取扱説明書

KEW2060BT

KEW2062BT

クランプパワーメータ

KEW 2060BT/2062BT
共立電気計器株式会社

目次

KEW 2060BT/2062BT

梱包内容の確認.................................................................................................................................................... 3
安全に関するご使用上の注意........................................................................................................................ 3
1章
製品の概略 ...................................................................................................................................... 72章
各部の名称、説明......................................................................................................................... 83章
基本操作........................................................................................................................................... 9
3.1 ファンクションスイッチ .................................................................................................................. 9
3.2 操作スイッチ .................................................................................................................................... 9
3.3 LCD に表示する記号.................................................................................................................11
3.4 測定値の単位 ...............................................................................................................................11
4章
測定前の準備 ...............................................................................................................................124.1 電源の投入 ....................................................................................................................................124.2 電池の確認 ....................................................................................................................................12
画面の表示／電池の残量.......................................................................................................13
乾電池のセット方法.................................................................................................................... 13
4.3 測定コードの接続（本体） .........................................................................................................14
4.4 測定対象への接続 .....................................................................................................................14
5章
設定...................................................................................................................................................16
項目の選択（移動） .....................................................................................................................16
結線方式.........................................................................................................................................17
ＶＴ比／ＣＴ比................................................................................................................................. 17
ＶＴ／ＣＴを使用した測定 ..........................................................................................................18
ブザー ON／OFF........................................................................................................................19
バックライト ON／OFF..............................................................................................................19公称電圧の周波数 .....................................................................................................................20
システムリセット............................................................................................................................20
6章
測定ファンクションごとの表示項目 ......................................................................................21
6.1 実効値・周波数測定...................................................................................................................21
電流実効値、周波数 ..................................................................................................................21
電圧実効値、周波数 ..................................................................................................................22
6.2 単相・三相（平衡）電力測定 ....................................................................................................23
単相２線式（１Ｐ２Ｗ）結線図...................................................................................................23
単相３線式（１Ｐ３Ｗ）結線図...................................................................................................23
三相３線式（３Ｐ３Ｗ）平衡 結線図.......................................................................................24
三相４線式（３Ｐ４Ｗ）平衡 結線図.......................................................................................24
表示切り換え................................................................................................................................. 25
6.3 三相（不平衡）電力測定............................................................................................................26
三相３線式（３Ｐ３Ｗ）不平衡...................................................................................................26
三相４線式（３Ｐ４Ｗ）不平衡...................................................................................................29-1-

KEW 2060BT/2062BT

KEW 2060BT/2062BT

目次

KEW 2060BT/2062BT

6.4

6.5
7章

8章
8.1
9章
9.1
9.2
9.3

高調波測定 ....................................................................................................................................32
電流高調波歪み率、含有率、実効値 .................................................................................32
電圧高調波歪み率、含有率、実効値 .................................................................................33
高調波歪み率 THD-R/THD-F ..............................................................................................35
検相...................................................................................................................................................36
その他の機能................................................................................................................................37「データホールド機能」...............................................................................................................37「自動バックライトオフ」..............................................................................................................37「オートパワーオフ」 ....................................................................................................................37「電流オートレンジ」.....................................................................................................................37
Bluetooth を使用する.................................................................................................................38
専用アプリ「KEW Power＊(アスタリスク)」の機能 ..........................................................39
仕様...................................................................................................................................................40
安全要求仕様 ...............................................................................................................................40
一般仕様 .........................................................................................................................................40
測定仕様 .........................................................................................................................................41交流電流ファンクション....................................................................................................41交流電圧ファンクション.....................................................................................................42

電力ファンクション.............................................................................................................43
高調波ファンクション......................................................................................................46
検相ファンクション..............................................................................................................48
アフターサービス.................................................................................................................................................49

KEW 2060BT/2062BT

KEW 2060BT/2062BT-2-

梱包内容の確認

KEW 2060BT/2062BT

梱包内容の確認
このたびは弊社クランプパワーメータ KEW 2060BT または KEW 2062BT をご購入していただき
ありがとうございます。
まずお手元に届きました本製品の梱包内容を確認してください。

梱包内容




1

本体

2

測定コード

3
4
5

電池
取扱説明書
ソフトケースKEW 2060BT またはKEW 2062BT：1 台
MODEL7290：1 セット
赤、黒、黄、各 1 本（ワニグチクリップ付き）
単３形アルカリ乾電池（LR6）：2 個：1 冊
MODEL9198：1 個

製品の間違い、品不足、破損、印刷不良等がございましたらお買上店（販売店）までご連絡をください。
本取扱説明書には保証書が付いています。大切に保管してください。

安全に関するご使用上の注意
本製品は IEC 61010：電子測定装置に関する安全規格に準拠して、設計・製造の上、検査合格した最良の状態で出荷しています。
この取扱説明書には、使用する方の危険をさけるための事項および本製品を損傷させずに長期間良好な状態で使用していただくための事柄が書かれていますので、お使いになる前に必ずこの取扱説明書をお読みください。

警告
取扱説明書について
 本製品を使用する前に必ずこの取扱説明書をよく読んでご理解ください。
 この取扱説明書は手近な所に保管し、必要なときにいつでも取り出せるようにしてください。
 製品本来の使用方法および取扱説明書で指定した使用方法を守ってください。
 取扱説明書の安全に関する指示に対しては、指示内容を理解の上、必ず守ってください。
以上の指示を必ず厳守してください。指示に従わないと、怪我や事故の恐れがあります。危険および警告、注意に反した使用により生じた事故や損傷については、弊社として責任と保証を負いかねます。
本製品に表示のマークは、安全に使用するため取扱説明書を読む必要性を表しています。尚、このマークには次の 3 種類がありますので、それぞれの内容に注意してお読みください。

危険： この表示を無視して誤った取扱いをすると、人が死亡または重傷を負う危険性が高い内容を示しています。

警告： この表示を無視して誤った取扱いをすると、人が死亡または重傷を負う可能性が想定される内容を表示しています。

注意： この表示を無視して誤った取扱いをすると、人が傷害を負う可能性が想定される内容および物的損害の発生が想定される内容を示しています。

-3-

̶3̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

安全に関するご使用上の注意

KEW 2060BT/2062BT
本製品に使用している安全記号

取扱説明書を参照する必要があることを示します。
二重絶縁または強化絶縁で保護されている機器を示します。
隣接表示の測定カテゴリに対する回路－大地間電圧以下であれば活線状態の裸導線をクランプできる設計であることを示しています。

～

交流(AC)を示します。
（機能）接地端子を示します。
本製品は、WEEE 指令（2002／96／EC）マーキング要求に準拠します。
この電気電子製品を一般家庭廃棄物として廃棄してはならないことを示します。

○測定カテゴリについて
安全規格 IEC61010 では測定器の使用場所についての安全レベルを測定カテゴリという言葉で規定し、以下のように O～CAT Ⅳの分類をしています。この数値が大きいほど過渡的なインパルスが大きい電気環境であることを意味します。CAT Ⅲで設計された測定器は CAT Ⅱで設計されたものより高いインパルスに耐えることができます。
O(None, Other) ： 主電源に直接接続しないその他の回路です
CAT II ： コンセントに接続する電源コード付機器の一次側の電気回路
CAT III ： 直接配電盤から電気を取り込む機器の一次側および分岐部からのコンセントまでの電路
CAT IV ： 引込み線から電力量計および一次過電流保護装置（配電盤）までの電路

KEW 2060BT/2062BT

KEW 2060BT/2062BT

-4-

̶4̶

安全に関するご使用上の注意

KEW 2060BT/2062BT
危険● 指定した操作方法および条件以外で使用した場合、本体の保護機能が正常に動作せず本器を破損したり感電等の重大な事故を引き起こす可能性があります。本器の使用前あるいは指示結果に対する対策を取る前に、既知の電源で正常な動作を確認してください。
● 被測定物やその周辺をさわると感電が想定される場所での測定には、絶縁保護具を着用してください。
● KEW2060BT の対地間最大定格電圧は、測定カテゴリに準じて CAT Ⅳで AC600V、CAT Ⅲで
AC1000V、KEW2062BT の対地間最大定格電圧は、測定カテゴリに準じて CAT Ⅳで AC300V、
CAT Ⅲで AC600V、CAT Ⅱで AC1000V です。これより高い対地間電圧のある回路では絶対に使用しないでください。
● 引火性ガスや爆発性のガスおよび、蒸気のある場所で使用すると大変危険ですので、使用しないでください。
● 本製品や手が濡れている状態や、湿気などの水滴が付着した状態では、絶対に使用しないでください。
測定について● 測定の際には、測定範囲を超える入力を加えないでください。
● 測定中は絶対に電池蓋を開けないでください。
電流センサの接続について● 測定電流と定格が合っているか必ず確認し、対地間最大定格電圧以下の電路で使用してください。
● 測定の際は指先等が、バリアを超えることのないよう充分注意してください。
バリア：操作中の感電事故を防ぐため、最低限必要な沿面および空間距離を確保するための目印です。
● 必ずブレーカーの二次側に接続してください。一次側は電流容量が大きく危険です。
● 電流センサを開いたとき、金属部で測定ラインの２線間を接触させないでください。
測定コードの接続について● 付属のものをご使用ください。
● 測定コードと本体の測定カテゴリが違っている場合は低い方の測定カテゴリを優先します。測定電圧と定格が合っているか必ず確認してください。
● 測定に必要のないものは絶対に接続しないでください。
● 本体に接続していない状態で測定ラインに接続しないでください。
● 測定の際は指先等が、バリアを超えることのないよう充分注意してください。
バリア：操作中の感電事故を防ぐため、最低限必要な沿面および空間距離を確保するための目印です。
● 測定中（測定ラインからの通電中）は絶対に本体の電圧入力端子から取りはずさないでください。
● 先端の金属部で測定ラインの２線間を接触させないでください。
● 先端の金属部には絶対に触れないでください。
電池について● 測定中は絶対に電池交換を行わないでください。

