Small Tool Instruments
and Data Management

Inspection Instrument for Indicators

i-Checker

Catalog No. E12015(4)For efficient in-house calibration

Various types of indicators can be inspected with a single tester.
Features• Inspection can be performed approx. 2.5 times fastercompared to the previous model.
• This instrument achieves the highest accuracy in its class(Mitutoyo survey, February 2016) and, therefore, guarantees
ultra-reliable inspection results.
• Digital type indicators equipped with a data output function
are checked very efficiently due to spindle positioning at theinspection points and recording of measurement results being
under fully automatic control.
• Analog type indicators are inspected in semi-automatic modewith the pointer of the indicator being manually adjusted ateach measuring point with automatic transfer of inspection
results and movement to the next measuring point.

• The various standards, such as the latest standard for dialindicators, ASME and DIN, are applied.
For the supported standards, see “Supported IndustrialStandards” on page 6.
• Inspection standards to a customer's own specification can beimplemented.

• A laptop PC can be connected using a USB cable, which contributes
to space-saving.

• All functions necessary for inspection are combined in the control box so thatthe operator need not rely on excessive eye movement to adjust the pointer.
1.【DATA】button

Inputs the current position data andmoves spindle to just before the next
inspection position automatically.

STOP button

Automatictransfer

2. Final positioning is by
Jog dial

RETRY button
DATA button

Inspection using analog type indicator

2Inspection using digital type indicator

Automatictransfer

Jog dial/shuttle control
Jog shuttle (continuous up/down movement)
Jog dial (fine incremental movement)

3.【DATA】button

Inputs the current position data andmoves spindle to just before the next
inspection position automatically.
Note: The distance at which spindle
movement is stopped before the next
graduation is user-definable.

CANCEL button

3

Optional Accessories and Applications

Optional Accessories and Applications

Applications for Bore Gage Accessory Set

Applications for Dial Test Indicator Accessory Set

Applications for bore gage measurement accessory set (type C)

Applications for dial test indicator accessory set (for horizontal type TI)

Application for mounting bore gage accessory set (type C)
Order No. 02ASU162

Applications for dial test indicator accessory set (for vertical type TI)

List of components for bore gage accessory set(type C) Order No.02ASU162

Dial Test Indicator Accessory Set (for ø6 stem)
Order No.02ASK000(1)

Unit: mm

9073

(3)

4148.5i-Checker mountable bore gages

240

100

Wooden box(4)4Range (mm)
ø6 to ø18
50 or more
120 or less(2)(1) Bore gage accessory(12)(6)(2) Stand B(11)(7)(3) V-shaped attachment B
(8)
(13)
(9)
(4) Bore gage stopper 5(5) Anvil (for calibration, M11 female)(10) V-shaped attachment C(6) Anvil (for calibration, M3.5×0.35 male) (11) Bore gage stopper 3(7) Anvil (for calibration, M3.5×0.5 male) (12) i-Checker's special(8) Anvil (for calibration, M5 male)contact point(9) Anvil (for calibration, M11 male)(13) Key wrench

59.5b(10)cabc(5)1525a27
46.5

5

Supported Industrial Standards
Dial Indicators
463:2006
R463:1965
B7503:2017
B7503-2011
B7503:1997
B7503:1992
2001:1998
2001:1994
2003:1994B89.1.10M:2001
B89.1.10M:1987
878:2006
878:1983

ISOJISJMASANSI / ASMEDIN
VDI/VDE /DGQ

2618
Blatt11:1991

BS
Mitutoyo Standard

907:1965—

Dial Test Indicators

Hicators

Bore Gages

Digimatic Indicators

Linear Gages

Electronic
Micrometers

9493:2010

—

—

—

—

—

B7533:2015
B7533:1990

B7519:1994

B7515:1982——

B7536:1982——

5009:1988——

5003:1962

B89.1.10M:2001
B89.1.10M:1987
2270:2017
2270:1985
2618
Blatt11.3:2002
2618
Blatt20:1991
2795:1981———

B89.1.10M:2001——

879:1983

—

—

—

—

2618
Blatt21:1991

———

2618
Blatt26:1991

—
—

—
—

—
4

—
4—
—

* “—” means that there are no applicable industrial standards.

