Manual Stage

Manual Rotary Stage Guidance
Can be positioned a sample rotation from coarse to fine adjustment.
Available angle scales on the side for repeatility positioning.
(Square type excluded.)

Features
Fitting type B43մ P.2-171֐
Fitting stage allows coarse 360 degree rotation and fine-control (micrometer head).
Can be widely used for R&D and integration in devices.
Stage size

Į24mm

Į38mm

Į60mm

Į85mm

Į110mmCross roller bearing type B44/BS43մ P.2-173֐
Stages that use crossed roller bearing allows coarse 360 degree rotation and fine-control(micrometerhead).
The rigidity is higher than a fitting type.
Selectable stages made of aluminum or stainless steel.
Stage size

Į60mmTransmission hole type B47մ P.2-173
Stage that use crossed roller bearing allows coarse 360 degree rotation and fine-control (micrometerhead). There is a transmission hole in the center of the stage for passing laser beam and organizationof the wires.
Stage size

Į100mmSquare type BREմ P.2-169֐
A square rotary stage that can be readity conbined with a square linear stage and goniometer.

Stage size2167

40×40mm

60×60mm

For use correctlyӨHow to mountӳPlate type…………Fix by supplied screw to the 4 places hole on the lower plate.
ӳNon-plate type…… Move the bolt hole on the top surface roughly to align it with bolt holes on the lower surface of the stage.

Plate type

Non-plate type

ӨAbout object on the upper or lower stage.
Stage surface might be deformed and mounting unflat object and set to the unflat place can affect tobe deformed stage surface and decreasing accuracy.

ӨPosition of stage mounting
All products SPEC shows must be shown flat setting condition.
Pay attention to mount such as up side down, vertical on the sideand horizontal on the side.
Load capacity and accuracy might be changed by the posioning.
Please feel free to ask us for more information.

Ceiling-hung, mounted
vertically use on the side٬Posture characteristic list for each products
Travel guide

Inverted and reversed

Side horizontal

vertically use on the side

Fitting
Cross roller
Ball bearingӤ
ӽ×

Ӥ
Ӥ×

Ӥ
Ӥ
×

Ӱ:Available under limit of load or moment
Ӥ:Accuracy might be decreased under limit of load or moment
×:Not available

How to align rotating center axis
The stage deliver their inherent performance by aligning the center axis of another device or work to be mounted
as much as possible.

Center of rotation

We recommend that you align the center axis using the method shown below.
٬Determine the position in which the center deviation becomes the smallest using the dial gauge byrotating the rotary stage.
٬Fix the stage or work. The center of axis can be fine-adjusted easily by combining XY stages.
˥No mounting reference surface on stage main body.

2168

Manual Stage

Rotary Stage BRE Series ӗ40ӗ60: BRE04020/BRE06020
Manual linear stage
BRE04020

XBRE06020

XYZ
Horizontal
ZXZ
Horizontal
XZ

1 ModelBRE04020

XYZ

1Horizontal
XYZ

2Goniometer
Rotary1 Stage table size

2 Travel distance0406020

40mm
60mm

20°

Unit
Accessories

LinearBall

CrossRoller

Dovetail

□25
□30
□40
□50
□60
□70
□80
□100
□120
Other

2169

Model
Stage table size
Travel distance
Minimum reading of micrometer
Travel guide
Load capacity
Parallelism
WeightMain materialSurface finishing
Provided screws (Hex socket screws)

SPEC
BRE04020
40×40mm±10°
͚1´51ˢFitting method
1kgf [9.8N]
50μm
0.14kgAluminum−Black alumite processing3 of M36BRE06020
60×60mm±10°
͚1´12ˢFitting method
3kgf [29.4N]
50μm
0.26kgAluminum−Black alumite processing
3 of M48CAD
3D・2D

Dimensional outline drawings

4038.87165φ6Clamp

18.51640

32160

φ1316

4-M3 6 DP

Manual linear stage

BRE04020

X

Center of rotation

2.5

32

97.520XY
Z

3-3.5THRU(For M3 bolt holes)

Horizontal
Z
XZ

BRE06020
Horizontal
XZ
60

36.85

Horizontal
XYZ
Goniometer

25

φ6602528.50Clamp

RotaryUnit

Center of rotation

97.5

20

Accessories350XYZ

25

φ1325

74-M4 6 DP

50

3-4.5THRU(For M4 bolt holes)

LinearBall

CrossRoller

Dovetail

□25
□30
□40
□50
□60
□70
□80
□100
□120
Other2170

Manual Stage

Rotary Stage (Fitting Type) Į24֐110: B43 Series
Manual rotation stage

B43-25

B43-38N

B43-85N

B43-110N

B43-60N

X
XY
Z
Horizontal
Z
XZ
ӖCross roller bearing stage that is available a rough motion 360
deg. rotation and micromotion (micrometer head).
մLow price.
Ideal for use in R&D, integrating device and much more.