-5-

̶5̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

安全に関するご使用上の注意

KEW 2060BT/2062BT

警告● 本製品を使用しているうちに、本体や測定コードに亀裂が生じたり金属部分が露出したときには使用を中止してください。
● 本製品を使用する前に必ず既知の電源で正常に測定できることを確認してください。
● 本製品の分解、改造、代用部品の取り付けは行わないでください。修理・調整が必要な場合は、弊社または取扱店宛にお送りください。
注意● 本製品の使用は住宅・商業用および軽工業の環境に制限されます。付近に強い電磁干渉装置や大電流による大きな磁界がある場合は、正確に測定できない場合があります。
● 被測定導線が高温の場合がありますので注意してください。
● 各ファンクションの測定範囲を超える電流や電圧を入力しないでください。
● 電源が OFF の状態で、測定コードや電流センサに電圧や電流を入力しないでください。
● 埃の多い場所や、水のかかる環境で使用しないでください。
● 強力な電磁波を発生したり、帯電したりしているものの近くでは使用しないでください。
● 振動や衝撃を与えたり、落下させたりしないでください。
測定コードについて● 端子にプラグを根元まで確実に差し込んでください。
● 強く引っ張ったり捻ったりしないでください。亀裂または断線する恐れがあります。
電池について● 銘柄や種類の違う電池を混ぜて使用しないでください。
使用後について● 使用後は必ずファンクションスイッチを、“OFF”にし、測定コードを外してください。
● 長期間ご使用にならない場合は、電池を取り外した状態で保管してください。
● 運搬の際には振動や衝撃を与えたり、落下させたりしないでください。
● 高温多湿、結露するような場所および直射日光の当たる場所に放置しないでください。
● クリーニングには研磨剤や溶剤を使用しないで、中性洗剤か水に浸した布を使ってください。
● 濡れているときは、乾燥後保管してください。
また、各章の危険、

KEW 2060BT/2062BT

KEW 2060BT/2062BT

警告、注意、注記の内容も必ず守ってください。

-6-

̶6̶

1章 製品の概略

KEW 2060BT/2062BT

1章 製品の概略
本製品は多彩な結線方式に対応したクランプパワーメータです。電圧・電流実効値、電力測定に加え、電源品質確認用の高調波解析機能と、検相機能とを備えています。また、Bluetooth 通信で測定デー
タや波形データを、市販のタブレット端末、スマートフォンからリアルタイムに確認したり、記録したりで
きます。
安全設計
KEW2060BT は安全規格 IEC 61010-1 CAT Ⅳ 600V／CAT Ⅲ 1000V、
KEW2062BT は安全規格 IEC 61010-1 CAT Ⅳ 300V／CAT Ⅲ 600V／CAT Ⅱ1000Vに準拠した安全設計です。
結線方式
単相２線式（単相３線式）、三相３線式（２電力計法）、三相４線式の各種測定ラインに対応しています。
大口径電流センサ
KEW2060BT は被測定導体径φ７５mm までの配線または、幅８０mm までのブスバーを安全にクランプできます。
KEW2062BT は被測定導体径φ５５mm までの配線を安全にクランプできます。
測定および演算真の電圧実効値、電流実効値と各周波数、有効／無効／皮相電力、力率、電圧電流位相差を測定
および演算します。
高調波測定
１次～３０次の各電圧／電流高調波の実効値および、含有率、総合高調波歪み率(THD-R／-F)を表示します。
検相
動力電源の回転方向および、欠相を判定します。
アプリケーション
Bluetooth 経由で測定データまたは波形データをタブレットやスマートフォンへ送信します。端末でロギングしたファイルは、アプリケーション（KEW Power＊(アスタリスク)）で簡単に参照することが可能です。

-7-

̶7̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

2章 各部の名称、説明

KEW 2060BT/2062BT

2章 各部の名称、説明①電流センサ※イラストは 2060BT です。

②トリガー（電流センサ開閉用レバー）
⑫
③

①

③バリア
操作中の感電事故を防ぐため、最低限必要な沿面および空間距離を確保するための目印です。
測定の際は、指先等がバリアを超えないように充分注意してください。
④ファンクションスイッチ
測定ファンクションの切り換えスイッチです。電源
スイッチを兼ねており、“ＯＦＦ”の位置で電源が切れます。

⑤⑤データホールドスイッチ
表示部の測定値を固定するためのスイッチです。
ホールド中は表示部にを表示します。

②④⑨⑥モードスイッチ※1､2
主に、表示する測定値を、MAX：最大値／MIN：最小値／AVG：平均値／|PEAK|：波高値（絶対値）の順で、ローテーションしながら切り換えます。

⑦ ⑥ ⑧

◀

⑦バックライトスイッチ⑪[ ：共通]

長押しでバックライトを点灯、または消灯します。

◀▶

⑩⑦⑧表示項目切換スイッチ[]※2
短押しで主に、表示項目をローテーションしながら切り替えます。

MAX：最大値／MIN：最小値／AVG：平均値／|PEAK|：波高値（絶対値）を表示している状態では、電流に関係するファンクションのレンジは固定です。瞬時値の表示に戻すとオートレンジに換わります。

※1※2

スイッチ⑥～⑧の操作内容は、⑦のバックライト操作を除いて、ファンクションごとに異なります。
詳しくは、9 頁「3.2 操作スイッチ」および各ファンクション項目にある説明を参照してください。

⑨表示部
バックライト付きの LCD（電界効果型液晶）です。
⑩交流電圧入力端子
測定コード（MODEL7290）の⑪プラグを差し込みます。測定する結線方式に合わせて接続してください。
⑪プラグ⑫ワニグチクリップ
KEW 2060BT/2062BT

KEW 2060BT/2062BT

-8-

̶8̶

3章 基本操作

KEW 2060BT/2062BT

3章 基本操作
3.1

ファンクションスイッチ
ファンクション

SETUP

W
～V
～A
3.2

概略

設定

結線方式、VT 比、CT 比、ブザーの ON/OFF、バックライトの ON/OFF、
公称周波数 50Hz/60Hz の設定を行います。システムリセットを行うと、すべての設定を出荷状態に戻します。

検相

動力電源の回転方向および、欠相を判定して表示します。

高調波

電圧・電流の 1 次基本波から 30 次高調波までの実効値、含有率、歪み率[THD-R/THD-F]を表示します。

電力

有効・無効・皮相電力、力率、電圧電流位相差、電圧・電流実効値を表
示します。

交流電圧

交流電圧の実効値、波高値と、周波数とを表示します。

交流電流

交流電流の実効値、波高値と、周波数とを表示します。

操作スイッチファンクション

－

スイッチ

操作内容
データホールドスイッチを押すと LCD 左上にを表示して、押した時点の表示を保持します。この状態では、入力が変化していても測定値を更新しません。再度データホールドスイッチを押すか、ファンクションを切り換えるとを消して、表示の更新を再開します。

データホールド
スイッチ
バックライトスイッチ[◀ ]長押し：バックライトを点灯、または消灯します。

スイッチ [◀▶ ]設定項目の変更および、設定値を変更します。

表示項目切換SETUP

モードスイッチ
表示項目切換
スイッチ
[◀▶ ]
[▶ ]
高調波
モードスイッチ

設定項目の選択および、設定した値を決定します。
短押し： ↔ THD-F ↔ THD-R ↔ 1 次基本波～30 次高調波 ↔
の表示をローテーションします。
長押し： 電圧・電流実効値の表示を、交互に切り換えます。
短押し：下記のように表示をローテーションします。
↔ 瞬時値 ↔ MAX（最大値） ↔ MIN（最小値）↔ AVG（平均値） ↔
長押し：
MAX、MIN、AVG の測定を一旦リセットして測定を再開します。

-9-

̶9̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

3.2 操作スイッチ

KEW 2060BT/2062BT
ファンクション

スイッチ

操作内容

表示項目切換
電力
1P2W
1P3W

スイッチ [◀▶ ]短押し：下記のように表示をローテーションします。
↔ 瞬時値 ↔ MAX（最大値） ↔ MIN（最小値） ↔ AVG（平均値） ↔
長押し：
MAX、MIN、AVG の測定を一旦リセットして測定を再開します。

モードスイッチ

表示項目切換
電力
3P3W
3P4W
平衡

スイッチ [◀▶ ]表示項目切換
[▶ ]
[◀▶ ]

電力
3P3W
不平衡
モードスイッチ

表示項目切換スイッチ [▶ ]

電力
3P4W
不平衡

モードスイッチ～V～A

モードスイッチ

短押し：下記のように表示をローテーションします。
↔ 有効電力・力率 ↔ 有効・皮相電力 ↔
有効・無効電力↔電流・電圧実効値 ↔
短押し：下記のように表示をローテーションします。
↔ 瞬時値 ↔ MAX（最大値） ↔ MIN（最小値） ↔ AVG（平均値） ↔
長押し：
MAX、MIN、AVG の測定を一旦リセットして測定を再開します。

モードスイッチ

スイッチ

短押し：下記のように表示をローテーションします。
↔ 有効電力・力率 ↔ 有効電力・電圧電流位相差 ↔
有効・皮相電力 ↔ 有効・無効電力 ↔ 電流・電圧実効値 ↔

測定中の短押し：測定対象を R（Ｌ１）相から T（Ｌ３）相へ移行します。
結果表示中の短押し：下記のように表示をローテーションします。
↔ 三相有効電力 ↔ R（Ｌ１）相有効電力 ↔ T（Ｌ２）相有効電力 ↔
測定中の短押し：有効電力と電圧・電流実効値の表示を、交互に切り換えます。
結果表示中の長押し：表示をリセットして再測定を開始します。
測定中の短押し：測定対象を R（Ｌ１）相→Ｓ（Ｌ２）相→T（Ｌ３）相へと移行します。
結果表示中の短押し：下記のように表示をローテーションします。
↔ 有効電力・力率 ↔ 有効・皮相電力 ↔ 有効・無効電力 ↔
測定中の短押し：有効電力と電圧・電流実効値の表示を、交互に切り換えます。
結果表示中の長押し：表示をリセットして再測定を開始します。
短押し：下記のように表示をローテーションします。
↔ 瞬時値 ↔ MAX（最大値） ↔ MIN（最小値）↔ AVG（平均値） ↔ |PEAK|（波高値※） ↔
長押し：
MAX、MIN、AVG、|PEAK|の測定を一旦リセットして測定を再開します。
※

