2300/20 and 2300/25 2300 Vibration
Monitors
Datasheet

Bently Nevada Machinery Condition Monitoring

105M0340 Rev. AB

Description
The 2300 Vibration Monitors provide cost-effective
continuous vibration monitoring and protection capabilities
for less critical and spared machinery. They are specificallydesigned to continuously monitor and protect essential
medium to low criticality machinery in a wide range ofindustries including: oil & gas, power generation, water
treatment, pulp and paper, manufacturing, mining, cement,
and other industries.
The 2300 Vibration Monitors deliver vibration monitoring andhigh vibration level alarming. They include two channels ofseismic or proximity measurement inputs from various
accelerometer, Velomitor and Proximitor types, a speed input
channel for time-synchronous measurements, and outputs
for relay contacts. The 2300/20 monitor features aconfigurable 4-20 mA output which interfaces more points to
a DCS. The 2300/25 monitor features System 1 Classic
connectivity for Trendmaster SPA interface which enables
users to leverage existing DSM SPA infrastructure.
The 2300 Vibration Monitors are designed for using on abroad range of machine trains or individual casings where
the sensor point count fits the monitor’s channel count andwhere advanced signal processing is desired.

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Caution: Two 4-20 mA outputs will
NOT work with an external powered
loop.

Monitor Key Features
2300/20
l

l
l

l

l

l

l

l

l

l

l

l

l
l

2300/25

Two 4-20 mA outputs with internalcurrent loop power supply.
Continuous monitoring and protection
Two acceleration/velocity/proximity
inputs with synchronized sampling foradvanced diagnostics.
One dedicated speed channel
supporting Proximity probes, Magnetic
pickup and Proximity switch type sensors.
Supports process variable on all three
input channels.
Key measurements (Acceleration pk,
Acceleration rms, Velocity pk, Velocity
rms, Displacement pp, Displacement rms,
Speed) real-time provided with alarmconfiguration.
Each channel has one measurement
group, and can add additional two
bandpass measurements and several nX
measurements (depends on the device
availability).
LCD and LED for real time value andstatus display.
Ethernet 10/100 Base-T communication
for configuration using Bently Nevada
Monitor Configuration software(Included) with RSA encryption.
Local contacts for positive engagementof monitor bypass, configuration lockout,
and latched alarm/relay reset.
Two relay outputs with programmablesetpoints.
Three buffered transducer outputs(including Keyphasor signal) providing
short circuit and EMI protection. Buffered
outputs for each signal are through BNC
connectors.
Modbus over Ethernet.
Alarm Data Capture

l
l
l

l

l

l

l

l

l

l

l

l

l
l

2/24

Trendmaster SPA interface.
Continuous monitoring and protection.
Two Acceleration/Velocity/Proximity
inputs with synchronized sampling foradvanced diagnostics.
One dedicated speed channel
supporting Proximity probes, Magnetic
pickup and Proixmity switch type sensor.
Support process variable on all three
input channels.
Key measurements (Acceleration pk,
Acceleration rms, Velocity pk, Velocity
rms, Displacement pp, Displacement rms,
Speed) real-time provided with alarmconfiguration.
Each channel has one measurement
group, and can add additional two
bandpass measurements and several nX
measurements (depends on the device
availability).
LCD and LED for real time value andstatus display.
Ethernet 10/100 Base-T communication
for configuration using Bently Nevada
Monitor Configuration software(included) with RSA encryption.
ocal contacts for positive engagement ofmonitor bypass, configuration lockout,
and latched alarm/relay reset.
Two relay outputs with programmablesetpoints.
Three buffered transducer outputs(including Keyphasor signal) providing
short circuit and EMI protection. Buffered
outputs for each signal are through BNC
connectors.
Modbus over Ethernet.
Alarm Data Capture2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Channel Types

