Instructions for Use

Color LCD Monitor

ImportantPlease read this “Instructions for Use”, and “Installation Manual”(separate volume) carefully to familiarize yourself with safe andeffective usage.
Please retain this manual for future reference.
• For monitor adjustment and settings, refer to the “Installation
Manual”.
• For the latest product information including the “Instructions forUse”, refer to our web site :http://www.eizoglobal.com

SAFETY SYMBOLSThis manual and this product use the safety symbols below. They denote critical information. Please readthem carefully.

WARNINGCAUTION

Failure to abide by the information in a
WARNING may result in serious injury
and can be life threatening.

Failure to abide by the information in a
CAUTION may result in moderate injury and /or property or product damage.

Indicates a warning or caution. For example,

indicates an “electrical shock” hazard.

Indicates a prohibited action. For example,

means “Do not disassemble”.

Indicates a mandatory action. For example,

means “Ground the unit”.

This product has been adjusted specifically for use in the region to which it was originally shipped. If
operated outside this region, the product may not perform as stated in the specifications.
No part of this manual may be reproduced, stored in a retrieval system, or transmitted, in any form or by
any means, electronic, mechanical, or otherwise, without the prior written permission of EIZO Corporation.
EIZO Corporation is under no obligation to hold any submitted material or information confidential unlessprior arrangements are made pursuant to EIZO Corporation’s receipt of said information. Although every
effort has been made to ensure that this manual provides up-to-date information, please note that EIZO
monitor specifications are subject to change without notice.

2

﻿PRECAUTIONSIMPORTANT• This product has been adjusted specifically for use in the region to which it was originally shipped.
If the product is used outside the region, it may not operate as specified in the specifications.
• To personal safety and proper maintenance, please read carefully this section and the caution
statements on the monitor.

Location of the Caution Statements

Symbols on the unit
SymbolThis symbol indicatesPower button:

Press to turn the monitor’s power on or off.

Alternating current
Alerting electrical hazard
CAUTION:

Refer to “SAFETY SYMBOLS” (page 2).

WEEE marking:Product must be disposed of separately; materials may
be recycled.

CE marking:EU conformity mark in accordance with the provisions ofCouncil Directive and/or Regulation (EU).

Manufacturer
Date of manufacture
Caution:Federal law (USA) restricts this device to sale by or on
the order of a licensed healthcare practitioner.

Medical device in EU
Importer in EU
Mark signifying compliance with UK regulations
UK Responsible Person
Authorised representative in Switzerland
Authorised representative in the European Community
PRECAUTIONS

3

WARNING
If the unit begins to emit smoke, smells like something is burning, or makes strange noises,
disconnect all power connections immediately and contact your EIZO representative for advice.
Attempting to use a malfunctioning unit may result in fire, electric shock, or equipment damage.
Do not disassemble or modify the unit.
Opening the cabinet or modifying the unit may result in fire, electric shock, or burn.
Refer all servicing to qualified service personnel.
Do not attempt to service this product yourself as opening or removing covers may result in fire, electric
shock, or equipment damage.
Keep small objects or liquids away from the unit.
Small objects accidentally falling through the ventilation slots into the cabinet or spillage
into the cabinet may result in fire, electric shock, or equipment damage. If an object or
liquid falls/spills into the cabinet, unplug the unit immediately. Have the unit checked bya qualified service engineer before using it again.
Place the unit at a sturdy and stable place.
A unit placed on an inadequate surface may fall and result in injury or equipment damage. If the unit falls,
disconnect the power immediately and ask your local EIZO representative for advice. Do not continue using
a damaged unit. Using a damaged unit may result in fire or electric shock.
Use the unit in an appropriate location.
Otherwise, fire, electric shock, or equipment damage may result.
• Do not place outdoors.
• Do not place in any form of transportation (ships, aircraft, trains, automobiles, etc.).
• Do not place in dusty or humid environments.
• Do not place in locations where water may be splashed on the screen (bathrooms, kitchens,etc.)• Do not place in locations where steam comes in direct contact with the screen.
• Do not place near heat generating devices or humidifiers.
• Do not place in locations where the product is subject to direct sunlight.
• Do not place in environments with inflammable gas.
• Do not place in environments with corrosive gases (such as sulfur dioxide, hydrogen sulfide,
nitrogen dioxide, chlorine, ammonia, and ozone)• Do not place in environments with dust, components that accelerate corrosion in theatmosphere (such as sodium chloride and sulfur), conductive metals, and so on

To avoid danger of suffocation, keep the plastic packing bags away from babies and children.
Use the enclosed power cord and connect to the standard power outlet in your country.
Be sure to use within the rated voltage of the power cord. Otherwise, fire or electric shock may result.
Power supply: 100-240Vac 50/60Hz
To disconnect the power cord, grasp the plug firmly and pull.
Tugging on the cord may damage and result in fire or electric shock.