KEW 2060BT/2062BT

KEW 2060BT/2062BT|ＰＥＡＫ|は瞬間の波高値を絶対値で表示します。

- 10 -

̶ 10 ̶

3.3 LCDに表示するマーク

3.3

KEW 2060BT/2062BT

LCD に表示する記号
表示時の状態マーク

電池の残量です。4 種類に変化します。
Bluetooth が使用できます。
画面の表示更新をホールドしています。
測定方法の設定は不平衡です。平衡の時は表示しません。
結線の設定です。単相では表示しません。
総合の電力です。「P1」または「P2」のみの表示は、表示に準じた単相の電力です。
ブザーを OFF しています。
総合高調波歪み率の種類です。
高調波の次数です。基本波 1 次（h - 1）から高調波 30 次（h - 30）までを表示します。
VT 比に 1/1 外を設定しています。
CT 比に 1/1 外を設定しています。
表示している測定値の種類上に、しるしをつけます。
高調波測定時に設定された公称周波数を示します。50Hz を設定していれば50Hz と表示されます。
各測定値の極性です。マイナスまたは符号なし(プラス)を表示します。
極性符号が示す状態は、「9.3 測定仕様」を参照してください。

3.4

測定値の単位
単位

V

電圧実効値

A

電流実効値

Hz

周波数

ｋW

有効電力

ｋVar

無効電力

kVA

皮相電力

PF

力率

deg

電圧電流位相差％

高調波含有率- 11 -

̶ 11 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

4章 測定前の準備

KEW 2060BT/2062BT

4章 測定前の準備
4.1

電源の投入
注記

●

ファンクションスイッチが測定機能の位置にある状態で電源が切れている時は、オートパワーオフの機能で、自動的に電源が切れた状態です。ファンクションスイッチを「OFF」の位置にして、再度測定ファンクションの位置に移動すると、本体の電源が入ります。
この状態でも電源が入らない場合には、電池が完全に消耗していると思われます。新しい電池に交換してから、電源を投入してください。
ファンクションスイッチを「OFF」以外の位置にセットすると、本体の電源をオンして、LCD のすべての表示項目を、約 1秒間点灯します。表示に欠けなどが、ないことを確認してく
ださい。

4.2

電池の確認
危険

●

測定中は絶対に電池交換を行わないでください。

●

電池の交換のため電池蓋を開けるときには、必ず測定コードをはずし、ファンクションスイッチをＯＦＦにしてください。
本製品が濡れている状態では、電池交換を行わないでください。
電池の電圧警告が点滅している状態では正確な測定ができません。ただちに使用を中止して電池を交換してください。また、電池が完全になくなっている場合は、LCD 表示が消えも表示しませんので注意してください。

警告

●
●

KEW 2060BT/2062BT

KEW 2060BT/2062BT- 12 -

̶ 12 ̶

画面の表示／電池の残量

KEW 2060BT/2062BT

注意
●
●
●

銘柄や種類の違う電池を混ぜて使用しないでください。
電池は古いものと混ぜて使用しないでください。
電池の極性をまちがえないよう、ケース内の表示に合わせて入れてください。

画面の表示／電池の残量表示

内容
電池が最適の状態です。

電池の残量

電池の残量に応じてアイコンが変化します。
電池残量が少なくなっています。早めの電池交換をお勧めします。

点滅

電池が消耗して正常に測定できない状態です。ただちに使用を中止して電池を交換してください。
この状態でも測定は継続しますが、Bluetooth の通信は停止します。

乾電池のセット方法
以下の手順で乾電池をセットします。

ネジ
電池蓋

単３形アルカリ乾電池：ＬＲ６１

測定コードを本体からはずし、ファンクションスイッチを OFF にします。

２

本体裏側の下部にある電池蓋のネジ 1 個を緩めて電池蓋をはずします。

３

古い電池をすべて取りはずします。

４

正しい極性で新しい単３形アルカリ乾電池：ＬＲ６を２本セットします。

５

電池蓋を取り付けて、ネジ１個を締めます。
- 13 -

̶ 13 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

4.3 測定コードの接続（本体）KEW 2060BT/2062BT

4.3

測定コードの接続（本体）
必ず確認してください。

危険
●
●
●
●

測定コードは付属の専用コードを使用してください。
測定に測定コードが必要ない場合には接続しないでください。
プラグ側を本体に接続していない状態で測定ラインに接続しないでください。
測定中（測定ラインからの通電中）は絶対に本体の電圧入力端子から取りはずさないでください。

警告● 使用しているうちに亀裂が生じたり、金属部分が露出したときは、ただちに使用を中止してください。

注意●●

本体の電源が OFF になっていることを確認してから接続してください。
接続は必ず先に本体側から行い、根元まで確実に差し込んでください。
以下の手順で測定コードを接続します。

１

本体に電源が入っていないことを確認します。

２

本体の交流電圧入力端子へ測定コードを接続
します。

※
※

接続する測定コードの数は、結線方式ごとに異なり
ます。

4.4

測定対象への接続
必ず確認してください

危険
●

●
●
●
●
●

KEW2060BT は測定カテゴリに準じて CAT Ⅳでは AC600V、CAT Ⅲでは AC1000V より高い電圧のある回路では絶対に使用しないでください。
KEW2062BT は測定カテゴリに準じて CAT Ⅳでは AC300V、CAT Ⅲでは AC600V、CAT Ⅱでは
AC1000V より高い電圧のある回路では絶対に使用しないでください。
測定コードは本製品専用のものをご使用ください。
測定コードは必ず測定物や電源よりも先に本体に接続してください。
測定コードと本体の測定カテゴリが違っている場合は低い方の測定カテゴリが優先されます。測定
電圧と定格が合っているか必ず確認してください。
測定に必要のない測定コードは絶対に接続しないでください。
電流センサは必ずブレーカーの二次側に接続してください。一次側は電流容量が大きく危険です。

KEW 2060BT/2062BT

KEW 2060BT/2062BT- 14 -

̶ 14 ̶

4.4 測定対象への接続

KEW 2060BT/2062BT

危険
●
●
●

結線時に測定コードの先端の金属部で電源ラインを短絡しないように注意してください。また、先端の金属部には絶対に触れないでください。
電流センサのコア先端部は被測定物を短絡しないような構造になっていますが、絶縁されていない
導線を測定する場合コアで被測定物を短絡しないように注意してください。
測定の際は指先等が、バリアを超えることのないよう充分注意してください。
バリア： 操作中の感電事故を防ぐため、最低限必要な沿面および空間距離を確保するための目印です。
正確に測定するために● 電流センサの中心に測定物があるときに精度を保証します。
● 電流センサの先端部で配線を挟まないように注意してください。
● 測定ラインと本製品の結線方式の設定は正しく行ってください。
● 電流センサは下記のように矢印を負荷側に向けてクランプしてください。
※
逆にクランプすると有効電力（Ｐ）の極性が逆転します。
● ２本の配線をクランプしないでください。

矢印- 15 -

̶ 15 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

5章 設定KEW 2060BT/2062BT

5章 設定
測定を始める前にあらかじめ、結線方式や被測定電圧の周波数、必要に応じて VT 比、CT 比を設定する必要があります。
ファンクションスイッチをに合わせて設定してください。

注記● 設定を“決定”する前に「ファンクションスイッチ」を切り換えると、設定中の値を反映しません。ご注意く
ださい。

項目の選択（移動）
本体正面下側の左右にある「表示項目切換スイッチ [◀▶ ]」で設
定項目を移動し、中央の「モードスイッチ」で選択します。
設定項目ごとの変更は「表示項目切換スイッチ [◀▶ ]」で値を変えてください。変更が完了した後に再度「モードスイッチ」を押すことで、設定値を決定して移動画面に戻ります。
表示切換スイッチ [◀▶ ] ： 設定項目の移動「モードスイッチ」 ： 選択
出荷時は下記の設定です。 システムリセットを行うと出荷時の設定に初期化します。
結線

VT 比

CT 比

[◀▶ ]

[◀▶ ]

システムリセット

ブザー

[◀▶ ]

[◀▶ ]

[◀▶ ]

[◀▶ ]

[◀▶ ]
公称電圧の周波数
KEW 2060BT/2062BT

KEW 2060BT/2062BT

バックライト- 16 -

̶ 16 ̶

結線方式

KEW 2060BT/2062BT

結線方式「結線」を表示し、中央の「モードスイッチ」を押すと、結線設定の変更が可能になります。測定対
象の結線に合わせて、５種類の結線方式の中から１つを選択してください。
※

単相３線式（１Ｐ３Ｗ）測定時には、単相２線式（１Ｐ２Ｗ）を選択し、各相（Ｌ１／Ｌ２）の電力を別々に測定してください。１Ｐ３Ｗの総合電力は表示できません。

表示切換スイッチ [◀▶ ] ： 結線の変更
単相２線式

平衡 三相３線式

不平衡 三相３線式
[◀▶ ]

[◀▶ ]

[◀▶ ]

[◀▶ ]

