シャッターコントローラ F77-7
取扱説明書
Ver 1.00（2011.7.1）

駿河精機株式会社 OST 事業部

1Index１．はじめに...................................................................................................................................................................................................... 4
1.1 安全にご使用頂くために...................................................................................................................................................................................4
1.2 製品概要、特長.............................................................................................................................................................................................................6
1.3 付属品.......................................................................................................................................................................................................................................6
1.4 設置 .............................................................................................................................................................................................................................................6

２．準備、使用例........................................................................................................................................................................................ 7
2.1 使用前の準備 ...................................................................................................................................................................................................................7
2.1.1 機器の接続 ...................................................................................................................................................................................................................8
2.1.2 USBドライバのインストール ...............................................................................................................................................................92.1.2.1 WINDOWSXP..................................................................................................................................................................................................92.1.2.2 WINDOWS7 ..................................................................................................................................................................................................16
2.1.3 F77-7 アプリケーションのインストール............................................................................................................................24
2.1.4 USBドライバのアンインストール ...............................................................................................................................................27
2.1.5 F77-7 アプリケーションのアンインストール................................................................................................................28

3．仕様、機能 ............................................................................................................................................................................................ 30
3.1 基本仕様 ............................................................................................................................................................................................................................30
3.2 各部の名称と機能 ..................................................................................................................................................................................................31
3.3 インターフェイス ..................................................................................................................................................................................................33
3.3.1
3.3.2
3.3.3
3.3.4
3.3.5

電磁シャッター、光路遮断器（ソレノイド）インターフェイス.......................................................................33
RS232C..................................................................................................................................................................................................................33
USB ............................................................................................................................................................................................................................34
I/O制御........................................................................................................................................................................................................................37
インターロック入力 .......................................................................................................................................................................................39

3.4 タイマー精度 ...............................................................................................................................................................................................................40

４．操作、制御方法 ............................................................................................................................................................................. 41
4.1 フロントパネルによる操作 ........................................................................................................................................................................41
4.1.1 起動、バージョン表示..................................................................................................................................................................................41
4.1.2 コントロールモード（CONTROL MODE） .............................................................................................................................41
4.1.3 シャッター操作 ...................................................................................................................................................................................................412

4.1.3.1 バルブモード（BULB MODE）................................................................................................................................................41
4.1.3.2 タイマーモード（TIMER MODE） ........................................................................................................................................41
4.1.4 ソレノイド（光路遮断器）操作 .........................................................................................................................................................424.1.4.1 バルブモード（BULB MODE）................................................................................................................................................424.1.4.2 タイマーモード（TIMER MODE） ........................................................................................................................................42

4.2

F77-7 アプリケーションによる操作 .......................................................................................................................................424.2.1 F77-7 アプリケーション起動 .........................................................................................................................................................42
4.2.2 シャッター操作 ...................................................................................................................................................................................................43
4.2.3 ソレノイドの操作...........................................................................................................................................................................................44
4.2.4 その他の操作.......................................................................................................................................................................................................44

4.3 ユーザプログラムを作成される場合 ..............................................................................................................................................45
4.3.4 通信コマンド一覧 .............................................................................................................................................................................................45
4.3.5 通信コマンド詳細 .............................................................................................................................................................................................47
4.3.5.1 開閉指令コマンド ..................................................................................................................................................................................47
4.3.5.2 問合せコマンド ........................................................................................................................................................................................48
4.3.5.3 ライトコマンド ........................................................................................................................................................................................50
4.3.5.4 その他コマンド ........................................................................................................................................................................................50
4.3.5.5 旧（F77-6）コマンド....................................................................................................................................................................51
4.3.6 エラーコード .........................................................................................................................................................................................................52
4.3.7 通信コマンド送受信手順............................................................................................................................................................................53

4.4 I/Oによる制御............................................................................................................................................................................................................54
4.4.1 シャッター操作 ...................................................................................................................................................................................................54
4.4.2 ソレノイド（光路遮断器）操作..........................................................................................................................................................54５．点検............................................................................................................................................................................................................... 55
６．故障診断と処置 ............................................................................................................................................................................. 55
７．保障とアフターサービス .................................................................................................................................................. 56● 付録............................................................................................................................................................................................................... 57■ DIPスイッチの設定.................................................................................................................................................................................................................57■ F77-7 外観図...........................................................................................................................................................................................................................58■ シャッター、ソレノイド接続ケーブル（型番：F116A-CABLE）...............................................................................................59＜お問い合わせ先＞........................................................................................................................................................................................... 60

3

１．はじめに
このたびは、当社製品をお買い上げ頂き、ありがとうございます。
正しくご使用頂くため、ご使用になる前に この取扱説明書をよくお読み下さい。
お読みになったあとは、いつでも見られるように必ず保管して下さい。
※製品の改良などのため、予告なしに仕様を変更する場合があります。ご了承下さい。
※本製品は一般産業機器向けに設計・製造されているため、以下の用途には使用しないで下さい。
−生命維持装置など人命に関わる装置、機器−航空宇宙機器、原子力システム、交通・輸送機器−上記に準ずる装置（各種安全装置など）

1.1 安全にご使用頂くためにご使用になる前に以下の注意事項を必ずお読み下さい。

！

マークは禁止の意味を表します。

注 意

ここに示された注意事項を必ずお守り下さい。この注意事項を守らなかった場合、けがをしたり、物的な損害を受けたりする可能性があります。
・安全上及び使用上の注意・感電からの保護のため、必ず保護接地端子を備えたコンセント（3P）へ電源ケーブルのプラグを接続してご使用下さい。
保護接地端子を備えていない延長コード（2P）を使用すると保護接地が無効になるので注意して下さい。
・停電時のシャッター動作について・シャッター開の状態で停電または電源 OFF した場合、シャッターは開いたままになります。停電時に安全を確保するため、レーザ電源を遮断するなどお客様の用途に応じて適切な対策を施して下さい。
・配線について・ ケーブルの着脱時は、機器の電源を切って下さい。内部回路が破損する恐れがあります。
・ 電磁シャッター接続コネクタ、光路遮断器接続コネクタは同じコネクタを採用していますので、誤配線のないよう注意して下さい。
・使用環境
・次のような場所でのご使用は避けて下さい。
−ほこりや粉塵（特に金属粉）の多いところ−直射日光の当たるところ−火気に近いところ−振動の大きいところ−水や油のかかるところ−傾きのある不安定なところ−腐食性ガス、可燃性ガスのあるところ
・管理／保管長時間使用しない時、本製品を移動させる時には、電源プラグをコンセントから抜いて下さい。
火災や感電などの思わぬ事故を予防します。

4

！

注 意

・電源について・本機は 交流 100〜240 ボルト（AC100〜240V 50/60Hz）の電源コンセント以外には接続しないで下さい。電源入力範囲を超えた場合、火災、機器破損の原因となります。
・分解／改造・製品の分解、改造、不当な修理は絶対に行わないで下さい。
火災、感電の原因となり、危険です。
・異常がある場合は、当社 OST 事業部カスタマサービスまでご連絡下さい。
・修理のご依頼
・次の場合は、ただちに電源プラグを抜いて下さい。
その後、当社 OST 事業部カスタマサービスまで修理をご依頼下さい。
そのまま使い続けると、火災や感電、けがの原因となります。
−異常な音がする、変な臭いがする、煙が出ているなどの異常な場合−電源ケーブルが傷んだ場合−本製品に水をこぼしたり、内部に異物が入った場合−本製品を落としたり、筐体を破損した場合

5

1.2 製品概要、特長
製品概要
F77-7（以下、本機とします。
）は、電磁シャッターコントローラ F77-6 の後継機種として開発された光学実験用シャッターコントローラです。F77-6 と比べ、以下の特長を有しています。
特長・ 電磁シャッター1 台と光路遮断器（ソレノイド）4 台を本機１台で独立制御
・ USB、RS232C 接続
・ USB ハブ使用で最大 4 台を制御可能
・ I/O 制御（フォトカプラ入力）
・ インターロック入力
・ 電源電圧 AC100〜240V

1.3 付属品
本機には、下記の物が同梱されています。開梱時にご確認下さい。
全てが揃っていない場合は、お手数ですが当社 OST 事業部カスタマサービスまでお問い合わせ下さい。
・ F77-7 本体 ：1 台・ 電源ケーブル（2m） ：1 本・ F77-7 CD-R（F77-7 アプリケーション、USB デバイスドライバ、取扱説明書（本書）） ：1 枚

1.4 設置
本機は、振動の少ない水平な場所に設置して下さい。また、放熱穴が本機の上部に設けてありますので、10mm以上のスペースを空けて設置して下さい。
複数台を横に並べる時には、コントローラ間にスペースを空ける必要はありません。

