Small Tool Instruments andData Management

ABS Digimatic Height Gage

Series 570

CatalogNo.
No.E12050
00000
CatalogAbsolute
DigimaticHeight Gage

Series 570

A new standard height gage providing reliable andsmart measurement

Main functions and features
Helps transform manufacturing sites into
smart factories

This is an update of the height gage standard type that has gained a strong reputation in the manufacturing field. We
have enlarged the DATA button and redesigned its button layout to improve operability. In addition, we have made iteasier to set up the "U-WAVE" wireless measurement data communication system, making the system even easier to use
in smart factories and other places where measurement data output is essential. Of course, we are committed to domestic
production in Japan and are continuing to pursue reliable Mitutoyo quality.

The design of the case allows for easy installation of U-WAVE (optional), which enables wirelesstransmission of measurement data, promoting smarter use of data in manufacturing sites.

The case features accents in Mitutoyo's standard colors.
This gives a harmonized design in spaces where multiple Mitutoyo products are used.

Supports smooth and reliable measurement operations
Measuring range
0~600 mm
No. 570-404

The handle for moving the slider enables smooth vertical movement. In addition, the highly rigid
pillars and large clamp levers provide smooth and reliable workability.

Easy to operate for everyone
We have enlarged the LCD to improve the visibility of numbers and other displays. In addition, we
have reduced the number of operation switches to a simple three-button system and have enlarged
the DATA/HOLD button to make it easier to press, ensuring more reliable operability.

Measuring range
0~300 mm
No. 570-402

Supports high quality
manufacturing throughefficient measurement

By placing the body and Dial Test Indicator on aflat surface plate, even complex shapes can bemeasured accurately and easily. In addition, thesmooth movement across the surface plate provides
excellent workability.

The holder arm and clamp are optional .

23

Special features
Supports smart data collection and management

Links with U-WAVE*
We have made it even easier to set up the"U-WAVE" wireless measurement data
communication system, which makes
measurement data entry easier and morereliable.
The reviewed design makes installation
faster and easier.

570-404

570-402*T he U-WAVE needs to be fixed in place using adouble-sided fastener/tape.

Ensures high visibility and improves work efficiency

Smooth operation and settings

Large LCD (liquid crystal display)Three-button operation

The 12 mm high character display—20% bigger than theprevious model—makes it easier to read the measured values.

The 3-button operation makes it consistent with other
Mitutoyo devices. And when used together with U-WAVE, it
makes data management operations easy.
The Absolute origin setting function
memorizes a surface plate, etc. as theabsolute origin.

10 mm

The Data button activates data output
when the data output option is connected.

New model

Previous model
Actual size

The INC (incremental) measurement
function allows the setting of the origin atany position.

12 mm

Actual size

The LCD screen has been enlarged and the typeface changed to bold,enhancing visibility over previous models.

The Hold function retains displayed values.
Other functions
The Measurement data output terminal can be used to build
statistical process management systems and measurement systems.

Order No.
570-402570-404
570-412570-413570-414
Unit (Display unit)
MetricInch/MetricMaximum permissible±0.03 mm±0.05 mm±0.0015 in/±0.03 mm±0.0020 in/±0.05 mm
error EMPE
0 to 12 in/0 to 300 mm 0 to 18 in/0 to 450 mm 0 to 24 in/0 to 600 mm
Measuring range
0 to 300 mm
0 to 600 mm
Resolution
0.01 mm
0.0005 in/0.01 mm
Functions
ON/OFF, ORIGIN, ZERO/ABS, HOLD/DATA, inch/mm conversion*
Drive method
Slider-feed wheel (manual)
Data output
DIGIMATIC d1
Max. response speedUnlimited
LCD character height
12 mmBattery lifeApprox. 20,000 hours with continuous use, 5 years under normal use*On Inch/Metric models only
Unit: mm
Order No.

570-40212570-404
35

1235

Contributes to reliable, accurate and repeatable measurements

Hardware design with emphasis on operability

827
97411598041155480

32

9116

116

Dimensions

97

The large, easy-to-grip lever
provides secure clamping.

516

Large clamp lever48

The slider-feed wheel handle
enables smooth verticalmovement of the slider.

64

32

Handle for moving theslider641004149120180

5

Option Lineup
Optional accessories

Optional accessories for data output

Contact Sensor

Mini-Printer Equipped with Data Logging Function
No. 900872

A ttached to both the workpiece*1 and height gage*2 when measuring heights using a height gagewith a scriber, the contact sensor is a convenient detector that lights up when the scriber touches
the workpiece.

Digimatic Mini-Processor DP-1VA LOGGER
DP-1VA LOGGER
No. 264-505*1 Conductive workpieces only. *2 Attach to a conductive part.

 agnet is incorporated.
MUse with a granite surface plate (precision stone surface plate). (For details, refer to our website.)Power supply: PR44 (air battery) × 2	The batteries for monitoring are not included with the product and must be purchased separately.

