Everest Mentor Visual iQVideoProbe™
Intelligent inspection solutions.

More precise and efficient inspections are now possible with Waygate
Technologies’ Mentor Visual iQ VideoProbe. Driven by advanced analytics
to inspect in entirely new ways, the more intelligent Mentor Visual iQleverages computer vision and machine learning to increase data quality
whilst reducing inspection times.

Unbeatable Clarity with TrueSight™

Real3D™ Measurement for Confident Decisions•Crystal clear live video and still images in highdefinition (HD)•Most advanced 3D capabilities for more precise
measurement from increased distances•

Unlocks the power of digital zoom on device with new
1.2 mega pixel image sensor•

Up to nine measurement types provide accurate, efficientand reliable measurement results•

Blends high resolution image and precision optics witha powerful suite of proprietary image transformation
functions•

Real time cursor adjustment in the fully surfaced Point Cloud•

Generate and measure larger areas of the inspected asset
with 3D Stitching

Intuitive Design

Enhanced Efficiency•

Portable, lightweight, rugged, and versatile••

QuickChange™ probes with optical tip adaptors•

Integrated WiFi and Bluetooth•On-screen keyboard•

Powered by high-capacity, rechargeable lithium ionbatteries

2Improve consistency and reliability of inspection withassisted defect recognition Analytics• Automate repetive tasks with Blade Counter•

Menu Directed Inspection (MDI) software that guides
inspectors through the inspection process, intelligently
names files and tags files, and creates inspection reports•

Specialty probes: LongSteer VideoProbe™ 2.2mm and 3.0mm
USB probes, Working Channel VideoProbe, Quartz fiberVideoProbes (UV and radiation tolerance) and Borescope
adapter (connect to rigid borescope and flexible fiberscope)Choose the right tool foryour inspection job.
Mentor Visual iQ VideoProbe is available in three customized platform configurations to meet yourinspection needs across a variety of industries and applications.

Mentor VisualiQ Inspect
Excellent quality andconnectivity
at a value price

Mentor Visual iQ Touch
Improve productivity
with QuickChangeprobes andtouchscreen

SystemInspect

Mentor Visual iQ Analyze
Our most capableVideoProbe with bundled
3D Measurement andProbability of Detection SWfor extreme image quality

TouchAnalyze

CapabilityHigh definition resolution

Y*Y

YConnectivity• Wi-Fi/Bluetooth connectivity, network drive mapping

YY

Y

PC Re-measurement with Inspection ManagerYYYInternal memory

16 GB

32 GB

32 GB

QuickChange probes

Y*YY

Compatible with Custom Probes, i.e., LongSteer, UV, 2.2mm USB, etc.

Y*

Y

Y

Menu Directed Inspection image management and reporting

UpgradeYY

Touchscreen interfaceUpgrade

YY

Real3D Measurement• 3D Phase Measurement with Fully Surfaced Point Cloud• 3D Stereo Measurement with Fully Surfaced Point Cloud• Projected plane measurement• Area depth profile measurement• Blade Tip Clearance auto measure

UpgradeUpgradeY

Radius gauge measurement

UpgradeUpgradeY

Traditional Measurement• Stereo Measurement• Comparison Measurement

UpgradeYUpgradeY

Y

UpgradeUpgradeY

InspectionWorks• Connect: Local+Control• Insight: push or send data to IW cloud portal• Store: download data from IW cloud portalY
UpgradeY

Y
UpgradeY

Y
UpgradeY

Turning Tool Support

UpgradeUpgradeY

USB Video-InY

YY

Always On DVR

YYY

90 Day Free Trial of all SW features

YY

Y

Image Rotation

YY

Y

3D Image StitchingUpgradeUpgradeY**Blade Counter

UpgradeUpgradeUpgradeAssisted Defect Recognition (ADR) analytics:
Gas Power assist, Aiir Edge Cold Section, Aiir Edge Hot Section

UpgradeUpgradeUpgradeProbability of Detection Suite (POD)• Variable High Dynamic Range (HDR) Images• Adaptive Noise Reduction (ANR)• Distortion Correction for Wide FOV tips
• Image Transform Presets• Dark Boost image settings* Requires interchangeable Inspect mode** Requires Real3D measurement
3

