ITER Product Catalog Total Pressure Measurement

ITER Product Catalogue
Total Pressure Measurement
V20210114draft

Page 1 / 17

ITER Product Catalog Total Pressure Measurement

Content
TPG 500 Passive Gauge Controller ............................................................................................................. 3
Pirani/Cold Cathode Measurement Board CP 300 C9 ................................................................................ 5
Pirani / Cold Cathode Measurement Board CP 300 T11 ............................................................................ 6
Pirani / Cold Cathode Measurement Board CP 300 T11L .......................................................................... 7
Interface and Relay Board .......................................................................................................................... 8IF 300 A ....................................................................................................................................................... 8Interface and Relay Board IF 300 P / IF 301 P ............................................................................................ 9
Pirani Gauge Head TPR 018 ...................................................................................................................... 10
Cold Cathode Gauge Head IKR 084 .......................................................................................................... 12
Cold Cathode Gauge Head IKR 085 .......................................................................................................... 14
Cold Cathode Gauge Head IKR 086 .......................................................................................................... 16

Page 2 / 17

ITER Product Catalog Total Pressure Measurement

TPG 500 Passive Gauge Controller
Modular plug-in circuit boards for measurement andcommunication for 4 gauge heads.
Ideal controller for 4 gauge heads under extreme
conditions such as high radiation environment andhigh temperature bake-out.

DIMENSIONS

Page 3 / 17

ITER Product Catalog Total Pressure Measurement

SPECIFICATION
Display Range

1000 … 1x10-11 mbar (depends on used gauge and boards)

Measured value display

15 mm character height / 7 segment

Unit of measurembar, Torr, Pa; hPa; micron; V; A

Set point / relays

Lower and upper threshold / 4 relays

Error display / relay

Red segment / 1 relay

Measurement rate

≥ 100/s

Display rate

≥ 10/sTemperature:

Storage-20 to +60° C

Operating+5 to +50° C

Radiation resistant

no*1

To be tested with TPG 500

Maximum magnetic field (with IF301P) *1

To be tested with TPG 500

Main requirement:

Voltage (range)

100 … 240 VAC ±10%Frequency (range)

50 … 60 Hz

Power consumption≤65 W

Maximum magnetic field (with IF300P)

Overvoltage category

IIRelative humidity≤80% at temperatures up to +31° C decreasing to 50% at +40° C

Use

Indoors only, height up to 2000 m

SafetyEuropean appliance connector IEC 320 C14; class 1

Weight (without boards)

1.45 kg

ORDERING INFORMATION
Controller Unit TPG 500

IO398400

Delivery time

3 weeks

ITER Status

TPG500 in qualification

ACCESSORIESBlind plate instead Measure board

BG441259

Blind plate for IF300

BG441017

Mains cable, German plug, 2.5 m

303-350

Mains cable, U.S. plug, 2.5 m

303-353

Mains cable, U.K. plug, 2.5 m303-354

Mains cable, Japan, 2.5 m

303-352

Mains cable, Korea plug, 2.5 m

303-351

Rack 19“ for 3 TPG500 modules

IT491211

CABLES
Please see page Fehler! Textmarke nicht
definiert. and following
1* Tested to 3 dB margin as per Test method for ITER equipment for static (d.c.) magnetic fields(ITER D 98IL4W)

Page 4 / 17

ITER Product Catalog Total Pressure Measurement

Pirani/Cold Cathode Measurement
Board CP 300 C9
For TPG 500 Measurement Unit

Measurement and control down to 10-9 mbar
Bayonet SHV Connector

SPECIFICATION
CP 300 C9Measurement Boards

Pirani / Cold CathodeNumber of measuring circuits

1 Pirani and 1 Cold Cathode

Gauge

ITER TPR018, IKR084 and IKR085

Display range max.

1000 mbar

Display range min.

