Eclipse® 5
with autoSAT®

User Manual (US)Eclipse 5User Controls & System Status Indicators
ISO 7000
Operator’s manual; operating instructions. Reg. # 1641
Keep away from rain, keep dry. Reg.
# 0626
Stacking limit by number. Reg. #2403
Name and address of manufacturer.
Reg. # 3082
The country and date of manufacture.
The “CC” identifies the two lettercountry code of the country of manufacture. The date of manufacture is in
YYYY-MM-DD format. Reg. # 6049

Type BF applied part (degree ofprotection against electric shock).
Reg. # 5333
Council Directive 93/42/EEC
Authorized representative in theEuropean Community
If the product unique device identifier(UDI) label has the CE#### symbolon it, the device complies with therequirements of Directive 93/42/EEC
concerning medical devices. The
CE#### symbol indicates notified
body number.
Additional Symbols
External Power Indicator
Power Cartridge (battery) Status
Gauge

Catalog Number. Reg. # 2493
Serial Number. Reg. # 2498
Storage or operating temperature
range. Reg. # 0632
Storage humidity range. Reg. # 2620

A

02

Power Cartridge Symbol
Amperes
Oxygen Output
Recycle Symbol

Atmospheric pressure limitation. Reg.
# 2621

FAA Approved Symbol: The U.S.
Federal Aviation Administration(FAA) has approved this device foruse on-board commercial aircraft.

This way up. Reg. # 0623

Flow Setting Indicator

Fragile, handle with care. Reg. # 0621

Pulse Mode Operation

Contains hazardous substances. Reg.
# 3723

Device operating normally; power
buttonImporter. Reg. # 3725

Increase Flow Setting

ISO 7010
The instruction manual must be read.
Reg. # M002
Keep away from open flame, fire,
sparks. Open ignition source andsmoking prohibited. Reg. # P003
Do not smoke near unit or while
operating unit. Reg. # P002

2 - US PN 21182453 G | User ManualDecrease Flow Setting
FCC Notice
Authorized representative in
Switzerland.
If the device bears the UKCA markas shown with UKCA#### indicating
the notified body number, this device
complies with UKCA regulations.

Eclipse 5
IEC 60417
Class II Equipment, Double InsulatedReg. # 5172
Alternating Current Reg. # 5032
Direct Current Reg. # 5031
21 CFR 801.15
Federal law restricts this device tosale by or on the order of a physician.
IEC 60601-1
Drip Proof
Council Directive 2012/19/EU
WEEE
This symbol is to remind the equipment owners to return it to a recycling
facility at the end of its life, per Waste
Electrical and Electronic Equipment(WEEE) Directive.
Our products will comply with therestriction of Hazardous Substances(RoHS) directive. They will not contain more than trace amounts of leador other hazardous material content.
IEC 60601-1-8
Low Priority Technical Alarm
ISO 15223-1
Medical Device. Reg. # 5.7.7

FCC NOTICE:
This device may contain CYBLE-022001-00, including the antenna 2450AT18B100 from Johnson
Technology, complies with par 15 of the FCC rules.
The device meets the requirements for modulartransmitter approval as detailed in FCC public Notice DA00-1407. Transmitter operation is subject tothe following two conditions (1) This device may
not cause harmful interference, and (2) This device
must accept any interference received, includinginterference that may cause undesired operation.
This product may be covered by one or more patents, US and international. Please visit our website
below for the listing of applicable patents. Pat.:www.caireinc.com/corporate/patents.
Aircraft Use:CAIRE confirms that this machine meets the Federal Aviation Administration (FAA) requirements(RTCA/DO-160, section 21, category M) for allphases of air travel.
Wireless Technology
Technology used

BluetoothConnection types

SSP, iAP2, GATT

Frequency

2402 to 2480 MHz

Max RF power output+4 dBmOperating range

10m (Class 2)

It is recommended that the machine is at a minimum
distance of 0.43” (1.1 cm) from the body during
operation. Not applicable to masks, air tubing, oraccessories.

Unique device identifier # 5.7.10
CSA Certification
Certified for both the U.S. and Canadian markets, to the applicable U.S.
and Canadian standards.
A S/NZS 4417.1, AS/NZS 4417.2
ACMA Regulatory ComplianceMark– indicates compliance with
ACMA regulations.

PN 21182453 G | User Manual US - 3

Eclipse 5
Quick Start Guide
1 Unpack Your Eclipse 5

2 Getting to Know Your Eclipse 5
Review all Warnings, Cautions and additional device information in the rest of this manual. Become
familiar with the key features of the Eclipse 5 andthe User Control Panel.

Eclipse 5

FRONT
HandleControl Panel

OxygenOutlet Port
FAA
Approval
Icon

Rating Label &Serial Number
Location

AC Power Supply

BACK
Universal Cart

Service Port
(Not forPatient Use)

Air Inlet
Filter

DC Power Supply
Cart
Attachment
Location

Power Cartridge (battery)

4 - US PN 21182453 G | User Manual

Exhaust Vent

Power Cartridge
Latch

Power
Cartridge(Battery)Eclipse 5
The Eclipse 5 User Control Panel displays important
operating information.

ON/OFF Button (Green) Indicator: Thisbutton powers the device ON or OFF. The
Green Indicator is illuminated when thedevice is ON and functioning properly.
Increase or Decrease Flow Setting
Buttons: Use these buttons to set the flow
to your prescribed setting.
Delivery Mode Button and Indicator: The
button toggles between Continuous Flowand Pulse Dose Mode. The Pulse Dose
Mode activates autoSAT Technology andallows a significant increase in the operating time while powered by the battery.
When the Pulse Dose Mode is activated,
the green Pulse Dose Mode Indicatorilluminates and a pulse of oxygen isdelivered with each inspiratory effort.
Low Priority Technical Alarm: Whenilluminated, this indicates a low priority
awareness condition. Continue to use
your system and refer to the trouble
shooting table for the proper response, or
contact your Home Care Provider.
Flow Setting Indicator: This is the main
focus on your control panel. Your homecare provider will correctly set yourprescribed flow for the Continuous Flow
Mode (LPM) and/or your Pulse Dose
Mode (mL) settings. Each time you power
the device ON, the previous mode and/or
setting has been saved and will be usedat start-up.
Power Cartridge (battery) Status Gauge:
This indicator displays the charge remaining in the battery. Each of the five horizontal gray bars represents approximately 20% of the total battery charge. Whenthe battery is being charged, the charge
indicator bars will blink in a waterfall-type
fashion. If the battery is not installed,
or if it is improperly installed, the Power
Cartridge (battery) Status Gauge will not
be illuminated or will flash on and off.
External Power Indicator: When the
Eclipse 5 is properly plugged in and isusing the AC or DC Power Supply, thisindicator will appear on the User Control
Panel.
Buzzer: An audible alarm (or buzzer) alerts you to theoperating condition of the device, either a warning or
failure, and confirms a valid key press.

3 Powering on the Eclipse 5
To connect to AC power source: Insert the AC
Power Supply cord securely into the ExternalPower Receptacle on the right side of the device.
If your power supply has three pins, plug the AC
power cord into a grounded AC outlet. If yourpower supply has a two pin plug, plug the AC cord
into a grounded or non-grounded AC outlet.
Step 2
Step 1

When the Eclipse 5 is plugged in properly, a green
indicator on the AC Power Supply will light andthe External Power Indicator, will appear on theuser control panel.
Install the Battery: Slide the battery into the empty
compartment until it locks into position and is flushwith the back of the device.

When the battery is properly installed and the
Eclipse 5 is ON, the Power Cartridge Status Gauge
will appear.

ON/OFF Button (Green) Indicator
Press and hold the “ON/OFF” button for two (2)seconds to power ON your Eclipse 5. A brief audible and visual power on self-test will occur. Verify
that all the indicators are illuminated and the buzzer sounds. Press and hold the “ON/OFF” button for
two (2) seconds to power OFF your Eclipse 5.

PN 21182453 G | User Manual US - 5

Eclipse 54 Select Flow Delivery Mode

6 Begin Using the Eclipse 5
Air Inlet Filter

Flow Mode Button and Indicator
Press the Flow Mode button to select your desired
flow mode. Pressing this button repeatedly will
toggle you back and forth between continuous
flow and pulse dose modes. In Continuous Flow
Mode the oxygen is provided at a constant flow rate
between 0.5 and 3.0 LPM. In Pulse Dose Mode, oxygen is supplied in a bolus at the beginning of eachinspiration, providing a selectable range of settings
1-9 with bolus sizes 16-192 mL. Pulse Dose can be
adjusted for sensitivity and breath bolus delivery.
Flow settings in both modes should be prescribedfor you by your physician.

5 Adjust Flow Setting to Prescribed Level
Increase or Decrease Flow Setting Button

Using the + or - buttons, adjust the setting to theprescription supplied by your physician.
Continuous Flow Mode Display

Your unit is equipped with an Air Inlet Filter (extra
filter provided). Check daily and clean once a week.
Connect the oxygen supply
tubing to the oxygen outlet andconnect the oxygen cannula
per the instructions provided
with the cannula, or per the
instructions from your homecare provider.
Breathe normally through the nasal cannula.

When operating on battery power, utilizing thePulse Dose Mode will achieve longer duration-of-use.

7 Universal Cart Operation
Align the locater holes on the bottom of the device
to the tabs on the Universal Cart. Align the large
threaded screw to the insert at the back of the
Eclipse 5. Then hand-tighten the knob on the cart
until the Universal Cart is tightly secured to thedevice. Depress the push-button on the telescoping
Universal Cart handle to adjust the height of thehandle.

Pulse Dose Mode Display

WARNING: IT IS VERY IMPORTANT TO
SELECT ONLY THE PRESCRIBED LEVEL OFOXYGEN. DO NOT CHANGE THE FLOW SELECTION UNLESS YOU HAVE BEEN DIRECTED TO DO SO BY A LICENSED CLINICIAN.
THE PORTABLE OXYGEN CONCENTRATOR
MAY BE USED DURING SLEEP UNDER
THE RECOMMENDATION OF A QUALIFIED
CLINICIAN. YOUR DELIVERY SETTINGSSHOULD BE PERIODICALLY REASSESSEDFOR EFFECTIVENESS OF THERAPY.

