PROXPAC XL Proximity Transducer
Assembly
Datasheet

Bently Nevada Machinery Condition Monitoring

178554 Rev. D

Description
The design of the PROXPAC XL Proximity Transducer Assemblyis similar to our 31000/32000 Proximity Probe Housing
Assemblies. The assembly offers the same advantages andfeatures as the 31000 and 32000 housings for accessing andexternally adjusting proximity probes. However, the housing
cover of the PROXPAC XL Assembly also contains its own 3300
XL Proximitor sensor. This design makes the PROXPAC XL
Assembly a completely self-contained proximity probe
system, and eliminates the need for an extension cablebetween the probe and its associated Proximitor sensor. It
also eliminates the need for a separate Proximitor housing, as
the field wiring connects directly between the monitors and
PROXPAC XL Assemblies.
The PROXPAC XL housing is made of Polyphenylene Sulfide(PPS), which is an advanced, molded thermoplastic. This
material replaces the steel and aluminum in previous
housings offered in the Bently Nevada product line. It also
incorporates glass and conductive fibers in the PPS tostrengthen the housing and more effectively dissipate
electrostatic charges. The PROXPAC XL housing is rated forType 4X and for IP66 environments and provides extra
protection in severe environments.

PROXPAC XL Proximity Transducer Assembly
Datasheet

178554 Rev. D

Specifications

target and is from 0.25 mm
to 2.3 mm (10 mils to 90 mils)(approximately –1 Vdc to –17
Vdc).

Electrical
Proximitor
Sensor Input

3300 XL 8 mm Proximity Probewith a 1-metre cable length
installed in the probe sleeve.

Power

Requires -17.5 Vdc to -26 Vdc
without barriers at 12 mA
maximum consumption, -23
Vdc to -26 Vdc with barriers.
Operation at a more positive
voltage than -23.5 Vdc canresult in reduced linear range.

SupplySensitivity

Less than 2 mV change in
output voltage per volt change
in input voltage.

Outputresistance50 Ω

Recommended
1.27 mm (50 mils)
Gap Setting

Incremental
Scale Factor(ISF)

7.87 V/mm (200 mV/mil)±5% includinginterchangeability error
when measured inincrements of 0.25 mm (10mils) over the 80 mil linear
range from 0 °C to +45 °C(+32 °F to +113 °F) at a
maximum altitude of 2000m.

Deviation from best fit straight line (DSL)
Standard 1
metre system

Less than ±0.025mm (±1 mil)with components at 0 °C to+45 °C (+32 °F to +113 °F).

Frequency
Response

0 to 10 kHz: +0, -3 dB, with up
to 305 metres (1000 feet) of
field wiring.

Probe Length R
PROBE (Ω)(m)

Minimum
Target Size

15.2 mm (0.6 in) diameter(flat target)

1.0

Shaft Diameter

Nominal Probe DC Resistance (RPROBE)
Resistance from the Center Conductor to theOuter Conductor

7.59 ± 0.50

Minimum

Field wiring

0.2 to 1.5 mm2 (16 to 24 AWG)
. Recommend using
3-conductor shielded triax
cable and tinned field wiring.
Maximum length of 305
metres (1,000 feet) between
the 3300 XL Proximitor
Sensor and the monitor. See
the frequency responsegraph (Figure 1, page 7) for
signal rolloff at highfrequencies when using
longer field wiring lengths.

Linear Range

2 mm (80 mils). Linear range
begins at approximately
0.25 mm (10 mils) from50.8 mm (2.0 in)
76.2 mm (3.0 in)

When gapped at the centerof the linear range, theinteraction between two
separate transducer
systems (cross-talk) will beless than 50 mV on shaft
Recommended diameters of at least 50 mm(2.0 in) or greater. Careminimumshould be taken to maintainminimum separation oftransducer tips, generally atleast 40 mm (1.6 in) for axial
position measurements or
38 mm (1.5 in) for radialvibration measurements tolimit cross-talk to 50 mV or

2/15

PROXPAC XL Proximity Transducer Assembly
Datasheet

178554 Rev. D

containing conductive fibers.

less. Radial vibration or
position measurements on
shaft diameters smaller
than 76.2 mm (3.0 in) will
generally result in a change
in scale factor. Consult
Performance Specification
159484 for additional
information.
Effects of 60 Hz Magnetic Files Up To 300
Gauss
Output Voltage in Mil pp/Gauss
Gap (mils)

1-metreProximitor
SensorProbe

100.01240.000450

0.01100.001490

0.01260.0045

Electrical Classification:
Complies with the European CE mark.

Mechanical
Housing
Ratings

IP66 rating verified by BASEEFA
report number T07/0709.

Probe Tip
Material

Polyphenylene Sulfide (PPS)

Probe Case
Material

AISI 304 stainless steel

Probe
Cable

1-metre length, 75 Ω triaxial,
fluoroethylene propylene (FEP)
insulated.

Probe
Connector

Gold-plated brass ClickLoc
connector.

Probe
TensileStrength

330 N (75 lb) between probe
cable and case, maximum.

Housing
Material

Ultraviolet (UV) resistant, glassreinforced polyphenylene
sulfide (PPS) thermoplastic

SleeveMaterial
andRetaining
Chain

AISI 304 stainless steel

Outer
Sleeve andRetaining
ScrewsAISI 303 stainless steel

Sleeve ORing
Material

Neoprene

GroundingLiner andRetaining
Plate
Material

AISI 304 Stainless Steel

Vibration
Isolation
Material

Extra Soft Cellular Silicone

Lid Label
Material

Gloss Radiant White Polyester

Recommended Torque

3/15

RetainingNut

29.5 N·m (260 in·lb)

Probe
SleeveLocknut

39.3 N·m (350 in·lb)

Housing
Strength
TypicalOuter sleeve was mounted on atest stand with its axis parallel tohorizontal and the housing
mounted on the outer sleevethrough an end hole. The
housing supported 912 N (205lbf) placed approximately 38
mm (1.5 in) from theunsupported end with the coverfastened in place andgrounding liner installed.

Housing
ImpactStrength

Certified by BASEEFA towithstand two separate 4 JoulePROXPAC XL Proximity Transducer Assembly
Datasheet

178554 Rev. D

strength of the plastic
housing. Longer contact with
hot water or steam may
weaken the plastic housing
during the first 6 to 8 weeks of
exposure and ultimately
reduce the housing strength
to approximately half of theinitial value. Tests of actual
housing performance after
contact with hot water andcondensing steam have notbeen conducted.

(5.4 ft·lb) impacts at -39°C (38°F) and at 115°C (239°F). CSA
verified that samples of the
housing and cover could
withstand a 7 Joule (9.5 ft·lb)
impact at ambient room
temperature.
Total
SystemWeight1.44 kg (3.2 lbm) typical with 0.3
metre (12 in) sleeve length.

Environmental Limits
Probe Temperature Range-52°C to +177°C(-62°F to +351°F)
OperatingTemperature

Exposing the probe
to temperatures
below –34°C(-29°F) may causepremature failure of
the pressure seal.

Probe
Pressure

Probe Housing and Proximitor Sensor
OperatingTemperature-52°C to +100°C(-62°F to +212°F)

Proxpac
Storage
Temperature-52°C to +105°C(-62°F to +221°F)

Relative
Humidity(PROXPAC XL
Sensor andprobe)

100% condensing, nonsubmersible when connectors
are protected. When properly
sealed, moisture should notenter the housing. Users
should take precautions toprevent moisture fromtraveling through the conduit
into the housing.

Hot Waterand Steam
ExposureEffects(Specification notguaranteed) Brief periods (upto one week) of contact with
hot water 95 °C (203 °F) and/or
condensing steam should not
significantly affect the

The PROXPAC XL design seals
differential pressure between
the probe tip and the housing
main body when used with a
3300 XL 8 mm probe. The
sealing material inside theprobe case consists of a Viton
O-ring, whereas the O-ring
between the sleeve and the
housing is Neoprene The
plastic housing design iscertified to seal against hosedirected water according toType 4X and IP66 standards
but does not resist internal orexternal pressure. Probes are
not pressure tested prior to
shipment.
Contact our custom design
department if you require atest of the pressure seal foryour application.

It is the responsibility of the customer
or user to ensure that their installatin
will contain and safely control all
liquids and gases should the
PROXPAC XL transducer leak.
Solutions with high or low pH values
may erode the tip assembly of theprobe, causing media to leak into
surrounding areas. Bently Nevada
LLC does not be held responsible forany damages resulting from leaking
Proximity Probe Housing Assemblies.

4/15

PROXPAC XL Proximity Transducer Assembly
Datasheet

178554 Rev. D

In addition, PROXPAC XL transducers
does not be replaced under theservice plan due to probe leakage.