不平衡 三相４線式

平衡 三相４線式[◀▶ ]設定する結線を表示して中央の「モードスイッチ」を押すと、決定して移動画面に戻ります。

ＶＴ比／ＣＴ比注意● ＶＴ、ＣＴを使用した表示範囲は、電圧・電流実効値 0.000～9999、電力 0.000k～9999k の範囲です。表
示範囲に合わせて適切なＶＴ比、ＣＴ比を設定してください。極端に大きな値、または小さい値を設定す
ると、表示がゼロまたは OL に飽和して変わらなくなります。
● ＶＴ比、ＣＴ比の設定に関わらず入力できるのは、交流電圧入力端子に 1100V、電流センサに 1100A までです。ＶＴ、ＣＴの出力がこれを超えていると表示は OL になります。
外部にＶＴ（変圧器）または、ＣＴ（変流器）が設置されている場合に設定します。設定したＶＴ
比とＣＴ比は、電圧と電流に関するすべての測定値に対して掛け合わせます。
「ＶＴ比」または「ＣＴ比」を表示して、中央の「モードスイッチ」を押すと表示が 4 桁に変わり、入力できる桁の数字を点滅します。設定は、０．００１～９９９９の範囲で行ってください。

点滅している数字が、変更中の位置です。

- 17 -

̶ 17 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

ＶＴ／ＣＴを使用した測定

KEW 2060BT/2062BT

表示切換スイッチ [◀▶ ]の短押しで、数字を 1 カウントアップまたは、1 カウントダウンします。表
示切換スイッチ [◀▶ ]を長押しすると、変更桁を 1 桁上位（左方向）または、1 桁下位（右方向）に移動します。なお、点滅している数字が左端の位置から変更桁を上位に移動した場合、または点滅している数字が右端の位置から変更桁を下位に移動した場合には、小数点の位置のみが移動します。変更中に中央の「モードスイッチ」を長押しすると、入力をキャンセルして設定を
1.000 に戻します。
変更後に中央の「モードスイッチ」を短押しすると、変更を決定して移動画面に戻ります。

ＶＴ／ＣＴを使用した測定
危険● 測定カテゴリに準じて 2060BT では CAT Ⅳ AC600V、CAT Ⅲ AC1000V より高い電圧、2062BT では
CAT Ⅳ AC300V、CAT Ⅲ AC600V、CAT Ⅱ AC1000V より高い電圧のある回路では絶対に使用しないでください。
● 本製品は必ずＶＴ（変圧器）、ＣＴ（変流器）の二次側で使用してください。
● 通電中はＣＴの二次側が開放しないよう充分注意してください。万一開放状態になりますと、二次側に高電圧が発生して大変危険です。

注意● ＶＴ、ＣＴを使用する場合には本製品の確度を保証できません。本製品の確度にＶＴ、ＣＴの確度、位相特性等を考慮してから測定してください。
測定ラインの電圧値や電流値が本製品の最大測定レンジを超える場合、下記のように特定ラインの電圧値、電流値に適した仕様のＶＴ、ＣＴを使用して二次側を測定することによって、一次側の値を表示させることができます。
単相２線（1P2W）の例

L
電源

負荷

N

CT

VT

i

V

KEW 2060BT/2062BT

KEW 2060BT/2062BT- 18 -

̶ 18 ̶

V

ブザー ON ／ OFF

KEW 2060BT/2062BT

KEW 2060BT/2062BT

3.6 測定コードの接続(本体)

ブザー ON／OFF
スイッチ操作した時、および検相結果のブザー音あり、なしを選択します。乾電池の警告ブザーと、
オートパワーオフの起動ブザーとは、この設定に関係しません。
「ブザー」を表示して中央の「モードスイッチ」を押すと、ＯＮ(on)/ＯＦＦ(oF)が点滅し初めて、変更が可能になります。
表示切換スイッチ [◀▶ ] ：on：ブザー音あり

oＦ：ブザー音なし[◀▶ ]変更後に中央の「モードスイッチ」を押すと、決定して移動画面に戻ります。

バックライト ON／OFF
無操作状態が続いた場合に、自動でバックライトをＯＦＦするか、ＯＦＦしないか選択します。
「バックライト」を表示して中央の「モードスイッチ」を押すと、ＯＮ(on)/ＯＦＦ(oF)が点滅し初めて、変更が可能になります。
表示切換スイッチ [◀▶ ] ：on：5 分後に OFF

oＦ：自動で OFF しない[◀▶ ]変更後に中央の「モードスイッチ」を押すと、決定して移動画面に戻ります。

KEW 2060BT/2062BT- 20 -

̶ 19 ̶

KEW 2060BT/2062BT

公称電圧の周波数

KEW 2060BT/2062BT

公称電圧の周波数
測定対象の電源周波数を設定します。

注記● 高調波は設定した周波数を基に演算しています。必ず測定対象の電源周波数と同じ周波数を設定してください。設定した周波数と公称電圧の周波数とが異なると、正確に測定できません。
「公称電圧の周波数」を表示して中央の「モードスイッチ」を押すと、.50〔Hz〕/.60〔Hz〕が点滅し初めて、変更が可能になります。
表示切換スイッチ [◀▶ ] ： 周波数の変更[◀▶ ]変更後に中央の「モードスイッチ」を押すと、決定して移動画面に戻ります。
設定した公称周波数は高調波測定時に左の画面のように表示されます。

システムリセット設定を出荷時の※状態に初期化します。

※

16 頁「項目の選択（移動）」参照「システムリセット」を表示して中央の「モードスイッチ」を押すと、n：キャンセルが点滅し初めて、選択が可能になります。
表示切換スイッチ [◀▶ ] ：.n：キャンセル.y：システムリセット[◀▶ ]「.ｙ」を表示して中央の「モードスイッチ」を押すと、システムリセットを行い移動画面に戻ります。
「.n」を表示して中央の「モードスイッチ」を押すと、操作をキャンセルし、なにもせずに移動画面に戻ります。

KEW 2060BT/2062BT

KEW 2060BT/2062BT- 20 -

̶ 20 ̶

6章 測定ファンクションごとの表示項目

KEW 2060BT/2062BT

6章 測定ファンクションごとの表示項目
6.1

実効値・周波数測定

Bluetooth の通信で、アプリケーション側（スマートフォン、タブレット）に「波形」を表示していると、本体は右の画面を表示して、測定値を表示しません。
本体に測定値を表示するには、アプリケーション側を「波形」から「測定値」に切り換えるか、Bluetooth の通信を切断してから確認してください。

電流実効値、周波数
ファンクションスイッチをに合わせてください。

電流レンジは、測定値によって自動で切り替わります。

「モードスイッチ」短押し ： 表示モードの切り換え
瞬時値、最大値(MAX)、最小値(MIN)、平均値(AVG)、波高値(|PEAK|)
上記それぞれの値は、押下時点からの測定値を計算します。

※「モードスイッチ」長押し ： 測定値のリセット（MAX、MIN、AVG、|PEAK|）
瞬時値

MAX（最大値）
モード
スイッチ

MIN（最小値）
モード
スイッチ

モードスイッチ

モードスイッチ|PEAK|（波高値）

AVG（平均値）
モード
スイッチ

MAX、MIN、AVG、|PEAK|を表示している状態のレンジは固定です。瞬時値の表示に戻すとオートレ
ンジに換わります。

- 21 -

̶ 21 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

電圧実効値、周波数

KEW 2060BT/2062BT

注記● PEAK 測定モードで測定する場合、電流センサを開閉すると表示値が変化することがあります。被測
定導体をクランプしたまま表示値を読み取るか、データホールド機能を使用して表示を固定した後、被測定導体から取り外して表示を読み取ってください。
● 本製品の PEAK 測定モードでは、1 周期分のサンプリングを行い、その中での PEAK 値（波高値）を表示しています。サンプリングは 0.5s に一度だけ行われるため、インラッシュカレントのような突発的な信号を測定することはできません。

電圧実効値、周波数
ファンクションスイッチをに合わせてください。

「モードスイッチ」短押し ： 表示モードの切り換え
瞬時値、最大値(MAX)、最小値(MIN)、平均値(AVG)、波高値(|PEAK|)
上記それぞれの値は、押下時点からの測定値を計算します。

※「モードスイッチ」長押し ： 測定値のリセット（MAX、MIN、AVG、|PEAK|）
瞬時値

MAX（最大値）
モード
スイッチ

MIN（最小値）
モード
スイッチ

モードスイッチ

モードスイッチ|PEAK|（波高値）

AVG（平均値）
モード
スイッチ

注記
● 本製品の PEAK 測定モードでは、1 周期分のサンプリングを行い、その中での PEAK 値（波高値）を表示しています。サンプリングは 0.5s に一度だけ行われるため、インラッシュカレントのような突発的な信号を測定することはできません。
KEW 2060BT/2062BT

KEW 2060BT/2062BT- 22 -

̶ 22 ̶

6.2 単相・三相（平衡）電力測定

6.2

KEW 2060BT/2062BT

単相・三相（平衡）電力測定
注記● 異容量三相 4 線式（Ｖ結線、Δ結線）は測定できません。単相に分けて測定してください。
ファンクションスイッチをに合わせてください。

設定画面での結線方式
単相２線式(１Ｐ２Ｗ)

三相３線式（３Ｐ３Ｗ） 三相４線式（３Ｐ４Ｗ）
平衡
平衡

単相２線式（１Ｐ２Ｗ）結線図

単相３線式（１Ｐ３Ｗ）結線図
単相３線式（１Ｐ３Ｗ）測定時には、単相２線式（１Ｐ２Ｗ）を選択して「Ｌ１」と「Ｌ２」の電力を別々に測定してください。１Ｐ３Ｗの総合電力は表示できません。

- 23 -

̶ 23 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

三相３線式（３Ｐ３Ｗ）平衡 結線図

KEW 2060BT/2062BT

三相３線式（３Ｐ３Ｗ）平衡 結線図

三相４線式（３Ｐ４Ｗ）平衡 結線図

KEW 2060BT/2062BT

KEW 2060BT/2062BT- 24 -

̶ 24 ̶

表示切り換え

KEW 2060BT/2062BT

表示切り換え「表示項目切換スイッチ [◀▶ ] 」
短押し ： 表示する測定値の切り換え
有効電力・力率、有効電力・電圧電流位相差、有効・皮相電力、有効・無効電力、電流・電圧実効値
有効電力・力率

有効電力・電圧電流位相差[◀▶ ]有効・皮相電力

[◀▶ ]

