‫ﻣﺸﺒﻚ ﻋﺪاد اﻟﻄﺎﻗﺔ‬‫‪KEW 2060BT / 2062 / 2062BT‬‬‫ﺷﻜﻞ اﻟﻔﻚ ﻣﻊ اﻟﺘﺮﻛﻴﺰ ﻋﻠﻰ اﻟﺴﻼﻣﺔ وﻗﺎﺑﻠﻴﺔ اﻻﺳﺘﺨﺪام‬‫‪KEW 2062/2062BT‬‬‫‪KEW 2060BT‬‬‫‪1000A rms‬‬‫‪1000V rms‬‬‫‪ 30‬اﻟﺘﻮاﻓﻘﻴﺎت ﺗﺼﻞ إﻟﻰ‬‫اﻟﺘﻴﺎر ﺣﺘﻰ‬‫ﺟﻬﺪ ﻳﺼﻞ إﻟﻰ‬‫● وﻇﺎﺋﻒ ﻗﻴﺎس ﻣﺨﺘﻠﻔﺔ‪ :‬اﻟﺘﻴﺎر‪ ،‬اﻟﺠﻬﺪ‪ ،‬اﻟﻄﺎﻗﺔ‪ ،‬اﻟﺘﻮاﻓﻘﻴﺎت واﻟﻜﺸﻒ ﻋﻦ اﻟﻄﻮر‬‫● ﻳﻤﻜﻦ ﻟﺸﺎﺷﺔ ‪ LCD‬ﻋﺮض ﻗﻴﻢ اﻟﺠﻬﺪ اﻟﻜﻬﺮﺑﻲ واﻟﺘﺮدد أو ﻣﻌﺎﻣﻞ اﻟﻄﺎﻗﺔ ﻓﻲ ﻧﻔﺲ اﻟﻮﻗﺖ‬‫● ﻳﻤﻜﻦ ﻋﺮض ﻛﻞ درﺟﺔ ﻣﻦ اﻟﺘﻮاﻓﻘﺎت ﻋﻠﻰ ﺷﺎﺷﺔ ‪ LCD‬ﻣﻦ اﻟﻤﺨﺘﺒﺮ‬‫● وﻇﻴﻔﺔ اﻻﺗﺼﺎل ﻋﺒﺮ ﺗﻘﻨﻴﺔ ®‪ 2060BT) Bluetooth‬و‪ 2062BT‬ﻓﻘﻂ(‬‫وﻇﺎﺋﻒ ﻗﻴﺎس ﻣﺘﻨﻮﻋﺔ‬‫〜‬‫ﺗﻴﺎر اﻟﺘﻴﺎر اﻟﻤﺘﺮدد‪ ،‬اﻟﺘﺮدد‬‫○ اﻣﻜﺎﻧﻴﺔ ﻗﻴﺎس اﻟﺘﻴﺎر‪ ،‬اﻟﺠﻬﺪ‪ ،‬اﻟﻄﺎﻗﺔ‪ ،‬اﻟﺘﻮاﻓﻘﻴﺎت‬‫واﻟﻜﺸﻒ ﻋﻦ اﻟﻄﻮر‪.‬‬‫〜‬‫○ ﺻﺤﻴﺢ ‪ :RMS‬ﻳﺸﻴﺮ إﻟﻰ ﻗﻴﻤﺔ ﺻﺤﻴﺤﺔ ﻋﻠﻰ ﺷﻜﻞ‬‫ﻣﻮﺟﺔ ﻣﺸﻮﻫﺔ‪.‬‬‫‪A‬‬‫‪V‬‬‫‪W‬‬‫اﻟﻄﺎﻗﺔ‬‫‪DATA‬‬‫‪HOLD‬‬‫إﻳﻘﺎف اﻟﺒﻴﺎﻧﺎت‬‫ﺟﻬﺪ اﻟﺘﻴﺎر اﻟﻤﺘﺮدد‪ ،‬اﻟﺘﺮدد‬‫اﻟﺘﻮاﻓﻘﻴﺎت‬‫○ وﻇﺎﺋﻒ اﻟﺬروة‪ ،‬اﻟﺤﺪ اﻷﻗﺼﻰ‪ ،‬اﻟﺤﺪ اﻷدﻧﻰ‪،‬‬‫اﻟﻤﺘﻮﺳﻂ ﻣﺘﻮﻓﺮة‪.‬‬‫ﺳﻬﻞ اﻟﺘﺤﻘﻖ ﻣﻦ اﻟﺘﻐﻴﻴﺮ اﻟﺰﻣﻨﻲ ﻓﻲ ﻣﺼﺪر اﻟﻄﺎﻗﺔ‪.‬‬‫ﻛﺸﻒ اﻟﻄﻮر‬‫‪SETUP‬‬‫○ ﻳﻤﻜﻦ ﻟﺸﺎﺷﺔ ‪ LCD‬ﻋﺮض ﻧﻔﺲ اﻟﻮﻗﺖ ﻗﻴﻢ اﻟﺠﻬﺪ‬‫واﻟﺘﺮدد أو اﻟﻄﺎﻗﺔ وﻋﺎﻣﻞ اﻟﻄﺎﻗﺔ‪.