6 - US PN 21182453 G | User Manual

Contact your equipment provider for available
training options. If you do not know your provider,contact CAIRE.
Note: For assistance, if needed, in setting up, or maintaining
the Eclipse 5, or to report unexpected operation or events,
contact CAIRE or CAIRE Representative.

Eclipse 5
Important!

Safety Instructions are defined as follows:
WARNING: IMPORTANT SAFETY INFORMATION FOR HAZARDS THAT MIGHT
CAUSE SERIOUS INJURY.
CAUTION: Important information forpreventing damage to the Eclipse 5.
Note: Information needing special attention.

Indications for Use
Intended Use

The CAIRE Eclipse 5 Oxygen Concentrator isintended for the administration of supplemental oxygen. The device is not intended for life support nor
does it provide any patient monitoring capabilities.
WARNING: IT IS VERY IMPORTANT
TO SELECT ONLY THE PRESCRIBEDLEVEL OF OXYGEN. DO NOT CHANGE
THE FLOW SELECTION UNLESS YOUHAVE BEEN DIRECTED TO DO SO BY A
LICENSED CLINICIAN. THE PORTABLEOXYGEN CONCENTRATOR MAY BE
USED DURING SLEEP UNDER THE
RECOMMENDATION OF A QUALIFIED
CLINICIAN.
WARNING: TO ENSURE RECEIVING THE
THERAPEUTIC AMOUNT OF OXYGEN DELIVERYACCORDING TO YOUR MEDICAL CONDITION,ECLIPSE 5 MUST BE USED ONLY AFTER ONE OR
MORE SETTINGS HAVE BEEN INDIVIDUALLY DETERMINED OR PRESCRIBED FOR YOU AT YOURSPECIFIC ACTIVITY LEVELS. ECLIPSE 5 MUST
BE USED WITH THE SPECIFIC COMBINATION OR
PARTS AND ACCESSORIES THAT ARE IN LINE
WITH THE SPECIFICATION OF THE CONCENTRATOR MANUFACTURER AND THAT WERE USED
WHILE YOUR SETTINGS WERE DETERMINED.
WARNING: FEDERAL (USA) LAW RESTRICTS THIS
DEVICE TO SALE OR RENTAL BY ORDER OF APHYSICIAN OR OTHER LICENSED HEALTH CAREPROVIDER.
WARNING: THIS UNIT IS NOT TO BE USED FOR
LIFE SUPPORT. GERIATRIC, PEDIATRIC, OR ANYOTHER PATIENT UNABLE TO COMMUNICATEDISCOMFORT CAN REQUIRE ADDITIONAL MONITORING AND/OR A DISTRIBUTED ALARM SYSTEMTO CONVEY THE INFORMATION ABOUT THE
DISCOMFORT AND OR THE MEDICAL URGENCY
TO THE RESPONSIBLE CARE GIVER TO AVOID
HARM. PATIENTS WITH HEARING AND/OR SIGHT
IMPAIRMENT(S) MAY NEED ASSISTANCE WITH
MONITORING ALARMS.

Contraindications for Use
WARNING: IN CERTAIN CIRCUMSTANCES, THE USE OF NON-PRESCRIBEDOXYGEN CAN BE HAZARDOUS. THIS
DEVICE SHOULD ONLY BE USED WHEN
PRESCRIBED BY A PHYSICIAN.
WARNING: NOT FOR USE IN THE PRESENCE OF
FLAMMABLE ANESTHETICS.
WARNING: AS WITH ANY ELECTRICALLYPOWERED DEVICE, THE USER MAY EXPERIENCEPERIODS OF NON-OPERATION AS A RESULT OF
ELECTRICAL POWER INTERRUPTION, OR THE
NEED TO HAVE THE ECLIPSE 5 SERVICED BY AQUALIFIED TECHNICIAN. THE ECLIPSE 5 IS NOTAPPROPRIATE FOR ANY PATIENT WHO WOULD
EXPERIENCE ADVERSE HEALTH CONSEQUENCES AS THE RESULT OF SUCH TEMPORARY
INTERRUPTION.

Safety Guidelines
WARNING: NO MODIFICATION OF THISEQUIPMENT IS PERMITTEDWARNING: THE MANUFACTURER RECOMMENDS
AN ALTERNATE SOURCE OF SUPPLEMENTALOXYGEN IN THE EVENT OF A POWER OUTAGE,
ALARM CONDITION, OR MECHANICAL FAILURE.
CONSULT YOUR PHYSICIAN OR EQUIPMENTPROVIDER FOR THE TYPE OF RESERVE SYSTEMREQUIRED.
WARNING: THIS DEVICE SUPPLIES HIGH-CONCENTRATION OXYGEN THAT PROMOTES RAPID
BURNING. DO NOT ALLOW SMOKING OR OPEN
FLAMES WITHIN TWO (2) METERS OF (1) THIS
DEVICE, OR (2) ANY OXYGEN-CARRYING ACCESSORY. FAILURE TO OBSERVE THIS WARNING CANRESULT IN SEVERE FIRE, PROPERTY DAMAGE,
AND/OR CAUSE PHYSICAL INJURY OR DEATH.
WARNING: DO NOT OPERATE UNIT IN A RESTRICTED OR CONFINED SPACE (I.E., A SMALL
CASE OR HANDBAG) WHERE VENTILATION CAN
BE LIMITED. THIS CAN CAUSE THE OXYGENCONCENTRATOR TO OVERHEAT AND IMPAIRPERFORMANCE.

PN 21182453 G | User Manual US - 7

Eclipse 5
WARNING: THE CONCENTRATORSHOULD BE LOCATED AS TO AVOIDSMOKE, POLLUTANTS OR FUMES.
WARNING: THE USE OF SOME OXYGENADMINISTRATING ACCESSORIES NOT SPECIFIEDFOR THIS OXYGEN CONCENTRATOR MAY
IMPAIR ITS PERFORMANCE. RECOMMENDEDACCESSORIES ARE REFERENCED WITHIN THISMANUAL.
WARNING: IF THE OXYGEN CONCENTRATOR HAS
BEEN DROPPED, DAMAGED OR EXPOSED TO
WATER PLEASE CONTACT YOUR HOME CAREPROVIDER FOR INSPECTION OR POSSIBLE
REPAIR OF THE DEVICE. DO NOT USE THE
OXYGEN CONCENTRATOR IF IT HAS A DAMAGEDPOWER CORD OR PLUG.
WARNING: PROPERLY SECURED, BELTOR OTHERWISE RESTRAIN THE OXYGENCONCENTRATOR WHEN IN A VEHICLE DURING
TRANSPORT TO PREVENT DAMAGE OR INJURY.
WARNING: DO NOT ALLOW EITHER THE AIRINTAKE OR THE AIR OUTLET VENTS TO BECOMEBLOCKED. DO NOT DROP OR INSERT ANYOBJECTS INTO ANY OPENINGS ON THE DEVICE.
THIS CAN CAUSE THE OXYGEN CONCENTRATORTO OVERHEAT AND IMPAIR PERFORMANCE.
WARNING: DO NOT OVERFILL THE OPTIONAL
HUMIDIFIER. FILL THE OPTIONAL HUMIDIFIERWITH WATER ONLY TO THE LEVEL SHOWN BY
THE MANUFACTURER OF THE HUMIDIFIER.
WARNING: DO NOT OPERATE THE ECLIPSE 5 AND
AMBULATE WHILE A HUMIDIFIER IS ATTACHED.
REMOVE THE HUMIDIFIER BOTTLE BEFORE
WALKING. DO NOT LAY THE ECLIPSE 5 DOWN
WHILE ATTACHED TO A HUMIDIFIER BOTTLE.
WARNING: THE US DEPARTMENT OF TRANSPORTATION (DOT) AND UNITED NATIONS (UN)REGULATIONS REQUIRE THE REMOVAL OF THE
BATTERY FROM THE DEVICE FOR ALL INTERNATIONAL AIRLINE TRAVEL WHEN THE OXYGENCONCENTRATOR IS CHECKED AS LUGGAGE.
WHEN SHIPPING THE OXYGEN CONCENTRATOR,
THE BATTERY MUST ALSO BE REMOVED FROM
THE DEVICE AND PACKAGED PROPERLY.
WARNING: DO NOT OPERATE THIS EQUIPMENTWITHOUT FIRST READING AND UNDERSTANDING
THIS MANUAL. IF YOU ARE UNABLE TO UNDERSTAND THE WARNINGS AND INSTRUCTIONS,CONTACT YOUR EQUIPMENT PROVIDER BEFOREATTEMPTING TO USE THIS EQUIPMENT; OTHERWISE, INJURY OR DAMAGE MAY RESULT.

