‫ﻣﺨﺘﺒﺮ ﺗﺜﺒﻴﺖ ﻣﺘﻌﺪد اﻟﻮﻇﺎﺋﻒ‬‫‪KEW 6516 / 6516BT‬‬‫اﻟﻜﺜﻴﺮ ﻣﻦ اﺧﺘﺒﺎر اﻟﻄﺎﻗﺔ ﻓﻲ أﻳﺪي‬‫ﻣﺤﺘﺮﻓﻲ اﻟﺘﺜﺒﻴﺖ اﻟﻜﻬﺮﺑﺎﺋﻴﺔ!‬‫ﻓﻘﻂ ‪6516BT‬‬‫واﺟﻬﺔ اﻻﺗﺼﺎل‬‫‪USB‬‬‫‪KEW Report‬‬‫®‪Bluetooth‬‬‫‪RCD‬‬‫دوارة‬‫ﻋﺰل‬‫‪10/30/100/300/500/1000mA‬‬‫‪2/20/200/2000Ω‬‬‫‪100/250/500/1000V‬‬‫اﻷرض‬‫‪PFC‬‬‫‪PSC‬‬‫‪20/200/2000Ω‬‬‫‪2000A/20kA 2000A/50kA‬‬‫‪2000A/20kA‬‬‫ﺗﺪوﻳﺮ اﻟﻄﻮر‬‫اﻻﺳﺘﻤﺮارﻳﺔ‬‫‪ACV‬‬‫‪20/200/2000Ω‬‬‫‪300/600V‬‬‫‪ ) SPD‬ﻓﺎرﻳﺴﺘﻮر (‬‫اﻟﺘﺮدد‬‫‪PAT‬‬‫‪KEW 6516 / 6516BT‬‬‫اﻟﻌﻤﻠﻴﺔ ﻓﻲ ‪ 3‬ﺧﻄﻮات ﺑﺴﻴﻄﺔ‬‫ﻗﻢ ﺑﺘﻌﻴﻴﻦ اﻟﻄﻠﺐ اﻟﺪوار إﻟﻰ ﻧﻄﺎق اﻻﺧﺘﺒﺎر‪.‬‬‫ﻗﻢ ﺑﺘﻮﺻﻴﻞ اﻵﻟﺔ ﺑﺎﻟﺘﺜﺒﻴﺖ ﻗﻴﺪ اﻻﺧﺘﺒﺎر‪.‬‬‫اﺿﻐﻂ زر اﻻﺧﺘﺒﺎر‪.‬‬‫اﺧﺘﺒﺎر ﺑﺤﺮ اﻟﻴﺪﻳﻦ‬‫ﺷﺎﺷﺔ ‪ LCD‬ﻛﺒﻴﺮة‬‫ﺑﻮاﺳﻄﺔ اﻟﻤﺴﺒﺎر ﻋﻦ ﺑﻌﺪ أو اﺳﺘﺨﺪام ﺗﺄﻣﻴﻦ‬‫وﻇﻴﻔﺔ زر اﻻﺧﺘﺒﺎر‪.‬‬‫ﻛﻞ ﺑﻴﺎﻧﺎت اﻻﺧﺘﺒﺎر ﻣﻮﺿﺤﺔ‬‫ﻓﻲ ﺷﺎﺷﺔ ﻛﺒﻴﺮة ﻣﻠﻮﻧﺔ‪.‬‬‫إﻏﻼق‬‫اﻟﻮﻇﻴﻔﺔ‬‫ﻋﻦ ﺑﻌﺪ‬‫ﻣﺴﺒﺎر‬‫زر اﻻﺧﺘﺒﺎر‬‫ﻟﻮﺣﺔ اﻟﻠﻤﺲ ﻻﻛﺘﺸﺎف وﺗﺤﺪﻳﺪ‬‫اﻟﺠﻬﺪ اﻷرﺿﻲ اﻟﻤﺤﺘﻤﻞ ﻟﻠﺨﻄﺮ‪.‬‬‫ﺗﻘﻨﻴﺔ ﻣﻀﺎدة ﻟﻠﺮﺣﻠﺔ )ﻣﻊ ‪ 2‬و ‪ 3‬أﺳﻼك(‬‫ﻟﻌﺪم وﺟﻮد رﺣﻠﺔ اﺧﺘﺒﺎر ‪ LOOP L-PE‬ﻋﻠﻰ ﺟﻤﻴﻊ ‪.RCDs‬‬‫ﻣﻊ ‪ 3‬أﺳﻼك )‪ ،(L, N, PE‬ﻟﻠﺤﺼﻮل ﻋﻠﻰ أﻓﻀﻞ ﻗﺮاءات اﻟﺪﻗﺔ‪.