31000 and 32000 Proximity Probe Housing
Assemblies
Datasheet

Bently Nevada Machinery Condition Monitoring141610 Rev. M

Description
The 31000/32000 Proximity Probe Housing Assemblies arerecommended when mounting proximity probes through themachine case and are typically used for radially mounted
transducers, whether vibration or Keyphasor measurements.
When using these housings to measure radialvibration, ensure that the machine casing is affixed
to the bearing in order to get an accurate relative
vibration signal.
When measuring shaft axial position with dual
proximity probes, use housing 21022 instead.
Consult datasheet (document 141601).
Use of a Proximity probe housing allows external access tothe proximity probe and its extension cable, permitting gap
adjustment or probe replacement without disassembly of themachine. The 31000/32000 Proximity Probe Housing Assemblyis made of polyphenylene sulfide (PPS), an advanced, highstrength, thermoplastic with excellent corrosion resistance.
Other elements of the housing assembly are made ofcorrosion-resistant stainless steel. The housing can beordered with installed 3300 XL Proximity Probes and a varietyof conduit fittings.
The 31000/32000 Proximity Probe Housing Assembly is fullycompliant with the American Petroleum Institute's (API) 670
Standard for externally mounted proximity probe housings.
When installed in conjunction with an approved transducer
system and appropriate I.S. barriers, the 31000/32000

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet

141610 Rev. M

Proximity Probe Housing Assembly can be used
in intrinsically safe hazardous area
applications.
The 31000/32000 Housing is intended
to provide mechanical andenvironmental protection only, and is
not an explosion-proof housing.
When an explosion proof proximity
probe housing assembly is required,use housing CA21000 or CA24701.
Consult the datasheet (document141600).

Related Documents
For probe information, refer to the following
manuals:
n

n

n

n

3300 XL 8mm & 3300 5mm Proximity
Transducer System User Guide(document 141078)
3300 XL NSv Proximity Transducer System
User Guide (document 147357)
3300 XL 11mm Proximity Transducer
System Installation User Guide(document 146255)
Radiation Resistant Probe and Proximitor
System (document TW8029407)

2/18

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet

141610 Rev. M

certified to seal against hose-directed wateraccording to NEMA 4X and IP66 standards but
is not designed to resist internal or externalpressure. Probes are not pressure tested prior
to shipment. Contact our custom design
department if you require a test of the pressure
seal for your application.

Specifications
Environmental
3300 XL 8 mm Probe
Operating andstorage-51°C to +177°C (-60°F to +351°F)

It is the responsibility of the customer
or user to ensure that all liquids andgases are contained and safely
controlled should leakage occur froma Proximity Probe Housing Assembly.
Solutions with high or low pH values
may erode the tip assembly of theprobe, causing media leakage into
surrounding areas. Bently Nevada,
LLC, does not be held responsible forany damages resulting from leaking
Proximity Probe Housing Assemblies.
In addition, Proximity Probe Housing
Assemblies and 3300 XL 8 mm
proximity probes does not be
replaced under the service plan due
to probe leakage.

3300 XL 11 mm Probe
Operating andstorage-51°C to + 177°C (-60°F to +351°F)

3300 NSV Probe
Operating-34°C to +177°C (-30°F to +351°F)

Storage-51°C to +177°C (-60°F to+351°F)

Hot Water and Steam Exposure Effects(Specification not guaranteed). Brief
periods (up to one week) of contact
with hot water (95°C [203°F]) and/or
condensing steam should not
significantly affect the strength of theplastic housing. Contact with thesebeyond this length of time may
eventually cause the strength of theplastic housing to permanently
decrease during the first 6 to 8 weeks
of exposure, and then level atapproximately half of its initial value.
Tests of actual housing performanceafter contact with hot water andcondensing steam have not been
conducted.

MechanicalProtection
Ratings

Type 4X rating certified by
Canadian Standards
Association (CSA). IP66
rating verified by SC1155821 (e) 106. CENELEC standard
EN50014 rating forelectrostatic dissipation ofa plastic material locatedin a hazardous area.

