Flue gas analyzer

HT-1210N/1210NT

11CO
H2補償付℃

0 〜 10000ppm

0 〜 650℃

OptionIR PrinterHT-1610Roll paper x 1AAbattery x 4Printer paper
HT-1636

５rollsAmbient airtemperature sensor
HT-1321

Measurement item
1210N ; CO
1210NT; CO・Temperature

H2 Compensation as standard functionMore accurate COfunction.

measurement by mean of H2 compensation

CO Peak・Average・Alarm function
Auto measurement of CO peak value and CO average value.
3 steps alarm both indication at display and beep sound.

Attach caseHT-1315

Made by aluminium
Size
350×460×155mm
2.9kg

Measurement
Software for PC
Online View 2000
HT-2084

RS232 transfer cable
USB cable
Recommended OS：
Windows7/8.1/10

Measurement
Software for PC
Online View 2000
HT-2094

RS232 transfer cable
Recommended OS：
Windows 7/8.1/10

Battery charge forabroadHT-1318

AC100 〜 240V

Sampling probe
HT-1001A

L=300mm φ6mm
Inc.. condensate trap
Flue gas temp.;
0 〜 650℃

Probe handle
HT-7201A

Sampling hose length
1800mmInc. condensate trap

Probe tube
HT-7231

L=180ｍｍ φ5mm
Flue gas temp.;
0 〜 650℃

Probe tube
HT-7235

L=500ｍｍ φ6mm
Flue gas temp.;
0 〜 650℃

Probe tube
HT-7232

L=750ｍｍ φ6mm
Flue gas temp.;
0 〜 650℃

Probe tube
HT-7233

L=750ｍｍ φ8mm
Flue gas temp.;
0 〜 1100℃

Integrated Data logger
Data storage up to 100 data.

0 〜 100℃
For air temperature

12

Speciﬁcation

Measurement items
CO＊1（H2 compensation）

TemperatureＧ（K. thermocouple）

Measurement range
Accuracy
Resolution
Response time
Measurement range
AccuracyStandard equipment ※Select one of probes
0 〜 10000ppm（0.000 〜 1.000％）
Measured value=0 〜 200ppm：±10ppm
Measured value=200ppm 〜：±5％reading
1ppm（0.001%）<30sec.
0 〜 650℃
0 〜 1100℃（depend on probe tube）±1℃ or ±1%（0 〜 650℃）reading＊2±2% reading （650.1 〜 1100℃）
0.1℃（0 〜 999.9℃） 1℃（1000 〜 1100℃）
0 〜 100℃±1℃
0.1℃

Integrated

Pump, Interface for PC (RS232)
Data logger (possible to store 100 data), Interface for printerAccessory
Battery, Sampling probe, Condensate trap, Reﬂection plate, Carrying case, Softcase, Filter set
Spec. / remarks
Description
Art. No.
Sampling probe/HT-1210N
HT-1229D L=180mm φ5mm, with condensate trap, w/o temp.measurement
Sampling probe( with ﬂue gas temp. sensor)/HT-1210N HT-1006 L=180mm φ5mm, with condensate trap, ﬂue gas temp.0 to 650℃

OptionCO（H2conpensation） Electrochemical
Temperature GK thermocouple
Temperature APt2000Ω

ProbesLong probe
HT-1235D L=690mm φ6mm, with condensate trap, w/o temp.measurement
Sampling probe
HT-1001A L=300mm φ6mm, with condensate trap, ﬂue gas temp.0 to 650℃
Ｌ-ProbeHT-1238D L=120mm φ5mm, with condensate trap, w/o temp.measurement
Ｌ-ProbeHT-1006L L=120mm φ5mm, with condensate trap, ﬂue gas temp.0 to 650℃
Probe handle
with hose, condensate trap
Probe handle
HT-7201A
Probe tube（for probe handle HT-7201A）
L=180ｍｍ φ5mm, ﬂue gas temp.0 to 650℃
Probe tube
HT-7231 L=500ｍｍ φ6mm, ﬂue gas temp.0 to 650℃
（With ﬂue gas temp. sensor）
HT-7235 L=750ｍｍ φ6mm, ﬂue gas temp.0 to 650℃

