MSR 160 mini Data Logger
with 4 analogue inputs and microSD card

Four analogue inputs, a high measurement rate and its gigantic memory capacity, allpacked into a small format are the key characteristics of the new MSR 160 data logger.
The analogue inputs of the robust mini logger can be used for conventional sensors
with analogue outputs (e.g. CO2, conductivity, pH etc). In addition, a multiple output
switching power supply is also provided.
This highly versatile data logger has a measurement rate of up to 1000/s. The analogue measuring instrument can be optionally fitted with additional internal or external
temperature, humidity and pressure sensors. With the optional microSD card over one
billion measurement parameters can be recorded and saved using the MSR 160. The
lightweight device (approx. 80 g) has an anodised aluminium case and a 900 mAh
rechargeable lithium polymer battery. All saved data can be rapidly transferred to a
PC or laptop via the USB interface. The user-friendly data acquisition and analysis
software is supplied free along with the data logger.
Basic specification and technical data
Analogue inputs:

Measurement/
Storage rate:
Housing:1000 /s to every 12 hSize & weight:

Anodised aluminium case, PC, encapsulated
waterproof (IP67), in blue, anthracite
39 x 23 x 72 mm, approx. 80 g

Key:

Configurable pushbutton functions

Memory capacity:
Power supply:
Interface:
PC software:

Operating conditions:
Lagerbedingungen:
MSR 160 withscrew terminal

4 analogue inputs with freely selectable input configuration: 0…20 mA; 4…20 mA; 0…3.0 V; 0.5…4.5 V;
0…5.0 V; 1.0…6.0 V; 0…10.0 V; 0…12.0 V; 0…24.0 V
Resolution: 12 BitStandards:

Over 2 000 000 measurement parameters – expandable
to over 1bn measurement parameters with optional
microSD card.
• Rechargeable lithium polymer battery 900 mAh• The battery is charged via the USB connection
USB
Free Setup, Reader, Viewer and Online software(Windows XP / Vista / 7 / 8 / 10) for data acquisition
and analysis.
• Temperature -20…+65 °C• Temperature +5…+45° C(ideal storage condition for the battery)• 10…95% relative humidity, non-condensing
The MSR 160 complies with EU-Directives RoHS/WEEE.

Measurement parameters of the available sensors (internal and external)

Measuredparameters

Working range

Accuracy

Measurement/
storage rate

Temperature

int.: -20…+65 °C±0.5 °C (-10…+65 °C)

1 /s to every 12 h

ext.: -55…+125 °C ±0.5 °C (-10…+65 °C)±2 °C (-55…+125 °C)
Relative humidity
with integrated
temperature

0…100%
rel. humidity-20…+65 °C±2% rel. humidity(10…85%, 0…+40 °C)±4% rel. humidity(85…95%, 0…+40 °C)

1 /s to every 12 h

Air pressure
absolute, withintegrated
temperature

0…2000 mbar
absolute-20…+65 °C±2.5 mbar
(750…1100 mbar
absolute, +25 °C)

1 /s to every 12 h

optional:
0…14 barabsolute-20…+65 °C±50 mbar (1…10 bar
absolute, +25 °C)

1 /s to every 12 h

3-axisacceleration(Position)±15 g-20…+65 °C±0.15 g (+25 °C)

1 /s to every 12 h

Options:~23
ø5You can specify your MSR 160 with up to four different internal sensors: the options aretemperature, humidity, pressure and attitude. If required you can also connect externalsensors in place of the internal ones (temperature, humidity or pressure) with the following cable lengths: 0.15 m, 0.4 m, 1.0 m, and 1.6 m (maximum 2 pcs.).

External temperature sensor

ø6~23

External humidity sensor

ø1955

External air pressure sensor
Expanding the memory capacity with a microSD memory card
On request the MSR 160 can be suppliedwith an interface for plugging in a standard
microSD card (one ≥4GB microSD card isincluded). The microSD card increases thememory of the MSR160 to over 1 billion andcan be replaced during operation.

Screw terminal

Please note that this expansion slot withplugged-in microSD-card only meets theprotection category IP60.
MSR Electronics GmbH
Mettlenstrasse 6CH-8472 Seuzach
Switzerland
Tel. +41 52 316 25 55
Fax +41 52 316 35 21
info@msr.ch
www.msr.ch

Distributor:

11.15

Contact us!
We would be pleased to provide you with prices and delivery terms!