Flow Meters
Product Guide
Measurements that matter.

Ensuring the
integrity of yourflare and flow
measurements.

We know that when it comes toflow measurement, accuracy
and reliability are two of thecritical factors when choosing
a flow meter. Because we serve
such a wide variety of industries,
applications, technologies, anduse cases, we’ve developed oneof the strongest portfolios of flow
meters on the market.

Panametrics, a Baker Hughes business, offers
a wide array of flow meters that improves plantsafety, efficiency, and profitability throughaccurate and reliable flow measurement andflare management solutions that optimize flare
operations and significantly reduces methane
emissions for customers across the globe.
Reliable and accurate, our products andservices are backed by decades of industry
experience and by knowledgeable experts.

02

03

Product type

Flare

Industry
applications

Portable
TransPort PT900
TransPort PT878GC
Clamp-On

Key to industries

AquaTrans AT600
AquaTrans AT868

Power

DigitalFlow DF868

Natural Gas

PanaFlow LC
DigitalFlow GC868

Oil & Gas (Upstream & Downstream)

Process Inline
Ultrasonic

Steel

Pharmaceutical

DigitalFlow GF868
DigitalFlow XGF868i

Panametrics’ flow and flare
solutions cover a wide range
of industries and applications.

Chemicalflare.IQ

Food & Beverages

DigitalFlow GM868
DigitalFlow XGM868i
DigitalFlow GS868
DigitalFlow XGS868i
PanaFlow ZxG

Hydrogen

PanaFlow Z3

BiogasPanaFlow XMT1000
PanaFlow HT

Semiconductor

PanaFlow LZ

Geothermal

Custody Transfer

Water & Wastewater

Sentinel LNG
Sentinel LCT8
Sentinel LCT4
Vortex

Sentinel LCT

04

MV84
MV82
05

Flare and Flow
Well-known for its high-technology
in-line and clamp-on flow meters,
Panametrics provides engineers andtechnicians with the tools designed
to handle the toughest environments
and most challenging applications.
In many industries, process control relies on knowing
exactly how much fluid flows through the pipelines inorder to optimize efficiency, maximize profit, and remain
in regulatory compliance. In a most direct way, flow
meters measure the flow rate, or quantity, of liquids,
gases, and steam using a variety of technologies.

Panametrics also offers a comprehensive
selection of advanced flow instrumentationthat measures all regulated and criticalflow rates to the flare tip, including air,steam and assisted fuel systems.

The results ofthis data ensuresproper controlof the emissionsof any purged gas
or flared gas.
06

07

FlareManagement
Panametrics’ flare.IQ technology provides
real-time methane emission quantification
and reporting with flaring in line with
OGMP2.0 requirements. It significantly
reduces methane slip emissions forassisted flares, minimizes costs from flaring
operations, and improves transparency
for flare operations. flare.IQ is a full-stream
flare solution, covering assisted flares
associated with downstream petrochemical
and refinery flare operations, all the way tounassisted flares associated with upstream
operations, that represent by far thebiggest number of flares in the industry.

Benefits:• 24/7/365 live emissions quantification• Technology fit for purpose (ultrasonicmeters) to be leveraged on the existing
asset infrastructure• In-situ measurement not prone to be
influenced by weather conditions• A quick hit in terms of actual monitoring
and reporting• Comply with OGMP level 4 for methane
emissions reduction guidelines• Control processes better thanperiodical checks• Easy to deploy and install

flare.IQ

Monitor, reduce and control youremissions with flare.IQ

Uses a proprietary model to monitor flare efficiency basedon ultrasonic flowmeter outputs. For assisted flares, it helps
optimizing the efficiency while reducing steam or air and fuelgas usage. Exists with a module that allows the flare meter to bevalidated on-line and in-process through remote communication
with a plant’s distributed control system. For unassisted flares it
provides a real-time measurement for CE/DRE and emissions.
APPLICATIONS
Oil & Gas
- Upstream explorationand oil production- Offshore and onshore- Conventional andunconventional gas- Midstream gas gatheringand processing plants- Downstream refineries- Petrochemical plants- LNG plants• Field proven and independently tested

FEATURES• Calculates 24/7/365 combustion
efficiency, destruction and removal
efficiency, CO2, CO2 equivalent, CO
and VOC• For un-assisted flares, it isa powerful system for realtime accurate reporting ofCO2eq emissions – rather
then assuming 98% DRE• For assisted flares, on top of
the previous feature, it helps in
optimizing the efficiency of theflare by calculating the set pointsof steam or air and fuel gas whichleads to increase the efficiency andhence reduce the methane slip
• It can be used for digital
verification of the flare meterby tracking the diagnostics andchecking, on a pre-defined basis,the flare gas speed of sound• It is compliant with OGMP 2.0 level 40809

FlareFlow Meters
Panametrics has offered ultrasonic flare flow
meters to industry leaders as a better way
to measure gas flare for over four decades.
Our flare gas mass flow meters provide
critical flow data and diagnostics reliably
over a wide flow range (turn-down ratio) withexcellent low-end resolution to help improvethe efficiency of oil and gas production
assets, pipelines, terminals, refineries andpetrochemical plants.

