Synos

HIGH POWER LASER BEAM PROFILER SYSTEM SOLUTION / FFP MEASUREMENT SYSTEM
FFP MEASUREMENT SYSTEM FOR ~10W CLASS HIGH POWER LASERFFP measurement system especially targeting for ~10W output class high power laser.

FFP measurement system for high power laser is suitable for measuring the emission angle
distribution of ~10W class high power laser. FFP measurement optics for high power laser
M-Scope type HF is used. The light ﬂux emitted from high power laser sample is attenuatedby beam sampler unit after passing through f-θ lens module, and is further attenuated by NDﬁlter at subsequent stage. The FFP image that has been attenuated to appropriate beam poweris captured and image processing analysis is performed.
【Features】○M-Scope type HF, FFP measurement optics for high power laser● Specially designed f-θ lens optics for high power laser measurement● Attenuation of incident beam with two-stage beam sampler and ND ﬁlters● Covers a wide range of measurement luminous ﬂux diameters of approx. 3mmφ
●Wide measurement angle coverage of approx. ±43°
○1” high resolution CMOS detector ISA061 is used as dedicated detector○Optical beam analysis module AP013, specially designed high-functional image processing software for optical beam proﬁle analysis● All-in-one package of PC, optical beam analysis software, detector driver, calibration data.
● High-performance image processing software for optical beam proﬁle measurement Optimetrics BA Standard is pre-installed.
【Large emission area compatible model】
It is possible to conﬁgure system using FFP measurement optics
for high power laser of large emitting area M-Scope type HF+.
○Optics●850nm~940nm : M-Scope type HF+/NIR*Please specify the measurement wavelength because AR
coating to some optical parts is required.
○Detector (recommended)● 400-1100nm : 2/3” digital CCD detector ISA011-01○Optical beam analysis module AP013● PC for image processing, optical beam analysis software
Optometrics BA Standard, detector driver, calibration
data, USB key○Accessories【Detector, angle coverage, pixel resolution (approx. value)】 ● Cables, instruction manuals, etc.

【Standard component】○Optics● 850~940nm : M-Scope type HF/NIR*Please specify the measurement wavelength because AR
coating to some optical parts is required.
○Available detector (dedicated)● 400~1100nm︓ 1” HR CMOS detector ISA061○Optical beam analysis module AP013● PC for image processing, optical beam analysis software
Optometrics BA Standard, detector driver, calibration
data, USB key○Accessories● Cables, instruction manuals, etc.
Detector
Spectral range
Total pixels
Pixels pitch
Meas. angle /pix. resolution

1” HR CMOS detector ISA061
400~1100nm
2048×2048 pixels
5.5μm sq.
Meas. angle
Pix. resolution±43°/N.A. 0.68
0.05°

*Pixel resolution: The measurement angle corresponding to 1 pixel of the detector,
calculated from the measurement angle range and the pixel pitch of the detector.

【Option】○ND ﬁlter (for M-Scope type HF, dedicated 35φ）
● NIR (700~1100nm): NDF NIR 35-5 (5 types per set)
○Optics bench● Optics bench for ﬁber measurement with manual stages● Vertical setting optics bench【Detector, angle coverage, pixel resolution (approx. value)】
Detector
Spectral range
Total pixels
Pixels pitch
Meas. angle /Pix resolution

2/3” digital CCD detector ISA011-01
400~1100nm
1392×1040 pixels
6.45μm sq.
Meas. angle
Pix. resolution±12°/N.A.0.2
0.026°*Pixel resolution: The measurement angle corresponding to 1 pixel of the detector,
calculated from the measurement angle range and the pixel pitch of the detector.

【Component selection of high power laser FFP measurement system】○Stages・optics bench○Optical beam analysis module AP013○Optics & detector selection

Sample stages
Optics stages●Personal computer
・Main unit
・Accessories●for 400~1100nm

Optics bench for ﬁber
measurement
FFP measurement optics for highpower laser M-Scope type HF

1” HR CMOS detector
ISA061●for 400~1100nm

Vertical setting
optics bench＊Can be combined with various
motorized/manual stages

https://www.synos.jp●Optical beam analysis software
Optometrics BA Standard●Detector driver●Calibration data●USB licence key○Accessories

FFP measurement optics for highpower laser M-Scope type HF＋

2/3” digital CCD detector
ISA011-01

25●35φ NIR ND ﬁlter

SYNERGY OPTOSYSTEMS CO., LTD