The equipment must be connected to a grounded main outlet.
Failure to do so may result in fire or electric shock.
Use the correct voltage.
• The unit is designed for use with a specific voltage only. Connection to another voltage than specified in this“Instructions for Use” may cause fire, electric shock, or equipment damage.
Power supply: 100-240Vac 50/60Hz• Do not overload your power circuit, as this may result in fire or electric shock.

4

PRECAUTIONSWARNING
Handle the power cord with care.
• Do not place the cord underneath the unit or other heavy objects.
• Do not pull on or tie the cord.

If the power cord becomes damaged, stop using it. Using of a damaged cord may resultin fire or electric shock.
The operator should not touch the patient while touching the product.
This product has not been designed to be touched by patients.
Never touch the plug and power cord if it begins to thunder.
Touching them may result in electric shock.
When attaching an arm stand, please refer to the user’s manual of the arm stand and install the unit
securely.
Otherwise, the unit may become detached, resulting in injury and/or equipment damage. Before installation,
make sure that desks, walls, or any other installation surface has adequate mechanical strength. If the unit
falls, please ask your local EIZO representative for advice. Do not continue using a damaged unit. Usinga damaged unit may result in fire or electric shock. When reattaching the tilt stand, please use the samescrews and tighten them securely.
Do not touch a damaged LCD panel directly with bare hands.
Liquid crystal is poisonous. If any part of your skin comes in direct contact with thepanel, wash thoroughly. If liquid crystal enters your eyes or mouth, immediately flushwith large amounts of water and seek medical attention.

PRECAUTIONS

5CAUTION
Handle with care when carrying the unit.
Disconnect the power cord and cables when moving the unit. Moving the unit with the power cord or cables
attached is dangerous and may result in injury.
Carry or place the unit according to the correct specified methods.
• When carrying, grasp and firmly hold the unit as shown in the illustration below.
• Monitors of size 30 inches and above are heavy. When unpacking and/or carrying the monitor, ensure at least two
people are utilized.

Dropping the unit may result in injury or equipment damage.

Do not block the ventilation slots on the cabinet.
• Do not place any objects on the ventilation slots.
• Do not install the unit in a place with poor ventilation or inadequate space.
• Do not use the unit laid down or upside down.

Blocking the ventilation slots prevents proper airflow and may result in fire, electric
shock, or equipment damage.
Do not touch the plug with wet hands.
Doing so may result in electrical shock.

Use an easily accessible power outlet.
This is to facilitate disconnecting the power in case of a problem.
Periodically clean the area around the power plug and the ventilation slot of the monitor.
Dust, water, or oil on the plug may result in fire.
Unplug the unit before cleaning it.
Cleaning the unit while it is plugged into a power outlet may result in electric shock.
If you plan to leave the unit unused for an extended period of time, disconnect the power cord fromthe wall socket after turning off the power switch for the safety and the power conservation.
For users in the territory of the EEA and Switzerland:
Any serious incident that has occurred in relation to the device should be reported to the
Manufacturer and the Competent Authority of the Member State in which the user and/or patient isestablished.

6

PRECAUTIONS

Notice for this Monitor
Intended Use
This Product is indicated for use in displaying radiological images for review, analysis, and diagnosis bytrained medical practitioners. The display is not intended for mammography.
Attention• This product may not be covered by warranty for uses other than those described in this manual.
• The specifications noted in this manual are only applicable when the following are used:- Power cords provided with the product- Signal cables specified by us• Only use EIZO accessories products specified by EIZO with this product.

Precautions for Use• Parts (such as the LCD panel) may deteriorate in the long-term. Periodically check that they areoperating normally.
• When the screen image is changed after displaying the same image for extended periods of time, anafterimage may appear. Use the screen saver or Power Save function to avoid displaying the sameimage for extended periods of time.
• If the monitor displays continuously over a long period of time, dark smudges or burn-in may appear. To
maximize the life of the monitor, we recommend the monitor be turned off periodically.
• An afterimage may appear even after a short time period has elapsed depending on the displayed
image. If this occurs, changing the image or leaving the power off for a few hours may solve the problem.
• The backlight of the LCD panel has a fixed lifetime. When the screen becomes dark, begins to flicker or
no longer lights up, please contact your local EIZO representative.
• The screen may have defective pixels or a small number of light dots on the screen. This is due to thecharacteristics of the panel itself, and is not a malfunction of the product.
• Do not press on the panel or edge of the frame strongly, as this may result in display malfunctions, suchas interference patterns and others. If pressure is continually applied to the panel, it may deteriorate
or damage your panel. (If pressure marks remain on the panel, leave the monitor with a black or white
screen. The symptom may disappear.)• Do not scratch or press on the panel with any sharp objects, as this may result in damage to the panel.
Do not attempt to brush with tissues as this may scratch the panel.
• When the monitor is cold and brought into a room or the room temperature goes up quickly, dew
condensation may occur on the interior and exterior surfaces of the monitor. In that case, do not turn themonitor on. Instead wait until the dew condensation disappears, otherwise it may cause some damage
to the monitor.