Glass-reinforced
Polyphenylene Sulfide
Housing Material (PPS) thermoplastic
containing conductive
fibers

Probe Pressure
The 31000/32000 Proximity Probe Housing
Assembly is designed to seal differential
pressure between the probe tip and the
housing main body when used with a 3300 XL 8
mm probe. The sealing material internal to theprobe case consists of a Viton O-ring; the Oring between the sleeve and the housing is a
Neoprene O-ring. The plastic housing is

Sleeve Material
and Retaining
Chain

AISI 304 stainless steel

Outer Sleeve and
AISI 303 stainless steel
Exterior Screws

3/18

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet

O-Ring Material

141610 Rev. M

Neoprene

Recommended
Torque(retaining 29.4 N·m (260 in·lb)
nut)Housing
Strength(typical)Outer sleeve was mounted
on a test stand with its axis
parallel to horizontal andthe housing mounted on
the outer sleeve throughan end hole. The housing
supported 912 N (205 lb)
placed approximately 38
mm (1.5 inches) from theunsupported end with thecover fastened in placeand grounding liner
installed.

Housing Impact
Strength

Certified by BASEEFA towithstand two separate 4
Joule (3.0 ft·lb) impacts at-39°C (-38°F) and at 115°C(239°F). Samples of the
housing and cover wereverified by CSA towithstand a 7 Joule (5.2
ft·lb) impact at ambientroom temperature.

Weight

1.2 kg (2.6 lb) typical

4/18

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet

141610 Rev. M

Hazardous Area Approvals
For the detailed listing of country andproduct-specific approvals, refer tothe Approvals Quick Reference Guide(108M1756).
For additional technicaldocumentation, please log in to
bntechsupport.com and access the
Bently Nevada Media Library.

21000 and 24701
CA21000 and CA24701 housings are certified by
CSA for explosion-proof applications in thefollowing areas:
North
AmericaClass I Divisions 1 and 2 Groups C and D
Class II Divisions 1 and 2 Groups E, F, and G
Circuit is non-incendive when connected
per drawing 140979
Class III
Enclosure Type 4
T5 @ Ta = -50°C to +40°C
T4 @ Ta = -50°C to +80°C

CA housings can only be ordered for North
American End Users. For all other end user
countries, housings 21000 and 24701 must beordered.

5/18

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet

141610 Rev. M

English Proximity Probe Housing
Assemblies

Ordering Information

31000 - AA-BB-CC-DDD-EE-FF

For the detailed listing of country andproduct-specific approvals, refer tothe Approvals Quick Reference Guide(108M1756).

A: Probe with Connector

For additional technicaldocumentation, please log in to
bntechsupport.com and access the
Bently Nevada Media Library.

Table 1: Maximum "C" Option plus "D" Option
for different "B" Options (probe cable length)
Probe
Cable
LengthMaximum C plus D

0.5 meter

394 mm (15.5 in)

1.0 meter

760 mm (30.0 in)

Table 2: Maximum "C" Option plus "D" Option
for different "B" Options (probe cable length)
where P/N and S/N Label on Probe Cable isvisible outside of probe sleeve
Probe
Cable
LengthMaximum C plus D with
Visible P/N and S/N Label

0.5 meter

64 mm (2.5 in)

1.0 meter

483 mm (19.0 in)

00

Probe not required

16

3300 XL 8 mm probe

26

3300 XL NSv probe

27

3300 XL NSv probe, multiple
approvals

28

3300 XL 8 mm probe, multiple
approvals

29

3300 XL 8 mm probe, with connector
protector

30

3300 XL 8 mm probe, with connector
protector, multiple approvals

31

3300 XL NSv probe with connector
protector

32

3300 XL NSv probe with connector
protector, multiple approvals

33

3300 XL 11 mm probe

34

3300 XL 11 mm probe, multiple
approvals

35

3300 XL 11 mm probe with connector
protector

363300 XL 11 mm probe with connector
protector, multiple approvals

B: Probe Cable Length

Conduit fittings are necessary whenhardline conduit or metal tubing isbrought into the housing. Flexible
conduit should be ordered withintegral 3/4-14 NPT fittings and do not
require additional conduit fittings withthe housing. If using flexible conduit,
order the "E" = 00 option.

