3300 XL 11 mm Proximity Transducer
SystemDatasheet

Bently Nevada Machinery Condition Monitoring

146256 Rev. P

Description
Transducer System
The 3300 XL 11 mm Proximity Transducer System consists of:lll3300 XL 11 mm probe
3300 XL 11 mm extension cable
3300 XL 11 mm Proximitor Sensor1

The 3300 XL 11 mm Proximity Transducer System has a 3.94
V/mm (100 mV/mil) output for non-contacting vibration anddisplacement measurements on fluid film bearing machines.
The large 11 mm tip enables this transducer system to have alonger linear range compared to our standard 3300 XL 8 mm
Transducer System. It is primarily used in the following
applications where the longer linear range is necessary:
l
l

l

l
l

Axial (thrust) position measurements
Ramp differential expansion measurements on steam
turbines
Rod position or rod drop measurements on
reciprocating compressors
Tachometer and zero speed measurements
Phase reference (Keyphasor) signals

The 3300 XL 11 mm Proximitor Sensor is designed to replace
the 7200-series 11 mm and 14 mm Transducer Systems. Whenupgrading from the 7200-series system to the 3300 XL 11 mm
system, every component must be replaced with 3300 XL 11
mm components. In addition, the monitoring system must be
updated. If using a 3500 Monitoring System, an updated
version of the configuration software that lists the 3300 XL 11mm Transducer System as a compatible option is required.
Existing 3300 Monitoring Systems may need a modification.

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

our patented CableLoc design that provides
330 N (75 lb) pull strength.

Contact your local sales and service
representative for assistance.

3300 XL Probes and Extension Cables can also
be ordered with a FluidLoc cable option. This
option prevents oil and other liquids fromleaking out of the machine through the cable’s
interior. The connector protector option
provides additional protection of theconnectors in a humid or moist environment.
Connector protectors are recommended for all
installations and provide increased
environmental protection2. Additionally, the3300 XL 11 mm probe comes standard with alocknut with pre-drilled safety wire holes.

The 3300 XL 11 mm Proximity
Transducer is designed for measuring
position or vibration within afrequency range of 0 to 8 kHz. Typicalapplications of this system include
radial vibration and position, axial
position and Keyphasor
measurements.
Although the terminals andconnector on the Proximitor sensor
have protection against electrostatic
discharge, take reasonableprecautions to avoid electrostatic
discharge during handling.

Notes:
1. Proximitor Sensors are supplied bydefault from the factory calibrated to AISI
4140 steel. Calibration to other target
materials is available upon request.

Proximitor Sensor
The 3300 XL 11 mm Proximitor Sensor has thesame advanced features found in the 3300 XL 8mm Proximitor Sensor. Its thin design allows it
to be mounted in either a high-density DIN-rail
installation or a more traditional panel mountconfiguration. Improved RFI/EMI immunity
allows the 3300 XL Proximitor Sensor to achieve
European CE mark approvals without any
special mounting considerations. This RFI
immunity also prevents the transducer systemfrom being adversely affected by nearby highfrequency radio signals. SpringLoc terminal
strips on the Proximitor Sensor require no
special installation tools and facilitate faster,highly robust field wiring connections.

2. Silicone tape is also provided with each3300 XL extension cable and can be usedinstead of connector protectors. Silicone
tape is not recommended in applications
where the probe-to-extension cableconnection will be exposed to turbine oil.

Proximity Probe and Extension Cable
The 3300 XL 11 mm probe comes in varying
probe case configurations, including armoredand unarmored ½-20, 5⁄8 -18, M14 X 1.5 and M16
X 1.5 probe threads. The reverse mount 3300 XL
11 mm probe comes standard with either 3⁄8-24
or M10 X 1 threads. All components of thetransducer system have gold-plated brass
ClickLoc connectors. ClickLoc connectors lock
into place, preventing the connection frombecoming loose. The patented TipLoc molding
method provides a robust bond between theprobe tip and the probe body. The probe cable
is securely attached to the probe tip utilizing