‬‬‫ﻣﻊ ‪ 2‬ﺳﻠﻚ ﻓﻘﻂ ‪ ،‬ﻣﻔﻴﺪ ﺟﺪا ﻓﻲ ﺣﺎﻟﺔ ﻋﺪم وﺟﻮد ﻣﺤﺎﻳﺪ )أي ﺧﻄﻮط ﻣﺤﺮك ‪ 3‬ﻣﺮاﺣﻞ(‪.‬‬‫ﻣﻔﺘﺎح اﻟﺘﻌﻠﻴﻤﺎت ]‪[HELP‬‬‫ﺳﺘﻈﻬﺮ وﻇﻴﻔﺔ اﻟﺘﻌﻠﻴﻤﺎت ﻛﻴﻔﻴﺔ ﺗﻮﺻﻴﻞ اﻵﻟﺔ وﻓﻘﺎ‬‫ﻟﻠﻮﻇﻴﻔﺔ اﻟﻤﺤﺪدة‪.‬‬‫اﺧﺘﺒﺎر دوارة ﻣﻦ ﻣﻘﺒﺲ اﻟﺤﺎﺋﻂ‬‫‪ 0.001‬دﻗﺔ‬‫ﺑﻔﻀﻞ ﺗﻴﺎر اﺧﺘﺒﺎر اﻟﺘﻴﺎر اﻟﻌﺎﻟﻲ اﻟﺒﺎﻟﻎ ‪ ،25A‬ﻳﺘﻢ ﻗﻴﺎس‬‫ﻣﺮﺣﻠﺔ ﻣﻘﺎوﻣﺔ اﻟﺪوارة إﻟﻰ اﻷرض ﺑﺪﻗﺔ ﻋﺎﻟﻴﺔ ﺗﺒﻠﻎ ‪0.001‬‬‫أوم‪ .‬ﻗﺪ ﻳﻜﻮن ﻫﺬا ﻣﻔﻴﺪا ﻋﻨﺪ إﺟﺮاء اﺧﺘﺒﺎر ﻓﻲ ﻟﻮﺣﺔ‬‫اﻟﻤﻔﺘﺎح اﻟﺮﺋﻴﺴﻲ ﺑﺎﻟﻘﺮب ﻣﻦ اﻟﻤﺤﻮل‪.‬‬‫اﻟﺘﺤﻘﻖ ﻣﻦ اﺳﺘﻤﺮارﻳﺔ اﻟﺘﺮ‬‫اﺑﻂ ﻣﺘﺴﺎوي اﻟﺠﻬﺪ‬‫اﻟﻜﺜﻴﺮ ﻣﻦ ﻃﺎﻗﺔ اﻻﺧﺘﺒﺎر ﻓﻲ أﻳﺪي ﻣﺤﺘﺮﻓﻲ اﻟﺘﺜﺒﻴﺖ اﻟﻜﻬﺮﺑﺎﺋﻲ!‬‫ﻳﻤﻜﻦ اﺧﺘﺒﺎر ﻣﺠﻤﻮﻋﺔ ﻣﺘﻨﻮﻋﺔ ﻣﻦ ‪RCDs‬‬‫اﺧﺘﺒﺎر ‪SPD‬‬‫اﻛﺘﺐ ﺗﻴﺎر ﻣﺘﺮدد و ‪ A‬و ‪ F‬و ‪) B‬ﻋﺎم واﻧﺘﻘﺎﺋﻲ( ‪ EV‬و ‪ RCDs‬اﻟﻤﺘﻐﻴﺮة‪ .‬اﺧﺘﺒﺎر‬‫واﺣﺪ وﺗﻠﻘﺎﺋﻲ ‪ ،‬اﺧﺘﺒﺎر ﻣﻨﺤﺪر وﺟﻬﺪ اﻻﺗﺼﺎل‪.‬‬‫ﻳﻤﻜﻦ اﺧﺘﺒﺎر ‪) SPD‬ﺟﻬﺎز اﻟﺤﻤﺎﻳﺔ ﻣﻦ زﻳﺎدة اﻟﺘﻴﺎر( اﻟﺬي‬‫ﻳﺤﺘﻮي ﻋﻠﻰ ﻓﺎرﻳﺴﺘﻮر ﻟﻘﻴﺎس ﺟﻬﺪ رﺣﻠﺔ دون إﺗﻼﻓﻪ‪.‬‬‫اﺧﺘﺒﺎر ‪PAT‬‬‫اﻻﺗﺼﺎل‬‫اﺧﺘﺒﺎر ‪ = PAT) PAT‬ﺟﻬﺎز اﺧﺘﺒﺎر اﻷﺟﻬﺰة‬‫اﻟﻤﺤﻤﻮﻟﺔ( ﻣﻦ اﻟﻤﻤﻜﻦ اﻟﺘﺤﻘﻖ ﻣﻦ ﻣﻘﺎوﻣﺔ‬‫اﻟﻌﺰل واﺳﺘﻤﺮارﻳﺔ اﻟﺮاﺑﻄﺔ اﻷرﺿﻴﺔ ﻟﻸﺟﻬﺰة‬‫اﻟﻤﺤﻤﻮﻟﺔ ﻟﻠﻔﺌﺔ اﻷوﻟﻰ واﻟﺜﺎﻧﻴﺔ‪.