8 - US PN 21182453 G | User Manual

WARNING: IF YOU FEEL DISCOMFORT OR ARE
EXPERIENCING A MEDICAL EMERGENCY, SEEKMEDICAL ASSISTANCE IMMEDIATELY.
WARNING: OPERATING THE OXYGEN CONCENTRATOR OUTSIDE OF THE OPERATIONALTEMPERATURE SPECIFICATIONS CAN LIMIT THE
CONCENTRATOR’S ABILITY TO MEET OXYGENCONCENTRATION SPECIFICATION. REFER TO THE
SPECIFICATION SECTION OF THIS MANUAL FORTEMPERATURE LIMITS.
WARNING: USE NO OIL, GREASE, OR
PETROLEUM-BASED OR OTHER FLAMMABLE
PRODUCTS WITH THE OXYGEN-CARRYINGACCESSORIES OR THE OXYGENCONCENTRATOR. ONLY WATER BASED,OXYGEN COMPATIBLE LOTIONS OR SALVESSHOULD BE USED. OXYGEN ACCELERATES THE
COMBUSTION OF FLAMMABLE SUBSTANCES.
WARNING: THE INCORRECT USE OF THE
BATTERY CAN CAUSE THE BATTERY TO GET HOT,IGNITE, AND MAY CAUSE SERIOUS INJURY. BE
SURE NOT TO PIERCE, STRIKE, STEP ON, DROP
THE BATTERY, OR OTHERWISE SUBJECT THE
BATTERY TO STRONG IMPACTS OR SHOCKS.
THE USE OF A DAMAGED BATTERY MAY CAUSEPERSONAL INJURY.
WARNING: THE OXYGEN CONCENTRATORSHOULD NOT BE USED ADJACENT TO OR
STACKED WITH OTHER EQUIPMENT. IF
ADJACENT OR STACKED USE IS UNAVOIDABLE,
THE DEVICE SHOULD BE OBSERVED TO VERIFYNORMAL OPERATION.
WARNING: USE OF ACCESSORIES, TRANSDUCERS AND CABLES OTHER THAN THOSE SPECIFIED OR PROVIDED BY THE MANUFACTURER OF
THIS EQUIPMENT COULD RESULT IN INCREASEDELECTROMAGNETIC EMISSIONS OR DECREASEDELECTROMAGNETIC IMMUNITY OF THIS EQUIPMENT AND RESULT IN IMPROPER OPERATION.
WARNING: THIS PRODUCT CAN EXPOSE YOU
TO CHEMICALS INCLUDING NICKEL, WHICH
IS KNOWN TO THE STATE OF CALIFORNIA TO
CAUSE CANCER. FOR MORE INFORMATION, GO
TO WWW.P65WARNINGS.CA.GOV.

Eclipse 5
WARNING: USE ONLY SPARE PARTSRECOMMENDED BY THE MANUFACTURER TO ENSURE PROPER FUNCTION AND TO AVOID THE RISK OF FIRE
AND BURNS.
WARNING: DO NOT LUBRICATE
FITTINGS, CONNECTIONS, TUBINGOR OTHER ACCESSORIES OF THE
OXYGEN CONCENTRATOR TO AVOIDTHE RISK OF FIRE AND BURNS.
WARNING: DO NOT ATTEMPT ANY
MAINTENANCE OTHER THAN THE POSSIBLE SOLUTIONS LISTED WITHIN THISMANUAL. DO NOT REMOVE COVERS,ONLY YOUR EQUIPMENT PROVIDER
OR A QUALIFIED SERVICE TECHNICIANSHOULD REMOVE THE COVERS ORSERVICE THIS DEVICE.
WARNING: ONLY ACCESSORIES RECOMMENDED
BY THE MANUFACTURER. USE OF ANY OTHER
MAY BE HAZARDOUS, CAUSE SERIOUS DAMAGE
TO YOUR OXYGEN CONCENTRATOR AND WILL
VOID THE WARRANTY.
WARNING: THE MANUFACTURER RECOMMENDS AN ALTERNATE SOURCEOF SUPPLEMENTAL OXYGEN IN THE
EVENT OF A POWER OUTAGE, ALARM
CONDITION, OR MECHANICAL FAILURE. CONSULT YOUR PHYSICIAN OR
EQUIPMENT PROVIDER FOR THE TYPE
OF RESERVE SYSTEM REQUIRED.
WARNING: IN THE EVENT THERE IS A SERIOUS
INCIDENT OCCURRING WITH THIS DEVICE, THE
USER SHOULD IMMEDIATELY REPORT THE
INCIDENT TO THE PROVIDER AND/OR THE MANUFACTURER. A SERIOUS INCIDENT IS DEFINEDAS AN INJURY, DEATH, OR POTENTIAL TO CAUSE
INJURY/DEATH SHOULD THERE BE A REOCCURRENCE OF THE INCIDENT. THE USER CAN ALSOREPORT THE INCIDENT TO THE COMPETENT AUTHORITY IN THE COUNTRY WHERE THE INCIDENTOCCURRED.
WARNING: SMOKING WHILE USING OXYGEN IS THE NUMBER ONE CAUSE OF
FIRE INJURIES AND RELATED DEATHS.
YOU MUST FOLLOW THESE SAFETYWARNINGS:WARNING: DO NOT ALLOW SMOKING, CANDLES,OR OPEN FLAMES IN THE SAME ROOM WITH THE
DEVICE OR THE OXYGEN-CARRYING ACCESSORIES.
WARNING: SMOKING WHILE WEARING AN OXYGEN CANNULA MAY CAUSE FACIAL BURNS AND
POSSIBLY DEATH.
WARNING: IF YOU SMOKE, THESE 3 STEPSMAY SAVE YOUR LIFE: TURN OFF THE OXYGENCONCENTRATOR, TAKE OFF THE CANNULA,
AND LEAVE THE ROOM WHERE THIS DEVICEIS LOCATED. IF UNABLE TO LEAVE THE ROOM,YOU MUST WAIT 10 MINUTES AFTER YOU HAVETURNED THE OXYGEN CONCENTRATOR OFF
BEFORE SMOKING.
WARNING: “NO SMOKING – OXYGEN IN USE”
SIGNS MUST BE PROMINENTLY DISPLAYEDIN THE HOME, OR WHERE OXYGEN IS IN USE.
PATIENTS AND THEIR CAREGIVERS MUST BE INFORMED ABOUT THE DANGERS OF SMOKING IN
THE PRESENCE OF, OR WHILE USING, MEDICALOXYGEN.
WARNING: DO NOT USE YOUR OXYGEN CONCENTRATOR IN THE PRESENCE OF FLAMMABLE
GASES. THIS CAN RESULT IN RAPID BURNINGCAUSING PROPERTY DAMAGE, BODILY INJURIESOR DEATH.
WARNING: REMOVING THE CANNULA AND
PUTTING IT ON CLOTHING, BEDDING, SOFAS,OR OTHER CUSHION MATERIAL WILL CAUSE A
FLASH FIRE WHEN EXPOSED TO A CIGARETTE,HEAT SOURCE, OR FLAME.
WARNING: DO NOT LEAVE A NASAL CANNULA
ON OR UNDER CLOTHING, BED COVERINGS OR
CHAIR CUSHIONS. IF THE UNIT IS TURNED ON
BUT NOT IN USE, THE OXYGEN WILL MAKE THE
MATERIAL FLAMMABLE. SET THE I/0 POWER
SWITCH TO THE 0 (OFF) POSITION WHEN THE
OXYGEN CONCENTRATOR IS NOT IN USE.
WARNING: PATIENT OR OTHERS MAY BE ENTANGLED IN CANNULA OR OTHER TUBING CAUSINGASPHYXIATION.
WARNING: DO NOT BLOCK ACCESS TO THE AC
POWER CORD, IN THE EVENT OF AN
EMERGENCY.
WARNING: ACCESSORIES MAYCONTAIN PHTHALATES. CHILDREN,
PREGNANT OR NURSING WOMENSHOULD NOT USE ACCESSORIES THATCONTAIN PHTHALATES.
PN 21182453 G | User Manual US - 9

Eclipse 5
CAUTION: Always disconnect AC power
supply from the wall before disconnection the AC power supply from theoxygen concentrator.
CAUTION: Always place oxygen supply tubing
and power cords in a manner that prevents a trip
hazard.
CAUTION: When using the Oxygen Concentrator in
an automobile, boat, or on other DC sources with
the DC power supply, make sure that the vehicle isstarted and running before connecting the Oxygen
Concentrator. If the DC power supply does notilluminate and requires resetting, disconnect the
DC power supply from the DC outlet, restart yourvehicle, and then reconnect your DC power supply
into the DC outlet. Failure to follow these instructions can result in the power supply not supplying
power to the Oxygen Concentrator.
CAUTION: When the automobile in which you
are using the Oxygen Concentrator unit is turnedoff, disconnect and remove the device from theautomobile. Do not store the Oxygen Concentrator
in a very hot or cold automobile or in other similar,
high-or low-temperature environments.
CAUTION: In the event of an alarm or
you observe the Oxygen Concentrator is
not working properly; consult the Troubleshooting section in this manual. If
you cannot resolve the problem, consult
your Equipment Provider.
CAUTION: To prevent a void warranty,
follow all manufacturers’ instructions.
Note: Portable and mobile RF communications equipment can effect medical electrical equipment.

What is the Oxygen
Concentrator
The air we breathe contains approximately 21%
oxygen, 78% nitrogen, and 1% other gasses. In
the Eclipse 5, room air is drawn into the machine
through the air intakes. It then passes through anadsorbent material called molecular sieve. This
material separates the oxygen from the nitrogen andallows only the oxygen to pass through. The result
is a flow of high-concentration oxygen delivered tothe user.
The Eclipse 5 combines advanced oxygen concentrator and oxygen conserving technologies tocreate a portable device that allows for maximum
portability and ambulation. The advanced pulse
flow delivery quickly senses when the user breathes
in and delivers a pulse of oxygen at the beginningof each inhalation. Pulse flow delivery is ideal for
activities and time away from home. It conserves
oxygen and extends battery life as opposed to continuous flow. The Eclipse 5 is lightweight enoughto be carried via handle or cart. It can be operated
on AC, DC (vehicle) or battery power.
The Essential Performance of the Eclipse 5 is toprovide oxygen concentrations between 82% to
96% at the prescribed flow setting. If Essential
Performance cannot be attained the Eclipse 5 Low
Priority Technical Alert will be activated. If the
Eclipse 5 is exposed to an extreme electromagnetic
disturbance the device may shut down without
activating the alert.
The Eclipse Oxygen System is intended for theadministration of supplemental oxygen to thepatient continuously in the end user’s home or for
their portable needs outside the home and can also
be used in institutions such as nursing homes or
sub-acute care facilities. The device is not intended
for life support nor does it provide any patientmonitoring capabilities. The instruction manual ofthe device recommends an alternate source of supplemental oxygen in the event of a power outage,alarm condition or mechanical failure.
The device is used by COPD patients or thosewith diminished breathing capacity. The device
is prescribed to the patient. Typically the device
is sold to a provider that is trained to operate andservice the Eclipse 5. The provider trains the user.
Those devices purchased through CAIRE’s website
receive setup instructions and the User Manual thatdefines setup and usage instructions.