Specifications

Process Variable Channel

InputsSupport most of unit with default on
TemperaturePower Input
DC Input

Channel Hardware Specification

18~36 Vdc, max 7.5 W
Channel Types

ICP Accelerometers
Configurable
Bandpass filter

0.2 Hz to 20 kHz

Scale Factor
range

5 to 1000 mV/g

Full scale range

2 to 80 g peak

Current Sink
Source3.3 mA ± 5%Open Circuit
Voltage-21 to -24 Vdc-0.25 to -22 V (greaterthan lower OK)

Configurable
Lower OK limit-0.25 to -22 V (less thanupper OK)

Accuracy: ±1% of full scale range
Independent 24-bit ADCs on input channels
Supports Bently transducer or 2/3 wires
custom transducer for Accelerometers,
Velomitor and Proximitor.
Speed/Keyphasor
Keyphasor transducers support multiple
events per revolution and event ratios for
speed inputs up to 20 kHz.
Threshold voltage 0.1 Vdc
resolution

Velocity
Configurable
Bandpass filter

0.2 Hz to 20 kHz

Scale Factor
range

5 to 1000 mV/in/s

Full scale range

0 to 50 in/s peakInterfaces
Proximity Transducer Interface

Radial Vibration
Configurable
Bandpass filter

0.2 Hz to 20 kHz

Scale Factor
range

5 to 1000 mV/mil

Full scale range

0 to 160 mil peak-peakSupply Voltage-22.8 to –25.2 Vdc

Maximum Rated
Current

15 mA

Short Circuit
Current

15.1 mA to 23.6 mA

Accuracy±1% of full scale range

Input Impedance 3-wire Voltage Mode, 10 kΩ

Thrust Channel
Scale Factor
range

Configurable
Upper OK limit

RPM range

1 to 120,000

Proximity Switch Interface

5 to 1000 mV/mil

Supply Voltage

3/24-10 to -24 Vdc

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Interfaces

Outputs

Lower Not Ok
limit-2.75 ±0.05 V

Buffered Output

Rpm range

1 to 60,000

Three buffered
outputs are available
on the monitor
through BNC
connectors

Other Inputs
Magnetic Pick up

Relays

Input voltage

Adapt to sensor 284947
outputRelays provide two
dry-contact outputs

RPM range

200 to 120,000

Configuration lock
Latched alarm/relay resetfunction
Monitor Bypass

Activate

0 to 10 kΩ

De-activate

150 kΩ to infinite

Relay Circuit Specification in
Non-Hazardous Area

Button Inputs

Type

Single pole, double
throw

Sealing

Epoxy sealed

Contact life

100,000 cycles @ 5
amps
250 Vac

External button to reset latched alarm andrelay
One buried
buttonprovides 3
functionslll200,000 @ 1 amp, 24
Vdc

Display monitor
information
LCD contrast
adjustment
Reset settings to default

Display Monitor Information
Reset listed
settings to
Defaultlll

May be normallyenergized orde-energized
No output feedback
determination

Contact Inputs
Monitorprovides 3
contact
capabilities
with input
terminals

2 Vibration Outputs
1 Speed Output

User account name
IP Address
FW/HW version

Jumper Between COM & Chassis GND
Jumpers are 2-pin terminal interfaces thatconnects COM to the Chassis ground (GND).
Alternatively, COM can be connected to an
earth ground separately through a terminal.

4/24

Insulationresistance1000 MΩ minimum @500 Vdc

Relay closed contact
resistance1 Ω maximum

Relay open contact
resistance

1 MΩ minimum

Maximum switchedcontact voltage

250 Vac / 250 Vdc

Maximum breaking
contact current

6 A @250 Vac / 6 A@24 Vdc

Maximum switched
power

1500VA AC / 150 Watts
DC2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Relay Circuit Specification in
Hazardous Area

Relay Circuit Specification in
Hazardous Area
Maximumswitched
contact
voltage andcurrent

LEDs

6 A @24 Vac / 5 A @ 30 Vac /5.8 A @24 Vdc / 4 A @30 Vdc

OK

Indicates when the monitor isoperating properly.

4-20mA Output (2300/20)

Protection
fault

Two 4-20 mA outputs with internal current
loop power supply

Indicates hardware fault thatis impacting alarm
determination.