Notice for this Monitor

7

To Use the Monitor for a Long Time● Quality control• The display quality of monitors is affected by the quality level of input signals and the degradation of
the product. Perform visual checks and periodic constancy tests to comply with medical standards /guidelines according to your application, and carry out calibration as necessary. Use of the RadiCS
monitor quality control software enables the user to perform high-level quality control that meets
medical standards / guidelines. For how to perform various tests and calibration, refer to the
RadiCS User's Manual.
• It takes about 30 minutes for the monitor display to stabilize. Please wait 30 minutes or more after
the monitor power has been turned on or the monitor has recovered from the power saving mode
before performing various tests for quality control, calibration, or screen adjustment of the monitor.
• We recommend that monitors be set to the recommended level or lower to reduce changes inluminosity caused by long-term use and maintain stable brightness.
Attention• The display status of the monitor may change unexpectedly due to an operating error or unexpected settingchange. Using the monitor with the control buttons locked is recommended after adjusting the screen of themonitor. For details on how to set, refer to the Installation Manual (on the CD-ROM).

● Cleaning

Periodic cleaning is recommended to keep the monitor looking new and to prolong its operation
lifetime.
Carefully wipe the cabinet and panel surface with a soft cloth moistened with water or the following
chemicals.

Chemicals that may be used for cleaning
Material name

Ethanol for disinfection
Isopropyl alcohol
Chlorhexidine gluconate
Sodium hypochlorite
Benzalkonium chloride
Alkyldiamino ethylglycine
Glutaral

Product name

Ethyl alcohol (Ethanol)
Isopropyl alcohol
Hibitane solution
PureloxWelpasTego 51
Cidexplus 28

Attention• Do not use chemicals frequently. Chemicals such as alcohol and antiseptic solution may cause gloss
variation, tarnishing, and fading of the cabinet or panel, and also quality deterioration of the image.
• Never use any thinner, benzene, wax, and abrasive cleaner, which may damage the cabinet or panel.
• Do not attach chemicals directly to the monitor.
Note• Using ScreenCleaner (optional) for cleaning of the cabinet and the panel surface is recommend.

To Use the Monitor Comfortably• Staring at the monitor for a long time tires your eyes. Take a 10-minute rest every hour.
• Look at the screen from a proper distance and from a proper angle.

8

Notice for this Monitor

CONTENTS
PRECAUTIONS....................................................... 3
IMPORTANT............................................................... 3

Notice for this Monitor........................................... 7
Intended Use . ........................................................... 7
Precautions for Use.................................................. 7
To Use the Monitor for a Long Time........................ 8● Quality control.................................................. 8● Cleaning........................................................... 8

To Use the Monitor Comfortably............................. 8

CONTENTS.............................................................. 9
Chapter 1
1-1.
1-2.

●

1-3.

Introduction...................................... 10

Features.........................................................10
Package Contents.........................................10
EIZO LCD Utility Disk......................................10Controls and Functions...............................11

Chapter 2

Installation / Connection................. 12

2-1.

Before Installing the Product......................122-2.

Connecting Cables.......................................13

2-3.

Turning On the Power...................................142-4.

Adjusting the Screen Height and Angle.....14● Installation Requirements...............................12

Chapter 3

No-Picture Problem......................... 15

Chapter 4

Specifications.................................. 16

4-1.

Specifications List........................................16

4-2.

Compatible Resolutions..............................17

4-3.

Accessories...................................................18● For Digital Signal Input (DisplayPort / DVI).....17● For Analog Signal Input (D-Sub).....................17

Appendix............................................................... 19
Medical Standard.....................................................19
EMC Information..................................................... 20
FCC Declaration of Conformity............................. 25

CONTENTS

9

Chapter 1 Introduction
Thank you very much for choosing an EIZO color LCD monitor.

1-1. Features• 19.0 inches• Supports a resolution of 1 M pixels (1280 dots x 1024 lines)• VA panel with 178˚ horizontal and vertical viewing angles• Supports triple-input system (DisplayPort, DVI, and D-Sub)• Equipped with the six CAL Switch modes, including the DICOM mode having DICOM ® Part
14-compliant grayscale characteristics
The CAL Switch mode can be switched according to a displayed image. For details, refer to the Installation
Manual (on the CD-ROM).

