Manostar digital sensor

QDP33

QDP33
List ofproducts

NEW

WO81

RoHS

WO71

FR51A

MS99

MS99S

MS61A-RA

QDP33

EMD8A

EMD7

EMT1

EMTGP1

QDP33
Installation example

EMT1H

EMT6

EMP5A

EMRT1

Horizontal connection

HWS15A

Product code

QDP33

1

D200

DEPa

1
4

4–20 mA

N
PkPa

1–5

Manostar digital sensor

QDP33

QDP33
List ofproducts

Model

QDP33

WO81

Analog output

Pressure unit
Pressuremeasurement
methodMeasured gas• QDP33

1• QDP33

4WO71

Sensor method
Pressure-receivingelement

FR51A

Mounting orientation
Operating ambient
temperaturePower voltage

MS99

Maximum consumption
current

Operating ambient
humidity

MS99S

Insulation resistance

Instrument bodywithstanding pressure

Withstand voltage

Withstanding pressureof pressure-receiving
element

Installation category

Display

Operating altitude

MS61A-RAProtection level

QDP33

Degree ofcontamination
Durable vibration

Zero adjustment
method

2

EMD8A

2

Durable impact
Exterior material

Comparison output

Pressure port

EMD7

• QDP33N
Polarity of piping
connector

EMT1

Connector
Mass• QDP33P

Approx. 30 g

Accessories

EMTGP1Pressure range code

Rating pressure range

LCD display

10255010020030050010000–10 Pa
0–25 Pa
0–50 Pa
0–100 Pa
0–200 Pa
0–300 Pa
0–500 Pa
0–1000 Pa

0 00–10 00
0 00–25 00
0 0–50 0
0 0–100 0
0–200
0–300
0–500
0–1000

120–1 kPa
0–2 kPa

0 00–1 00
0 00–2 00

1010 Pa
2525 Pa
50
50 Pa
100
100 Pa200
200 Pa300
300 Pa500
500 Pa1000
1000 Pa

10 00 10 00
25 0 25 0
50 0 50 0
100 100
200 200
300 300
500 500
1000 1000

102550100
200300
500
1000
12

1
2

1 kPa
2 kPa

1 00
2 00

..........

.......................
........................................................
.................................

1 00
2 00

Comparison output

Analog output

EMT1H

EMT6

EMP5A
4–20 mAoror
1–5

EMRT1

HWS15AManostar digital sensor

QDP33

QDP33
List ofproducts

WO81

External dimension drawing

WO71

FR51A

MS99

MS99S

MS61A-RA

QDP33

EMD8A

EMD7

EMT1

EMTGP1EMT1H

Operation panel

EMT6Unit

EMP5A

Main display
Sub display

EMRT1

Comparison output
1 indication lamp

Comparison output
2 indication lamp
Down key

HWS15AUp keyMode key

Manostar digital sensor

QDP33

QDP33
List ofproducts

Terminal arrangement drawing

WO8112345120

WO71

24FR51A

12MS99MS99S

MS61A-RA

Connection example
QDP33N type
CN1 (RITS connector 6P)

DC powersource

External device

DC powersource

CommonPowercircuit

Input

EMD8A
Load

EMD7Input

Analog output circuit

NPN transistor

QDP33

Input

Comparison output 1 circuit

EMT1

NPN transistor

Input

Comparison output 2 circuit

EMTGP1
External input circuit

EMT1H
External device

QDP33P type
CN1 (RITS connector 6P)

Powercircuit

EMT6

DC power source
Input

EMP5A
LoadAnalog output circuit

PNP transistor

Input
Input

EMRT1Input

HWS15A

Comparison output 1 circuit

PNP transistor
Comparison output 2 circuit

External input circuit

CommonManostar digital sensor

QDP33

QDP33
List ofproducts

WO81

Panel mounting diagram and panel cutting dimensions

WO71

Rear face panel adapter

FR51A

MS99

MS99S
PanelMS61A-RA
QDP33

QDP33

Front face panel adapter 31 mm square

EMD8A

EMD7
Front face panel adapter

Rear face panel adapter

EMT1

EMTGP1EMT1H

EMT6Mounting plate thickness 1 to 6

EMP5A

EMRT1

Panel cutting dimensions

For mounting one piece

When n pieces are installed horizontally

When n pieces are installed vertically55 or more

HWS15AManostar digital sensor

QDP33

QDP33
List ofproducts

Panel mounting diagram and panel cutting dimensions

WO81

Rear face panel adapter

WO71

FR51A

MS99

MS99S
Panel

MS61A-RA

QDP33

QDP33

EMD8A

Front face panel adapter 36 mm square

EMD7
Front face panel adapter

Rear face panel adapter

EMT1

EMTGP1EMT1H

EMT6Mounting plate thickness 1 to 6

EMP5A

Panel cutting dimensions

EMRT1
For mounting one piece

When n pieces are installed horizontally

When n pieces are installed vertically55 or more

HWS15AManostar digital sensor

QDP33

QDP33
List ofproducts

WO81

Metallic bracket set for mounting use (separately sold) Installation diagram
1

WO71

FR51A

MS99

MS99S
Supplied M3 screws
are used.