[◀▶ ]

[◀▶ ]

電流・電圧実効値

有効・無効電力[◀▶ ]「モードスイッチ」
短押し ： 表示モードの切り換え

瞬時値、最大値(MAX)、最小値(MIN)、平均値(AVG)※
上記それぞれの値は、押下時点からの測定値を計算します。
長押し ： 測定値のリセット（MAX、MIN、AVG）
例） 有効電力・力率※の表示画面※
上段と下段に表示している、それぞれの測定値が、すべての画面で両方同時に切り換ります。
瞬時値

MAX（最大値）
モード
スイッチ

モードスイッチ

モードスイッチ

AVG（平均値）

MIN（最小値）
モード
スイッチ

MAX、MIN、AVG を表示している状態のレンジは固定です。瞬時値の表示に戻すとオートレンジに換わります。

- 25 -

̶ 25 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

6.3 三相（不平衡）電力測定

KEW 2060BT/2062BT

6.3

三相（不平衡）電力測定
注記● 異容量三相 4 線式（Ｖ結線、Δ結線）は測定できません。単相に分けて測定してください。
ファンクションスイッチをに合わせてください。

設定画面での結線方式
三相３線式（３Ｐ３Ｗ）
不平衡

三相４線式（３Ｐ４Ｗ）
不平衡

三相３線式（３Ｐ３Ｗ）不平衡
測定手順と構成
Ｒ（Ｌ１）相への結線[▶ ]モードスイッチ

KEW 2060BT/2062BT

KEW 2060BT/2062BT

測定結果の表示[◀▶ ]Ｔ（Ｌ３）相への結線

[▶ ]

[▶ ]

モードスイッチ- 26 -

̶ 26 ̶

[▶ ]三相３線式（３Ｐ３Ｗ）不平衡

KEW 2060BT/2062BT

Ｒ（Ｌ１）相への結線
１回目の測定指示画面を表示した状態で、下「結線図」のように結線します。

結線後に「表示項目切換スイッチ [▶ ] 」を短押しすると、R（Ｌ１）相の有効電力を表示し、「モードスイッチ」の短押しごとに、R（Ｌ１）相の有効電力と電圧・電流実効値とを切り換えて表示します。
モード
スイッチ

再度「表示項目切換スイッチ [▶] 」を短押しすると、測定対象を R（Ｌ１）相から T（Ｌ３）相へ移行します。

Ｔ（Ｌ３）相への結線
２回目の測定指示画面を表示した状態で右「結線図」のように、電流センサのみをＴ（Ｌ３）相に移動します。測定コードは移動しないでください。

結線後に「表示項目切換スイッチ [▶ ] 」を短押しすると、Ｔ（Ｌ３）相の有効電力を表示し、「モードスイッチ」の短押しごとに、Ｔ（Ｌ３）相の有効電力と電圧・電流実効値とを切り換えて表示します。
モード
スイッチ

再度「表示項目切換スイッチ [▶ ] 」を短押しすると、測定結果の表示画面に移行します。

- 27 -

̶ 27 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

三相３線式（３Ｐ３Ｗ）不平衡

KEW 2060BT/2062BT

測定結果の表示「表示項目切換スイッチ [◀▶ ] 」
短押し ： 表示する測定値の切り換え
総合有効電力R（Ｌ１）相の有効電力[◀▶ ]Ｔ（Ｌ２）相の有効電力

[◀▶ ]

[◀▶ ]
「モードスイッチ」を長押しすると、測定値をリセットして１回目の測定指示画面に戻ります。

KEW 2060BT/2062BT

KEW 2060BT/2062BT- 28 -

̶ 28 ̶

三相４線式（３Ｐ４Ｗ）不平衡

KEW 2060BT/2062BT

三相４線式（３Ｐ４Ｗ）不平衡
測定手順と構成
Ｒ（Ｌ１）相への結線

Ｓ（Ｌ２）相への結線[▶ ]モードスイッチＴ（Ｌ３）相への結線

[▶ ]

[▶ ]

[▶ ]

[▶ ]

モードスイッチ[▶ ]モードスイッチ

測定結果の表示[◀▶ ]- 29 -

̶ 29 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

三相４線式（３Ｐ４Ｗ）不平衡

KEW 2060BT/2062BT

Ｒ（Ｌ１）相への結線
１回目の測定指示画面を表示した状態で、右「結線図」のように結線します。

結線後に「表示項目切換スイッチ [▶ ] 」を短押しすると、
R（Ｌ１）相の有効電力を表示し、「モードスイッチ」の短押
しごとに、R（Ｌ１）相の有効電力と電圧・電流実効値とを切り換えて表示します。
モード
スイッチ
再度「表示項目切換スイッチ [▶ ] 」を短押しすると、測定対象を R（Ｌ１）相からＳ（Ｌ２）相へ移行します。

Ｓ（Ｌ２）相への結線
２回目の測定指示画面を表示した状態で、電流センサと測定コード（赤）とを、右「結線図」のようにＳ（Ｌ２）相に移動します。

結線後に「表示項目切換スイッチ [▶ ] 」を短押しすると、Ｓ（Ｌ２）相の有効電力を表示し、「モードスイッチ」の短押しごとに、Ｓ（Ｌ２）相の有効電力と電圧・電流実効値とを切り換えて表示します。
モード
スイッチ
再度「表示項目切換スイッチ [▶] 」を短押しすると、測定対象をＳ（Ｌ２）相からＴ（Ｌ３）相へ移行します。

KEW 2060BT/2062BT

KEW 2060BT/2062BT- 30 -

̶ 30 ̶

三相４線式（３Ｐ４Ｗ）不平衡

KEW 2060BT/2062BT

Ｔ（Ｌ３）相への結線
３回目の測定指示画面を表示した状態で、電流センサと測定コード（赤）とを、右「結線図」のようにＴ（Ｌ３）相に移動します。

結線後に「表示項目切換スイッチ [▶ ] 」を短押しすると、Ｔ（Ｌ３）相の有効電力を表示し、「モードスイッチ」の短押
しごとに、Ｔ（Ｌ３）相の有効電力と電圧・電流実効値とを切り換えて表示します。
モード
スイッチ
再度「表示項目切換スイッチ [▶ ] 」を短押しすると、測定結果の表示画面に移行します。

測定結果の表示「表示項目切換スイッチ [◀▶ ] 」
短押し ： 表示する測定値の切り換え
有効電力・力率

有効・皮相電力[◀▶ ]有効・無効電力
[◀▶ ]

[◀▶ ]

[◀▶ ]

Ｔ（Ｌ3）相の有効電力

S（Ｌ２）相の有効電力[◀▶ ]R（Ｌ１）相の有効電力[◀▶ ]「モードスイッチ」を長押しすると、測定値をリセットして１回目の測定指示画面に戻ります。

- 31 -

̶ 31 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

6.4 高調波測定

KEW 2060BT/2062BT

6.4

高調波測定
ファンクションスイッチをに合わせてください。

高調波ファンクションにすると、右の画面のように公称周波数が表示されます。公称周波数
は SET UP ファンクションで 50Hz または 60Hzに切り替えることができます。（20 頁「公称電圧の周波数参照）
Bluetooth の通信中には右の画面を表示して、本体に測定値を表示しません。
測定値は、通信しているアプリケーション側（スマートフォン、タブレット）で確認す
るか、Bluetooth の通信を切断してから確認してください。

電流高調波歪み率、含有率、実効値
電流レンジは、測定値によって自動で切り替わります。
表示単位が「V」のときは、「電圧高調波」の測定画面です。中央の「表示項目切換スイッチ[▶ ]」を長押しして、表示単位を「A」にしてから測定してください。
[▶ ]
長押し

「表示項目切換スイッチ [◀▶ ] 」
短押し ： 表示する測定値の切り換え
実効値・高調波歪み率 THD-F、実効値・高調波歪み率 THD-R、
1 次基本波実効値・含有率～30 次高調波実効値・含有率
実効値・高調波歪み率 THD-F

実効値・高調波歪み率 THD-R
[◀▶ ]

[◀▶ ]

[◀▶ ]

30 次基本波実効値・含有率～～

1 次基本波実効値・含有率[◀▶ ]上段は高調波次数表示(1h～30h)と、各高調波実効値とを、1 秒ごとに切り換えて表示します。
KEW 2060BT/2062BT

KEW 2060BT/2062BT- 32 -

̶ 32 ̶

電圧高調波歪み率、含有率、実効値

KEW 2060BT/2062BT「モードスイッチ」
短押し ： 表示モードの切り換え

瞬時値、最大値(MAX)、最小値(MIN)、平均値(AVG)※
上記それぞれの値は、押下時点からの測定値を計算します。
長押し ： 測定値のリセット（MAX、MIN、AVG）
例） 実効値・高調波歪み率 THD-F※の表示画面※
上段と下段に表示している、それぞれの測定値が、すべての画面で両方同時に切り換ります。
瞬時値

MAX（最大値）
モード
スイッチ

モードスイッチ

モードスイッチ

AVG（平均値）

MIN（最小値）
モード
スイッチ

MAX、MIN、AVG を表示している状態のレンジは固定です。瞬時値の表示に戻すとオートレンジに換わります。

電圧高調波歪み率、含有率、実効値
表示単位が「A」のときは、「電流高調波」の測定画面です。中央の「表示項目切換スイッチ[▶ ]」を長押しして、表示単位を「V」にしてから測定してください。
[▶ ]
長押し

- 33 -

̶ 33 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

電圧高調波歪み率、含有率、実効値

KEW 2060BT/2062BT「表示項目切換スイッチ [◀▶ ] 」

短押し ： 表示する測定値の切り換え
実効値・高調波歪み率 THD-F、実効値・高調波歪み率 THD-R、
基本波実効値・含有率～30 次高調波実効値・含有率
実効値・高調波歪み率 THD-F

実効値・高調波歪み率 THD-R
[◀▶ ]

[◀▶ ]

[◀▶ ]