• The quality control software "RadiCS LE" is included.
- Performs calibration and daily checkups and manages the history.
- Allows the user to operate the monitor, such as switching the CAL Switch mode or input signal
by using the mouse or keyboard.
• Employs stand with wide range of movement.
Allows you to adjust the monitor to the optimal position for easy operation and minimizing fatigue.
(Tilt: 30˚ up/0˚ down, swivel: 35˚ right /35˚ left, adjustable height: 100 mm)

1-2. Package Contents
Check that all of the following items are included in the packaging box. If any of these are missing,contact your dealer or local EIZO representative.
Note• Please keep the packaging box and materials for future movement or transport of the monitor.
• Monitor• Power cord

• Digital signal cable: PP300
DisplayPort - DisplayPort• USB cable: UU300• EIZO LCD Utility Disk (CD-ROM)• Instructions for Use• VESA Mounting Screws (M4 x 12 mm, 4 pcs)

● EIZO LCD Utility Disk

The following items are included in the EIZO LCD Utility Disk (CD-ROM). Refer to Readme.txt file on
the CD-ROM for software startup procedures or file reference procedures.
• Readme.txt file
• RadiCS LE monitor quality control software (for Windows)• User's Manual- Monitor Installation Manual- RadiCS LE User's Manual• Outline dimensions
Note• For information on how to install and use RadiCS LE, refer to RadiCS LE User's Manual. When using RadiCSLE, connect the monitor to your PC using the supplied USB cable.

10

Chapter 1 Introduction

1-3. Controls and Functions

9
10

8

1

2

1. Ambient Light
Sensor

2. Control button

3.
button4. Power Indicator
5. Power connector
6. Input Signal
Connector

7. USB upstream port
8. Security lock slot
9. Stand10. Cable holder

3

4

5

6

7

This sensor measures illumination with the RadiCS / RadiCS LE function that monitors
changes in illumination. Refer to RadiCS / RadiCS LE User’s Manual for details.
The values measured by this sensor are not reflected on the RadiCS environmental
illumination because the sensor is simplified one.
Displays the operation guide. Set menus according to the operation guide.
For details on the operation guide and the menus, refer to the Installation Manual (on the
CD-ROM).
Turns the power on or off.
Indicates the monitor’s operation status.
Green: Operating Orange: Power saving mode Off: Power off
Connects the power cord.
Connect signal cables.
Left: DisplayPort connector
Center: DVI-D connector
Right: D-Sub mini 15-pin connector
Connects the USB cable to use the software that needs USB connection.
Complies with Kensington’s MicroSaver security system.
Used to adjust the height and angle (tilt, swivel) of the monitor screen.
Covers the monitor cables.

Chapter 1 Introduction

11

Chapter 2 Installation / Connection
2-1. Before Installing the Product
Carefully read “PRECAUTIONS” (page 3) and always follow the instructions.
If you place this product on a lacquer-coated desk, the color may adhere to the bottom of the stand dueto the composition of the rubber. Check the desk surface before use.

● Installation Requirements

When installing the monitor in a rack, ensure that there is adequate space around the sides, back andtop of the monitor.

Attention• Position the monitor so that there is no light to interfere with the screen.

12

Chapter 2 Installation / Connection

2-2. Connecting Cables
Attention• Check that the monitor and the PC are powered off.
• When replacing the current monitor with this monitor, be sure to change the PC settings for resolution and verticalscan frequency to those that are available for this monitor by referring to “4-2. Compatible Resolutions” (page17), before connecting the PC.

1. Connect signal cables.

Check the shapes of the connectors, and connect the cables.
After connecting the DVI cable or the D-Sub cable to the monitor, tighten the fasteners to secure theconnector.

MD-C87 (optional)

DD300 (optional)

PP300 (supplied)

2. Plug the power cord into a power outlet and the power connector on themonitor.

Insert the power cord fully into the power connector on the monitor.

3. When using RadiCS / RadiCS LE, connect the USB cable between the monitor's
USB upstream port and the PC.

Chapter 2 Installation / Connection13

2-3. Turning On the Power
1. Pressto turn on the monitor.

The monitor’s power indicator lights up green.
If the indicator does not light up, refer to “Chapter 3 No-Picture Problem” (page 15).

2. Turn on the PC.

The screen image appears.
If an image does not appear after turn-on, refer to “Chapter 3 No-Picture Problem” (page 15) for
additional advice.
Attention• For the maximum power saving, it is recommended that the Power button be turned off. Disconnecting thepower outlet cuts the power consumption completely, when not using the monitor.
Note• When turning on the monitor and PC for the first time with the analog signal, the Auto Adjustment function
works to adjust the clock, phase, and display position automatically.
• In order to maximize the monitor's lifespan by impeding brightness degradation and to reduce power
consumption, carry out the following:- Use the power saving function of the computer.
- Turn off the monitor and PC after using them.

2-4. Adjusting the Screen Height and Angle
Hold left and right edge of the monitor with both hands, and adjust the screen height, tilt and swivel of thescreen to the best condition for working.

Attention• After the adjustment is finished, make sure that the cables are correctly connected.

14

Chapter 2 Installation / Connection

Chapter 3 No-Picture Problem
ProblemPossible cause and remedy

1. No picture• Power indicator does not light.

• Check whether the power cord is connected properly.
.
• Press• Power indicator lights green.

• Increase "Brightness", "Contrast", or "Gain" in the Setting
menu. For details, refer to the Installation Manual (on the
CD-ROM).

• Power indicator is lighting orange.

• Switch the input signal with
. For details, refer to the
Installation Manual (on the CD-ROM).
• Move the mouse or press any key on the keyboard.
• Check whether the PC is turned on.
• This problem may occur when the PC is connected via the
DisplayPort connector. Reconnect the power cord of themonitor, or reboot the PC.