2/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Extension Cable dc ResistanceSpecifications
Unless otherwise noted, the following
specifications are for a 3300 XL 11 mm
Proximitor Sensor, extension cable and probe
between 0°C and +45°C (+32°F to +113°F) at a
maximum altitude of 2000m, with a -24 Vdc
power supply, a 10 kΩ load, our supplied AISI
4140 steel target that is 31 mm (1.2 in) diameteror larger, and a probe gap of 2.5 mm (100mils). The system accuracy andinterchangeability specifications do not apply
when using a transducer system calibrated to
any target other than our AISI 4140 steel target.

Electrical

Length ofExtension
Cable (m)Resistancefrom Center
Conductor toCenterConductor(RCORE)(ohms)

Resistancefrom Coaxial
Conductor to
CoaxialConductor(RJACKET)(ohms)

4.0

1.0 ± 0.25

0.3 ± 0.1

8.0

2.0 ± 0.5

0.6 ± 0.2

Extension cablecapacitance

69.9 pF/m (21.3 pF/ft)
typicalField wiring

0.2 to 1.5 mm2 (16 to 24
AWG) [0.25 to 0.75 mm2 (18to 23 AWG) with ferrules].
Recommend using threeconductor shielded triad
cable. Maximum length of
305 meters (1,000 feet)between the 3300 XL
Proximitor Sensor and themonitor. See the frequency
response graph for signal
rolloff at high frequencies
when using longer fieldwiring lengths.

Linear Range

4.0 mm (160 mils). Linear
range begins atapproximately 0.5 mm (20mils) from target and isfrom 0.5 to 4.5 mm (20 to
180 mils) (approximately –1
to –17 Vdc). Extended
Deviation from StraightLine (DSL) range is fromapproximately 0.5 to 5.0
mm (20 to 200 mils) (–1 to–19 Vdc). The extended DSL
range is not applicablewhen using zener barriers.

Recommended
Gap Setting

2.5 mm (100 mils)

Proximitor Sensor
Input

Accepts one non-contacting
3300 XL 11 mm Proximity Probeand Extension Cable.

Power

Requires -17.5 Vdc to -26 Vdc
without barriers at 12 mA
maximum consumption, -23
Vdc to -26 Vdc with barriers.
Operation at a more positive
voltage than -23.5 Vdc canresult in reduced linear range.

SupplySensitivity

Less than 2 mV change in
output voltage per volt change
in input voltage.

Outputresistance50 Ω

Probe dc Resistance
Probe
Length (m)Resistance from the Center
Conductor to the Outer
Conductor (RPROBE) (ohms)

1.0

5.9 ± 0.55.0

7.2 ± 0.89.0

8.5 ± 1.1

3/27

3300 XL 11 mm Proximity Transducer System
Datasheet

Incremental
Scale Factor(ISF)

146256 Rev. P

Cross-talk Limitation:

3.94 V/mm (100 mV/mil)±10% includinginterchangeability error
when measured inincrements of 0.5 mm (20mils) over the 4.0 mm (160mil) linear range.

Target Size - Flat Surface
Diameter
Recommended
Minimum

Deviation from best fit straight line (DSL)
Standard DSL
range

Less than ±0.10 mm (±4mils).

Extended DSL
range

Less than ±0.15 mm (±6mils).

SystemPerformanceover extended
temperatures

Frequency
Response

30.5 mm (1.2 in)

Shaft Diameter
Recommended
Minimum

152 mm (6.0 in)

When operating below
recommended limits, there will becross-talk between the probes. To
minimize this effect, close the gap asmuch as possible while still keeping
the needed linear range.

Over a probe temperature
range of -35°C to +120°C (31°F to +248°F) with theProximitor Sensor andextension cable between
0°C to +45°C (+32°F to+113°F), the ISF remains
within ±25% of 3.94 V/mm
(100 mV/mil), the DSL
remains within ±0.51 mm(±20 mils) and theextended range DSL
remains within ±0.59 mm(±23 mils)

A shaft diameter of 76.2 mm (3.0 in)
can keep the cross-talk below 50 mV
by having the minimum separation oftransducer tips; it needs to be at least
64 mm (2.5 in) for axial position
measurements and 58 mm (2.3 in)for radial vibration measurements.
Many factors affect the cross-talkwhen operating below therecommended limit. Consult
Performance Specification, document
144979 for additional information.