Compliance andCertifications
FCC
This device complies with part 15 of the
FCC Rules. Operation is subject to thefollowing two conditions:llThis device may not cause harmfulinterference.
This device must accept any
interference received, includinginterference that may cause
undesired operation.

EMC
EN 61000-6-2
EN 61000-6-4
EMC Directive 2014/30/EU

RoHS
RoHS Directive 2011/65/EU

Maritime
ABS 2009 Steel Vessels Rules
1-1-4/7.7,4-8-3/1.11.1,4-9-7/13

5/15

PROXPAC XL Proximity Transducer Assembly
Datasheet

178554 Rev. D
140979.

Hazardous Area Approvals(see Temperature Schedule table tofollow)

ATEX/IECEx

For the detailed listing of country andproduct specific approvals, refer tothe Approvals Quick Reference Guide(108M1756) available from Bently.com.

3300 XL Proximitor Sensoria
II 1 GEx ia IIC T4/T5 Ga
Ex ia IIIC T90C/T105C Dc
For EPL Dc:
T105C @ Ta = -55°C to 100°C
T90C @ Ta = -55°C to +85°C

CSA/NRTL/C
3300 XL Proximitor Sensor
ia
When installedwith
intrinsicallysafe zener
barriers per
drawing 141092
or wheninstalled with
galvanic
isolators.

nA, ecWhen installed
without
barriers per
drawing
140979.

Class I, Zone 0: AEx/Ex ia IIC T4/T5 Ga;
Class I, Groups A, B, C, and D,Class II, Groups E, F and G,Class III;
T5 @ Ta= -55°˚C to + 40°˚C.
T4 @ Ta= -55°˚C to + 80°C.

nA, ecWhen installed
without
barriers per
drawing

Ii= 140mA

Io= 140mA

Pi= 0.91W

Po= 0.742W

Ci- 47nF

Co= 1.5nF

Li= 1460µH

Lo= 610µH

II 3 GEx nA IIC T4/T5 Gc
Ex ec llC T4/T5 Gc

Class I, Zone 2: AEx/Ex nA IIC T4/T5
Gc;
Class I, Division 2, Groups A, B, C, andD;Class I, Zone 2: AEx/Ex ec llC T4/T5
Gc;
Class I, Division 2, Groups A, B, C, andD;

Ui= -28V
li= 140 mA
T5 @ Ta= -55˚ C to + 40˚C
T4 @ Ta= -55 ˚C to + 80˚C

3300 XL Probe
Probe entity parameters are met
when used with BN extension cables
and connected to BN Prox.

3300 XL Probe

When installedwith
intrinsicallysafe zener
barriers per
drawing 141092
or wheninstalled with
galvanic
isolators.

Uo= -28V

nA,ec

T5 @ Ta= -55°C to + 40°C
T4 @ Ta= -55°C to + 80°C

ia

Ui= -28V

Class I, Zone 0: AEx/Ex ia IIC T5...T1 Ga;
Class 1, Groups A, B. C, and D,Class II, Groups E, F, and G,Class III;ia
II 1 GEx ia IIC T5…T1 Ga,(see Temperature Schedule
table to follow)(see Temperature Schedule table tofollow)

Ex ia IIIC T90°C ... T280°C Dc
For EPL Dc:

Class I, Zone 2: AEx/Ex nA IIC T5…T1 Gc;
Class 1, Division 2, Groups A, B, C, D;
Class 1, Zone 2: AEx/Ex ec llC T5…T1
Gc; Class I, Division 2, Groups A, B, C,
and D;

Ui= -28V

Ci = 1.5 nF

Ii = 140 mA

Li =610 μH

Pi = 0.91 W

6/15

PROXPAC XL Proximity Transducer Assembly
Datasheet

178554 Rev. D

nA,ecII 3 GEx nA IIC T5…T1 Gc,
Ex ec llC T5…T1 Gc,(see Temperature Schedule
table to follow)
Ui= -28V

Hazardous Area Conditions ofSafe Use
CSA/NRTL/C:

li= 140 mAia

Temperature Schedule
Temperature
Classification

Install per Bently Nevada drawing 141092.
nA, ecAmbient Temperature(Probe Only)

Install per Bently Nevada drawing 140979.

