24.1" Color LCD Monitor

24.1-inch widescreen medical monitor withDICOM Part 14 calibration for viewing medical images.
®

View More with Widescreen

Display Both Monochrome and Color

The monitor has a resolution of 1920 x 1200 pixels and 16:10
aspect ratio. This provides more space for comfortably working
with images, tool palettes, and other windows compared toconventional square monitors.

The Hybrid Gamma PXL function automaticallydistinguishes between monochrome and color
images pixel by pixel, creating a hybrid display
where each pixel has optimum grayscale.

HybridGamma PXL

1.2x

More Space

Automatically Distinguish & Display as Color
Gamma 2.2

Square Monitor

Widescreen Monitor

Automatically Distinguish & Display as Monochrome
DICOM Part 14

Specifications
PanelElegant Cabinet Design
The monitor’s cabinet is designed with rounded corners andelegant curves on the back to present a gentle aesthetic andfacilitate a comforting environment for patients and medicalstaff.

Save Work Space with Sleek Cabinet Design
With the new monitor cabinet and stand design, compared to its
predecessor, the monitor’s width and depth are reduced by 23mm and 45 mm respectively. This results in a monitor that takes
up 24% less space, giving you more room for other tasks.

Make the Precise Diagnosis
EIZO carefully measures and sets each grayscale tone to createa monitor compliant with DICOM Part 14. This ensures theconsistent shading possible, allowing for the accurate diagnosis.

Maintain the Precision
Perform a simplified calibration compliant with DICOM Part 14
using the bundled RadiCS LE quality control software. RadiCS LE
corrects the brightness and grayscale tones of the monitor tomaintain image accuracy and consistency over time.

Select the Ideal Mode for Modalities
The CAL Switch function allows you to choose various modes fordifferent modalities such as CR, CT, and endoscopy. Using thebundled RadiCS LE software, modes can be set to automaticallyswitch to optimal image viewing conditions.

Hassle-Free Multi-Monitor Configuration
Using the DisplayPort connection, you can drive several monitors
in a daisy chain sequence. This allows you to configure a multimonitor setup without the complicated hassle of excessivecabling.

TypeBacklight
Size
Native Resolution
Viewable Image Size
(H x V)
Pixel Pitch
Display Colors

Color (IPS)
LED24.1” (61 cm)
1920 x 1200 (16:10 aspect ratio)
518.4 x 324.0 mm

0.270 x 0.270 mm
10-bit (DisplayPort): 1.07 billion froma palette of 543 billion (13-bit) colors
8-bit: 16.77 million from a palette of543 billion (13-bit) colors
Viewing Angles (H / V, typical) 178° / 178°
Brightness (typical)
410 cd/m²
Recommended
220 cd/m²
Brightness for Calibration
Contrast Ratio (typical) 1350:1
Response Time (typical) 22 ms (black-white-black)Video
Input Terminals
DisplayPort, DVI-DSignals
Output Terminals
DisplayPort (daisy chain)
Digital Scanning Frequency (H / V) 31 - 76 kHz / 59 - 61 Hz
USBUpstream
USB 2.0: Type-B
Downstream
USB 2.0: Type-A x 2
PowerPower Requirements
AC 100 - 240 V: 50 / 60 Hz
Typical Power Consumption 26 W
Maximum Power Consumption 56 W
Power Save Mode
0.6 W or lessSensorBacklight Sensor
Features &Brightness Stabilization Yes
Functions
Digital Uniformity Equalizer Yes
Hybrid Gamma PXL
YesWork-and-Flow
Point-and-Focus
Preset Modes
CAL Switch (DICOM, CAL1, CAL2, Custom,sRGB, Text)
OSD Languages
English, German, French, Italian,Japanese, Simplified Chinese, Spanish,Swedish, Traditional Chinese
PhysicalNet Weight
7.8 kgSpecifications Net Weight (Without Stand) 4.9 kg
Hole Spacing (VESA Standard) 100 x 100 mm
Certifications & Standards
CE / UKCA EN60601-1 (Medical Device),(Please contact EIZO for the latest information)
ANSI/AAMI ES60601-1, CAN/CSA-C22.2
No. 60601-1, IEC60601-1, VCCI-B, FCC-B,CAN ICES-3 (B), RCM, RoHS, China
RoHS, WEEE, CCC, EAC
FDA
510(k) Clearance for General
Radiography*
Dedicated
Monitor Quality
Supported
SoftwareControl Software
RadiCS
Signal Cables
DisplayPort (3 m)
SuppliedAccessories
OthersAC power cord (3 m), USB Type-A (May vary by country.
USB Type-B cable (3m), Utility Disk
Please contact EIZO for(RadiCS LE, PDF instructions for use,details.)
PDF installation manual), instructions
for useWarranty
5 Years* Display of mammography images for diagnosis is not supported.

A graphics board that supports daisy chain is necessary.

Conserve to Preserve

Dimensions (Unit: mm)
5°552

EIZO is moving away from using plastic and styrofoam inits packaging to reduce environmental impact. The monitor
is safely packed using molded pulp, which is made fromrecycled cardboard and newspaper, and cables are wrapped inenvironmentally friendly pulp sheet instead of plastic bags.

361.6
430.8~520.8

Cushioning Environmental Impact

30°

64.5

°
90The monitor is made of approximately 19% recycled plastic.
This cuts down on the amount of plastic waste going into theenvironment, conserves resources, and promotes reuse ofmaterials for preserving our natural ecosystems.

70°
200

EIZO, the EIZO Logo, RadiForce and RadiCS are registered trademarks of EIZO Corporation in Japan and other153 Shimokashiwano, Hakusan, Ishikawa 924-8566 Japan
Phone +81-76-277-6794, Fax +81-76-277-6793

https://www.eizoglobal.com

countries. DICOM is the registered trademark of the National Electrical Manufacturers Association for its standards publications relating to digital communications of medical information. All other company and productnames and logos are trademarks or registered trademarks of their respective owners. Specifications are subject
to change without notice.
Copyright © 2022 EIZO Corporation. All rights reserved. (220308B)