Over a Proximitor Sensor
and extension cabletemperature range of 35°C to +65°C (-31°F to+149°F) with the probe
between 0°C to +45°C(+32°F to +113°F), the ISF
remains within ±25% of
3.94 V/mm (100 mV/mil),the DSL remains within±0.51 mm (±20 mils) andthe extended range DSL
remains within ±0.59 mm(±23 mils)

Effects of 60 Hz Magnetic Fields Up to 300
Gauss (5 meter system)
Output voltage in mil pp/gauss

0 to 8 kHz: +0, -3 dB typical,
with up to 305 meters(1000 feet) of field wiring.

4/27

Gap

Proximitor
Sensor

Probe

Ext.
Cable

0.5 mm (20
mil)

0.006

0.001

0.001

2.5 mm (100
mil)

0.033

0.009

0.005

4.6 mm (180
mil)

0.033

0.027

0.007

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

MechanicalConnector-to-connector torque

Probe Tip
Material

Polyphenylene sulfide (PPS).

Recommended
torqueFinger tight

Probe Case
Material

AISI 304 stainless steel (SST).

Maximum
torque0.565 N•m (5 in•lb)

Probe Cable
Specification

75 Ω triaxial, fluoroethylene
propylene (FEP) insulated
probe cable in the following
total probe lengths: 1, 5 or 9
meters.

Minimum Bend
Radius (with orwithout SST
armor)

25.4 mm (1.0 in)

System Weight (typical)

Extension Cable 75 Ω triaxial, fluoroethylene
Material
propylene (FEP) insulated.

Probe

Proximitor
Sensor Material

A380 aluminum

70 g (2.5 oz) (minimum length case, 1m lead,no armor)

System Length

5 or 9 meters including
extension cable

170 g (6.0 oz) (minimum length case, 1m lead,with armor)

Extension Cable
Flexible AISI 302 SST with FEP
Armorouter jacket.
(optional)

For longer case lengths add 1.1 g/mm (1.0oz/in).
For 5 m probe length add 180 g (6 oz) for nonarmored probe or 560 g (20 oz) for armoredprobe.

330 N (75 pounds) probe
Tensile Strength case to probe lead. 270 N(maximum(60 pounds) at probe lead
rated)
to extension cableconnectors.
Connector
Material

For 9 m probe length add 360 g (12 oz) for
non-armored probe or 1120 g (40 oz) for
armored probe.

Gold-plated brass

Extension Cable 45 g/m (0.5 oz/ft)

Probe Case TorqueMaximum
Rated

Armored140 g/m (1.5 oz/ft)
Extension Cable

Recommended

Proximitor
Sensor

15.2 N•m
1/2-20 and 5/8- 45.2 N•m
18 probe cases (400 in•lb) (133 in•lb)
M14x1.5 andM16x1.5 probe
cases
Reverse mountprobes63.3 N•m

Thread Engagement Limits

21.1 N•m(560 in•lb) (187 in•lb)
22.6 N•m

255 g (9 oz)

7.5 N•m(200 in•lb) (66 in•lb)

5/27

Probe
Case
ThreadMaximum Length of Thread
Engagement

1/2-200.750 in

5/8-180.938 in

3300 XL 11 mm Proximity Transducer System
Datasheet

Probe
Case
ThreadMaximum Length of Thread
Engagement

M14x1.5

21 mm

M16x1.5

24 mm

146256 Rev. P

Maximum thread engagement
lengths are per the industry standardof 1.5 times the nominal thread
diameter. A fit class matching that of
the external probe thread is assumed
for all internal threads. Applications
with thread engagement lengths
exceeding the values in the table
above may exhibit binding during
installation. Contact your Bently
Nevada representative if you require
probe thread engagement lengths
exceeding the values above. Bently
Nevada does not replace proximity
probes under warranty due toexcessive thread engagement
lengths.