30 次基本波実効値・含有率～～

基本波実効値・含有率[◀▶ ]上段には高調波次数表示(1h～30h)と、各高調波実効値とを、1 秒ごとに切り換えて表示します。
「モードスイッチ」
短押し ： 表示モードの切り換え

瞬時値、最大値(MAX)、最小値(MIN)、平均値(AVG)※
上記それぞれの値は、押下時点からの測定値を計算します。
長押し ： 測定値のリセット（MAX、MIN、AVG）
例） 実効値・高調波歪み率 THD-F※の表示画面※
上段と下段に表示している、それぞれの測定値が、すべての画面で両方同時に切り換ります。
瞬時値

MAX（最大値）
モード
スイッチ

モードスイッチ

モードスイッチ

AVG（平均値）

MIN（最小値）
モード
スイッチ

KEW 2060BT/2062BT

KEW 2060BT/2062BT- 34 -

̶ 34 ̶

高調波歪み率THD-R/THD-F

KEW 2060BT/2062BT

高調波歪み率 THD-R/THD-F
THD： 総合高調波歪み率の算出方法には２種類あり、基本波を基準とした総合高調波歪み率が「THD-F」、全実効値を基準とした総合高調波歪み率が「THD-R」です。
THD-F [%] =THD-R [%] =高調波実効値（2 次～）
基本波実効値(1 次)×100

高調波実効値（2 次～）
基本波実効値+高調波実効値

×100

どちらも電圧波形または電流波形の高調波レベルを定量化するのに使用する性能指標ですが、
THD-R で用いる演算式では、高調波が多い(歪みが大きい)と比較的誤差が大きくなります。つまり、測定環境の歪みが小さければ測定誤差は THD-R/F どちらも同じですが、歪みが大きいと THD-R に比べ THD-F の演算式の方が誤差は少なくなります。
古い測定器では THD-F の演算式に必要な基本波（1 次)のみの実効値を正確に測定することが難しかったために THD-R が性能指標の主流でしたが、今ではこれを正確に測定できるため、THD-R は
簡易的な測定に使用し、測定確度に対して高調波成分量による影響が少ない THD-F の歪み率を、一般的に多く使うようになりました。
負荷側の高調波原因を探査する現場では THD-R を使うことが多く、電源品質の管理を行う現場では
THD-F を主に使用する傾向にあります。

- 35 -

̶ 35 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

6.5 検相

KEW 2060BT/2062BT

6.5

検相
注記

● 異容量三相 4 線式（Ｖ結線、Δ 結線）のように、各相間電圧が不平衡の結線では、相順を正しく判定できません。
● SETUP ファンクション内の 19 頁「ブザー ON／OFF」を OFF に設定すると、判定後のブザー音がなりません。ブザーでの検相判定が必要な場合には、「ブザー ON／OFF」を ON に設定してください。
ファンクションスイッチをに合わせてください。

三相３線式、三相４線式の被測定結線の状態に準じて、下表のように判定結果を表示します。
各数字は結線した先の相順を表しています。
結線状態
正相

R(L1)

S(L2)

判定結果
ブザー

T(L3)

表示

活線
または
活線
接地
欠相、周波数異常、電圧有効入力範囲外、不平衡

１．２．３

断続：ピッピッピッ

３．２．１

連続：ピ――――

－.－.－

なし

活線

逆相
判定不能

KEW 2060BT/2062BT

KEW 2060BT/2062BT- 36 -

̶ 36 ̶

7章 その他の機能

KEW 2060BT/2062BT

7章 その他の機能「データホールド機能」
データホールドスイッチを押すと LCD 左上にを表示して、現在の表示を保持します。この状態でも測定は継続していますが、表示は更新しません。また、最大値、最小値、平均値、波高値等の表示を切り替えてデータホールド時の測定値を参照できます。再度データホールドスイッチを押すとを消して、表示の更新を再開します。
なお、ファンクションを切り換えると、データホールドを解除してから切り換え後のファンクションで測定を開始します。

「自動バックライトオフ」
なにも操作しない状態が 5 分間経過すると自動的にバックライトを消灯します。再点灯する場合には、表示項目切換スイッチ[◀ ]を長押しすることで、再度 5 分間点灯します。なお、点灯中に表示項目切換ス
イッチ[◀ ]を長押しするとバックライトは消灯します。
SETUP ファンクション内の 19 頁「バックライト ON／OFF」を OFF に設定すると、なにも操作しない状態
が 5 分間経過しても点灯し続けます。この状態からバックライトを消灯するには、表示項目切換スイッチ[◀ ]を長押してください。

「オートパワーオフ」
注記● ファンクションスイッチが測定機能の位置にある状態で電源が切れている時は、オートパワーオフの機能で、自動的に電源が切れている状態です。
Bluetooth で通信している状態を除いて、なにも操作しない状態が 15 分間経過すると、ブザーを 4 回鳴動し(ピー､ピー､ピー､ピー)自動的に本体の電源をオフします。再度電源をオンしたい場合には、ファン
クションスイッチを一旦 OFF の位置にしてから、測定ファンクションの位置に移動してください。

「電流オートレンジ」
測定中の電流実効値に準じて、自動で電流レンジを切り替えます。
切り替えは、現在レンジの実効値 110%または PEAK 値 300％を越える場合にレンジをアップし、現レンジにおいて下位レンジの実効値 90%未満になった場合に、レンジをダウンします。
ただし、表示モードで MAX(最大値)、MIN(最小値)、AVG(平均値)、|PEAK|(波高値)を選択した場合には、選択時のレンジに固定します。

- 37 -

̶ 37 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

KEW 2060BT/2062BT

8章 Bluetoothを使用する

8章 Bluetooth を使用する
警告● Bluetooth 通信で使用する電波は医療電子機器などの動作に影響を与える恐れがあります。医療機器のある場所で使用する場合は誤作動を起こさないか充分注意してください。

注記● 本器もしくはタブレットの付近に無線 LAN(IEEE802.11.b/g)を搭載した機器がある場合、電波干渉によって接続ができなくなったり、本器からタブレットへのデータ転送速度が遅くなり、本器指針の動作とタブレットの表示更新に著しい時間差が生じる場合があります。この場合、無線 LAN 機器を本器・タブレットからなるべく遠ざけるか、電源を切る、もしくは本器とタブレットの距離をなるべく近づけてください。
● 本器もしくはタブレットのどちらかを金属製のボックス内に入れて通信を行おうとすると、接続が確立できない場合があります。測定場所を変更するか、本器・タブレット間に金属板がない状態で使用してください。
● Bluetooth 通信時に情報の漏えいが発生しましても、弊社としては一切の責任を負いかねますのでご了承ください。
● 専用アプリが動作するタブレットであっても、機種によっては本器との接続が確立できない場合があります。ほかのタブレットで接続を試しても接続ができない場合、故障が考えられますので、弊社サービスセ
ンターもしくはお買い上げの販売店までご連絡ください。
● Bluetooth ワードマークは、Bluetooth SIG ,Inc.の所有です。共立電気計器はライセンスに基づき使用しています。
● Android、Google Play ストア、Google Map は Google Inc.の商標または登録商標です。
● iOS は Cisco の商標または登録商標です。
● App Store は Apple Inc.のサービスマークです。
●™マーク、 ®マークは明記していません。

Bluetooth 通信により、Android/iOS のタブレットと通信を行うことができます。タブレットに専用アプリ「KEWPower＊(アスタリスク)」をインストールすることにより、KEW 2060BT または KEW2062BT から離れた場所
からでも手元のタブレットで判定結果を確認することができます。
本機能を使用するためには、タブレットをインターネットに接続し、専用アプリ「KEW Power＊」をダウンロ
ードしていただく必要があります。また、一部機能についてはインターネットに接続した状態でなければご利用になれません。詳しくは、「8.1 専用アプリ「KEW Power＊(アスタリスク)」の機能」をご覧ください。

Bluetoothタブレット端末
スマートフォン

KEW 2060BT/2062BT

KEW 2060BT/2062BT- 38 -

̶ 38 ̶

8.1 専用アプリ「KEW Power *（アスタリスク）」の機能

8.1

KEW 2060BT/2062BT

専用アプリ「KEW Power＊(アスタリスク)」の機能

専用アプリ「KEW Power＊」は Android のタブレットでは Google Play ストア、iOS のタブレットでは App
Store で無料配信しています（インターネットに接続する必要があります)。ダウンロードや特定機能の使用にかかる通信費、インターネット接続料等はお客様の負担となりますのでご了承ください。なお、「KEWPower＊」は記録メディアによる配布は行っておりません。
「KEW Power＊」の代表的な機能としては、以下のようなものがあります。
(1)
(2)
(3)
(4)
(5)

測定結果をタブレットに表示することが可能
測定結果のデータ保存・表示機能
電圧と電流の入力波形表示機能
高調波の実効値と含有率のグラフ表示機能
測定値の良否判定機能- 39 -

̶ 39 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

9章 仕様KEW 2060BT/2062BT

9章 仕様
9.1

安全要求仕様使用環境： 屋内使用 高度 2000m 以下
確度保証温湿度範囲 ： 23℃±5℃ 相対湿度 85%以下（結露しないこと）
使用温湿度範囲：-10℃～50℃ 相対湿度 85%以下（結露しないこと）
保存温湿度範囲：-20℃～60℃ 相対湿度 85%以下（結露しないこと）
耐電圧：KEW2060BT は、AC7000V／5 秒間 電流センサと外装間または電気回路と外装間
KEW2062BT は、AC5160V／5 秒間 電流センサと外装間または電気回路と外装間
絶縁抵抗：50MΩ 以上／1000V 電気回路と外装間
適合規格： IEC 61010-1/-2-032(本体)/-031(測定コード)
KEW2060BT は、測定カテゴリ CAT Ⅳ600V CAT Ⅲ1000V 汚染度 2、KEW2062BT は、測定カテゴリ CAT Ⅳ300V CAT Ⅲ600V CAT Ⅱ1000V
汚染度２
IEC61326（EMC) Class B、EN50581(RoHS)、
EN 301 489-1、EN 300 328、EN 62479
防塵/防水性：IEC 60529 IP40