00

Probe cable not required (Option A
must also be 00)05

0.5 m (20 in)

10

1.0 m (39 in)

C: Standoff Adapter Length (Option C
Dimension)

6/18

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet

141610 Rev. M

Must be ordered if Standoff
Adapter Length option is not
00.
15

Minimum length 1.5 in (38 mm)

75

Maximum length 7.5 in (191 mm)
Recommendation
Order in increments of 0.5 in (13 mm)
Examples03

Two plugs, one 3/4-14 NPT fitting
One 3/4-14 to 1/2-14 NPT SST reducer
One cable seal grip with grommets forthe following cable sizes:• 1/8 to 3/16 inches• 1/4 to 5/16 inches• 5/16 to 3/8 inches

06

One 3/4-14 NPT fitting
One 3/4-14 NPT to 1/2-14 NPR SST
reducer
Two plugs

F: Mounting Thread Option

1.5 in (38 mm) = 15
No standoff adapter = 00

00

No outer sleeve, retainer, or retaining
nut

D: Probe Penetration Option (Option D
Dimension)

02

3/4-14 NPT (Required if ordering
Standoff Adapter Option).

05

7/8-14 UNF 2A

10Minimum length 1.0 in (25.4 mm)
Recommendation
Order in increments of 0.1 in (3 mm)
ExamplesNo probe sleeve = 000
3.7 in (94 mm) = 037
2.4 in (569 mm) = 224"C" plus "D" dimensions greater than 12
in (305 mm) require additional sleeve
support near the probe to prevent
resonance from occurring. Sleeveadjustment range of Probe
Penetration Option "D" is ±0.5 in (13mm).

For probe penetration lengths between 1.0 and
2.0 inches, it may be necessary to counter bore
the machine case to reduce probe side view orrear view effects.
E: Fittings
00

Without fittings

01

One 3/4-14 NPT fitting, two plugs.

02

Two 3/4-14 NPT fittings, one plug.

7/18

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet

141610 Rev. M

Metric Proximity Probe Housing
Assemblies

Must be ordered if Standoff
Adapter Length option is not
00.

32000 - AA-BB-CC-DDD-EE-FF
A: Probe with Connector

04

Minimum length 40 mm

00

Probe not required

20

Maximum length 200 mm

163300 XL 8 mm probe

Recommendation

26

3300 XL NSv probe

Order in increments of 10 mm

273300 XL NSv probe, multiple
approvals

D: Probe Penetration Option (Option D
Dimension)

28

3300 XL 8 mm probe, multiple
approvals

760

Maximum length 760 mm

025

Minimum length 25 mm

293300 XL 8 mm probe, with connector
protector

30

3300 XL 8 mm probe, with connector
protector, multiple approvals

Order in increments of 1.0 mm

313300 XL NSv probe with connector
protector

No probe sleeve = 000

323300 XL NSv probe with connector
protector, multiple approvals

760 mm = 760

333300 XL 11 mm probe

34

3300 XL 11 mm probe, multiple
approvals

35

3300 XL 11 mm probe with connector
protector

363300 XL 11 mm probe with connector
protector, multiple approvals

Recommendation

Examples

50 mm = 050"C" plus "D" dimensions greater than 12
in (305 mm) require additional sleeve
support near the probe to prevent
resonance from occurring. Sleeveadjustment range of Probe
Penetration Option "D" is ±0.5 in (13mm).
For probe penetration lengths between 25 and50 millimeters, it may be necessary to counterbore the machine case to reduce probe sideview or rear view effects.

B: Probe Cable Length00

Probe cable not required (Option A
must also be 00)05

0.5 m (20 in)

E: Fittings

10

1.0 m (39 in)

00

Without fittings; two plugs and two
washers.

01

One M25 fitting, two plugs.

02

Two M25 fittings, one plug.