10 - US PN 21182453 G | User ManualEclipse 5
Recommended Operating Environments
For proper use of the device, the following chart
provides important information concerning therecommended operating environments, or operating
conditions.

Specifications
OperatingTemperature
Operating
Humidity
Transport/
Storage
Temperature

50º F to 104º F (10º C to 40º C)
10% - 95% at an 82.4° F (28° C)
dew point-4º F to 140º F (-20º C to 60º C)

Humidity: Up to 95% Non-Condensing
Electrical
Use no extension cords. Use noelectrical outlets controlled by aswitch.
Altitude0 – 13,123 feet (0 - 4,000 meters)
Placement
DO NOT block the air inlet (top backcover) or the exhaust vent (backbottom left). Place device a minimumof 3 inches (7.5 cm) away from walls,
draperies, furniture, etc.
Environment Must be smoke, pollutant, and fume
free.
Operating Time 24-hours a day when connected toan external AC or DC power source.
Oxygen90%+5.5%/-3% @ sea level
Concentration
Dimensions 19.3 high x 12.3 wide x 7.1 deep(inches), 49.0 height x 31.2 wide x
18.0 deep (CM)WeightEclipse 15 lbs., Battery 3.4 lbs.
Power
AC Power (100-240 VAC, 50-60 Hz;
DC Power (12V nominal);
Battery (Lithium Ion)
Nominal
1.8 to 5.0 hrs recharge time toBattery
achieve 80% capacity (dependent
Recharge Time upon the flow)
Alarms/Alerts Loss of power, Low Battery, Low
Therapeutic O2 Output, O2 flow
outside normal limits. No Inspiration
detected in Pulse Dose Mode, Unit
Malfunction
O2
Green Light=Normal
Concentration Yellow=Caution <85%
Indicator
OutletNominal: 7.0 psig
Pressure
Maximum: 14.0 psig

Sound Level
Applied Part

48 dBA at 3.0 LPM Continuous Flow
Mode; 40 dBA at 3.0 Pulse Dose
Setting; 59 dBA while alarming
CannulaThe expected service life is a minimum of five years.

WARNING: USE OF THIS DEVICE AT AN
ALTITUDE ABOVE 13,123 FEET (4,000
METERS) OR OUTSIDE TEMPERATURE
OF 50º F TO 104º F (10º C TO 40º C) OR
A RELATIVE HUMIDITY ABOVE 95% IS
EXPECTED TO ADVERSELY AFFECTTHE FLOWRATE AND THE PERCENTAGE OF OXYGEN AND CONSEQUENTLY
THE QUALITY OF THE THERAPY.
Note: If the Oxygen Concentrator has been stored forand extended period of time outside its normal operating temperature range, the unit should be allowed
to return to normal operating temperature before use.
The recommended time is 3 hours.

Available Settings

The Eclipse 5, including the cannula and otheraccessories, are specified for use at the following
specific flows.
Continuous Flow
Settings in LPM
0.5
1.0
1.5
2.0
2.5
3.0
-

Pulse Dose Setting Bolus
Size (mL)
16
24
32
40
48
56
64
72
80
88
96
128
160
192

Minimum oxygen concentration (82%).

PN 21182453 G | User Manual US - 11

Eclipse 5
Proper Placement of the Eclipse 5

Select a location for the device that avoids theintake of smoke, fumes and pollutants. Correct
placement of the device should allow intake of air
through the air inlet filter at the top rear of the cabinet and allow exhaust air to freely leave the exhaustvent at the bottom left of the device.
Place the device such that the alarms can be heard.
Position the oxygen supply tubing in such a waythat it does not kink or occlude.
WARNING: DO NOT USE YOUR OXYGENCONCENTRATOR IN THE PRESENCE
OF FLAMMABLE GASES. THIS CANRESULT IN RAPID BURNING CAUSINGPROPERTY DAMAGE, BODILY INJURIESOR DEATH. USE NO OIL, GREASE,OR PETROLEUM-BASED OR OTHER
FLAMMABLE PRODUCTS WITH THE
OXYGEN-CARRYING ACCESSORIES OR
THE OXYGEN CONCENTRATOR. ONLYWATER BASED, OXYGEN COMPATIBLE LOTIONS OR SALVES SHOULD
BE USED. OXYGEN ACCELERATESTHE COMBUSTION OF FLAMMABLE
SUBSTANCES.
WARNING: THIS DEVICE SUPPLIES HIGH-CONCENTRATION OXYGEN THAT PROMOTES RAPID
BURNING. DO NOT ALLOW SMOKING OR OPEN
FLAMES WITHIN THE SAME ROOM OF (1) THIS
DEVICE, OR (2) ANY OXYGEN-CARRYING ACCESSORY. FAILURE TO OBSERVE THIS WARNING CANRESULT IN SEVERE FIRE, PROPERTY DAMAGE,
AND/OR CAUSE PHYSICAL INJURY OR DEATH.
Note: When the automobile in which you are using
the Oxygen Concentrator unit is turned off, disconnect
and remove the device from the automobile. Do notstore the Oxygen Concentrator in a very hot or cold
automobile or in other similar, high-or low-temperatureenvironments. DO NOT leave the Oxygen Concentrator or the Power Supply plugged into the vehicle if theignition is in the off position. Doing so may drain thevehicle’s battery.
CAUTION: Do not operate unit in a
restricted or confined space (i.e., a small
case or handbag) where ventilation canbe limited. This can cause the Oxygen
Concentrator to overheat and impair
performance. Do not allow either the air
intake or the air outlet vents to become
blocked. This can cause the Oxygen
Concentrator to overheat and impair
performance.
12 - US PN 21182453 G | User Manual

Operating Instructions
Before OperatingThis users manual serves as your reference to help
you operate and maintain the device. If you haveany questions or concerns please call your homecare provider.
Important! DO NOT attempt to operate the Eclipse
5 without first reading the Safety Guidelines sectionof this manual. Please follow all of the operating
instructions. Please observe all Warnings on thedevice and in the Users Manual. In order to reduce
the risk of fire, personal injury and serious damage
to the Eclipse 5, please observe all of the safety
precautions.
WARNING: CLEAN THE CABINET, CONTROL PANEL, AND POWER CORD ONLY
WITH A MILD HOUSEHOLD CLEANERAPPLIED WITH A DAMP (NOT WET)
CLOTH OR SPONGE, AND THEN WIPE
ALL SURFACES DRY. DO NOT ALLOWANY LIQUID TO GET INSIDE THE
DEVICE. PAY SPECIAL ATTENTION TO
THE OXYGEN OUTLET FOR THE CANNULA CONNECTION TO MAKE SURE IT
REMAINS FREE OF DUST, WATER, AND
PARTICLES.
CAUTION: Do not allow either the air
intake or the air outlet vents to become
blocked. This can cause the Oxygen
Concentrator to overheat and impair
performance. Do not operate the Oxygen
Concentrator without the air intake filterin place. If a second filter is provided,
insert the “replacement” filter before you
clean the dirty filter. Clean the dirty filter
in a warm soap and water solution thendry thoroughly prior to use.

Eclipse 5Powering on your Eclipse 5

AC POWER SUPPLY WITH TWO PINS

The device is capable of being
operated directly from three (3)different power sources:• AC Power Supply with
NEMA Power Cord
• DC Power Supply• Rechargeable Power
Cartridge (Battery)

GeneralCharging
Information

The Eclipse
5 charges thebattery while
using AC power and when the battery temperature
is within safe charging temperature. When external
power is disconnected, the device will automatically switch over to the battery if the battery is
installed and charged. When external power isrestored by plugging into an AC outlet, the device
will accept power from the external power source
and replenish the battery. It is not mandatory
for the battery to be installed for the Eclipse 5 tooperate.
Proper connection of the AC or DC Power Supply
requires aligning the power plug with the recessed
external power receptacle of the device. The
recessed external power receptacle is located on
the right side of the device when facing the Control
Panel. Push the power plug into the recessed
receptacle.
WARNING: THE ECLIPSE BATTERYCARTRIDGE AND DC POWER SUPPLYMUST BE PERIODICALLY CHECKED.
WHEN AC POWER IS NOT AVAILABLE
THESE ALTERNATE POWER SOURCES
MAY BE USED.

AC Two-Pin Power Supply
Item# 21334759

The Eclipse 5 includes a universal AC Power
Supply for use at home or wherever standard AC
power is available. To connect to an AC powersource, be sure the AC Power Supply cord issecurely inserted into the recessed power receptacleon the side of the device and that the cord from the
AC Power Supply is plugged into an AC outlet.
When the Eclipse 5 is plugged in properly, a green
indicator on the AC Power Supply will light and theExternal Power Indicator,
will appear on theuser control panel.

Step 1

Step 2

Note: Use only electrical voltage specified on thespecification label affixed to the device.
CAUTION: Use only accessories recommended by the manufacturer. Use of any
other may be hazardous, cause serious
damage to your oxygen concentrator
and will void the warranty.
CAUTION: Do not use extension cords with thisunit or connect too many plugs into the same
electrical outlet. The use of extension cords could
adversely affect the performance of the device. Toomany plugs into one outlet can result in an overload to the electrical panel causing the breaker/
fuse to activate or fire if the breaker or fuse fails to
operate.
CAUTION: Use of cables and adapters other thanthose specified, with the exception of cables andadapters sold by the manufacturer of the medical
electrical equipment as replacement parts for internal components, may result in increased emissionsof decreased immunity of the Oxygen Concentrator.

