XDP

Panametrics
Explosion-proof display
Applications
Explosion-proof display/controller for use with• XMO2 thermoparamagnetic oxygen transmitter• XMTC thermal conductivity hydrogen/gas transmitter• OXY.IQ electrochemical oxygen transmitter• Any other 4 to 20 mA output transmitter

Features• Explosion-proof for Class I, Division 1, Group B, C and D
hazardous areas, Nema 4X• Flameproof for ATEX II 2 G Ex d IIC T6 Gb IECEX Ex d IIC
T6 Gb IP66
• Magnetic through-the-glass keypad• Universal AC input, Optional 24VDC input• 24 VDC power supply for XMO2, XMTC or OXY.IQ• Software for measurement of percent or ppm oxygenor hydrogen• Three-curve software for hydrogen-cooled generator• Programmable process relay contacts

XDP Explosion-proof display
The XDP explosion-proof display provides the measurement ofpercent or ppm oxygen or hydrogen gas. The explosion-proof
XDP is certified for use in Class 1, Division 1, Groups B, C and D
as well as Zone 1 ATEX/IECEX Ex d IIC T6 Gb hazardous areas.
The XDP features advanced microprocessor-based electronics,a magnetic, through-the-glass keypad, a universal power
supply (magnetic), one 0/4 to 20 mA or 0 to 2 VDC analog
output, four process alarms, and a fault alarm.

Auto-verification/auto-calibration
The XDP provides long-term, hands-off operation with thisoptional feature. When initiated, the XDP controls solenoid
valves in the sample system to bring zero and span gases tothe transmitter. Then the XDP software compares calibration
gas readings with factory data to verify proper calibration.
If an adjustment is necessary, the XDP makes corrections
automatically and notifies the user via the front panel display
and alarm contacts.

XDP specifications
Functional
Analog output
Linearized isolated 4 to 20 mA, 0 to 20 mA or NAMUR userselectable, field-programmable output for any range
from 0 to 100 percent or 0 to 10,000 ppm.
Input power
85 to 264 VAC, 47 to 63 Hz, 40 W
24 VDC(±0.2V), 40W
Fuse
1.25 A (AC units)
2.0 A (DC units)
Analog input
4 to 20 mA
Output power supply
24 VDC ±2 VDC at 1.2 Amp

Display accuracy±0.25% of full scale

Ambient temperature range
14°F to 140°F (–10°C to 60°C)
KeypadMagnetic, through the glass, six keys
Display
Four-line, backlit liquid crystal display

Relay outputs• Contact ratings: 2 A, 28 VDC, SPDT
• Programmable as fail-safe or nonfail-safe

Front View

Side View

Back View
XDP dimensions in inches (mm)XDP specifications
Standard software for O2/gas analysis

Environmental

Measurement Range Examples• 0 to 10; 100; 1,000; or 10,000 ppm O2• 0% to 1%; 10%; 21%; 25%; or 100% O2• Weather proof/explosion proof enclosure type 4X/IP66. FM/CSA Class I, Division 1, Group B, C and DRelays• Four process alarms• One fault alarm• Two automatic calibration contacts• Two calibration alarms

Hydrogen-cooled generators
Three ranges• 0% to 100% H2 in air• 0% to 100% H2 in CO2• 0% to 100% air in CO2Relays• Two process alarms• One fault alarm• One normal alarm

PhysicalDimensions (w x h x d)
9 in x 10 in x 9 in. (229 mm x 254 mm x 229 mm)
Weight15 lb (6.8 kg)• Flameproof: KEMA 01ATEX2128 II 2 GD EEx d IIC T6 T85°C
European compliance• Complies with EMC directive 2014/30/EU and low voltage
directive 2014/35/EU
Conduit/cable gland entry
Six 1/2 in NPTF conduit ports
Mounting holes
Four 3/8 in (10 mm) holes

Order information
Record selected option in blank indicated at bottom of form.
XDP explosion-proof display for use with XMO2, XMTC and OXY.IQ

B

Package
2 Explosion-proof/weatherproof enclosure

CPower
2 100-240VAC

D

5 24VDC
Keypad1E

Magnetic keypad
Software1Standard2

Hydrogen-cooled generator		
3 SpecialZ		

0 None		S Special

XDP: _ _ _ _ Use this number to order productPanametrics, a Baker Hughes Business, provides solutions in
the toughest applications and environments for moisture,oxygen, liquid and gas flow measurement. Experts in flaremanagement, Panametrics technology also reduces flareemissions and optimizes performance.
With a reach that extends across the globe, Panametrics’
critical measurement solutions and flare emissions
management are enabling customers to drive efficiency
and achieve carbon reduction targets across criticalindustries including: Oil & Gas; Energy; Healthcare; Water andWastewater; Chemical Processing; Food & Beverage andmany others.
Join the conversation and follow us on LinkedIN
linkedin.com/company/panametricscompany

Copyright 2021 Baker Hughes Company. All rights reserved.
BHCS38783(02/2021)

panametrics.com