The VideoProbe of the future.
The Mentor Visual iQ is changing the way remote visualinspections are being performed. Leveraging artificialintelligence and cloud-based digital tools to help improve
productivity, accuracy, and reliability of inspection data.
Powerful analytics
Artificial intelligence is being utilized across industries andfunctions, the Mentor Visual iQ is designed to enable a user toleverage analytical models from multiple sources.
• Waygate Technologies Analytics. Gas Power assist andBlade Counter are analytics developed to improvereliability and data integrity• Third Party Partner Analytics. Available directly on theMentor Visual iQ, users can leverage Aiir Innovations’aviation commercial engine models for defect detection• Third Party Private Analytics. For customers developing
their own analytics, the Mentor Visual iQ enables
deployment to a global teamDatasheets are available for each analytic to guide users
through applicability and operating procedures.
When and where you need them
The Mentor Visual iQ connects directly to the InspectionWorksStore to provide access to workflows and reference libraries.
Available to all is the Public store which contains manuals,
analytic data sheets, MDI templates and other referencematerials. The Private store is available to InspectionWorks
subscribers providing a secure repository for custom MDI
workflows, analytics, reference documents, or any otherresources your team may need to access and download onto their Mentor Visual iQ.
With Push2IW, data collected during an inspection is
automatically pushed to the InspectionWorks Insight cloud,ensuring against data loss and reducing after-inspection
data management tasks.

4

The Blade Counter analytic showing entry, exit lines and Blade number.

Technical Specifications - Handset
Operating Temperature
Tip-25ºC to 100ºC (-13ºF to 212ºF). Reduced articulation below 0ºC (32ºF)

System-25° to 46°C (-13°F to 115°F). LCD may require warm-up period below 0°C (32°F)

Storage Temperature-25º to 60ºC (-13ºF to 140ºF)

Relative Humidity

95% max, non-condensing

Waterproof

Insertion tube and tip to 14.7 psi (1 bar, 10.2 m of H2O, 33.5 ft of H2O)

CameraDiameter Probe

4.0 mm (0.16”), 6.1 mm (0.24”), 6.2mm (0.24”), 8.4 mm (0.33”)

Image Sensor

1/6” Color Super HAD CCD camera (6.1, 8.4mm), 1/10” Color Super HAD CCD camera (3.9mm, 4.0mm, 6.2mm)

Pixel CountSD 3.9, 4.0, 6.1, 6.2, 8.4 mm - 440k pixels
HD 6.1, 8.4 mm - 1200k pixels

HousingTitanium

SystemSystem Dimensions

17.1 x 19.7 x 38.1 cm (6.75 x 7.75 x 15”)

Case Dimensions

Small carry-on case 54.9x 34.6x 23.6 cm (21.61x13.62x29”)
Medium size dual probe case 55.8 x 33.0 x 63.5 cm (22 x 13 x 25”)
Large workstation case 57.2 x 56.3 x 41.2 cm (22.52 x 22.16 x 16.22”)

System Weight

In small case with contents: 10.2 kg (22.4 lbs). In medium case with contents: 10.9 kg (24 lbs). In large case with contents: 19.6 kg (43.2 lbs)
Out of Case: 6120 system - 3.0 kg (6.7 lbs). 61100 system - 4.0 kg (8.7 lbs)

Construction

Magnesium and Polycarbonate housing with integrated elastomeric bumpers

LCD Monitor

Integrated (6.5”) active matrix XGA color LCD, daylight readable display, with optically bonded, multipoint capacitive
Dragontrail™ touchscreen (Pinch and zoom gesture support, on-screen radio and slider button activation)

Joystick Control

360º All-Way® tip articulation, menu access, and navigation

Button SetAccess user functions, measurement, and digital functions

AudioCompatible with bluetooth headset/microphone

Internal Memory

16 GB (Inspect model) and 32 GB (Touch and Analyze model) internal SSD for user data storage

Data I/O Ports

Two USB 3.0 host ‘A’ ports, one USB 3.0 client micro ‘B’ port

Video Out

DisplayPort

Brightness Control

Auto and Variable

Illumination TypeWhite LEDLong Exposure

Auto – up to 16 seconds

White Balance

Factory default or user definedPower Supply
Lithium Ion Battery

10.8 V (nominal), 73 Wh, 6.8 Ah, two-hour typical run timeCurrent
AC100-240 VAC, 50-60 Hz, <1.5 A RMS

DC18 V, 3.34 A

Standards Compliance and Classifications
MIL-STD-810G

United States Department of Defense Environment Tests Sections 501.5, 502.5, 506.5, 507.5, 509.5, 510.5, 511.5, 514.6, 516.6, 521.3

