ＬＥＤ Light Source

LUMINAR ACE
Model LA-HDF5010RL
LA-HDF7010RL

Ｉｎｓｔｒｕｃｔｉｏｎ Ｍａｎｕａｌ
Thank you for purchasing our product.
Prior to using it, please read the instruction manual thoroughly forsafe and correct operation.
Be sure to keep the instruction manual carefully at hand.

ＬＥＤ ＯＮＰＯＷＥＲ

ＲＥＭＯＴＥ

４
２

６

８

ＬＥＤ
ＬＥＤ ＯＦＦＭＡＮＵＡＬ
０

ＴＥＭＰ．
ＦＩＮＥ

１０

ＬＩＧＨＴ ＣＯＮＴＲＯＬＡＬＡＲＭ

HAYASHI-REPIC CO.,LTD.
E-1

This LED light source is used for the illumination of image processing, microscope,science observation.

CAUTION
Caution labels and statements provided in this manual are the description of potentiallyhazardous situation which if not avoided could result in moderate or minor injury.
It could also advise safety precautions against unsafe practices.

CAUTION
Please carefully follow the instruction as misuse of the product may lead an accident of fire,electric shock or malfunction of the product.

CAUTION
Please use the light guide specifically provided for the product by manufacturer.
It may cause malfunction or insufficient performance of the product in case you use no lightguide or non-specified light guide.

CAUTION
Please use the AC power supply included.
When using something other than the included AC power supply,
please use the high-performance products more than accessory.

CAUTION
AC power supply cable must be properly grounded.
It may lead an accident or malfunction of the product if you do not make the proper ground.

CAUTION
Do not move while it is being illuminated.
Do not drop, or give a shock to the light source apparatus. (A trouble may result)
Do not disassemble or modify the light source apparatus. (A trouble may result.)

1. Accessories
Accessories
Options

Instruction Manual, Ethernet installation instructions,Command manual, AC power supply（AC100V）
AC power supply:
US type (AC120V), UK type (AC240V), European type (AC230V)

E-2

2. Part Names
Front Panel

Light Guide Socket
This is a socket to introduce a light from the LED lamp built
in this apparatus to the light guide.
* Please Using Hayashi standard light guide.
Control function change switch
Front switch [UP：REMOTE / DOWN：MANUAL]* Regarding the switching of control function, please see to[ 3. Operation method ].

ＬＥＤ ＯＮＲＥＭＯＴＥ

ＰＯＷＥＲ

４
２

６

８

ＬＥＤ
ＬＥＤ ＯＦＦＭＡＮＵＡＬ
０

ＴＥＭＰ．
ＦＩＮＥ

Light quantity control knob (LIGHT CONTROL)This knob is used when controlling brightness of the LED lamp
manually. A light quantity is increased by turning the knob in theclockwise direction.

１０

ＬＩＧＨＴ ＣＯＮＴＲＯＬ

ＡＬＡＲＭ

Indicator
POWER（Green）：When the power ON, the light up.
LED（Orange）：When an abnormality occurs for LED ramp, the light up.
TEMP.（Red）：When an abnormal temprature rise by inside the light box,
the light up.
* LED ramp light is turn off.
Current adjustment volume (FINE)
For maintenance, so usually do not touch.
LED ON/LED OFF change switch
This function is enabled when the control function change switch is "MANUAL".
LED ON ： LED ramp light is light up.
LED OFF ： LED ramp light is turn off.
* Even LED is switched off, the cooling fan still remains to work as the main power supply is
active yet.

Rear Panel

REMO TEIN

・INPUT connector (REMOTE IN)
・RS232C connector
・LAN port
This is a connector for external control. Please connect with afunction to be used.

RS232CL A N

SERIAL

L A N

ON

ＡＣ１００‑２４０Ｖ

RS232CSTANDARD

１
２

P O W E R

Power Switch (POWER)
This switch is used to turn ON/OFF the power.
Control function change switch
・Rear switch 1 UP：SERIAL DOWN：STANDARD
・Rear switch 2 UP：LAN
DOWN：RS232C* Regarding the switching of control function, please see to[ 3. Operation method ].
AC inlet
It is an AC power cord socket.
*When the input voltage is other than AC100V, please use anotherpower supply cord provided as an option.