For EPL Ga and Gc
T1-55°C to +232°C

T2-55°C to +177°C

T3-55°C to +120°C

T4-55°C to +80°C

T5-55°C to +40°C

ATEX/IECEx:
ia
Install per Bently Nevada drawing 141092.
nA, ecThe Prox must be installed so as to provide theterminals with a degree of protection of at least
IP54.

For EPL Dc
T280°C @ Ta

-55°C to +232°C

T225°C @ Ta

-55°C to +177°C

T170°C @ Ta

-55C to +120°C

T130°C @ Ta

-55°C to +80°C

T105°C @ Ta

-55°C to +100°C

T90°C @ Ta

-55°C to +40°C

7/15

PROXPAC XL Proximity Transducer Assembly
Datasheet

178554 Rev. D

Ordering Information

probe side view and/or rear view
effects.

For the detailed listing of country andproduct specific approvals, refer tothe Approvals Quick Reference Guide(108M1756) available from Bently.com.

Order in increments of 0.1 in (2 mm).
Minimum length: 1.0 in (25 mm)
Maximum length: 30 in (762 mm )ExamplesOrder -00 or –01 (for multi approvals)
for the A option and –00 or -000 for allother options to receive just a spare
housing with Proximitor Sensor.

PROXPAC XL Proximity Transducer,
English000

No probe sleeve037

3.7 in (94 mm)

224

22.4 in (569 mm)

D: Fittings Option
For 1/2-14 NPT fittings, order
option -03 or spare 2665001 reducers for either
option -01 or -02.

330880-AA-BB-CCC-DD-EE
A: Probe and Approvals Option
00

No probe; Proximitor sensor without
approvals

00

01

No probe; Proximitor sensor with
Multiple Approvals

No fittings; two plugs and two
washers01

One 3/4-14 NPT fitting, two plugs

16

3300 XL 8 mm probe without
approvals

02

Two 3/4-14 NPT fittings, one plug

3300 XL 8 mm probe with Multiple
Approvals

03

One 3/4-14 NPT fitting, one 3/4-14 NPT
to 1/2-14 NPT SST reducer and two
plugs

28

E: Mounting Thread Option

B: Standoff Adapter Option (B Dimension)
Order in increments of 0.5 in (13 mm).
Minimum length: 1.5 in (38 mm)
Maximum length: 7.5 in (191 mm )Examples00

No standoff adapter

15

1.5 in (38 mm)

C: Probe Penetration Option (C Dimension)
For penetration lengths between 1.0
and 2.0 inches, counterbore may berequired in machine case to reduce

8/15

00

No outer sleeve assembly

02

3/4-14 NPT (Required if ordering
Standoff Adapter Option.)05

7/8-14 UNF-2A

PROXPAC XL Proximity Transducer Assembly
Datasheet

178554 Rev. D

PROXPAC XL Proximity Transducer,
Metric760

D: Fittings Option (supplied as a kit)

330881-AA-BB-CCC-DD-EE
A: Probe and Approvals Option
00

No probe; Proximitor sensor
without approvals

01

No probe; Proximitor sensor with
Multiple Approvals

16

3300 XL 8 mm probe without
approvals

28

760 mm

3300 XL 8 mm probe with
Multiple Approvals

B: Standoff Adapter Option (B Dimension)00

No fittings; two plugs and two
washers

01

One M25 fitting, two plugs

02

Two M25 fittings, one plug

03

One M20 fitting, two plugs

05

One PG21 to PG11 reducer, two
plugs

06

One 3/4-14 NPT fitting, one 3/4-14
NPT to 1/2-14 NPT SST reducer
and two plugs

07

One PG21 x M20 fitting, two plugs
Two PG21 x M20 fittings, one plug

Order in increments of 10 mm.

Conduit fittings arenecessary whenhardline conduit or
metal piping isbrought into the
housing. If using
flexible conduit, you
should order withintegral 3/4-14 NPT
fittings so that you do
not require additional
conduit fittings withthe housing. If using
flexible conduit, orderthe D = 0 0 option.

Minimum length: 40 mm
Maximum length: 200 mm
Examples00

No standoff adapter

04

40 mm

20

200 mm08

C: Probe Penetration Option (C Dimension)
For penetration lengths
between 25 and 50 mm,
counter bore may be required inmachine case to reduce probeside view and/or rear view
effects.