‬‬‫إﻋﺪاد‬‫ﻳﻤﻜﻦ اﺳﺘﺨﺪام ﻣﺤﻄﺎت‬‫اﻹدﺧﺎل ﺣﺘﻰ ‪.3‬‬‫ﻗﻢ ﺑﺘﻌﻴﻴﻦ ﺧﻴﻮط اﻻﺧﺘﺒﺎر‬‫وﻓﻘﺎ ﻟﺘﻜﻮﻳﻦ اﻷﺳﻼك‪.‬‬‫ﻗﻴﺎس اﻟﻄﺎﻗﺔ‬‫ﻗﻴﺎس اﻟﻄﺎﻗﺔ ﻋﻠﻰ‬‫أي ﻧﻈﺎم أﺳﻼك ﻣﻤﻜﻦ‪.‬‬‫* ﻋﻠﻰ ﺳﺒﻴﻞ اﻟﻤﺜﺎل‪:‬‬‫‪3P4W‬‬‫)‪R(L1‬‬‫)‪S(L2‬‬‫ﺣﻤﻮﻟﺔ‬‫اﳌﺼﺪر‬‫أﺣﻤﺮ‬‫)‪T(L3‬‬‫‪N‬‬‫أﺳﻮد‬‫ﻳﻤﻜﻦ ﻟـ ‪ KEW 2060BT‬و‪ KEW 2062‬و‪KEW 2062BT‬‬‫إﺟﺮاء ﻗﻴﺎﺳﺎت أﺣﺎدﻳﺔ اﻟﻄﻮر ﻟﺴﻠﻜﻴﻦ ‪ 3 /‬أﺳﻼك أﺣﺎدﻳﺔ‬‫اﻟﻄﻮر وﻗﻴﺎﺳﺎت اﻟﺘﻮازن وﻋﺪم اﻻﺗﺰان ﻟﺜﻼث أﺳﻼك ﺛﻼﺛﻴﺔ‬‫اﻟﻄﻮر ‪ 4 /‬أﺳﻼك ﺛﻼﺛﻴﺔ اﻟﻄﻮر‪.‬‬‫ﻳﻤﻜﻦ ﻟﻠﺸﺎﺷﺔ اﻟﻤﺰدوﺟﺔ أن ﺗﻌﺮض ﻓﻲ وﻗﺖ واﺣﺪ اﻟﻌﺪﻳﺪ‬‫ﻣﻦ اﻟﻤﻌﻠﻤﺎت ﻣﺜﻞ & ‪W & PF, W & deg, W & VA, W‬‬‫‪ ،Var, V & A‬إﻟﺦ‪.‬‬‫ﻋﻦ ﻃﺮﻳﻖ ﺗﻌﻴﻴﻦ ‪ CT‬و ‪ VT‬اﻟﻨﺴﺒﺔ‬‫وﻗﻴﺎس اﻟﺠﺎﻧﺐ اﻟﺜﺎﻧﻮي ﻟﻠﻤﺤﻮل‪ ،‬ﻳﻤﻜﻦ‬‫اﻟﺤﺼﻮل ﻋﻠﻰ ﻗﻴﻤﺔ اﻟﺠﺎﻧﺐ اﻷﺳﺎﺳﻲ‪.‬‬‫ﻳﻤﻜﻦ ﻗﻴﺎس إﺟﻤﺎﻟﻲ اﻟﻄﺎﻗﺔ ﺛﻼﺛﻴﺔ‬‫اﻟﻄﻮر ﺑﺨﻄﻮات ﺑﺴﻴﻄﺔ‪.‬‬‫ﻳﻤﻜﻦ ﻗﻴﺎس وﺣﺪات ﻣﺨﺘﻠﻔﺔ ﻣﺜﻞ‪:‬‬‫اﻟﻄﺎﻗﺔ اﻟﻨﺸﻄﺔ‪/‬اﻟﻤﺘﻔﺎﻋﻠﺔ‪/‬اﻟﻈﺎﻫﺮﻳﺔ‪،‬‬‫وﻋﺎﻣﻞ اﻟﻄﺎﻗﺔ واﺧﺘﻼف اﻟﻄﻮر‬‫)‪ 1P2W‬ﻗﻔﻂ( اﻟﻤﻄﻠﻮﺑﺔ ﻟﻘﻴﺎس‬‫اﻟﻄﺎﻗﺔ‪.‬‬‫ﻗﻴﺎس اﻟﺘﻮاﻓﻘﻴﺎت‬‫أﻋﻄﺎﻻ ﻣﺨﺘﻠﻔﺔ ﻓﻲ اﻟﻤﺠﺎل‪.‬‬‫ً‬‫ﻳﻤﻜﻦ ﻗﻴﺎس ﺗﻮاﻓﻘﻴﺎت اﻟﺠﻬﺪ واﻟﺘﻴﺎر‪ ،‬واﻟﺘﻲ ﺗﺴﺒﺐ‬‫ﻣﻦ اﻟﻤﻤﻜﻦ ﻗﻴﺎس وإﻇﻬﺎر ﻛﻞ ﺗﻮاﻓﻘﻴﺎت اﻟﺠﻬﺪ واﻟﺘﻴﺎر ﺣﺘﻰ ‪.30‬‬‫إﺟﻤﺎﻟﻲ ﻋﺎﻣﻞ اﻟﺘﺸﻮه اﻟﺘﻮاﻓﻘﻲ ﻣﻮﺟﻮد ﻓﻲ ﻧﻮﻋﻴﻦ ‪ THD-R‬و ‪.THD-F‬‬‫ﻳﻤﻜﻦ ﻋﺮض ﻛﻞ درﺟﺔ ﻣﻦ درﺟﺎت اﻟﺘﻮاﻓﻘﻴﺎت ﻋﻠﻰ ﺷﺎﺷﺔ ‪ LCD‬اﻟﺨﺎﺻﺔ ﺑﺠﻬﺎز‬‫اﻻﺧﺘﺒﺎر‪.