Benefits:• Our flare gas flow meters use a proprietaryalgorithm that determines the molecularweight and mass flow rate of the flare andsweep gas

DigitalFlow GF868

Flare gas mass ultrasonic flowmeter
The DigitalFlow GF868 ultrasonic flowmeter uses the patented
Correlation Transit-Time™ technique, and an accurate and patented
method of calculating molecular weight. Add to these features theinherent advantages of ultrasonic flow measurement reliability withno routine maintenance, high accuracy, wide range ability, and theDigitalFlow GF868 flowmeter is the clear choice for flare gas applications.
APPLICATIONS• Our meters conserve energy and reduce
product loss by identifying sources of leaksin the flare system

Power Plant• Our flare flow meters adhere to local
regulatory requirements both for reportingand proper control of the flare system- Flare gas- Stack gas
Oil & Gas
- Fuel gas- Boil off gas- Acid/sour gas- Recycle gas- Hydrogen gas
Steel Plant- Coke gas and blast furnace gas- Feed gas- Stack gas
Chemical- Flare gas

FEATURES• Measures all flare gases and cancope with a wide variation of gas
composition with CO2, H2, N2, CO, etc• The widest turndown ratio with 4000:1(from 0.03 to 120 m/s or 0.1 to 400 ft/s)• Exists in 1- or 2-channel configuration• Transducers with temperaturecapability from -180°C to 300°C

10• NHV output (option)• Many current outputs available(6 or more)• Digital communication(HART, Modbus, FF, etc)

• Largest installed base in the industry• Typical accuracy velocity ±0.75% to
2.5%, MW ±1.8% to 2%, NHV ±2% to 5%

11

DigitalFlow XGF868i

Flare gas mass ultrasonic flowmeter
The DigitalFlow XGF868i ultrasonic flowmeter uses the patented
Correlation Transit-Time™ technique, and an accurate and patented
method of calculating molecular weight. Add to these features theinherent advantages of ultrasonic flow measurement reliability withno routine maintenance, high accuracy, wide range ability, and theDigitalFlow GF868 flowmeter is the clear choice for flare gas
applications. It comes in a transmitter type electronics.
APPLICATIONSPower Plant- Stack gas
Oil & Gas
- Flare gas- Fuel gas- Boil off gas-A cid/sour gas- Recycle gas- Hydrogen gas
Steel Plant- Coke gas and blast furnace gas- Feed gas- Stack gas
Chemical- Flare gas

FEATURES• Measures all flare gases and can cope witha wide variation of gas composition with
CO2, H2, N2, CO, etc• The widest turndown ratio with 4000:1(from 0.03 to 120 m/s or 0.1 to 400 ft/s)• Exists in 1- or 2-channel configuration• Transducers with temperature capability
from -180°C to 300°C• NHV output (option)• 2x current outputs available• Digital communication (HART, Modbus, FF, etc.)• Largest installed base in the industry• Typical accuracy velocity ±0.75% to 2.5%,
MW ±1.8% to 2%, NHV ±2% to 5%
1213

Portable
Flow Meters
The portables battery powered line fromPanametrics, a Baker Hughes business, offers
both ultrasonic liquid and gas clamp-on flow
meters: the TransPort PT878GC for gas; andthe TransPort PT900 for liquid. These ultrasonicflow meters utilize ultrasound transit time
to measure the velocity of a given fluid, andonce it is multiplied by the cross-sectionof the pipe gives the volumetric flow – andare used in a wide variety of liquid and gas
applications, from (waste) water and various
liquids to natural gas and compressed air.
These meters are attached to the outsideof the pipe to measure the flow within thepipe. Quick installation and meter readings
with advanced diagnostics enable our users
to continue their process without having toshut down. The meter’s portability allows thedevice to move to numerous measurement
locations, making it easy to spot-check theflow, verify the flow meter, or temporarily
install the device and provide reports using
its embedded data logger, without the needof a power supply.