C: Standoff Adapter Length (Option C
Dimension)

8/18

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet03

Two plugs, one M20 fitting
One cable seal grip with grommets forarmored probe cable

05

One DIN PG11 fitting, two plugs

07

One PG21 x M20 fitting
Two plugs

08

Two PG21 x M20 fittings
One plug

141610 Rev. M

Terminal Housing
106769-AA
The 106769 housing consists of a 31000-style
PPS housing with two terminal mounting blocks(each terminal block has four terminals)
mounted in each housing. Sixteen ring lugs aresupplied loose inside the housing forconnecting transducer cables. Conduit fittingsare 3/4-14 NPT chrome-plated zinc fittings.

F: Mounting Thread Option
00

No outer sleeve, retainer, or retaining
nut

01

M24 X 302

3/4-14 NPT (Required if ordering
Standoff Adapter Option.)

A: Conduit Fitting Option

9/18

00

No fittings

01

One fitting

02

Two fittings

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet

141610 Rev. M

AAA Option
for 31000
Prox Probe
Housing

Accessories
Part
NumberDescription

37948-01

Probe Support / Oil Seal
Recommended for sleeves
longer than 305 mm (12 in)

12420031000 and 32000 User Guide

Housing AAA
option for3300 XL NSv
probe option(A: 26 or A:
27)

English Probe Sleeve for 31000
Proximity Probe Housings
108883-AAA

Standoff adapter option fromoriginal housing (31000 optionC) + Probe penetration option
from original housing (31000
option D) + 0 2 6.
ExampleOriginal part number 3100027-10-20-035-03-02
AAA option for replacement
sleeve (020 + 035 + 026) = 0 8
1.
Standoff Adapter Option fromoriginal housing (31000 optionC) + Probe penetration option
from original housing (31000
option D) + 0 1 7.

A: Measured Probe Sleeve Length
Recommendation

Housing AAA
option for3300 XL 11 mmprobe option Example(A: 33 or A:
Original part number 3100034)
33-10-30-113-01-02

Order in increments of 0.1 in (3mm)
The individual probe sleeve length does notinclude the distance from the end of the sleeveto the probe tip or the gap from the probe tip to
the target material. If only the part number ofthe original housing is known, and the sleevecannot be measured, use the following table todetermine the sleeve length:
AAA Option
for 31000
Prox Probe
Housing

Description

AAA option for replacement
sleeve (030 + 113 + 017) = 1 6 0.

Minimum Probe Sleeve Length

Description

Standoff adapter option fromoriginal housing (31000 optionC) + Probe penetration option
from original housing (31000
option D) + 0 2 5.

Housing AAA
option for3300 XL 8 mm Example
probe option(A: 16 or A: 28) Original part number 3100016-10-15-035-03-02

3300 XL 8 mm probes

3.5 in (89 mm) = 0 353300 XL NSv probe

3.6 in (91 mm) = 0 363300 XL 11 mm probe

2.7 in (69 mm) = 0 2
7

Maximum Probe Sleeve Length

AAA option for replacement
sleeve (015 + 035 + 025) = 075

10/183300 XL 0.5 meter 8
mm probe

18.0 in (457 mm) =180

3300 XL NSv 0.5 meterprobe

18.1 in (460 mm) =181

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet

3300 XL 0.5 metre 11mm probe

141610 Rev. M

Metric Probe Sleeve for 32000
Proximity Probe Housings
108882-AAA

16.0 in (406 mm) =160

3300 XL 1.0 meter 8 mm 32.5 in (826mm) =
probe
325
3300 XL NSv 1.0 meterprobe

A: Measured Probe Sleeve Length

32.6 in (828mm) =326

Recommendation
Order in increments of 1 mm
(0.04 in).

3300 XL 1.0 meter 11 mm 31.7 in (805 mm) =
probe
317

The individual probe sleeve length
does not include the distance fromthe end of the sleeve to the probe tipor the gap from the probe tip to thetarget material. If only the partnumber of the original housing is
known, and the sleeve cannot bemeasured, use the following table todetermine the sleeve length:
AAA Option
for 32000
Prox Probe
Housing

Housing AAA
option for3300 XL 8
mm probe
option (A: 16
or A: 28)

Description

Standoff adapter option fromoriginal housing (32000 optionC) *10 + Probe penetration
option from original housing(32000 option D) + 063.
ExampleOriginal part number 3200016-10-08-205-03-02
AAA option for replacement
sleeve (080 + 205 + 063) = 348

11/18

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet

AAA Option
for 32000
Prox Probe
Housing

Housing AAA
option for3300 XL NSv
probe option(A: 26 or A:
27)

141610 Rev. M

3300 XL 1.0 meter 8 mm 823 mm (32.4 in) =
probe
823

Description

3300 XL NSv 1.0 meter
probe

Standoff adapter option fromoriginal housing (32000 optionC) *10 + probe penetration
option from original housing(32000 option D) + 066.

