Infrared Thermal Imaging Camera

High Resolution Infrared Image for Professional Thermographer

InfReC R550seriesHigh speed sampling at maximum
speed 120Hz（R550Pro）■ 120Hz：640×120 with windowing■ 60Hz ：640×240 with windowing■ 30Hz ：640×480 at full window

Automatic movie recording function by PC

Realizes the highest
speed sampling（*）
in the portable
VGA class equipment*Research by Avio in August 2018

■ Software makes movie recording start automatically byexternal trigger input.

1.2M pixels Infrared Thermal Imaging Camera■ Super Resolution Mode (SR MODE) :1280x960 pixel(Standard MODE) : 640x480 pixel■ Spatial Resolution: equivalent to 0.58 mrad

Variety of Lens Lineup Makes Play an Active
Role in Various Measuring Scene■ 21um Closeup Lens ■ 52um Closeup Lens ■ 2x Telephoto Lens
■ 2x Wide Angle Lens ■ 3x Wide Angle Lens

Selectable 2 models for your application■ R550Pro: Measuring range: -40 to 2000℃
Suitable for use in R&D, for making high temperature
measurements, and for measuring high temperature■ R550: Measuring range: -40 to 650℃
Excellent choice for inspection of electrical facilities andremotely located pipes.

For monitoring of welding process.
By high speed sampling time of 120Hz（R550Pro）, occurrence ofsmall spatter and thermal affection can be checked.
120Hz

Page 130Hz

POINT

Page 1

Image of laser welding

Page 2

Page 3

Page 4

Page 5High speed sampling eliminate
missing acquisition of data.

Page 2

Example of laser welder monitoring (Object: Stainless steel, Laser output: 250W, Scan 50mm)

Automatic Movie Recording Feature Built In

Trigger Signal

Start recording bytrigger signal input

R550ProStart RecordingUSB2.0V

AutomaticA

Stop Recording

Start Recording

Trigger Signal Input

Measuring Functions

Recording Movie to PC automatically

Image Display

Stop Recording

Sophisticated On-Line
Software（NS9500Pro）

Power Supply

Real Time Transfer

機能Maximum 1.2million pixel recoding (SR mode)
0.3 million pixels(Take by standard mode withdigital zoom x2.0)

Infrared Detector
Spectral Range
Measuring Range
Sensitivity (NETD)
Accuracy
Frame Rate
Detector Pixels
Recording Pixels
Field of View
Spatial Resolution
Focal Distance
Focus
Auto Function
Color Pallets
Gradation
Visible Camera
Visible/Thermal Fusion
Display Functions
Image Quality Improvement
Point TemperatureTemperature Display in Assigned Region
Line ProﬁleAlarm Function
Temperature Correction
EmissivityInterface

Drift Stabilizer
Storage Device
Data Format
Data Storage
Super Resolution (SR)Quick Panorama
SD Movie Recording
Interval Recording
External Trigger Recording
Voice Recording
Text Annotation
Mass Storage
Movie Data Transfer
Windowing Data Transfer
W1：640×240
W2：640×120
External Trigger Input
Video Output
Alarm Output
External Trigger Input

OthersWith use of 21um close up lens (Spatial resolution 21um)

R550Pro-D

R550

R550-S
R550-D

R550

Uncooled Focal Plane Array (Microbolometer)
8 to 14μm
CC−40 to 2000°−40 to 650°
0.025°C at 30°
C (with S/N improvement)±1°
C *130Hz / 60Hz* 2 / 120Hz*3
7.5Hz
30Hz
7.5Hz
640(H)× 480(V) pixels
Standard mode： 640(H) × 480(V) , Super Resolution (SR mode) mode：1280(H) × 960(V) *4(H) × 24°(V)(with standard lens)32°Standard mode： 0.87mrad , Super Resolution (SR mode) mode ：0 .58mrad equivalent *5
10cm to inﬁnity (with standard lens) *6
Auto / Manual
Auto Scale, Auto Focus, Full Auto
7 pallets (Rainbow, Brightness, Hot-white, Hot-black, etc.)
256 / 32 / 16 / 8 grade
CMOS camera 5M pixels
Side-by-side, Fusion（transparency changeable）, Picture-In-Picture（ transparency changeable）
1 to 8 times continuous zoom (with display positioning scroll), Grid Overlay, 9 images multi-display (replay mode)Denoising, Averaging (with ghost rejection), Edge enhancement
10 Movable Points, Temperature search: MAX/MIN x 1 each, Delta T
MAX, MIN and AVG in BOX (Up to 5 boxes)
N /AHorizontal, Vertical, Horizontal & Vertical
Alarm Display, Alarm Sound, Color Alarm, Alarm Recording,N /A
Alarm Signal Output
Emissivity, Environmental/Background, DistanceMulti-point Correction, Emissivity Table
N /AEmissivity Reverse Calculation
N /AYes
SD card, Conforms to SDHC
Still image：JPEG *7 with temperature data,14bit（with visible image） Movie：SVX ﬁle（Dedicated format）
Still image(JPEG *7 with temperature data,14bit）, Recorded with visible image
Yes
Horizontal equivalent to 100°/ Vertical equivalent to 75°
N /A
Max 3Hz
3 sec to 60 min interval, with Visible image recorded
N /AYes
30sec Recording, replay per a Thermal image
Annotate up to 128 Characters per a Thermal Image. Characters imported from SD Card
Yes
Thermal image: Max：30Hz、Recorded with visible image *8
Thermal image
N /A
W1：60Hz