User inhibit

4 to 20 mA output values are proportional to
the full-scale of the associatedmeasurement.

Indicates the alarm/relays
have been intentionallyinhibited from operation.

BypassSoftware configuration may determine thevariable of each output. Voltage compliance:
0 to +12 Vdc range across load.

Indicates user initiated bypass
action.

Relay status

Indicates if relays have been
activated.

Load resistance: 0 to 600 Ω

TX/RX

Indicates the Ethernet status
and monitor communicatingwith remote software.

SPEED/AUX
channelstatusIndicates the speed channel
has valid speed signal input
OR operating correctly when
AUX.

Resolution: 0.3662 µA
Accuracy: 1% over operating temperature
range
Update rate: 100 ms
Configurable with default 2 mA clamp current

Channel Alarm Status

No output feedback determination
Caution: Two 4-20 mA outputs will
NOT work with an external poweredloop.
SPA Output (2300/25)
Input signal
range

High AC: 8 Vpp Low AC: 1.6 Vpp
DC GAP: 0 to -20 Vdc (maxmeasurable AC signal is 1
Vpp).

Accuracy

High/Low AC: ±1% of Full-Scaleat 100 Hz DC GAP: ±0.5 V(measurable AC accuracy:±20 mV)

Frequency
response

10 Hz to 3000 Hz ±5%

Alert LED

Engages if any channel is inalert state.

Danger LED

Engages if any channel is indanger state.
LCD Display

Allows viewing machine speed, vibration
measurements value, setpoints, andconfiguration information.

5/24

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Trend Measurements:

Communications
Ethernet

Ethernet, 10Base-T and
100Base-TX. Conforms toIEEE802.3
RJ-45 for 10Base-T/100Base-TX
Ethernet cabling
Cable length: 100 meters (328 ft.)
maximum

DurationIntervals

10 minutes

1 second

20 seconds

100
milliseconds

Post-Event Data 10 seconds

100
milliseconds

Pre-Event Data

Environmental Limits
OperatingTemperature-30°C to +65°C(-22°F to +149°F)

Storage
Temperature-40°C to +85°C(-40°F to +185°F)

Humidity

Up to 95%, non-condensing

Vibration
Limitation

3g

Battery Life forReal Time
Clock

Powered: 38 years @ 50°C
(122°F)
Un-powered: 12 years @50°C (122°F)

1 minute

Spectrums/Waveforms:

127 mm x 127 mm x 76.2 mm(5 in. x 5 in. x 3 in.)

Weight1.03 kg (2.26 lbs)

Mounting

Panel mount or DIN rail
(adapter included)

DurationIntervals

Pre-Event Data

2.5 minutes

10 seconds

Post-Event Data

1 minute

10 seconds

In the event of a network disruption
between the 2300 and System 1, the2300 can store up to at least 5 alarm
data captures. (This requires System 1
Evolution 21.1 or newer and Firmware
release 3.2 or newer for 2300.)

PhysicalDimensions(Width x
Depth x
Height)1 second

System 1 Evolution Connectivity
2300 monitors connect to System 1 Evolution
and support current value and time-based
data collection of all static values, waveforms,
and spectral data. This includes System1
Software's full suite of plots and tools forcondition monitoring and asset management.
When an alarm is triggered on the 2300
monitor, the following high-resolution alarm
data is forwarded to System 1.

6/24

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Compliance and Certifications
Hazardous Area Approvals

FCC
This device complies with part 15 of the
FCC Rules. Operation is subject to thefollowing two conditions:ll

For the detailed listing of country andproduct-specific approvals, refer tothe Approvals Quick Reference Guide(108M1756).

This device may not cause harmfulinterference.
This device must accept any
interference received, includinginterference that may cause
undesired operation.

For additional technicaldocumentation, please log in to
bntechsupport.com and access the
Bently Nevada Media Library.

EMC

CSA/NRTL/C

EN 61000-6-2 : 2005

Class I, Division 2/ Zone 2

EN 61000-6-4: 2007 +A1

AEx nA nC [ic] IIC T4 Gc
Class I, Division 2, Groups A, B, C,
and D; T4

EN 61326-1: 2013
EN 61326-2-3: 2013

ATEX/IECEx

EMC Directive 2014/30/EU

Electrical Safety

2300/20

EN 61010-1: 2010
LV Directive 2014/35/EU

II 3 GEx na nC [ic] IIC T4 Gc
Ex ec nC [ic] IIC T4 Gc

ATEX
EN 60079-0: 2012/A11:2013

T4 @ Ta = -30°C to +65°C

EN 60079-11: 2012
EN60079-15: 2010

2300/25

EN60079-7: 2015
ATEX Directive 2014/34/EU

II 3 GEx na nC ic [ic] IIC T4 Gc
Ex ec nC ic [ic] IIC T4 Gc

RoHS
RoHS Directive 2011/65/EU

T4 @ Ta = -30°C to +65°C

7/24

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB
001

Ordering Information

2 Sensors and 1 Housingll

For the detailed listing of country andproduct-specific approvals, refer tothe Approvals Quick Reference Guide(108M1756).

l

For additional technicaldocumentation, please log in to
bntechsupport.com and access the
Bently Nevada Media Library.

ll2300 Series Vibration Monitorl2300/20-AA: Monitor with 4-20 ma
Outputs002(including DIN rail mount assembly, manualand monitor configuration software)

00

None

02

Multiple Explosive Atmosphere
Certifications(ATEX/IECEx/CSA)

1 Sensor and 1 Housinglll

A: Approvals Optionl2300/25-AA: Monitor with SPA Outputsl(including DIN rail mount assembly, manualand monitor configuration software)l003

A: Approvals Option
00

None

02

Multiple Explosive Atmosphere
Certifications (ATEX/IECEx/CSA)

l

l

l
l

Bently Nevada 2300/20 Condition Monitoring
System Kit
A: Configuration

8/24

1 - 2300/20 or 2300/25 Monitor
1 - 6 ft. (1.8 m) shielded Ethernetcable
1 - Housing Kit: 105M6193-01(fiberglass housing for
nonhazardous area) or
105M6193-03 (stainless steel
housing for hazardous area) 18.03
x 15.04 x 6.3 in.
1 - Accelerometer sensor(AM3100T2-Z2)
1 - 17 ft. (5.2 m) cable (9571)(Excluding Keyphasor sensor and
24 Vdc power supply1)

2 Sensorsl

2300/20 KIT-AAA-BB

1 - 2300/20 Monitor
1 - 6 ft. (1.8 m) shielded Ethernetcable
1 - Housing Kit: 105M6193-01(fiberglass housing for
nonhazardous area) or
105M6193-03 (stainless steel
housing for hazardous area) 18.03
x 15.04 x 6.3 in.
2 - Accelerometer sensors(AM3100T2-Z2)
2 - 17 ft. (5.2 m) cables (9571)(Excluding Keyphasor sensor and
24 Vdc power supply1)

1 - 2300/20 Monitor
1 - 6 ft. (1.8 m) shielded Ethernetcable
2 - Accelerometer sensors(AM3100T2-Z2)
2 - 12 ft. (3.6 m) cables (9571)(Excluding Keyphasor sensor and
24 Vdc power supply1)

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet004

details in the Accessories section tofollow.

2 Velomitors and 1 Housing
l
l

l

l
l
l

005

105M0340 Rev. AB

1 - 2300/20 Monitor
1 - 6 ft. (1.8 m) shielded Ethernetcable
1 - Housing Kit: 105M6193-01(fiberglass housing for
nonhazardous area) or
105M6193-03 (stainless steel
housing for hazardous area) 18.03
x 15.04 x 6.3 in.
2 - Velomitor sensors (330500)
2 - 17 ft. (5.2 m) cable (9571)(Excluding Keyphasor sensor and
24 Vdc power supply1)

