| CONNECTION DIAGRAM

FRONTBOTTOM VIEW

NC

F.1

24VDC (OUT)

F.2

GND

F.3

RS-485A+ (RX/TX)

F.4

B- (RX/TX)

F.5

SDMS-VMS-E

1.4 SHIELD
1.3 TEMP
1.2 GND
1.1 VIB+PMI

D.2

D

GND D.3

2.3 TEMPWE MAKE INDUSTRY SMARTER

2.2 GND
2.1 VIB+POWER
24 VDC +24V A.1
GND A.2+24V D.1

2.4F

3.4 SHIELD

A
3.3 TEMP
3.2 GND

USB HOSTB

B

3.1 VIB+4.4 SHIELDETHERNETVIBRATIONSENSOR
14.3 TEMP
4.2 GND
4.1 VIB+VIBRATIONSENSOR
2SDMS-VMS-E

PHASE
MARKER

Vibration Module
Quick Reference Guide

USB DEVICEEEVIBRATIONSENSOR
3

VIBRATIONSENSOR4

DIN rail

ELMODIS Sp. z o.o.

Aleja Pokoju 1

elmodis@elmodis.com

www.elmodis.com

31-548 Kraków, Polandtel. +48 531 507 668

Vibration Module is a part of Elmodis system used to monitor electric induction motor
driven machines. Vibration module constitutes an extension of systems with Electro
Module as a basic module. The task of Vibration Module is to read and preprocess datafrom vibration sensors. Vibration sensor signals are sampled with the frequency of 32kHz. In order to synchronize electrical measurements with vibration measurements, themodule is equipped with additional key phasor input installed at the machine shaft. Thisenables spectral analysis of a signal within a wide range of frequencies and detectionof vibration pattern anomalies with reference to a specific angular position of the shaft.
Communication between Electro and Vibration modules is executed via Ethernet.

| BASIC FEATURESХUp to four vibration sensors
supportedХ

Support of IEPE (IntegratedElectronic Piezoelectric)
sensors.

ХTemperature measurementfrom vibration sensors with an
integrated temperature sensor

Х

Sampling of signal with thefrequency of 32 kHzХ

Ethernet interface used
for communication with theInternet and otherSDMS modules

Х

24 VDC supplyХInstallation on DIN TS35 rail

SymbolDescription| OPERATION

Vibration measurement range:0 ÷ 30 [g] (for a sensor withthe sensitivity of 100 mV/g)

Nominal supply current ofvibration sensors:

4.7 mA

Measurement
track according to:

ISO 2954 standard

Temperature
measurement range:

10 ÷ 120 [°C] (for a sensor withthe sensitivity of 10 mV/°C)

Temperature measurement
accuracy:±0.5 °C

Power supply (SELV):

24 VDCPower consumption:< 7.5W

Communication:

10/100 Mbit Ethernet

Operation temperature:-20 ÷ 50 [°C]

Sensor circuit shorted /sensor damaged

Storage temperature:-20 ÷ 85 [°C]

Operating humidity range:

5% to 90% without condensation

Machine OK

Maximum altitude:

2000 mMounting method:

TH35 rail (acc. to PN-EN 60715 standard)

Dimensions:46 x 115 x 125 [mm]Weight:0.4 kgStable communication
with the Cloud
No communication withthe Cloud
No Internet accessCollecting vibration
measurements

Prior to the commencement of installation works
read the ELMODIS installation and start-up manual. Prior to the commencement of installation
works it is absolutely required to switch off theelectrical supply of machine at which the Elmodis
system is installed in the switching station. Failure to comply with the above requirement may result in permanent disability or death. Special careshould be taken during installation in order not todamage the existing devices or interrupt their operation.

Sensor properly connected,
in stand-by modeNo vibration sensor or open
circuit

Module identification
CASING
BACK-LIGHT

ELMODIS.COM

IEPE (Integrated Electronic Piezoelectric)

Service required - contact
Elmodis customer service

Installation may only be performed byadequately authorized persons.

| TECHNICAL DATA
Vibration sensor standard:

Failure-free operation of thesystem| SAFETY

The symbol designating selective collection of electronic equipment. It is forbidden to dispose used equipment withother waste.

Status

Machine-related alarmIntermittent

Static