MIL-STD-461G

U.S. Department of Defense – Electromagnetic Compatibility with RE102, and RS103 – ABOVE DECKStandards ComplianceGroup 1 Class A; EN61326-1, IEC CB Scheme, UL/EN/CSA-C22.2 61010-1, IEC 62133, UN/DOT T1-T8, EU RoHS 2, EU RED Directive, ISTA 2G

IP Rating

IP65 (assembled), IP55 (disassembled)

5Technical Specifications - Handset
SoftwareOperating System

Embedded multi-tasking operating system

User Interface

Menu-driven and soft button operation; menu navigation using either touchscreen or joystick

File Manager

Embedded file manager supports the following operation on files and folders: copy, cut, paste, create, rename, delete, filter,sort, and image recall. USB and internal flash storage

Audio Data

PC Compatible ACC (.M4A file) format

Image Control

Brightness, Long Exposure, Invert, Inverse+, Single View, Illumination, Color Saturation Control, Distortion Correction, AdaptiveNoise Reduction(ANR), High Dynamic Range (HDR) imaging, Dark Boost, and Image Transform Presets

Digital Zoom

Continuous (5x)Image Formats

Bitmap (.BMP), JPEG (.JPG)

Video Format

MPEG4 AVC/H.264 (.MP4 file)

Text Annotation

Built-in on screen text overlay generator with selectable font size

Graphic Annotation

User placement of arrows

Articulation Control

User-selectable steer and Steer-and-Stay* articulation; tip “Home” return to neutral forward-tip orientation

Software Updates

Field upgradeable via upgradeable WiFi or USB ThumbDrive®

Languages

English, Spanish, French, German, Italian, Russian, Japanese, Korean, Portuguese, Chinese, Polish, Hungarian, Finnish and Swedish

Application Software

Menu Directed Inspection* (MDI) software digitally guides inspectors through the inspection process, intelligently names
files and tags files and creates Microsoft® Word and PDF compatible inspection reports (MDI included on Analyze and Touch
models, available as upgrade to Inspect model)

Technical Specifications - Probes and Tips
Probe Measurements

Mentor Visual iQ 4.0 mm tips

Diameter

Length

2.2 mm (0.086”)

1.0m, 1.5m, and 2.0m (3.2, 4.9, 6.6 ft.)Forward View3.0mm (0.118”)

1.0, 1.5, 2 m (3.2, 4.9, 6.6 ft.)

4.0 mm (0.15”)

2.0, 3.0 m (6.6, 9.8 ft.)

6.1 mm (0.24”)T

2.0, 3.0, 3.5, 4.5, 6.0, 10.0 m(6.6, 9.8, 14.8, 19.7, 32.8 ft.)

6.2 mm (.24”)

3.5 m (11.5 ft)

Side View

8.4 mm (0.33”)

2.0, 3.0, 4.5, 6.0, 10.0 m (6.6, 9.8, 14.8, 19.7, 32.8 ft)

Longer lengths up to 30 m are available.
Contact your sales representative for details.

Insertion Tube Length Articulation
4.0mm, 6.1mm, and 8.4mm
Insertion Tube Length

Articulation

2 m – 4.5 m

Up/Down 160° min, Left/Right 160° min

6 m – 10 m

Up/Down 140° min, Left/Right 140° min

Note: Typical articulation exceeds minimum specifications.
6.2mm
Insertion Tube Length

Articulation

3.5m

Up/Down 130° min, Left/Right 130° min

Note: Typical articulation exceeds minimum specifications.

6Part No.

Color

FOV (deg)

DOF mm(in)T4080FF**

None

80

35-inf(1.38-inf)

T40115FNBlack

115

4-inf

(.16-inf)

T40115SN

Red1151-30(.04-1.18)

T40120SFBlue

1206-inf(.24-inf)

StereoProbe™ MeasurementTT
TM405555FGTT

Black

55/55-FWD

5-inf(.20-inf)

TM405555SGTT

Blue

55/55-SIDE

4-inf(.16-inf)** Indicates tip with maximum brightness
TIndicates HD optimized
TT
Stereo and 3D Stereo use same lenses

Mentor Visual iQ 6.1 mm tips
Part No.

Color

FOV (deg)

DOF mm(in)Forward View

Mentor Visual iQ 8.4 mm tips
Part No.