上面の放熱穴を塞いで設置すると、本機の内部が高温となり、火災や機器損傷の原因となりますのでご注意下さい。

6

２．準備、使用例
2.1 使用前の準備
F77-7 を使用する前の準備の流れを説明します。

機器の準備、接続

USB ドライバのインストール（USB を使用する場合）

F77-7 アプリケーションのインストール（PC 制御する場合）

・
・
・
・
・

F77-7
電源ケーブル（付属品）
電磁シャッター F77-4
光路遮断器 F573-T、F573-R、F116-1、F116-2
専用ケーブル F116A-CABLE（電磁シャッター、光路遮断器付属品）＜外部制御する場合＞
・ RS232C ケーブル（クロス）：（例）D100-R9-2
・ USB ケーブル（B タイプ）・ I/O ケーブル・ 上位機器（パソコン、PLC など）
・ F77-7 CD-R（付属品）

・ F77-7 CD-R（付属品）

動作準備完了※ 弊社の光路遮断器は、ソレノイドを使用しています。便宜上、電磁シャッターを“シャッター”、光路遮断器を“ソレノイド”と表します。

7

2.1.1 機器の接続
F77-7 の接続方法を説明します。

PC

上位機器
PLC

RS232C

RS232Cケーブル
D100-R9-2

USB

I/O

USBケーブル（Bタイプ）oror

I/OケーブルS HUT TER CONTROLLER F 77-7
SHUTTERS OL ENO ID
Up

C ontrol ModeR eadyRem oteL ocal
T/B

ms

I/O

T/B

Down

CH1

C H1

C H2

C H4

CH1〜4

CH1

付属電源ケーブル

CH3

ケーブル
F116A-CABLE

AC100V

AC100V

電磁シャッター
F77-4

①
②
③

光路遮断器（ソレノイド）
F573-T、F573-R
F116-1、F116-2

電磁シャッター、光路遮断器（ソレノイド）を専用ケーブルにて接続します。
PC 等の上位機器により外部制御する場合は、RS232C、USB、または I/O ケーブルを接続します（通信ケーブル、I/O ケーブルは本機に付属しませんので、必要に応じて別途ご用意下さい。
）。
AC インレットと AC100V コンセントを付属の電源ケーブルで接続します（AC200V 系電源を使用する場合は、別途、適切な電源ケーブルをご用意下さい。
）。

8

2.1.2

USB ドライバのインストール

パソコンへの USB ドライバのインストール方法を説明します。本機と PC を接続すると「USB Serial
Converter」、「USB Serial Port」の 2 つデバイスが認識されますので、指示に従いインストール作業を行って下さい。
尚、本機の USB ドライバは以下の OS に対応しています。
＜対応 OS＞
・ Windows 7・ Windows 7 x64
・ Windows Vista・ Windows Vista x64
・ Windows XP
・ Windows XP x64
・ Windows 2000

2.1.2.1 WindowsXP
1.
2.
3.
4.

パソコンの電源を入れ、WindowsXP を起動します。
パソコンの CD ドライブに“F77-7 CD-R”をセットします。
パソコンの USB ポートと F77-7 の USB コネクタを USB ケーブルにて接続します。
F77-7 の電源を入れます。
パソコンが F77-7 の接続を認識すると、以下のメッセージが表示されます。

5.

[一覧または特定の場所からインストールする(詳細)]を選択し、[次へ]ボタンをクリックします。

9

6.

[検索しないで、インストールするドライバを選択する]にチェックを入れ、[次へ]ボタンをクリックします。

7.

ハードウェアの一覧が表示されるので、[USB コントローラ]を選択し [次へ]ボタンをクリックします。

8.

[ディスク使用]ボタンをクリックします。

10

9.

[参照]ボタンから USB ドライバが保存されたフォルダ「F77-7 USB Driver」を指定し、[OK]ボタンをクリックします。

10. 「SURUGA SEIKI F77-7 USB Serial Converter」が表示されることを確認し、[次へ]ボタンをクリックします。

11. ハードウェアのインストール画面が表示されるので、[続行]ボタンをクリックします。

1112. インストールが始まります。
新しいハードウェアの検出ウィザード画面が表示されます。

13. [完了]ボタンをクリックします。

14. 引き続き、USB Serial Port のインストールを行います。

1215. [一覧または特定の場所からインストールする(詳細)]を選択し、[次へ]ボタンをクリックします。

16. [検索しないで、インストールするドライバを選択する]にチェックを入れ、[次へ]ボタンをクリックします。

17. ハードウェアの一覧が表示されるので、[ポート（COM と LPT）]を選択し [次へ]ボタンをクリックします。

13

18. [ディスクの使用]ボタンをクリックします。

19. [参照]ボタンから USB ドライバが保存されたフォルダ「F77-7 USB Driver」を指定し、[OK]ボタンをクリックします。

20. ハードウェアのインストール画面が表示されるので、[続行]ボタンをクリックします。

14

21. インストールが始まります。
新しいハードウェアの検出ウィザード画面が表示されます。

22. [完了]ボタンをクリックします。

23. インストールが正常に完了すると以下の画面が表示されます。

15

24. デバイスマネージャーにて COM ポート番号を確認することができます（例は COM6）。

2.1.2.2 Windows7
1.
2.
3.
4.

パソコンの電源を入れ、Windows7 を起動します。
パソコンの CD ドライブに“F77-7 CD-R”をセットします。
パソコンの USB ポートと F77-7 の USB コネクタを接続します。
F77-7 の電源を入れます。
パソコンが F77-7 の接続を認識すると、以下のメッセージが表示されます。

この動作は Windows7 の仕様ですので不具合ではありません。

16

5.

スタートメニューから「コントロールパネル」を開きます。

6.

「コントロールパネル」の一覧から「デバイスマネージャー」を選択します。

17

7.

ハードウェアの一覧の中から、「FT232R USB UART」を選択し 右クリックで[ドライバーソフトウェ
アの更新]を選択します。

8.

[コンピュータを参照してドライバーソフトウェアを検索します]を選択します。

18

9.

[参照]ボタンから USB ドライバが保存されたフォルダ「F77-7 USB Driver」を指定し、[次へ]ボタンをクリックします。

10. Windows セキュリティー画面が表示されたら、[このドライバーソフトウェアをインストールします]を選
択します。

19

11. インストールが始まります。

12. 正常にインストールされると、以下の画面が表示されます。

20

13. 引き続き、USB Serial Port のインストールを行います。

14. ハードウェアの一覧の中から、「USB Serial Port」を選択し 右クリックで[ドライバーソフトウェアの更新]を選択します。

15. [コンピュータを参照してドライバーソフトウェアを検索します]を選択します。

21

16. [参照]ボタンから USB ドライバが保存されたフォルダ「F77-7 USB Driver」を指定し、[次へ]ボタンをクリックします。

17. Windows セキュリティー画面が表示されたら、[このドライバーソフトウェアをインストールします]を選
択します。

22

18. インストールが始まります。

19. 正常にインストールされると、以下の画面が表示されます。

2320. デバイスマネージャーにて COM ポート番号を確認することができます（例は COM4）
。

2.1.3

F77-7 アプリケーションのインストール

パソコンへの F77-7 アプリケーションのインストールの方法について説明します。
尚、本アプリケーションは以下の OS に対応しています。
＜対応 OS＞
・ Windows 7・ Windows Vista・ Windows XP※いずれも、64bit OS では動作保障していません。
① パソコンのドライブに F77-7 CD-R をセットします。
② CD-R（F77-7 application フォルダ）内の setup.exe ファイルをダブルクリックします。
セットアップウィザード画面が起動します。

24

③ [次へ]ボタンをクリックします。
インストールフォルダの選択画面が表示されます。
④ インストール先のフォルダを選択し、[次へ]ボタンをクリックします。
プログラムフォルダの選択画面が表示されます。
※ インストール先のフォルダは、初期設定が C：\Program Files\SURUGA\F77-7 application となります。変更が必要な場合は、[参照]ボタンを押し、任意のフォルダを指定して下さい。
※ 変更の必要の無い場合は[次へ]ボタンをクリックして次の手順に進んで下さい。

25

⑤ インストールの確認画面が表示されます。
インストールを進める場合、[次へ]ボタンをクリックします。

⑥ F77-7 プログラムをインストールします。

26

⑦ インストール完了の画面が表示されます。
[閉じる]ボタンをクリックし、F77-7 アプリケーションのインストールは終了です。

2.1.4

USB ドライバのアンインストール

USB ドライバのアンインストールの方法について説明します。ここでは WindowsXP を例に説明しますが、Windows7 の場合も同様です。
① パソコンと本機を USB ケーブルで接続し電源を入れます。
② デバイスマネージャーを起動し、ハードウェアの一覧の中から、「SURUGA SEIKI F77-7 USB Serial
Port(COM□)」を選択し 右クリックで[削除]を選択します。

※

必ず、「SURUGA SEIKI F77-7 USB Serial Port(COM□)」から削除してください。

③ 「デバイスの削除の確認」画面が表示されたら、[OK]ボタンを押します。

27

④ 同様にハードウェアの一覧の中から、「SURUGA SEIKI F77-7 USB Serial Converter」を選択し 右ク
リックで[削除]を選択します。

⑤ 「デバイスの削除の確認」画面が表示されたら、[OK]ボタンを押します。
アンインストールが完了します。

2.1.5

F77-7 アプリケーションのアンインストール

F77-7 アプリケーションのアンインストールの方法について説明します。ここでは WindowsXP を例に説明しますが、Windows7 の場合も同様です。
① パソコンのドライブに F77-7 CD-R をセットします。
② CD-R 内の setup.exe ファイルをダブルクリックします。
セットアップウィザード画面が起動します。