5x10-9 mbar

Temperature: Operating

5 to 50° C

Radiation resistantnoMax. applicable magnetic field

See TPG500

Sensor cable length< 100 mReaction time of the output signal: Rise
Pirani:

50ms (Sudden pressure changes: Rise (10-90% from 10-3 to 103 mbar)

Cold Cathode:

10 ms (Sudden pressure changes: Rise (10-90% from 10-9 to 10-3 mbar)

Weight0.21 kg

ORDERING INFORMATION
Measurement Board

IO441000

Delivery time

2 weeks

ITER Status

Qualified

ACCESSORIESPirani Connector

In qualification

Cold Cathode connector

In qualification

Lab plug 2mm LS205

IT4711152MA

Red isolation KT205

IT4711152N2

Black isolation KT205

IT4711152NZCABLES
Please see page Fehler! Textmarke nicht
definiert. and following

Page 5 / 17

ITER Product Catalog Total Pressure MeasurementPirani / Cold Cathode Measurement
Board CP 300 T11
For TPG 300 Measurement Unit
Measurement and control down to 10-11 mbar
Triaxial Push & Pull Connector

SPECIFICATION
CP 300 T11
Measurement Boards

Pirani / Cold CathodeNumber of measuring circuits

1 Pirani and 1 Cold CathodeGauge

ITER TPR018 and IKR086

Display range max.

1000 mbar

Display range min. *1x10-11 mbar

Temperature: Operating

5 to 50° C

Radiation resistantnoMax. applicable magnetic field

See TPG500

Sensor cable length max.

Pirani 100 m, Cold cathode 500m

Reaction time of the output signal: RisePirani:

50ms (Sudden pressure changes: Rise (10-90% from 10-3 to 103 mbar)

Cold Cathode:50 ms (Sudden pressure changes: Rise (10-90% from 10-9 to 10-3 mbar)

Weight0.25 kg

ORDERING INFORMATION
Measurement Board

IO441080

Delivery time

2 to 3 weeks

ITER Status

Available, in qualification

ACCESSORIESPirani connector Amphenol C91

In qualification

Cold Cathode connector

In qualification

Lab plug 2mm LS205

IT4711152MA

Red isolation KT205

IT4711152N2Black isolation KT205

IT4711152NZCABLES
Please see page Fehler! Textmarke nichtdefiniert. and following

Page 6 / 17

ITER Product Catalog Total Pressure Measurement

Pirani / Cold Cathode Measurement
Board CP 300 T11L
For TPG 500 Measurement Unit
Measurement and control down to 10-11 mbar
Pirani long cable version up to 500m
Triaxial Push & Pull Connector

SPECIFICATION
CP 300 T11L
Measurement Boards

Pirani / Cold CathodeNumber of measuring circuits

1 Pirani and 1 Cold Cathode

Gauge

ITER TPR018 and IKR086

Display range max.

1000 mbar

Display range min. *1x10-11 mbar

Temperature: Operating

5 to 50° C

Radiation resistantnoMax. applicable magnetic field

See TPG500

Sensor cable length max.

Pirani 500 m, Cold cathode 500m

Reaction time of the output signal: RisePirani:

50ms (Sudden pressure changes: Rise (10-90% from 10-3 to 103 mbar)

Cold Cathode:50 ms (Sudden pressure changes: Rise (10-90% from 10-9 to 10-3 mbar)

Weight0.25 kg

ORDERING INFORMATION
Measurement Board

IO441120

Delivery time

2 to 3 weeks

ITER Status

Available, in qualification

ACCESSORIESPirani connector Amphenol C91

In qualification

Cold Cathode connector

In qualification

Lab plug 2mm LS205

IT4711152MA

Red isolation KT205

IT4711152N2

Black isolation KT205

IT4711152NZCABLES
Please see page Fehler! Textmarke nicht
definiert. and following

Page 7 / 17

ITER Product Catalog Total Pressure Measurement

Interface and Relay Board
IF 300 A
RS communication to control controller measurement system
Hard wired set points relay outputs.

SPECIFICATION
IF 300 A
Relay Numbers

4 mechanical relay for set point, 1 error relay all with change overcontacts potential free.

Relaiy connector, 15 pin

D-Sub

Relay: Switching voltage max.

30 V DC / 50 V AC

Relay: Switching current max.

