Accessories
List ofproducts

WO81

RoHS

Flange set to be mounted on Pitot tube

WO71
Itemnumber

FR51A

Applicable Pitottube
PTKPTF

MS99

NameMaterial
1
NBR

MS99S

NBR

PTS
PTTQ’ty

PTK110101

PTF 414PTS4PTT

4MS61A-RA

QDP33

For vinyl pipe

EMD8A

EMD7

EMT1

EMTGP1EMT1H
Item number

EMT6

EMP5Amm

PTTS

300 S300PTTS

500 S500

PTTS

800 S

800

Material Panel cutting
10 5

For metallic pipe

EMRT1

HWS15A

Item numbermm

PTTS MT 300 S

300PTTS MT 500 S500

PTTS MT 800 S

800

Material Panel cutting
10 5Accessories
RoHS

List ofproducts

WO81

Dimension drawing

Characteristics brought by inclination ofWO71

FR51A

MS99

MS99S

MS61A-RA

QDP33

EMD8A

EMD7

EMT1Pressure detector
Static pressure hole (construction at the customer)

EMTGP1

EMT1H

EMT6

EMP5A

EMRT1

Precautions on use of pressure detector

HWS15A

Accessories
How to use the Pitot tube
List ofproducts

WO81

Measurement method of static pressure and dynamic pressure
Measurement method of static pressure

WO71

1Diagram for explanation of total pressure, staticpressure, and dynamic pressure

FR51A

MS99

MS99SMeasurement method of dynamic pressure

MS61A-RAPt (total pressure) = Ps (static pressure) +Pd (dynamic pressure)

QDP33
1.

EMD8A

EMD7

2.

Measurement method of static pressure and dynamic pressure

EMT1

EMTGP1

EMT1H

the Pitot tube
2

EMT6

3

EMP5A
3EMRT1

HWS15A

Measurement points
in Pitot tube

10

Composite Pitot tube

20

=0.9

r1
r2
r30
R
0 548R
0 707R

r4
r50 837R
0 949RNEW AEROEYE

Accessories
How to use the Pitot tube
List ofproducts

Guide for Pitot tube installation location

WO81

WO71

FR51A

MS9990

D4DMS99S

MS61A-RA

10D

4D

QDP33

EMD8A

19D

4D

EMD7

EMT1

EMTGP1
D4D
EMT1H

EMT6
12D

4D
EMP5A

EMRT1

HWS15A

WarrantyWarranty period
The warranty period for our product is one (1) year from delivery to the location specified by the orderer who makes a direct transaction
with us.

If any failure or defect attributable to us becomes clear during the above warranty period, we will repair the product or supply asubstitute product free of charge. However, even during the warranty period, we will exclude the product from the scope of the warranty
if the failure or defect corresponds to any of the following:(1) The failure or defect was caused by an unreasonable condition, environment, handling, or usage not mentioned in theinstruction manual, specifications, and our product catalog.
(2) The failure or defect was caused by a factor other than our product.
(3) The failure or defect was caused by a modification or repair conducted by a party other than us.
(4) The failure or defect was caused by an event that could not be foreseen at the scientific and technical levels at the time ofproduct shipment from us.
(5) The failure or defect was caused by an external factor not attributable to us, such as acts of God and disasters.
Please note that the warranty mentioned here means the warranty for our individual product, and damage provoked by a failure or
defect of the product is excluded from the scope of the warranty.
*This warranty is valid only in Japan.

Application and usage
Our products are designed and manufactured as general-purpose instruments for general industries.
Therefore, our products are not intended for the following uses, and our products used in such a manner are outside the scope ofapplication.
(1) Equipment that is anticipated to greatly affect lives and properties, such as nuclear power generation, aviation, railways, marine
vessels, vehicles, and medical devices(2) Utilities that include electricity, gas, and service water(3) Use in outdoor locations and under similar conditions or environments other than those stipulated in the instruction manual(4) Usage to which considerable safety consideration and attention equivalent to (1) and (2) above need to be given

Service
Because the product price does not include service expenses, such as the dispatch of engineers, we will separately charge for theexpenses in the following cases:(1) Instruction for installation and adjustment and a witnessed test run(2) Maintenance inspection, adjustments, and repairs(3) Technical guidance and technical education(4) Witnessed inspections of products at our factory<<Note>> The product specifications and information in this catalog are subject to change without prior notice for product improvement orother reasons.

For order placement, contactGeneral
agent

Manostar Co., Ltd.

1-2-3 Nishishiriike-cho, Nagata-ku, Kobe City, Hyogo 653-0031
TEL. +81-78-621-7000 FAX. +81-78-621-7788

ManufacturerYamamoto Electric Works Co., Ltd.

1-2-3 Nishishiriike-cho, Nagata-ku, Kobe City, Hyogo 653-0031
TEL. +81-78-631-6000 FAX. +81-78-631-6020

SEPTEMBER 2021 YAMAMOTO ELECTRIC WORKS CO., LTD., JAPAN