‬‬‫‪L‬‬‫ﺣﻤﻮﻟﺔ‬‫اﻟﻤﺼﺪر‬‫‪N‬‬‫ﻳﻤﻜﻦ ﻋﺮض ﻛﻞ ﻣﻦ ﻗﻴﻤﺔ اﻟﺘﻴﺎر )أو ﻗﻴﻤﺔ اﻟﺠﻬﺪ( وﻧﺴﺒﺔ اﻟﻤﺤﺘﻮى ﻋﻠﻰ ﻧﻔﺲ‬‫اﻟﺸﺎﺷﺔ‪.‬‬‫ﺷﻜﻞ اﻟﻔﻚ ﻣﻊ اﻟﺘﺮﻛﻴﺰ ﻋﻠﻰ اﻟﺴﻼﻣﺔ‬‫وﻗﺎﺑﻠﻴﺔ اﻻﺳﺘﺨﺪام‬‫ﺻﻒ ﻧﻮﻋﻴﻦ ﻣﻦ اﻟﻔﻚ ﻟﺘﻠﺒﻴﺔ ﺑﻴﺌﺔ اﻻﺧﺘﺒﺎر‬‫○ ﻳﺘﻤﻴﺰ ‪ KEW 2060BT‬ﺑﺸﻜﻞ ﻓﻚ ﺧﺎص ﻣﺼﻤﻢ ﺣﺪﻳﺜًﺎ‬‫ﺑﺎﺳﺘﺨﺪام ﻓﻲ ﺑﺴﺒﺎر ﻛﺒﻴﺮ‪ .‬ﻓﻚ ﻛﺒﻴﺮ ﺟﺪا ﻣﻊ‬‫ﻳﻤﻜﻦ ﻟﺸﻜﻞ ﻗﻄﺮة اﻟﻤﺴﻴﻞ ﻟﻠﺪﻣﻮع ﺗﺜﺒﻴﺖ ﺷﺮﻳﻂ ﻛﺒﻴﺮ ﺑﺄﻣﺎن‪.‬‬‫)ﺣﺠﻢ اﻟﻤﻮﺻﻞ ‪ ،75mm‬ﺑﺴﺒﺎر ‪ 30mm × 80‬ﻛﺤﺪ أﻗﺼﻰ(‬‫○ ‪ KEW 2062‬و‪ KEW 2062BT‬ﻟﺪﻳﻬﻤﺎ ﺗﻤﺰق‬‫ﻓﻚ اﻟﺸﻜﻞ اﻟﻤﺴﻘﻂ‪ ،‬واﻟﺤﺠﻢ ﻣﻨﺎﺳﺐ‬‫ﻟﻼﺳﺘﺨﺪام ﻓﻲ ﻣﻜﺘﺐ ﺻﻐﻴﺮ وﻣﺼﻨﻊ‪.‬‬‫)ﺣﺠﻢ اﻟﻤﻮﺻﻞ ‪(55mm‬‬‫ﻳﻤﻜﻦ ﻟﻠﻔﻚ ذو اﻟﺸﻜﻞ اﻟﻤﺴﻴﻞ‬‫ﻟﻠﺪﻣﻮع أن ﻳﺒﻘﻲ اﻟﻤﺸﻐﻞ ﺧﺎﻟﻴًﺎ ﻣﻦ اﻟﺨﻄﺮ‬‫ﻣﻦ ﻟﻤﺲ ﺑﺴﺒﺎر‪.‬‬‫اﻟﺼﻮرة‪2060BT:‬‬‫ﻛﺸﻒ اﻟﻄﻮر‬‫ﻳﻤﻜﻦ ﺗﺄﻛﻴﺪ اﻟﻤﺮﺣﻠﺔ اﻹﻳﺠﺎﺑﻴﺔ واﻟﻤﺮﺣﻠﺔ اﻟﺴﻠﺒﻴﺔ ﻣﻦ ﺧﻼل ﺻﻮت اﻟﺠﺮس واﻟﺸﺎﺷﺔ دون ﺗﻐﻴﻴﺮ‬‫اﺗﺼﺎل اﻟﺨﻴﻮط‪.‬‬‫)‪R (L1‬‬‫)‪S (L2‬‬‫)‪T (L3‬‬‫ﺣﻤﻮﻟﺔ‬‫أﺻﻔﺮ‬‫اﻟﻤﺼﺪر‬‫أﺣﻤﺮ‬‫‪N‬‬‫أﺳﻮد‬‫*‪KEW Power‬‬‫اﻟﺮﺟﺎء اﻟﺒﺤﺚ‬‫”*‪“KEW Power‬‬‫ﻗﺪ ﻳﺘﻢ ﻓﺮض رﺳﻮم ﻋﻠﻰ اﻻﺗﺼﺎﻻت ﺗﺘﻜﺒﺪ‬‫ﺑﺸﻜﻞ ﻣﻨﻔﺼﻞ ﻟﺘﻨﺰﻳﻞ اﻟﺘﻄﺒﻴﻖ‪.