Motorized Stage Update

Motorized Goniometer Stage：KGW04/KGW06
Precision specifications and Price unchanged by the update.
2-axis (KAW04/KAW06) changes as same as 1-axis(KGW04/KGW06).

Specification changes/Improvements① Additional connector option② Improved sensor board③ Back of top plate④ Dimensions⑤ Model code system①

Additional connector option
Current model

New model

Pigtail

Pigtail

Panel Mount

Added panel mount specifications.
You can choose according to the usage situation such as cable routing and installation space.

②Improved sensor board
Photosensor：EE-SX4320
Power-supply voltage
Control output ( ※ )※ NPN open collector output.

DC5-24V ±10%DC5-24V ±5%

DC5-24V 8mA or less

DC30V 10mA or less

Added following function.
Reverse power supply connection protection,overcurrent protection function on the sensor board
We aim to reduce failures and defects due to incorrect wiring.

Motorized Stage Update

Motorized Goniometer Stage：KGW04/KGW06③Back of top plate

Change the surface treatment on the back of the goniometer stage top plate. This change does not affect performance orquality.

Motorized Stage Update

Motorized Goniometer Stage：KGW04④※ For dimensions other than those specified, please contact us for further information.

Dimensions

Current model
116.8

3676140
322020
φ1240

32

32

φ1213

4-3.5 THRU φ8 C'BORE(Bolt hole for M3)

9.8

Knob
φ4 (H7) 4 DP
φ4 from the opossite (H7) 3 DP

4-M3 4 DP
4-M2 4 DP

Center of rotation

Center of rotation

149.5

4.5

KGW04040-L

28

3.5
2028

1420

32

5.5
8.5

Scale 1 degree equallvernier 0.1 degree

60

2.540

Scale 1 degree equallvernier 0.1 degree32
KGW04060-L

New model

Center of rotation

1428
Scale 1 degree equallvernier 0.1 degree60

3.59.5201428

φ4 (H7) 4DP
φ4 from the opossite (H7) 3DP

4.5

φ124-M2 4DP

2.540
8.56761

Knob

4-M3 4DP

32

Center of rotation

17134032
20116.8

Knob
φ4 (H7) 4DP
φ4 from the opossite (H7) 3DP

Scale 1 degree equallvernier 0.1 degree20

φ124-M2 4DP

4032203

φ129.8326761

134-M3 4DP

4-3.5THRU φ6C'BORE(Bolt hole for M3)

116.8

40
3220
φ12324-3.5THRU φ6C'BORE(Bolt hole for M3)

340
32
20

5.5

9.832

Motorized Stage Update

Motorized Goniometer Stage：KGW06※ For dimensions other than those specified, please contact us for further information.

Dimensions

Current model
136.5
11.5560
50
32

65
5950

60
50
32
φ15

φ12134-4.5 THRU φ8 C'BORE(Bolt hole for M4)Knob

φ4 (H7) 4 DP
φ4 from the opossite (H7) 3 DP

4-M4 5 DP
4-M3 5 DP

Center of rotation

75

Center of rotation

3.550

4Scale 1 degree equallvernier 0.1 degree
14282510.51428251050

Scale 1 degree equallvernier 0.1 degree

KGW06050-L50
KGW06075-L

Center of rotation

Scale 0.5 degree equallvernier 0.05 degree

4.250
KGW06100-L

1428

142825
9.8

Scale 0.5 degree equallvernier 0.05 degree

25
10.4
3.6

125

100

Center of rotation50
KGW06125-L

New model
136.5
6559

11.560
50
32

136.5

65
59

1360
50
32

135

φ15

11.56050
32
φ15
φ126050
324-M3 5DP

Knob
φ4 (H7) 5DP
φ4 from the opossite (H7) 3DP

Scale 0.5degree equallvernier 0.05degree14282510.5142875

Center of rotation

3.550

Scale 0.5degree equallvernier 0.05degree

Scale 0.5degree equallvernier 0.05degree

125Center of rotation

14281428

Center of rotation

25
9.84100
25
10.4
3.6

�-M� �DP

Scale 1degree equallvernier 0.1degree5050

�-M� �DP

Center of rotation

25
1050

4-4.5 THRU φ8 C'BORE(Bolt hole for M4)

φ4 (H7) 5DP
φ4 from the opossite (H7) 3DP

4.2

4-M4 5DP

φ12Knob50④50

Motorized Stage Update

Motorized Goniometer Stage：KGW04⑤

Model code system

Current modelNew modelK GW04040 - □□11 AxisGA

23CodeLR□40mm

3 Height of center rotation (W.D.)
04006054 Sensor cover location

1-axis2-axis

2 Stage size04

440mm60mm※2-Axis［A］is available for only 040.