-52°C to +105°C (-62°F to+221°F)

Relative
Humidity

100% condensing, nonsubmersible whenconnectors are protected.
Tested to IEC 68-2-3 damp
heat.

Probe
Pressure

3300 XL probes are designed
to seal differential pressure
between the probe tip andcase. The probe sealingmaterial consists of a Viton
O-ring. Probes are notpressure tested prior to
shipment. Contact ourcustom design department if
you require a test of thepressure seal for yourapplication

It is the responsibility of the customer
or user to ensure that all liquids andgases are contained and safely
controlled should leakage occur froma proximity probe. In addition,
solutions with high or low pH values
may erode the tip assembly of theprobe causing media leakage into
surrounding areas. Bently Nevada not
be held responsible for any damages
resulting from leaking 3300 XL
proximity probes. In addition, 3300 XL
proximity probes does not be
replaced under the service plan due
to probe leakage.

Environmental Limits
Probe Temperature Range
Operating and-52°C to +177°C (-62°F to
Storage+351°F)
Temperature
Exposing the probe to temperatures
below -34°C (30°F) may causepremature failure of the pressure
seal.
Extension Cable Temperature Range
Operating and-52°C to +177°C (-62°F to
Storage+351°F)
Temperature
Proximitor Sensor Temperature Range
OperatingTemperature

Storage
Temperature-52°C to +100°C (-62°F to+212°F)

6/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Hazardous Area Approvals

Compliance and Certifications

For the detailed listing of country andproduct-specific approvals, refer tothe Approvals Quick Reference Guide(108M1756).

FCC
This device complies with part 15 of the
FCC Rules. Operation is subject to thefollowing two conditions:ll

For additional technicaldocumentation, please log in to
bntechsupport.com and access the
Bently Nevada Media Library.

This device may not cause harmfulinterference.
This device must accept any
interference received, includinginterference that may cause
undesired operation.

cNRTLus3300 XL Proximitor Sensor

EMC

ia

European Community Directive:

When installedwith
intrinsicallysafe zener
barriers per
drawing 141092
or wheninstalled with
galvanic
isolators.

EMC Directive 2014/30/EU
Standards
EN 61000-6-2; Immunity for
Industrial Environments
EN 61000-6-4; Emissions for
Industrial Environments

RoHSnA, ecWhen installedwith nonincendive
circuitconnected per
drawing
140979.

European Community Directive:
RoHS Directive 2011/65/EU

Maritime
2019 Rules for Conditions of Classification,

Class I, Zone 0: AEx/Ex ia IIC T4/T5 Ga;
Class I, Groups A, B, C, and D,Class II, Groups E, F and G,Class III;
T5 @ Ta= -55°C to + 40°C.
T4 @ Ta= -55°C to + 80°C.

Class I, Zone 2: AEx/Ex nA IIC T4/T5 Gc;
Class I, Division 2, Groups A, B, C, andD;Class I, Zone 2: AEx/Ex ec llC T4/T5 Gc;
Class I, Division 2, Groups A, B, C, andD;
T5 @ Ta= -55°C to + 40°C
T4 @ Ta= -55°C to + 80°C

3300 XL Probe

Part 1, 1-1-1/7.7, 1-1-A3, 1-1-A4
2019 Rules for Conditions of Classification,iaPart 1, Offshore Units and Structures

When installedwith
intrinsicallysafe zener
barriers per
drawing 141092
or wheninstalled with
galvanic
isolators.

1-1-4/9.7, 1-1-A2, 1-1-A3

Functional Safety
SIL 2, HFT = 0
SIL 3, HFT = 1

7/27

Class I, Zone 0: AEx/Ex ia IIC T5...T1 Ga;
Class 1, Groups A, B. C, and D,Class II, Groups E, F, and G,Class III;(see Temperature Schedule table tofollow)

3300 XL 11 mm Proximity Transducer System
DatasheetnA, ecWhen installedwith nonincendive
circuitconnected per
drawing
140979.