‬‬‫*ﺗﻘﻨﻴﺔ ®‪ Bluetooth‬ﻫﻲ ﻋﻼﻣﺔ ﺗﺠﺎرﻳﺔ أو ﻋﻼﻣﺔ ﺗﺠﺎرﻳﺔ‬‫ﻣﺴﺠﻠﺔ ﻟﺸﺮﻛﺔ ﺗﻘﻨﻴﺔ ‪ .Bluetooth® SIG‬ﺷﺮﻛﺔ‬‫™‪ Android‬ﻫﻲ ﻋﻼﻣﺔ ﺗﺠﺎرﻳﺔ أو ﻋﻼﻣﺔ ﺗﺠﺎرﻳﺔ ﻣﺴﺠﻠﺔ‬‫ﻟﺸﺮﻛﺔ ‪ Google Inc. iOS‬ﻫﻲ ﻋﻼﻣﺔ ﺗﺠﺎرﻳﺔ أو ﻋﻼﻣﺔ‬‫ﺗﺠﺎرﻳﺔ ﻣﺴﺠﻠﺔ ﻟﺸﺮﻛﺔ ‪ ،Cisco Technology‬ﻓﻲ اﻟﻮﻻﻳﺎت‬‫اﻟﻤﺘﺤﺪة وﺑﻠﺪان أﺧﺮى‪.‬‬‫ﺗﻨﺰﻳﻞ ﻋﻠﻰ‬‫ﺗﻘﻨﻴﺔ ®‪ 2060BT) Bluetooth‬و‪ 2062BT‬ﻓﻘﻂ(‬‫○ ﺗﻄﺒﻴﻖ ﻣﺨﺼﺺ "*‪ "KEW Power‬ﻳﺪﻋﻢ ﻛﻼً ﻣﻦ ™‪ Android‬و‪.iOS‬‬‫○ ﻳﻤﻜﻦ ﻓﺤﺺ ﺷﺎﺷﺔ ‪ LCD‬ﻋﻠﻰ اﻷﺟﻬﺰة اﻟﺬﻛﻴﺔ‪.‬‬‫○ ﻳﻤﻜﻦ ﺗﺨﺰﻳﻦ اﻟﻘﻴﻢ واﻟﺮﺳﻮﻣﺎت اﻟﺒﻴﺎﻧﻴﺔ اﻟﻤﻘﺎﺳﺔ ﺑﻀﻐﻂ زر‪.‬‬‫○ ﻳﻤﻜﻦ إﻇﻬﺎر اﻟﺠﻬﺪ واﻟﺘﻴﺎر ﻓﻲ ﺷﺎﺷﺔ اﻟﻌﺮض ﻋﻠﻰ ﺷﻜﻞ ﻣﻮﺟﻲ‪ .‬وﺟﻮد‬‫اﻟﺘﻮاﻓﻘﻴﺎت ﻳﻤﻜﻦ اﻟﻌﺜﻮر ﻋﻠﻴﻪ ﺑﺴﻬﻮﻟﺔ‪.‬‬‫○ ﻳﻤﻜﻦ ﺗﻌﻴﻴﻦ ﻋﺘﺒﺔ ﻛﻞ ﻗﻴﻤﺔ ﻳﺘﻢ ﻗﻴﺎﺳﻬﺎ ﻓﻲ اﻟﺘﻄﺒﻴﻖ‪ ،‬وﻣﻦ اﻟﻤﻤﻜﻦ‬‫اﻟﺤﻜﻢ ﺑﺎﻟﻨﺠﺎح واﻟﻔﺸﻞ‪.‬‬‫ﺗﻄﺒﻴﻖ ‪iOS‬‬‫اﻟﺘﻄﺒﻴﻖ ﻟﻨﻈﺎم ‪iOS‬‬‫ﺑﺮﻧﺎﻣﺞ ‪ iOS‬اﻟﻤﺠﺎﻧﻲ "*‪"KEW Power‬‬‫ﻣﺘﻮﻓﺮ ﻋﻠﻰ "ﻣﺘﺠﺮ اﻟﺘﻄﺒﻴﻘﺎت"‬‫ﺗﻄﺒﻴﻖ أﻧﺪروﻳﺪ ﻋﻠﻰ‬‫ﺗﻄﺒﻴــﻖ ™أﻧﺪروﻳﺪ‬‫ﺗﻄﺒﻴﻖ ﻟﻨﻈﺎم ™‪Android‬‬‫ﺑﺮﻧﺎﻣﺞ ‪ Android‬اﻟﻤﺠﺎﻧﻲ "*‪"KEW Power‬‬‫ﻣﺘﻮﻓﺮ ﻓﻲ "ﻣﺘﺠﺮ ﺟﻮﺟﻞ ﺑﻼي"‬‫● اﻟﻤﻠﺤﻘﺎت‬‫● ﻣﻮاﺻﻔﺎت ‪KEW 2060BT / 2062 / 2062BT‬‬‫‪2062/2062BT‬‬‫‪2060BT‬‬‫‪1P2W, 1P3W, 3P3W, 3P4W‬‬‫اﻟﺠﻬﺪ اﻟﻜﻬﺮﺑﻲ‪ ،‬اﻟﺘﺮدد‪ ،‬اﻟﻄﺎﻗﺔ اﻟﻨﺸﻄﺔ‪ ،‬اﻟﻄﺎﻗﺔ اﻟﺘﻔﺎﻋﻠﻴﺔ‪ ،‬اﻟﻄﺎﻗﺔ اﻟﻈﺎﻫﺮة‪ ،‬ﻋﺎﻣﻞ اﻟﻄﺎﻗﺔ )‪ ،(cosθ‬زاوﻳﺔ اﻟﻄﻮر‪ ،‬اﻟﺘﻮاﻓﻘﻴﺎت)‪ ،(THD-R/THD-F‬دوران اﻟﻄﻮر‬‫ﺗﻮﺻﻴﻼت اﻷﺳﻼك‬‫اﻟﻘﻴﺎﺳﺎت واﻟﻤﻌﻠﻤﺎت‬‫‪ACV‬‬‫‪1000V‬‬‫اﻟﻨﻄﺎق‬‫‪ 40.0)±0.7%rdg±3dgt‬إﻟﻰ ‪ 70.1)±3.0%rdg±5dgt (70.0Hz‬إﻟﻰ ‪(1kHz‬‬‫دﻗﺔ‬‫‪ 1.7‬أو أﻗﻞ‬‫ﻋﺎﻣﻞ اﻟﺬروة‬‫‪ACA‬‬‫‪ 3) 40.00/400.0/1000A‬ﻧﻄﺎﻗﺎت ﺗﻠﻘﺎﺋﻴﺔ(‬‫اﻟﻨﻄﺎق‬‫‪ 40.0)±1.0%rdg±3dgt‬إﻟﻰ ‪ 70.1)±2.0%rdg±5dgt (70.0Hz‬إﻟﻰ ‪(1kHz‬‬‫دﻗﺔ‬‫ﻋﺎﻣﻞ اﻟﺬروة‬‫‪ 3‬أو أﻗﻞ ﻋﻠﻰ ﻧﻄﺎق ‪ 40.00A/400.0A، 3‬أو أﻗﻞ ﻣﻦ ‪ 1500A‬اﻟﺬروة ﻋﻠﻰ ﻧﻄﺎق ‪1000A‬‬‫اﻟﺘﺮدد‬‫‪ 40.0‬إﻟﻰ ‪999.9Hz‬‬‫ﻧﻄﺎق اﻟﻌﺮض‬‫‪±0.3%rdg±3dgt‬‬‫دﻗﺔ‬‫اﻟﻄﺎﻗﺔ اﻟﻨﺸﻄﺔ‬‫‪40.00/400.0/1000kW‬‬‫اﻟﻨﻄﺎق‬‫‪ ،PF1) ±1.7%rdg±5dgt‬ﻣﻮﺟﺔ ﺟﻴﺒﻴﺔ‪ 45 ،‬إﻟﻰ ‪(65Hz‬‬‫دﻗﺔ‬‫ﻃﺎﻗﺔ ﻇﺎﻫﺮة‬‫‪40.00/400.0/1000kVA‬‬‫اﻟﻨﻄﺎق‬‫‪ ±1dgt‬ﻣﻘﺎﺑﻞ ﻛﻞ ﻗﻴﻤﺔ ﻣﺤﺴﻮﺑﺔ‪ ،‬اﻟﻤﺠﻤﻮع‪ :‬إﺿﺎﻓﺔ أﺧﻄﺎء ﻓﻲ ﻛﻞ ﻗﻨﺎة‪(3P3W: ±2dgt, 3P4W: ±3dgt ،‬‬‫دﻗﺔ‬‫ﻃﺎﻗﺔ ﺗﻔﺎﻋﻠﻴﺔ‬‫‪40.00/400.0/1000kVar‬‬‫اﻟﻨﻄﺎق‬‫‪ ±1dgt‬ﻣﻘﺎﺑﻞ ﻛﻞ ﻗﻴﻤﺔ ﻣﺤﺴﻮﺑﺔ‪ ،‬اﻟﻤﺠﻤﻮع‪ :‬إﺿﺎﻓﺔ أﺧﻄﺎء ﻓﻲ ﻛﻞ ﻗﻨﺎة‪(3P3W: ±2dgt, 3P4W: ±3dgt ،‬‬‫دﻗﺔ‬‫ﻋﺎﻣﻞ اﻟﻄﺎﻗﺔ‬‫‪ 1.000‬إﻟﻰ ‪ 0.000‬إﻟﻰ ‪1.000+‬‬‫ﻧﻄﺎق اﻟﻌﺮض‬‫‪ ±1dgt‬ﻣﻘﺎﺑﻞ ﻛﻞ ﻗﻴﻤﺔ ﻣﺤﺴﻮﺑﺔ‪ ،‬اﻟﻤﺠﻤﻮع‪ :‬إﺿﺎﻓﺔ أﺧﻄﺎء ﻓﻲ ﻛﻞ ﻗﻨﺎة‪(3P3W: ±2dgt, 3P4W: ±3dgt ،‬‬‫دﻗﺔ‬‫زاوﻳﺔ اﻟﻄﻮر )‪ 1P2W‬ﻓﻘﻂ(‬‫‪ 180.0‬إﻟﻰ ‪ 0.0‬إﻟﻰ ‪179.9+‬‬‫ﻧﻄﺎق اﻟﻌﺮض‬‫‪±3.0°‬‬‫دﻗﺔ‬‫ﺗﻮاﻓﻘﻴﺎت ‪) RMS‬ﻣﻌﺪل اﻟﻤﺤﺘﻮى(‬‫ﺗﺮﺗﻴﺐ اﻟﺘﺤﻠﻴﻞ‬‫اﻟﺘﺮﺗﻴﺐ ﻣﻦ ‪ 1‬إﻟﻰ ‪30‬‬‫دﻗﺔ‬‫)ﻣﻦ ‪ 21‬إﻟﻰ ‪) ±20％rdg±10dgt (30‬ﻣﻦ ‪ 11‬إﻟﻰ ‪) ±10％rdg±10dgt (20‬ﻣﻦ ‪ 1‬إﻟﻰ ‪±5.0％rdg±10dgt (10‬‬‫ﻣﺠﻤﻮع اﻟﺘﻮاﻓﻘﺎت ‪THD-R/THD-F‬‬‫‪ 0.0%‬إﻟﻰ ‪100.0%‬‬‫ﻧﻄﺎق اﻟﻌﺮض‬‫دﻗﺔ‬‫‪ ±1dgt‬ﻣﻘﺎﺑﻞ اﻟﻨﺘﺎﺋﺞ اﻟﻤﺤﺴﻮﺑﺔ ﻟﻜﻞ ﻗﻴﻤﺔ ﻣﺤﺴﻮﺑﺔ‬‫‪ ACV 80‬إﻟﻰ ‪ 1100V‬ﻓﻮﻟﺖ )‪ 45‬إﻟﻰ ‪(65Hz‬‬‫دوران اﻟﻄﻮر‬‫®‬‫اﻟﺤﺪ اﻷﻗﺼﻰ‪/‬اﻟﺤﺪ اﻷدﻧﻰ‪/‬اﻟﻤﺘﻮﺳﻂ‪/‬اﻟﺬروة‪ ،‬إﻳﻘﺎف اﻟﺒﻴﺎﻧﺎت‪ ،‬ﺗﻘﻨﻴﺔ ﺑﻠﻮﺗﻮث )‪ 2060BT‬و‪ 2062BT‬ﻓﻘﻂ(‪ ،‬اﻹﺿﺎءة اﻟﺨﻠﻔﻴﺔ‪ ،‬إﻳﻘﺎف اﻟﺘﺸﻐﻴﻞ اﻟﺘﻠﻘﺎﺋﻲ‬‫اﻟﻮﻇﺎﺋﻒ اﻷﺧﺮى‬‫ﻋﺎم‬‫واﺟﻬﺔ اﻻﺗﺼﺎﻻت‬‫ﺗﻘﻨﻴﺔ ‪ 2060BT) Bluetooth®5.0‬و‪ 2062BT‬ﻓﻘﻂ(‬‫‪LR6(AAA)(1.5V)×2‬‬‫ﻣﺼﺪر اﻟﻄﺎﻗﺔ‬‫زﻣﻦ اﻟﻘﻴﺎس اﻟﻤﺴﺘﻤﺮ‬‫ﺣﻮاﻟﻲ ‪ 58‬ﺳﺎﻋﺔ‬‫ﺣﺠﻢ اﻟﻤﻮﺻﻞ‬‫‪ Φ55mm‬ﻛﺤﺪ أﻗﺼﻰ‬‫‪ Φ75mm‬ﻛﺤﺪ أﻗﺼﻰ )ﺑﺴﺒﺎر ‪(80mm×30mm‬‬‫‪ / 247(L)×105(W)x49(D)mm‬ﺗﻘﺮﻳﺒًﺎ ‪) 490g‬ﺑﻤﺎ ﻓﻲ ذﻟﻚ اﻟﺒﻄﺎرﻳﺎت(‬‫‪ / 283(L)×143(W)x49(D) mm‬ﺗﻘﺮﻳﺒًﺎ ‪) 590g‬ﺑﻤﺎ ﻓﻲ ذﻟﻚ اﻟﺒﻄﺎرﻳﺎت(‬‫اﻟﺒﻌﺪ ‪ /‬اﻟﻮزن‬‫)‪IEC 61010-1, IEC 61010-2-032, IEC 