Torque Meter

Specifications
Dimensions

Digital Torque Meter

20 - 140

TME3-G

35HB
20 - 154

Designed to measure rotating torque of precision parts
Ideal for Cap TestingΦDL’Accuracy F.S. +/-0.3% +1 digitcN mModel
TME20CN3-G
TME200CN3-G
TME500CN3-G
TME1000CN3-G
TME2000CN3-G

Nm

gf cm

kgf cmozf inlbf inRange

1 digitRange

1 digitRange

1 digitRange

1 digitRange

1 digit

2-20

0.02

--200-2000

20.2-20.002

2.84-280.02 0.178-1.77 0.002

20-200

0.2

0.2-2

0.002

--2-200.02

28.4-280

0.2

1.78-17.6 0.02

50-500

0.5

0.5-5

0.005

--

5-50

0.05

71-7000.5

4.4-440.05

100-1000

11-100.01--10-1000.1

142-1400

1

8.8-880.1

200-2000

22-20

0.02

--

20-2000.2

284-28002

17.8-177

0.2

Range

1 digit

L’B

H

ΦD

273.5

217

116.5

7-190

Weight[kg]

3.5

Standard Accessories: AC Adapter (BA-7), USB power cable, Pole x 4, Support plate (TME500NN3-G to 2000CN3-G only)

Common Specifications
4 digits

CounterDisplay

6 digitsValues: Run mode

Lighting display by ▼: Peak mode
: 1st, 2nd peak (Double peak) mode
: Data calling / Parameter setting
cNm, Nm, gfcm, kgfcm, ozfin, lbfin

Unit
External output

USB connector (Type C) * Applicable to serial communication

Power supply

DC 5V 1A

AC adapter

AC 100 - 240V +/-10% 50/60HzPower consumption
Operating Temperature

5W or less

TME200CN3-G

0 - 40℃ (No condensation)

Figure of Calibration kit
TMTCL2 with TME3-G

Optional Accessories
Calibration Kit

Connecting Cable
Model586
587

Description
USB type C- type C
USB type C- type A

ModelTMTCL2Applicable Models
TME3-G✓ "Double peak” hold mode for bottle cap measurement

TOHNICHI MFG. CO., LTD.
URL http://www.global-tohnichi.com
E-mail overseas@tohnichi.co.jp✓ Teardrop Shape saddle rubber allows chucking from Φ 7 mm of objects✓ Multiple torque units, cN m, N m, gf cm, kgf cm, ozf in, lbf in

TOHNICHI AMERICA CORP.
URL
http://www.tohnichi.com
E-mail inquiry@tohnichi.com✓ Accuracy of full scale +/- 0.3% + 1 digit✓ Five models cover the range of 20 to 2000 cN m

TOHNICHI SHANGHAI MFG. CO., LTD.
URL
http://www.tohnichi-sh.com
E-mail sales@tohnichi-sh.com
TOHNICHI EUROPE TECHNICAL SUPPORT OFFICE (TETSO)
E-mail europe@tohnichi.com
TOHNICHI ASIA TECHNICAL SUPPORT OFFICE (TATSO)
E-mail tatso@tohnichi.com
The appearance and specifications are subject to change for improvement without prior notice.
All right reserved. No reproduction or republication without written permission.
TOHNICHI Mfg. CO., LTD. All Rights Reserved.

7000.03.24.EN.DRoHS

Digital Torque Meter

TME3-GLoadingDirectionFeatures✓ Easy operation with intuitive icon keys

PEAKRUN✓Special teardrop shape allows chucking from a minimum
diameter of 7 mm to maximum 190 mm.

1ST, 2ND PEAK

✓ The 1st and 2nd peaks holding function,
especially for bottle cap measurement

For small objects

Cosmetic caps

CapMolded bridge
Ring
For standard objects
Bottle caps

1st Peak

2nd PeakDrop width

Potentiometers

Peak holding point

orData transfer

Detection of 1st and 2nd peak torque
1st peak: Starting point of mold bridge detachment
2nd peak: Starting point of cap movement
Drop width: Threshold for preserving 1st torque2 x type C USB port
Power port / Data port

CSV output

Power source

AC adapter

Graph depiction

Torque Meter

Specifications
Dimensions

Digital Torque Meter

20 - 140

TME3-G

35
H
B
20 - 154

Designed to measure rotating torque of precision parts
Ideal for Cap TestingΦDL’

Accuracy F.S. +/-0.3% +1 digit
cN m

Model
TME20CN3-G
TME200CN3-G
TME500CN3-G
TME1000CN3-G
TME2000CN3-G

Nm

gf cm

kgf cmozf inlbf inRange

1 digitRange

1 digitRange

1 digitRange

1 digitRange

1 digit

2-20

0.02--200-2000

20.2-20.002

2.84-280.02 0.178-1.77 0.002

20-2000.2

0.2-20.002

-

-

2-20

0.02

28.4-280

0.2

1.78-17.6 0.02

50-5000.5

0.5-50.005

-

-

5-50

0.05

71-7000.5

4.4-440.05

100-1000

1

1-10

0.01

--

10-1000.1

142-1400

18.8-880.1

200-2000

2

2-20

0.02

-

-

20-2000.2

284-28002

17.8-177

0.2

Range

1 digit

L’B

H

ΦD

273.5

217

116.5

7-190

Weight[kg]

3.5

Standard Accessories: AC Adapter (BA-7), USB power cable, Pole x 4, Support plate (TME500NN3-G to 2000CN3-G only)

Common Specifications4 digits

CounterDisplay

6 digits

Values: Run mode

Lighting display by ▼: Peak mode
: 1st, 2nd peak (Double peak) mode
: Data calling / Parameter setting
cNm, Nm, gfcm, kgfcm, ozfin, lbfin

Unit
External output

USB connector (Type C) * Applicable to serial communication

Power supply

DC 5V 1A

AC adapter

AC 100 - 240V +/-10% 50/60HzPower consumption
Operating Temperature

5W or less

TME200CN3-G

0 - 40℃ (No condensation)

Figure of Calibration kit
TMTCL2 with TME3-G

Optional Accessories
Calibration Kit

Connecting Cable
Model586587Description
USB type C- type C
USB type C- type A

Model
TMTCL2

Applicable Models
TME3-G✓ "Double peak” hold mode for bottle cap measurement

TOHNICHI MFG. CO., LTD.
URL http://www.global-tohnichi.com
E-mail overseas@tohnichi.co.jp✓ Teardrop Shape saddle rubber allows chucking from Φ 7 mm of objects✓ Multiple torque units, cN m, N m, gf cm, kgf cm, ozf in, lbf in

TOHNICHI AMERICA CORP.
URL
http://www.tohnichi.com
E-mail inquiry@tohnichi.com✓ Accuracy of full scale +/- 0.3% + 1 digit✓ Five models cover the range of 20 to 2000 cN m

TOHNICHI SHANGHAI MFG. CO., LTD.
URL
http://www.tohnichi-sh.com
E-mail sales@tohnichi-sh.com
TOHNICHI EUROPE TECHNICAL SUPPORT OFFICE (TETSO)
E-mail europe@tohnichi.com
TOHNICHI ASIA TECHNICAL SUPPORT OFFICE (TATSO)
E-mail tatso@tohnichi.com
The appearance and specifications are subject to change for improvement without prior notice.
All right reserved. No reproduction or republication without written permission.
TOHNICHI Mfg. CO., LTD. All Rights Reserved.

7000.03.24.EN.DRoHS