PN 21182453 G | User Manual US - 13

Eclipse 5
DC POWER SUPPLYDC Power Supply
Item# 5942-SEQ

A DC Power Supply allows the system to operatefrom DC outlets, such as those found in motor
vehicles.
1. Start your vehicle.
2. Insert the DC cord into the recessed power
receptacle on the side of the device.
3. Insert the DC Power Supply plug into the
DC power outlet in your motor vehicle.
4. Start the unit on the lowest flow setting
and slowly increase the flow setting to theprescribed flow rate. (Starting the unit at a
setting of 3 LPM on DC power can cause atemporary overload of the power supply).
When the device is properly connected andreceiving power from the DC power source,agreen indicator light on the Power Supply and theExternal Power Indicatoron the control panel
will illuminate.
Eclipse 5 DC Supply Connection
When the Eclipse 5 is connected to the DC supply
in a vehicle, the following applies:• The Eclipse 5 is able to run at all continuous flow
settings up to 3.0LPM and all pulse flow settings
up to 9.0 (192 mL) while being operated with
the DC supply in a car.*• The Eclipse 5 is able to charge the battery while
on DC Supply in a car while running on continuous flow at 2LPM or less, and at all pulse
settings.*• If the Eclipse 5 battery is being charged, thebattery symbol will waterfall. If the Eclipse 5battery is not charging, the battery symbol will
be displayed solid.
*Capabilities of the unit on DC power are dependent on the vehicle electrical system. Somefeatures or flow settings may not be available
depending on the power available from thevehicle’s DC power outlet.
WARNING: DO NOT ALLOW EITHER
THE AIR INTAKE OR THE AIR OUTLETVENTS TO BECOME BLOCKED. THISCAN CAUSE THE OXYGEN CONCENTRATOR TO OVERHEAT AND IMPAIRPERFORMANCE.
14 - US PN 21182453 G | User Manual

WARNING: IF THE VEHICLE’S DC POWER SOURCEDROPS BELOW 11.5 VOLTS, THE ECLIPSE 5 WILL
REVERT TO THE BATTERY OPERATION (IF THE
BATTERY IS PRESENT AND CHARGED). (THE
EXTERNAL POWER INDICATOR MAY REMAIN ON.)CAUTION: The DC Power Supply isdesigned for 12VDC minimum (150
Watt) vehicle electrical systems. Do not
attempt to operate with 6V, 24V, or othervehicle electrical system. Device performance may be impacted by operating on
a 120 Watt DC system.
Note: While operating the device from a DC power
supply, the Battery may not charge if the DC powersource does not supply sufficient power. The Eclipse
will dedicate power to running the device first. Any
leftover power is used to supply a trickle charge tothe battery.
Note: To ensure that the device is utilizing the DC
power, check that the External Power Indicator is illuminated. If a beep is heard, this could indicate a lossof external power. Check if the green Battery Power
Verification LED is on or off. If the battery is removed
on DC power, it will ensure no loss of charge while theunit is running. If the indicator is not illuminated, thebattery is powering the device.
Settings Available When Operating on DC for Eclipse 5
Flow Setting
Bolus Size
Continuous
Pulse
(LPM)
(mL)
0.5
Yes
1.0
Yes
16
Yes
1.5
Yes
24
Yes
2.0
Yes
32
Yes
2.5
Yes
40
Yes
3.0
Yes
48
Yes
56
Yes
64
Yes
72
Yes
80
Yes
88
Yes
96
Yes
128
Yes
160
Yes
192
Yes
WARNING: ACTUAL PERFORMANCE OF
THE ECLIPSE 5 MAY VARY DEPENDING
UPON USAGE CONDITIONS OF THE MOTOR VEHICLE AND THE PERFORMANCE
SPECIFICATIONS OR CONDITION OF ITSELECTRICAL SYSTEM.

Eclipse 5
Rechargeable Power Cartridge (Battery)

Rechargeable Power
Cartridge (Battery)
Item#7082-SEQ

The Eclipse 5 can be powered by the rechargeable
Power Cartridge (Battery), which is supplied withthe device.
Installing the Battery: Align the Battery so that it
will slide into the empty Power Compartment. Push
the Battery into the empty Power Compartment
until it locks (clicks) into position and is flush withthe back of the device. When the Battery is properly
installed and the Eclipse 5 is ON, the Battery Status
Gauge will appear on the Control Panel. (If you DO
NOT have a Battery installed, the Power Cartridge
Status Gauge will not illuminate). The unit will still
operate on external power.

To remove the Battery, pull down on the black release lever. While holding DOWN the release lever,pull the Battery away from the Eclipse 5.
Battery Operating Times: The duration-of-use of
the Battery is displayed at the top of the Control
Panel. A variety of factors, such as flow setting,Pulse or Continuous Flow Mode, and breath rate
will impact the operating time. The table Typical
New Power Cartridge Operating Times provides
operating time estimates for the Eclipse 5 using a
new, fully charged Battery based on flow settingsand operating conditions as indicated.

WARNING: FOR OXYGEN CONCENTRATORS EQUIPPED WITH BATTERIES:STORE IN A COOL AND DRY LOCATIONTO HELP ENSURE THE LONGEVITY OF
YOUR BATTERY. STORING YOUR OXYGEN CONCENTRATOR FOR EXTENDEDPERIODS OF TIME AT HIGH TEMPERATURES OR WITH A FULLY CHARGED ORCOMPLETELY DISCHARGED BATTERYCAN DEGRADE ITS OVERALL BATTERY
LIFE. DO NOT ATTEMPT TO OPEN THE
BATTERY; THERE ARE NO SERVICEABLE PARTS INSIDE THE BATTERY.
KEEP BATTERIES AWAY FROMCHILDREN.
Initial Battery Charging
The new Battery supplied with your Eclipse 5 is not
fully charged when it is shipped from the factory.
Before using your Eclipse 5 for the first time, youmust first fully charge the Battery.
With the AC Power Supply plugged in, and theBattery correctly installed in the Power Compartment of the device, allow the Battery to completely
charge. The Battery is fully charged when the Power Cartridge Status Gauge, on the Control Panel, is
no longer cascading.
The initial battery charge can occur with the
Eclipse either on or off and connected to AC or
DC power. The Eclipse can be used while either
discharging or recharging the Battery. It may takebetween 2 and 5 hours, dependent upon the flow
setting, to achieve 80% capacity from a fullydischarged Battery. The battery will also recharge
when the Eclipse is plugged into AC or DC power
and turned off.
CAUTION: When the automobile inwhich you are using the Oxygen
Concentrator unit is turned off, disconnect and remove the device from theautomobile. Do not store the Oxygen
Concentrator in a very hot or cold
automobile or in other similar, high-or
low-temperature environments. DO NOT
leave the Oxygen Concentrator or thePower Supply plugged into the vehicle if
the ignition is in the OFF position. Doingso may drain the vehicle’s battery.
CAUTION: Only use the Manufacturer’s provided
batteries. For proper battery disposal, contact
your Equipment Provider or your local government
agency for disposal requirements.
PN 21182453 G | User Manual US - 15

Eclipse 5
Typical Battery Recharge Time

* Batteries are recyclable. *In the event of an AC or DC power interruption,
the Eclipse 5 will beep and automatically switch toBattery operation, as long as the Battery is present
and charged. When AC power is restored, theEclipse 5 will beep and the Battery will automatically start recharging, if not fully charged or
too warm. If the Battery is not present, or fullydischarged, during an AC power interruption, theEclipse 5 will shut down and alarm. The Battery is
fully discharged when the Power Cartridge Status
Gauge on the control panel is no longer illuminated. The device will display a yellow light andsound a beep for 120 seconds. The yellow light willremain on and the beep will repeat after 20 seconds
until the device completely shuts down.
Note: Press and hold the Eclipse 5 ON/OFF button
for a minimum of 5 seconds, to silence the Loss ofPower alarm. Connecting the Eclipse 5 to an external
AC power supply for 2 minutes will also eliminate thisalarm.
Typical New Power Cartridge Operating Times
Continuous
Flow

Battery
Duration

Pulse
Setting

0.5 LPM
1.0 LPM
2.0 LPM
3.0 LPM4.4 hours
3.7 hours
2.0 hours
1.3 hours

16mL–1.0
32mL–2.0
48mL–3.0
64mL–4.0
80mL–5.0
96mL–6.0
128mL–7160mL–8
192mL–9

Battery
Duration(12 BPM)5.4 hours
5.1 hours
4.9 hours
4.0 hours
3.7 hours
3.5 hours
2.5 hours
2.0 hours
1.7 hours

Note: Operating the Oxygen Concentrator outsideof its normal operating temperature range canaffect performance and decrease battery run time
and/or increase battery charge time. (Refer to theSpecifications section in this manual.) You will achieve
longer operating time on the Battery if you operate
your Eclipse 5 in the Pulse Mode (Physician approvedmode).

16 - US PN 21182453 G | User Manual

The typical time to recharge your Battery toachieve 80% capacity from a fully discharged
Battery is between 2 and 5 hours, depending upon
the device flow setting. When the battery is fully
charged, the battery gauge will cease cascading.
The battery can be charged in the unit when the
Eclipse 5 is turned off and plugged in.
If the Battery becomes too warm during discharging, recharging will not begin until the Battery sufficiently cools. Removing the Battery and allowing
it to cool may expedite this cooling process.
When operating the Eclipse 5 on DC power, thebattery will recharge while the unit is running at
2.0LPM or less if sufficient power is available.
Typical Power Cartridge Operating Times at 80%
ChargeContinuous
Flow

Battery
Duration(12 BPM)

Pulse
Setting

Battery
Duration(12 BPM)

0.5 LPM4.0 hours

16mL–1.0

3.0 hours

1.0 LPM3.5 hours

24mL–1.5

3.0 hours

1.5 LPM2.5 hours

32mL–2.0

3.0 hours

2.0 LPM2.0 hours

40mL–2.5

3.0 hours

2.5 LPM1.5 hours

48mL–3.0

3.0 hours

3.0 LPM1.3 hours

56mL–3.5

3.0 hours

64mL–4.0

3.0 hours

72mL–4.5

3.0 hours

80mL–5.0

2.5 hours

88mL–5.5

2.5 hours

96mL–6.0

2.5 hours

128mL–7.0

2.0 hours

160mL–8.0

1.5 hours

192mL–9.0

1.5 hours

Eclipse 5First Time Eclipse 5 Operation
Step 1: Positioning Your Eclipse 5 for Use
Place the Eclipse 5 in a well-ventilated, well lit
area. Be sure the air inlet and
Air Inlet Filter
exhaust vents are not obstructed.
Position the Eclipse 5 so that all
audible and visual indicators or
alarms can be easily seen andheard.
Be sure the air inlet filter is inplace before operating your
Eclipse 5. If the air inlet filter is
Exhaust Vent
missing, contact your home careprovider. If the air inlet filter isdirty, wash with warm soapy water, rinse with clear
water and allow it to dry before placing the air inlet
filter back in the device. If necessary, replace it witha new, clean air inlet filter.
Plug the device into an appropriate AC Power outlet, a DC Power source, or be sure there is a fully
charged Battery installed.