9.2

一般仕様
表示更新周期： , , ,
ファンクション…約 0.5 秒
ファンクション…約 1 秒
被測定可能導体
KEW2060BT：Φ75ｍｍまでの円形導体及び 80×30mm 以下の柱状導体(ブスバー)
KEW2062BT：Φ55ｍｍまでの円形導体
外形寸法
KEW2060BT：283（Ｌ）×143（Ｗ）×49（Ｄ）mm
KEW2062BT：247（Ｌ）×105（Ｗ）×49（Ｄ）mm
質量
KEW2060BT：約 590g（電池含む）
KEW2062BT：約 490g（電池含む）
付属品： 測定コード MODEL7290(赤､黒､黄､各１本（ワニグチ付き））······ 1 ｾｯﾄ
取扱説明書········································································································ 1 冊
単３形ｱﾙｶﾘ乾電池（LR6）············································································ 2 個
ソフトケース MODEL9198 ··········································································· 1 個
連続使用時間：約 58 時間(
ファンクション、連続、無負荷、バックライト OFF、
単三形アルカリ乾電池（LR6）×2 個)
消費電流
外部通信

KEW 2060BT/2062BT

KEW 2060BT/2062BT： 35mA typ.(@3.0V、
ファンクション)：Bluetooth® Ver5.0 準拠- 40 -

̶ 40 ̶

9.3 測定仕様

9.3

KEW 2060BT/2062BT

測定仕様
交流電流ファンクション
交流電流実効値（ACA）〔Arms〕・波高値（絶対値） 〔A|peak|〕
レンジ

表示桁数
サンプリング期間
サンプリング
周波数
有効入力範囲表示範囲

40.00A／400.0A／1000A※
レンジ固定なしのオートレンジのみ
現レンジの実効値 110%または PEAK 値 300％を越える場合にレンジをアップする。現レンジにおいて下位レンジの実効値 90%未満になった場合にレンジをダウンする。ただし、表示モードで MAX(最大値)、MIN(最小値)、AVG(平均値)、|PEAK|(波高値)を選択した場合には、現レンジ固定になる。
4桁
1 周期／500ms
32.8kHz(30.5μs 間隔)
PEAK 値：周波数測定値 40.0Hz～70.0Hz 間のみ 9 ポイントの移動平均
40.00A レンジ 実効値：0.60A～40.00A、 PEAK 値：±(0.6A～56.57A)
400.0A レンジ 実効値：6.0A～400.0A、
PEAK 値：±(6.0A～565.7A)
1000A レンジ 実効値：60A～999.9A、
PEAK 値：±(60A～1414A)
実効値：
40.00A レンジ ：0.30～44.00A
400.0A レンジ ：3.0(36.0A※)～440.0A
1000A レンジ ：30(360A※)～1100A※
オートレンジでは”()”内の範囲を表示、0.30A 未満はゼロ、1100A を超えるとOL を表示する
PEAK 値(絶対値)：

クレストファクタ
確度

40.00A レンジ ：0.30A～120.0A
400.0A レンジ ：3.0A～1200A
1000A レンジ ：30A～1500A
40.00A／400.0A レンジは 3 以下，1000A レンジは 3 以下 1500Apeak
実効値：正弦波の測定波形において 40.0～70.0Hz： ±1.0％rdg±3dgt
70.1～1kHz： ±2.0％rdg±5dgt※
40～70Hz の正弦波以外は確度に±0.5％rdg±5dgt を加算する
PEAK 値(絶対値)：
40.0～70.0Hz： ±2.5％rdg±5dgt
70.1～1kHz： ±4.0％rdg±5dgt

演算式
n-1

1
A=�� ��(Ai )2 ��ni ：サンプリングポイント№
n：1 周期間でのサンプル数

i=0

電流周波数（Ａｆ） 〔Hz〕
表示桁数

4桁

確度

±0.3％rdg±3dgt (40.0Hz～999.9Hz、正弦波)

測定有効入力範囲

ACA の有効入力範囲

表示範囲

40.0～999.9Hz(表示範囲外または、ACA の表示範囲外では ”----” 表示)- 41 -

̶ 41 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

交流電圧ファンクション

KEW 2060BT/2062BT

交流電圧ファンクション
交流電圧実効値（ACV）〔Vrms〕・波高値（絶対値） 〔V|peak|〕
レンジ
表示桁数
サンプリング期間
サンプリング
周波数
有効入力範囲表示範囲
クレストファクタ
確度

入力
インピーダンス
演算式

1000V
4桁
1 周期／500ms
32.8kHz(30.5μs 間隔)
PEAK 値：周波数測定値 40.0Hz～70.0Hz 間のみ 9 ポイントの移動平均
実効値：30.0V～999.9V
PEAK 値：±(30.0V～1414V)
実効値：30.0V～1100V
PEAK 値(絶対値)：30.0V～1555V※
下限値未満は Lo、上限値を超えると OL を表示する
1.7 以下
実効値：正弦波の測定波形において 40.0～70.0Hz： ±0.7％rdg±3dgt
70.1～1kHz： ±3.0％rdg±5dgt※
40～70Hz の正弦波以外は確度に±0.5％rdg±5dgt を加算する
PEAK 値(絶対値)：
40.0～70.0Hz： ±2.5％rdg±5dgt
70.1～1kHz： ±4.0％rdg±5dgt
約４MΩ ※各端子間の抵抗値

n-1

1
V=�� ��(Vi )2 ��n

L＝V1, N＝V3 と接続し測定
i ：サンプリングポイント№
n：1 周期間でのサンプル数

i=0

電圧周波数（Ｖｆ） 〔Hz〕
表示桁数

4桁

確度

±0.3％rdg±3dgt (40.0Hz～999.9Hz、正弦波)

測定有効入力範囲

ACV の有効入力範囲 (実効値：30.0V～999.9V)

表示範囲

40.0～999.9Hz(表示範囲外または、ACV の表示範囲外では ”----” 表示)

信号ｿｰｽ

KEW 2060BT/2062BT

KEW 2060BT/2062BT

V1-V3 (端子間電圧) または A- 42 -

̶ 42 ̶

電力ファンクション

KEW 2060BT/2062BT

電力ファンクション
有効電力（P）〔W〕
レンジ
表示桁数
サンプリング期間
サンプリング
周波数
有効入力範囲※40.00kW／400.0 kW／1000 kW
電流レンジに準じて切り替える
4桁(確度保証周波数外では ”----” 表示)
1 周期／500ms

表示範囲

40.00ｋＷレンジ ：0.00～44.00kW
400.0kW レンジ ：0.0(36.0kW)～440.0kW
1000kW レンジ ：0(360kW)～1210kW※
表示モードで MAX(最大値)、MIN(最小値)、AVG(平均値)を選択した場合には、現レンジ固定になる。

32.8kHz(30.5μs 間隔)
電圧実効値、電流実効値の有効入力範囲でかつ、45～65Hz の周波数範囲

※

確度
極性表示
演算式

オートレンジでは”()”内の範囲を表示、電圧実効値、電流実効値の表示範囲外

および、有効入力周波数外では”----” を表示する
正弦波、力率 1 において±1.7%rdg±5dgt
位相角の影響±3.0°以内※
Sum 値は各測定チャンネルの誤差を合算した確度、3P3W は 2 倍、3P4W は 3 倍
消費(流入)：符号なし、回生(流出)：－V 基準としたn-1
i：サンプリングポイント№1
n：1
周期間でのサンプル数
P = ��(Vi ×Ai )�

n

対象結線

i=0

表示値

接続先

1P2W・1P3W

P

平衡 3P3W
平衡 3P4W
不平衡 3P3W

Psum(=Ssum×cos(θ))
Psum(＝P×3)
P1、P2 、Psum(＝P1+P2)

不平衡 3P4W

各相測定時のみ表示：P1 、P2 、P3
合算値：Psum(＝P1+P2+P3)

1P2W：L＝V1・A，N=V3
1P3W：L1/L2＝V1・A ，N=V3
R=V1・A，S=V3
R=V1・A，N=V3
P1：R=V1・A，S=V2，T=V3
P2：R=V1 ，S=V2，T=V3・A※
2 回接続先を変更して測定(2 電力計法)
P1：R=V1・A，N=V3
P2：S=V1・A，N=V3
P3：T=V1・A，N=V3※
3 回接続先を変更して測定- 43 -

̶ 43 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

電力ファンクション

KEW 2060BT/2062BT

皮相電力(S)〔VA〕
レンジ
表示桁数
有効入力範囲
確度

極性表示
演算式

有効電力と同じ
有効電力と同じ
有効電力と同じ
各測定値からの演算に対して±1dgt※
Sum 値は各測定チャンネルの誤差を合算した確度、3P3W は±2dgt、3P4W は±3dgt
極性なし
S＝V×A

※

P＞S 時は P＝S とする。

対象結線
1P2W・1P3W
平衡 3P3W
平衡 3P4W

表示値
S
Ssum(＝S×√3)

不平衡 3P4W

Ssum(＝S1+S2+S3)

接続先
有効電力と同じ

Ssum(＝S×3)※

Sn：n 回目測定の皮相電力値

無効電力（Q）〔Var〕
レンジ
表示桁数
有効入力範囲
確度
極性表示
演算式

有効電力と同じ
有効電力と同じ
有効電力と同じ
各測定値からの演算に対して±1dgt※
平衡 3P3W 測定時は±2dgt、平衡 3P4W 測定時は±3dgt
遅れ：符号なし、進み：－
Q＝�S2 -P2※

|P|＞S 時は Q＝0 とする。
極性符号は電圧電流位相差(θ)によって下記のように付加する
0°～－90°～180°の範囲：符号なし (＋) 遅れ
0°～＋90°～180°の範囲：マイナス (－) 進み
表示値
接続先
Q
有効電力と同じ※