Benefits:• Installation within minutes without
process interruptions• Embedded diagnostics to ensure
accurate and reliable measurements• Ability to be installed on very small
pipes (1/2") up to very large sizes (300")
for liquid applications• Permanent meter verification

TransPort PT900

Portable liquid clamp-on flow meter

It easily attaches to the outside of the pipe to measure flow within
the pipe. Users have no need to cut into it which causes delays
and downtime. The PT900 can also be used to spot-check othermeters or deploy for temporary installations. It can provide energy
measurement with optional temperature probes to monitor theenergy efficient in campus energy and building management systems.
APPLICATIONSPower Plant

Hydrogen- Boiler feed water- Demineralizedwater- Water- Brine- Campus energy:
Water/glycol• Pump or compressor performance check- Penstock• Cover all applications across all industries

Oil & Gas
- Water injection- MeOH and otherchemicals injection- Ultrapure water
Geothermal- Brine Flow
Water &
Wastewater- Water- Potable water- Water and cooling water- Waste water- Sea water intake- Raw water- Pure chemicals- Chemicals- Refined chemicals
Steel Plant- Tar- WaterPharmaceutical- Ultrapure water- Cleaning in place
Food & Beverages- Ultrapure water- Animal fat- Liquid product flow:
milk, juice

14

Semiconductor

FEATURES• Fluid types: All liquids• Flow measurement
patented Correlation
Transit-Time™ mode

• Pipe sizes from 0.5" up to 300"
• Pipe wall thickness up to 3"• Pipe material: All metals
and most plastics including
GRP/FRP pipes• Accuracy ±1% of reading±2% of reading• Optional thickness gauge

15TransPort PT878GC

Portable gas clamp-on flow meter

The TransPort PT878GC flowmeter is a highly versatile, self-contained,
portable gas clamp on transit-time flow meter with options andaccessories to meet a variety of gas flow measurement needs. It easily
attaches to the outside of the pipe to measure flow within the pipe.
Users have no need to cut into it which causes delays and downtime.
Its compact size, lightweight, rechargeable battery pack, and universal
power supply charger make it the ideal go anywhere flowmeter.
APPLICATIONSPower Plant- Loading fuel tanks​-C ompressed air- Fuel to boiler/turbine​
Oil & Gas
- Dehydration- Landfill gas- Gas lift-G as injection-G as compression- Air- Nitrogen
Steel Plant- Air- Oxygen- Argon

Pharmaceutical-C ompressed air
Food & Beverages-C ompressed air
Semiconductor- Air- Nitrogen

FEATURES• No moving parts• No pressure drop

• Flow measurement patented
Correlation Transit-Time™ mode

• Standard volumetric flow calculation• Small, lightweight and easy to use• Large, backlit LCD display• Rechargeable battery pack

• Logs over 100,000 flow data points• Accuracy ±1% of reading ±2% of reading• Optional thickness gauge

16

17

Custody Transfer,
Allocation andLeak Detection
Our Sentinel flow meters deliver custodytransfer-level measurement, easy
integration, rugged construction, andreliable performance – even in theunstable conditions of typical pipelines.
When you spec a Sentinel flow meter, you
can rest easy knowing you’re getting thehigh-accuracy measurements and insights
you need to operate cost-effectively
and free from false alarms.

Benefits:• Deliver high-accuracy measurements –even in the unstable conditions typicalof pipelines• Low maintenance• No pressure drop = reduce flashing risk

Sentinel LNG

High accuracy ultrasonic custody transfer
liquid flow meter for cryogenic application

4-path high accuracy liquid meter with no moving parts, a robust
path configuration, and bundled waveguide technology, the signals
can enter the process while protecting the transducers fromcryogenic temperatures.
APPLICATIONS
Oil & Gas
- LNG- Leak detection- Allocation metering- Fiscal and custodytransfer measurement• No interruption to operations after
installation thanks to transducers
online removability• Long term stability and drift
free characteristics• Deliver high-accuracy measurements

FEATURES• Improved performance, reduced
maintenance and dynamic flow
measurement for cryogenic liquids• Fully welded construction with no moving parts• Ultrasonic transducers protected from cryogenic
temperatures using Bundle Waveguide™
Technology and field replaceable without
the need for recalibration• Full bore design, zero pressure drop

• High accuracy to overcome limitations of tanklevel gauging• Robust Path Configuration™ through extensive
use of CFD (Computational Fluid Dynamics)• Accuracy ±0.25% of reading
viscosity independent

18

19

Sentinel LCT8

Sentinel LCT4High accuracy ultrasonic custodytransfer liquid flow meter

High accuracy ultrasonic custodytransfer liquid flow meter

8-path liquid meter uniquely designed to meet the strict
requirements for custody transfer of hydrocarbons. Its robust
path configuration provides trouble-free operation that will
not drift, even when the liquid product changes in the line.

