Optical Measuring

Toolmakers' Microscopes

TM Series

Catalog No. E14013(3)SPECIFICATIONS

TM-505B
176-818-11

Model No.
Order No.
Optical tube
Observation image
Eyepiece protractor
Eyepiece
Objective lens
Microscope Maximum height of workpiecehead
Focusing method
Illumination Transmitted illumination
unit
Surface illumination

TM-1005B
176-819-11

Monocular type (Depression angle: 60°)
Erect
Resolution (graduation) : 1°, Rotation angle: 360°, Minimum reading by Vernier: 6', Adjustable zero point
Standard accessory: 15X (field number: 13), Options: 10X, 20X
Standard accessory: 2X, Options: 5X, 10X
115 mm
107 mmManual (Coarse feed)
Stepless brightness adjustment, White LED light source, With green filter
Oblique single-source type, Stepless brightness adjustment, White LED light source
100×50 mm(An optional 50 mm gauge block* is required to cover full range.
Measuring range
50×50 mm
A CERA block is recommended.)
Cross-travel
Table size
152×152 mm
240×152 mmstage
Usable area of the stage glass
96×96 mm
154×96 mm
Maximum stage glass loading
5 kg
Linear measurement method
Micrometer head*
Depends on the micrometer head specifications*Resolution(for MHD-50MB (164-163): 0.001 mm)
Micrometer head travel range
for MHD-50MB (164-163): 50 mm
Power source
100 to 240 VAC 50/60 Hz Maximum power consumption: 4.2 W
14 kg
Main unit mass
15 kg
* Micrometer heads and gauge blocks (CERA blocks) are optional.

DIMENSIONS

TM-505B
176-818-11* Shown with optional Digimatic micrometer heads

313343

12186
210

28

Newly designed LED illuminators provide clear
observation with outstanding color reproduction.

Eyepiece
protractor

Conventional illumination

LED illumination

Installation of optional Digimatic micrometer heads(164-163) enables easy digital measurement.

2

Optional LED circular illuminatoravailable for applications requiring
all-round lighting.

TM-505B

1581296447

186
210492313
343

15

SYSTEM DIAGRAM

TM-1005B

For TM505B

A vernier scale mounted outside
the eyepiece enables simple anglemeasurement.

127
2525139
1220

15492

379

Eyepiece 10X
176-115FEATURES

50

131
12Max. workpiece height
107
67 Working distance

15850

min.343 max.463

1272525392

Max. workpiece height
115
67 Working distance2525(Unit:mm)

TM-1005B

min343 max.463

Bright and ClearField of View
Superior Performancein Small-model Class

392

TM-505B

Eyepiece 15X
176-116

Rotary table
176-106

Eyepiece 20X
176-117For digital
measurement

Objective lens 2X
176-138
Objective lens 5X
176-139

TM-505B/1005B main unit

Objective lens 10X
176-137

Dial indicator attachment
for Z-axis
measurement
176-204

Power cord (essential)
02ZAA000 (Japan)
02ZAA010 (USA)
02ZAA020 (EU)
02ZAA030 (UK)
02ZAA040 (China)
02ZAA050 (Korea)

For TM1005B

LED circular illumination unit
176-208

Dial indicator with stem
diameter of ø8 (or lineargage) and gauge block

Micrometer headtravel range: 50mm
MHD-50MB
164-163×2pcs.

Swivel center
176-105

Rotary table with fine feed knob Swivel center
172-197
172-198

Rotary table
172-196

V-block with clamp
172-378

For TM1005B
CERA block 50mm (613675-04) x 1pc.
Connection cable
959149(1m)×2pcs.
or959150(2m)×2pcs.

For analog measurement
Micrometer head travel range: 25mm
For X-axis: MHG1-25X2 152-390
For Y-axis: MHG1-25Y2 152-389

MUX-10F
264-002

Connection
cable C
12AAD194

2D Data
Processing Unit
QM-Data200
264-160-11 (mm)264-160-13 (mm/inch)

For TM505B

CERA block 25mm613635-04×2pc1.

For TM-1005B

Holder with clamp
176-107

CERA block 25mm 613635-04×2pcs.
CERA block 50mm 613675-04×1pc
Standard accessory
Refer to the QM-DATA200 (Catalog No.E14008) for more details.

Notes: 1) One or two micrometer heads are required for measuring dimensions.
2) If a measuring range longer than the micrometer head travel range is to be used, a gauge block accessory is required.

3

2404 (JP10), Printed in Japan

Coordinate Measuring Machines

Vision Measuring Systems

Form Measurement

Optical Measuring

Whatever your challenges are, Mitutoyo
supports you from start to finish.
Mitutoyo is not only a manufacturer of top quality
measuring products but one that also offers qualified
support for the lifetime of the equipment, backed up
by comprehensive services that ensure your staff canmake the very best use of the investment.
Sensor Systems

Test Equipment

Digital Scale and DRO Systems

Small Tool Instruments
and Data Management

Apart from the basics of calibration and repair,
Mitutoyo offers product and metrology training,as well as IT support for the sophisticated software
used in modern measuring technology. We can also
design, build, test and deliver measuring solutions
and even, if deemed cost-effective, take your criticalmeasurement challenges in-house on a sub-contract
basis.

Find additional product literatureand our product catalogue
https://www.mitutoyo.co.jp/global.html
Mitutoyo Corporation
Notes on Export Regulations:
Do not commit an act, which could directly or indirectly, violate any law or regulation of Japan,
your country or any other international treaty, relating to the export or re-export of any commodities.

20-1, Sakado 1-Chome,
Takatsu-ku, Kawasaki-shi,Kanagawa 213-8533, Japan

Note: Product illustrations are without obligation. Product descriptions, in particular any and all technical specifications,are only binding when explicitly agreed upon.
MITUTOYO and MiCAT are either registered trademarks or trademarks of Mitutoyo Corp. in Japan and/or other countries/regions.
Other product, company and brand names mentioned herein are for identification purposes only and may be thetrademarks of their respective holders.
All product information contained in this brochure is current as of April 2024.

T +81 (0) 44 813-8230
F +81 (0) 44 813-8231
https://www.mitutoyo.co.jp