testo 330i · Flue gas measuring instrument
testo 330i LX · Flue gas measuring instrument

Startup & SafetyRegister your Testo product at www.testo.com/register and receive a
one-year free warranty extension.
The product registration is valid for 30 days after purchase.
For product registration terms and conditions and participating
countries, please go to www.testo.com/register2

1 Commissioning

1
1.1.

1.2.

Commissioning
Installing the App

A mobile terminal device (tablet or smartphone) with the Android oriOS operating system and Bluetooth® interface is required tooperate the testo 330i / testo 330i LX measuring instrument.
The testo 330i App also needs to be installed on this mobile
device:1. Open the AppStore on the mobile terminal device.
2. Enter the search term Testo SE to start searching.
3. Select the testo 330i App and install.
Information on the compatibility of the App with your mobile
terminal device can be found in the respective AppStore.
Observe the documentation for your mobile terminal device when
installing the App!

Establishing a connection✓ The testo 330i App must be installed on your mobile terminaldevice.
1. Switch on the measuring instrument: Press thekey.

-Connecting mode is activated: The blue LED flashes (0.5 s on /0.5 s off).
2. Start up the testo 330i App on your mobile terminal device.
- The Find Device function is activated: Any compatible devices
within radio range are displayed (product name + last 4 digits ofthe device serial number).
3. Tap instrument designation to select a measuring instrument.
- It may take around 30 s to establish the connection.
- A Bluetooth® connection is established: The blue LED isconstant.
If it is not possible to establish a connection, various
corrective measures are displayed. If you are unable toimplement these successfully, please refer to the section
Tips and assistance and/or contact your dealer or Testo
Customer Service. For contact details, please visit
www.testo.com/service-contact.
-The first time a connection is established with a measuring
instrument the Country version function opens.

3

1 Commissioning
The Country version configuration affects the measurement
parameters, fuels, fuel parameters, and the basis of and formulas
for calculations activated in the measurement box.
4. Tap the Country version to be activated.
5. Tap Continue to perform a restart.
6. Restart the testo 330i App.
7. Tap instrument designation to select a measuring instrument.
- The Flue gas function opens.
- The instrument is operational.

1.3.

Further information
It is paramount that you read and observe the warning and safety
instructions in the Safety and the environment section before using
the measuring instrument!!
Tutorial
A description of the most important operating controls and functions
can be opened in the App.

Instruction manual
For further information please download the instruction manual as aPDF file from our website:
www.testo-international.com/330imanuals
The website can also be called up directly from your mobile
terminal device (internet connection required):>4| Help | Help online

2 Safety and the environment

2Safety and the environment
Use> Please read this documentation through carefully andfamiliarize yourself with the product before putting it to use. Pay
particular attention to the safety instructions and warning advice
in order to prevent injuries and damage to the products.
> Keep this document to hand so that you can refer to it whennecessary.
> Hand this documentation on to any subsequent users of theproduct.

2.1.

Ensure safety
Please note for transportation and use:
Strong holding magnets on the back of the instruments!
WARNING
Magnetic fieldMay be harmful to those with pacemakers.
> Keep a minimum distance of 15 cm between pacemaker andinstrument.
ATTENTION
Magnetic field
Damage to other devices!
> Keep a safe distance away from products that could bedamaged by the effects of magnetism (e.g. monitors,computers or credit cards).
For differential pressure measurements please note:
WARNINGDangerous mixture of gases
Danger of explosion!
> Make sure there are no leaks between the sampling point andthe measuring instrument.
> Do not smoke or use naked flames during measurement.

5

2 Safety and the environment

Please note when emptying condensate:
CAUTION
Condensate entering the gas path.
Damage to sensors and flue gas pump!
> Do not empty the condensate container while the flue gas
pump is in operation.
Safety related symbols on the instrument
Representation

Explanation
If the product is not used in strict compliancewith this documentation, the intended
protection may be impaired.
> Operate the product only as described inthis documentation.
> Please consult your dealer or themanufacturer if in doubt.