E: Mounting Thread Option

Order in increments of 1 mm.
Minimum length: 25 mm
Maximum length: 760 mm
Examples000

No probe sleeve

050

50 mm

9/15

00

No outer sleeve assembly

01

M24 X 302

3/4-14 NPT (required if ordering
Standoff Adapter Option)

PROXPAC XL Proximity Transducer Assembly
Datasheet

178554 Rev. D

English Probe Sleeve (Spare)

Accessories
159484

Performance Specification 3300 XL Proximity Transducer
System178761

Operation manual

178850-00

Upgrade Kit, Non-approved(Includes housing lid, 1-metre
Proximitor sensor, and support
table)

178850-05

Upgrade Kit, Multi-Approvals(Includes housing lid, 1-metre
Proximitor sensor, and support
table)

178644-01

Proximitor sensor support table

330180-1200

Spare 3300 XL Proximitor sensor,
non-approved

330180-1205

Spare 3300 XL Proximitor sensor,
Multiple Approvals

108883 –AAA
This is the measured probe sleeve length.
Order in increments of 0.1 in (3 mm). Note thatthe individual probe sleeve length does notinclude the distance from the end of thesleeve to the probe tip or the gap from theprobe tip to the target material. If only the partnumber of the original housing is known andthe sleeve cannot be measured, use thefollowing formula to determine the sleeve
length:
A: Standoff Adapter Option from original
housing (330800 option B) + Probe
penetration option from original housing(330800 option C) + 025.
Example: Original part number is 330800-1615-035-03-02. A option for replacement
sleeve is (015 + 035 + 025) = 0 7 5.

330105-02- Spare 3300 XL 8 mm probe,
12-10-02-00 English, non-approved

Minimum Probe Sleeve Length:
3.5 in (89 mm)= 0 3 5
Maximum Probe Sleeve Length:32.5 in (826 mm) = 3 2 5330105-02- Spare 3300 XL 8 mm probe,
12-10-02-05 English, approved

Metric Probe Sleeve (Spare)

330106-0530-10-0200

Spare 3300 XL 8 mm probe,
metric, non-approved

330106-0530-10-0205

Spare 3300 XL 8 mm probe,
metric, approved

02120015

Bulk Field Wire
1.0 mm² (18 AWG), 3-conductor,
twisted shielded cable with drain
wire. Specify length in feet.

37948-01

Probe Support / Oil Sleeve
Provides seal along probe
sleeve. May be used as a probe
sleeve support in certain
installations.

108882 –AAA
This is the measured probe sleeve length.
Order in increments of 1 mm. Note that theindividual probe sleeve length does notinclude the distance from the end of thesleeve to the probe tip or the gap from theprobe tip to the target material. If only thepart number of the original housing is known
and the sleeve cannot be measured, use thefollowing formula to determine the sleeve
length:AStandoff Adapter Option from original
housing (330801 option B) * 10 + Probe
penetration option from original housing(330801 option C) + 063.
Example: Original part number is 330801-1608-205-03-02. A option for replacement
sleeve is (080 + 205 + 063) = 3 4 8.

10/15

PROXPAC XL Proximity Transducer Assembly
Datasheet

178554 Rev. D

Heavy Duty Cable Fittings

Minimum Probe Sleeve Length:
88 mm (3.5 in) = 0 8 8
Maximum Probe Sleeve Length:
823 mm (32.4 in) = 8 2 3English Standoff Adapter (Spare)Hex = 1-3/8 in; threads = 3/4-14 NPT
109319 –AAAOrder in increments of 0.5 in (13 mm).
Minimum length: 1.5 in (38 mm)
Maximum length: 7.5 in (191mm)

03813103

Chrome-plated zinc conduit
fitting, 3/4-14 NPT

03818100AISI 316 stainless steel conduit
fitting, 3/4-14 NPT

03818101AISI 316 stainless steel conduit
fitting, PG21 x M25

03818102AISI 316 stainless steel conduit
fitting, PG21 x M20

03818111

Nickel-plated brass conduit
fitting, PG21 x M20

26650-01

AISI 303 stainless steel reducer
3/4-14 NPT to 1/2-14 NPT

Example: 0 2 0 = 2 in (51 mm)

Sealtite Flexible Conduit

Metric Standoff Adapter (Spare)
Wrench flats = 35 mm; threads = 3/4-14 NPT.
109318 –AAA

Maximum length: 99 ft (30.2 m)

Example: 0 5 = 50 mm

Example: 0 5 = 5 ft (1.5 m)

English sleeve plug threaded,
303 stainless steel.
Metric Sleeve Plug Threaded, 303
stainless steel.
Plugs fill opening when sleeve isremoved from machine case.
English Blanking Plug
Metric Blanking Plug.