146256 Rev. P
ia

Class I, Zone 2: AEx/Ex nA IIC T5…T1 Gc;
Class 1, Division 2, Groups A, B, C, D;
Class 1, Zone 2: AEx/Ex ec llC T5…T1 Gc;
Class I, Division 2, Groups A, B, C, andD;

II 1 GEx ia IIC T5…T1 Ga,(see Temperature Schedule
table to follow)(see Temperature Schedule table tofollow)

Ex ia IIIC T90°C ... T280°C Dc
For EPL Dc:

ATEX/IECEx

Ui= -28V

Ci = 1.5 nF

Ii = 140 mA

Li =610 μH

Pi = 0.91 W

3300 XL Proximitor Sensor

nA,ec
II 3 GEx nA IIC T5…T1 Gc,
Ex ec llC T5…T1 Gc,(see Temperature Schedule
table to follow)ia
II 1 GEx ia IIC T4/T5 Ga
Ex ia IIIC T90C/T105C Dc
For EPL Dc:
T105C @ Ta = -55°C to 100°C
T90C @ Ta = -55°C to +85°C
Ui= -28V

Uo= -28V

Ii= 140mA

Io= 140mA

Pi= 0.91W

Po= 0.742W

Ci- 47nF

Co= 1.5nF

Li= 1460µH

Lo= 610µH

Ui= -28V

li= 140 mA

Temperature Schedule
Temperature
Classification

Ambient Temperature(Probe Only)

For EPL Ga and Gc

nA,ec
II 3 GEx nA IIC T4/T5 Gc
Ex ec llC T4/T5 Gc
Ui= -28V
li= 140 mA
T5 @ Ta= -55˚ C to + 40˚C
T4 @ Ta= -55 ˚C to + 80˚C

T1

-55°C to +232°C

T2-55°C to +177°C

T3-55°C to +120°C

T4-55°C to +80°C

T5-55°C to +40°C

For EPL Dc

3300 XL Probe
Probe entity parameters are met
when used with BN extension cables
and connected to BN Prox.

8/27

T280°C @ Ta

-55°C to +232°C

T225°C @ Ta

-55°C to +177°C

T170°C @ Ta

-55C to +120°C

T130°C @ Ta

-55°C to +80°C

T105°C @ Ta

-55°C to +100°C

T90°C @ Ta

-55°C to +40°C

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Hazardous Area Conditions ofSafe UsecNRTLus:ia
Install per Bently Nevada drawing 141092.
nA, ecInstall per Bently Nevada drawing 140979.

ATEX/IECEx:
ia
Install per Bently Nevada drawing 141092.
nA, ecThe Proximitor must be installed so as toprovide the terminals with a degree ofprotection of at least IP54.

9/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

C: Total Length Option

Ordering Information

10

1.0 meter (3.3 feet)

For the detailed listing of country andproduct-specific approvals, refer tothe Approvals Quick Reference Guide(108M1756).

50

5.0 meters (16.4 feet)

For additional technicaldocumentation, please log in to
bntechsupport.com and access the
Bently Nevada Media Library.

90

9.0 meters (29.5 feet)
Five meter probes aredesigned for use withthe five meter
Proximitor Sensor only.

D: Connector and Cable-Type Option

3300 XL 11 mm Proximity Probes

01

Miniature coaxial ClickLoc
connector with connector
protector, standard cable

330702 3300 XL 11 mm Probe, 1/2-20 UNF
thread, with armor

02

Miniature coaxial ClickLoc
connector, standard cable

330707 3300 XL 11 mm Probe, 5/8-18 UNF
thread, without armor

11

Miniature coaxial ClickLoc
connector with connector
protector, FluidLoc cable

330708 3300 XL 11 mm Probe, 5/8-18 UNF
thread, with armor

12

Miniature coaxial ClickLoc
connector, FluidLoc cable

Part Number-AA-BB-CC-DD-EE-

E: Agency Approval Option

A:Unthreaded Length Option00

Not required

05

Multiple Approvals

330701 3300 XL 11 mm Probe, 1/2-20 UNF
thread, without armor

Unthreaded lengthmust be at least 1.0
inch less than the caselength.