Main unit

Temp. probe

TemperatureＡ（Pt2000Ω）＊3

Resolution
Measurement range
Accuracy
Resolution

SensorTemperature
DisplaySize
WeightPowerOperating：0℃ 〜 ＋45℃
Storage：−20℃ 〜 ＋60℃
Dotmatrix 4 lines（W×H×D）80×150×35 mm
Approx.330g
Battery charger（AC100V 50/60Hz DC12V 100mA）
Integrated Nicd rechargeable battery（Max.8hours continuous operation）

HT-7232 L=750ｍｍ φ8mm, ﬂue gas temp.0 to 1100℃
HT-7233 φ3×130L, 0〜950℃,
Kthermocouple HT-1251a for air/liquids
φ1.5×130L, 0〜950℃,
HT-1252a for air/liquids
φ3×130L, 0〜400℃,
HT-1253a for air/liquids/food, centric top
130L, 0〜400℃,
HT-1254a for surface /rifts/air/liquid
φ4×130L, 0〜650℃,
HT-1255a for surface /air/liquid0〜450℃, magnetic probe for surface
HT-1256a 0〜180℃, Pliers probe for plates, tube
HT-1257a 0〜100℃, for air
HT-1382 0〜100℃
HT-1321 AC100〜240V
HT-1318 Aluminum, dimension : 350×460×155,weight : 2.9kg
HT-1315 Online View 2000
HT-2094 （Recommended OS：Windows 7/8.1/10）

Air temp.probe
Pt2000Ω
Ambient air temp. sensor Pt2000Ω
Battery charger
Hard case
Measurement software（with RS232cable）
Online View 2000
Measurement software
HT-2084 （Recommended OS：Windows 7/8.1/10）（with RS232 cable・USB conversion cable）
Roll paper ×1 AA battery× 4
Infrared printerHT-1610 5 rolls
Roll paper for printerHT-1636＊1：Compensate the measured value when measure ﬂue gas include H2.
＊2：The higher value is applied as accuracy.
＊3：Ambient air temp. (optional items) sensor is needed.
＊Accuracy is based on cylinder gas.

Speciﬁcation might be changed without notice.

HT-1210N
Standard set

HT-1210NT
Standard set

・Main unit・Sampling probeHT-1229D
・Battery charger
・Carrying case・Softcase・Reflecting plate
・Filter set
・Operating manual

・Main unit・Sampling probeHT-1006
・Battery charger
・Carrying case・Softcase・Reflecting plate
・Filter set・Operating manualPrice 140,000 yen

Price 154,000 yen（tax excluded）

13(tax excluded)

Flue gas analyzer

HT-1210N/1210NT
Dimension

HT-1210N
φ61.6
Inlet port gas

Charge LED

Reﬂection plate
35

Battery charger port
13.0Interface RS232

φ5.2

Ambient temp. sensor port

φ32
K

IR printer Interface

φdL

Gas inlet
Gas sampling cable
Metal cone

FilterCondensate trap

150169Silica gel

L(mm) K(mm) φd(mm) Sampling hose length(mm) Flue gas temp.（℃）
Art.No.
5−HT-1229D 0 〜 180 315
1500
6−HT-1235D 0 〜 690 7901500Description
Sampling probe
Long probe

80HT-1210NT
Inlet port gas

φ61.6

Charge LED

35

Battery charger port
Reﬂection plate

Interface RS232
Ambient temp. sensor port
13.0

φ5.2

IR printer Interface

φ32
KφdL

Gas inlet
Flue gas temp.sensor
Metal cone

Gas sampling cable
FilterDescription
Sampling probe

150

Condensate trap

169

Silica gel

Art. No. L(mm) K(mm) φd(mm) Sampling hose length(mm) Flue gas temp.（℃）
50 〜 650
HT-1006 0 〜 180 290
1800

80
Flue gas temp. connector
Flue gas temp.compensation cable

14