CablesDepth Gage Attachment

No. 06AFZ050
USB cable (Type A - micro B)
No. 905338
Connection cable (1 m) Type F
No. 905409
Connection cable (2 m) Type F

No. 900764: With metric-type holding bar (9x9 mm cross-section)
No. 900878: With inch-type holding bar (0.25x0.5 in cross-section)

 ttaches to a height gage for measuring groove and hole depth.
A
Minimum measurable hole diameter: 5.5 mm
Maximum distance from the bottom of the holding bar to the contact point: 80 mm (metric type),
2.95 in (inch type)D ial indicator contact points are usable. (For details, refer to our website.)
Cross-section of mounting bar: 9×9 mm
Holding bar length: 100 mm

Center Probe
No. 951144: With metric-type holding bar (9x9 mm cross-section)
No. 900581: With inch-type holding bar (0.25x0.5 in cross-section)

Swivel clamp
Holding bar*(with dovetail groove)* (9×9×50)

Measurement data collection software

USB-ITPAK V2.1

USB-ITPAK V2.1 creates a procedure to input data from gages
equipped with Digimatic output to Excel sheets via IT-016U or USB-ITNor U-WAVE. Using together with USB-ITPAK V2.1 will further improve
the operational efficiency of repetition inspection work. Best suited forkeeping track of inspection data of mass-produced products.
Automatically calls Excel sheet.
Automates cursor movement after data entry, eliminating the need for line
feed and other cell movement operations.
Input range can be specified per Digimatic gage, which reduces improper
input.
The last data input can be canceled by a single operation (foot switch,function key etc.)
Data input or cancellation can be performed at once in multiple-point
simultaneous measurement.
<TIP> Refer to the USB-ITPAK V2.1 user's manual on the Mitutoyo website.
https://www.mitutoyo.co.jp/eng/6

 sed to import data from Height Gage withU
Digimatic output function to a computer using
wireless communication.

U-WAVE-RE asy installation without cables that may get in
the way thanks to wireless communication (up to20 m).

USB-ITPAK V2.1

U-WAVE-T

No. 953638 (metric)
No. 953639 (inch)*For mounting test indicators, etc.

USB-ITPAK V2.1 further streamlines inspection work. Ideal for recording inspection data ofmass-produced products that are repeatedly produced using the same procedure every day!!

No. 02AZD880G

U-WAVE-TCB
No. 264-625

U-WAVE-T connection cable
No. 02AZD790F
U-WAVE-T dedicated connection cable Type F

USB dongle
A USB dongle must beconnected to the PC
running the software.

Digimatic Gage/PC Data Input Device

USB Input ToolInterface used to export calculation results tospreadsheet software on a computer via a USB
cable. Calculation results (values) can be exported
in one operation.

Operating environment
Order No.
Compatible OS*
Supported Excelversions
Hard disk
CD-ROM drive
USB port
Monitor resolution*T he connection cable to the Height Gage is notsupplied. A connection cable sold separately isrequired.

U-WAVEWith USB 2.0 cable (1 m)

No. 902053 (metric)
No. 900322 (inch)*For mounting test indicators, etc.

T he data logger function allows storage of up
to 1,000 pieces of data in memory and batch
transfer stored data to an Excel-format inspection
certificate, etc., by connecting to a PC with a USB
cable (optional).

Wireless Measurement Data Communication System

No. 02AZD810D

 llows quick measurement of center-toA
center distance between holes.
Measurable hole diameters: 1 to 38 mm
Cross-section of mounting bar: 9×9 mm

T he versatile DP-1VA LOGGER printer not only
prints measurement data, but performs a variety
of statistical analyses, draws histograms andD-charts and also performs complex operations on
Xbar-R control charts.

06AFM386
Windows 2000 SP4, Windows XP SP2 or later, Windows
Vista, Windows 7, Windows 8, Windows 8.1, Windows 10
Excel 2002, Excel 2003, Excel 2007,Excel 2010, Excel 2013, Excel 2016
Free space of more than 20 MB
For program installation
2 ports or more (for USB dongle and USB-ITN)
800×600, 256 colors or more

32-bit, 64-bit OS supported
The OS language must be the same as that of USB-ITPAK V2.1.

USB Direct Input Tool
No. 06AFM380F
USB-ITN-F

USB Keyboard Signal Converter TypeNo. 264-016-10
IT-016U

1m No. 905338
2m No. 905409

Note: O ne of the cables on the upper right (sold separately) is
required to connect the testing machine to the IT-016U.

7

101 2202(1)g-(TS)NE-(JP1), Printed in Japan

Coordinate Measuring Machines

Vision Measuring Systems

Form Measurement

Optical Measuring

Whatever your challenges are, Mitutoyo
supports you from start to finish.
Mitutoyo is not only a manufacturer of top quality
measuring products but one that also offers qualified
support for the lifetime of the equipment, backed up
by comprehensive services that ensure your staff canmake the very best use of the investment.
Sensor Systems

Test Equipment

Digital Scale and DRO SystemsSmall Tool Instruments
and Data Management

Apart from the basics of calibration and repair,
Mitutoyo offers product and metrology training,as well as IT support for the sophisticated software
used in modern measuring technology. We can also
design, build, test and deliver measuring solutions
and even, if deemed cost-effective, take your criticalmeasurement challenges in-house on a sub-contract
basis.

Find additional product literatureand our product catalogue
https://www.mitutoyo.co.jp/global.html
Mitutoyo Corporation
Our products are classified as regulated items under Japanese Foreign Exchange and Foreign Trade Law.
Please consult us in advance if you wish to export our products to any other country.
If the purchased product is exported, even though it is not a regulated item (Catch-All controls item), the customer service
available for that product may be affected. If you have any questions, please consult your local Mitutoyo sales office.
Note: Product illustrations are without obligation. Product descriptions, in particular any and all technicalspecifications, are only binding when explicitly agreed upon.
MITUTOYO and MiCAT are either registered trademarks or trademarks of Mitutoyo Corp. in Japan and/or other countries/regions.
Other product, company and brand names mentioned herein are for identification purposes only and may be thetrademarks of their respective holders.

20-1, Sakado 1-Chome,
Takatsu-ku, Kawasaki-shi,
Kanagawa 213-8533, Japan
T +81 (0) 44 813-8230
F +81 (0) 44 813-8231
https://www.mitutoyo.co.jp