Sensor Systems

Non-contact Line-laser Sensor

SurfaceMeasure-S Series

Catalog No. E13023(3)SurfaceMeasure-S Series

Line-laser sensor providing stablemeasurements with simple operations

Product structureThis line-laser sensor combines the opticaltechnology and sensor technology ofMitutoyo.

Laser diode
Collimating lens

Image sensor

High-accuracy
cylindrical lensCamera lens

CLASS 2 blue laser

Three product features(1) High-accuracy and environment-resistant sensor
The sensor itself is guaranteed for accuracy and has a z repeatability of 0.5 µm or less. Ithas also achieved the IP67 protection level, providing stable measurements.

(2) Supporting automated measurement
System configuration (example)
This is an example of thesystem configuration of
the Sur faceMeasure-SSeries. Various other system configurations arealso available.

PC(Can also beconnected to PLC*)

Encoder signal

SurfaceMeasure-S Series

Master cable

1

The SurfaceMeasure-S Series can obtain the profiles and three-dimensional shapes ofmeasurement workpieces at high speed (a maximum frame rate of 10 kHz) and make anautomatic judgement inside the sensor.
It is also equipped with a parts matching function that allows the measurement tool to beapplied throughout, regardless of the orientations of the parts being measured. Measurements can be taken without performing alignment.

Robot control(You can also configure the systemusing a stage, conveyor, or otherdrive mechanism)

MasterEthernet

Access the user interface via your Web
browser to set up measurement
conditions and judgement conditions.
* PLC protocol: Modbus TCP, EtherNet/IP, PROFINET(3) Simple operability
The software supports intuitive operation and is built into the sensor (software installationis not required), so you can use it immediately after mounting.

2APPLICATIONS
Automobile industry

Rubber and tire industries

Panel gap inspection

Tire shape inspection

Electrical and electronic industries

Automobile industry

Dimensional measurement on precision resin molded components

Condition inspection after welding

SOFTWARE
Electric vehicle battery industry

Electrical and electronic industries

Cell assembly inspection

Adhesive inspection

Powerful interface with excellent operability and functionality

OKNG- Excellent operability simply by using a mouse- Simple and intuitive interface- Web browser-based, no need to install software
Electrical and electronic industries

Food industry

Connector pin inspection

Standard inspection of food, etc.

Easy-to-configure measuring system
Measurement tool

Pattern matching

Emulator

3- Various built-in measurement tools- 2D and 3D data can be obtainedCountersink HoleA variety of measurement tools are available.

OCR

Gap & Flush

GrooveSize

Using the emulator makes it possible to consider measurement conditions or make an analysis with obtained
data even when offline.

4FEATURESOPTIONAL ACCESSORIES

Automatic pass/fail judgment is possible

Sensor networking hub Master

The equipment can obtain the profiles and three-dimensional shapesof measurement workpieces at high speed (a maximum frame rate of10 kHz) and make an automatic judgement inside the sensor.

SurfaceMeasure1008S
The sensor itself is guaranteed for an accuracyof 20 µm (1 σ) and a Z axis repeatability of 0.5µm. It has also achieved the IP67 protection level,
providing stable measurements.

Measurement is possible without alignment

Device used for distributing power to sensors and synchronizing the sensors in the multi-sensor system
Input: Power supply 24 to 48 V, Laser Enable input, Encoder input, External input
Master810

Master cable
SurfaceMeasure0303S

Simple operation

This is a non-contact line-laser sensor for inline
measurement with high resolution (13 to 17 µmin X direction) and high measuring accuracy (1 σ= 9 µm).
As with the SM1008S, the sensor itself isguaranteed for traceability and measurement
accuracy.

The software supports intuitive operation and is built into the sensor(software installation is not required), so you can use it immediately
after mounting.