‬‬‫ﻳﻤﻜﻦ ل ‪ KEW 6516‬ﺗﻨﺰﻳﻞ ﺑﻴﺎﻧﺎت اﻻﺧﺘﺒﺎر ﻋﻦ ﻃﺮﻳﻖ ﺗﻮﺻﻴﻞ‬‫ﻣﺤﻮل ‪) USB‬ﻃﺮاز ‪ (USB-8212‬ﺛﻢ ﻃﺒﺎﻋﺔ ﺗﻘﺎرﻳﺮ اﻻﺧﺘﺒﺎر اﻟﻜﺎﻣﻠﺔ‬‫ﺑﻮاﺳﻄﺔ ﺟﻬﺎز ﻛﻤﺒﻴﻮﺗﺮ‪ .‬ﺑﻴﻨﻤﺎ ﻳﻤﻜﻦ ل ‪ KEW 6516BT‬ﻧﻘﻞ ﺑﻴﺎﻧﺎت‬‫اﻻﺧﺘﺒﺎر إﻟﻰ ﺟﻬﺎز ﻟﻮﺣﻲ أو ﻫﺎﺗﻒ ذﻛﻲ ﻋﺒﺮ ‪ .Bluetooth‬ﻳﻤﻜﻦ‬‫ﺣﻔﻆ ﺑﻴﺎﻧﺎت اﻻﺧﺘﺒﺎر ﻫﺬه وﻣﺸﺎرﻛﺘﻬﺎ وإرﺳﺎﻟﻬﺎ ﺑﺎﻟﺒﺮﻳﺪ اﻹﻟﻜﺘﺮوﻧﻲ‪.‬‬‫ﻳﻤﻜﻦ أﻳﻀﺎ ﺗﺮﺗﻴﺐ ﻣﺤﻮل ‪ USB‬ﻛﺨﻴﺎر‪.‬‬‫‪USB‬‬‫‪Zs/Ra Limit‬‬‫ﻳﺘﻢ ﺗﺒﺴﻴﻂ اﻟﺘﺤﻘﻖ ﻣﻦ ﻣﺘﻄﻠﺒﺎت‬‫اﻟﺴﻼﻣﺔ ﻋﻠﻰ اﻟﺘﺜﺒﻴﺖ اﻟﻜﻬﺮﺑﺎﺋﻲ‬‫ﺑﺎﺳﺘﺨﺪام وﻇﻴﻔﺔ ‪.Zs/Ra Limit‬‬‫ﺳﺘﺘﺤﻘﻖ ﻫﺬه اﻟﻮﻇﻴﻔﺔ ﺗﻠﻘﺎﺋﻴﺎ ﻣﻤﺎ‬‫إذا ﻛﺎﻧﺖ ﻣﻘﺎوﻣﺔ اﻟﺪوارة اﻟﻤﻘﺎﺳﺔ‬‫ﻟـ ‪) TN‬أو اﻟﺪوارة اﻷرﺿﻴﺔ ل ‪(TT‬‬‫ﻣﻨﺨﻔﻀﺔ ﺑﻤﺎ ﻳﻜﻔﻲ ﻟﺮﺣﻠﺔ )ﻓﺼﻞ(‬‫‪ / MCB‬اﻟﺼﻤﺎم ‪ RCD /‬ﻣﻤﺎ‬‫ﻳﻌﻄﻲ ﻧﺘﻴﺠﺔ ﻳﻤﺮ )✓( أو ﻳﻔﺸﻞ‬‫)×( ﻋﻠﻰ اﻟﺸﺎﺷﺔ‪.‬‬‫ﻛﻤﺒﻴﻮﺗﺮ‬‫‪KEW 6516‬‬‫‪Bluetooth‬‬‫‪KEW 6516BT‬‬‫ﻳﺄﺗﻲ ‪ KEW 6516‬و ‪ KEW 6516BT‬ﻣﻜﺘﻤﻠﻴﻦ ﺑﻜﻞ ﻣﺎ ﺗﺤﺘﺎﺟﻪ ﻻﺧﺘﺒﺎر اﻟﺘﺜﺒﻴﺖ اﻟﻜﻬﺮﺑﺎﺋﻲ‬‫اﻟﻤﻠﺤﻘﺎت‬‫اﻟﻤﻠﺤﻘﺎت اﻻﺧﺘﻴﺎرﻳﺔ‬‫‪MODEL 8017A‬‬‫اﻣﺘﺪاد ﻃﻮﻳﻞ‬‫‪MODEL 7272‬‬‫ﻳﺘﻢ إﻋﺎدة ﺗﺮﻛﻴﺐ ‪ 2‬ﺳﻠﻚ ﻣﻊ ﻣﺠﺲ اﻻﺧﺘﺒﺎر‪،‬‬‫‪ 2‬اﻟﻤﺴﺎﻣﻴﺮ ‪ ،‬ﺟﻬﺪ اﺧﺘﺒﺎر اﻷرض‪،‬‬‫ﺣﻘﻴﺒﺔ ﺣﻤﻞ‪.