MODEL 3340 G-LOGGERDATA ACQUISITION SYSTEM••••••••••••Start and stop data acquisitions with button and/or scheduleCollect information on peak
View a list of peak events and event data in any display mode
Add a GPS module to also collect GPS data
Uses internal battery power, external power packs or AC power
Compatible with Silicon Designs accelerometers
Automatic and manual calibration routines via +/-1G flip
Adjustable filters and FFT for live or post collection dataanalysis
Included software features familiar and convenient userinterface built on a LabView platform
Three input channels for three 1- axis modules or one 3-axis
module16-bit sample rates from 1 to 10,000 samples/second per axis
Automatic setup in less than 5 minutes upon wiring of Silicon
Designs accelerometer modules

SPECIFICATIONSPHYSICAL
Case Size
WeightCase Material

ENVIRONMENTAL
5.5" x 4.25" x 2.5"
Operating Temperature
425 grams / 15 oz. + batteries Storage Temperature
Die Cast Aluminum, Plastic
Humidity

OPERATIONAL
Connection
USB Connection
Memory TypeMax SD Card Size

25 Pin Female D-Sub
Micro USB (B)SD Card, Micro SD w/ Adaptor
32 GB

PC REQUIREMENTS
Operating Systems
Host Connection
Power Supply
Max Power Consumption
TCP/IP Remote Operation

0°C to +55°C (max)-40°C to +85°C (max)
0% - 90% Non-condensing

Windows 10, 8, 7, XP
USB2 Type A
USB/AC Power/AA Batteries
750 mW
Network Connection Req.

ZERO (DC) TO MEDIUM FREQUENCY APPLICATIONS





ADDITIONAL FEATURES• Real time data monitoring• Collect data in G or volts• Display shows from 100ms to 2 minutes of data
• FFT (Fast Fourier Transform) analysis is anadvanced feature usually found on much more
expensive DAQ systems• Independent scale G/Div settings expand or shrink
each channel’s input for better visibility• collection• PAUSE, RWD, FWD without interrupting data
• Optional offsets provide a staggered display for no overlap• Independent scale G/Div per channel• Oscilloscope (Sweep, Scope, Strip) and Volt Meter modes• Hide or show any or all of the 3 channels• View data from remote locations on network via TCP/IP• Optional offset setting per channel• Export time–stamped data to Excel, MatLab etc.

^W/&/d/KE^^h:ddK,E't/d,KhdEKd/^ŝůŝĐŽŶĞƐŝŐŶƐ͕/ŶĐ͘ͻϭϯϵϬϱEϭϮϴƚŚ^ƚƌĞĞƚ͕<ŝƌŬůĂŶĚtϵϴϬϯϰͻWŚŽŶĞ͗ϰϮϱͲϯϵϭͲϴϯϮϵͻ&Ăǆ͗ϰϮϱͲϯϵϭͲϬϰϰϲǁǁǁ͘ƐŝůŝĐŽŶĚĞƐŝŐŶƐ͘ĐŽŵ΀ƉĂŐĞϭ΁ϭϱͲ:ƵůǇͲϮϬϮϬ

MODEL 3340 G-LOGGERDATA COLLECTION

Data can be collected live or recorded for playback later. Modifying the horizontal scale expands or contracts theperiod of time displayed on the screen from 100ms up to 2 minutes. Each axis is one channel, and these can behidden or offset (but will still be recorded) as desired.

SPECTRUM (FFT)
SPECTRUM displays the FFT of the data. This is a more advanced feature of the G-logger 3340. You can analyze
the data to see at which frequencies the maximum vibrations are occurring.

SPECIFICATIONS SUBJECT TO CHANGE WITHOUT NOTICE
Silicon Designs, Inc. • 13905 NE 128th Street, Kirkland WA 98034 • Phone: 425-391-8329 • Fax: 425-391-0446
www.silicondesigns.com[page 2]15-July-2020



MODEL 3340 G-LOGGERPEAK VALUESThe METERS screen provides DC, RMS, and peak values in either Volts or Gs. These values are calculated over thetime interval selected by the horizontal scale selected on the Data tab. The values are updated at that same interval
as well.

CALIBRATE
The default calibration parameters are supplied automatically, or unit-specific calibration parameters can be
manually entered. Manual calibration can be done any time using gravity and performing a simple +/-1G flip.

SPECIFICATIONS SUBJECT TO CHANGE WITHOUT NOTICE
Silicon Designs, Inc. • 13905 NE 128th Street, Kirkland WA 98034 • Phone: 425-391-8329 • Fax: 425-391-0446
www.silicondesigns.com[page 3]15-July-2020

MODEL 3340 G-LOGGERPEAK EVENT REPORTINGREPORTINGtŚĞŶĞǀĞŶƚĚĞƚĞĐƚŝŽŶŝƐĞŶĂďůĞĚ͕ƚŚĞϯϯϰϬǁŝůůŝĚĞŶƚŝĨǇĂŶĚǁƌŝƚĞĞǀĞŶƚƐǁŚĞŶĞǀĞƌĚĂƚĂĂĐƋƵŝƐŝƚŝŽŶŽĐĐƵƌƐ͘dŚĞƉĞĂŬĞǀĞŶƚƐĂƌĞǁƌŝƚƚĞŶƚŽĂĨŝůĞŽŶƚŚĞ^ĐĂƌĚ͕ǁŚŝĐŚŝƐƚŚĞŶĂƵƚŽŵĂƚŝĐĂůůǇƐƉĞĐŝĨŝĞĚĂƐƚŚĞĞǀĞŶƚĨŝůĞƵƉŽŶƐĂǀŝŶŐ͘

CONNECTOR PIN LAYOUT
TOP ROW PIN NUMBERS
123CH 0CH 0CH 10 Volt
AON0 Volt
Bottom Row Pin Numbers

4CH 1AON

5CH 20 Volt

6CH 2AON

14CH 0AOP

17
CH 1
8-32 V18
CH 2AOP19
CH 2
8-32 V15
CH 0
8-32 V

16CH 1AOP78910111213XXXXXXX202122232425XXXX

X

X

Included 25 Pin D-Sub
Connector for Accelerometer
Connection

SPECIFICATIONS SUBJECT TO CHANGE WITHOUT NOTICE
Silicon Designs, Inc. • 13905 NE 128th Street, Kirkland WA 98034 • Phone: 425-391-8329 • Fax: 425-391-0446
www.silicondesigns.com[page 4]15-July-2020



MODEL 3340 G-LOGGER
CABLE WIRINGSPECIFICATIONS SUBJECT TO CHANGE WITHOUT NOTICE
Silicon Designs, Inc. • 13905 NE 128th Street, Kirkland WA 98034 • Phone: 425-391-8329 • Fax: 425-391-0446
www.silicondesigns.com[page 5]15-July-2020