DIGITAL MULTIMETER

KEW MATE 2000A / 2001A / 2012RA
TRUERMS

Unique Open Jaw TechnologyMAX

Φ12 120A

Φ6Robust cable protection

TRUERMSMAXMAX60AMAX

Φ12 120A

TRUE

Φ10 100ARMSMAX

Φ12 120A

Φ6MAX60A
Φ6MAX60A
KEW MATE

2000AMAXΦ10 100A

KEW MATE 2012RA

 Cable strength increased with robustcable protection
 Test probe can be fixed to the holster
 Can measure AC/DC current and voltageMAXΦ10 100A

KEW MATE 2001A
 Pocket size and heavy duty design
 Test lead cap to protect from short circuitaccident
 The open jaws are thin, perfect to clamp
wires even in tight spaces KEW MATE 2000A / 2001A / 2012RA Specification
2000A

2001A

2012RA

340.0mV/3.400/34.00/340.0/600V (Input impedance: 10MΩ)±1.5%rdg±4dgt
3.400/34.00/340.0/600V (Input impedance: 10MΩ)±1.5%rdg±5dg (50 to 400Hz)
60.0A
100.0A
±2.0%rdg±5dgt±2.0%rdg±5dgt

DC VAC V
DC A

600.0mV/6.000/60.00/600.0V (Input impedance: Approx. 10MΩ)±1.0%rdg±3dgt
6.000/60.00/600.0V (Input impedance: Approx. 10MΩ)±1.5%rdg±5dgt (45 to 400Hz)
60.00/120.0A±2.0%rdg±8dgt (60A)±2.0%rdg±5dgt (120A)
60.00/120.0A±2.0%rdg±5dgt (45 to 65Hz)
600.0Ω/6.000/60.00/600.0kΩ/6.000/60.00MΩ
±1.0%rdg±5dgt (600Ω/6/60/600kΩ)±2.0%rdg±5dgt (6MΩ)±3.0%rdg±5dgt (60MΩ)

60.0A
100.0A
±2.0%rdg±5dg (50/60Hz)±2.0%rdg±5dg (50/60Hz)
340.0Ω/3.400/34.00/340.0kΩ/3.400/34.00MΩ
±1.0%rdg±3dg (340Ω/3.4/34/340kΩ)±5.0%rdg±5dg (3.4MΩ)±15.0%rdg±5dg (34MΩ)AC AΩContinuity buzzer

Buzzer sounds below 30±10Ω (Continuity buzzer works on 340Ω range only)

Diode test—

—Capacitance——

Buzzer sounds below 35±25Ω (Continuity buzzer works on 600Ω range only)
2.000V ±3.0%rdg±5dgt, Open circuit voltage: Approx. 2.7V

Input sensitivity

Current: more than 15A
Voltage: more than 30V

Current: more than 25A
Voltage: more than 30V

400.0nF/4.000/40.00μF±2.5%rdg±10dgt(AC A) 99.99/400.0Hz±0.2%rdg±2dgt (100Hz)±0.1%rdg±1dgt (400Hz)(AC V) 99.99/999.9Hz/9.999/99.99/300.0kHz±0.2%rdg±2dgt (100Hz)±0.1%rdg±1dgt (1000Hz/10/100/300kHz)
Current: more than 6A
Voltage: more than 6V [-10kHz]/more than 20V [10 to 300kHz])

Conductor size

ø6 mm max.

ø10 mm max.

ø12 mm max.

Applicable standards

IEC 61010-1 CATⅢ 300V / CATⅡ 600V Pollution degree 2, IEC 61010-031, IEC 61010-2-032, IEC 61326-1, EN 50581(RoHS)

Frequency

Power source(AC A) 3.400/10.00kHz±0.1%rdg±1dgt(AC V) 3.400/34.00/300.0kHz±0.1%rdg±1dgt

R03(AAA)(1.5V) × 2* Continuous measuring time: Approx. 45hours(Auto power save: Approx. 10 minutes)

R03(AAA)(1.5V) × 2* Continuous measuring time:
DC V: Approx. 150 hours, AC A: Approx. 25 hours(Auto power save: Approx. 15 minutes)

Dimension

128(L) × 87(W) × 24(D) mm

128(L) × 92(W) × 27(D) mm

WeightApprox. 210g (including batteries)

Approx. 220g (including batteries)

Accessories

R03(AAA) × 2, Instruction manual

Optional accessories

9107 (Soft case) Application examples
Test Probe can be fixed to the holster

DC measurement in solar panels

Measuring current in a Switchboard

Servicing a car Selection GuideModel2000A

2001A

2012RA

Detection method

Average value

Average value

True RMS value

Conductor size

ø6 mm

ø10 mm

ø12 mmAC/DC A

60A

100A

120A

Diode test--üCapacitance--üPlease read the "Safety Warnings" in the instruction manual supplied with the instrument thoroughly and completelycorrect use. Failure to follow the safety rules can cause fire, trouble, electrical shock, etc. Therefore, make sure
Safety Warnings : forto operate the instrument on a correct power supply and voltage rating marked on each instrument.

For inquiries or orders :2-5-20, Nakane, Meguro-ku, Tokyo, 152-0031 Japan
Phone:+81-3-3723-0131
Fax:+81-3-3723-0152

This leaflet was produced in January 2024. The contents of this leaflet are subject to change without notice.

LC_2000A_2001A_2012RA_2_0_Jan24_AD_en