2300/25 KIT-AAA-BB
Bently Nevada 2300/25 Condition Monitoring
System Kit
A: Configuration
001

l
l

1 Velomitor and 1 Housing
l
l

l

l
l
l

2 Sensors and 1 Housingl

1 - 2300/20 Monitor
1 - 6 ft. (1.8 m) shielded Ethernetcable
1 - Housing Kit: 105M6193-01(fiberglass housing for
nonhazardous area) or
105M6193-03 (stainless steel
housing for hazardous area) 18.03
x 15.04 x 6.3 in.
1 - Velomitor sensor (330500)
1 - 17 ft. (5.2 m) cable (9571)(Excluding Keyphasor sensor and
24 Vdc power supply1)

l

l
l

002

1 Sensor and 1 Housingl006

2 Velomitors
l
l

l
l
l

l

1 - 2300/20 Monitor
1 - 6 ft. (1.8 m) shielded Ethernet
cable
2 - Velomitor sensors (330500)
2 - 12 ft. (3.6 m) cable (9571)(Excluding Keyphasor sensor and
24 Vdc power supply1)l

B: Approvals Options

00

None

02

Multiple Explosive Atmosphere
Certifications (ATEX/IECEx/CSAlll

1 Three kinds of power supplies

provide different temperature andpower ranges. Verify the necessary

9/24

1 - 2300/25 Monitor
1 - 6 ft. (1.8 m) shielded
Ethernet cable
1 - Housing Kit: 105M6193-01(fiberglass housing for
nonhazardous area) or
105M6193-03 (stainless steel
housing for hazardous area)
18.03 x 15.04 x 6.3 in.
2 - Accelerometer sensors(200350)
2 - 17 ft. (5.2 m) cables (9571)(Excluding Keyphasor sensor
and 24 Vdc power supply 1)

1 - 2300/25 Monitor
1 - 6 ft. (1.8 m) shielded
Ethernet cable
1 - Housing Kit: 105M6193-01(fiberglass housing for
nonhazardous area) or
105M6193-03 (stainless steel
housing for hazardous area)
18.03 x 15.04 x 6.3 in.
1 - Accelerometer sensor(200350)
1 - 17 ft. (5.2 m) cable (9571)(Excluding Keyphasor sensor
and 24 Vdc power supply1)

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

003

l

l

l
l

1 - 2300/25 Monitor
1 - 6 ft. (1.8 m) shielded
Ethernet cable
2 - Accelerometer sensors(200350)
2 - 12 ft. (3.6 m) cables (9571)(Excluding Keyphasor sensor
and 24 Vdc power supply1)

l

l

l

l
l

l

l

l
l

l

l

l
l
l

1 - 2300/25 Monitor
1 - 6 ft. (1.8 m) shielded
Ethernet cable
2 - Velomitor sensors(330500)
2 - 12 ft. (3.6 m) cable (9571)(Excluding Keyphasor sensor
and 24VDC power supply1)

B: Approvals Option

1 - 2300/25 Monitor
1 - 6 ft. (1.8 m) shielded
Ethernet cable
1 - Housing Kit: 105M6193-01(fiberglass housing for
nonhazardous area) or
105M6193-03 (stainless steel
housing for hazardous area)
18.03 x 15.04 x 6.3 in.
2 - Velomitor sensors(330500)
2 - 17 ft. (5.2 m) cable (9571)(Excluding Keyphasor sensor
and 24VDC power supply1)

00

None

02

Multiple Explosive Atmosphere
Certifications (ATEX/IECEx/CSA
1 Three kinds of power supplies

provide different temperature andpower ranges. Verify the necessary
details in the Accessories section tofollow.