E-3

3. How to use
1) Insert the light guide in to the light guide socket of the apparatus.
2) Please insert the power supply into the AC inlet.
3) Switch the control function selector switch to the desired control function.
Controlfunction

MANUALREMOTERS232C

LAN

Front switch

MANUALREMOTE--

Rear switch 1

STANDARD

STANDARDSERIALSERIAL

Rear switch 2--

RS232CLAN

Switch・ MANUAL ： Manual light quantity control (using light control knob at Front panel).
・ REMOTE ： Remote light quantity control (Input the 8 bit parallel signal)
・ RS232C ： Remote light quantity control （256 gradation command input from 0 to 255）
・ LAN

： Remote light quantity control （256 gradation command input from 0 to 255）

* Be sure to turn off the power supply switch before switching the control switch.
4) Insert the plug of the AC cord in the outlet. (Please take the ground)5) Turn on the power switch.
6) Control the light quantity to your desired level to operate the lighting apparatus.
* LED lamp may slightly light on during minimum light volume operation.
* LED lamp may be un- stable during minimum light volume operation.
7) To finish, turn off the power switch.

CAUTION
Avoid looking irradiation light through light guide or light source.
It may cause the pain of eyes or the sight injury.

E-4

4. External Control
4.1 REMOTE(parallel communication ）The interface (INPUT) allows an external control that turns on and off the product and adjust the lightvolume as the SELECTOR SWITCH is set REMOTE.
* Use as shorter shielded cable(2m max) as possible for the external control. Long cable may result in anoise interfusion, and a noise may hinder the product from working properly as the cable is placed near
the source of noise.

PIN ASSIGNMENT for the interface(INPUT)

Interface: Dsub female(socket), 25pins* An external control I/O interface is not provided.

※ Example of the recommendation for the external control I/O connector
Connector : Japan Aviation Electronics Industry model Co., Ltd. : DB-25PF-N
Resin hood ： Japan Aviation Electronics Industry DB-C8-J10-F1-1R us Co., Ltd.
These items are also available from us. Please contact with your sales representative.

CAUTIONThe 2nd pin, the 3rd pin, the13th pin, the 17th pin and the 24th pin are pins for services work.
Never connect those pins by yourselves. It will cause malfunction.

Pin No.
1
2
3
4
5
6
7
8
9
10
11

PIN ASSIGNMENT
LED abnormal output

No conect
No conect
Temperature error output

Light quantity control 1
Light quantity control 2
Light quantity control 3
Light quantity control 4
Light quantity control 5
Light quantity control 6
Light quantity control 7

Pin No. PIN ASSIGNMENTLight quantity control 8
12No conect
13
COM
14
COM
15
COM
16
No conect1718
19Supply power supply＊1
20
21
SET signal
22

E-5

PIN ASSIGNMENT
Pin No.
Supplypower supply ＊1
23
No conect
24
LED lamp on/off
25＊1. 18,19,20,21,23 are common

1) LED lamp on/off function
It is a function to turn on / off the LED lamp.
It is valid during the control function selector switch is set to "REMOTE setting".
It is turned on by shorting the No.25(REMOTE IN) and No.23(GND(-)) pins at the
INPUT connector.
Examples of connecting

IN P UT c o nn ec t or

I N PU T c on ne c to r

Ds u b co n ne ct o r ma l e( Pi n )D su b c on n ec to r f em a le (S o ck et )

23 23
＋
−

resistance

DC power supply
Ｄ Ｃ５〜 ２４Ｖ
25 25

photocoupler

switchSwitch short:light on
Switch open :light off

LED light source
insideou ts id e

( Ex .1 )Us e o f Sw i tc h

I N PU T c on n ec to r

IN P UT c o nn e ct or

D s ub c o nn e ct or ma le ( Pi n )Ds u b co n ne c to r f em al e (S o ck et )