4-path liquid meter designed specifically for high accuracy
measurement of crude oil, other liquid refined hydrocarbon
products, and other non-hydrocarbon liquids, it delivers
extremely reliable and repeatable results and meets thestrict performance requirements of OIML R117-1.

APPLICATIONSAPPLICATIONSPower Plant- Feed oil​Power Plant- Diesel fuelOil & Gas

- Loading fuel tanks​- Leak detection- Fuel to boiler/turbine-A llocation metering- Fiscal and custodytransfer measurement

Oil & Gas
- Leak detection- Allocation metering- Fiscal and custodytransfer measurement- Water and cooling water

FEATURES• No drifting or required
periodic calibration• Field replaceable transducers
without the need for recalibration• Optimized path configuration• No pressure drop

• No restriction in the pipe

• No moving parts and no filtersor strainers• Optionally built-in flow computerfor API MPMS 11.1 corrections fortemperature, pressure and density• Accuracy (linearity) ±0.12% ofreading viscosity independent

20

FEATURES• No drifting or required periodic calibration• Field replaceable transducers without theneed for recalibration• Optimized path configuration• No pressure drop

• No restriction in the pipe

• No moving parts and no filters or strainers• Optionally built-in flow computer for API
MPMS 11.1 corrections for temperature,pressure and density• Accuracy ±0.15% of reading
viscosity independent

21

Sentinel LCTHigh accuracy ultrasonic custodytransfer liquid flow meter4-path liquid meter designed specifically for the custody transfer
measurement of crude oil and other liquid refined products.
APPLICATIONSPower Plant- Feed oil
Oil & Gas
- Leak detection-A llocation metering- Fiscal and custodytransfer measurement- Oil after separation- Water and cooling water- High end process application:
Paraxylene production

FEATURES• No drifting or required
periodic calibration• Field replaceable transducers
without the need for recalibration• Optimized path configuration• No pressure drop

• No restriction in the pipe

• Fully welded construction withno moving parts and no filtersor strainers• Optionally built-in flow computerfor API MPMS 11.1 corrections
for temperature, pressureand density• Accuracy ±0.15% of reading
viscosity independent

22

23

Process Flow Meters
Ultrasonic Inline
Panametrics is a leader in flare gas, steam
and liquid metering applications. Our portfolio
offers both inline and clamp-on flow meterscapable of delivering flow measurement
insights directly to your plant or for remotemonitoring. Our in-line flow meters are well
suited for permanent installations whereaccurate, dependable and reliable flow
measurements are required.
Our broad portfolio of in-line flow meters
and readily available expertise enables you
to optimize your processes and meet yourenvironmental and performance goals.

Benefits:• Accurate flow measurements of various
process fluids• Delivers digital insights directly to your plant

DigitalFlow GM868 and XGM868i

Gas flowmeter

The flow meter includes wide-range transducers to measure any gas.
This accurate ultrasonic meter ensures reliable gas flow that requires
little or no maintenance.
APPLICATIONS
Oil & Gas
- Fuel gas- Associated gas• Provides reliable measurements withembedded diagnostics- Dehydration• Can address very challenging anddemanding applications from premium
alloys, high pressure and/or hightemperature, etc- Wet gas- Landfill gas- Boil off gas- Acid/sour gas- Recycle gas- Hydrogen gas- Quench gas- Gas after separation- Allocation metering- Flare gas- AirFEATURES- Gas injection• Full-featured flowmeter package using
a transmitter type electronics- Gas lift

- Gas compression- Natural gas- Acid/sour gas- Hydrogen gas- Gas compression
Steel Plant- Coke gas and blast furnace gas- Feed gas- Stack gas- Air- Nitrogen
Hydrogen- Pure and blended hydrogen- Power gen for air, nitrogen• Transducer removable under linepressure (depending on meter set-up)• No moving parts• No pressure drop

• Wide rangeability with up to 1500
to 1 turndown ratio (application andtransducer dependent)• Non-obstructive flow measurement• Tolerance to dirty streams• Low maintenance• Suitable for high temperatures• Two-path measurement available
for better accuracy• Binary gas mixture content
determination (optional)• Accuracy ±1% of reading ±2% of reading
24

25

DigitalFlow GS868 and XGS868i

PanaFlow Z1G and PanaFlow Z2GSteam ultrasonic flowmeter

Gas ultrasonic flow meter systems

Both the GS868 and XGS868i are designed to measure the mass flowrate of saturated or superheated steam. It offers a unique combination of
no pressure drop, wide rangeability, ease of installation, low maintenanceand high accuracy in a full-feature flowmeter package. The XGS868i
offers a rugged transducer, weatherproofing, and is approved for use inhazardous environments. It offers the same performances as the GS868
in a more cost-effective set up with only fewer I/Os possibilities.