2. The message below appears.

This message appears when the signal is not input correctly
even when the monitor functions properly.

• This message appears when no signal is
input.
Example:• The message shown left may appear, because some PCs do
not output the signal soon after power-on.
• Check whether the PC is turned on.
• Check whether the signal cable is connected properly.
• Switch the input signal with
. For details, refer to the
Installation Manual (on the CD-ROM).

• The message indicates that the input signal
is out of the specified frequency range.
Example:• Check whether the PC is configured to meet the resolution
and vertical scan frequency requirements of the monitor (see“4-2. Compatible Resolutions” (page 17)).
• Reboot the PC.
• Change to the appropriate setting using the graphics board's
utility. For details, refer to the User's Manual of the graphics
board.

Chapter 3 No-Picture Problem

15Chapter 4 Specifications
4-1. Specifications List
LCD PanelVideo Signals

Type

VA

Backlight

LED

Size

48 cm (19.0 inch) (48.1 cm diagonal)

Resolution

1280 dots × 1024 lines

Display Size (H x V)

376.3 mm x 301.0 mm

Pixel Pitch (H x V)

0.294 mm x 0.294 mm

Display Colors

Max. 1073.74 million colors

Viewing Angles(H / V, typical)

178˚ / 178˚Contrast Ratio(typical)2000 : 1

Response Time (typical)

20 ms (black -> white -> black)

Input Terminals

DisplayPort x 1, DVI-D (Single Link) x 1,
D-Sub mini 15-pin x 1

Horizontal scanningfrequency

DVI, DisplayPort: 31 kHz - 64 kHz

Vertical Scan Frequency

DVI, DisplayPort: 59 Hz - 61 Hz (720 x 400: 69 Hz - 71 Hz)

D-Sub: 24.8 kHz - 80.0 kHz
D-Sub: 50.0 Hz - 75.0 Hz

USB
Power

PhysicalSpecifications

Synchronization Signal

Separate

Dot Clock (Max.)

DVI, DisplayPort: 108 MHz
D-Sub: 135 MHz

Port

Upstream x 1

Standard

USB Specification Revision 2.0

Input

100 - 240 VAC ±10 %, 50 / 60 Hz, 0.50A - 0.30A

Maximum Power
Consumption

28 W or lessPower Save Mode

0.6 W or less *1

Standby Mode

0.6 W or less *2

Outside dimensions(Width × Height × Depth)

405 mm x 406.5 mm - 506.5 mm x 205 mm (Tilt: 0˚)
405 mm x 438.8 mm - 538.8 mm x 227.7 mm (Tilt: 30˚)

Outside dimensions(Without Stand)(Width x Height x Depth)

405 mm × 334 mm × 61.5 mm

Net Weight

Approx. 6.0 kg

Net Weight (Without Stand) Approx. 4.2 kg

Operating
Environment
Requirements

16

Height Adjustment Range

100 mmTilt

Up 30˚, down 0˚SwivelRight 35˚, Left 35˚Rotation

90˚ (clockwise)

Temperature

0 ˚C - 35 ˚C

Humidity

20 % - 80 % R.H. (no condensation)

Air Pressure

540 hPa - 1060 hPa

Chapter 4 Specifications

Transportation /Storage
Environmental
Requirements

Temperature-20 ˚C - 60 ˚C

Humidity

10 % - 90 % R.H. (no condensation)

Air Pressure

200 hPa - 1060 hPa*1When DVI input is used, the USB upstream port is not connected, "Auto Input Detection": "Off", "Power
Save": "On" and "DP Power Save": "On"*2 The USB upstream port is not connected and "DP Power Save": "On"4-2. Compatible Resolutions
The monitor supports the following resolutions.

● For Digital Signal Input (DisplayPort / DVI)
Resolution

Vertical Scan Frequency

640 x 480

60 Hz

720 x 400

70 Hz

800 x 600

60 Hz

1024 x 768

60 Hz

1280 x 1024 *1

60 Hz*1 Recommended resolution.

● For Analog Signal Input (D-Sub)
Resolution

Vertical Scan Frequency

640 x 480

~75 Hz720 x 400

70 Hz

800 x 600

~75 Hz1024 x 768~75 Hz1152 x 864

75 Hz1280 x 960

60 Hz

1280 x 1024 *1~75 Hz640 x 400

70 Hz

*1 Recommended resolution.