3300 XL 11 mm Proximity Probes,
MetricOrder in increments of 0.1 inLength configurations:
Maximum unthreaded length:
8.9 inMinimum unthreaded length:
0.0 inExample: 0 4 = 0.4 in

330703 3300 XL 11 mm Probe, M14 x 1.5 thread,
without armor
330704 3300 XL 11 mm Probe, M14 x 1.5 thread,with armor
330709 3300 XL 11 mm Probe, M16 x 1.5 thread,
without armor

B: Overall Case Length Option
Order in increments of 0.1 inThreaded length configurations:
Maximum case length: 9.9 in
Minimum case length: 1.0 in
Example: 2 4 = 2.4 in

330710 3300 XL 11 mm Probe, M16 x 1.5 thread,with armor
Part Number-AAA-BBB-CC-DD-EE

10/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

A:Unthreaded Length Option

12

Unthreaded lengthmust be at least 26
mm less than thecase length.

Miniature coaxial ClickLoc
connector, FluidLoc cable

E: Agency Approval Option

Order in increments of 2 mm.
Length configuration:
Maximum unthreaded
length: 224 mm
Minimum unthreaded length:
0 mm
Example: 0 6 0 = 60 mm

00

Not required

05

Multiple Approvals

3300 XL 11 mm Reverse Mount
Probe
330705-02-18- CC-DD-EE 3/8-24 UNF
threads

B: Overall Case Length Option

330706-005-046- CC-DD-EE M10 x 1 threads

Order in increments of 2 mm.
Metric thread configurations:

C: Total Length Option

Maximum length:250 mm

10

1.0 meter (3.3 feet)

Minimum length:26 mm

50

5.0 meters (16.4 feet)

Example: 0 6 0 = 60 mm

9.0 meters (29.5 feet)

C: Total Length Option
10

1.0 meter (3.3 feet)50

5.0 meters (16.4 feet)
9.0 meters (29.5 feet)90

D: Connector Option

Five meter probes
are designed for use
with the five meter
Proximitor Sensor
only.

02

Miniature coaxial ClickLoc
connector with connector
protector, standard cable

02

Miniature coaxial ClickLoc
connector, standard cable

11

Miniature coaxial ClickLoc
connector with connector
protector, FluidLoc cable

Miniature ClickLoc coaxial
connector

E: Agency Approval Option

D: Connector and Cable-Type Option
01

Five meter probes aredesigned for use withthe five meter
Proximitor Sensor only.

9000

Not required

05

Multiple Approvals

For a shorter delivery time, order commonlystocked probes. Currently, stocked probes
consist of the following part numbers: 33070100-10-10-02-00, 330701-00-20-10-02-00,
330703-000-050-10-02-00, 330705-02-18-1002-00, 330706-005-046-10-02-00.

11/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

3300 XL 11 mm Proximitor Sensor
330780-AA-BB

00

Standard cable

01

Armored cable

02

Standard cable with connector
protector

03

Armored cable with connector
protector

10

FluidLoc cable

11

Armored FluidLoc cable

A: Total Length and Mounting Option
50

5.0 meter(16.4 feet) system
length, panel mount51

5.0 meter (16.4 feet) system
length, DIN mount52

5.0 meter (16.4 feet) system
length, no mounting hardware

12

FluidLoc cable with connector
protector

90

9.0 meters (29.5 feet) system
length, panel mount13

Armored FluidLoc cable withconnector protector

91

9.0 meters (29.5 feet) system
length, DIN mount92

9.0 meters (29.5 feet) system
length, no mounting hardware

C: Agency Approval Option00

Not required

05

Multiple Approvals

B: Agency Approval Option00

Not required

05

Multiple approvals

Accessories

Country specific approvals may beavailable, consult your local
Customer Care Representative formore information.

