NEWRIEGL VZ -2000i®

2.5 km
1.2MHzoptional

Based on a future-oriented, innovative newprocessing architecture, internet connectivity, and RIEGL’s latest waveform processing
LiDAR technology the RIEGL VZ-2000i LongRange 3D Laser Scanning System combines proven user friendliness in the fieldwith fast and high accurate data acquisition.
The processing architecture enables
execution of different backgroundtasks (such as point cloud registration, geo-referencing, orientation via
integrated Intertial Measurement
Unit, etc.) on-board in parallel tothe acquisition of scan data.
RIEGL‘s unique Waveform-LiDAR
technology enables such highspeed, long range, high accuracy measurements even inpoor visibility and demanding
multi-target situations anddelivers reliable data evenin harsh environments likeopen-pit mining.

NEW RIEGL VZ -2000i®

Long Range, Very High Speed
3D Terrestrial Laser Scanning System
Typical Applications• Topography and Mining • Natural Hazard Surveying • Construction Site Monitoring
Documentation • City Modeling • Tunnel Surveying • Civil Engineering • Research• Archeology & Cultural Heritagewww.riegl.com

RIEGL LMS GmbH, Austria

RIEGL USA Inc.

RIEGL Japan Ltd.

RIEGL China Ltd.

NEW RIEGL VZ -2000iat a glance®

RIEGL VZ-2000i Main Features• range up to 2,500 m, accuracy 5 mm
• orientation sensor for pose estimation• advanced flexibility through support for external peripherals and accessories,e.g. integrated GNSS unit for high accurate RTK solution, SIM Card slot for
3G/4G LTE, WLAN, LAN, USB• cloud connectivity via LAN, Wi-Fi, and 3G/4G LTE• easy to operate even in harsh environments (protection class IP64)• fully compatible with the RIEGL VMZ Hybrid Mobile Laser Mapping System• RiSCAN PRO standard processing software (included),RiMINING software package offering an optimized workflow
for open-pit mining (optional)

RIEGL VZ-2000i in the field

NEW Automatic On-board Registration
With two processors on-board, the RIEGL VZ-2000i is able to perform different
processes in real-time such as automatic on-board registration in parallel tothe scan data acquisition.
Processor 1

Processor 2• scan data acquisition• conversion of scan data into RIEGL
data base• acquisition of photographs

RIEGL VZ-2000i
monitoringmonnitoring in open-pit
open pit mining• on-board multiple time aroundresolution• pose estimation(using GNSS/IMU/environment
sensors)• registration of scan data as a
background process

data,, range-coloredRIEGL VZ-2000i
VZ 2000i sscancan data
range
g colored
first scan
position

next scan
positionbackgroundprocess

registeredpointcloud

RIEGL VZ-2000i Technical Data
2.5 km max. measurement

range

1.2MHzoptional camera
optional

pulse repetition
rate PRR

onlinewaveformprocessing

multiple target
capability

Laser Class 1Wi-Fi and3G/4G LTE

Laser Pulse Repetition Rate PRR (peak)

50 kHz

100 kHz300 kHz600 kHz

1,20000 kHz

Max. Effective Measurement Rate (meas./sec)

21,00042,000125,000

250,000

0,000
500,000

Max. Measurement Range (≥ 90 %)

2,500 m1,850 m

1,100 m

800 m

00 m
600

Max. Measurement Range (≥ 20 %)1,300 m

950 m

540 m

380 m

29090 m

2m

1.5 m

1.5 m

1.0 m

1.00m

Minimum Range

Accuracy / Precision

5 mm / 3 mm

Field of View (FOV)100° vertical / 360° horizontal

Eye Safety Class
Main Dimensions (width x height) / Weight

Laser Class 1 (eyesafe)
206 mm x 308 mm / 9.8 kg

Further details to be found on the current RIEGL VZ-2000i Data Sheet.

Copyright RIEGL Laser Measurement Systems GmbH © 2017– All rights reserved.
Use of this data sheet other than for personal purposes requires RIEGL’s written consent.
This data sheet is compiled with care. However, errors cannot be fully excluded and alternations might be necessary.

www.riegl.com
Preliminary Info Sheet, RIEGL VZ-2000i, 2017-09-18