28

③ [F77-7apprication の削除]を選択し、[完了]ボタンをクリックすると削除を開始します。

④ 正常に削除されると、以下の画面が表示されます。
[閉じる]ボタンをクリックし、F77-7 アプリケーションの削除は完了です。

※「コントロールパネル」の「プログラムの追加と削除」からも削除できます。

29

3．仕様、機能
3.1 基本仕様【シャッターコントローラ F77-7】
型番
使用環境
保存環境
電源電圧
外形寸法
重量
電磁シャッター型番
電磁シャッターch 数
シャッタースピード（タイマー）
電磁シャッター用コネクタ
光路遮断器型番
光路遮断器 ch 数
光路遮断器タイマー
光路遮断器用コネクタ
RS232CUSB
外部制御入力(I/O)
通信コマンドエラーコードInterlock 端子
メモリバックアップ
RoHS

F77-7
0〜40℃ 20〜80%RH（非結露）-20〜60℃ 20〜80%RH（非結露）AC85〜264V 50/60Hz
W×D×H＝210.6×170×62.2(mm)
突起物（ゴム足、ネジ）含まず
1.6kg
F77-4
1ch
11〜9,999,999ms（但し、フロントパネル操作は、11〜9,999ms）
HR10A-7R-4S(73) （ヒロセ電機）
F116-1、F116-2、F573-T、F573-R
4ch
各 ch 100〜9,999,999ms（但し、フロントパネルからは変更不可）
HR10A-7R-4S(73) （ヒロセ電機）
Dsub9 ピン（オス）/ボーレート：4800/9600/19200/38400bps（選択）USB2.0 （仮想 COM ポート） ハブ接続により最大 4 台接続
Dsub9 ピン（メス）／フォトカプラ入力
新コマンド体系（F77-6 互換モードあり）〇〇（通常時短絡）
シャッター開閉モード、シャッター・ソレノイドのタイマー値を保持する。
（書き換え 100 万回）〇

30

3.2 各部の名称と機能【F77-7 フロントパネル】
3

4

5

6

7

8

SHUTTER CONTROLLER F77-7
SHUTTE R

SOLENOIDUp

Control Mode

Ready
Remote
Local

T/B

ms

I/O

T/B

Down

CH1

CH1

1

2

9

CH2

10

CH3

11

CH4

12

説明
本体の電源スイッチです。
シャッタースピード（タイマー）を表示します（単位：ms）。
注）ソレノイドのタイマー値は Local モードでは変更できません。
③ シャッタースピード（タイマー） 押すとシャッタースピード（タイマー）が上がります。
設定ボタン（アップ）
1 秒以上の長押しで値が連続的にカウントアップします。
④ シャッタースピード（タイマー） 押すとシャッタースピード（タイマー）が下がります。
設定ボタン（ダウン）
1 秒以上の長押しで値が連続的にカウントダウンします。
⑤ レディー表示
レディー（シャッター開準備完了）時に点灯します。
⑥ 開閉モード切替ボタン
押す度に開閉モードが交互に切り替わります。
（シャッター）
T：Timer Mode（タイマーモード）※選択時、点灯（橙）B：Bulb Mode（バルブモード）※選択時、消灯⑦ シャッターCH1 開閉ボタン
閉時に押すとシャッターが指定の開閉モードで開きます。
開時に押すと閉じます。シャッター開時に点灯（緑）します。
⑧ 開閉モード切替ボタン
押す度に開閉モードが交互に切り替わります。
（ソレノイド 4CH 共通）※2
T：Timer Mode（タイマーモード）※選択時、点灯（橙）B：Bulb Mode（バルブモード）※選択時、消灯⑨ ソレノイド CH1 開閉ボタン
OFF 時に押すとソレノイドCH1 が指定の開閉モードでON になります。ON 時に押すと OFF になります。ソレノイド CH1 ON 時に点灯（緑）します。
⑩ ソレノイド CH2 開閉ボタン
OFF 時に押すとソレノイドCH2 が指定の開閉モードでON になります。ON 時に押すと OFF になります。ソレノイド CH2 ON 時に点灯（緑）します。
⑪ ソレノイド CH3 開閉ボタン
OFF 時に押すとソレノイドCH3 が指定の開閉モードでON になります。ON 時に押すと OFF になります。ソレノイド CH3 ON 時に点灯（緑）します。
⑫ ソレノイド CH4 開閉ボタン
OFF 時に押すとソレノイドCH4 が指定の開閉モードでON になります。ON 時に押すと OFF になります。ソレノイド CH4 ON 時に点灯（緑）します。
⑬ 制御モード切替スイッチ
制御モードを Remoto、Local、I/O の 3 段階で切り替えます。
①②

名称
電源スイッチシャッタースピード表示 ※1

13※1 ソレノイドのタイマー値は Local モードでは変更できません。
※2 Remote モードの時は、最後に設定したチャンネルの開閉モードが表示されます。
31

【F77-7 リアパネル】
8

9

10

11

SHUTTE R

2

1

SOLENO ID

CH1

RS2 32C

12

CH1

CH2

USB

DIPSW

1

2 3

4

CH3

CH4

In terlock

I/O

6

7

5 6

F U S E( 2 5 0 V / 1 A)
5 .2 × 2 0 . 0 mm
V o l ta g e
A C1 0 0 - 2 4 0 V
W a t ta g e
m ax 4 0 W
F r e qu e n c y
5 0 /6 0 H z

ON

3

4

5

①

名称
AC インレット②

ヒューズホルダ／ヒューズ③④⑤

RS232C コネクタ
USB コネクタ
DIPSW（6bit）⑥

Interlock コネクタ

⑦
⑧
⑨
⑩
⑪
⑫

I/O コネクタ
シャッターCH1 接続コネクタ
ソレノイド CH1 接続コネクタ
ソレノイド CH2 接続コネクタ
ソレノイド CH3 接続コネクタ
ソレノイド CH4 接続コネクタ

説明
AC100〜240V 50/60Hz を入力します。
AC100V の場合は、本機付属の電源ケーブルをご使用ください。
AC200V 系の場合は、適切な電源ケーブルを別途ご用意下さい。
250Ｖ、1Ａのガラス管ヒューズ（5.2×20mm）を使用しています。
RS232C 通信用コネクタです（Dsub9P オス（インチネジ））。
USB 通信用コネクタです（B タイプ）。
RS232C ボーレート(2bit)、USB ID(2bit)、
コマンドレスポンス(1bit)、旧コマンド有効/無効(1bit)を設定する
ディップスイッチです。
インターロック入力用コネクタです（ソフトウェアインターロッ
ク、B 接点）。出荷時はジャンパー線により解除されています。
I/O 用コネクタです（Dsub9P メス（M2.6 ネジ））。
電磁シャッター（CH1）を接続します。
ソレノイド（CH1）を接続します。
ソレノイド（CH2）を接続します。
ソレノイド（CH3）を接続します。
ソレノイド（CH4）を接続します。

32

3.3 インターフェイス
3.3.1 電磁シャッター、光路遮断器（ソレノイド）インターフェイス
電磁シャッター接続用コネクタと光路遮断器（ソレノイド）接続用コネクタは以下の通りです。
●電磁シャッター
ピン番号
1
2
3
4

端子機能
電磁シャッター ＋（青）電磁シャッター −（黄）
N.C.
N.C.

【電磁シャッター接続用コネクタ】
コネクタ型番： HR10A-7R-4S(73) （ヒロセ電機）
適合プラグ： HR10A-7P-4P(73) （ヒロセ電機）※光路遮断器用と同じ※光路遮断器用と同じ●光路遮断器（ソレノイド）
ピン番号
1
2
3
4

端子機能
ソレノイド（DC24V+）
ソレノイド（GND）
N.C.
N.C.