Cable for the connection between sensor and MasterM16 connector
Sensor side (Straight or 90°upwardpointing can be selected)
Specifications Power supply
RJ45 (Connect to Master)
sideCommunication
RJ45 (Ethernet connection)
sideSpecifications

Max. 10 kHz
CLASS 2(IEC 60825-1:2014, EN 60825-1:2014+A11:2021)
CLASS 2 (JIS C 6802: 2014)
Semiconductor laser
405 nm (visible)
2.2 mW
650 g
0 ℃ to 40 ℃
RH 20 to 80%, non-condensing-30 ℃ to 70 ℃
RH 20 to 95%, non-condensing
IP 67 *2
24 to 48 VDC (15 W)

EN / IECJIS
Laser medium
WavelengthMax. output4818.51221.2°

Power and Ethernet cable
Cable to supply and control power without using the Master for the sensor
M16 connector
Sensor side (Straight or 90°upwardpointing can be selected)
Specifications Power supply
Flying leadsideCommunication
RJ45 (Ethernet connection)
side74

Mounting size

1213 20
80

540

18.5

74

Remarks
1 × Open wire end, 1 × RJ45 end
1 × Open wire end, 1 × RJ45 end
1 × Open wire end, 1 × RJ45 end
1 × Open wire end, 1 × RJ45 end
1 × Open wire end, 1 × RJ45 end
1 × Open wire end, 1 × RJ45 end1 × Open wire end, 1 × RJ45 end, 90°connector
1 × Open wire end, 1 × RJ45 end, 90°connector
1 × Open wire end, 1 × RJ45 end, 90°connector
1 × Open wire end, 1 × RJ45 end, 90°connector
1 × Open wire end, 1 × RJ45 end, 90°connector
1 × Open wire end, 1 × RJ45 end, 90°connector

198071

M16 connector(Straight or 90°upward-pointing can be selected)
Specifications I/O device side Flying lead
Encoder input (2-phase square-wave signals Phase A, Phase B,Phase Z), Trigger input, Digital output, RS-485 output
Sensor side

13 20

2380
1108151246
16 15 2×Ø4.4 456.5
3480

56.5
34

4×M3depth3.5
30 20

2328.75
32.5

Product Name
2 m Power and Ethernet
5 m Power and Ethernet
10 m Power and Ethernet
15 m Power and Ethernet
20 m Power and Ethernet
25 m Power and Ethernet
2 m Power and Ethernet 90deg
5 m Power and Ethernet 90deg
10 m Power and Ethernet 90deg
15 m Power and Ethernet 90deg
20 m Power and Ethernet 90deg
25 m Power and Ethernet 90deg

Cable to connect the external I/O device to the sensor

3×M4depth4.7

13 20
19571

12.5

Mounting size
80
11030 °
81
12Code No.
02AQL367
02AQL368
02AQL369
02AQL37002AQL371
02AQL372
02AQL385
02AQL38602AQL387
02AQL388
02AQL389
02AQL390

I/O cable

4040

60

1246
16 15 2×Ø4.4 4
18.5
25
12.5

30 20

548

19
21.2°

100

Measuring range
3×M4深さ4.7

4×M3深さ3.580
1108171

30 20

74

1246
16 15 2×Ø4.4 480

4×M3depth3.5

21.2°

100

SurfaceMeasure0303SRemarks
2 × RJ45 ends
2 × RJ45 ends
2 × RJ45 ends
2 × RJ45 ends
2 × RJ45 ends
2 × RJ45 ends
2 × RJ45 ends, 90°connector
2 × RJ45 ends, 90°connector
2 × RJ45 ends, 90°connector
2 × RJ45 ends, 90°connector
2 × RJ45 ends, 90°connector
2 × RJ45 ends, 90°connector

Code No.
02AQL420
02AQL421
02AQL422

Product Name
GoMax NX Power plug type: B
GoMax NX Power plug type: F
GoMax NX Power plug type: I

Code No.
02AQL361
02AQL362
02AQL363
02AQL364
02AQL365
02AQL366
02AQL379
02AQL380
02AQL381
02AQL382
02AQL383
02AQL384

Product Name
2 m I/O
5 m I/O
10 m I/O
15 m I/O
20 m I/O
25 m I/O
2 m I/O 90deg
5 m I/O 90deg
10 m I/O 90deg
15 m I/O 90deg
20 m I/O 90deg
25 m I/O 90deg

Remarks

56.534408048

12Product Name
2 m Power and Ethernet Master5 m Power and Ethernet Master10 m Power and Ethernet Master15 m Power and Ethernet Master20 m Power and Ethernet Master25 m Power and Ethernet Master2 m Power and Ethernet Master 90deg
5 m Power and Ethernet Master 90deg
10 m Power and Ethernet Master 90deg
15 m Power and Ethernet Master 90deg
20 m Power and Ethernet Master 90deg
25 m Power and Ethernet Master 90deg