23 23
＋
−
Input signal

resistance

DC power supply
ＤＣ ５〜２ ４Ｖ
transistor

25 25

photocoupler

resistanceH(1) input signal:
L(0) input signal:

light onlight off

LED light source
insideo ut s id e

(E X . 2 )U s e o f t r a ns i s t or

E-6

2) Light quantity control (REMOTE)
It is valid during the control function selector switch is set to "REMOTE setting".
The following table shows a connection example and the light quantity at input.
The light quantity can be changed by entering SET signal(<1ms pulse signal) with ainterval more than 2ms after confirming the light quantity adjustment signal 1-8.
IN PU T c on n ec t or
D s ub c o nn e ct or ma l e( Pi n )−D su b c o nn ec t or ma le ( Pi n )I NP UT co n ne ct o r
Ds ub co n ne ct o r f em al e (S o ck et )

23 23

23 23

21 21

21 21

DC power supply
20 20
ＤＣ５ 〜２４ Ｖ＋−

DC power supply
20 20
ＤＣ５ 〜２４ Ｖ

19 19

19 19

18 18

18 18

16 16

SET signal

IN P UT co nn e ct o r

Ds u b co n ne c to r f em a le (S o ck e t)ＣＯＭ（ＧＮＤ）

16 16

15 15

15 15

14 14

14 14

switchresistance

22 22
photocoupler

SET signal＋

IN PUT c o nn ec t ortransistor

ＣＯＭ（ＧＮＤ）

resistance

22 22
resistancephotocoupler

resistance

12 12

７

11 11

６

10 10

５

９９

４

８８

３

７７

２

６６

１

５５

photocouplero u ts i d e( Ex . 1 )The light quantity adjustment signal

The light quantity adjustment signal

switch８

LED light source
insidetransistor

８

resistancephotocoupler

７

11 11

６

10 10

５

９９

４

８８

３

７７

２

６６

１

５５

o u ts i d e

U s e of Sw i t c hresistance

12 12(E X .2 )LED light source
insideUs e o f t r an s i s to r

Note 1.

Please observe the polarity of the supply power supply output at the time of the connection.
LED light supply and the power supply may be damaged by the wrong polarity.

Note 2.

The external power supply the range is DC +5V - DC +24V.
LED light supply may become unstable or be damaged if it is out of the range.

Note 3.

Use the external power supply with the capacity of the output electric current morethan 100 mA.

E-7

*The Light intensity table at input
INPUTconnector

No.5 pin No.6 pin No.7 pin No.8 pin No.9 pin No.10 pin No.11 pin No.12 pin Pin No.
Light quantity（Low）（High）
Light quantity Min(0）
0
0
0
0
0
0
0
0
1
0
0
0
0
0
0
0
0
1
0
0
0
0
0
0
1
1
0
0
0
0
0
0
0
0
1
0
0
0
0
0
1
0
1
0
0
0
0
0
Light quantity UP
0
1
1
0
0
0
0
0
Course
1
1
1
0
0
0
0
0
↓
0
0
0
1
0
0
0
0
1
0
0
1
0
0
0
0
0
1
0
1
0
0
0
0
1
1
0
1
0
0
0
0
0
0
1
1
0
0
0
0
1
0
1
1
0
0
0
0
0
1
1
1
0
0
0
0
1
1
1
1
0
0
0
0
：
：
：
：
：
：
：
：
Light quantity Max
1
1
1
1
1
1
1
1
0 ： Short circuit
1 ： Open3) LED abnormal output
In the case of abnormality such as the opening circuit of a LED, it will internally make a short circuit
between the first pin of the INPUT connector (REMOTE IN) and the 14th pins.
The LED (orange) of the front indicator also turns on.
If the LED abnormality output appears, stop to use it and ask the manufacturer for repair.
１１

ＰＣＣ
LED abnormal signal＊Abnormal：SHORT＊Normal ：OPEN
ＰＣＥ

14 14
15 15
16 16
LED light source
insideo uts i d e4) Temperature error output
When the temperature in the device rose abnormally, it will internally make a short circuit between the
4th pin and 15th pin of Input connector (REMOTE IN).
LED lamp will be light off and the indicator TEMP(red) in the front will be also light on.
If the temperature abnormality output appears, Stop the operation and contact the manufacturer.
４４

14 14

ＰＣＣ
Temperature error signal＊Abnormal：SHORT＊Normal ：OPEN
ＰＣＥ

15 15
16 16
LED light source
insideo uts id eE-8

4.2 RS232C、LAN(Serial communication）
1) Communication specification

Communications protocol
Baudrate
Data length
Parity bit
Stop bit

RS232C
9600bps
8bit
none
1bit

* All the data for transmission and reception except STX and ETX should be
half-width capital letter. (ASCII code)

2) Connector specification

・RS232C Connector* Use a straight cable to connect to a PC.