【光路遮断器接続用コネクタ】
コネクタ型番： HR10A-7R-4S(73) （ヒロセ電機）
適合プラグ： HR10A-7P-4P(73) （ヒロセ電機）※電磁シャッター用と同じ※電磁シャッター用と同じ

シャッター、
ソレノイド共に同じコネクタを採用していますので、挿し間違えのないよう十分ご注意下さい。

3.3.2 RS232C
本機のリアパネルにあるディップスイッチによりボーレートの設定を行い、ＲＳ２３２ＣコネクタとパソコンのRS232C コネクタとを RS232C クロスケーブル（例：D100-R9-2）により接続します。
・ ディップスイッチの設定は必ず本機の電源投入前に行って下さい。電源投入後のディップスイッチの変更は誤動作の原因となります。
・ ケーブルの接続の前に、本機と周辺機器の電源が切れていることを確認して下さい。
電源が入った状態でケーブルの取り付け、取り外しは機器破損の恐れがありますので絶対に行わないで下さい。

33

【RS232C コネクタ】
RDED-9P-LNA(4-40)(55) （ヒロセ電機製：Dsub9P オス（インチネジ）
）
ピン番号
１
２
３
４
５
６
７
８
９

名称
−
ＲｘＤ（ＲＤ）
ＴｘＤ（ＳＤ）−
ＧＮＤ（ＳＧ）−−−−【通信パラメータ】
伝送手順
ボーレート
データ長
パリティ
ストップビット
Ｘパラメータハンドシェーク
デリミタ

機能
未使用
受信データ（入力）
送信データ（出力）
未使用
信号用接地
未使用
未使用
未使用
未使用

調歩同期
4,800、9,600、19,200、38,400bps
８ビット
なし
１ビット
なし
なし
ＣＲ

【DIP スイッチの設定】

NO
１ ２ ３ ４ ５ ６

DIPSW● RS232C ボーレートの設定
1
2
ボーレート
OFF
OFF
4,800bps
ONOFF
9,600bps
OFFON
19,200bps
ONON
38,400bps（初期値）※工場出荷時の設定は 38,400bps です。

3.3.3

USB

RS232C ポートを持たないノート PC などから本機を外部制御する場合、USB 接続が利用できます。付属のUSB デバイスドライバをインストールして、PC と本機を接続すると、仮想 COM ポートとして PC に認識されます。
尚、USＢケーブルは市販の B タイプ USB ケーブルをご使用下さい。
※USB デバイスドライバのインストール方法は、２．１．２節を参照下さい。
※ 本機の USB 接続は仮想 COM 接続です。USB-RS232C 変換用 IC を使用しデータを変換しているため、3．3．２節の RS232C 接続より内部処理速度は若干下がります。システムのタクトタイムを追求される場合は、RS232C のご使用をお奨めします。

34

【USB コネクタ】
B タイププラグ

XM7B-0442 （オムロン）

ピン番号
１
２
３
４
【通信パラメータ】
USB 規格
ボーレート
データ長
パリティ
ストップビット
デリミタ

名称
Ｖbus
−DATA
＋DATA
ＧＮＤ
USB2.0 準拠（仮想 COM 接続）
38,400bps
８ビット
なし
１ビット
ＣＲ

【ベンダーID，プロダクト ID】
ベンダーID は、ベンダー毎の ID となります。
ベンダーID（ｄｅｃｉｍａｌ ３５８１ ｈｅｘ ０ＤＦＤ）を ROM に記憶しています。
プロダクト ID は、機種の管理ＩＤとなります。本機のプロダクト ID（ｈｅｘ ０００3）を ROM に記憶しています。
パソコン側の USB ドライバは、デバイス側のベンダーID 及びプロダクト ID を認識し、プラグアンドプレイで、動作します。

【USB ハブ接続（複数台制御）】
2 台以上のコントローラを一台の PC から制御する場合、USB ハブ接続により本機を接続して使用します（最大 4 台）。
ハブ接続台数USB ID

最大 4
0〜3（DIP スイッチにより設定）

重複しない USB ID を設定し、USB ハブ、USB ケーブルを接続後、電源を入れると、空いている COM ポー
ト番号に順次割り当てられます。
＜COM ポート番号と USB ID の関係＞
COM ポート番号とUSB ID の関係を確認するためには、「SURUGA SEIKI F77-7 USB Serial Port(COM□)」
として割り当てられた各ポートに通信コマンド“USBID”を送って下さい。すると USB ID 設定値”0“〜”3“のいずれかが返信され、COM ポート番号と USB ID の関係が分かります。一度、COM ポート番
号が決まると、ドライバを削除するか、以下の方法で COM ポート番号を変更するまで固定となります。

35

＜COM ポート番号を変更する場合＞
デバイスマネージャーの「ポート（COM と LPD）」の下の「SURUGA SEIKI F77-7 USB Serial Port(COM□)」のプロパティーを開くと以下の画面が表示されます。

※ 設定値の変更は不要です（初期値で動作します。
）。

詳細設定ボタンを押すと下記のような画面が表示され、未使用の COM ポート番号へ任意に変更できます。

36

【DIP スイッチの設定】
ＳＷ1 の 3 ビット目、4 ビット目を設定します。

NO
１ ２ ３ ４ ５ ６

DIPSW● USB ID の設定
3
4
OFF
OFF
ON
OFF
OFF
ON
ON
ON
※工場出荷時の設定は OFF です。

USB ID0（初期値）
1
2
3

3.3.4 I/O 制御
外部入力によるシャッター、ソレノイドの開閉を行います。
【I/O コネクタ】
コネクタ型番： XM3B-0922-112 （オムロン製：Dsub9P メス（M2.6 ネジ））
適合プラグ一例： XM3A-0921 （オムロン製：Dsub9P オス）
ピン番号
名称
機能
1
DC24V+
DC24V+出力
2
DC24V+
DC24V+出力
3
電源入力
DC5〜24V 電源入力
4
SO1
ソレノイド CH1 開閉
5
SO2
ソレノイド CH 2 開閉
6
SO3
ソレノイド CH 3 開閉
7
SO4
ソレノイド CH 4 開閉
8SH1シャッターCH 1 開閉
9GND (SG)
シグナルグランド37

【接続例】●F77-7 内部電源（DC24V）を使用する場合【 ユー ザ 回 路】

【 F77-7】+24V
接 続 は 1 番 , 2番 の い ず れ か で 可

ソ レ ノイ ド CH1

ソ レ ノイ ド CH2

1

1

2

2

3

3

4

4

5

5

6

6

7

7

8

8

9

9

1.8kΩ 1W

PC
SO1

TLP621

GND
SO2

GND

SO3

ソ レ ノイ ド CH3

SO4

ソ レ ノイ ド CH4

SH1

シ ャ ッタ ー CH1●外部電源（DC5〜24V）を使用する場合【 ユー ザ 回 路】

【 F77-7】+24V+5〜24Vソ レ ノイ ド CH1

ソ レ ノイ ド CH2

1

1

2

2

3

3

4

4

5

5

6

6

7

7

8

8

9

9

1.8kΩ 1W

PC
SO1

TLP621

GND
SO2

GND

SO3

ソ レ ノイ ド CH3

SO4

ソ レ ノイ ド CH4

SH1

シ ャ ッタ ー CH1

GND

38

3.3.5 インターロック入力
インターロックコネクタの 1 ピンと 2 ピンをオープンにすることで、シャッターおよび全ての光路遮断器を閉じることができます。
【インターロックコネクタ】
コネクタ型番： XW4A-02B1-H1 （オムロン）
適合ソケット： XW4B-02B1-H1 （オムロン）
ピン番号
1
2

信号名
INTERLOCK (+)
INTERLOCK (-)

I/O
I
-

※付属品

機能
INTERLOCK 信号入力(B 接)
GND【インターロック回路、接続例】
+24V
2.4kΩ
INT

ｶﾊﾞｰｽｲｯﾁ等

1

1

2

2

TLP621
GND

GND※ インターロック信号を使用しない場合は、付属の解除コネクタを常時挿してご使用下さい。

ソフトウェアのインターロック機能ですので、システムの最終保護機能として使用しないで下さい。

39

3.4 タイマー精度
本機のタイマー精度は以下の通りです。尚、シャッターの開閉動作を優先しているため、ソレノイドの精度、応答性はシャッターより若干劣ります。
【シャッター】
コントロールモード
Local（タイマー）

Remote（タイマー）

I/O（バルブ）

I/O（タイマー）

タイマー精度
11〜19ms の設定に対する誤差：±2.5％以内
20〜99ms の設定に対する誤差：±2％以内
100〜9,999,999ms の設定に対する誤差：±1％以内
11〜19ms の設定に対する誤差：±2.5％以内
20〜99ms の設定に対する誤差：±2％以内
100〜9,999,999ms の設定に対する誤差：±1％以内＜通信応答時間 参考値＞
RS232C（SHUTTER1:TIMER1000:OPEN の場合）
：約 19ｍｓ
RS232C（SH1:O の場合）
：約 7ｍｓUSB（SHUTTER1:TIMER1000:OPEN の場合）
：約 19〜39ｍｓ
USB（SH1:O の場合）
：約 7〜27ｍｓ※USB の場合は、RS232C に比べ、0〜20ms 程、応答が遅れます。
入力信号に対する開信号の遅れ：1〜3ms
入力信号に対する閉信号の遅れ：1〜3ms※パルス幅が短いほど、誤差の割合が大きくなります。
50ms 以下の場合、タイマーモードの使用を推奨します。
（例）11ms 入力→誤差±15%以内
20ms 入力→誤差±10%以内
50ms 入力→誤差±3%以内
100ms 入力→誤差±2%以内
入力信号に対する開信号の遅れ：1〜3ms
11〜19ms の設定に対する誤差：±2.5％以内
20〜99ms の設定に対する誤差：±2％以内
100〜9,999ms の設定に対する誤差：±1％以内【ソレノイド ※1】
コントロールモード

タイマー精度
100〜499ms の設定に対する誤差：±3％以内
Local（タイマー）
500〜9,999,999ms の設定に対する誤差：±2％以内
100〜499ms の設定に対する誤差：±3％以内
Remote（タイマー）
500〜9,999,999ms の設定に対する誤差：±2％以内
入力信号に対する開信号の遅れ：1〜6ms
I/O（バルブ）
入力信号に対する閉信号の遅れ：1〜6ms
入力信号に対する開信号の遅れ：1〜6ms
I/O（タイマー）
100〜499ms の設定に対する誤差：±3％以内
500〜9,999,999ms の設定に対する誤差：±2％以内※1 F77-7 から出力される駆動電圧の ON/OFF 精度を表しています。光路遮断器の遮光板の動作精度は表しておりません。