対象結線
1P2W・1P3W
平衡 3P3W
平衡 3P4W
不平衡 3P4W

Qsum(＝�Ssum 2 − Psum 2 )Qsum(＝Q×3)

Qsum(＝Q1+Q2+Q3)※

Qn：n 回目測定の皮相電力値

KEW 2060BT/2062BT

KEW 2060BT/2062BT- 44 -

̶ 44 ̶

電力ファンクション

KEW 2060BT/2062BT

力率(PF)
有効入力範囲
表示範囲
確度
極性表示
演算式

有効電力と同じ-1.000～0.000～1.000
各測定値からの演算に対して±1dgt※
平衡 3P3W 測定時は±2dgt、平衡 3P4W 測定時は±3dgt
遅れ：符号なし、進み：－
PF＝�P�S� ただし、平衡 3P3W のみ PF = cos (θ)※
三相回路測定時は sum 値から算出する。
※
S＝0 時は非表示とする。
※
極性符号は電圧電流位相差(θ)によって下記のように付加する
0°～－90°～180°の範囲：符号なし(＋) 遅れ
0°～＋90°～180°の範囲：マイナス(－) 進み※平均値は力率=1 を基準とし、進み・遅れの量を平均化して演算する。
【演算例】
PF=0.99、-0.92、+0.96 の 3 つの測定値ではそれぞれ、
0.99 と 1 との差=-0.01 進み、-0.92 と 1 との差=+0.08 遅れ、
0.96 と 1 との差=-0.04 進み、となり、各差分を合計すると-0.01+0.08+(-0.04)=0.03 遅れとなる。
これを測定数の 3 回で割り平均化すると 0.03÷3=0.01 遅れとなり、求める PF の平均値は 1 に対し 0.01 遅れているため、-0.99 進みが PF の平均値になる。

電圧電流位相差(θ)〔deg〕 (単相２線式測定時のみ)
表示範囲
極性表示
測定方式

※
-180.0～0.0～+179.9
有効電力の表示範囲外では ”----” を表示する
進み：符号なし、遅れ：－
電圧波形に対する電流波形のゼロクロス位置を比べて差を測定する※
S＝0 時は非表示とする。
※
極性符号は電圧の位相を基準(0°)とした電流の位相角を表示する
符号なし(＋) 進み、マイナス(－) 遅れ- 45 -

̶ 45 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

高調波ファンクション

KEW 2060BT/2062BT

高調波ファンクション
測定方式

接続先

周波数固定サンプリング
50/60Hz それぞれの入力 1 周期間に 256 回サンプリングを行い FFT 演算を実施する。サンプリング周波数は公称周波数設定値に応じて切り替える。
50Hz 設定…12.8ksps(78μs 間隔)，60Hz 設定…15.4ksps(65μs 間隔)
L＝V1，N=V3，L・R・S・T(電力配線へクランプ)=A

有効周波数50/60Hz

解析次数1～30 次ウインドウ幅1 周期

ウインドウの種類

レクタンギュラ

解析データ数256 ポイント

解析レート

1 回/1 秒

高調波電圧実効値（Vk：１次基本波～30 次高調波）〔Vrms〕
レンジ、表示桁数、有効入力範囲
電圧実効値に同じ
表示範囲
電圧実効値に同じ ※含有率 0.0%～100.0% 基本波に対する割合
確度
実効値
1～10 次 ：±5.0％rdg±10dgt
11～20 次 ：±10％rdg±10dgt
21～30 次 ：±20％rdg±10dgt
含有率
各測定値からの演算に対して±1dgt
演算式
L＝V1, N＝V3 と接続し測定
1

Vk = � �(Vkr)2 + (Vki)2
n=-1

含有率 =

※

KEW 2060BT/2062BT

KEW 2060BT/2062BT

Vk×100V1(基本波)- 46 -

̶ 46 ̶

k ：高調波の次数
Vr ：電圧 FFT 変換後の実数成分
Vi ：電圧 FFT 変換後の虚数成分

高調波ファンクション

KEW 2060BT/2062BT

高調波電流実効値（Ak：１次基本波～30 次高調波）〔Arms〕
レンジ、表示桁数、有効入力範囲
電流実効値に同じ
表示範囲
電流実効値に同じ ※含有率 0.0%～100.0% 基本波に対する割合
確度
実効値
1～10 次 ：±5.0％rdg±10dgt
11～20 次 ：±10％rdg±10dgt
21～30 次 ：±20％rdg±10dgt
含有率
各測定値からの演算に対して±1dgt
演算式
1Ak＝� �(Akr)2 + (Aki)2
n=-1

※

含有率＝

k ：高調波の次数
Ar ：電流 FFT 変換後の実数成分
Ai ：電流 FFT 変換後の虚数成分

Ak×100A1(基本波)

総合高調波電圧歪み率(V THD-F) 〔%〕
表示桁数

4桁

表示範囲

0.0%～100.0%

確度

各測定値からの演算に対して±1dgt

演算式

V THD-F ＝2
�∑30
k=2(Vk) ×100

V ：高調波電圧k ：高調波の次数V1(基本波)

総合高調波電流歪み率(A THD-F) 〔%〕
表示桁数

4桁

表示範囲

0.0%～100.0%

確度

各測定値からの演算に対して±1dgt

演算式

A THD-F =2
�∑30
k=2(Ak) ×100

A : 高調波電流k : 高調波の次数A1(基本波)- 47 -

̶ 47 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

検相ファンクション

KEW 2060BT/2062BT

総合高調波電圧歪み率(V THD-R) 〔%〕
表示桁数

4桁

表示範囲

0.0%～100.0%

確度

各測定値からの演算に対して±1dgt

演算式

V THD-R=2�∑30
k=2(Vk) ×1002�∑30
k=1(Vk)V : 高調波電圧k : 高調波の次数

総合高調波電流歪み率(A THD-R) 〔%〕
表示桁数

4桁

表示範囲

0.0%～100.0%

確度

各測定値からの演算に対して±1dgt

演算式

A THD-R =検相ファンクション
有効入力範囲表示

KEW 2060BT/2062BT

KEW 2060BT/2062BT2�∑30
k=2(Ak) ×1002�∑30
k=1(Ak)A : 高調波電流k : 高調波の次数

測定波形 45～65Hz の正弦波において、電圧実効値(ACV)80～1100V
各相電圧間の位相誤差がない状態において、電圧振幅 MAX－MIN の差が 10%以内
相電圧平衡状態において 位相誤差 3P4W (三相 4 線式) ±30°以内
3P3W (三相 3 線式) ±15°以内：正相、全相活線(1.2.3) ブザー断続鳴動：ピッピッピッピッ：逆相、全相活線(3.2.1) ブザー連続鳴動：ピ――――：判定不能(-.-.-) ブザーなし
欠相、周波数異常、電圧有効入力範囲外、不平衡等- 48 -

̶ 48 ̶

アフターサービス

KEW 2060BT/2062BT

アフターサービス●修理・校正を依頼されるにはお買い上げいただいた販売店または弊社サービスセンターにお送りください。
●製品のご使用に関するお問い合わせは弊社お客様相談室にご連絡ください。
●校正周期について
本製品を正しくご使用いただくため、定期的（推奨校正周期１年）に校正することをおすすめいたします。
●補修用部品の保有期間
本製品の機能・性能を維持するために必要な補修部品を製造打ち切り後、５年間を目安に保有しています。
この取扱説明書に記載している事項を断りなく変更することがあります。
ご了承ください。

■ホームページのご案内
www.kew-ltd.co.jp●新製品情報●取扱説明書／ソフトウェア／単品カタログのダウンロード●販売終了製品情報ご使用に関するお問い合わせは共立電気計器
電話受付時間

修理・校正に関するお問い合わせは

お客様相談室

共立電気計器営業時間

9:00 ～ 12:00、13:00 ～ 17:00

サービスセンター

8:40 ～ 12:00、13:00 ～ 17:30（土・日・祝日・年末年始・夏季休暇を除く）（土・日・祝日・年末年始・夏季休暇を除く）

0894-62-1172

0120-62-1172※折り返しお電話させていただくことがございますので発信者番号の通知にご協力いただきますようお願いいたします。
※フリーコールをご利用いただけない場合は、お手数をおかけしますが最寄りの弊社営業所へおかけ下さい。

修理を依頼される場合は事前に電池の消耗、ヒューズや測定コードの断線を確認してから輸送中に損傷しないように十分梱包した上で弊社サービスセンターまでお送りください。

送付先：〒797-0045 愛媛県西予市宇和町坂戸４８０- 49 -

̶ 49 ̶

KEW 2060BT/2062BT

KEW 2060BT/2062BT

保 証 書KEW 2060BT/2062BT
保証期間

ご購入日（

製造番号
年

月

日）より１年間共立製品をお買い上げいただきありがとうございます。保証期間内に正常なご使用状態で万一故障が生じた場合は、保証規定により無償修理をさせていただきます。
本書を添付の上ご依頼ください。
お名前
ご住所
ＴＥＬ
◎本保証書に製造番号、ご購入日、およびお名前、ご連絡先をご記入の上、大切に保管してください。
◎本保証書の再発行はいたしません。
◎本保証書は日本国内でのみ有効です。This warranty is valid only in Japan.
保証規定
保証期間内に生じました故障は無償で修理いたします。
ただし、下記事項に該当する場合は対象から除外させていただきます。
1． 取扱説明書と異なる不適切な取扱い、または使用方法が原因で発生した故障。
2． お買い上げ後の持ち運びや輸送の間に、落下させるなど異常な衝撃が加わって生じた故障。
3． 弊社サービス担当者以外による改造、修理が原因で生じた故障。
4． 火災、地震、水害、公害およびその他の天変地異が原因で生じた故障。
5． 傷など外観上の変化。
6． その他弊社の責任と見なされない故障。
7． 電池など消耗品の交換、補充。
8． 保証書のご提出がない場合。

10-21
03-21

92-2409
92-2409