61326-1, 2-2(EMC), IEC 60529(IP40‬‬‫اﻟﻤﻌﺎﻳﻴﺮ اﻟﻤﻌﻤﻮل ﺑﻬﺎ‬‫‪ CAT Ⅳ 300V / CAT Ⅲ 600V / CAT II 1000V‬درﺟﺔ اﻟﺘﻠﻮث ‪2‬‬‫‪ CAT Ⅳ 600V / CAT Ⅲ 1000V‬درﺟﺔ اﻟﺘﻠﻮث ‪2‬‬‫‪) 7290‬ﻣﺠﻤﻮﻋﺔ ﺳﻠﻚ اﺧﺘﺒﺎر اﻟﺠﻬﺪ( ‪) 9198‬ﺣﻘﻴﺒﺔ اﻟﺤﻤﻞ( ‪ ،LR6(AAA)×2‬دﻟﻴﻞ اﻟﺘﻌﻠﻴﻤﺎت‬‫اﻟﻤﻠﺤﻘﺎت‬‫‪i‬‬‫‪i‬‬‫‪ 7290‬ﻣﺠﻤﻮﻋﺔ ﻣﺠﺲ ﻻﺧﺘﺒﺎر اﻟﺠﻬﺪ‬‫‪ 9198‬ﺣﻘﻴﺒﺔ ﺣﻤﻞ‬‫● اﻟﻘﻴﺎس ﻓﺌﺎت )‪(CAT‬‬‫ﻳﻈﻬﺮ اﻟﺸــﻜﻞ ﻣﺜﺎﻻ ﻟﻠﻘﻴﺎس اﻟﻔﺌﺔ‪.‬‬‫أدوات اﻟﻘﻴــﺎس اﻟﻤﺼﻤﻤﺔ ﻟـ ‪ CAT IV‬اﻟﺒﻴﺌﺎت ﻳﻤﻜﻦ‬‫اﺳــﺘﺨﺪاﻣﻬﺎ ﻓﻲ اﻟﺒﻴﺌﺎت ﻣﻦ ‪ CAT III‬و‪ CAT II‬و‪.CAT O‬‬‫‪ :O‬دواﺋﺮ اﻟﻘﻴﺎس‬‫ﺑﺪون ﻗﻴﺎس‬‫اﻟﻔﺌﺔ‪.‬‬‫ﻣﻘﺒﺲ‬‫● دﻟﻴﻞ اﺧﺘﻴﺎر ﻋﺪادات اﻟﻄﺎﻗﺔ‬‫ﻣﺸﺒﻚ ﻋﺪاد اﻟﻄﺎﻗﺔ‬‫‪2060BT‬‬‫‪2062‬‬‫‪2062BT‬‬‫ﻋﺪاد اﻟﻄﺎﻗﺔ‬‫ﻣﺤﻠﻞ ﺟﻮدة اﻟﻄﺎﻗﺔ‬‫‪6305‬‬‫‪6315‬‬‫ﺣﺠﻢ اﻟﻤﻮﺻﻞ‬‫اﻟﺬاﻛﺮة‬‫ﻋﺪد ﻗﻨﻮات اﻹدﺧﺎل‬‫واﺟﻬﺔ اﻻﺗﺼﺎﻻت‬‫‪300V‬‬‫‪CAT Ⅳ‬‬‫اﻟﻤﺮﺣﻠﺔ اﻻﺳﻤﻴﺔ إﻟﻰ‬‫اﻟﺠﻬﺪ اﻷرﺿﻲ‬‫ﻗﻴﺎس ﻓﺌﺔ‬‫ﻣﺜﺎل ﻗﻴﺎس ﻓﺌﺔ‬‫ﺣﺬر‬‫ﻣﻈﻬﺮ‬‫اﻟﺠﻬﺪ ］‪［V‬‬‫اﻟﺘﻴﺎر］‪［A‬‬‫اﻟﻄﺎﻗﺔ ］‪［W‬‬‫اﻟﺘﺮدد ］‪［Hz‬‬‫اﻟﻄﺎﻗﺔ］‪［Wh‬‬‫اﻟﺘﻮاﻓﻘﻴﺎت‬‫اﻧﺘﻔﺎخ‬‫اﻧﺨﻔﺎض‬‫اﻟﻄﺎﻗﺔ‬‫اﻧﻘﻄﺎع‬‫اﻟﺠﻮدة‬‫ﻋﺎﺑﺮﻳﻦ‬‫ﺗﺪﻓﻖ اﻟﺘﻴﺎرات‬‫اﻷﺳﻼك اﻟﺪاﺧﻠﻴﺔ‬‫ﺳﻠﻚ وارد‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫‪Φ75mm‬‬‫‪Φ55mm‬‬‫‪Φ55mm‬‬‫ﻳﺨﺘﻠﻒ ﺣﺴﺐ أﺟﻬﺰة اﺳﺘﺸﻌﺎر‬‫اﻟﻤﺸﺒﻚ اﻻﺧﺘﻴﺎرﻳﺔ‪ .