Compression Load Cell

TC-AR(T) n n KN- G6/8

Accuracy
Body
Material

Linearity 0.1%
Stainless Steel

Compression Load Cell
Benefit

Mounting
MethodEasy to install on the existing
facilities/systems.

Four M5 or M8 screws to mount.
(Screw size varies by models.)

DurableRobot Cable
standardized

Plug & Playwithbuilt-in TEDS

Enhanced durability against bendingthat occurs in moving parts withfrequent repetitive motion, such asindustrial robots and machine tools.
High stability and reliability are realized.

With the TD series indicators,
equivalent input calibration,
likely to forget in manual setting,can be performed automaticallyand help prevetion.

(See the reverse page for detail on TEDS)

Specifications
TypeCopression Load Cell

Model

TC-AR(T) n n KN - G6/8

(Embeded in the body)(10 substances)

Line up

TC-AR(T)20KN-G6

TC-AR(T)30KN-G6

TC-AR(T)50KN-G8

TC-AR(T)100KN-G8

TC-AR(T)200KN-G8

Rated Capacity (R.C.)

20kN

30kN

50kN

100kN

200kN

Natural Frequency

23kHz

23kHz

7.7kHz11kHz

50 kHzWeight (Approx.)

0.8kg

0.8kg

1.8kg

1.8kg

3.1kg

Safe overload rating150 ％ R.C.

Rated Output (R.O.)

2mV/V ±1％

Linearity

0.1% R.O.

Hysterisis

0.15％ R.O.

Repeatability

0.1％ R.O.

Safe Excitation Voltage

15V

Input Terminal Resistance

425 ±50Ω

Output Terminal Resistance

350 ±5ΩInsulation Resistance

1000MΩ or more (50V DC)

Compensated Temperature Range–10℃ to 70℃

Permissible Temperature Range–30℃ to 80℃

Temperature Effect on ZeroBalance

0.1% R.O. / 10℃

Temperature Effect on Output
Cable

0.1％ R.C. / 10℃
Φ6, 6-core shielded,
5m direct connection robot cable with bare lead wires

Φ8, 6-core shielded,
5m direct connection robot cable with bare lead wires

Mounting Method

Screw holes

Body Material

Stainless SteelDimensional drawings (Units: mm)

Head Olate, Base Plate
Model
TC-AR (T)- G6 20kN
TC-AR (T)- G6 30kN

Head Plate(Weight)
TF-UR102F

TF-LR060F(approx. 0.13kg)(approx. 0.6kg)

TF-UR050F(approx. 2.9kg)

TC-AR (T)- G8 50kN
TC-AR (T)- G8 100kN

Base Plate(Weight)

TF-LR101F(approx. 1.53kg)

TF-LR121F

TC-AR (T)- G8 200kN(approx. 5.8kg)Bold w/Hex holeAØBØC

ØDØEØF

ØG

ØH

ØJ

KLM

ØN

12 x M555

53

38

6.698

80

6.660

4215251560

12 x M555

53

38

6.698

80

6.660

4215251560

25 x M5

98

118

100

11148

124

9100

80

30

40

28

100

25 x M598

118

100

11148

124

9100

80

3040

28100

30 x M5

113

118

100

11168

144

14120

90

40

45

28120

Related Products (Indicators and Signal Conditioners)

NewNew

EtherNet/IP™
model

CC-Linkmodel

92 x 92mm
Panel opening size

Color Graphics Digital Indicator

Digital Indicator

RS-485 model
EtherNet/IP™ modelCC-Link modelStandard model
CC-Link model
RS-485 model

TD-9000T

Attaches tocommon DIN rails

92 x 45 mm
Panel opening size

TD-700T

Weighs only 320g(incl. batteries)

Portable Digital Indicator

Signal Conditioner

TD- 01 Portable

TD-SC1
D/A model
RS-485 modelHigh performance modelwith large LCD

Excellent model with compact
and high functionality

Slim and light-weightsignal conditioner

On-site checking toolwith versatility

Supporting two inputs, force sensor
and displacement sensor, various
comparison judgments function,

Supporting five key functions in one unit,
numeric display, graph display, TEDS function,
static strain display, and signal conditioner.
This small and cost-effective TD-700T
achieves equal or even higher performance toupper-class models, with high-visibility color
LCD and various hold functions.

Supporting high-speed samplingof 20,000 times/second, PC-based
configuration via USB connection,
selectable network, and TEDS
calibration function.

Supporting various functions
that equal to embeded systems,
in hand-held size, allowing you
to take measurements anytime
anywhere, according to yourpurpose.

and direct saving of waveform
data onto large capacity internalmemory.

*** Under planning

EtherNet/IP is a trademark of ODVA, Inc. Other company names, product names and logos in this document are the trademarks or registered trademarks of their respective holders.

TEAC CORPORATIONTEAC America, Inc.,

1-47 Ochiai, Tama-shi, Tokyo
206-8530, JapanTEAC EUROPE GmbH.

E-mail: cs_ipd@teac.jp
Web: https://loadcell.jp/en/

https://loadcell.jp/en/products/loadcell/tc-ar-g6.html
https://loadcell.jp/en/products/loadcell/tc-ar-g8.html

E-mail: datarecorder@teac.com
E-mail: info@teac.eu

TEAC SALES & TRADING (ShenZhen) CO., LTD.
E-mail: teacservice3@teac.com.cn© Copyright TEAC CORPORATION 2022 2204TCJ-PDF / ISD-194