3300 XL 1.0 meter 11 mm 802 mm (31.6 in) =
probe
802

ExampleOriginal part number 3200027-10-10-105-03-02
AAA option for replacement
sleeve (100 + 105 + 066) = 271.

Housing AAA
option for3300 XL 11
mm probe
option (A: 33
or A: 34)

Standoff Adapter Option fromoriginal housing (32000 optionC) * 10 + Probe penetration
option from original housing(32000 option D) + 042.
ExampleOriginal part number 3200033-10-10-105-03-02
AAA option for replacement
sleeve (100 + 105 +042)=247

Minimum Probe Sleeve Length
3300 XL 8 mm probes

88 mm (3.5 in) = 088

3300 XL NSv probe

91 mm (3.6 in) = 091

3300 XL 11 mm probe

67 mm (2.6 in) = 067

Maximum Probe Sleeve Length
3300 XL 0.5 meter 8
mm probe

457 mm (18.0 in) =457

3300 XL NSv 0.5 meterprobe

460 mm (18.1 in) =460

3300 XL 0.5 metre 11mm probe

436 mm (17.2 in) =436

826 mm (32.5 in) =
826

12/18

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet

141610 Rev. M

English Standoff Adaptor
109319-AAA

The standoff adapter covers 38 mm (2.0 in) of
the probe sleeve plus an additional 13 mm (0.5in).

A: Individual Standoff Adapter Length

Therefore, the effective probe penetration
length drops to 13 mm (0.5 in).

Hex

1 3/8 in; threads = 3/414 NPT

015

Minimum length 1.5 in(38 mm)075

Maximum length 7.5 in(191 mm)

Recommendation

Order in increments of
0.5 in (13 mm)

If you use effective penetration lengths of lessthan 1.0 inches, signal effects are likely due toprobe side view or rear view of metalcomponents.

Metric Standoff Adaptor
109318-AA
A: Individual Standoff Adapter Length
Wrench Flats

35 mm; threads = 3/414 NPT

04

Minimum length 40 mm

20Maximum length 200
mm

Recommendation

Order in increments of
10 mm

Example120 mm = 12

For desired probe penetration lengthsof less than 25 mm (1.0 in), order aseparate Individual Standoff Adapter.
The effective probe penetration length will thenbe reduced by the length of the IndividualStandoff Adapter, plus an additional 13 mm (0.5in) due to the NPT thread engagement.
Example:To create a probe penetration length 13 mm(0.5in), order a 31000 housing with DDD (probe
penetration) option of 030 [76 mm (3 in)] and aseparate individual standoff adapter that is 51
mm (2.0 in) in length (part number 21003-020).

13/18

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet

141610 Rev. M

Spare 3300 XL 8 mm Reverse
Mount Probe, 3/8-24 UNF threads

Spare 3300 NSV Reverse Mount
Probe, 3/8-24 UNF threads

330105-02-12-CC-DD-EE

330906-02-12-CC-DD-EE

Spare 3300 XL 8 mm Reverse
Mount Probe, M10 X1 threads

Spare 3300 NSV Reverse Mount
Probe, M10 X1 threads

330106-05-30-CC-DD-EE

330907-05-30-CC-DD-EE

C: Total Length

C: Total Length05

0.5 meter (1.6 feet)05

0.5 meter (1.6 feet)10

1.0 meter (3.3 feet)10

1.0 meter (3.3 feet)15

1.5 meter (4.9 feet)50

5.0 meters (16.4 feet)20

2.0 meters (6.6 feet)90

9.0 meters (29.5 feet)50

5.0 meters (16.4 feet)

D: Connector

90

9.0 meters (29.5 feet)00

Connector not installed

02

Miniature ClickLoc coaxial
connector

D: Connector
00

Connector not installed

02

Miniature ClickLoc coaxial
connector

E: Agency Approval00

Not required

E: Agency Approval05

Multiple approvals

00

Not required

Sleeve and Blanking Plugs

05

Multiple approvals
104968-01

English Sleeve Plug,threaded, 303 stainlesssteel104968-02

Metric Sleeve Plug,threaded, 303 stainlesssteel.