Horizontal
XZ
XYZ

٬Square rotation type (BRE series)

Horizontal
XYZ
Goniometer

P.2-169֐

RotaryUnitAccessories
How to use a rotary stage
Micromotion positioning after roughpositioning
ΖA rough adjustment to the target angle withfeeding knob
①Coarse motion 360°
ΗSqueeze a rough clamp and fix.
ΘA micromotion adjustment to the target
angle with micrometer.
ΙSqueeze a micromotion clamp and fix.

Fitting
Type④Micromotion clamp②Rough clamp③Fine motion ±5°

CrossRoller
SPECModel(Opposite hand)Stage table size
Travel distance
Vernier minimum reading
Minimum reading capability
Guide
Load capacity
Allowable load for moment
Moment rigidity
Parallelism
Eccentricity amount
Runout amount
WeightMain materialSurface finishing
Provided screws (Hex socket screws)

φ 24
φ 38
φ 60
φ 85
φ 100
φ 110

2171

B43-25
B43-25R
Į24mm
Coarse motion 360°մFine motion ±3°
Vernier scale 0.5°
͚1.50°/Rotation

B43-38N
B43-38NR
Į38mm

B43-60N
B43-85N
B43-60NR
B43-85NR
Į60mm
Į85mm
Coarse motion 360°մFine motion ±5°

Vernier scale0.2°͚1ˡ26ˢ/Scale͚55ˢ/Scale
Fitting method
1.0kgf [9.8N]
3.0kgf [29.4N]
0.12N٬m
0.3N٬m0.7N٬m8.11ˢ/N٬cm
3.56ˢ/N٬cm
0.41ˢ/N٬cm
50μm
20μm
50μm
50μm
20μm
20μm
0.03kg0.09kg0.28kgAluminum−Black alumite processing
4 of M264 of M38
4 of M410B43-110N
B43-110NR
Į110mmVernier scale0.1°
͚43ˢ/Scale͚34ˢ/Scale
4.0kgf [39.2N]
1.2N٬m
0.22ˢ/N٬cm

5.0kgf [49.0N]
1.5N٬m0.17ˢ/N٬cm

0.48kg0.75kg4 of M48

4 of M48

CAD
3D・2D

Dimensional outline drawings

B43-38N
8.7

2416207 7Į616

38.5±2.5

4-M2.6 5 DP Į4

Į9.3
24

15'FFEJOHTDSFXFeed Knob60

B43-85N

15M16P=1 6 DP
Transmission hole dia.Į148-M3 6 DP

37.5
Į1225
16XY

1116
ӗ32

4-3.5THRUĮ6C'BORE(For M3 bolt holes)

Micromotion clamp

Z

With vernier scale(0.2°)

Horizontal
Z
XZ

48
2525

Į12

Micromotion clamp

3225

47.8±3.2

Hole for bolt Į8.5

Horizontal
XYZ

15

Į15
37.5

8-M4 6 DP

63±6.5Į415

45±3.2

25

25
16
4-M4 6 DP

RotaryĮ4Į13Feed Knob

Į6

Micrometer head48
Į85

Rough clamp

With vernier scale(0.1°)
28
Rough clamp

With vernier scale(0.2°)1344-4.5THRUĮ8C'BORE(For M4 bolt holes)

20

11.5

12.5

22.5

2525
22.5
1315
ӗ50

Unit
Accessories

Į10Į604.5

Goniometer

Micrometer head

Feed Knob

4-4.5THRUĮ8C'BORE(For M4 bolt holes)

Horizontal
XZ
XYZ

60

2516

52.3±6.5

4-M3 6 DP

322516

M16P=1 6 DP
Transmission hole dia.Į144.5

13

B43-60N

17.520
ӗ


Į
5)36Į$#03&'PS.CPMUIPMFTXĮ38

Rough clamp

3PVHIDMBNQ

Į6Į

8JUIWFSOJFSTDBMF n

Micrometer head

10.5

.%1

Micromotion clamp

7 71640162012-2 4 DP Į5)36Į)%140

11.5
M6P=1 6 DP
Transmission hole dia.Į337.4±2.5
 .JDSPNPUJPODMBNQManual rotation stage

B43-25

ӗ50FittingType

B43-110N

61.5

M16P=1 6 DP
Transmission hole dia.Į1470
25

4-M3 6 DP

Į1225
32

Micromotion clamp
52.3±6.5

32

CrossRoller

63±6.5257025

Hole for bolt Į8.5

12-M4 6 DP
Į415

Feed Knob

Į1361.5

With vernier scale(0.1°)
29.5
Rough clamp(For M4 bolt holes)

25
ӗ50

11.5

4-4.5THRUĮ8C'BORE

φ24φ38
φ60

13425

22.5

Į10Į110Micrometer head

φ85
φ100
φ110

2172

Manual Stage

Rotary Stage (Cross Roller Bearing Type) Į85٬100: B44/B47 Series
Manual Rotation Stage

B44-85N

B47-100AN
ӖA v a i l a b l e c o a r s e m o t i o n 3 6 0 d e g r e e a n d f i n emotion(micrometer head).
Rigidity is better than fitting type.