‬‬‫‪MODEL 8259‬‬‫ﻣﺤﻮل ﻟـ‬‫ﻃﺮﻓﻴﺔ اﻟﻘﻴﺎس‬‫]أﺣﻤﺮ‪ ،‬أﺻﻔﺮ‪ ،‬أﺧﻀﺮ‪ 1/‬ﻣﺠﻤﻮﻋﺔ[‬‫ﻣﺠﻤﻮﻋﺎت ﻣﺤﻮل ‪ MFT‬و ‪EVSE‬‬‫‪KEW 6516-EV2 : KEW 6516×1, KEW 8602×1‬‬‫‪KEW 6516BT-EV2 : KEW 6516BT×1, KEW 8602×1‬‬‫‪KEW 8601‬‬‫ﻣﺤﻮل ‪EVSE‬‬‫ﻗﺎﺑﺲ ﻣﻦ اﻟﻨﻮع ‪1‬‬‫‪KEW 8602‬‬‫ﻣﺤﻮل ‪EVSE‬‬‫ﻗﺎﺑﺲ ﻣﻦ اﻟﻨﻮع ‪2‬‬‫● ﻣﻮاﺻﻔﺎت ‪KEW 6516 / 6516BT‬‬‫‪) SPD‬ﻣﺘﻐﻴﺮ(‬‫ﻣﻘﺎوﻣﺔ اﻟﻌﺰل‬‫ﺟﻬﺪ اﻻﺧﺘﺒﺎر‬‫ﻧﻄﺎﻗﺎت اﻟﻘﻴﺎس‬‫‪100V‬‬‫دﻗﺔ‬‫ﺗﻴﺎر اﻟﻤﻘﺪر‬‫ﺗﻴﺎر ﻣﺎس ﻛﻬﺮﺑﺎﺋﻰ‬‫ﻣﻘﺎوﻣﺔ اﻟﺪوارة‬‫اﻟﻮﻇﻴﻔﺔ‬‫‪500V‬‬‫‪1000V‬‬‫اﻟﺤﺪ اﻷﻗﺼﻰ ‪1000V‬‬‫‪2.000/20.00/200.0MΩ‬‬‫)اﻟﻨﻄﺎق اﻟﺘﻠﻘﺎﺋﻲ(‬‫‪20.00/200.0/1000MΩ‬‬‫)اﻟﻨﻄﺎق اﻟﺘﻠﻘﺎﺋﻲ(‬‫‪20.00/200.0/2000MΩ‬‬‫)اﻟﻨﻄﺎق اﻟﺘﻠﻘﺎﺋﻲ(‬‫‪)0-1000V‬ﺗﺮﺗﻔﻊ ﺑﻤﻘﺪار ‪(1V‬‬‫)‪±2%rdg±6dgt (2.000/20.00MΩ‬‬‫)‪±5%rdg±6dgt (200.0MΩ‬‬‫‪1.0-1.2mA‬‬‫‪1.0-1.2mA‬‬‫‪@0.1MΩ‬‬‫‪@0.25MΩ‬‬‫)‪±5%rdg±5dgt ±2%rdg±6dgt (20.00/200.0MΩ) ±2%rdg±6dgt (20.00/200.0MΩ‬‬‫)‪±5%rdg±6dgt (1000MΩ‬‬‫)‪±5%rdg±6dgt (2000MΩ‬‬‫‪250V‬‬‫‪1.0-1.2mA‬‬‫‪@1MΩ‬‬‫‪1.0-1.2mA‬‬‫‪@0.5MΩ‬‬‫–‬‫‪ 1.5mA‬ﻛﺤﺪ أﻗﺼﻰ‬‫–‬‫ﺗﻜﺮار ‪ATT‬‬‫ﺟﻬﺪ ﻣﻘﻨﻦ‬‫ﻣﺪى