Chapter 4 Specifications17

4-3. Accessories
The following accessories are available separately.
For the latest information about the optional accessories sold separately and information about thecompatible graphics board, refer to our web site.
http://www.eizoglobal.com

18

Panel protector

EIZO "FP-702"

Arm, StandEIZO "LS-HM1-D": Dual height adjustable stand
EIZO "LA-011-W": Wall-hung arm
EIZO "AAH-02B3W": Wall-hung arm

Calibration Kit

EIZO "RadiCS UX2" Ver. 4.6.3 or later

Network QC Management Software

EIZO "RadiNET Pro" Ver. 4.6.3 or later
EIZO "RadiNET Pro Lite" Ver. 4.6.3 or later

Cleaning Kit

EIZO "ScreenCleaner"

Signal Cable (Display Port - Display Port)

PP200

Signal Cable (DVI-D - DVI-D)

FD-C39, DD300

Signal Cable (D-Sub - D-Sub)

MD-C87Signal Cable (DVI-I - D-Sub)

MD-C16

USB cable

FD-C93

Chapter 4 Specifications

Appendix
Medical Standard• It shall be assured that the final system is in compliance to IEC60601-1-1 requirement.
• Power supplied equipment can emit electromagnetic waves, that could influence, limit or result in
malfunction of the monitor. Install the equipment in a controlled environment, where such effects
are avoided.

Classification of Equipment-

Type of protection against electric shock : Class I
EMC class: IEC60601-1-2 Group 1 Class B
Classification of medical device (EU): Class I
Mode of operation : Continuous
IP Class : IPX0

Appendix

19

EMC Information
The RadiForce series has a performance that appropriately displays medical images.

Environments of Intended Use
The RadiForce series is intended to be used in the following environments.
• Professional healthcare facility environments such as clinics and hospitals
The following environments are not suitable for the RadiForce series to be used:• Home healthcare environments• In the vicinity of high-frequency surgical equipments such as electrosurgical knives• In the vicinity of short-wave therapy equipments• RF shielded room of the medical equipment systems for MRI• In shielded location Special environments• Installed in vehicles including ambulances• Other special environment

WARNINGThe RadiForce series requires special precautions regarding EMC and need to be installed. You
need to carefully read EMC Information and the “PRECAUTIONS” section in this document, andobserve the following instructions when installing and operating the product.
The RadiForce series should not be used adjacent to or stacked with other equipment. If adjacentor stacked use is necessary, the equipment or system should be observed to verify normal
operation in the configuration in which it will be used.
When using a portable RF communication equipment, keep it 30 cm (12 inches) or more away fromany part, including cables, of the RadiForce series. Otherwise, degradation of the performance of
this equipment could result.
Anyone who connects additional equipment to the signal input part or signal output parts,
configuring a medical system, is responsible that the system complies with the requirements of
IEC60601-1-2.
Do not touch the signal input/output connectors while using the RadiForce series. Otherwise, thedisplayed image may be affected.
Be sure to use the cables attached to the product, or cables specified by EIZO.
Use of cables other than those specified or provided by EIZO of this equipment could result inincreased electromagnetic emissions or decreased electromagnetic immunity of this equipment
and improper operation.
Cable
Signal cable (DisplayPort)
Signal cable (DVI-D)
Signal cable (D-Sub)
USB cable
Power cord (with earth)20

Appendix

EIZO Designated
Cables

PP300 / PP200
DD300 / FD-C39
MD-C87
UU300 / MD-C39-

Max. Cable Length
3m
3m
1.8 m
3m
3m

ShieldingShielded
Shielded
Shielded
Shielded
Unshielded

Ferrite Core
With Ferrite Cores
With Ferrite Cores
With Ferrite Cores
With Ferrite Cores
Without Ferrite Cores

Technical Descriptions
Electromagnetic emissions

The RadiForce series is intended for use in the electromagnetic environment specified below.
The customer or the user of the RadiForce series should assure that it is used in such an environment.

Emission test

ComplianceElectromagnetic environment - Guidance

RF emissions
CISPR11

Group 1The RadiForce series uses RF energy only for its internal function.
Therefore, its RF emission are very low and are not likely to cause any interference innearby electronic equipment.

RF emissions
CISPR11Class B

Harmonic emissions
IEC61000-3-2Class D

The RadiForce series is suitable for use in all establishments, including domestic
establishments and those directly connected to the public low-voltage power supply
network that supplies buildings used for domestic purposes.

Voltage fluctuations /flicker emissions
IEC61000-3-3Complies

Electromagnetic immunity
The RadiForce series has been tested at the following compliance levels (C) according to the testing requirements (T)for Professional healthcare facility environments defined in IEC60601-1-2.
The customer or the user of the RadiForce series should assure that it is used in such an environment.

Immunity test

Test level (T)Compliance level (C)Electromagnetic environment GuidanceElectrostatic
discharge (ESD)
IEC61000-4-2±8 kV contact discharge±15 kV air discharge±8 kV contact discharge±15 kV air discharge

Floors should be wood, concrete or ceramic tile.
If floors are covered with synthetic material, therelative humidity should be at least 30%.

Electrical fasttransients / bursts
IEC61000-4-4±2 kV power lines±1 kV input / output lines±2 kV power lines±1 kV input / output lines

Mains power quality should be that of a typicalcommercial or hospital environment.

Surges
IEC61000-4-5±1 kV line to line
±2 kV line to ground±1 kV line to line
±2 kV line to ground

Mains power quality should be that of a typicalcommercial or hospital environment.