NVIDIA module Jetson Xavier NXCPU6-core NVIDIA Carmel ARM v8.2
GPU
Volta GPU, 384 CUDA core, 48 Tensor core
MemoryOnboard 8 GB LPDDR4Storage
Onboard 16 GB eMMc
Specifications
I/O2-port Ethernet
Power supply
12 to 24 V, Max. 15 W
Weight2.1 kgOperating temperature -15 to 55℃
Mounting
DIN rail, wall mounting

Unit: mm3×M4depth4.7
23

Measuring range

18.5

Code No.
02AQL373
02AQL374
02AQL375
02AQL376
02AQL377
02AQL378
02AQL391
02AQL392
02AQL393
02AQL394
02AQL395
02AQL396

Arithmetic unit to accelerate measurement processing without a PC

100

External dimensionsSurfaceMeasure1008S

Remarks
Accepting a maximum of 8 sensors
Accepting a maximum of 24 sensors

GoMax NX

40MassTemperature
Operatingenvironment Humidity
Temperature
Storageenvironment Humidity
IP codePower supply (power consumption)*1 Accuracy inspection environment:
Temperature 20 ℃ ± 1 ℃ , Humidity
50%RH ± 10%RH
Measurement workpieces: Specified
reference ball for inspection ( ф25 mm)
Inspection method: Determined by
Mitutoyo-specified inspection method.
The operating environment and the storage
environment are different from theguaranteed accuracy environment.
*2 Measuring accuracy may deteriorate if any
water droplet or dust particle adheres to theoptical path.

553-110
SurfaceMeasure0303S
32.5 mm
25 mm
60 mm
9 µm (Sphere measurement)
0.4 µm80

Line laser553-100
SurfaceMeasure1008S
100 mm
80 mm
80 mm
20 µm (Sphere measurement)
0.5 µm40

Laser class

Product Name
Master810
Master2410

Master2410

The system is equipped with a parts matching function that allows
the measurement tool to be applied throughout, regardless of theorientations of the parts being measured. Measurements can be taken
without performing alignment.

Code No.
Product NameMaximum measuring width
Measuring range
Working distance
Scanning error (1 σ ) *1
Z axis repeatability
Frame rate

Code No.
02AQL401
02AQL402

Note: Sensors cannot be synchronized by the signals input to andoutput from each sensor through this cable.

Remarks
Open wire end
Open wire end
Open wire end
Open wire end
Open wire end
Open wire end
Open wire end, 90°connector
Open wire end, 90°connector
Open wire end, 90°connector
Open wire end, 90°connector
Open wire end, 90°connector
Open wire end, 90°connector

62408(JP3), Printed in Japan

Coordinate Measuring Machines

Vision Measuring Systems

Form Measurement

Optical Measuring

Whatever your challenges are, Mitutoyo
supports you from start to finish.
Mitutoyo is not only a manufacturer of top quality
measuring products but one that also offers qualified
support for the lifetime of the equipment, backed up
by comprehensive services that ensure your staff canmake the very best use of the investment.
Sensor Systems

Test Equipment

Digital Scale and DRO Systems

Small Tool Instruments
and Data Management

Apart from the basics of calibration and repair,
Mitutoyo offers product and metrology training,as well as IT support for the sophisticated software
used in modern measuring technology. We can also
design, build, test and deliver measuring solutions
and even, if deemed cost-effective, take your criticalmeasurement challenges in-house on a sub-contract
basis.

Find additional product literatureand our product catalog
https://www.mitutoyo.co.jp/global.html
Mitutoyo Corporation
Notes on Export Regulations:
Do not commit an act, which could directly or indirectly, violate any law or regulation of Japan,
your country or any other international treaty, relating to the export or re-export of any commodities.

20-1, Sakado 1-Chome,Takatsu-ku, Kawasaki-shi,
Kanagawa 213-8533, Japan

Note: Product illustrations are without obligation. Product descriptions, in particular any and all technical specifications,are only binding when explicitly agreed upon.
MITUTOYO and MiCAT are either registered trademarks or trademarks of Mitutoyo Corp. in Japan and/or other countries/regions.
Other product, company and brand names mentioned herein are for identification purposes only and may be thetrademarks of their respective holders.
All product information contained in this brochure is current as of August 2024.

T +81 (0) 44 813-8230
F +81 (0) 44 813-8231
https://www.mitutoyo.co.jp