Step 2:
Power ON the Device andAllow it to Warm-Up

Press and hold the “ON/OFF” Button for two(2) seconds to power ON your Eclipse 5. A brief
audible and visual power-on self-test will occur.
Verify that all the power verification indicators are
illuminated and that a buzzer sounds to indicate thedevice is operating properly.
The Eclipse 5 has an oxygen concentration statusindicator (OCSI) built in to the device. The OCSI
continually monitors the oxygen output of thedevice and incorporates a green light (normal),yellow light (alert), and a buzzer. These indicators
will all illuminate during the power-on self-test at
start-up. After initially powering ON the device,please allow up to five (5) minutes for the device toreach its performance specifications. When only thegreen indicator light is illuminated, the device hasreached its performance specifications.

Step 3: Connect the Oxygen Supply Tubing or Nasal Cannula to the Oxygen Outlet
Replace the cannula and oxygen supply tubing
regularly, as recommended by your home careprovider. Consult your home care provider for cannula and supply tubing cleaning and replacement
information.
Note: Ensure the cannula is fully inserted and secure.
This ensures that the Oxygen Concentrator canproperly detect inspiration for oxygen delivery. During
inhalation, you should hear or feel oxygen flow to theprongs of the nasal cannula. The proper placement
and positioning of the prongs of the nasal cannula inyour nose is critical to the amount of oxygen delivered
to the respiratory system of the end user.

Step 4:

Select the Flow Mode

Press the Flow Mode button to select the desired
mode. Pressing this button repeatedly will toggle
you back and forth between Pulse Dose Mode andContinuous Flow Mode. The proper mode andproper setting should be prescribed for you by yourphysician.
Continuous Flow Mode Operation:
When operating in the Continuous Flow Mode, acontinuous supply of oxygen measured in liters per
minute (LPM) will flow through your tubing andnasal cannula.
Pulse Dose Mode Operation:
When operating in Pulse Mode, a bolus of oxygen,measured in milliliters (mL), is delivered upon
inspiration. DO NOT use a humidifier in the Pulse
Dose Mode.
To ensure proper saturations, secure a prescription
for pulse dose from your physician with consideration to your needs at rest, during exercise,
and when traveling at altitude. Contact your careprovider for prescription and boarding forms.
When operating in pulse dose mode, the Eclipse
5 supplies a continuous flow of oxygen if a breath
is not detected. The device will continually checkfor an inspiratory effort every 15 seconds. Oncean inspiration is detected, the device reverts tomeasured bolus delivery. To exit Pulse Dose Modeand engage Continuous Flow Mode, push the flowmode button.

PN 21182453 G | User Manual US - 17

Eclipse 5
Pulse Dose Mode and autoSAT Features
Your Eclipse 5 has a feature, called autoSAT® thatdelivers a consistent-sized bolus of oxygen up to
40 breaths per minute (see Max Breath Rate tablebelow). As your breath rate increases, ie. during
ambulation, the autoSAT feature servo-controls
the device, delivering the set bolus size. autoSAT
Technology is proprietary to CAIRE and is utilized
during Pulse Mode only.
Max Breath Rate

Bolus Volume (+/- 15%)

Pulse
Dose
Setting
1.0
2.0
3.0
4.0
5.0
6.0
7
8
9

Bolus
Size
(±
15%)
mL
16
32
48
64
80
96
128
160
192

AC Power
Supply &Power
Cartridge(Battery)Max Breath
Rate
40
40
40
40
37
31
23
18
15

DC Power
SupplyMax Breath
Rate
40
40
40
31
25
20
15
12
10

Note: Bolus volume decreases as breath rate exceeds
published range.
Note: The pulse dose setting number (1-9) is notequal to LPM.
WARNING: PULSE DOSE MODE
SETTINGS MUST BE DETERMINED FOR
EACH PATIENT INDIVIDUALLY FOR
THEIR NEEDS AT REST, DURING EXERCISE AND WHEN TRAVELING.
WARNING: THE ECLIPSE 5 MAY NOT BE ABLE TO
DETECT ALL INSPIRATORY EFFORTS IN PULSE
MODE. IF THE ECLIPSE 5 DOES NOT RESPOND TO
YOUR INSPIRATION EFFORTS, ASK YOUR HOME
CARE PROVIDER TO CHECK THE SENSITIVITYSETTING.

Note: To Equipment Provider: The following oxygen
administration accessories are recommended for use
with the Eclipse 5:• Nasal Cannula: CAIRE Part Number 5408-SEQ• Firebreak: CAIRE Part Number 20629671
A firebreak is required for use with any cannula.
• CAIRE offers a firebreak intended to be used in conjunction with the oxygen concentrator. The firebreak is
a thermal fuse to stop the flow of gas in the event thatthe downstream cannula or oxygen tubing is ignited
and burns to the firebreak. It is placed in-line with thenasal cannula or oxygen tubing between the patient
and the oxygen outlet of the Eclipse 5. For proper useof the firebreak, always refer to the manufacturer’s
instructions (included with each firebreak kit).
• Additional recommended accessories information isavailable online at www.caireinc.com.

Step 5:Adjust the Flow Setting to the
Prescribed Level
Using the Flow Setting Buttons, adjust the flow
setting to the prescribed setting supplied by yourphysician.

WARNING: IT IS VERY IMPORTANT
TO SELECT ONLY THE PRESCRIBEDLEVEL OF OXYGEN. DO NOT CHANGE
THE FLOW SELECTION UNLESS YOUHAVE BEEN DIRECTED TO DO SO BY A
LICENSED CLINICIAN.

Step 6: Begin Using Your Eclipse 5

Breathe normally through the nasal cannula.

Step 7: Universal Cart Operation

The Universal Cart is designed for use on pedestrian-type surfaces.
Align the locater holes on the bottom of the device
to the tabs on the Universal Cart.
Align the large threaded screw to the insert at
the back of the Eclipse 5. Then hand-tighten theknob on the cart until the Universal Cart is tightly
secured to the device.
Depress the push-button on the
Universal Cart handle to adjust
the height of the handle.

18 - US PN 21182453 G | User ManualEclipse 5Power Offrun for a minimum of 2 hours per month to ensure
proper function of the unit.

Press and hold the “ON/OFF”
Button for two (2) seconds topower OFF your Eclipse 5.

Note: DO NOT operate the Eclipse 5 and ambulate
while a humidifier is attached. Remove the humidifier
bottle before walking. DO NOT lay the Eclipse 5 down
while attached to a humidifier bottle.

Step 8:the Device

Store the Eclipse 5 in a cool and dry location for aperiod no longer than 1 month. The Eclipse 5 must

User Troubleshooting TableDo Not Ignore Alarms
WARNING: THE MANUFACTURER RECOMMENDS AN ALTERNATE SOURCE OF SUPPLEMENTALOXYGEN IN THE EVENT OF A POWER OUTAGE, ALARM CONDITION, OR MECHANICAL FAILURE.
CONSULT YOUR PHYSICIAN OR EQUIPMENT PROVIDER FOR THE TYPE OF RESERVE SYSTEMREQUIRED.

Symptom

Possible Cause

The yellow light is
on solid. There is no
audible alarm.

The Eclipse 5 hasdetected a problem
with the connection to
the power cartridge.

The yellow light ison. Battery symbol is
flashing on LCD.

The Eclipse 5 power
cartridge may be
warm.

The yellow light ison. Battery symbol is
flashing on LCD.

The Eclipse 5 power
cartridge voltage may
be low.

The yellow light is on.
The green light is off.

Yellow LightYour Action
1) Remove power cartridge and re-install to ensure that it issecure in the concentrator.
2) If possible, install a separate power cartridge in the Eclipse 5.
3) If this problem persists, contact healthcare provider.
1) Re-attach the Eclipse 5 to external power or install a fully-charged power cartridge in the Eclipse 5. Allow warm power
cartridge to cool outside the concentrator for 30 minutes.
2) Re-charge the power cartridge using AC power
3) Replace the power cartridge if another is available.
4) If the condition persists, contact your healthcare provider.
1) Plug Eclipse 5 into AC power supply and recharge or replace
power cartridge with fully charged power cartridge.

1) Clean and replace cabinet inlet filter
2) Ensure that the cannula is not kinked or blocked. If used witha humidifier bottle, ensure that it is filled properly and not
The Eclipse 5 may
creating a blockage.
have detected a flow
3) Ensure that the Eclipse 5 has proper ventilation. It needs to be
restriction causing low
at least 3 inches from any surface to ensure the vents are notor blocked flow.
blocked.
4) If the problem persists, switch to an alternate source of oxygenand contact healthcare provider for assistance.

PN 21182453 G | User Manual US - 19

Eclipse 5
Alarms Table
Alarm

LCD Text

Green

Yellow

Audio

Code

All OK, warm-up done

NO ALARM

ON

OFF

OFF

000

Warm-up O2 < 85%

O2 LOWON

ON

No Beeps008

No breath detected (less than 45
seconds)

NO ALARM

ON

OFF

OFF

000

No breath detected (exceeding
45 seconds)

P <--> CONON

1 alarm beep, repeat
every 20 seconds

200

One wire communication loss

ONE WIRE

ON OR
OFF

ON

1 alarm beep, repeat
every 20 seconds

100

Cannot charge battery

CHARGER

ON OR
OFF

ON

1 alarm beep, repeat
every 20 seconds

001

Ambient pressure fail

AMB PRESS

ON OR
OFF

ON

1 alarm beep, repeat
every 20 seconds004

O2 < 85% (After warm-up)

O2 LOWOFFON

1 alarm beep, repeat
every 20 seconds

008

Flow rate error > +/-10%

FLOWRATE

OFFON

1 alarm beep, repeat
every 20 seconds

020

Warm battery > 60°C

WARM BATON

ON

1 alarm beep, repeat
every 20 seconds002

Low battery

LOW BATONON

1 alarm beep, repeat
every 20 seconds

010

Loss of power********

OFFON

1 alarm beep, repeat
every 20 seconds for 200
seconds

040

Malfunction

FAIL XXOFFON

1 alarm beep, repeat
every 20 seconds for 200
seconds

080(5% remaining battery capacity)

20 - US PN 21182453 G | User ManualEclipse 5
Other Alarm Conditions

Symptom

Possible Cause

While in pulse
mode, thecompressor speeds
up and the pulse
mode LED turns on.