40

４．操作、制御方法
4.1 フロントパネルによる操作
4.1.1 起動、バージョン表示
本機の電源を入れると本機のファームウェアバージョンが約 1 秒間表示されます。その後、シャッタータイマー
値（シャッタースピード）表示に切り替わりパネル操作可能な状態となります。

ms例）バージョン表示（V100）

ms→

シャッタータイマー値表示（0100）

4.1.2 コントロールモード（Control Mode）
フロントパネルから操作を行う時は、Control Mode を Local に合わせて下さい。Remote、I/O Mode では、操作できません（I/O 時の開閉モード、シャッタータイマー値は変更可）。

4.1.3 シャッター操作
シャッター操作は、ボタンを押すと開き、再度押すと閉じるバルブモード、一回ボタンを押すと設定された時間だけ開いて自動的に閉じるタイマーモードがあります。

4.1.3.1 バルブモード（Bulb Mode）
T/B ボタンが消灯している状態がバルブモードです。点灯している場合は、T/B ボタンを押してバルブモード（消灯）に切り替えます。CH1 ボタンを押すとシャッターが開きます。再度 CH1 ボタンを押すとシャッターが閉じます。

4.1.3.2 タイマーモード（Timer Mode）
T/B ボタンが点灯している状態がタイマーモードです。消灯している場合は、T/B ボタンを押してタイマーモード（点灯）に切り替えます。タイマー値を変更する場合は、up ボタンまたは down ボタンを押して希望する値に合わせます。CH1 ボタンを押すとシャッターが開き、指定の時間が経過すると自動的に閉じます。タイマー時
間経過前に再度 CH1 ボタンを押すと、強制的にシャッターを閉じます。
※ シャッターが閉じて、次に開くことができる状態になるまで、
約 1 秒かかります。
準備完了すると Readyが点灯（緑）します。
※ シャッターの開閉モード、タイマー値は、CH1 ボタンを押すと EEPROM に保存されるため、電源をOFF しても保持されます。但し、タイマー値、開閉モードを変更しても CH1 ボタンを押さなければ保持されません.。
※ フロントパネルから設定できるシャッターのタイマー値は、11〜9999ms です。9999ms を超えた
値で使用する場合は、付属のアプリケーションまたは、ユーザプログラムにて設定して下さい。尚、9999ms を超えた値を Remote モードにて設定した場合、タイマー値表示部は、“OVER”と表示されます。
※ OPEN 中に Control Mode を Remote または I/O に切り替えた場合、タイマー値表示部に“E100”が表示されますので、Local に戻して操作を行って下さい。
※ シャッターOPEN 中は、開閉モードの切り替えはできません。

41

4.1.4 ソレノイド（光路遮断器）操作
ソレノイド操作は、シャッターと同様にボタンを押すと開き、再度押すと閉じるバルブモード、一回ボタンを押すと設定された時間だけ開いて自動的に閉じるタイマーモードがあります。

4.1.4.1 バルブモード（Bulb Mode）
T/B ボタンが消灯している状態がバルブモードです。点灯している場合は、T/B ボタンを押してバルブモード（消灯）に切り替えます。CH1 ボタンを押すとソレノイドが ON になります。再度 CH1 ボタンを押すとソレノ
イドが OFF になります。

4.1.4.2 タイマーモード（Timer Mode）
T/B ボタンが点灯している状態がタイマーモードです。消灯している場合は、T/B ボタンを押してタイマーモード（点灯）に切り替えます。タイマー値の変更は、付属のアプリケーションまたはユーザプログラムにより行います。CH1 ボタンを押すとソレノイドが ON になり、指定の時間が経過すると自動的に OFF になります。タイ
マー時間経過前に再度 CH1 ボタンを押すと、強制的にソレノイドを OFF にします。
※ ソレノイド ON 中は、開閉モードの切り替えはできません。
※ ソレノイド ON 中に Control Mode を Remote または I/O に切り替えた場合、シャッタータイマー値
表示部に“E100”が表示されますので、Local に戻して操作を行って下さい。

4.2

F77-7 アプリケーションによる操作

F77-7 アプリケーションを使用すると、シャッターの開閉、ソレノイドの ON/OFF を PC から簡単に制御することができます。本アプリケーションを使用する場合は、コマンドレスポンスを有効（DIPSW5 を ON）、旧コ
マンドを無効（DIPSW6 を OFF）に設定して下さい（いずれも初期設定）
。

4.2.1

F77-7 アプリケーション起動

２．１．３節の手順で F77-7 アプリケーションをインストールすると Windows のスタートメニューから
F77-7 アプリケーションを起動できます。起動すると以下の画面が表示されます。通信ケーブルを接続し、電源が ON になっている状態で、Port Number（ポート番号）と Baud Rate（ボーレート）を合わせ、[OPEN]ボタ
ンをクリックします。

本機と正常に通信が行われると、本機に設定されている値を読み込み、以下のように操作可能となります。
42

4.2.2 シャッター操作
シャッターの動作には画面左の SHUTTER 表示部を使用します。
＜シャッター操作画面＞

③

②

④
⑥

①
②
③
④
⑤
⑥

⑤

①

開閉モードを選択します。→バルブモードの場合は、⑤へ。
タイマー値（シャッタースピード）を設定します。
タイマー値のカウントアップ、ダウンを行います。
表示されているタイマー値を本機にセットします（省略可能）。
指定のモードでシャッターの開閉を行います。
レディー状態を表示します。

＜基本操作＞
i. 開閉モード（Timer または Bulb）を選択します。
ii. [Up]、[Down]ボタン、またはキーボードによる直接入力によりタイマー値（シャッタースピード）を設定します。
iii. [SET]ボタンを押し、本機へ設定します（省略可能）。
iv. [CH1]ボタンを押すと、指定の開閉モードでシャッターが開きます（[SET]ボタンを押さなくても表示されているタイマー値で開閉します。
））。
※ シャッターOPEN 中は、[CH1]ボタンが緑色に変わります。
※ シャッターOPEN 中に再度[CH1]ボタンを押すと、CLOSE します。
※ レディー表示は、レディー（準備完了）時、緑色の表示となります。

43

4.2.3

ソレノイドの操作

ソレノイドの動作には画面右の SOLENOID 表示部を使用します。
＜ソレノイド操作画面＞
ここでは、CH1 の操作を説明しますが、CH2〜4 も同様に操作して下さい。

②

③

⑤

④
①

① 開閉モードを選択します。→バルブモードの場合は、⑤へ。
② タイマー値を設定します。
③ タイマー値のカウントアップ、ダウンを行います。
④ 表示されているタイマー値を本機にセットします（省略可能）。
⑤ 指定のモードでソレノイドの ON/OFF を行います。
＜基本操作＞
i. 開閉モード（Timer または Bulb）を選択します。
ii. [Up]、[Down]ボタン、またはキーボードによる直接入力によりタイマー値を設定します。
iii. [SET]ボタンを押し、本機へ設定します（省略可能）。
iv. [CH1]ボタンを押すと、指定の開閉モードでソレノイドが ON します（[SET]ボタンを押さなくても表示されているタイマー値で ON/OFF します。
））。
※ ソレノイド ON 中は、[CH1]ボタンが緑色に変わります。
※ ソレノイド ON 中に再度[CH1]ボタンを押すと、OFF します。

4.2.4

その他の操作＜Command → LEDOFF＞
フロントパネルの LED を消灯します。
＜Command → LEDON＞
フロントパネルの LED を点灯します。
＜Command → WRITE＞
全てのタイマー値、シャッターの開閉モード、ソレノイド CH1 の開閉モードを EEPROM に書込みます（電源 OFF 時も保持する。
）。
＜Command → RESET＞
全てのタイマー値、シャッター、ソレノイドの開閉モードを初期化します。
※ RESET コマンドだけでは EEPROM には書込みません。必要に応じて、RESET コマンド送信後にWRITE コマンドを送ってください。
＜Input Command＞
通信コマンドを直接入力し、開閉動作や受信データの確認ができます。

※ここでの操作はメイン画面には反映されませんので、必要に応じて、「Screen Update」ボタンを押して下さい。
＜Help → About F77-7 application＞
F77-7 アプリケーションのバージョンを表示します。
44

4.3 ユーザプログラムを作成される場合
本機は RS232C または、USB を使用して PC と通信コマンドを送受信することにより、シャッターおよびソレノイド（光路遮断器）を自由に制御することができます。