اﻟﻨﻄﺎق‬‫ﺗﻜﺮار ﻋﺎﻟﻲ‬‫‪ 3)L-PE/L-N‬ﺳﻠﻚ(‬‫‪ 2)L-PE‬ﺳﻠﻚ(‬‫)‪L-PE(0.01ΩRes‬‬‫)‪100-260V(50/60Hz‬‬‫)‪48-260V(50/60Hz‬‬‫)‪48-260V(50/60Hz‬‬‫)‪100-260V(50/60Hz‬‬‫‪20.00/200.0/2000Ω‬‬‫)اﻟﻨﻄﺎق اﻟﺘﻠﻘﺎﺋﻲ(‬‫‪2.000Ω‬‬‫)‪L-PE(0.001ΩRes‬‬‫‪L-N/L-L‬‬‫)‪48-500V(50/60Hz‬‬‫‪20.00Ω‬‬‫‪±3%rdg±10dgt‬‬‫‪±3%rdg±4dgt‬‬‫‪±3%rdg±25mΩ‬‬‫‪±3%rdg±4dgt‬‬‫‪L-PE:15mA‬‬‫‪20Ω:6A/20ms‬‬‫‪200Ω:0.5A/20ms‬‬‫‪2000Ω:15mA/500ms‬‬‫‪25A/20ms‬‬‫‪6A/20ms‬‬‫)‪2000A/20kA(PFC‬‬‫)‪2000A/50kA(PFC‬‬‫)‪2000A/20kA(PSC‬‬‫‪20.00/200.0/2000Ω‬‬‫)اﻟﻨﻄﺎق اﻟﺘﻠﻘﺎﺋﻲ(‬‫دﻗﺔ‬‫‪±3%rdg±6dgt‬‬‫اﺧﺘﺒﺎر اﻟﺘﻴﺎر اﻻﺳﻤﻲ ﻋﻨﺪ اﻟﺤﻠﻘﺔ اﻟﺨﺎرﺟﻴﺔ ‪L-N:6A/60ms :0Ω‬‬‫اﻟﺤﺠﻢ‪/‬اﻟﻤﺪة ﻋﻨﺪ ‪230V‬‬‫‪N-PE:10mA‬‬‫‪PSC/PFC‬‬‫اﻟﻨﻄﺎق‬‫دﻗﺔ‬‫‪RCD‬‬‫ﺟﻬﺪ ﻣﻘﻨﻦ‬‫اﻟﻮﻇﻴﻔﺔ‬‫))‪2000A/20kA(PFC) 2000A/20kA(L-N(PSC)/L-PE(PFC‬‬‫ﺗﺴﺘﻤﺪ دﻗﺔ ‪ PSC/PFC‬ﻣﻦ ﻣﻮاﺻﻔﺎت ﻣﻘﺎوﻣﺔ اﻟﺪوارة وﻣﻮاﺻﻔﺎت اﻟﺠﻬﺪ اﻟﻤﻘﺎس‬‫)‪100-260V(50/60Hz‬‬‫‪ x1/2, x1, x5,‬ﻣﻨﺤﺪر‪ ،‬اﻟﺘﻠﻘﺎﺋﻲ‪Uc ،‬‬‫‪/6/10/30/100/300/500/1000mA‬اﻟﻤﺘﻐﻴﺮة‬‫ﻧﻮع ‪RCD‬‬‫إﻋﺪاد ﺗﻴﺎر اﻟﺮﺣﻠﺔ‬‫دﻗﺔ‬‫ﺗﻴﺎراﻟﺮﺣﻠﺔ‬‫وﻗﺖاﻟﺮﺣﻠﺔ‬‫)‪AC(G/S‬‬‫)‪A(G/S‬‬‫‪10/30/100/300/500mA 