The Eclipse 5 isunable to detect theuser’s breathingeffort. See page
18 for pulse mode
information.

Your Action
1) Pulse mode may not work correctly if the user has any condition
that causes blocked/restricted nasal passages. (Illness, deviatedseptum, etc.)
2) Ensure cannula is no longer than 7 feet in length.
3) Ensure that there are no kinks or blockages in cannula tubing.
4) If a humidifier bottle is connected, remove the humidifier bottle andconnect cannula directly to outlet port. Pulse mode will not workwith a humidifier bottle installed.
5) If the condition persists, contact your healthcare provider.
1) Verify that the outlet is providing power. Do not connect the power
supply to a dimmer circuit or a power strip.
2) Check that cable connections on power supplies are secure to thewall/vehicle and concentrator. If using the AC power supply, checkthe cable connection on the power supply transformer as well.
3) Ensure that the power verification lights are on.

Power Cartridge
The Eclipse 5 is notreceiving power fromis draining while
plugged in to AC/DC the attached power - There will be a green light on the power supply box.
power supply
supply.
- There will be a green light on the Eclipse 5 concentrator control
panel in the shape of a power cord.
4) If power verification LEDs are not lit, remove all connections of thepower supply for 20 seconds and reconnect.
5) If the condition persists, contact your healthcare provider.
1) Connect to AC power to recharge the battery. Verify that thebattery charges for 2-5 hours and the battery icon is full and not
The power cartridge
flashing before use.
was not fully charged
2) Refer to the actions for “Power Cartridge is Not Charging” Symptom below.
The Eclipse 5 was
1) Ensure that you are using the flow rate prescribed by your physiThe Powernot operating at thecian and that pulse or continuous flow is being used as required.
Cartridge (battery)
anticipated flow rate.
charge is not lasting1) Attempt to drain the battery cartridge completely. Do this by
as long as it should.
running the Eclipse 5 on battery power until the machine shuts
off completely. Then remove the battery and allow it to cool for
The power cartridge
approximately 30 minutes. After this time, re-insert the battery in
is not preforming to
the Eclipse 5 and recharge it using AC power.
specifications.
2) If the battery still does not last as long as it should after following
step 1, contact your healthcare provider.

PN 21182453 G | User Manual US - 21

Eclipse 5
Symptom

Possible Cause

The Eclipse 5 is notreceiving external
AC or DC power tocharge the battery.

The Power Cartridge (battery) is
not charging.

Your Action
1) Verify that the AC or DC outlet is providing power. Do not connect
the power supply to an AC dimmer circuit or a power strip.
2) Check that cable connections on power supplies are secure to thewall/vehicle and concentrator. If using the AC power supply, checkthe cable connection on the power supply transformer as well.
3) Ensure that the power verification lights are on.
- There will be a green light on the power supply box.

- There will be a green light on the Eclipse 5 concentrator control
panel in the shape of a power cord.
4) If power verification LEDs are not lit, remove all connections of thepower supply for 20 seconds and reconnect.
5) If the condition persists, contact your healthcare provider.
1) Remove power cartridge and re-install to ensure that it is secure in
the concentrator.
2) Inspect the Eclipse 5 for a solid yellow light when the battery is
The Eclipse 5 is notinstalled, or a flashing battery icon when attempting to charge. If
properly commueither of these symptoms is present, continue to step 3
nicating the power3) If possible, install a separate power cartridge in the Eclipse 5.
cartridge to charge it.
If the replacement cartridge works properly, then the original
cartridge needs to be replaced.
4) If this problem persists, contact healthcare provider.

22 - US PN 21182453 G | User ManualEclipse 5
Cleaning, Care & Routine MaintenanceRoutine Maintenance

Servicing of the internal components inside thecabinet of the Eclipse 5 must be conducted by a
CAIRE trained and qualified service technician asneeded.
Preventative Maintenance (PM) to be completed
on an annual or yearly basis. Consult your provider
to arrange PM schedule. Your home care provider
or qualified technician will perform inspections of
the alarms, internal 9-volt battery, battery, internalfilters and other internal parts only as needed.

User Care and Cleaning of the Device

Cannula Replacement
Replace your supply tubing and cannula on aregular basis as recommended by your home careprovider. Your physician or home care provider
will provide you with cleaning, disinfection andreplacement information.
Note: Always follow the cannula manufacturer’s
instructions for proper use. Replace the disposable
cannula as recommended by the cannula manufacturer or your Equipment Provider. Additional supplies areavailable from your Equipment Provider.
Air Inlet Filter
Ambient air is drawn into the device through the air
inlet located on the top rear of the device. Cleaning
the air inlet filter is the most important maintenanceactivity you will perform as a user. Cleaning the
air inlet filter will keep your Eclipse 5 performing
properly. Check this air inlet filter daily and clean it
at least once a week using the following procedure:

CAUTION: Do not operate the Oxygen
Concentrator without the air intake filterin place. If a second filter is provided,
insert the “replacement” filter before you
clean the dirty filter. Clean the dirty filter
in a warm soap and water solution thendry thoroughly prior to use.
Note: The Manufacturer does not recommend thesterilization of this equipment.
Note: Do not operate the Oxygen Concentrator without the air intake filter in place. If a second filter is provided, insert the “replacement” filter before you clean
the dirty filter. Clean the dirty filter in a warm soap andwater solution then dry thoroughly prior to use.
The air inlet filter should be replaced annually. If
the Eclipse 5 is used in a dusty environment, thefilter may need to be replaced more frequently. You
should not operate the Eclipse 5 without an air inlet
filter installed for more than 5 minutes.
AC Power Supply, DC Power Supply, Power Cord andOuter Case:
Turn OFF the Eclipse 5 and disconnect from AC or
DC power before any cleaning or disinfection activity. DO NOT spray the outer case directly. Use adamp (not wet) cloth or sponge. Spray the cloth orsponge with a mild detergent solution to clean thecabinet and power supplies. To disinfect the Eclipse
5, use Lysol® Brand II disinfectant (or equivalent).
Proceed as directed by the cleaner manufacturer.

1. Remove the air inlet filter
2. Wash the filter inwarm tap waterusing a mild
soap detergent
solution
3. Rinse the filter
thoroughly withclear tap waterand squeeze out
the excess water
4. Allow the filter to
air dry
5. Reinsert the filterin the cabinet
Cleaning tip: Keep a clean second filter as a replacement for use when one filter is drying.
PN 21182453 G | User Manual US - 23

Eclipse 5
WARNING: ELECTRICAL SHOCKHAZARD. TURN OFF THE UNIT AND
DISCONNECT THE POWER CORD FROMTHE ELECTRICAL OUTLET BEFOREYOU CLEAN THE UNIT TO PREVENT
ACCIDENTAL ELECTRICAL SHOCK AND
BURN HAZARD. ONLY YOUR EQUIPMENT PROVIDER OR A QUALIFIEDSERVICE TECHNICIAN SHOULD REMOVE THE COVERS OR SERVICE THE
UNIT. DO NOT USE LIQUID DIRECTLY
ON THE UNIT. A LIST OF UNDESIRABLE
CHEMICAL AGENTS INCLUDES BUTIS NOT LIMITED TO THE FOLLOWING:
ALCOHOL AND ALCOHOL-BASEDPRODUCTS, CONCENTRATED CHLORINE-BASED PRODUCTS (ETHYLENECHLORIDE), AND OIL- BASED PRODUCTS (PINE-SOL®, LESTOIL®). THESE
ARE NOT TO BE USED TO CLEAN THE
PLASTIC HOUSING ON OXYGEN CONCENTRATOR, AS THEY CAN DAMAGE
THE UNIT’S PLASTIC.
WARNING: ELECTRICAL SHOCK HAZARD.
DISCONNECT THE POWER CORD FROM THE
ELECTRICAL OUTLET BEFORE YOU CLEAN
THE UNIT TO PREVENT ACCIDENTAL ELECTRICAL SHOCK AND BURN HAZARD. ONLY YOUREQUIPMENT PROVIDER OR A QUALIFIED SERVICETECHNICIAN SHOULD REMOVE THE COVERS ORSERVICE THE UNIT.
WARNING: CARE SHOULD BE TAKEN TO PREVENT THE OXYGEN CONCENTRATOR AND THE
BATTERY FROM GETTING WET OR ALLOWINGFLUIDS TO ENTER THE UNIT. THIS CAN CAUSE A
MALFUNCTION OR SHUT DOWN, AND CAUSE AN
INCREASED RISK FOR ELECTRICAL SHOCK ORBURNS.
WARNING: CLEAN THE CABINET, CONTROLPANEL, AND POWER CORD ONLY WITH A MILD
HOUSEHOLD CLEANER APPLIED WITH A DAMP(NOT WET) CLOTH OR SPONGE, AND THEN WIPE
ALL SURFACES DRY. DO NOT ALLOW ANY LIQUIDTO GET INSIDE THE DEVICE. PAY SPECIAL ATTENTION TO THE OXYGEN OUTLET FOR THE CANNULA CONNECTION TO MAKE SURE IT REMAINS
FREE OF DUST, WATER, AND PARTICLES.
Optional Humidifier
If used, clean your humidifier daily to reduce thepossibility of contamination. Follow the cleaning
recommendations provided by the home careprovider, or the manufacturer of the humidifier.
Consult your home care provider for suitable
humidifier options and proper placement. Replace
24 - US PN 21182453 G | User Manualthe humidifier monthly, or as recommended by the
manufacturer, or your home care provider. (Do not
overfill the humidifier.)
CAUTION: DO NOT operate the Eclipse
5 and ambulate while a humidifier is
attached. Remove the humidifier bottle
before walking. DO NOT lay the Eclipse
5 down while attached to a humidifier
bottle.
Universal Cart
Before cleaning the Universal Cart, remove it fromthe Eclipse 5. Use a damp (not wet) cloth or sponge
and mild detergent solution to clean the Universal
Cart.
Power Cartridge (Battery)
The Battery in the Eclipse 5 requires special careto assure a longer life and the highest level of performance. The CAIRE Battery is the only approved
Battery recommended for use with the Eclipse 5.
Use a damp (not wet) cloth or sponge to clean theBattery. First spray the cloth or sponge with a mild
detergent then clean the Battery case and the latch.
CAIRE Recommended Monthly Battery Service:
Run the Eclipse 5 on Battery completely down topower loss condition (yellow light and alarm). Plug
Eclipse 5 into AC power and recharge until full on
indicator. The device does not have to be running tocharge (which allows for shorter charge time).
Battery Disposal
Your Battery is rechargeable and can be recycled.
Always return it to your home care provider for
proper disposal. You can also contact your local
city or town offices for instructions on proper
disposal of the Battery.