4.3.4 通信コマンド一覧
本機の通信コマンドは、開閉指令コマンド、問合せコマンド、ライトコマンド、リセットコマンド、旧コマンドに大別されます。
コマンド
内容
開閉指令コマンドチャンネル指定、モード指定、タイマー設定、開閉動作を行います。
問合せコマンド
シャッター、ソレノイドの状態（ステータス）を要求するためのコマンドです。
ライトコマンド
EEPROM（不揮発性メモリ）へ開閉モード、タイマー値を書込みます（４．３．
５．3 節参照）。
リセットコマンド開閉モード、タイマー値を初期化します（４．３．５．4 節参照）。
旧コマンド
旧コントローラ（F77-6）のコマンドです。DIPSW（6 番）にて旧コマンドモー
ドを有効にした場合のみ機能します。
・ コマンド中の□は設定データを表します。
・ コマンド中の：（コロン）はチャンネル指定が必要なコマンドを意味します。
・ コマンドは、大文字・小文字ともに可能とします（混在可能です。
）。
・ 記載されているコマンドの小文字表記部は省略可能です。
（ ）内は、小文字表記の省略されたコマンドです。

機能

データ範囲

初期値

SHutter□（SH□）

電磁ｼｬｯﾀｰ CH 設定
1⇒ｼｬｯﾀｰ CH1 指定

1

−

1〜4
または、
ALL

−

SOlenoid□（SO□）

ｿﾚﾉｲﾄﾞ CH 設定
1⇒ｿﾚﾉｲﾄﾞ CH 1 指定
2⇒ｿﾚﾉｲﾄﾞ CH 2 指定
3⇒ｿﾚﾉｲﾄﾞ CH 3 指定
4⇒ｿﾚﾉｲﾄﾞ CH 4 指定
ALL⇒全ｿﾚﾉｲﾄﾞ指定:Bulb

ﾊﾞﾙﾌﾞﾓｰﾄﾞ設定−−

11〜9999999
ﾀｲﾏｰﾓｰﾄﾞ設定（SH□）（ｽﾋﾟｰﾄﾞ（開時間）設定）
1000 〜 9999999
単位：ms
（SO□）
−
電磁ｼｬｯﾀｰ開、ｿﾚﾉｲﾄﾞ ON

RST

開閉モード、タイマー値の初期化

O

電磁ｼｬｯﾀｰ開−−C

電磁ｼｬｯﾀｰ閉−−□

指定時間の電磁ｼｬｯﾀｰ開単位：ms

11〜9999999−

旧ｺﾏﾝﾄﾞ

:Open（:O）:Close（:C）

電磁ｼｬｯﾀｰ閉、ｿﾚﾉｲﾄﾞ OFF

−

−

−

−

45

備考

100
（SH1）
1000
（SO□）
−

開閉

:Timer□
（:T□）

RST

モード

CH 指定

＜開閉指令コマンド＞
種
コマンド
類

旧ｺﾏﾝﾄﾞﾓｰﾄﾞ選択時のみ有効

＜問合せｺﾏﾝﾄﾞ＞
種
コマンド
類

ｽﾃｰﾀｽ

:Status?

Status1?
（S1?）

旧ｺﾏﾝﾄﾞ

VersionUSB

ReaDY?
（RDY?）
INTerlock?
（INT?）

機能
指定 CH のｽﾃｰﾀｽ
0⇒ﾊﾞﾙﾌﾞﾓｰﾄﾞでｸﾛｰｽﾞ中
1⇒ﾊﾞﾙﾌﾞﾓｰﾄﾞでｵｰﾌﾟﾝ中
2⇒ﾀｲﾏｰﾓｰﾄﾞでｸﾛｰｽﾞ中
3⇒ﾀｲﾏｰﾓｰﾄﾞでｵｰﾌﾟﾝ中4⇒Interlock でｸﾛｰｽﾞ中
全 CH のｽﾃｰﾀｽ 1 要求
Bit1⇒Interlock（ｲﾝﾀｰﾛｯｸ中で 1）
Bit2⇒Ready（未ﾚﾃﾞｨｰで 1）
Bit3⇒SH1（ｵｰﾌﾟﾝで 1）
Bit4⇒SO1（ｵｰﾌﾟﾝで 1）
Bit5⇒SO2（ｵｰﾌﾟﾝで 1）
Bit6⇒SO3（ｵｰﾌﾟﾝで 1）
Bit7⇒SO4（ｵｰﾌﾟﾝで 1）
Bit8⇒未使用（常時 0）
Ready 状態要求
0⇒ﾚﾃﾞｨｰ中／1⇒未ﾚﾃﾞｨｰ
ｲﾝﾀｰﾛｯｸ状態要求
0⇒ｲﾝﾀｰﾛｯｸ解除／1⇒ｲﾝﾀｰﾛｯｸ中

備考

0〜3

0〜127

CH 指定なし

0〜1

CH 指定なし

0〜1

CH 指定なし

0〜3
USBID?

USBID 要求ｺﾏﾝﾄﾞ(DIPSW 3,4bit の値を返信)

VER?

ｺﾝﾄﾛｰﾗのﾊﾞｰｼﾞｮﾝ要求ｺﾏﾝﾄﾞ
F77-7̲0.00⇒Version 0.00
F77-7̲9.99⇒Version 9.99

Ｉ

電磁ｼｬｯﾀｰ ｽﾃｰﾀｽ要求
Bit1⇒電磁ｼｬｯﾀｰ（閉で 1）
Bit2⇒Ready（ﾚﾃﾞｨｰ状態で 1）
Bit3⇒未使用（常時 0）
Bit4⇒未使用（常時 0）
Bit5⇒未使用（常時 0）
Bit6⇒未使用（常時 0）
Bit7⇒未使用（常時 0）
Bit8⇒未使用（常時 0）＜その他コマンド＞
種
コマンド
類
WRITE

応答データ範囲

WRITE

F77-7̲0.00〜
F77-7̲9.99

旧ｺﾏﾝﾄﾞﾓｰﾄﾞ選択時のみ
有効

機能

EEPROM 書き込み

LEDON

CH 指定なし

0〜3

データ範囲

初期値

シャッター、ソレノイド
CH1 の開閉モード、およびシャッター、各ソレノイドのタイマー設定値

-

-

-

-

-

LED 消灯LED

LEDOFFCH 指定なし

LED 点灯46

備考

4.3.5 通信コマンド詳細
4.3.5.1 開閉指令コマンド（A） チャンネル指定コマンド（1） ＳＨｕｔｔｅｒ□
シャッターの開閉、開閉モードの指定、ステータスの要求を行うときにチャンネル指定コマンドを使用してチャンネル指定を行います。
コマンド
コマンド内容□ ＜データ＞
SHutter□
チャンネル指定
１（または、SH□） コマンド※ コマンドと設定データ間にスペースは不要です。
※ データ内容は、次のようになっています。
＜データ＞
内容
１
電磁シャッターCH1 を指定（2） ＳＯｌｅｎｏｉｄ□
ソレノイドの開閉（ON/OFF）、開閉モードの指定、ステータスの要求を行うときにチャンネル指定コマンドを使用してチャンネル指定を行います。
コマンド
コマンド内容□ ＜データ＞
SOlenoid□
チャンネル指定
１〜４（または、SO□） コマンド※ コマンドと設定データ間にスペースは不要です。
※ データ内容は、次のようになっています。
＜データ＞
内容
１
ソレノイド CH1 を指定
２
ソレノイド CH2 を指定
３
ソレノイド CH3 を指定
４
ソレノイド CH4 を指定
ＡＬＬ
全 CH を指定（開閉コマンドのみ有効）（B） 開閉モード指定コマンド（1） ：Ｂｕｌｂ
指定したチャンネルの開閉モードをバルブモードに指定します。
コマンド
:Bulb
（または、:B）

コマンド内容バルブモード指定□ ＜データ＞
なし

バルブモード → OPEN コマンドで開き、CLOSE コマンドで閉じる。
（例）
SHutter1:Bulb:Open →シャッター1 を開く（Close コマンドを受けるまで閉じない。
）。
（SH1:B:O）47

（2） ：Ｔｉｍｅｒ□
指定したチャンネルの開閉モードをタイマーモードに指定します。
コマンド:Timer□（または、:T□）

コマンド内容タイマーモード
指定□ ＜データ＞
11〜9999999 （SH□の場合）
1000〜9999999（SO□の場合）

タイマーモード → タイマーで指定した時間だけ開き、自動で閉じる。
（注）データがない場合は、開かない。
（例）
SHutter1:Timer123:Open →シャッター1 を 123ms 開く。
（SH1:T123:O）（C） 開閉コマンド
チャンネル指定の後に、開閉コマンドを送ることにより指定したチャンネルが開閉します。
（1） ：Ｏｐｅｎ
指定したチャンネルをオープンします（シャッター開、ソレノイド ON）。
コマンド
コマンド内容□ ＜データ＞:Open
なし
オープンコマンド（または、:O）（注）オープンコマンドには、ＣＨ指定と開閉モード指定が必要です。
（例）
Solenoid2:Bulb:Open →ソレノイド 2 を開く（Close コマンドを受けるまで閉じない。
）。
（SO2:B:O）（2） ：Ｃｌｏｓｅ
指定したチャンネルをクローズします（シャッター閉、ソレノイド OFF）。
コマンド
コマンド内容□ ＜データ＞
なし:Closeクローズコマンド（または、:C）（注）クローズコマンドには、ＣＨ指定が必要です。
（例）
Solenoid2:Close →ソレノイド 2 を閉じる。
（SO2:C）