High-performance, affordable solutions for a variety of gas flow
applications. The PanaFlow Z1G provides a one-path configuration
gas flow meter, while the PanaFlow Z2G is a two-path gas flow meter.
Both offer a broad range that is accurate and reliable, and available
in robust, sleek designs.
APPLICATIONS
Oil & Gas

APPLICATIONS- Fuel gas

Oil & Gas

- Flare gas-S aturated steam
and superheated steam- Gas after separation- Associated gas- Process steam- Dehydration

Steel Plant- Coal seam gas- Landfill gas-S aturated steam- Wet gas

Geothermal- Air- Gas compression- Nitrogen- Steam flow

- Recycle gas- Power for saturated- Quench gas- Superheated steam
measurement- Hydrogen blending
Steel Plant- Feed gas- Air- Oxygen

FEATURES• Full-featured flowmeter package
using a transmitter type electronics• Built-in steam tables for mass
flow calculation• Transducer removable underline pressure (depending on
meter set-up)• No moving parts• No pressure drop

- Natural gas​​• Non-obstructive flow measurement• Tolerance to dirty streams• Low maintenance• Suitable for high temperatures• Two-path measurement available forbetter accuracy• Accuracy ±1% of reading ±2%of reading

Hydrogen- Hydrogen blending
Biogas/Biofuels- Biogas with methane
content determination

FEATURES• No drifting, no periodic
calibration required• No pressure drop

• No restriction in the pipe
• No filters or strainers• No moving parts• Explosion-prooftransducer design• Bi-directional measurement• Binary gas mixture content
determination (optional)• Accuracy ±1% of reading±1.5% of reading• Wide rangeability with up to 1500 to
1 turndown ratio (application andtransducer dependent)
26

27

PanaFlow Z3

Three-path liquid ultrasonic flow meter
Designed specifically for dependable, accurate, and repeatableflow measurement of process liquids with a very tight accuracy
down to +/- 0.25% of reading under reference conditions.
APPLICATIONS

PanaFlow LZ

Ultrasonic flow meter system
for liquid applications
Offered as a one- or two-path wetted, ultrasonic flow
meter that brings all of the advantages of ultrasonictechnology at a very affordable value.
APPLICATIONSPower Plant- Water/glycol

Power Plant- Boiler feed water

Oil & Gas

- Water/glycol (facility mgt/
campus energy)-C ondensate (HC)- Produced water-W ater injection

Oil & Gas-A
 llocation metering- Leak detection- Condensate (HC)-B lending lines- Water injection- Oil- Produced water-W ater after separation- Leak detection- Water- Allocation metering- LPG and other light HC- Fuel oil- Oil- Water and cooling water

FEATURES

Steel Plant• No drifting, no periodic
calibration required- Water- TarHydrogen-D emineralized water
Water & Wastewater• No pressure drop

• No restriction in the pipe
• No filters or strainers• Bi-directional measurement• No moving parts- Potable water• Field replaceable transducers- Waste water• Explosion-proof transducer design- Raw water- Chemicals• All cast design- LPG and other light HC

FEATURES- Fuel oil• One or two-channel- Blending Lines- Oil after separation- Water after separation- Water and cooling water- Quench oil
Steel Plant- Tar- Water
Water & Wastewater• All modern communication protocols(HART, Modbus, FF FISCO)- Potable water• SIL rated (optional)- Chemicals• Accuracy ±0.25% of reading- Raw water- Waste water• No drifting, no periodic calibration required• No pressure drop

• No restriction in the pipe
• No filters or strainers• Bi-directional measurement• No moving parts• Field replaceable transducers• Explosion-proof transducer design• All modern communication protocols(HART, Modbus, FF FISCO)• Suitability for a wide range ofpipe sizes and materials• Accuracy ±0.5% of reading• SIL certification (option)• Flow measurement in extremely highand low temperature conditions
from 73K to 873K applications
28

29

PanaFlow HT SIL

Reliable extreme temperatureliquid ultrasonic flow meter
Ideal for harsh applications, such as delayed coker units, fluidic
catalytic cracking, vacuum distillation, crackers, hydrotreaters,
visbreakers, crude oil, and LNG. It is rated for hazardous areas
and extreme processes with temperatures from -200°C to 600°C.