1 Velomitor and 1 Housingl

2 Velomitorsl

2 Velomitors and 1 Housing
l

005

006

2 Sensorsl004

105M0340 Rev. AB

1 - 2300/25 Monitor
1 - 6 ft. (1.8 m) shielded
Ethernet cable
1 - Housing Kit: 105M6193-01(fiberglass housing for
nonhazardous area) or
105M6193-03 (stainless steel
housing for hazardous area)
18.03 x 15.04 x 6.3 in.
1 - Velomitor sensor (330500)
1 - 17 ft. (5.2 m) cable (9571)(Excluding Keyphasor sensor
and 24 Vdc power supply1)

10/24

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

System 1 Software

105M6193-01 Fiberglass NEMA 4X/IP66
weatherproof housing withwindow in door (includes
mounting plate for monitor)

2300/20 can interface to System 1 V16.2 orhigher for expanded condition monitoring andanalysis. System 1 software and the 2300
device connectivity (P/N 3071/13) are soldseparately. Refer to document 108M5214 for
System 1 detailed information.

3071/13-AA-BB
System 1 2300 Series Device Import
A: Not available for 2300 monitor
00
B: Quantity of 2300 Monitoring Systems##

Numeric [1->n]

Accessories
110M7102-01

Power supply for DIN rail
mounting, 100/240 AC to
24 DC/1.3 A
Certifications (CID2 by UL)(-25°C ~70°C,
22.5 x 99 x 107 mm) (One power
can drive max 4 monitors.)

106M669401

Power supply for DIN rail
mounting, 110/220AC to
24VDC/5A
Certifications (ATEX, IECEx, CID2
by UL) (-40°C ~70°C,
40 x 130 x 12 mm) (One power
can drive max 10 monitors.)

106M760701

Power supply for DIN rail
mounting, 100/240AC to
24DC/2.5A
Certifications (ATEX) (-25°C~70°C, 40 x 99 x 114 mm) (Onepower can drive max 6
monitors.)

11/24

105M619302

Stainless Steel Housing for 2300
KIT (can only be used in nonhazardous area)

105M619303

Stainless Steel Housing for 2300
KIT (can be used in hazardousarea)2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Proximity Transducer System

The CB2W100 cable is notrecommended for use with the200350 Accelerometer. The O-ringwill not form a proper seal with theaccelerometer.

Refer to the following proximity transducer
system datasheets for details.
1720363300 5mm

1411943300 XL 8mm

1462563300 XL 11mm

1473853300 XL NSV

02120015

9571-AA2

A: Cable Length

Bulk Cable from Proximity sensor tomonitor (500 ft.)Low cost cable for accelerometer

A: From “02” to “99” Increments of 1.0 foot015

15 ft. (4.8 m)

032

32 ft. (9.8 m)

064

64 ft. (19.5 m)

112

112 ft. (34.1 m)

125

125 ft. (38.1 m)

150

150 ft. (45.7 m)

200

200 ft. (61.0 m)

250

250 ft. (76.2 m)
2 Cable lengths greater than 30

meters (100 feet) will experiencesome attenuation of amplitudes athigher frequencies when using theAM3100T2-Z2 Accelerometer.
84661-AA2

Splash Proof Cable for 2-wiretransducer

Armored cable for 2 -wiretransducer

A: From “03” to “99” Increments of 1.0 foot9571 Mod : 285031-AA2
Cable for 2-wire extension with Splash Proof
Connection. This cable assembly provides anequivalent IP66 level of protection.
A: Cable Lengths

CB2W100AAA

Cable for 2-wire transducer

12/24

16

16 ft. (4.8 m)

3232 ft. (9.8 m)

6464 ft. (19.5 m)

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

For conducted RF performance, metal
conduit with both ends grounded isrequired for Proximitor 3300-NSV
cables and Accelerometer 330400
cable.
2 Cable lengths greater than 30
meters (100 feet) will experiencesome attenuation of amplitudes athigher frequencies when using theAM3100T2-Z2 Accelerometer.

286244

106M2223

5-pin contact input connector(Alarm Reset)

106M3408

5-pin contact input connector(Alarm Inhibit, Config lock)106M3211

16-pin transducer input
connector

106M32126-pin relay output connector

106M2231

3-pin power input connector

Accessories

Magnetic mounting base ¼-28
threaded hole

Ethernet Cables

02120015

Bulk Cable from Proximity sensor tomonitor (500 ft.)9571-AA2

Low cost cable for 2-wire transducer

2 Cable lengths greater than 30

138131-AAA

meters (100 feet) will experiencesome attenuation of amplitudes athigher frequencies when using theAM3100T2-Z2 Accelerometer.