4.3.5.2 問合せコマンド（A） ステータス要求コマンド（1） ：Ｓｔａｔｕｓ？
指定したシャッターまたはソレノイドの状態をアスキーコードで返信します。
＜応答データ＞
内容
バルブモードでクローズ中（OFF）
０
バルブモードでオープン中（ON）
１
タイマーモードでクローズ中（OFF）
２
タイマーモードでオープン中（ON）
３
インターロックでクローズ中（OFF）
４

48

（2） Ｓｔａｔｕｓ１？
全シャッター、ソレノイドの状態を１バイトバイナリコードで一括返信します。
※ このバイナリーコードは 0〜127 までの数値をとりアスキーコードに変換して出力します。
＜応答データ＞
Bit8
Bit7
Bit6
Bit5
Bit4
Bit3
Bit2
Bit1
ｲﾝﾀｰﾛｯｸ
SO4 開
SO3 開
SO2 開
SO1 開
SH1 開
準備中
１
中
ｲﾝﾀｰﾛｯｸ
０ 常時０
SO4 閉
SO3 閉
SO2 閉
SO1 閉
SH1 閉 準備完了
解除中

Ｂｉｔ １

インターロック

Ｂｉｔ ２

レディー

Ｂｉｔ ３

シャッター1

Ｂｉｔ ４

ソレノイド 1

Ｂｉｔ ５

ソレノイド 2

Ｂｉｔ ６

ソレノイド 3

Ｂｉｔ ７

ソレノイド 4

Ｂｉｔ ８

未使用

１
０
１

インターロック中（インターロックが働いている。
）
インターロック解除中（開閉可能状態）
シャッター開閉動作準備中、または開いている０

シャッター開閉動作準備完了 ※1

１
０
１
０
１

開いている
閉じている
開いている（ON）
閉じている（OFF）
開いている（ON）

０

閉じている（OFF）

１

開いている（ON）

０

閉じている（OFF）

１

開いている（ON）

０

閉じている（OFF）

１
０

常時０

※1 インターロック中の場合でも、内部的にシャッター開閉動作準備が完了している状態では、“0”が返信されます。
（3） ＲｅａＤＹ？
シャッターのレディー信号の状態をアスキーコードで返信します。
＜応答データ＞
内容
レディー中（開閉動作準備完了）
０
未レディー中（開閉動作準備未完了）
１（4） ＩＮＴｅｒｌｏｃｋ？
インターロック信号の状態をアスキーコードで返信します。
＜応答データ＞
インターロック解除中（短絡）０インターロック中（開放）
１

49

内容

（B） USB 関連コマンド（1） USBID？
USBID をアスキーコードで返信します。
＜応答データ＞
DIPSW3 OFF、DIPSW4 OFF
０
DIPSW3 ON、DIPSW4 OFF
１
DIPSW3 OFF、DIPSW4 ON
２DIPSW3 ON、DIPSW4 ON
３内容（C） バージョン要求コマンド（1） VER？
本機のファームウェアバージョンをアスキーコードで返信します。
＜応答データ＞
内容
F77-7̲0.00〜
Version 0.00〜9.99
F77-7̲9.99

4.3.5.3 ライトコマンド
本機に設定するシャッターの開閉モードおよびシャッター、ソレノイドのタイマーを内部 EEPROM に保存しま
す。EEPROM の書き換え回数は 約 100 万回と制限があるため、コマンドを受信するたびに EEPROM に保存するという方式は採用しておりません。電源 OFF 時に設定したタイマー値を保持したい場合は、電源を OFF する
前にライトコマンド（”WRITE”）を送信して下さい。ライトコマンド受信時にのみ、シャッター、ソレノイド CH1の開閉モードおよびシャッター、ソレノイドのタイマー設定値を EEPROM に書込みます。
コマンド
WRITE

コマンド内容EEPROM 書き込み

（重要）・ ライトコマンドを送らずに電源を OFF した場合、通信コマンドにより設定した開閉モード、タイ
マー値は保持されません。
・ ライトコマンドを送った後は、40ms 以上、間隔を空けてから次のコマンドを送って下さい。
・ フロントパネルから設定したシャッターの開閉モード、タイマー値は、保持されます（OPEN した場合のみ）。
4.3.5.4 その他コマンド（1） RST
電磁シャッター、ソレノイドの開閉モード、およびタイマー値を初期値に戻します。
項目
シャッター開閉モード
ソレノイド開閉モード
シャッタータイマー値
ソレノイドタイマー値

リセット後
バルブモードバルブモード
100ms
1000ms（4 チャンネル全て）※ リセットコマンドを送ると各値が初期値に戻りますが、EEPROM には書込みません。必要に応じてライトコマンドを続けて送ってください（40ms 以上間隔を空けて下さい。
）。

50

（2） LEDOFF
フロントパネルの LED を全て消灯します。光学実験などわずかな光量でも遮光したい場合などに使用します。
※ LED が消灯していても、全てのコマンドは有効です。
※ フロントパネルから LED を消灯することはできません。
※ インターロックが働くと、「----」と表示されます。インターロックを解除しても表示したままとなりま
すので、再度消灯する場合は改めて“LEDOFF”を送って下さい。
※ Control Mode を Remote から、Local に変更すると、LED が点灯します。
（3） LEDON
フロントパネルの LED を点灯します。
※ Control Mode を Remote から、Local に変更しても、LED が点灯します。

4.3.5.5 旧（F77-6）コマンド（1） Ｏ
電磁シャッターをオープンします。
コマンド
Ｏ
（例）

コマンド内容オープンコマンド（バルブモード）

O →電磁シャッターを開く（C コマンドを受けるまで閉じない。
）
。

（2） Ｃ
電磁シャッターをクローズします。
コマンド
コマンド内容Ｃ
クローズコマンド（例）
C →電磁シャッターを閉じる。
（3） １１〜９９９９９９９
電磁シャッターをオープンします。
コマンド
コマンド内容１１〜９９９９９９９
オープンコマンド（タイマーモード）（例）12345 →12345ms 電磁シャッターを開く。
（指定時間前に C コマンドを受付けるとその時点で閉じる。
）（4） Ｉ
電磁シャッターの状態を１バイトバイナリコードで返信します。
※ このバイナリーコードは０〜３までの数値をとりアスキーコードに変換して出力します。
＜応答データ＞
Ｂｉｔ
Ｂｉｔ
Ｂｉｔ
Ｂｉｔ
Ｂｉｔ
Ｂｉｔ
Ｂｉｔ
Ｂｉｔ
２
８
７
６
５
４
３
１
１
準備 OK
ｼｬｯﾀｰ閉
動作中またはｼｬｯﾀｰ開
０ 常時０
常時０
常時０
常時０
常時０
常時０
準備中

Ｂｉｔ １

シャッターステータス

Ｂｉｔ ２

レディー

１
０
１

シャッターが閉じている
シャッターが開いている
準備完了

０

動作中または準備中

51

4.3.6 エラーコード
コマンドレスポンス機能有効時、通信コマンドが不適切だった場合、以下のコードをアスキーコードで返信します。また、正常受付完了時には必ず“>”(0x3E)を返します。
（１）正常時コード
機能説明>
正常受付完了受信したコマンドを正常処理した場合、必ず上位に返信します。
但し、応答値のある要求コマンドは、“>”を返しません。
（２）エラー時
コード
機能
E00
動作（シャッター開）中
E01
E02

準備中
インターロック中

E20
E21
E22

コマンド文法エラー
デリミタ未送信エラー
設定範囲エラー

原因
電磁シャッターOPEN、または、ソレノイド ON 中に OPEN コ
マンドを送信した。
シャッター開閉動作準備中に OPEN コマンドを送信した。
インターロック機能が働いている時に:S?、S1?、INT?以外のコマンドを送信した。
コマンドの書式に間違いがあった。
デリミタ（CR）がない。または正しくない。
設定値が設定範囲を超えている。

※ コマンドレスポンス機能が無効の時は、通信コマンドが適切であっても不適切であっても、何も返信しません（問合せコマンドは除く。
）。
【DIP スイッチの設定】
ＳＷ1 の 5 ビット目を設定します。

NO
１ ２ ３ ４ ５ ６

DIPSW● コマンドレスポンスの設定
5
コマンドレスポンスの有無
OFF
レスポンス無し
ON
レスポンス有り（初期値）※ 工場出荷時の設定は ON です。

52

4.3.7 通信コマンド送受信手順
前節までの通信コマンドを使用し、PC などの上位機器と通信する場合、以下の手順を守って下さい。この手順が守られない場合、正常に動作しない可能性があります。
【送受信タイミング】

T1
T2

T3

コマンド文字数や通信方式により異なります。コマンド 1 文字当り 0.26〜0.28ms
程度かかります（RS232C 38,400bps の場合）。
本機の応答時間です。
RS232C（38.4kbps）：3〜20ms 程度（コマンド種類、文字数により異なります。
）
USB：3〜40ms 程度（コマンド種類、文字数により異なります。
）
本機が受信準備完了までに必要な時間です。必ず 20ms 以上、空けて下さい。

【コマンド送受信一例】
PC
Shutter1:Bulb:Open⇔→
3〜40ms←
20ms 以上→
3〜40ms←
20ms 以上→
3〜40ms←
20ms 以上→
3〜40ms←Solenoid1:Timer1500:Open

Status1?