MS61A-RA

QDP33

EMD8A

EMD7
Metallic mountingbracket

EMT1

EMTGP12
EMT1H

EMT6

EMP5A

EMRT1

HWS15AManostar digital sensor

QDP33

QDP33
List ofproducts

Mode changeover (1)WO81
Sub setting mode (SET2-MODE)

Before turning on of power supply

WO71[Turn on power supply]

FR51A

Output 1 function Sets the operation mode of output 1 function.

<For 100 Pa range>

Hysteresis mode
Window mode

Minimum scale indication(one second)

MS99
Output 1 mode Sets the output mode of output 1.
Maximum scale indication(one second)

Maximum value indication

MS99S
Zero adjustment

EverysecondSetting value
protection

Output 1 on-delay time

Sets the on-delay time of output 1 function.

Protection setting

Holding down
for two seconds
Clearing ofretained value

Minimum value indication
EverysecondHolding down
for two seconds
Clearing of retained value

Measurement pressure indication
Normal display

MS61A-RA

Blinks onceInversion of main
display color

Output 1 off-delay time

Alternates between
Set and Cancel

Sets the off-delay time of output 1 function.

QDP33

0 to 600 seconds(in units of 1 second)

Cancellation ofprotection
Output 1 power supply on-delay time

Sets the on-delay time of output 1 power supply.

Blinks once

Holding down for
two seconds

Pressure indication mode

0 to 600 seconds(in units of 1 second)

EMD8A

0 to 20 minutes(in units of 1 minute)

EMD7

Holding down for
two seconds

When no key has been pressed for 15 seconds,
the mode automatically returns to the pressure
indication mode.
(Excluding sub setting mode and detailed
setting mode)
Upper limit over range indication
Single pressure range: +110% FS or higher

Output 2 function Sets the operation mode of output 2 function.
Hysteresis mode
Window mode

Holding down
for two seconds

Holding down
for two seconds

EMT1

Output 2 mode Sets the output mode of output 2.
OFFN.C.
N.O.

EMTGP1

Lower limit over range indication

Display during sleep
Output 2 on-delay time

Sets the on-delay time of output 2 function.

0 to 600 seconds(in units of 1 second)

Holding down for two seconds

Main setting mode

Output 2 off-delay time

Sets the off-delay time of output 2 function.

EMT1H

EMT6

0 to 600 seconds(in units of 1 second)

EMP5A
Output 1 function
In hysteresis mode

Output 1 function
In window mode

Output 2 power supply on-delay time

Sets the output 2 power supply on-delay time.

0 to 20 minutes(in units of 1 minute)

Error indication (when an error occurs: when two errors haveoccurred at the same time)
Measurement
Measurement
Error code indication
Error code indicationpressure indicationpressure indication

Output 2 function
In hysteresis mode

Output 2 function
In window mode
After every indication of pressure, the error codes are displayed in order at intervals of one second.

Sub display when the key is held down

Explanation of marks
Press the <<Mode>> key.
Press the <<Up>> key.
At the pressure setting points (P1, P2, Hi, and Lo), pressure values are set.
Setting range: Within the pressure range Units: Pa, kPa
P1: Pressure setting point in hysteresis mode
Hi: Upper limit pressure setting point in window modeP2: Hysteresis pressure setting point in hysteresis mode
Lo: Lower limit pressure setting point in window mode

Press the <<Down>> key.
Press the indicated keys simultaneously.
Blinking

EMRT1

HWS15AManostar digital sensor

QDP33

QDP33
List ofproducts

WO81

Mode changeover (2)Detailed setting mode (SET3-MODE)

WO71

FR51A
Sub display
indication setting

Number setting

Set a two-digit number.

MS99

MS99S

Pressure
indication mode

Holding down
for fourseconds
Holding down
for twoseconds

Pressing <<Up>> key and <<Down>> key changes the character, and pressing the <<Mode>> key switches the digit. For thecharacters that can be set, refer to the list of setting characters below.
Pressing the <<Mode>> key when the cursor is at the rightmost digit confirms the entered characters, and proceeds to the next mode.

MS61A-RA
List of characters that can be set

QDP33

EMD8A
Indication filter

Sets the average move time of the indication filter.

0.2 seconds
1.0 seconds
2.0 seconds
5.0 seconds
10.0 secondsSets the average move time of the output filter.

0.2 seconds
1.0 seconds
2.0 seconds
5.0 seconds
10.0 seconds

EMD7
Output filter

EMT1

EMTGP1

EMT1H

Indication low cut Sets the indication low cut value.
0% to 5% FS(in units of 1% FS)

Output low cut

Sets the output low cut value.
0% to 5% FS(in units of 1% FS)

EMT6
Main display color Sets the display color during indication of measured pressure.
White (measured pressure in white, others in red)Red (measured pressure in red, others in white)EMP5A

Low power consumption Sets the power consumption mode.
Low consumption OFF
Low consumption ON
Sleeps in three minutes.