104288-02

3/4-14 NPT assembly

Minimum length: 1 ft (0.3 m).

Maximum length: 200 mm

104288-01

14848-AAAOrder in increments of 1 ft (0.3 m).

Minimum length: 40 mm

104968-02

1/2-14 NPT assembly

A: Length Option

Order in increments of 10 mm.

104968-01

14847-AAA

Blanking plugs are included with
the Fittings Option "D". Spareplugs fill conduit holes in plastic
housing where needed.

11/15

PROXPAC XL Proximity Transducer Assembly
Datasheet

178554 Rev. D

Graphs and Figures
Note: All dimensions in millimeters (inches) except as noted.

Figure 1: Frequency Response with Different Field Wiring Lengths without Barriers (1 m System)

12/15

PROXPAC XL Proximity Transducer Assembly
Datasheet

178554 Rev. D

Figure 2: Dimensions for PROXPAC XL Proximity Transducer Assembly
1. Depth = 92.7 (3.65)
2. Fitting “D”
3. PROXPAC XL Housing
4. 42 (1.75) hexagonal
5. Dimension “B”
6. Standoff adaptor (optional)
7. 3/4-14 NPT
8. Dimension “C”
9. 12.7 (0.5) diameter stainless steel sleeve
10. Probe
11. Target surface

13/15

PROXPAC XL Proximity Transducer Assembly
Datasheet

178554 Rev. D

Figure informationlll"B" plus "C" dimensions greater than 305 mm (12.0 in) require additional sleeve support near theprobe to stiffen the assembly and avoid the influence of resonance.
For desired probe penetration lengths of less than 51 mm (2.0 in), order a separate IndividualStandoff Adapter. The effective probe penetration length will then be reduced by the length of the
Individual Standoff Adapter, plus an additional 13 mm (0.5 in) due to the NPT thread engagement.
o Example:The customer desires a probe penetration length of 25 mm (1.0 in). To do this,they order a 330800 housing with CXXX (probe penetration) option of 0 3 0 [76 mm (3 in)]and a separate individual standoff adapter that is 38 mm (1.5 in) in length (part number
109319-015). The standoff adapter would cover 38 mm (1.5 in) of the probe sleeve, plus anadditional 13 mm (0.5 in). Therefore, the effective probe penetration length would drop to 25
mm (1.0 in).
"B" plus "C" dimension represents mid-setting distance between mounting surface and target
surface. Threaded sleeve allows ±12.7 mm (0.5 in) adjustment from this point. "B" plus "C"
dimension is 760 mm (30 in) maximum.

14/15

PROXPAC XL Proximity Transducer Assembly
Datasheet

178554 Rev. D

Copyright 2021 Baker Hughes Company. All rights reserved.

Bently Nevada, Proximitor, Proxpac and Orbit Logo are registered trademarks of Bently Nevada, a Baker Hughes
Business, in the United States and other countries. The Baker Hughes logo is a trademark of Baker Hughes Company.
All other product and company names are trademarks of their respective holders. Use of the trademarks does not
imply any affiliation with or endorsement by the respective holders.
Baker Hughes provides this information on an “as is” basis for general information purposes. Baker Hughes does notmake any representation as to the accuracy or completeness of the information and makes no warranties of any
kind, specific, implied or oral, to the fullest extent permissible by law, including those of merchantability and fitness
for a particular purpose or use. Baker Hughes hereby disclaims any and all liability for any direct, indirect,
consequential or special damages, claims for lost profits, or third party claims arising from the use of theinformation, whether a claim is asserted in contract, tort, or otherwise. Baker Hughes reserves the right to makechanges in specifications and features shown herein, or discontinue the product described at any time without
notice or obligation. Contact your Baker Hughes representative for the most current information.
The information contained in this document is the property of Baker Hughes and its affiliates; and is subject tochange without prior notice. It is being supplied as a service to our customers and may not be altered or its content
repackaged without the express written consent of Baker Hughes. This product or associated products may becovered by one or more patents. See Bently.com/legal.

1631 Bently Parkway South, Minden, Nevada USA 89423
Phone: 1.775.782.3611 (US) or Bently.com/support
Bently.com

15/15