Voltage dips, shortinterruptions andvoltage variations on
power supply input
lines
IEC61000-4-11

0 % UT (100 % dip in UT)
0.5 cycles and 1 cycle
70 % UT (30 % dip in UT)
25 cycles at 50 Hz
0 % UT (100 % dip in UT)
250 cycles at 50 Hz

0 % UT (100 % dip in UT)
0.5 cycles and 1 cycle
70 % UT (30 % dip in UT)
25 cycles at 50 Hz
0 % UT (100 % dip in UT)
250 cycles at 50 Hz

Mains power quality should be that of a typicalcommercial or hospital environment. If the userof the RadiForce series requires continued
operation during power mains interruptions, it
is recommended that the RadiForce series be
powered from an uninterruptible power supply
or a battery.

Power frequency
magnetic fields
IEC61000-4-8

30 A/m(50 / 60 Hz)

30 A/mPower frequency magnetic fields should be atlevels characteristic of a typical location in a
typical commercial or hospital environment. The
product should be kept at least 15 cm awayfrom the source of power frequency magnetic
fields during use.

Appendix21Electromagnetic immunity
The RadiForce series has been tested at the following compliance levels (C) according to the testing requirements (T)for Professional healthcare facility environments defined in IEC60601-1-2.
The customer or the user of the RadiForce series should assure that it is used in such an environment.

Immunity test

Test level (T)Compliance level(C)

Electromagnetic environment GuidancePortable and mobile RF communications
equipment should be used no closer to any partof the RadiForce series, including cables, than
the recommended separation distance calculated
from the equation applicable to the frequency of
the transmitter.
Recommended separation distance

Conducted
3 Vrmsdisturbances induced 150 kHz - 80 MHz
by RF fields
IEC61000-4-6

Radiated RF fields
IEC61000-4-3

3 Vrmsd = 1.2√P

6 Vrms
ISM a) bands between
150 kHz and 80 MHz

6 Vrmsd = 1.2√P

3 V/m
80 MHz - 2.7 GHz

3 V/m

d = 1.2√P, 80 MHz - 800 MHz
d = 2.3√P, 800 MHz - 2.7 GHz
Where “P” is the maximum output power ratingof the transmitter in watts (W) according to
the transmitter manufacturer and “d” is therecommended separation distance in meters (m).
Field strengths from fixed RF transmitters, as
determined by an electromagnetic site survey b),
should be less than the compliance level in eachfrequency range c).
Interference may occur in the vicinity of equipment
marked with the following symbol.

Note 1

UT is the a.c. mains voltage prior to application of the test level.

Note 2

At 80 MHz and 800 MHz, the higher frequency range applies.

Note 3

These guidelines regarding conducted disturbances induced by RF fields or radiated RF fields may not apply in all
situations. Electromagnetic propagation is affected by absorption and reflection from structures, objects and people.

a)

The ISM (industrial, scientific and medical) bands between 150 kHz and 80 MHz are 6.765 MHz to 6.795 MHz, 13.553
MHz to 13.567 MHz, 26.957 MHz to 27.283 MHz, and 40.66 MHz to 40.70 MHz.

b)

Field strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones and land mobile
radios, amateur radio, AM and FM radio broadcast and TV broadcast cannot be predicted theoretically with accuracy.
To assess the electromagnetic environment due to fixed RF transmitters, an electromagnetic site survey should beconsidered. If the measured field strength in the location in which the RadiForce series is used exceeds the applicable RFcompliance level above, the RadiForce series should be observed to verify normal operation. If abnormal performance isobserved, additional measures may be necessary, such as reorienting or relocating the RadiForce series.

c)Over the frequency range 150 kHz to 80 MHz, field strengths should be less than 3 V/m.

22

Appendix

Recommended separation distances between portable or mobile RF communication equipment
and the RadiForce Series
The RadiForce series is intended for use in an electromagnetic environment in which radiated RF disturbances
are controlled. The customer or the user of the RadiForce series can help prevent electromagnetic interference by
maintaining a minimum distance (30 cm) between portable and mobile RF communications equipment (transmitters)
and the RadiForce series.
The RadiForce series has been tested at the following compliance levels (C) according to the testing requirements (T)of immunity to proximity fields from the following RF wireless communication services.