Plugs fill opening when sleeve isremoved from machine case.
104288-01

English Blanking Plug

104288-02

Metric Blanking Plug

Plugs fill extra holes in plastic
housing where needed.

14/18

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet

141610 Rev. M

Heavy Duty Conduit and Cable
Fittings
03813103

Chrome-plated Zinc Conduit
Fitting, 3/4-14 NPT

03818100AISI 316 Stainless Steel Conduit
Fitting, 3/4-14 NPT

03818101AISI 316 Stainless Steel Conduit
Fitting, PG21 x M25

03818102AISI 316 Stainless Steel Conduit
Fitting, PG21 x M20

03818103AISI 316 Stainless Steel Conduit
Fitting, PG21 x PG11

03818104AISI 303 Stainless Steel Cable
Gland, PG11

03818105AISI 316 Stainless Steel Cable
Gland, M20

03818111

Nickel-plated Brass Conduit
Fitting, PG21 x M20

26650-01

AISI 303 Stainless Steel Reducer,
3/4-14 NPT to 1/2-14 NPT

15/18

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet

141610 Rev. M

Graphs and Figures

Figure 1: Dimensions for 31000 and 32000 Proximity Probe Housings
Dimensions are in millimeters (inches).

16/18

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet

141610 Rev. M

All 4 holes in housing base, 1 per side, will accept sleeve or conduit fittings and cable glands.
Fittings are supplied with housing depending on English, metric or DIN type.
Hole plugs are provided to seal unused holes.

Installation Procedures
1. Install outer sleeve into machine case.
2. Insert probe sleeve and adjust probe gap.
3. Disconnect probe cable and fit housing over outer sleeve.
4. Slide retainer under retaining nut. Tighten nut.
5. Re-connect probe cable and Connector Protector.
6. Place housing cover on housing and tighten captive screws.
7. If hole plugs are used, tighten hole plug nuts to 0.5 N-m (5 in-lbs).
Figure 2: Vertical and Horizontal Profile Views of the 31000 and 32000 Proximity Probe Housings

17/18

31000 and 32000 Proximity Probe Housing Assemblies
Datasheet

141610 Rev. M

Copyright 2024 Baker Hughes Company. All rights reserved.

Bently Nevada, Keyphasor, M2, Proximitor and Orbit Logo are registered trademarks of Bently Nevada, a Baker Hughes
business, in the United States and other countries. The Baker Hughes logo is a trademark of Baker Hughes Company.
All other product and company names are trademarks of their respective holders. Use of the trademarks does not
imply any affiliation with or endorsement by the respective holders.
Baker Hughes provides this information on an “as is” basis for general information purposes. Baker Hughes does notmake any representation as to the accuracy or completeness of the information and makes no warranties of any
kind, specific, implied or oral, to the fullest extent permissible by law, including those of merchantability and fitness
for a particular purpose or use. Baker Hughes hereby disclaims any and all liability for any direct, indirect,
consequential or special damages, claims for lost profits, or third party claims arising from the use of theinformation, whether a claim is asserted in contract, tort, or otherwise. Baker Hughes reserves the right to makechanges in specifications and features shown herein, or discontinue the product described at any time without
notice or obligation. Contact your Baker Hughes representative for the most current information.
The information contained in this document is the property of Baker Hughes and its affiliates; and is subject tochange without prior notice. It is being supplied as a service to our customers and may not be altered or its content
repackaged without the express written consent of Baker Hughes. This product or associated products may becovered by one or more patents. See Bently.com/legal.

1631 Bently Parkway South, Minden, Nevada USA 89423
Phone: 1.775.782.3611 (US) or Bently.com/support
Bently.com

18/18