Color

FOV (deg)

DOF mm(in)XLG3T8440FF**None40250-inf(9.84-inf)

XLG3T8440FG

White4080-500(3.15-19.68)

Forward View

T6150FFNone

5050-inf(2.36-inf)

XLG3T6150FG

White

5012-200(.47-7.87)

XLG3T61120FG

Black

1205-120(.20-4.72)

XLG3T6180FN

Orange80

3-20(.12-.79)

XLG3T6190FF

Yellow

9020-inf(.79-inf)

XLG3T6150FB

Purple50 (45 DOV)

12-80(.47-3.15)

Side View

Orangeand BlueXLG3T8440SF**

Brown

40250-inf(9.84-inf)

T6165FF**T

65

65-inf(2.56-inf)

XLG3T8480SG

Green8025-500(.98-19.68)

T61120FF

Gray

12020-inf.79”-inf

XLG3T84120SN

Blue

1204-200(.16-7.87)

XLG3T8480FG

Yellow8025-500(.98-19.68)

T84120FF

Orangeand Blue

12020-inf(0.79-inf)Black

1205-200(.20-7.87)

XLG3T84120FN

StereoProbe MeasurementTT

Side View
XLG3T6150SF

Brown

5045-inf(1.77-inf)

XLG3T6150SG

Green

50

9-160(.35-6.30)

XLG3T61120SG

Blue

120

4-100(.16-3.94)

XLG3T6180SN

Red

80

1-20(.04-.79)

XL4TM61105FGT

Black

105

8-250

(.31-9.84)

XL4TM61105SG

Blue

105

7-250

(.27-9.84)

XL4TM61105FN

Orange105

3-120(.12-4.72)

XL4TM61105SF

Green

105

15-inf(0.59-inf)

XL4TM61105SN

Red

1052-20(.08-.79)

XL4TM61105FF

Yellow

10515-inf(0.59-inf)

XLG3TM616060FGTT

Black

60/60-FWD

4-80(.16-3.15)

XLG3TM615050SGTT

Blue

50/50-SIDE

2-50(0.8-1.97)

T

XLG3TM846060FGTT

Black

60/60-FWD

4-50(.16-1.97)

XLG3TM846060SG

Blue

60/60-SIDE

4-50(.16-1.97)**Indicates tip with maximum brightness
T
Provides increased image detail with HD probes
TT
Stereo and 3D Stereo use same lenses

3D Phase Measurement

TT

StereoProbe Measurement

Mentor Visual iQ 6.2 mm tips
Tip View

Color

FOV (deg)

DOF mm(in)Standard Tip Optics
Forward View
T6280FF

Silver ring80

35 - inf(1.38- inf)

T62120FN

Black120

2 - 200(0.08 - 7.87)

T62120FF

Yellow

120

15 - inf(0.59 - inf)

T62120SN

Blue

120

6 - inf(0.24 - inf)

T6280SFGreen80

15 - inf(0.59 - inf)

Black

60/60

2 - 200(0.08-7.87)

Blue60/60

2 - 200(0.08-7.88)

Side View

StereoProbe Measurement Tip
Forward View
TM626060FN
Side View
TM626060SN

7Technical Specification - Analytics
Analytic Part No.

Description

ApplicabilityMVIQ-BLADECOUNT*Live Analytic: detects, tracks and trigger blade edges
moving in real time

Gas turbine blades (compressor and turbine)

MVIQ-AIIRADR1AN(1 year license)*

Still and live ADR Analytic (x2): automatically detect
defects within aerospace gas turbines

Gas turbine blades (compressor and turbine)
Combustion and TBC coated HPT components
CF6, CFM56 5A/B/C, CFM56 7B, V2500, PW 1100, GEnx, CF34, GE90, LEPA, PW 2000,Trent 700, Trent 900, Trent 1000

ADR analytic capable of detecting and categorizing
defects during live inspections

Gas turbine blades (compressor and turbine)
Combustion and TBC coated HPT components
Gas Power assist

UG-MVIQ-AIIRADR1AN(add to existing device,1 year license)*
MVIQGASPWRADR(perpetual)*
MVIQ GASPWRADR1AN(1 year license)*
UG-MVIQ-GASPWRADR(add to existing device,perpetual)*
UG-MVIQ-GASPWRADR1AN(add to existing device,
1 year license)**Refer to analytic datasheet for more information
Note: UG part numbers will provide a thumb drive to the customer for installation locally, or they can update over-the-air (InspectionWorks).

Copyright 2023 Baker Hughes Company. All rights reserved.
BHCS31352F(06/2023)

waygate-tech.com