1.5 ARelay: Switching power max.

45W, 75VA

Relay: Resistance (with connector)

125 m Ohm

Interface

RS232 CInterface: Connection

D-sub-con. 9-pin, maleInterface: TypRS232 C asynchronInterface: Baud rates

300 to 9600 baudInterface: Data format

ASCII, 1 start, 8 Dates-, 1 Stop bit no Parity bit

Interface: Cable length max.

30 mWeight≈ 0.14 kg

ORDERING INFORMATIONInterface and Relay Board

IG441130-T

ACCESSORIESMating connector, D-Sub, male, 9poles399-977

Mating connector, D-Sub, female, 15
poles for IF 300 A/C, relay output

399-973

Interface cable
Relay connector complete

Page 8 / 17

ITER Product Catalog Total Pressure Measurement

Interface and Relay Board IF 300 P / IF301 P
For TPG 500 Measuring systems
ProfiBus communication with control system andSet point relay outputs.

SPECIFICATION
IF 300 P

IF 301 P

Relay Connector

D-sub 15 pin, male

D-sub 15 pin, male

Relay Numbers

4 mechanical relay for set point, 1error relay all with change overcontacts potential free.

4 photo MOS relay for set point, 1 error
relay all with change over contacts
potential free.

Max. applicable magnetic field

See TPG500See TPG500

Relay: Switching voltage max.

30 V DC / 50 V AC

30 V DC / 50 V AC

Relay: Switching current max.

1.5 A

0.5 A

Relay: Switching power max.

45W, 75VA

25 VA

Relay: Resistance (withconnector)

125 m Ohm

2.5 Ohm

Interface

ProfiBus-DP

ProfiBus-DP

Interface: Connection

D-sub-connector, 9-pin, male

D-sub-connector, 9-pin, maleInterface: Baud rates

9600, 19200

9600, 19200

Interface: Data format

ProfiBus protocols

ProfiBus protocols

Interface: Cable length max.

30 m

30 m

Weight≈ 0.16 kg≈ 0.15kg

ORDERING INFORMATIONInterface and Relay Board

IO441395

IO441396

Delivery time

2 weeks

2 weeks

ITER Status

Qualified for lab use only

In qualification

Mating connector, D-Sub, male,
9 poles

IT941145

IT941145

Mating connector, D-Sub,female, 15 poles for IF 300 A/C,
relay output

IT441129

IT441129ACCESSORIESPage 9 / 17

ITER Product Catalog Total Pressure Measurement

Pirani Gauge Head TPR 018
For TPG 500 measuring system
Bake able, magnetic field- and radiation-resistantHigh Vacuum-compatible Gauge HeadPush & Pull connector

DIMENSIONS

Page 10 / 17ITER Product Catalog Total Pressure Measurement

SPECIFICATION
FlangeTPR 018

TPR 018

DN 16 ISO-KF

DN 40 CF-F

5x10-4 ... 1000 mbar *3

Pressure range
Operating temperature, high temperature sensor cable

0 to 120°C

Operating temperature, standard sensor cable

0 to 80°C

Bake-out temperature (with high temp. cable or without cable)

0250°C
to 80°C

Mechanical Integrity

300°C *4

Radiation resistance

1x106 Gy

Pressure deviation due to external magnetic field@ 10-2 mbar
@ 1 mbar@ 102 mbar

Up to 480 mT: < 5%
Up to 300 mT: < 2.5%Always positive *5

Accuracy:

at room temp. and cable length <20m=±10% of reading in the range of 1x10-2 … 100 mbar

Accuracy:

fully specified up to +70°C and within the=±20% of reading in the range of 1x10-2 … 100 mbar

pressure range and cable length
Accuracy:

at 0 … 120°C and specified cable length=±5% of reading in the range of 1x10-2 … 100 mbar

Repeatability with airMaterials:=±35% of reading in the range of 1x10-2 … 100 mbar

Internal seal

Nimonic 90, Al

Filament / holder

W/Ni

Chamber wall, Housing feed throughW/Ni 1.4306
St. Steel 1.4435,
St.AlSteel
2O3Insulator
Brazing both feed through vacuum
atmosphere

Vacon 10, NiCo29-18 (0.379g Co)
Vacon 10, NiCo29-18 (0.360g Co)

Overpressure≤9 bar (limited to inert gases)

Protection category≤9 bar (ltd.
IPto40inert gases)

Mounting orientation40
We recommendIPmounting
the gaugealigned with the magnetic field.