PanaFlow XMT1000 SIL

Ultrasonic flow transmitter for liquids
Certified for hazardous installations, such as petrochemical
and chemical processing, and supports up to three paths
for more accurate measurement.
APPLICATIONSPower Plant

APPLICATIONS- Boiler feed water- WaterPower Plant- Brine- Boiler feed water​

Oil & Gas

Oil & Gas

- Oil after separation- Heavy residue- Water after separation- L iquid sulfur- Condensate (HC)-S
 lurry oil- Produced water- Hydrocarbon liquid- Water injection- Chemicals- Sea water intake- Safety critical liquid flow

- Oil- LNG- LPG and other light HC

FEATURES- Liquid sulfur• One, two or three-channel operation- Heavy residue

FEATURES• One or two-channel, in single double,
triple or quadruple set up• No drifting, no periodic
calibration required• No pressure drop

• No restriction in the pipe
• No filters or strainers• Bi-directional measurement• No moving parts• Field replaceable transducers- Fuel oil• Explosion-proof transducer design• All modern communication protocols(HART, Modbus, FF FISCO)• Suitability for a wide range of pipe sizes
and materials.
• Accuracy ±0.5% of reading• Flow measurement in extremely highand low temperature conditions from
73K to 873K applications• SIL certification- Water- Leak detection• No drifting, no periodic calibration required• No pressure drop

- Allocation metering• No restriction in the pipe

- Quench oil• Bi-directional measurement- Water and cooling water• No filters or strainers- Blending lines• No moving parts• Field replaceable transducers• All cast design• Explosion-proof transducer design• All modern communication protocols(HART, Modbus, FF FISCO)• Suitability for a wide range of pipe sizes
and materials.
• Accuracy ±0.3% of reading• SIL certification30

31

Ultrasonic Clamp-On
Flow Meters
Panametrics, offers a comprehensive
clamp-on portfolio consisting of AquaTrans,
DigitalFlow, and PanaFlow models. They all
exhibit common benefits such as: no moving
part, nothing to wear and tear over time, nopressure drop being contactless with theprocess fluid.
Ultrasonic clamp-on flow meters
measure fluid flow in many applications.
By using sound waves to determine thetransit time of a gas or liquid, ultrasonic flow
meters can accurately monitor any changes
in flow. Ultrasonic flow meters offer additional
benefits such as no moving parts, thus
reducing failure occurrences and extending
the life of the flow meter, plus the ability tosimply clamp the flow meter to the outsideof a pipe, so there is no need to shut down
production, install an inline meter, calibrate it,
and restart. With our ultrasonic clamp-on flow
meters you can choose where to measure,without having to disrupt production at all.

Benefits:• Easy to install without process shutdown• Accurate and versatile for a variety ofindustries, fluids, and pipe thicknesses• The meters maintain a reading of+- 1 to 2% and are contamination-free• Reliable measurement thanks to
its diagnostics

AquaTrans AT600

Liquid flow ultrasonic transmitter

The AquaTrans AT600 liquid clamp on flow ultrasonic transmitter
combines state-of-the-art flow measurement capability with acost-effective transmitter package that can be installed right at theprocess measurement point. It’s designed specifically for Water &Wastewater applications in full pipes and can be used in safe areas.
APPLICATIONSPower Plant

Food & Beverages- Boiler feed water- Ultrapure water- Brine- Liquid product flow:
milk, juice- Water- Combined cycle- Animal fat- Boiler water
blow down

Hydrogen- Campus energy:
Water/glycol

Semiconductor- Water from dam​

Oil & Gas
- Water injection- Demineralized water- Ultrapure water
Geothermal- MeoH and otherchemicals injection- Brine- Sea water intake

Water & Wastewater- Water- Water andcooling water
Steel Plant- Tar- Water- Blast furnace
water Cooling​
Pharmaceutical- Ultrapure water- Cleaning in place- Cooling water- Potable water- Raw water- Waste water- Chemicals

FEATURES• Economical non-intrusive
flow measurement• Extremely simple setup
and installation• Suitable for a wide range ofpipe sizes and materials• Suitable for lined pipes• Velocity, volumetric, andtotalized flow outputs• HART and Modbus
communication optional• Permanent solid
couplant available• Accuracy ±1% of reading±2% of reading

32

33

AquaTrans AT868

DigitalFlow DF868

Liquid flow ultrasonic transmitter

Full-featured, fixed-installation liquid flowmeter

Combines state-of-the-art liquid clamp on flow measurement
capability with a 2-path capability that can be installed right at theprocess measurement point. It’s designed specifically for Water &Wastewater applications in full pipes and can be used in safe areas.