IC2000100 mm0.01 μm(0.1+0.4L / 100) μm L=Arbitrary length (mm)(0.15+0.6L / 100) μm L=Arbitrary length (mm)Maximum 10 mm/s
Motordrive,
semi-automatic, fully automatic
Drive method(Fully automatic only for Indicator with SPC data output)
Measuring Unit
Separate type Linear Encoder
Semi-automatic measurement
Fully automatic measurement
Measurement method(only when using an indicatorequipped with data output function)*1*2
Mass
20 kg
Operating temperature range
20 °C±0.5 °C
Model No.
Measuring Range
Resolution
Accuracy vertical position(20 °C) lateral position
Feed speed• Dial Indicators• Dial Test Indicators• Mitutoyo Hicators• Bore Gages• Digimatic Indicators• Linear Gages• Electronic Micrometers (Mu-Checker)

Dimensions—7379.5

Usage and remarks

02ASK040 Stem bush ø602ASJ856 Stem bush ø8
02ASK150 Stem bush ø8, short

For attachment of cartridge head MCHP-341

02ASL150 Stem bush ø1002ASK050 Bush ø9.5 to ø15For attachment of LGB2 fixing nut type withstem diameter ø9.5Stem bush (ø15 mm) is required separately.

02ASK060 Stem bush ø1202ASK070 Stem bush ø1502ASK080 Stem bush ø20
02ASK710 Stem bush ø28
For attachment of indicator with stem diameterø9.525 mmbox for stem Storage of 7 stem bushes and 1 bush
02ASK130 Wooden
bushes(ø9.5 to ø15 mm)
OthersUsed instead of [DATA] button in the control
937179T Foot switch
box.
02ASK730 Observation mirror
02ASK090 Stem bush ø3/8 inConnecting cables

965014937387965013Connecting cable (1 m)*1 Used for connecting with Digimatic Indicators
Connecting cable (2 m)*1 (ID-C, ID-S, ID-U, and IDU)
Connecting cable (1 m)*1 Used for connecting with Digimatic Indicators(ID-F, IDF, Digimatic power supply unit andConnecting cable (2 m)*1 between the testers)
Connecting cable (1 m)*1 Used for connecting with Digimatic Indicators
Connecting cable (2 m)*1 (IDC and IDA)

21EAA194 Connecting cable (1 m) Used for connecting with Digimatic Indicators
21EAA190 Connecting cable (2 m) (ID-B and ID-N)

Stroke 100QTY.
1111111111

Item

Stem bushes

936937ø8 / ø9.525, Standard accessory

965275

Digimatic power supply Used for the Digimatic Indicators that require
unit*2
external power supply*1 Use the cable described in the user's manual.
*2 Confirm the specification in the user's manual.

384.8559.5

Tester main unit
Controller ICMC-2
Control box
EXT. Signal Cable
EXT. Motor Cable
EXT. USB2.0 Cable
Stem bush 8 mm
Stem bush 9.525 mm
Dust cover
User's Manual (English)
Inspection certificate, Certification of calibration andTraceability chart (English)* Fully automatic measurement requires the indicator's connecting cable.
Additionally some form of indicator, along with the normally connected accessory(the optional accessory for the indicator such as a Digimatic power supply unit in
an EF counter) will be required.

Order No.