PIN ASSIGNMENT for the interface(RS232C) Pin No.
5

9

1

6

Interface: Dsub female(socket),25pins* An external control I/O interface is not provided.

E-9

1
2
3
4
5
6
7
8
9

SignalWiring insidecontent
light source
No connect
RXD To light source reception
TXD To light source transmitter
Short circuit of 4pin - 6pin
Grand
GND
Short circuit of 7pin - 8pin
No connect-

・LAN port (X-PORT produced by Lantronix)* Use a crossing cable to connect to a PC and use a straight cable to connect to the hub.

Right LED

Left LED
No light:No connection
Green light :100BASE-T connection state
Orange light:10BASE-T connection state

No light:No communication
Green light :Full duplex communication state
Orange light:Half duplex communication state

CONTACT 8

CONTACT 1

Default value

Item
Communications protocol TCP/UDP

IP address
Subnet mask
Default gatewayPort

Set value
TCP
192.168. 0.101
255.255.255. 0
0. 0. 0. 0
10001

Note
At the factory
At the factory
At the factory
At the factory
At the factory

The initial setting of the PC side is necessary to operate with the PC.
Please see "Ethernet introduction procedure for initial setting"
Please see "command manual" for the commands in detail.

E-10

5. Installation
CAUTION
Do not use the product in an enclosed area, where may result in malfunction.
Do not Install to a product upside down.
Make room for ventilation so that the product's vent hole is not blocked off.
* Make sure an open upper space.
60mm and up

Ｉ ＥＣ ６ ２ ４７ １

ＲＩＳＫ ＧＲＯＵＰ ２

Ｉ ＥＣ ６ ２ ４７ １

ＷＡＲＮＩＮＧ

50mm and up

ＲＩＳＫ ＧＲＯＵＰ ２

ＷＡＲＮＩＮＧIR emitted from this product.
Do not stare at operating lamp.

IR emitted from this product.
Do not stare at operating lamp.

ＣＡＵＴＩＯＮＣＡＵＴＩＯＮ

Possibly hazardous optical radiation emitted
from this product.Do not stare at operatinglamp.May be harmful to the eyes.

Possibly hazardous optical radiation emitted
from this product.Do not stare at operatinglamp.May be harmful to the eyes.

50mm and up

50mm and up

CAUTION
The AC power supply plug or AC inlet should be set for quick disconnection for emergency case.
It is customer s responsibility to take care of the emergency case if the product assembled in thecustomer s system.

CAUTION
Do not use in the place exposed to dust, oil, or iron powder, where a liquid or powder issplashed. It causes the trouble.
Do not install the LED light source near a strong electromagnetic field.

E-11

6. Outer dimensions drawing
ＩＥＣ６２４７１

ＲＩＳＫ ＧＲＯＵＰ ３

IR emitted from this product.
Do not stare at operating lamp.

ＷＡＲＮＩＮＧ

Possibly hazardous optical radiation emitted
from this product.Do not stare at operatinglamp.May be harmful to the eyes.

