L1L2| CONNECTION DIAGRAM

L3

FRONT

BOTTOM VIEW

Recommended
100mA fuses
A.4

L2A.3

L3A.2

N

A.1

SDMS-MMS

300V CATIII
L1USB HOST
POWER
24 VDC

A+24V

E.1

GND

E.2+24V

B.5
B.4
MULTILEVELINSTALLATIONTERMINAL
BLOCKS

B.3
B.2

ETHERNET

C

M

MOTOR

A+

H.1

SYNC BI/O NC
NC

H.2
H.3
H.4

PHASE
MARKER

SYNC
I/O

V1W

F.4

SIG

F.5

GND

F.6

DO-NO

F.7

DO-COMMON

F.8

SDMS-MMSFElectro Module
H

A+

I.1

BNC
NC

I.2
I.3

Quick Reference GuideI

I.4

USB DEVICE

BKE

F.3

TEMPERATURESENSOR 1-WireRELAY
OUTPUTWE MAKE INDUSTRY SMARTER

F.2

GND

B

DF.1

PMI

ELMODIS

B.7
B.6

DG

GWH
BKWH
BK

DIN RAIL
EARTHEDWH

DIN RAILELMODIS Sp. z o.o.

Aleja Pokoju 1

elmodis@elmodis.com

www.elmodis.com

31-548 Kraków, Polandtel. +48 531 507 668

Electro module is a part of Elmodis system used to monitor electric induction motor-supplied machines. Measurements of physical quantities used by the system (current, voltage,key phasor) are connected to the module. The device is used for collecting and processing
measurement data, communicating with other SDMS modules and communicating on-linewith Elmodis application located in the Cloud.

| BASIC FEATURES
Х

Direct voltage measurements withthe use of voltage dividers

ХConnection of motor current
measurement through Rogowski
coils

Х

Connection of key phasor signal(inductive or photoelectric sensor)

ХKey phasor outlets forsynchronization of measurementswith other devices

Х

Ethernet interface used for thecommunication with the Internetand other SDMS modulesХ

Two-state output (relay)ХTemperature measurement by1-Wire standard sensors

Х24 VDC supplyХInstallation on DIN TS35 rail

SymbolStatus Description| OPERATION

300V CATIII

Nominalmeasurement voltage:Three-phase four-wire systems (ULN/ULL): 277/480V
Three-phase three-wire systems unearthed (ULL): 480V

Measurement
current range:

3.5 kA (RMS) for coils with sensitivity 100mV/kA @ 50Hz
3 kA (RMS) for coils with sensitivity 120mV/kA @ 60Hz

Accuracy:

1.5% per measurement range

No Internet accessTemperature
measurement:

up to 8 1-WIRE sensors (DS18B20+ standard)

Collecting electricalmeasurements

Relay output (SELV):

1 A / 30 VDC (resistive)

Phase marker:

NPN/PNP sensors support

Power supply (SELV):

24 VDCPower consumption:<6W

Key phasor highEthernet:

10/100 Mbit

Temperature measurement
by sensors

Operating temperature:-20 ÷ 50 [°C]

Storage temperature:-20 ÷ 85 [°C]

Service required - contact
Elmodis customer service
Stable communication
with the Cloud

Installation may only be performed by adequately authorized persons.

No communication with the Cloud

Prior to the commencement of installation
works read the ELMODIS installation andstart-up manual. Prior to the commencement of installation works it is absolutely
required to switch off the electrical supplyof machine at which the Elmodis system
is installed in the switching station. Failure to comply with the above requirement
may result in permanent disability ordeath. Special care should be taken during
installation in order not to damage the existing devices or interrupt their operation.

No electrical measurements
Correct operationof the key phasor

No response from a sensor

Operating humidity range: 5% to 90% without condensation

Communication

The symbol designating selective collection of electronicequipment. It is forbidden todispose used equipment withother waste.

Pollution degree:2

No communication

Maximum altitude:

2000 mMachine OK

Mounting method:

TH 35 rail (acc. to PN-EN 60715 standard)

Dimensions:

37 x 115 x 125 [mm]Weight:0.3 kgModule identification
CASING
BACK-LIGHT

ELMODIS.COM

Measurement category:

Failure-free operationof the system| SAFETY| TECHNICAL DATA

Machine-related alarmIntermittent

Static