Specification
L position
Oppsite hand

5 Calbe option
CodeBlankA
B
C
DEFGH
J

Specification
Cable is not included (Standard)
2m
2m One end loose
4m4m One end loose
Only connector (Cable is not included)
Robot cable 2m
Robot cable 2m one end loose
Robot cable 4m
Robot cable 4m one end loose

K GW04040M - L C - □11 AxisGA

21-axis2-axis

KGW04040T - LC - A

67６ Motor option
CodeSpecification
CStandard

3 Height of center rotation (W.D.)

BlankAB
C
DEFGH
J

□40mm

040
060

40mm
60mm

T

PigtailM

Panel Mount

4 Connector specifications

LR→

5７ Cable option

04

Code・KGW04040 - LA

42 Stage size

5 Sensor cover location◆ Example of model code change

3Specification
L position
Oppsite hand

CodeSpecification
Cable is not included (Standard)
2m2m One end loose
4m
4m One end loose
Only connector (Cable is not included)
Robot cable 2m
Robot cable 2m one end loose
Robot cable 4m
Robot cable 4m one end loose

Motorized Stage Update

Motorized Goniometer Stage：KGW06⑤Model code system

Current model

New model

KGW 06050 - □□11 Axis23

4

5

K GW06050M - L C - □1 AxisGA

1-axis2-axis

06

□60mm

050
075
100
12550mm
75mm
100mm
125mm

CodeSpecification
L position
Oppsite hand21GA

1-axis2-axis

06

□60mm

050
075
100
12550mm
75mm
100mm
125mm

32 Stage size

3 Height of center rotation (W.D.)

3 Height of center rotation (W.D.)

LR

CodeSpecification

７ Cable option
CodeBlankA
BC
DEFGHJ◆ Example of model code change

・KGW06050 - LA→

KGW06050T - LC - A

7PigtailM

Panel Mount

5 Sensor cover location

６ Motor option
CodeSpecification
CStandard

Cable is not included (Standard)
2m2m One end loose
4m4m One end loose
Only connector (Cable is not included)
Robot cable 2m
Robot cable 2m one end loose
Robot cable 4m
Robot cable 4m one end loose

6TCode5 Calbe option
BlankABCDEF
GHJ

54 Connector specifications

2 Stage size

4 Sensor cover location

4Specification
Cable is not included (Standard)
2m2m One end loose
4m4m One end loose
Only connector (Cable is not included)
Robot cable 2m
Robot cable 2m one end loose
Robot cable 4m
Robot cable 4m one end loose

L
RSpecification
L position
Oppsite handMotorized Stage Update

Motorless Goniometer Stage：KGW04V/KGW06V⑤Model code system

Current model

New model

KGW04040 Ｖ - L - Ｐ28
1

1 Height of center rotation (W.D.)
040
060

40mm
60mm

23KGW04040 T V-L P28
1

1 Height of center rotation (W.D.)

2 Sensor cover location
CodeSpecification

040060

40mm
60mm

LR

3 Sensor cover location
CodeSpecification

L position
Opposite hand

LRL position
Opposite hand

Code
P28
S38
S40

□28 Stepping motor specification□38 Servo motor specification□40 Servo motor specification

3 Application Motor

Code
P28
S38
S40→

250mm
75mm
100mm
125mm

3Panel Mount

Specification

KGW06050 T V-L P28
1

1 Height of center rotation (W.D.)
050
075
1001252 Sensor cover location
Specification

50mm
75mm
100mm
125mm

2342 Connector specifications
TPigtailM

Panel Mount

3 Sensor cover location
Specification

コード
LRL position
Opposite hand

3 Application Motor

L position
Opposite handSpecification□28 Stepping motor specification□38 Servo motor specification□40 Servo motor specification

・KGW06050V - L - P28

M

New model

1 Height of center rotation (W.D.)◆ Example of model code change

Pigtail

KGW04040TV - LP28

KGW06050 Ｖ - L- Ｐ28

コード
P28
S38S40T□28 Stepping motor specification□38 Servo motor specification□40 Servo motor specification

Current model

コードLR

44 Application Motor

・KGW04040V - L - P28

132 Connector specifications

Specification◆ Example of model code change

05007510012524 Application Motor

コードP28
S38S40→KGW06050TV - LP28Specification□28 Stepping motor specification□38 Servo motor specification□40 Servo motor specification