ＣＡＵＴＩＯＮR EMOTEINR S232CL A N

S E RI A L

L A N

ＲＥＭＯＴＥＭＡＮＵＡＬ

ＰＯＷＥＲ

４

109mm

ＬＥＤ ＯＮＬＥＤ ＯＦＦ

６

２

P O W E R

RS232CSTANDARD

155mm

ON

１
２

ＡＣ１００‑２４０Ｖ

８

ＬＥＤ
０

ＴＥＭＰ．
ＦＩＮＥ

１０

ＬＩＧＨＴ ＣＯＮＴＲＯＬ

ＡＬＡＲＭ

LA‑HDF5010RL:265mm
LA‑HDF7010RL:315mm

120mm※ Bottom surface:LA‑HDF7010RL

7. Specifications（Subject to change without prior notice due to product improvement）

Input rated Voltage AC100V-AC240V ±10％
Input rated frequency 50/60HzPower Consumption LA-HDF5010RL: 95W / LA-HDF7010RL: 155W
LED ColorWhite
LA-HDF5010RL: 1,400,000Lx *1
Illuminance
LA-HDF7010RL: 1,800,000Lx *1
Color Temperature(typ.) 6,500K (typ.) *2
LED Lifetime

30,000H *3

Lightquantity control

MANUAL ： Consecutive dimming by adjusting the light quantity control knob.
REMOTE ： 8 bit digital light quantity control
ＲＳ２３２Ｃ ： 256 gradation command input
LAN ： 256 gradation command input

Function

1) LED lamp ON / OFF function : External input and Front switch
2) LED abnormal output : External output and Front indicator3) Temperature error output : External output and Front indicator

Cooling System

Fan

Environmental
Conditions
Dimmension
WeightAmbient Temperature ： 0℃˜ +40℃
Ambient Humidity ： 20%˜ 80%RH （No dew condensation）
Indoor Use, Altitude up to 2,000m
LA-HDF5010RL: 120mm(W)×163mm(H)×265mm(D) （except protrusions）
LA-HDF7010RL: 120mm(W)×163mm(H)×315mm(D) （except protrusions）
LA-HDF5010RL: 4.0kg / LA-HDF7010RL: 4.6kg*1. Illuminance ： L=500mm, reference value measured at 50mm distance from the 11Φ optical fiber end.（reference value）*2. Color temperature. (reference value)*3. LED Lifetime ： The lifetime may differ as you use the product under specific environment and temperature.

E-12

8. Trouble shooting
Check again before asking for repair.
Malfunctions

The LED cannot be
illuminated even if the poweris turned on.

Check

Handing

A power supply cord may be
unplugged.

Plug in.

A light quantity control knob
may be set on zero level.

Turn the control knob (LIGHT
CONTROL) to adjsut the light volume.

The light quantity cannot be Is the control function selector
Set the control function change switch tocontrolled by turning the light switch set to a setting other
MANUAL.
quantity control knob.
than MANUAL?
Phenomenon:
The temperature abnormality
output appeared.

The temperature might be
Use it within the operational temperature
beyond the limit of therange. The device can be reset after
operational temperature range cooling the device by turning off thedue to in the closed place etc. power supply.

CAUTION
In case of malfunction, disconnect the power supply immediately.
The repair service should be performed by manufacturer.
The product must be shipped to manufacturer for repair.

CAUTION
This product contains no replaceable parts. Please send the product to manufacturer in case youneed replacement for such as LED lamp etc.

E-13

9. After-sale services☆ When you ask for a repair, let us know the following:
・ Model Name : LA-HDF5010RL or LA-HDF7010RL
・ Serial Number : Inscribed on the serial number label pasted to this product.
・ Details of trouble : Detailed information such as a period of use, the operating
condition at the time of trouble.
・ Date of purchase : Date of purchasing this product.
・ Your name, your company name, address, phone number and fax number.

HAYASHI-ＲＥＰＩＣ CO.,LTD.
Division-1
1-28, Kitaohtsuka, Toshima-Ku, Tokyo 170-0004 JAPANTEL : +81-3-3918-5623 FAX : +81-3-3918-5683

KORIN ELECTRONIC CO.,LTD.
Room 1005, Fook Yip Bldg., 53-57 Kwai Fung Crescent, Kwai Chung,
HONG KONG
TEL : (852)-2418-1122 FAX : (852)-2418-1235

HAYASHI TRADING (SHENZHEN) CO.,LTD.
Room 27B07, 27F, Jueshi Building, 4018 Jiabin Road,
Luo Hu District, Shenzhen 518001, CHINATEL : (86)-755-3685-6361 FAX : (86)-755-3685-6364

2020.05.08

IM-135-Rev.2

E-14