ＬＥＤ Light Source

LUMINAR ACE
Model LA-HDF5010RL
LA-HDF7010RL

Command Manual

HAYASHI-REPIC CO.,LTD.

J-1

１．Command List
Setting Command
Set the Light Quantity Value

ＬＥＤ OFF
Memory the Light Quantity ValueNG Signal ResetExternal ON/OFF Control

Command No.

STX
STX
STX
STX
STX
STX

ModeW
W
W
W
W
W

Command No.

STX
STX

Mode
RR

Check Command
Check the Light Quantity Value

Check the Status1414
10080000

1408

Unit No.
000000000000

Data
****1
****000000
00000
00001
00000

CS
CS
CS(08)
CS(0F)
CS(08)
CS(07)

ETX
ETX
ETX
ETX
ETX
ETXUnit No.
00
00

Data
00000
00000

CS(07)
CS(0A)ETX
ETX・ Response to Setting Command and Check Command
In the case of Set Command
If Signal Control is OK
If Signal Control is NG

STX
STX

If Signal Control is OK, Output the ACK. If the Signal Control is NG, Output the NAK.

ModeWW

Command No.

Unit No.
00
00

ACK
NAK

CS
CSETX
ETX

ModeR
R

Command No.

Unit No.
00
00

Data
0****000

CS
CS

ETX
ETX

Outqut the NAK
Mode Command No. Unit No.
R
1400

NAK

CS

ETX****In the case of Check Command
Check the Light Quantity Value

Check the Status

STX
STX

In the case of Signal Control is NG
If the Signal Control is NG

STX

1408

２．About Control Contents by Command
Setting Command (Mode is W）
Command No.
Set the Light Quantity Value It Sets the Light Quantity Value and the LED ON.
14
ＬＥＤ OFF
Turn OFF the LED.
14
MemorytheLightQuantity
Value
It Memorize the Light Quantity Value.
10NG Signal Reset
Reset the NG Signal of LED and Temperature Sensor.
08
External ON/OFF Control It is a Setting to Control the LED ON / OFF by a External Signal.
00
Check Command (Mode is R)
Command No.
Check the Light Quantity Value Check to the Light Quantity Value.
14
Check the Status Check to the NG Signal of LED and Temperature Sensor.
08・ About the Command Format
FormatSTX

ModeCommand No.

Unit No.

DataCSETXSTX (0x02), ETX (0x03) are ASCII Control Codes, CS is a Checksum. (How to get CS will be Described Later)

The Command is ASCII, Described in the Half-Width Capital Letter
The Data is ASCII, Described in the Half-Width Capital Letter.
It does not use the Unit Number. Please Set to the "00".

J-2

３．Command Detail
Setting Command (Mode is W）

Data
STX
Mode Command No. Unit No.
CS
****1
STXW
1400
CS
**** is Light Quantity Value
The Light Quantity Value is 256 Gradation from 0 to 255.
For example, if the Light Quantity Value is 100, Please Set to the "0100".
The "1" of "**** 1" Indicate the LED ON.

・ Set the Light Quantity Value

・ LED OFF

STX
Mode Command No.
STXW
14**** is Light Quantity Value
The "0" of "**** 0" Indicate the LED OFF.

Unit No.

STXSTX

Command No.

Unit No.

1000

・ Memory the Light Quantity Value

ModeW00

ETXETX

Data****0CSCSETXETX

Data
00000

CSCS

ETXETXMemorize the Light Quantity Value. Next time, If Power ON, It light up with this Memorized Light Quantity Value.

If the Light Quantity Value is Changed, the Memorized Light Quantity Value will be Reset.
After Reset, If it was not Memorized by the time of Power OFF,The next time the Power on, It does not LED ON.

Data
STX
Mode Command No. Unit No.
CS
ETX
00000
STX
W
0800
CS
ETX
It is Reset the NG Signal (LED, Temperature Sensor NG). After Removing the NG Factor, Please run.
・ NG Signal ResetNote) If it arise the NG Signal, the Output a Signal and turn ON the alarm LED in front panel of Light Source.

Regarding Connection with External Equipment, Please refer to the Instruction Manual.

DataSTX
Mode Command No. Unit No.
CS
ETX0000*STXW0000
CSETXNote) The Product is set to "Do not Perform External ON / OFF Control" as at the Factory Shipped.
About the ON / OFF Control from External, Please setting this.
If the ON / OFF control is run from the External, Please Setting the Data "00001".
If the ON / OFF control is not run from the External, Please Setting the Data "00000".
If the ON / OFF control is run from the External, Please Setting the REMOTE/MANUAL Switch on theFront Panel of Light Source to "REMOTE". This setting is Keeped. Please operate once if necessary.

・ External ON/OFF Control

・ Response to Command
STX
Mode Command No. Unit No.
CSETX
Set the Light Quantity Value
ACK/NAK
STX
W
1400
CSETX
ACK/NAKＬＥＤ OFF
STX
W
1400
CS
ETX
Memory the Light Quantity Value
ACK/NAK
STX
W
1000
CS
ETX
ACK/NAKNG Signal Reset
STX
W
0800
CSETX
External ON/OFF Control
ACK/NAKSTXW0000
CSETXNote) If Signal Control is OK, Output the ACK(0x06). If the Signal Control is NG, Output the NAK(0x15).

Check Command（Mode is Ｒ）
J-3

・Check the light quantity value
STX
ModeSTX
R
Response toSTX
Mode
STXR

Command No.

Unit No.

1400

Command No.

Unit No.

1400

Data
00000

CS
CS

ETX
ETX

Data
****

CS
CS

ETX
ETX

**** is Light Quantity Value. If the setting Light Quantity Value is "100", it Indicate the "0100".

・ Check the Status
STX
STX

ModeR

Command No.

Unit No.

0800

Data
00000

CS
CS

ETX
ETX

Response toData
STX
Mode Command No. Unit No.
CS
ETX
*000
STX
R
0800
CSETX
"*" Indicates the Content of the NG.
NG Contents is Indicate in BIT units. "0 BIT" is Temperature Sensor, "1 BIT" is LED.
"00" = OK for both Temperature Sensor and LED, "01" = Temperature Sensor NG,"10" = LED OK, "11" = NG for both Temperature Sensor and LED"*" Is an ASCII indication, "0": OK for Both Temperature Sensor and LED, "1": Temperature Sensor NG"2": LED NG, "3": OK for Both Temperature Sensor and LED

If the Signal Control is NG, Output the NAK(0x15).
STX
Mode Command No.
STXW
14Unit No.

00

NAKCSCSETX
ETX・ About the Checksum (CS) Calculation Method
For example, use the Error Reset Command,STX ¦ W ¦ 08 ¦ 00 ¦ 00000 ¦ CS ¦ ETXConvert ASCII code of "W08 00 00000" part to Hexadecimal No.
W→0x57, 0→0x30, 8→0x38
After Conversion, add it.
0x57 + 0x30 + 0x38 + 0x30+0x30 + 0x30+0x30+0x30+0x30+0x30 = 0x20F
The Lower 1 Bite of the Calculation Result is CS.
As a result, the Error reset command is "STX W 08 00 00000 0F ETX".

2020/5/7 Rev.0J-4