10/30/100/300/500/1000mA(G) x1/2,x1,Uc‬‬‫)‪10/30/100/300/500mA(S‬‬‫)‪F(G/S‬‬‫)‪B(G/S‬‬‫‪EV‬‬‫‪10/30/100/300/500mA‬‬‫‪10/30/100/300mA‬‬‫‪1×)6mA‬ﻓﻘﻂ(‬‫‪x5‬‬‫‪10/30/100mA‬‬‫‪10/30/100mA‬‬‫‪10/30/100mA‬‬‫‪10/30mA‬‬‫–‬‫ﻣﻨﺤﺪر‬‫‪10/30/100/300/500mA‬‬‫‪10/30/100/300/500mA‬‬‫‪10/30/100/300/500mA‬‬‫‪10/30/100/300mA‬‬‫‪6mA‬‬‫‪x1/2‬‬‫‪-8% - -2%‬‬‫‪-10% - 0%‬‬‫‪-10% - 0%‬‬‫‪-10% - 0%‬‬‫–‬‫‪x1‬‬‫‪+2% - +8%‬‬‫‪0% - +10%‬‬‫‪0% - +10%‬‬‫‪0% - +10%‬‬‫‪0% - +10%‬‬‫‪x5‬‬‫‪+2% - +8%‬‬‫‪0% - +10%‬‬‫‪0% - +10%‬‬‫‪0% - +10%‬‬‫–‬‫ﻣﻨﺤﺪر‬‫‪-4% - +4%‬‬‫‪-10% - +10%‬‬‫‪-10% - +10%‬‬‫‪-10% - +10%‬‬‫‪-10% - +10%‬‬‫‪x1/2‬‬‫‪2000ms(G/S):±1%rdg±2ms‬‬‫–‬‫‪x1‬‬‫‪550ms(G):±1%rdg±2ms,1000ms(S):±1%rdg±2ms‬‬‫‪10.5s:±1%±2ms‬‬‫‪x5‬‬‫‪410ms(G/S):±1%rdg±2ms‬‬‫اﻻﺳﺘﻤﺮارﻳﺔ‬‫اﻟﻨﻄﺎق‬‫ﺟﻬﺪ اﻟﺪاﺋﺮة اﻟﻤﻔﺘﻮﺣﺔ )ﺗﻴﺎر ﻣﺴﺘﻤﺮ(‬‫‪200mA‬‬‫ﻗﻴﺎس‬‫ﺗﻴﺎر‬‫‪15mA‬‬‫دﻗﺔ‬‫ﺗﺪوﻳﺮ اﻟﻄﻮر‬‫ﺟﻬﺪ ﻣﻘﻨﻦ‬‫اﻟﻤﻼﺣﻈﺎت‬‫–‬‫ﻓﻮﻟﻂ‬‫‪) 20.00/200.0/2000Ω‬اﻟﻨﻄﺎق اﻟﺘﻠﻘﺎﺋﻲ(‬‫‪7-14V‬‬‫اﻟﻨﻄﺎق‬‫ﻧﻄﺎﻗﺎت