LAB2BOzone Generator

OZONE

RESEARCH

BENCH-TOP

VARIABLE Applications- Research and Development- Education

 Main characteristics- Ozone generator producing up to 10 g/h forlaboratory use

The LAB2B is a corona discharge type ozone
generator with variable ozone output. Producing up
to 4g/h using air and 10g/h using oxygen. The LAB2B
is designed specifically for laboratory research.

MAIN FEATURES
Variable ozone output up to 10g/h
Operate under vacuum or at maximum
pressure of 10psig
Illuminating switches indicating ozone
production and faults

Air cooledOperating and maintenance manual included
performance graphs
Full twelve month warranty and technicalbackup facilities

OZONE TECHNOLOGY: LAB2B OZONE SYSTEMSThe LAB2B ozone generator is a small air-cooled unit specificallydesigned for bench use incorporating function indicators, feed gas
flowmeter and variable output control.
Output variation is manually adjustable using a control knob
mounted on the front panel.

Operating on various feed gases such as dried air or oxygen theLAB2B is capable of producing concentrations up to 10% volume.

HOW IT WORKS
Ozone is produced when dry air or oxygen gas is passed over theceramic dielectric of an ozone generating module. The module ispowered by a high voltage/high frequency power board.
The electronic power board is designed for either intermittent orcontinuous operation.

The ceramic dielectric is housed within a finned heat sink block
which is air cooled by fan assisted atmospheric air.

TECHNICAL DATAPower Supply

PowerConsumption

ltr/minOxygen%v/ph/HzW

2-5

15-100230/1ph/50 or
115/1ph/60

105Ozone Output(2)Feed Gas Flow Rate

g/h

lb/h

g/h

lb/hltr/min
air

4.0

0.14

10.00.35

4-10Model

LAB2B

VariableOutput ControlOzone Output(1)(1) Feed Gas: Dry Air-60°C Dewpoint(2) Feed Gas: 100% Oxygen

Operating method:
Module Cooling Medium:
Connections:Vacuum or Pressure (10psi max.)
Ambient Air (fan assisted)
PVDF compression Fitting to suit 8 mm (0.31 inch) OD PIPEStandards


Remote controls and alarms- BS EN ISO 9002:94- CE Approval- Ozone On-Off: green illuminator switch- Fault Red:illuminator switch- Flowmeter:
2–10 L/min
MaterialsOptions- Enclosure: mild steel, epoxy coated- Module:
316 stainless steel electrode assembly inside aceramic dielectric tube supported by P.T.F.E end caps

- Additional LAB2B units for larger ozone output

DIMENSIONS

lxhxw

Model
LAB2B

Weightmm

inchkglb

350 x 160 x 300

13.8 x 6.3 x 11.8

613.23

Width

LengthFLOWMETERHeightON-OFFFAULT
VARIABLE

FRONT VIEW

Contacts

SIDE VIEW

WWW.DEGREMONT-TECHNOLOGIES.COM

REAR VIEW

Manufacturers’ Representative:

Ozonia North America• info-ozonia@degtec.com• + 1 201 794 3100

Ozonia International Ozone• info-ozoniaCH@degtec.com• + 41 44 801 8511

Ozonia France• info-ozoniaFR@degtec.com• + 33 1 46 253 950

Ozonia Triogen UK• info-triogen@degtec.com• + 44 141 810 4861

Ozonia Russia OOO• info-ozoniaRU@degtec.com• + 7 831 220 3256

Ozonia Korea• info-ozoniaKR@degtec.com• + 82 31 701 9036

Ozonia China• info-china@degtec.com• + 86 10 659 73 860

Ozonia Japan• info-japan@degtec.com• + 81 3 544 46 361

Copyright © 2009 Ozonia North America Inc., Degremont Technologies - DISO22102EN-V2-08/2009 - Subject to change without prior notice, contact Degremont Technologies for more information.