EMRT1

HWS15A

Measurement indication
Sets the symbol inversion for measurement indication value.
symbol inversion
OFFON

Analog output inversion

Sets inversion of analog output.
OFFON

Test modeEnters the test mode, and sets a simulated pressure value. (Unit: Pa or kPa)
Pressure setting range:

Blinking
Setting clear

Blinking

Returns the setting value to the initial setting.
NO (does not clear the setting value)
YES (clears the setting value)Manostar digital sensor

QDP33

QDP33
List ofproductsComparison function operation diagrams

WO81

WO71

FR51A

MS99

MS99S

MS61A-RA

QDP33

EMD8A

EMD7

EMT1

EMTGP1

EMT1H

EMT6

EMP5A

EMRT1

HWS15AManostar digital sensor

QDP33

QDP33
List ofproducts

WO81

WO71

QDP33 accessories

RoHS

QDP33.

FR51A
Item number

Cover color

Core wire
insulation sheathcolor

Terminal
numberMS99
CAB-RITS5-15

MS99S
1.0 mm
24

MS61A-RA(TE Connectivity)

QDP33

EMD8A
Item number

EMD7

EMT1

EMTGP1

EMT1H

EMT6

EMP5A

EMRT1

HWS15A-50.1–0.5 mm2

1.0–1.15 mm

Manostar digital sensor

QDP33

QDP33
List ofproducts

QDP33 accessories

RoHS

VT base (Nihon Pisco Co., Ltd.)

WO81

PT base (Nihon Pisco Co., Ltd.)PR base (Nihon Pisco Co., Ltd.)
WO71

FR51A

MS99
Item number

Material

Item number

5

Material

Item number

5Material

5MS99SMS61A-RA

Front face panel adapter 31 mm square

Front face panel adapter 36 mm square

Rear face panel adapter

QDP33

EMD8A

EMD7

Item number

Material

ADP33-31

Item number

Material

ADP33-

Item number

Material

EMT1

ADP33-B

EMTGP1EMT1HMetallic bracket set for mounting use
High-pressure side

Low-pressure side

EMT6

EMP5A

EMRT1
Item number

BRKT-QDP

Material

Item number-H

Material

Item number

Material-L

HWS15A

Warranty
Warranty period
The warranty period for our product is one (1) year from delivery to the location specified by the orderer who makes a direct transaction
with us.

If any failure or defect attributable to us becomes clear during the above warranty period, we will repair the product or supply asubstitute product free of charge. However, even during the warranty period, we will exclude the product from the scope of the warranty
if the failure or defect corresponds to any of the following:(1) The failure or defect was caused by an unreasonable condition, environment, handling, or usage not mentioned in theinstruction manual, specifications, and our product catalog.
(2) The failure or defect was caused by a factor other than our product.
(3) The failure or defect was caused by a modification or repair conducted by a party other than us.
(4) The failure or defect was caused by an event that could not be foreseen at the scientific and technical levels at the time ofproduct shipment from us.
(5) The failure or defect was caused by an external factor not attributable to us, such as acts of God and disasters.
Please note that the warranty mentioned here means the warranty for our individual product, and damage provoked by a failure or
defect of the product is excluded from the scope of the warranty.
*This warranty is valid only in Japan.

Application and usage
Our products are designed and manufactured as general-purpose instruments for general industries.
Therefore, our products are not intended for the following uses, and our products used in such a manner are outside the scope ofapplication.
(1) Equipment that is anticipated to greatly affect lives and properties, such as nuclear power generation, aviation, railways, marinevessels, vehicles, and medical devices(2) Utilities that include electricity, gas, and service water(3) Use in outdoor locations and under similar conditions or environments other than those stipulated in the instruction manual(4) Usage to which considerable safety consideration and attention equivalent to (1) and (2) above need to be given

ServiceBecause the product price does not include service expenses, such as the dispatch of engineers, we will separately charge for theexpenses in the following cases:(1) Instruction for installation and adjustment and a witnessed test run(2) Maintenance inspection, adjustments, and repairs(3) Technical guidance and technical education(4) Witnessed inspections of products at our factory<<Note>> The product specifications and information in this catalog are subject to change without prior notice for product improvement orother reasons.

For order placement, contactGeneral
agent

Manostar Co., Ltd.

1-2-3 Nishishiriike-cho, Nagata-ku, Kobe City, Hyogo 653-0031
TEL. +81-78-621-7000 FAX. +81-78-621-7788

Manufacturer

Yamamoto Electric Works Co., Ltd.

1-2-3 Nishishiriike-cho, Nagata-ku, Kobe City, Hyogo 653-0031
TEL. +81-78-631-6000 FAX. +81-78-631-6020

SEPTEMBER 2021 YAMAMOTO ELECTRIC WORKS CO., LTD., JAPAN