اﻟﻘﻴﺎس‬‫‪ 200mA‬أو أﻛﺜﺮ )‪ 2Ω‬أو أﻗﻞ(‬‫‪) 15mA±3mA‬دارة ﻗﺼﻴﺮة(‬‫دﻗﺔ‬‫‪±2%rdg±8dgt‬‬‫‪) 300.0/600V‬اﻟﻨﻄﺎق اﻟﺘﻠﻘﺎﺋﻲ(‬‫ﻓﻮﻟﻂ‬‫‪2-600V‬‬‫اﻟﺘﺮدد‬‫‪45-65Hz‬‬‫ﻓﻮﻟﻂ‬‫‪±2%rdg±4dgt‬‬‫اﻟﺘﺮدد‬‫‪±0.5%rdg±2dgt‬‬‫اﻷرض‬‫)‪48-600V(50/60Hz‬‬‫ﻋﺮض اﻟﺘﺴﻠﺴﻞ اﻟﻄﻮري اﻟﺼﺤﻴﺢ‪ "1.2.3" :‬وﻋﻼﻣﺔ‬‫اﻟﺘﺴﻠﺴﻞ اﻟﻄﻮري اﻟﻤﻌﻜﻮس‪ :‬ﻳﻌﺮض "‪ "3.2.1‬وﻳﻮﺿﻊ ﻋﻼﻣﺔ‬‫اﻟﻨﻄﺎق‬‫‪) 20.00/200.0/2000Ω‬اﻟﻨﻄﺎق اﻟﺘﻠﻘﺎﺋﻲ(‬‫دﻗﺔ‬‫)‪±2%rdg±0.08Ω(20.00Ω‬‬‫)‪±2%rdg±3dgt(200.0/2000Ω‬‬‫ﻋﺎم‬‫ﻣﻌﻴﺎر ﻗﺎﺑﻞ ﻟﻠﺘﻄﺒﻴﻖ‬‫واﺟﻬﺔ اﻻﺗﺼﺎل‬‫ﻣﺼﺪر اﻟﻄﺎﻗﺔ‬‫اﻟﺒﻌﺪ‬‫اﻟﻮزن‬‫اﻟﻤﻠﺤﻘﺎت‬‫اﻟﻤﻠﺤﻘﺎت اﻻﺧﺘﻴﺎرﻳﺔ‬‫‪ IEC 61010-1 CAT Ⅳ 300V / CAT Ⅲ‬درﺟﺔ اﻟﺘﻠﻮث ‪IEC 61326(EMC) ،IEC 60529(IP40) ،IEC 61557-1،2،3،4،5،6،7،10 ،IEC 61010-2-034 ،600V 2‬‬‫‪ ،USB, Bluetooth 5.0*¹‬اﻧﺪروﻳﺪ ‪ 5.0‬أو أﺣﺪث‪ iOS 10.0 ،‬أو أﺣﺪث‬‫‪LR6(AA)(1.5V) × 8‬‬‫‪136(L) × 235(W) × 114(D)mm‬‬‫‪) 1350g‬ﺑﻤﺎ ﻓﻲ ذﻟﻚ اﻟﺒﻄﺎرﻳﺎت(‬‫ﻣﺠﺲ اﺧﺘﺒﺎر اﻟﺘﻴﺎر اﻟﻜﻬﺮﺑﺎﺋﻲ*‪)7281 ،2‬ﻣﺠﺲ اﻻﺧﺘﺒﺎر ﻣﻊ ﻣﻔﺘﺎح اﻟﺘﺤﻜﻢ ﻋﻦ ﺑﻌﺪ(‪)7246 ،‬ﻣﺠﺲ اﺧﺘﺒﺎر ﻟﻮﺣﺔ اﻟﺘﻮزﻳﻊ(‪)7228A ،‬ﻣﺠﺲ اﺧﺘﺒﺎر ﻣﻘﺎوﻣﺔ اﻷرض(‪،‬‬‫‪)8041‬ﻣﺴﺎﻣﻴﺮ اﻷرض