Periodontal Probing Instrument

Spec : Pam main unit

Spec : accuracy and measurement

Size

16 mm x 185.8 mm

Press in pressure

Weight48 gStandard voltage

5 VDC (USB bus power)

Measurement
range

Consumption current

0.1 A (Max.)

Minimum display unit

1 mm

USB cord length2m

Display accuracy+/-0.5 mm

Components

General name:- Pam main unit

x1

- Pam disposable tip 7 mmx2

- Pam disposable tip 13 mmx2- PERIOSTUDIO DVDx1

Max. 25 g or less
7 mm

0 – 6 mm

13 mm

0 – 12 mm

JMDN 70697000 Periodontal pocket measurement apparatus

Classification of devices: Specially designated maintenance required medical devices, Class II
Authentication No.:

228AIBZX00027000

Pam - Probing Instrument

N e w & E as y D i g i t ali ze d P r ob i n g
USB single wired footswitch

USB double wired footswitch

USB single wireless footswitch*2

USB double wireless footswitch*2

Input settings*1

Input settings*1

Input settings*1

Input settings*1

Normal
Hold

Enter
Bleed

NormalHold

Pam main unit+
USB single wired footswitch

LeftMemory
BackRightEnter
Bleed

Pam main unit+
USB double wired footswitch

NormalHoldEnter
Bleed

NormalHold

Pam main unit+
USB single wireless footswitch

LeftMemory
BackRightEnter
Bleed

Pam main unit
+USB double wireless footswitch*1 The settings are factory default. They can be changed as desired.
*2 Equipped with “EnOcean” wireless communication technology.

System Requirements for PERIOSTUDIO
Supported OSAttachment tip

Windows 7 SP1
Windows 8.1 or Windows 10*The latest service pack required.

Storage
Display resolution
Drive

Pam disposable tip 7 mm (100 pcs)

Minimum 1GB available disk space for installation

Pam disposable tip 13 mm (100 pcs)

*Additional disk space is required for data storage.

1024 x 768 or above

Pam disposable tip assort

DVD-ROM drive or compatible drive(50 pcs X 2 types)*When the amount of stored data increases, use additional hard disk or other media for storingdata. *Even if the above system requirements are met, the product may not work properly
according to your environment. *Microsoft and Windows are either registered trademarks ortrademarks of Microsoft Corporation in the United States and/or other countries.

For more details NARCOHM

Search*Please note that product specifications and appearance are subject change without prior notice.
*The price is as of February 2019. Consumption tax is not included.

Contact information for this product
Manufacturer and sales company

NARCOHM Co., Ltd.
Head Office/ Factory:
Sales Brunch:1251-239, Yomogida, Saku-city, Nagano, 384-2107, Japan
7F Matsudo Honcho 2nd Bldg, 20-8, Honcho, Matsudo-city, Chiba,271-0091, Japan
E-mail: Info@narcorhm.co.jp URL: https://www.narcohm.co.jp

Pam entirely transforms periodontal probing
System Overview (When USB wireless footswitch in use)

Quick, easy and accurate!

Pam

Depending on the skill and experience probing data
could differ in various way in manual probing.
However, by this new tool.
Pam anyone perform easy and accurate periodontal
probing with 20 - 25g of pressure (recommended) .

Measurement

Transmit input
information

Automatic value collection
Wireless

ReceiverFootswitch

No assistants!
Only press the footswitch, the data obtained by Pam
will automatically be collected in the system.
No longer need to bing back the probe and pen,
Focusing on the probing and save more time by Pam.

UsageInsert Pam to an appropriate position

Disposable tip
Tip length 7mm / 13mm

Tap the footswitch and confirm measurement

Automatic input into the linked application

Tap the footswitch to send pocket
depth data to the application.
Tap the other one to send “bleeding”.

The measured value of pocket depth is automaticallyrecorded in the supplied “PERIOSTUDIO”.

Print and utilize

Tip needle recedes at over
20 - 25 grams of pressure.
Pocket depth is measured
nearby millimeter.

Pocket depth

Fully insert themetal part of tip.

Max 20 – 25g
Pocket depth is measured
by correct pressure always.

Print out also available. The patient data canbe used for futher explaination.