‬اﻟﻤﺴﺘﺨﺪﻣﺔ‪.‬‬‫）‪4ch（V3,A1‬‬‫ﺗﻘﻨﻴﺔ ®‪Bluetooth‬‬‫）‪4ch（V3,A1‬‬‫）‪4ch（V3,A1‬‬‫ﺗﻘﻨﻴﺔ ®‪Bluetooth‬‬‫● ﺻﺪر ﻫﺬا اﻟﻤﻨﺸــﻮر ﻓﻲ ﻧﻮﻓﻤﺒﺮ ‪ .2023‬ﺗﺨﻀﻊ ﻣﺤﺘﻮﻳﺎت ﻫﺬا اﻟﻤﻨﺸــﻮر ﻟﻠﺘﻐﻴﻴﺮ دون إﺷﻌﺎر‪.‬‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫✓‬‫ﻳﺨﺘﻠﻒ ﺣﺴﺐ أﺟﻬﺰة اﺳﺘﺸﻌﺎر‬‫اﻟﻤﺸﺒﻚ اﻻﺧﺘﻴﺎرﻳﺔ‪ .‬اﻟﻤﺴﺘﺨﺪﻣﺔ‪.‬‬‫ﺑﻄﺎﻗﺔ ‪SD‬‬‫ﺑﻄﺎﻗﺔ ‪SD‬‬‫）‪6ch（V3,A3‬‬‫）‪7ch（V3,A4‬‬‫®‬‫ﻳﻮ اس ﺑﻲ‪ ،‬ﺗﻘﻨﻴﺔ ®‪ Bluetooth‬ﻳﻮ اس ﺑﻲ‪ ،‬ﺗﻘﻨﻴﺔ ‪Bluetooth‬‬‫ﻗﻴﻤــﺔ اﻟﺠﻬــﺪ اﻟﺘــﻲ ﺗﻠــﻲ اﻟﻔﺌــﺔ ﺗﻌﻨــﻲ ﺣــﺪ اﻟﻄــﻮر ﻋﻠــﻰ‬‫اﻷرض اﻟﺠﻬــﺪ‪ .‬ﻟﻴــﺲ اﻟﺤــﺪ اﻷﻗﺼــﻰ ﻟﻠﻤﺮﺣﻠــﺔ اﻟﺠﻬــﺪ‪.‬‬‫ﻟﺬﻟــﻚ‪ ،‬ﻓــﻲ ﺣﺎﻟــﺔ اﻟﻘﻴــﺎس أداة ﺗﺘﻮاﻓــﻖ ﻣــﻊ ‪CAT IV‬‬‫‪ ،300V‬ﻳﻤﻜــﻦ اﺳــﺘﺨﺪاﻣﻪ ﻓــﻲ اﻟﺪاﺋــﺮة اﻟﻜﻬﺮﺑﺎﺋﻴــﺔ ﺣﺘــﻰ‬‫‪ 520V‬ﻣﺮﺣﻠــﺔ إﻟــﻰ ﻣﺮﺣﻠــﺔ ﻋﻨــﺪ اﻟﺨــﻂ ﺛﻼﺛــﻲ اﻟﻄــﻮر‬‫اﻟﻤﺘﺼﻞ ﺑﻮاﺳﻄﺔ اﺗﺼﺎل اﻟﻨﺠﻤﺔ‪.‬‬‫ﺗﻨﻈﻢ ﺑﻌﺾ اﻟﺒﻠﺪان اﻻﻣﺘﺜﺎل ﻟﻘﺎﻧﻮن اﻟﺮادﻳﻮ‬‫اﻟﺨﺎص ﺑﻬﺎ ﻓﻴﻤﺎ ﻳﺘﻌﻠﻖ ﺑﺎﻟﻤﻨﺘﺠﺎت اﻟﻤﺠﻬﺰة ﺑﺘﻘﻨﻴﺔ‬‫®‪.Bluetooth‬‬‫ﻳﺮﺟﻰ ﺗﺄﻛﻴﺪ ذﻟﻚ ﻣﻊ ﻣﻮزﻋﻚ ﻗﺒﻞ ﺷﺮاء ﻣﻨﺘﺠﺎﺗﻨﺎ‬‫‪ -‬ﻣﺰود ﺑﺘﻘﻨﻴﺔ ®‪.Bluetooth‬‬‫‪LC_2060BT_2062_2062BT_2_1_Nov23_SS_ar‬‬