

a.The intensity of a magnetic field from fixed transmitters such as a radio (mobile/radio) telephone base station,
land mobile radio, amateur radio, AM and FM radio broadcasting, and TV broadcasting cannot be theoretically
predicted accurately. To determine the electromagnetic environment for a fixed RF transmitter, consider
investigating the electromagnetic field in the location where the bed is used. If the intensity of the magnetic
field in the location where the PD-200 Series is used exceeds the above RF compliance level, make sure tomonitor if the PD- Series operates appropriately. If abnormal movement is found, take additional
measures as needed, such as changing the direction or position of the PD-200 Series.
b.These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption andreflection from structures, objects, and people.

50

1.When requesting repairs
First check the items described in “14. TROUBLESHOOTING” (p. 42-43) in this instruction manual. If
the problem persists, disconnect the bed's power plug from the power outlet and contact your dealer.
 Details necessary when requesting repair
・Product name, model number
・Product ID label number (see p.43)
・Date of purchase・Details of fault or problem (please provide as much details as possible)
・Your name, organization, address, and phone number
 Consumable parts
The mattress, casters, handheld controller, and rubber parts are consumable parts.
2.Minimum stock availability period
Paramount Bed keeps replacement parts (necessary to maintain bed performance) in stock for at least 8
years after the discontinuation of production.
3.Questions regarding after-sales service
Please contact your dealer.

51

When the PD-200 Electric delivery bed has reached the end of its usable life, discard it according tolocal standards.

For any queries, please feel free to contact your dealer or to the following addresses.
Website: https://www.paramount.co.jp/english/

Location in Japan (Manufacturer)

Location in Indonesia (Manufacturing premises)

Paramount
Bed Co., Ltd.
●M anufacturer
14-5, 2-chome, Higashisuna, Koto-ku,
Tokyo, 136-8670, Japan
TEL: +81-(0)3-3648-2961●Dateof M anufacturer
FAX : +81-(0)3-3648-5781

PT. Paramount Bed Indonesia

MM2100 INDUSTRIAL TOWN, BLOCK M-1-1,
GANDAMEKAR, CIKARANG BARAT,BEKASI 17530
JAWA BARAT, INDONESIATEL : +62-(0)21-8981051
FAX : +62-(0)21-89981611●Authorised representativein the european community●Serial number●Batch code52

2023-01