W2：120Hz
N /A
Automatic recording function by external trigger input
NTSC / PAL Switchable
Contact Signal. No Voltage
N /A
Pulse Signal
N /A
3.5" LCD Monitor (with tilt and brightness adjustment), Color View Finder (with tilt adjustment)
DisplayLaser Pointer (Red, class 2, IEC 60825-1 2007/2014), LED Light, Remote Controller
Auxiliary
Operating Temperature & Humidity −15°C to 50°C, 90%RH (non-condensing)
Storage Temperature & Humidity −40°C to 70°C, 90%RH (non-condensing)
29.4m/sec 2 (3G), 294m/sec 2 (30G)
Vibration & Shock
Conforms to CE regulations (Class A)
EMC

Environment Resistance

Clear imaging and measurement from wide to micro with use of option lensUSB2.0Storage & Output

1.2 million pixels(Taken by SR mode withdigital zoom x2.0)

R550-Pro
R550ProFeatureBasic Performance■Recording movie connected PC automatically by external trigger
input to R550Pro■Data recording linked with test equipment and field facility is available
without configuring I/O system

Specifications

Dust & splash proofBattery Operation
AC Power
Dimensions
WeightStandard Accessories

Protection class IP54 equivalent2.5hours (Typ), Rechargeable Li-Ion battery
100V − 220V AC, 50 / 60Hz
Approx. H121mm×W105mm×D195mm (excluding projection)
Approx. 1.3kg (including Battery Pack)
Wired remote Controller x 1, SD Card x 1, USB Cable x1, Neck Strap x 1, Grip Beltx 1,Rechargeable Li-Ion Battery x 1, Software x 1, Operation Manual x 1, and Carrying Case x 1

InfReC Analyzer NS9500Professional
Standard Software
InfReC Analyzer NS9500Standard *9*1 Only the Range 1 at the environmental temperature from 20 to 30°
C（In other condition, it is ±2°
C or ±2%.）*2 Windowing Mode W1（640×240） *3 Windowing Mode W2（640×120） *4 Still Image Only
*5 This increased resolution results from detecting characteristic within all frames acquired by the SR process and removing such eﬀects as those caused by hand vibration.
*6 For temperature accuracy : 30cm to inﬁnity *7 SAX format for USA and European countries. JPEG is not available.
*8 Thermal image only when image transfer speed at 30Hz
*9 In order to transfer thermal image movie data by R550 /R550-D, you need to upgrade to InfReC Analyzer NS9500 Professional (optional software).

Power Kit

R550 series is not including AC adapter, battery charger and AC cables.
Please order below "Power Kit A" with R550 series main unit.
Item

13mm

Capture detailed image of ﬁne pattern

Measuring Distance and F.O.V

Field of View and Spatial Resolution are the same magnification withmeasuring distance.
2x Telephoto Standard
2x Wide3x WideLens
Lens
Angle Lens Angle Lens
Field of View（H）×（V）
29×22cm 57×42cm 128×92cm 211×149cm
0.44mm
0.9mm
1.74mm
3.1mm
L＝1m Spational Normal ModeResolution Super Resolution（SR mode)*3 0.30mm
0.6mm
1.16mm2.1mm

Power Kit

Model

Speciﬁcation/remarks

PKB-110VGM

For 100 to 120V region

PKB-220VGM

For 220 to 240V region

AC adapter x 1,battery charger x 1
and AC cable x 2Options

Lens

Lens TypeAccessory

Options
2x Telephoto Lens
2x Wide Angle Lens
3x Wide Angle Lens
21μm Closeup Lens
52μm Closeup Lens
AC adaptor
3-terminal Battery Pack
3-terminal Battery Charger
LCD Hood

Model
IRL-TX02D
IRL-WX02D
IRL-WX03D
IRL-C021UB20
IRL-C052UB
SDI50-9-UD-P14-110V / 220V
2NCR18650BF-G-0077

BC2S1P-110V / 220V
IRU-F01A

Speciﬁcation/remarks
16°（H）
×12°
（V）
64°
（H）
× 48°
（V）
×73°
（V）
93°
（H）
13mm(H) × 10mm(V), Working Distance 22mm
33mm(H) × 25mm(V), Working Distance 56mm
110V or 220V

2800mAh
110V or 220V

█ Listed speciﬁcations, appearance and design are subject to change without notice. █ Company and commodity names are trade names or registered trade marks of each company. █ NIPPON AVIONICS Co., Ltd. will not be
responsible for any damage of infrared detectors due to incoming strong light (e.g. laser) through lens(es). █ This product is subject to Japanese Export Control Law. Depending on its destination, prior assessment and authorization may
be required. When exporting from country of initial purchase destination, please be sure to follow that country's export regulations as it may require an export permit beforehand.

Sales Department
Sensing Solution Division
Shimamura-Building, 4475, Ikonobe-cho,
Tsuzuki-ku, Yokohama-shi,
224-0053, Japan
TEL: +81-45-287-0303 FAX: +81-45-287-0307
E-mail : product-irc-e@ml.avio.co.jphttps://www.avio.co.jp/english/
CAT.NO.430-022-E

2209-02-BP-Rv3-R