Eclipse 5
EMC Testing
Medical equipment needs special precautions regarding EMC and needs to be installed and put into service according to the EMC information provided in this section.

Guidance and Manufacturer’s Declaration­—Electromagnetic Emissions
The Eclipse 5 is intended for use in the electromagnetic environment specified below. The customer or the user ofthe Eclipse 5 should assure that it is used in such an environment.
Emissions Test
RF emissions
CISPR 11
RF emissions
CISPR 11
Harmonic emissions
IEC 61000-3-2
Voltage fluctuations/
flicker emissions
IEC 61000-3-3ComplianceElectromagnetic Environment - GuidanceGroup 1The device uses RF energy only for its internal function. Therefore, its RF emissions are very low and are not likely to cause any
interference in nearby electronic equipment.

Class BClass A
Complies

The device is suitable for use in all establishments, including domestic establishments and those directly connected to the public
low-voltage power supply network that supplies buildings used fordomestic purposes.

WARNING: THE DEVICE SHOULD NOT BE USED ADJACENT TO OR STACKED WITH OTHER
EQUIPMENT. IF ADJACENT OR STACKED USE IS NECESSARY, THE DEVICE SHOULD BE OBSERVED TO VERIFY NORMAL OPERATION IN THE CONFIGURATION IN WHICH IT WILL BE USED.
WARNING: THE USE OF ACCESSORIES OTHER THAN THOSE SPECIFIED FOR THE DEVICE IS NOT RECOMMENDED. THEY MAY RESULT IN INCREASED EMISSIONS OR DECREASED IMMUNITY OF THE DEVICE.

PN 21182453 G | User Manual US - 25

Eclipse 5
Guidance and Manufacturer’s Declaration—Electromagnetic Immunity

The Eclipse 5 is intended for use in the electromagnetic environment specified below. The customer or the user ofthe Eclipse 5 should assure that it is used in such an environment.
Immunity test
Compliance level
Electromagnetic environment – guidanceElectrostatic discharge±8 kV contact
Floors should be wood, concrete or ceramic tile. If(ESD) IED 61000-4-2
floors are covered with synthetic material, the relative±15 kV air
humidity should be at least 30 %.
Electrical fast transient/
burst

±2 kV

IEC 61000-4-4
Surge

Mains power quality should be that of a typical commercial or hospitalenvironment.

±1 kV

IEC 61000-4-5±2 kV

Mains power quality should be that of a typical commercial or hospitalenvironment.
Mains power quality should be that of a typical commercial or hospital environment. If the user of the
Eclipse 5 requires continued operation during power
mains interruptions, it is recommended that the Eclipse
5 is powered from an uninterruptible power supply(UPS).
Power frequency magnetic fields should be at levels
characteristic of a typical location in a typical commercial or hospital environment.

Voltage dips, short in100V-240V
terruptions and voltage
variations on power supply
input lines IEC 61000-4-11
Power frequency magnetic 30A/m
field IEC 61000-4-8
Conducted RF

10 Vrms

IEC 61000-4-8

150 kHz to 80 MHz

Radiated RF

9 V/m to 85 V/m

IEC 61000-4-6

At frequencies, up to
5.785GHzThe Eclipse 5 complies with all applicable electromagnetic compatibility requirements (EMC) according toIEC 60601-1-2:2014, for residential, commercial andlight industry environments. Portable and mobile RFcommunications equipment should be used no closer
to any part of the machine, including cables, than therecommended 10cm separation distance. The Eclipse
5 has been designed to meet EMC standards. However, should you suspect that the machine performance(e.g. pressure or flow) is affected by other equipment,
move the machine away from the possible cause of
the interference.
The Eclipse 5 complies with Part 15 of the FCC rules
and industry Canada license-exempt RSS standards.
Operation is subject to the following two conditions:this machine may not cause harmful interference, andthis machine must accept any interference revived,
including interference that may cause undesired
operation.
FCC ID: WAP2001

26 - US PN 21182453 G | User ManualEclipse 5

During taxi, take-off, and landing, the user must
be in a seat location that does not restrict anyother passenger’s access to, or use of, any required
emergency or regular exit, or the aisle(s) in thepassenger compartment of the aircraft if the Eclipse
5 is used.
Users are not permitted to be seated in an exit row
if using the Eclipse 5.
If the Eclipse 5 is used when decompression of thecabin occurs and the oxygen cabin oxygen system
deploys, then the user is to discontinue use of the
Eclipse 5 and use the aircraft supplemental oxygen.
The Eclipse 5 unit is to be turned off after securing
the aircraft supplemental oxygen.
Prior to travel, the user needs to inspect the Eclipse
5 to ensure it is in good operational condition.
CAUTION: The US Department ofTransportation (DOT) and United Nations(UN) Regulations require the removal of
the battery from the device for all international airline travel when the oxygen
concentrator is checked as luggage.
When shipping the oxygen concentrator,
the battery must also be removed fromthe device and packaged properly.

These tables calculate flight times only.
Please allow for ground time (preflight check in,security check in, etc.) and layovers.
Most airlines require that you havean adequate number of fully charged
batteries to power the device for atleast 150% of the expected maximum trip duration(flight times, all ground time, before and after
flight, and during connections and for unexpected
delays).
It is suggested to always carry an
extra battery with you on trips.
Each airline has their own requirements
for traveling with oxygen.
Contact your airline at least 48 hours prior to
departure to inform them you will be traveling with
the Eclipse 5.
Continuous Flow

FLIGHT LENGTH (in Hrs)*

SETTING

During taxi, take-off, and landing, the Eclipse 5
must be turned off and stowed under the seat orin another approved stowage location so as to notblock the aisle way or the entry way into the row if
the Eclipse 5 will not be used.

Power Cartridge &Airline Travel Table

1.0 2.0 3.0 4.0 5.0 6.0 7.0 8.0 9.0 10.0
0.5 1 1 1 2 2 2 3 3 4 4
1.0 1 1 2 2 2 3 3 4 4 5
1.5 1 1 2 2 3 3 4 4 5 5
2.0 1 2 2 3 4 4 5 5 6 7
2.5 1 2 3 4 5 5 6 7 8 9
3.0 2 3 4 5 6 7 9 10 11 12
ESTIMATED # OF POWER CARTRIDGES NEEDED*Table accounts for the typical 150% duration requirement.

Pulse Dose Mode

FLIGHT LENGTH (in Hrs)*

PULSE SETTING

General Airline Travel
Information

1.0 2.0 3.0 4.0 5.0 6.0 7.0 8.0 9.0 10.0
1 1 1 1 2 2 2 3 3 4 4
1.5 1 1 2 2 2 3 3 3 4 4
2 1 1 2 2 3 3 3 4 4 5
2.5 1 1 2 2 3 3 4 4 5 5
3 1 1 2 2 3 3 4 4 5 5
3.5 1 2 2 3 3 4 4 5 5 6
4 1 2 2 3 3 4 4 5 6 6
4.5 1 2 2 3 3 4 5 5 6 6
5 1 2 2 3 4 4 5 6 6 7
5.5 1 2 2 3 4 5 5 6 7 7
6 1 2 3 3 4 5 5 6 7 8
7
8
9

1
1
1

Table is based on 20 Breaths Per Minute,for Pulse Dose Mode only.

2
2
2

2
3
3

3
3
4

3
4
5

4
5
6

5
6
7

5
6
8

6
7
8

Table is based on 15 Breaths Per Minute,for Pulse Dose Mode only.

6
8
9

PN 21182453 G | User Manual US - 27

www.cairein c .c o m

CAIRE Inc.
2200 Airport Industrial Dr., Ste. 500
Ball Ground, GA 30107 U.S.A.
Medical Product Service GmbH
Borngasse 20
35619 Braunfels, Germany
Tel: +49 (0) 6442-962073
E-mail: info@mps-gmbh.eu
Accumed Sagl Viale
Serfontana 106834 Morbio Inferiore
Switzerland
CAIRE and CAIRE Inc. are registered trademarks of CAIRE Inc.
Please visit our website below for a full listing of trademarks. Trademarks: www.caireinc.com/corporate/trademarks/.
Salter Labs® is a registered mark of Salter Labs, Arvin, CA 92303
Lysol® is a registered trademark of Reckitt Benckiser, UK.
Pine-Sol® and Lestoil® are registered trademarks of The Clorox
Company.
Copyright © 2023 CAIRE Inc.. CAIRE Inc. reserves the right todiscontinue its products, or change the prices, materials, equipment,
quality, descriptions, specifications and/or, processes to its products
at any time without prior notice and with no further obligation orconsequence. All rights not expressly stated herein are reserved byus, as applicable.

|21182453-Gs~

6 Feb 2023 PN 21182453 Rev G