Status1?

F77-7

＞

＞

14

14

※ コマンドレスポンス機能無効で返信のないコマンドを連続して送信する場合は、送信コマンドの間隔を
100ms 以上空けて下さい。

53

4.4 I/O による制御
4.4.1 シャッター操作
3.3.4 節の I/O 仕様を参照し配線を行った後、以下の要領で入力信号を受付けるとタイミングチャートに従ってシャッターの開閉を行います。
【バルブモードのタイミングチャート】
On

SH1(8番ﾋﾟﾝ) Off
1〜3ms

Open

1〜3ms

シャッター Close【タイマーモードのタイミングチャート】
3ms≦
On

SH1(8番ﾋﾟﾝ) Off
1〜3ms

設定値

Open

シャッターClose※ シャッターClose 後、1 秒以上間隔を空けてから次の信号を送ってください。
※ 50ms 以下でより高いタイマー精度（シャッタースピード）を要求される場合はタイマーモードをご使用下さい。

4.4.2 ソレノイド（光路遮断器）操作
シャッターと同様に 3.3.4 節の I/O 仕様を参照し配線を行った後、以下の要領で入力信号を受け付けるとタイミ
ングチャートに従ってソレノイドの ON/OFF を行います。
【バルブモードのタイミングチャート】
On

SO1(4番ﾋﾟﾝ) Off
1〜6ms

On

1〜6ms

ソレノイドCH1 Off【タイマーモードのタイミングチャート】
20ms≦
On

SO1(4番ﾋﾟﾝ)

Off
1〜6ms

設定値

On

ソレノイドCH1

Off※ タイマーの値は F77-7 アプリケーションまたはユーザプログラムより変更してください。
54

５．点検
運転後は、定期的に次の項目について点検することをお奨めします。異常がある時は使用を中止し、当社ＯＳＴ事業部営業グループにお問い合わせ下さい。
（点検項目）
・
接続されているケーブルに傷、ストレスや接続部にゆるみがないか。
・コントローラの通風穴に、埃がついていたり、目詰まりしていたりしていないか。
・
異臭や異音がないか。
・その他、異常はないか。

６．故障診断と処置
ご使用時、設定や接続の誤りなどでシャッターやソレノイドが正常に動作しないことがあります。正常に動作しない時には、この項をご覧になり適切な処置を行って下さい。
それでも正常に運転できない時は、当社ＯＳＴ事業部カスタマサービスまでお問い合わせ下さい。
現象予想される原因
処置
電源を入れて
電源ケーブルを本体から抜き、リアパネルのヒューズホルダ
も何も表示さ ヒューズが切れている。
に内蔵されているヒューズが切れていないか確認をして下されない。
い。
コネクタが正常に刺さっていコネクタの勘合を確認して下さい。
シャッター、ない。
ソレノイドがインターロック入力が ON になっていないか（2 ピン間が動作しない
インターロックが働いている。
開放されていないか）確認して下さい。
RS232C ケーブル違い
RS232C 通
信 が で き な Control Mode の違い
い。
ボーレートの違い

本機と PC の接続には、クロスケーブルをご使用下さい。
ストレートケーブルでは、動作しません。
Control Mode が Remote になっているか確認して下さい。
本機のボーレート設定（DIPSW）と、プログラムの設定を合わせてください。

USB 通信には、付属の USB ドライバを PC にインストール
ドライバがインストールされする必要があります。インストールの手順は、2.1.2 節を参ていない。
照して下さい。
USB 通信がControl Mode の違いできない。
ボーレートの違い
光路遮断器の 遮光板への負荷が大きい
遮光板が途中で止まってしまう。
ソレノイドの劣化
ソレノイドがON 時間が長い。
熱いControl Mode が Remote になっているか確認して下さい。
USB 通信（仮想 COM）のボーレートは、38.4kbps 固定です。プログラムの設定を合わせて下さい。
遮光板にミラーなどを貼り付けて使用することは避けて下さい。動作不良の原因となります。
ソレノイドの寿命（耐久回数）は約 2 万回です。これを超えている場合は、ソレノイドの劣化が考えられます。
ON 時間を短くするか、OFF 時の時間が長くなるように光学系を組んで下さい。

55

７．保障とアフターサービス●保証について保証期間 ： 工場出荷日から１年間
お問い合わせ時は、製品の後面または前面にあるシリアルナンバーをご連絡下さい。
但し、次の場合は保証対象外となり、有償修理とさせて頂きます。
−使用上の誤り及び当社以外の者による改造、修理に起因する故障、損傷の場合−輸送、移動時の落下等、お取扱いが不適当なために生じた故障、損傷の場合−火災、塩害、ガス害、異常電圧及び地震、雷、風水害、その他の天災地変等による故障、損傷の場合−説明書記載方法及び注意書きに反するお取扱いによって生じた故障、損傷の場合●アフターサービスについて
ご不明な点等ございましたら、当社ＯＳＴ事業部カスタマサービスまでお問い合わせ下さい。
《保証期間中》取扱説明書の注意書きに従った正常な使用状態で故障した場合には、無償で修理致します。
上記の保証対象外の故障につきましては、有償修理とさせて頂きます。
《保証期間が過ぎた場合》修理によって機能が維持できる場合は、ご要望により有償修理致します。
●修理可能期間について
本製品の補修用性能部品（機能を維持するために必要な部品）の最低保有期間は、製造打ち切り後１年です。
この期間を修理可能期間とします。尚、部品の保有期間を過ぎた後でも修理可能な場合がありますので、
当社ＯＳＴ事業部カスタマサービスまでお問い合わせ下さい。

＜改訂履歴＞
版

改定日Ver1.00

2011.7.1

改訂内容
初版

56

F77-7

アプリケーション

Ver1.00

Ver1.00

● 付録■ DIP スイッチの設定
RS232C ボーレート(2bit)、USB ID(2bit)、コマンドレスポンス(1bit)、旧コマンド有効/無効(1bit)を設定します。

NO
１ ２ ３ ４ ５ ６

DIPSW● RS232C ボーレートの設定
1
2
RS232C ボーレート
OFFOFF
4,800bps
ONOFF
9,600bps
OFFON
19,200bps
ONON
38,400bps（初期値）※USB のボーレートは、38,400bps 固定です。
● USB ID の設定
3
OFF
ON
OFF
ON

4
OFF
OFF
ON
ON

USB ID0（初期値）
1
2
3● コマンドレスポンスの設定
5
コマンドレスポンスの有無
OFF
レスポンス無し
ON
レスポンス有り（初期値）● 旧コマンドの設定
6
OFF
ON

旧コマンド有効/無効
旧コマンド無効（初期値）
旧コマンド有効※工場出荷時の設定は 1,2,5 番が ON で、他は OFF です。

57

1 6. 358
2 10 . 6

1 6. 3

5

CH1

CH1

CH2

CH3

CH4

Down
ms

T/B

T/B

6 2. 2

I/O
Local
RemoteUp

Ready

SHUTTE R

SOLENOID

Control Mode

SHUTTER CONTROLLER F77-7
15
1 70

15
SHUTTE R

SOLENOID

CH1

CH1

CH2

USB

CH3

DIPSW

4 5

CH4

In terlock

6 2. 2RS2 32C

1 2 3

F U S E( 2 5 0 V / 1 A)
5 .2 × 2 0 . 0 mm
V o l ta g e
A C1 0 0 - 2 4 0 V
W a t ta g e
m ax 4 0 W
F r e qu e n c y
5 0 /6 0 H z

I/O

6

ON

■ F77-7 外観図

■ シャッター、ソレノイド接続ケーブル（型番：F116A-CABLE）

2000±30
30
20

10

20

ｺﾈｸﾀ:HR10A-7P-4P(73)(HRS)

0.3mm2 ×2P
ｺﾈｸﾀ:DF1E-2S-2.5C(HRS)
ｺﾝﾀｸﾄ:DF1B-2428SC(HRS)

端子名
SHUT̲ASHUT̲B
N.C.
N.C.

No.
黒
1
白
2
3
4

No.
1
2

59

端子名
SHUT̲ASHUT̲B＜お問い合わせ先＞

ミスミグループ

駿河精機 株式会社 OST 事業部 カスタマサービス
TEL：0120−789−446
FAX：0120−789−449
http://www.surugaost.jp/
E-mail ost@suruga-g.co.jp

東京営業所〒１０8−００75
東京都港区港南 2−4−12 港南 YK ビル 3Ｆ
ＴＥＬ ０３−6711−5012 FAX ０３−6711−5022
関西営業所〒５６９−００７１
大阪府高槻市城北町１−５−２５ ＦＪＹビル
ＴＥＬ ０７２−６６１−３５００ ＦＡＸ ０７２−６６１−３６２２
本社〒４２４−８５６６
静岡県静岡市清水区七ツ新屋５05
ＴＥＬ ０５４−３４４−0332 ＦＡＸ ０５４-３4６−１１９６60