3300 XL 11 mm Extension Cable
330730-AAA-BB-CC
Make sure that the extension cable lengthand the probe length, when added together,equal the Proximitor Sensor Total length.
A: Cable Length Option
040
080

1462553300XL 11 mm User Guide

144979Performance Specification

02120015

Bulk field wire. 1.0 mm2 (18 AWG),
3 conductor, twisted, shieldedcable with drain wire. Specify
length in feet.

02173009

Bulk field wire. 1.0 mm2 (18 AWG),
3 conductor, twisted, shieldedcable. Specify length in feet.

138492-01

Replacement panel-mount
mounting pad138493-01

Replacement DIN-mount
mounting pad

01609137

BNC (F) to banana plugs

01609138

Proximitor Connector Test Pin
wiring (two test pins to a BNC (F)connector)

4.0 meters (13.1 feet)
8.0 meters (26.2 feet)

B: Connector and Cable Option

12/27

3300 XL 11 mm Proximity Transducer System
Datasheet

40971-04

04310310

146256 Rev. P

50 Ω cable with two BNC (M)connectors. Use this cable incombination with adapter
01609137 and adapter 01609138
when checking performance ofthe transducer system from theProximitor Sensor test pin holes.
3300XL Proximitor Sensor Panelmount Screws. Package
includes four 6-32 UNC thread
forming mounting screws(Supplied standard with 3300 XL
Proximitor Housings [3300 XL
option]).

03200006

Silicone Self-fusing tape. A 9.1
meter (10 yard) roll of silicone
tape to protect connectors. It iseasy to install and provides
excellent electrical isolation andprotection from theenvironment. It is notrecommended for use inside thecasing of the machine.

40113-02

Connector Protector Kit.
Connector Protector Kit for 3300
XL probes and extension cables,
including connector protectors
and installation tools.
Connector Protector Adapter.

136536-01

Makes our previous 3300
connector protector kitscompatible with 3300 XL probes
and extension cable connectors.

40180-02

Connector Protectors. Package
contains 10 pairs of connector
protectors.

03839410

Male Connector Protector.
Placed on the extension cable toconnect to the female
connector protector on theprobe and provide
environmental protection ofconnectors.

13/27

03839420

Female Connector
Protector. Placed on the probelead to connect to the maleconnector protector on theextension cable and provide
environmental protection ofconnectors. Also placed on theextension cable to slide over theProximitor Sensor connection
and protect it from theenvironment.

330153-01

3300 XL Connector Kit. Used on
3300 XL probes and extension
cables. Contains one set of maleand female ClickLoc connectors,
sleeves and one strip of silicone
tape.

163356

Connector Crimp Tool Kit.
Includes one set of 75 Ω 3300 XL
ClickLoc inserts and connector
installation instructions.
Supplied with carrying case.

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Graphs and Figures

Figure 1: Typical 3300 XL 11 mm 5 m System Over Ambient Testing Range

14/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Figure 2: Typical 3300 XL 11 mm 9 m System Over Ambient Testing Range

15/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Figure 3: Typical 3300 XL 11 mm Probe + 1 m Cable @ Low Temperature(Proximitor Sensor + 4m of Extension Cable @ 25 °C)

16/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Figure 4: Typical 3300 XL 11 mm Probe + 1 m Cable @ High Temperature(Proximitor Sensor + 4m of Extension Cable @ 25 °C)

17/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Figure 5: Typical 3300 XL 11 mm 5 m Proximitor Sensor with 4 m of Extension Cable @ Cold
Temperature(Probe is at 25°C)

18/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Figure 6: Typical 3300 XL 11 mm 5 m Proximitor Sensor with 4 m Extension Cable @ High
Temperature(Probe is at 25°C)

19/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Figure 7: Typical 3300 XL 11 mm 9 m Proximitor Sensor with 8 m of Extension Cable @ Low
Temperature(Probe is at 25°C)

20/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Figure 8: Typical 3300 XL 11 mm 9 m Proximitor with 8 m of Extension Cable @ High Temperature(Probe is at 25°C.)