Weight0.6 kgxxxxxxx

0.85 kg

ORDERING INFORMATION
Gauge

IOG15020

IOG15024

Optional with Material certificate

211-800

211-800

Delivery time

4 weeks

4 weeks

ITER StatusQualified

Qualified*3 Limited measurement and switching accuracy at pressures above 100 mbar and below 10-3 mbar
*4 Mechanical integrity, regarding vacuum tightness*5 Tested to 3 dB margins as per Test method for ITER equipment for static (d.c.) magnetic fields (ITER D 98IL4W)

Page 11 / 17

ITER Product Catalog Total Pressure Measurement

Cold Cathode Gauge Head IKR 084
For TPG 500 measuring system
Bakeable and radiation-resistantHigh Vacuum Gauge Head
Magnetically fully shieldedBayonet SHV connector for coaxial cable
Designed for use in double containment interiorDIMENSIONS

Page 12 / 17

ITER Product Catalog Total Pressure Measurement

SPECIFICATIONIKR 084
FlangeIKR 084

DN 40 ISO-KF

DN 40 CF-F-9-2

Display range

5x10 … 1x10 mbar

Measurement range

1x10-8 … 5x10-3 mbar

Operating temperature, with normal cable+5° C … +80° C

Operating temp., high temp. cable and Magnet+5° C … + 230° C

Bake out temperature with -Magnet

230°C

Mechanical Integrity

250°C

Radiation resistanceAccuracy in measurement range at 25°C

1x107 Gy
Approx. ± 30%Max. external magnetic field, axial direction180 mT (<10% deviation), 280 mT (< factor 2) *6dev.)*5 max
150 mT (<10% deviation), 180 mT (< factor 2 dev.)*6

Max. external magnetic field, radial direction
Fully reversible recovery from external magnetic field,
any orientation
Materials:

Tested up to 600 mT

Vacuum connection

St. steel (1.4306)

Measuring chamber

St. steel (1.4306)

Feed through isolation

Ceramic (Al2O3)

Anode

Molybdenum

Ignition aid

St. steel (1.4330, 1.4435 ESU)

Internal Vacuum Seal

Ni90, Ag plating

Brazing feed through atm. vacuum

AU82Ni 950Ionization Chamber: wall, pole disks

St. steel (1.4301, 1.4016)

MagnetHigh Temperature NdFeB

Magnetic protection shield

Fe 99.9% (soft iron) Ni electro plated ( 7 to10µm)

Electrical connection (Connector/Type)
Overpressure

Bayonet SHV connector for coaxial cable≤ 9 bar, limited to gases and temp. <55° C

Protection category

IP 40

Weight0.6 kg0.85 kg

Gauge

IOG18770

IOG18771

with optional Material certificate 3.1

211-800

211-800

Delivery time

2 weeks

2 weeks

ITER Status

QualifiedQualified

ORDERING INFORMATION*6 Tested to 3 dB margins as per Test method for ITER equipment for static (d.c.) magnetic fields (ITER D 98IL4W)

Page 13 / 17

ITER Product Catalog Total Pressure Measurement

Cold Cathode Gauge Head IKR 085
For TPG 500 measuring system
Bakeable and radiation resistantHigh Vacuum Gauge Head
Magnetically fully shielded, triaxial connector
Push & Pull connector for coaxial cable
For use in moderate-radiation environments

DIMENSIONS

Page 14 / 17

ITER Product Catalog Total Pressure Measurement

SPECIFICATIONIKR 085
FlangeIKR 085

DN 40 ISO-KF

DN 40 CF-F-9-2

Display range

5x10 … 1x10 mbar

Measurement range

1x10-8 … 5x10-3 mbar

Operating temperature, with normal cable+5° C … +80 °C

Operating temp., high temp. cable and Magnet+5° C … + 230 °C

Bake out temperature with Magnet

230 °C

Mechanical Integrity

250 °C

Radiation resistanceAccuracy in measurement range at 25°C

1x107 Gy
Approx. ± 30%

Max. external magnetic field, axial direction

180 mT (<10% deviation), 280 mT (< factor 2) *7dev.)*5 max
150 mT (<10% deviation), 180 mT (< factor 2 dev.)*7Max. external magnetic field, radial directionFully reversible recovery from external magnetic field,
any orientation
Materials:

Tested up to 600 mT

Vacuum connection

St. steel (1.4306)

Measuring chamber

St. steel (1.4306)

Feed through isolation

Ceramic (Al2O3)

Anode

Molybdenum

Ignition aid

St. steel (1.4330, 1.4435 ESU)

Internal Vacuum Seal

Ni90, Ag plating

Brazing feed through atm. vacuum

AU82Ni 950

Ionization Chamber: wall, pole disks

St. steel (1.4301, 1.4016)

MagnetHigh Temperature NdFeB

Magnetic protection shield

Fe 99.9% (soft iron) Ni electro plated ( 7 to10µm)

Electrical connection (Connector/Type)
Overpressure

Triaxial connector (used as coax) and SHV cableconnector≤ 9 bar, limited to gases and temp. <55° C

Protection category

IP 40

Weight0.6 kg0.85 kg

Gauge

IOG18772

IOG18773

with optional Material certificate 3.1

211-800

211-800

Delivery time

2 weeks

2 weeks

ITER Status

QualifiedQualified

ORDERING INFORMATION*7 Tested to3 dB margins as per Test method for ITER equipment for static (d.c.) magnetic fields (ITER D 98IL4W)

Page 15 / 17

ITER Product Catalog Total Pressure Measurement

Cold Cathode Gauge Head IKR 086
For TPG 500 measuring system
Bakeable and radiation-resistantHigh Vacuum Gauge Head
Magnetically fully shielded, triaxial connector
Push & Pull connector for triaxial cable
For locations requiring long-distance cabling

DIMENSIONS

Page 16 / 17

ITER Product Catalog Total Pressure Measurement

SPECIFICATIONIKR 086
FlangeIKR 086

DN 40 ISO-KF

DN 40 CF-F-9-2

Display range

1x10 … 1x10 mbar

Measurement range

1x10-8 … 5x10-3 mbar

Operating temperature, with normal cable+5 °C … +80 °C

Operating temp., high temp. cable and Magnet+5 °C … + 230 °C

Bake out temperature with -Magnet

230 °C

Mechanical Integrity

250 °C

Radiation resistanceAccuracy in measurement range at 25°C

1x107 Gy
Approx. ± 30%

Max. external magnetic field, axial direction

180 mT (<10% deviation), 280 mT (< factor 2) *8dev.)*5 max
150 mT (<10% deviation),
180 mT (< factor 2) *8

Max. external magnetic field, radial directiondev.)*5Tested up to 600 mT

Fully reversible recovery from external magnetic field,
any orientation
Materials:

Vacuum connection

St. steel (1.4306)

Measuring chamberSt. steel (1.4306)

Feed through isolation

Ceramic (Al2O3)

AnodeMo

Ignition aid

St. steel (1.4330)

Internal vacuum sealNi90, Ag plating

Brazing feed through atm. vacuum

AU82Ni 950

Ionization Chamber : (Wall + pole disks)

St. steel (1.4301, 1.4016)

MagnetHigh Temperature NdFeB

Magnetic protection shield

Fe 99.9% (soft iron) Ni electro plated ( 7 to10µm)

Electrical connection (Connector/Type)

Triaxial connector on gauge and controller

Overpressure≤ 9 bar, limited to gases and temp. <55 °C

Protection category

IP 40

Weight0.6 kg

0.85 kgGauge

IOG18776

IOG18777

with optional Material certificate 3.1

211-800

211-800

Delivery time

2 weeks

2 weeks

ITER Status

In qualification

In qualification

ORDERING INFORMATION*8 Tested to3 dB margins as per Test method for ITER equipment for static (d.c.) magnetic fields (ITER D 98IL4W)*5 Tested to3 dB margin as per Test method for ITER
equipment for static (d.c.) magnetic fields (ITER D98IL4W)
ent for static (d.c.) magnetic fields (ITER D 98IL4W)

Page 17 / 17