Designed to meet all your flow metering and energy measurement
needs. Its patented Correlation Transit-Time™ digital signal processing
provides drift-free measurements in ultraclean and most ‘dirty’ liquids
with its optional energy feature, it can deliver an all-in-one solution forenergy metering.

APPLICATIONSPower Plant

Food & Beverages- Boiler feed water- Ultrapure water
Animal fat- Water- Brine-C ampus energy:
Water/glycol- Combined cycle- Boiler water
blow down- Water from dam​
Oil & Gas-W
 ater injection

Semiconductor- Water- Blast furnace
water cooling​-O xygen furnace​-C ontinuous casting​Pharmaceutical- Ultrapure water- CIP- Boiler feed water- Water- Ultrapure water

Oil & Gas

Geothermal- Water after
separation

Water & Wastewater- TarSteel Plant- Campus energy:
Water/glycol
Hot water
Chilled water- Demineralized water- Sea water intake

Steel Plant

Power Plant- Brine

Hydrogen- Brine- Water andcooling water34- Liquid product flow:
milk, juice- MeoH and otherchemicals injection- Water

APPLICATIONS- Cooling water-P otable water- Raw water-W aste water- Chemicals- Oil after separation

FEATURES• Non-intrusive flow
measurement• Simple setup and installation• Suitable for wide range ofpipe sizes and materials• Suitable for lined pipes• Two-channel/two-pathversion available• Velocity, volumetric andtotalized flow outputs• Internal keypad for field
programming• Accuracy ±1% of reading±2% of reading- Condensate (HC)- Produced water- Water injection- Sea water intake- Water- Leak detection- Oil- LNG- MeoH and otherchemicals injection- Water and coolingwater- LPG and otherlight HC- Blending Lines- TarPharmaceutical- Ultrapure water
Food & Beverages- Ultrapure water- Animal fat

Semiconductor- Ultrapure water
Water & Wastewater- Potable water- Raw water- Waste water- Chemicals

FEATURES• Non-intrusive flow
measurement• Hazardous (classified)location certifications• Simple set-up
and installation• Suitable for wide range ofpipe sizes and materials• Two-channel/two-pathversion available• Accuracy ±1% of reading±2% of reading• Energy option withtemperature supply
and return andcalculation algorithm

35

PanaFlow LC

Permanent clamp-on ultrasonic flow meter
PanaFlow LC uses the XMT1000 electronics with a fast response time (3 Hz), in
1, 2 or 3 path configuration, a wide range of transducer models and frequency
and a user friendly human machine interface (Panaview Plus). It comes with
its clamping fixture and selectable cable configuration and length. Suitable
for all Newtonian liquids. SIL certification available as an option.
APPLICATIONS
Oil & Gas

- Oil after separation- Water after
separation- Condensate (HC)- Produced water-W ater injection- MeoH and otherchemicals injection- Sea water intake- Water- Leak detection- LNG- Sea water intake- Water andcooling water- Oil- LPG and other light HC-B lending Lines- Heavy residue- L iquid sulfur

DigitalFlow GC868

The DigitalFlow GC868 clamp-on gas flowmeteris a complete ultrasonic flow metering system
The DigitalFlow GC868 flowmeter can be used to measure the flowof any gas. It is especially useful for metering erosive, corrosive, toxic,
high-purity or sterile gases, or in any application where penetrating
the pipe wall is undesirable.
APPLICATIONS

Steel Plant

Power Plant- Water- Compressed air- TarPharmaceutical- Ultrapure water
Food & Beverages- Ultrapure water- Animal fat

Oil & Gas
- Dehydration- Landfill gas- Gas lift

- Gas injection- Gas compression- Air- Semiconductor- Ultrapure water

Steel Plant- Air- Argon- Water &Wastewater- Oxygen​-P otable water- Raw water

Pharmaceutical- Compressed air-W aste water- Chemicals

Food & Beverages- Compressed air- Fuel oil- Quench oil

Hydrogen- Demineralized water

FEATURES• Wide selection of transducers
suitable for many applications• Hazardous area certification• Improved accuracy and repeatability
through enhanced• Signal processing• HART and Foundation Fieldbus FISCO
compliant digital protocols
36