Attachments
02ASK000 Test indicator accessory set For attachment of ø6 stem test indicator(for ø6 mm stem)
Test indicator accessory set For attachment of ø8 stem test indicator
02ASK180 (for ø8 mm stem)
Test indicator holder
02ASK370 (for ø6 stem)
Holder to fix ø6 stem to 02ASK180
Testindicatorholder
02ASK380
Holder to fix ø8 stem to 02ASK000(for ø8 stem)
Stem to attach Mu-Checker with knurled
21CZB128 ø6 dovetail grooved stem clamp ring (lever head type) to test indicator
accessory set (02ASK000)
Stem to attach Mu-Checker with knurled
21CZB129 ø8 dovetail grooved stem clamp ring (lever head type) to test indicator
accessory set (02ASK180)
gage accessory set Accessory set for holding grip type bore gage
02ASU162 Bore(type C)

905409189Item• Basic inspection functions· Creation of inspection standard conformed to the industrial standard· Creation and edition of original inspection standard, by customizing theindustrial standard· Inspection of indicator using the inspection standard(inspection for accuracy and repeatability)· Graphical display of measurement result· Creation, edition, and printing of simplified inspection certificate

Optional accessories
Attachments/Stem bush/Others/Connecting cables

905338370.5

Order No.
——−
02ASU001
02ASU002
02ASU003A
02ASJ856
02ASK091*3
601614-2
99MBD063A

Unit: mm

ø160

Standard accessories supplied• Supported Industrial Standards
ISO, JIS, JMAS, ANSI/ASME, DIN, VDI/VDE/DGQ, BS• Input/Output specifications· Digimatic data connector (rear side of i-Checker main unit)*· PC serial-data input port (RS-232C)· Generating, editing and printing of simplified inspection certificates

Note 1: It cannot inspect the indicator with accuracy that is higher than themain unit accuracy of the tester.
Note 2: Resolution of electronic micrometer (Mu-Checker) is greater than 1 μm
(e.g. 1 μm / 5 μm / 10 μm/ 50 μm).

*1 Fully automatic measurement requires the indicator’s connecting cable.
Additionally some form of indicator, along with the normally connected accessory(the optional accessory for the indicator such as a Digimatic power-supply unit in
an EH counter) will be required.
*2 The indicator measured via RS-232C has the capability to receive data from themain unit and output the counter value.

• Supported OS
Windows10 (64 bit)

• Functions· Retry "measurement positioning"· Re-measurement· Data cancellation· Change of speed for jog shuttle

Supported indicators
Specifications

Required optional accessories
Dedicated software (i-Pak specification)

1*3 When purchasing as single part, use the order No. 02ASK090.
1646

ø2007

176 2110(4)e-(PA)NE-(JP7), Printed in Japan

Coordinate Measuring Machines

Vision Measuring Systems

Form Measurement

Optical Measuring

Whatever your challenges are, Mitutoyo
supports you from start to finish.
Mitutoyo is not only a manufacturer of top quality
measuring products but one that also offers qualified
support for the lifetime of the equipment, backed up
by comprehensive services that ensure your staff canmake the very best use of the investment.
Sensor Systems

Test Equipment

Digital Scale and DRO Systems

Small Tool Instruments
and Data Management

Apart from the basics of calibration and repair,
Mitutoyo offers product and metrology training,as well as IT support for the sophisticated software
used in modern measuring technology. We can also
design, build, test and deliver measuring solutions
and even, if deemed cost-effective, take your criticalmeasurement challenges in-house on a sub-contractbasis.

Find additional product literatureand our product catalogue
https://www.mitutoyo.co.jp/global.html
Mitutoyo Corporation
Our products are classified as regulated items under Japanese Foreign Exchange and Foreign Trade Law.
Please consult us in advance if you wish to export our products to any other country.
If the purchased product is exported, even though it is not a regulated item (Catch-All controls item), the customer service
available for that product may be affected. If you have any questions, please consult your local Mitutoyo sales office.
Note: Product illustrations are without obligation. Product descriptions, in particular any and all technicalspecifications, are only binding when explicitly agreed upon.
MITUTOYO and MiCAT are either registered trademarks or trademarks of Mitutoyo Corp. in Japan and/or other countries/regions.
Other product, company and brand names mentioned herein are for identification purposes only and may be thetrademarks of their respective holders.

20-1, Sakado 1-Chome,
Takatsu-ku, Kawasaki-shi,
Kanagawa 213-8533, Japan
T +81 (0) 44 813-8230
F +81 (0) 44 813-8231
https://www.mitutoyo.co.jp