Test frequency(MHz)

Bandwidth a)(MHz)

Service a)

Modulation b)

Compliancelevel (C)(V/m)

Test level (T) c)
(V/m)

385

380 - 390TETRA 400

Pulse modulation b) 27
18 Hz

27

450

430 - 470

GMRS 460,
FRS 460

FM±5 kHz deviation
1 kHz sine

28

28

710

704 - 787

LTE Band 13, 17

Pulse modulation b) 9
217 Hz9800 - 960

GSM 800 / 900,TETRA 800,
iDEN 820CDMA 850,
LTE Band 5

Pulse modulation b) 28
18 Hz

28

1700 - 1990

GSM 1800;
CDMA 1900;
GSM 1900;
DECT;
LTE Band 1, 3, 4,
25;
UMTS

Pulse modulation b) 28
217 Hz28

2450

2400 - 2570

Bluetooth,
WLAN,
802.11 b/g/n,
RFID 2450,
LTE Band 7

Pulse modulation b) 28
217 Hz28

5240

5100 - 5800

WLAN 802.11 a/n

Pulse modulation b) 9
217 Hz

9

745
780
810
870
930

1720
1845
1970

5500
5785
a)

For some services, only the uplink frequencies are included.

b)

The carrier is modulated using a 50 % duty cycle square wave signal.

c)

The test levels were calculated using the maximum power, separation distance of 30 cm.

The customer or the user of the RadiForce series can help prevent interference from proximity magnetic fields bymaintaining the minimum distance (15 cm) between RF transmitters and the RadiForce series.
The RadiForce series has been tested at the following conformance levels(C) according to the testing requirements(T)of immunity to proximity magnetic fields in the table below.

Test frequency

Test level (T)(A/m)

Modulation a)Compliance level (C)(A/m)

134.2 kHz

Pulse modulation a)
2.1 kHz65

65

13.56 MHz

Pulse modulation a)
50 kHz

7.5

7.5

a)

The carrier is modulated using a 50 % duty cycle square wave signal.

Appendix23For other portable and mobile RF communication equipments (transmitters), minimum distance between portable andmobile RF communications equipment (transmitters) and the RadiForce series as recommended below, according to
the maximum output power of the communications equipment.

Rated maximum
output power oftransmitter(W)

Separation distance according to frequency of transmitter(m)
150 kHz to 80 MHz
d = 1.2√P

80 MHz to 800 MHz
d = 1.2√P

800 MHz to 2.7 GHz
d = 2.3√P

0.01

0.12

0.12

0.23

0.1

0.38

0.38

0.73

1

1.2

1.2

2.3

10

3.8

3.8

7.3

100

12

12

23

For transmitters rated at a maximum output power not listed above, the recommended separation distance “d” in meters (m)can be estimated using the equation applicable to the frequency of the transmitter, where “P” is the maximum output power
rating of the transmitter in watts (W) according to the transmitter manufacturer.
Note 1

At 80 MHz and 800 MHz, the separation distance for a higher frequency range applies.

Note 2

These guidelines regarding conducted disturbances induced by RF fields or radiated RF fields may not apply in all
situations. Electromagnetic propagation is affected by absorption and reflection from structures, objects and people.

24

Appendix

FCC Declaration of Conformity
For U.S.A., Canada Only
FCC Declaration of Conformity
We, the Responsible Party

EIZO Inc.
5710 Warland Drive, Cypress, CA 90630
Phone: (562) 431-5011

declare that the product

Trade name: EIZO
Model: RadiForce MX194

is in conformity with Part 15 of the FCC Rules. Operation of this product is subject to the following two
conditions: (1) this device may not cause harmful interference, and (2) this device must accept any
interference received, including interference that may cause undesired operation.
This equipment has been tested and found to comply with the limits for a Class B digital device, pursuant
to Part 15 of the FCC Rules. These limits are designed to provide reasonable protection against
harmful interference in a residential installation. This equipment generates, uses, and can radiate radio
frequency energy and, if not installed and used in accordance with the instructions, may cause harmfulinterference to radio communications. However, there is no guarantee that interference will not occur ina particular installation. If this equipment does cause harmful interference to radio or television reception,
which can be determined by turning the equipment off and on, the user is encouraged to try to correct
the interference by one or more of the following measures.
* Reorient or relocate the receiving antenna.
* Increase the separation between the equipment and receiver.
* Connect the equipment into an outlet on a circuit different from that to which the receiver is connected.
* Consult the dealer or an experienced radio/TV technician for help.
Changes or modifications not expressly approved by the party responsible for compliance could void theuser’s authority to operate the equipment.
Note
Use the attached specified cable below or EIZO signal cable with this monitor so as to keep interference
within the limits of a Class B digital device.
- AC Cord- Shielded Signal Cable (enclosed)

Canadian NoticeThis Class B information technology equipment complies with Canadian ICES-003.
Cet équipement informatique de classe B est conforme à la norme NMB-003 du Canada.

Appendix25

153 Shimokashiwano, Hakusan, Ishikawa 924-8566 Japan
Carl-Benz-Straße 3, 76761 Rülzheim, Germany
中国苏州市苏州工业园区展业路 8 号中新科技工业坊 5B
1 Queens Square, Ascot Business Park, Lyndhurst Road,
Ascot, Berkshire, SL5 9FE, UK
Moosacherstrasse 6, Au, CH-8820 Wädenswil, Switzerland

www.eizoglobal.com
Copyright © 2018 - 2023 EIZO Corporation. All rights reserved.

00N0N383B1
IFU-MX194
2nd Edition - September 29th, 2023 Printed in Japan.