Semiconductor• Wider flow range for handlingmore diverse flow measurements• Velocity, volumetric, mass, and totalizer
flow rate measurements- Nitrogen- Oxygen- AirFEATURES• Clamp-on unobstructed installation• Two-channel/two-pathversion available• No moving parts• No pressure drop

• Sound speed output• Simple installation• Standard volumetric flow calculation• Suitable for a wide range oftemperatures and pressures• Accuracy ±1% of reading±2% of reading• Based on legacy Panametrics technology
for reliable flow measurements• Accuracy ±1% of reading ±2% of reading37Vortex
Flow Meters
Our PanaFlow vortex meter offers a multivariable design that contains velocity with
RTD temperature sensors and a solidstate pressure transducer for measuringmass flow rate of steam, gases and liquids.
The accuracy of these meters enables
velocity, temperature and pressure to
be measured at the same location.

Benefits:• Reliable and accurate• Saturated and superheated steam
mass flow calculation with pressure
and temperature inputs

PanaFlow MV84

In-line multi-variable vortex flow meter
In-line multi-variable flow meters for volume, mass, temperatureand pressure. PanaFlow MV84’s multi-variable design consists of a vortex
shedding velocity sensor, an RTD temperature sensor and a solid-statepressure transducer that measures the mass flow rate of steam, gases
and liquids. Pressure and temperature sensors are optional.
APPLICATIONSPower Plant• Easy installation- Saturated andsuperheated steam• Low maintenance costs- Nitrogen• Wide flow range and a low pressure drop
• Available as an insertion model- Air- Oxygen- Water

Oil & Gas
- Saturated andsuperheated steam- Air- Nitrogen- Oxygen- WaterSteel Plant- Saturated andsuperheated steam- Air- Nitrogen- Oxygen- Water

FEATURES• Built-in flow computer for measuring volumetricflow, temperature, pressure, density, energy,
and mass flow using a single meter• No need to recalibrate• Advanced design and digital signal processing
for vibration isolation• Cost effective, accurate and reliable• Energy management through accurate
measurement of both temperature and mass
flow simultaneously• Remote monitoring and integration
to DCS using HART, Modbus, and BACnet
communication protocols• Accuracy ±1% of reading ±2% of reading

38

39

PanaFlow MV82

Insertion multi-variable mass vortex flowmeter
PanaFlow MV82’s multi-variable design consists of a vortex shedding
velocity sensor, an RTD temperature sensor and a solid state pressuretransducer that measures the mass flow rate of steam, gases andliquids through an insertion point. This insertion vortex meter installs ineither existing lines with a packing gland or compression fitting design
or for new installations.
APPLICATIONSPower Plant-H T/MT/LT steam
Oil & Gas
-H
 T/MT/LT steam
Steel Plant- Blast furnace​- Oxygen furnace​-C ontinuous casting​

FEATURES• Built-in flow computer for measuring volumetricflow, temperature, pressure, density, energy,
and mass flow using a single meter• Insertion flow meter with online
retraction capability (option)• No need to recalibrate• Advanced design and digital signal processing
for vibration isolation• Cost effective, accurate and reliable• Energy management through accurate
measurement of both temperature andmass flow simultaneously• Remote monitoring and integration
to DCS using HART, Modbus, and BACnet
communication protocols
40

41

Learn about Panametrics’
industry solutions

42

Food &
BeverageDistrict
EnergyRefinery

Steel

Geothermal

Power
Generation

Healthcare
and Pharma

Water &
Wastewater
SolutionsHydrogenRenewable
Natural Gas

LNG

Semiconductor
Industry43

Panametrics, a Baker Hughes business,
provides solutions in the toughest
applications and environments for moisture,oxygen, liquid and gas flow measurement.
Experts in flare management, Panametrics
technology also reduces flare emissions
and optimizes performance.
With a reach that extends across the globe,
Panametrics’ critical measurement solutions

and flare emissions management are
enabling customers to drive efficiency andachieve carbon reduction targets acrosscritical industries including: Oil & Gas; Energy;
Healthcare; Water & Wastewater; Chemical
Processing; Food & Beverage and many others.
Join the conversation and follow us on LinkedIn
linkedin.com/company/panametricscompany

Contact us

For more information please contact yourlocal Panametrics representative, or visit:
panametrics.com© Copyright 2021. Baker Hughes Company. This material contains
one or more registered trademarks of Baker Hughes Company andits subsidiaries in one or more countries. All third-party productand company names are trademarks of their respective holders.