Standard 10 Base-T/100 Base-TX Shielded
Category 5 Cable with RJ-45 connectors (solid
conductor)

Software

A: Cable Length006

6 ft. (1.8 m)

010

0 10 ft. (3.0 m)

025

25 ft. (7.6 m)

040

40 ft. (12.2 m)

050

50 ft. (15.2 m)

075

75 ft. (22.9 m)

085

85 ft. (25.9 m)

100

100 ft. (30.5 m)

100M946501

l

l

l

BNMC version 5.2 orgreater
2300 series monitor
firmware(DVD includes 2300
Series Software Guide)

Additional Informationl

Sparesl

105M6203-01 35mm DIN rail mount andscrews (included with 2300/20
monitor)ll

106M3210

BN Monitor Configuration
SW/FW DVD10-pin 4-20mA output
connector

13/24

2300 Series Operation and Maintenance
Manual (Document 105M0341)
2300 Field Wiring Diagram (Document
106M5801)
2300 Series Software Guide (Document
107M7626)
2300 Series Monitor Installation Guide(Document 121M3029)

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Graphs and Figures

Figure 1: 105M6193-03 Weatherproof Housing

14/24

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Figure 2: 105M6193-01 Weatherproof Housing

15/24

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Figure 3: 200350 Accelerometer Sensor

16/24

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Figure 4: AM3100T2-Z2 Accelerometer Sensor

Figure 5: 330400/330425 Accelerometer Sensor

17/24

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Figure 6: 330500 Velomitor

Figure 7: 330505 Velomitor

18/24

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Figure 8: 330525 Velomitor

Figure 9: 190501 Velomitor

19/24

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Figure 10: 100M0741 Proximity Switch

Figure 11: 284947 Magnetic Pickup

20/24

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Figure 12: 2300 Series Monitor Recommended Clearance

21/24

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Wiring Diagram

Figure 13: 2300/20 Wiring Diagram
2300/20 and 2300/25 use the same interface connector for recorder output or SPA output.

22/24

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Figure 14: 2300/25 Wiring Diagram
2300/20 and 2300/25 use the same interface connector for recorder output or SPA output.

23/24

2300/20 and 2300/25 2300 Vibration Monitors
Datasheet

105M0340 Rev. AB

Copyright 2024 Baker Hughes Company. All rights reserved.

Bently Nevada and Orbit Logo are registered trademarks of Bently Nevada, a Baker Hughes business, in the UnitedStates and other countries. The Baker Hughes logo is a trademark of Baker Hughes Company. All other product andcompany names are trademarks of their respective holders. Use of the trademarks does not imply any affiliation
with or endorsement by the respective holders.
Baker Hughes provides this information on an “as is” basis for general information purposes. Baker Hughes does notmake any representation as to the accuracy or completeness of the information and makes no warranties of any
kind, specific, implied or oral, to the fullest extent permissible by law, including those of merchantability and fitness
for a particular purpose or use. Baker Hughes hereby disclaims any and all liability for any direct, indirect,
consequential or special damages, claims for lost profits, or third party claims arising from the use of theinformation, whether a claim is asserted in contract, tort, or otherwise. Baker Hughes reserves the right to makechanges in specifications and features shown herein, or discontinue the product described at any time without
notice or obligation. Contact your Baker Hughes representative for the most current information.
The information contained in this document is the property of Baker Hughes and its affiliates; and is subject tochange without prior notice. It is being supplied as a service to our customers and may not be altered or its content
repackaged without the express written consent of Baker Hughes. This product or associated products may becovered by one or more patents. See Bently.com/legal.

1631 Bently Parkway South, Minden, Nevada USA 89423
Phone: 1.775.782.3611 (US) or Bently.com/support
Bently.com

24/24