21/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Figure 9: Frequency Response, magnitude of typical 3300 XL 11 mm Systemwith various lengths of field wiring, no barriers

Figure 10: Frequency Response, phase change of typical 3300 XL 11 mm Systemwith various lengths of field wiring, no barriers

22/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Figure 11: 3300 XL 11 mm Proximity probes, Standard Mount
330701, ½-20 UNF-2A, without armor
330702, ½-20 UNF-2A, with armor
330703, M14X1.5 thread, without armor
330704, M14X1.5 thread, with armor
330707, 5/8-18 UNF-2A, without armor
330708, 5/8-18 UNF-2A, with armor
330709, M16X1.5 thread, without armor
330710, M16X1.5 thread, with armor

Note: Connector Protector only installed on female end when optioned. Both ends available asaccessories.
Figure 12: Installed Connector Protectors

23/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Figure 13: 3300 XL 11 mm Proximity Probes, Reverse Mount
330705, 3/8-24 UNF-2A threads
330706, M10X1 threads

Figure 14: 330730, 3300 XL 11 mm Extension Cable

24/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Figure 15: Panel Mount 3300 XL 11 mm ProximitorSensor

25/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Figure 16: DIN Mount 3300 XL 11 mm Proximitor Sensor
Notes:
1. All dimensions on figures are in millimeters (inches) unless otherwise noted.
2. Standard mount ½-20 UNF thread probes are supplied with ¾ inch lock nut and 7⁄16 wrench
flats.
3. Standard mount M14x1.5 thread probes are supplied with 22 mm lock nut and 12 mm
wrench flats.
4. Standard mount 5⁄8-18 UNF thread probes are supplied with 15⁄16 inch lock nut and 9⁄16
wrench flats.
5. Standard mount M16x1.5 thread probes are supplied with 24 mm lock nut and 14 mm
wrench flats.
6. Reverse mount probes are not available with armor, FluidLoc cable or connector protector
options.
7. Letters inside quotation marks on figures refer to probe ordering options.
8. Stainless steel armor is supplied with FEP outer jacket.
9. FEP jacket is standard on all non-armored probes.
10. Probes ordered with 5 or 9 meter integral cables have a length tolerance of +20%, -0%.

26/27

3300 XL 11 mm Proximity Transducer System
Datasheet

146256 Rev. P

Copyright 2024 Baker Hughes Company. All rights reserved.

Bently Nevada, Keyphasor, M2, Proximitor and Orbit Logo are registered trademarks of Bently Nevada, a Baker Hughes
business, in the United States and other countries. The Baker Hughes logo is a trademark of Baker Hughes Company.
All other product and company names are trademarks of their respective holders. Use of the trademarks does not
imply any affiliation with or endorsement by the respective holders.
Baker Hughes provides this information on an “as is” basis for general information purposes. Baker Hughes does notmake any representation as to the accuracy or completeness of the information and makes no warranties of any
kind, specific, implied or oral, to the fullest extent permissible by law, including those of merchantability and fitness
for a particular purpose or use. Baker Hughes hereby disclaims any and all liability for any direct, indirect,
consequential or special damages, claims for lost profits, or third party claims arising from the use of theinformation, whether a claim is asserted in contract, tort, or otherwise. Baker Hughes reserves the right to makechanges in specifications and features shown herein, or discontinue the product described at any time without
notice or obligation. Contact your Baker Hughes representative for the most current information.
The information contained in this document is the property of Baker Hughes and its affiliates; and is subject tochange without prior notice. It is being supplied as a service to our customers and may not be altered or its content
repackaged without the express written consent of Baker Hughes. This product or associated products may becovered by one or more patents. See Bently.com/legal.

1631 Bently Parkway South, Minden, Nevada USA 89423
Phone: 1.775.782.3611 (US) or Bently.com/support
Bently.com

27/27