اﻟﻤﺴﺎﻋﺪة]‪ 2‬ﻣﺴﺎﻣﻴﺮ‪ 1/‬ﻣﺠﻤﻮﻋﺔ[(‪)9084 ،‬ﺣﻘﻴﺒﺔ ﻧﺎﻋﻤﺔ(‪)9142 ،‬ﺣﻘﻴﺒﺔ ﺣﻤﻞ(‪)9151 ،‬ﺣﺰام ﻛﺘﻒ(‪)9199 ،‬وﺳﺎدة ﻛﺘﻒ(‪ ،LR6(AA) × 8 ،‬دﻟﻴﻞ اﻟﺘﻌﻠﻴﻤﺎت‪،‬‬‫‪)USB-8212‬ﻣﺤﻮل ‪ USB‬ﻣﻊ "‪)KEW Report‬ﺑﺮﻧﺎﻣﺞ("(*‪ ،3‬ﺷﻬﺎدة اﻟﻤﻌﺎﻳﺮة‬‫‪)USB-8212‬ﻣﺤﻮل ‪ USB‬ﻣﻊ "‪) KEW Report‬اﻟﺒﺮﻧﺎﻣﺞ(''(‪)8259 ،*³‬ﻣﺤﻮل ﻟﻄﺮﻓﻴﺔ اﻟﻘﻴﺎس(‪)7272 ،‬ﻣﺠﻤﻮﻋﺔ ﺳﻠﻚ ﻗﻴﺎس اﻟﺪﻗﺔ(‪)8017A ،‬ﺗﻤﺪﻳﺪ ﺑﺮود ﻃﻮﻳﻞ(‬‫‪)8601‬ﻣﺤﻮل ‪)8602 ،(EVSE‬ﻣﺤﻮل ‪(EVSE‬‬‫*‪ 6516BT ¹‬ﻓﻘﻂ‬‫*‪ :7187A ²‬ﻗﺎﺑﺲ ﺑﺮﻳﻄﺎﻧﻲ‪ :7218A ،‬ﻗﺎﺑﺲ ﺷﻮﻛﻮ أوروﺑﻲ)‪ ،(EU‬ﻗﺎﺑﺲ )‪ 7221A(SA‬ﺟﻨﻮب أﻓﺮﻳﻘﻲ‪ :7222A ،‬ﻗﺎﺑﺲ )‪(AU‬أﺳﺘﺮاﻟﻲ‬‫*‪ :USB-8212 ³‬ﻣﻠﺤﻖ ﻗﻴﺎﺳﻲ ﻟـ ‪ ، 6516‬ﻣﻠﺤﻖ اﺧﺘﻴﺎري ﻟـ ‪6516BT‬‬‫‪ Bluetooth‬ﻫﻲ ﻋﻼﻣﺔ ﺗﺠﺎرﻳﺔ أو ﻋﻼﻣﺔ ﺗﺠﺎرﻳﺔ ﻣﺴﺠﻠﺔ ﻟﺸﺮﻛﺔ ‪.Bluetooth SIG‬‬‫‪ Android‬ﻫﻲ ﻋﻼﻣﺔ ﺗﺠﺎرﻳﺔ أو ﻋﻼﻣﺔ ﺗﺠﺎرﻳﺔ ﻣﺴﺠﻠﺔ ﻟﺸﺮﻛﺔ ‪.Google Inc‬‬‫‪ iOS‬ﻫﻲ ﻋﻼﻣﺔ ﺗﺠﺎرﻳﺔ أو ﻋﻼﻣﺔ ﺗﺠﺎرﻳﺔ ﻣﺴﺠﻠﺔ ﻟﺸﺮﻛﺔ ‪ .Cisco Technology, Inc‬ﻓﻲ اﻟﻮﻻﻳﺎت اﻟﻤﺘﺤﺪة وﺑﻠﺪان أﺧﺮى‪.‬‬‫● ﺻﺪر ﻫﺬا اﻟﻤﻨﺸﻮر ﻓﻲ ﺳﺒﺘﻤﺒﺮ ‪ .٢٠٢٣‬ﺗﺨﻀﻊ ﻣﺤﺘﻮﻳﺎت ﻫﺬا اﻟﻤﻨﺸﻮر ﻟﻠﺘﻐﻴﻴﺮ دون إﺷﻌﺎر‪.‬‬‫‪KEW_6516/6516BT-3E_Sep23_SS_1.ar‬‬