Transmission hole type٬Transmission hole XY-axis cross roller guide
P.2-077֐
մB27 series

XCAD
3D・2D

Dimensional outline drawings

XY

B44-85N

Į85

With vernier scale(0.1°)
28
Rough clamp

Į6Į10

Rough clamp

13

2.5

20
ӗ50

11.5

4-4.5THRUĮ8C'BORE(For M4 bolt holes)

4-4.5THRUĮ8C'BORE(For M4 bolt holes)

UnitAccessories

How to use a rotary stage①Coarse motion 360°
Micromotion positioning after roughpositioning
ΖA rough adjustment to the target angle withfeeding knob
ΗSqueeze a rough clamp and fix.
ΘA micromotion adjustment to the target
angle with micrometer.
ΙSqueeze a micromotion clamp and fix.

FittingType③Fine motion ±5°④Micromotion clamp②Rough clamp

Cross
RollerModel(Opposite hand)Stage table size
Travel distance
Vernier minimum reading
Minimum reading of micrometer
GuideLoad capacity
Allowable load for moment
Moment rigidity
Parallelism
Eccentricity amount
Runout amount
WeightMain materialSurface finishing
Provided screws (Hex socket screws)

φ 24
φ 38
φ 60
φ 85
φ 100
φ 110

2173

52.8±6.5
63.5±6.5

Transmission hole dia Į50

Į13Į4

Micrometer head

65

Feed Knob

25
22.5
3.5

Rotary5045

Micrometer head

Į12
Micromotion clamp

Į100

SPEC
B44-85N
B47-100AN
B44-85NR
B47-100ANR
Į85mm
Į100mmCoarse motion 360°մFine motion ±5°
Vernier scale0.1°
͚43ˢ/Scale͚32ˢ/Scale
Cross roller bearing
6.0kgf [58.8N]
5.0N٬m0.36ˢ/N٬cm
0.13ˢ/N٬cm
50μm
50μm
20μm
0.43kg0.45kgAluminum−Black alumite processing
4 of M484 of M46With vernier scale(0.1°)

9

Į1348

Horizontal
XYZ
Goniometer

4563±6.5
Į450

15Feed Knob

65
45

20
17.5
10XYZ

8-M4 6 DP

15

Horizontal
XZ

21.510045

4-M4 6 DP

25XZ

Micromotion clamp

100

2532

Horizontal
Z

254825 Į1232

52.3±6.5Z

10.5
4-M4 5 DP

B47-100AN

M16P=1 6 DP
Transmission hole dia.Į124-M3 6 DP
Hole for bolt Į8.520
ӗ90

Rotary Stage (Cross Roller Bearing Type (Stainless Type)) Į60: BS43 Series
Manual rotation stage

BS43-60

ӖAvailable coarse motion 360degree and fine motion(micrometer head).
High rigidity of materials because of stainless made.
CAD
3D・2DX
XY

19.5

60

1525
25
16 16

8-M3 6 DP

Z4-M4 6 DP Į4Feed Knob

HorizontalZ
XZ

44±3.2

60
2525
16 16

M16P=1 6 DP
Transmission hole dia.Į1235BS43-60

Micrometer head

Į1537.5

15

XYZ

Micromotion clamp
Rough clamp

With vernier scale(0.2°)
Į10Horizontal
XZ

Į60

Horizontal
XYZ

4-4.5THRUĮ8C'BORE(For M4 bolt holes)

15

12.5

4.5

25
22.5
13

Goniometer

ӗ50

RotaryUnitAccessories

How to use a rotary stage
Micromotion positioning after roughpositioning
ΖA rough adjustment to the target angle withfeeding knob
ΗSqueeze a rough clamp and fix.
ΘA micromotion adjustment to the target
angle with micrometer.
ΙSqueeze a micromotion clamp and fix.

④Rough clamp

Fitting
Type②Micromotion clamp①Coarse motion 360°
③Fine motion ±5°

CrossRoller
SPECModel(Opposite hand)Stage table size
Travel distanceVernier minimum reading
Minimum reading of micrometer
Guide
Load capacity
Allowable load for moment
Moment rigidity
Parallelism
Eccentricity amount
Runout amount
WeightMain materialSurface finishing
Provided screws (Hex socket screws)

BS43-60
BS43-60R
Į60mm
Coarse motion 360° Fine motion ±5°
Vernier scale0.2°
͚55ˢ/Scale
Cross roller bearing
5.0kgf [49.0N]
5.0N٬m0.15ˢ/N٬cm
50μm
50μm
20μm
0.58kgStainless4 of M48

φ24
φ38
φ60
φ85
φ100
φ1102174