Fundamental safety instructions> Only operate the product properly, for its intended purpose andwithin the parameters specified in the technical data. Do not
use any force.
> Do not operate the instrument if there are signs of damage at
the housing, mains unit or feed lines.
> Do not perform contact measurements on non-insulated, liveparts.
> Do not store the product together with solvents. Do not use anydesiccants.
> Carry out only the maintenance and repair work on this
instrument that is described in the documentation. Follow theprescribed steps exactly. Use only original spare parts
from Testo.
>

Any further or additional work must only be carried out by
authorised personnel. Testo will otherwise refuse to accept
responsibility for the proper functioning of the measuring
instrument after repair and for the validity of certifications.

> Only use the device in closed, dry rooms and protect it from rainand moisture.

62 Safety and the environment> Temperatures given on probes/sensors relate only to themeasuring range of the sensors. Do not expose handles andfeed lines to any temperatures in excess of 70 °C unless they
are expressly permitted for higher temperatures.
> The testo 330 must be checked before commissioning for any
visible damage. Do not commission the testo 330 if there aresigns of damage on the housing, mains unit or supply lines.
Electrical risk.
> Dangers may also arise from the systems being measured or
the measuring environment: Note the safety regulations valid inyour area when performing the measurements.
Use distilled water, or alternatively mild solvents such as
isopropanol to clean the flue gas analyzer. If using
isopropanol, please refer to the instruction leaflet for theproduct. Isopropanol fumes have a slight narcotic effect,
and typically cause irritation of the eyes and sensitive
mucous membranes. When using it, please ensure that
there is adequate ventilation.
Do not store any objects that have come into contact withsolvents and/or degreasers (e.g. isopropanol) in the case.
Evaporating or leaking solvents and/or degreasers may
cause damage to the instrument and to the sensors.
The use of strong or harsh alcohol or brake cleaner can
result in damage to the instrument.

For products with Bluetooth® (optional)
Changes or modifications that have been made without the explicit
consent of the responsible approval authority, may cause theretraction of the type approval.
Data transfer may be disturbed by equipment that uses the same
ISM-band, e.g. WLAN, microwave ovens, ZigBee.
The use of radio communication links is not permitted, among
others, in aeroplanes and hospitals. For this reason the followingpoints must be ensured before entering:> Switch off the device:> Isolate the device from any external power sources (mainscable, external rechargeable batteries, ...).

7

2 Safety and the environment

2.2.

Product-specific safety instructions
CAUTION
Acid in the sensors.
May cause chemical burns.
> Do not open the sensors.
Eye contact: Rinse the affected eye thoroughly under runningwater for 10 minutes, keeping the eyelids wide open andprotecting the unaffected eye. Remove contact lenses
wherever possible.
CAUTION
Acid in the sensor filters.
May cause irritation to the skin, eyes or respiratory tract.
> Do not open the sensor filters.
Eye contact: Rinse the affected eye thoroughly under runningwater for 10 minutes, keeping the eyelids wide open andprotecting the unaffected eye. Remove contact lenses
wherever possible.
Skin contact: Remove the injured person’s contaminatedclothing, ensuring self-protection. Rinse affected skin areas
under running water for at least 10 minutes.
Inhalation: Move to fresh air and make sure that breathing isunrestricted.
Ingestion: Rinse mouth out and spit out liquid. If conscious,
drink 1 glass of water (approx. 200 ml). Do not induce vomiting.

2.3.

Protecting the environment> Dispose of faulty rechargeable batteries/spent batteries inaccordance with the valid legal specifications.
> At the end of its useful life, send the product to the separate
collection for electric and electronic devices (observe localregulations) or return the product to Testo for disposal.
> The button cell used in the instrument contains 1,2Dimethoxyethane (CAS 110-71-4). See EC Regulation No.
1907/2006 (REACH) Art. 33.
>

8WEEE Reg. no. DE 75334352Testo SE & Co. KGaA
Testo-Straße 1, 79853 Lenzkirch
Telefon: +49 7653 681-0
Fax: +49 7653 681-100
E-Mail: info@testo.de
Internet: www.testo.com

0970 3340 en 06