DC MILLIAMP CLAMP LOGGER

KEW 2510

Measures DC4-20mA signals without breaking the loop
"Advanced model with recording function!"

Actual size

 Measures 4 to 20 mA instrumentationand processes signals without breakingthe loop

 Memory function stores up to 192,000

data useful for long-time signal monitoring and fault finding Transfer data to PC via Bluetooth®

 Top class measurement with 0.2% accuracy∅6mm clamp jaw easy to use in tight places LED light for illuminating the measure-ment spot Analog output terminal to allow the con-nection to a recorder or a digital multimeter● KEW 2510 Speciﬁcation
20/100mA(Auto-ranging)±0.2%rdg±5dgt(0.00 to 21.49mA)±1.0%rdg±5dgt(21.0 to 120.0mA)
Conductor size
φ6mm max.
Analog output
Recorder: DC1000mV against DC100mA
Communication Interface
BluetoothⓇ
Applicable standards
IEC 61010-1 Pollution degree 2
IEC 61010-2-032, IEC 61326-1(EMC)
IEC 60529 IP40, EN50581(RoHS)
Operating temperature and humidity-10 to +50℃, relative humidity 85% or less (no condensation)
range
When using AC adapter :0 to +40℃, relative humidity 85% or less (no condensation)
Storage temperature and humidity range -20 to +60℃, relative humidity 85% or less (no condensation)
Power source
R6/LR6(AA) (1.5V) x4 (Alkaline LR6 is recommended.)External supply (AC adapter 8320)
Battery life
Approx. 50 hours continuous with alkaline batteries(with Backlight, LED light and Bluetooth feature OFF)
Dimension
111(L)x61(W)x46(D)mm : Display unit
104(L)x34(W)x20(D)mm : Sensor
700mm : Sensor cable
WeightApprox. 310g(including batteries)
Accessories
8320(AC adapter), KEW Windows for 2510(Software)
9096(Carrying case), LR6(AA)x4, Instruction manual
Software installation manual
Optional accessories
7256(Output cord)

For signal measurement in process andbuilding instrumentation

DC mA

Memory function
Communication function

Fluorescent buttons foreasy operation in dark areas

Five selectable
recording intervals

Measured data can be transferred to a PC
Memory capacity up to192,000 data

● Accessories

Analyzing and processing themeasured data with a PC
Recording Max recording
intervalperiod
1 sec. 53 hours
5 sec. 11 days
10 sec. 22 days
30 sec. 66 days
60 sec. 133 days➡

External supplyLED light & Backlight Display

Bluetooth®

System requirements
OS:Windows®11/10
Display: XGA（Resolution 1024×768 dots）or more
Required HDD space: 1Gbyte or more
Other: With CD-ROM drive , .NET Framework (4.6.1 or later)● Optional accessory

8320
AC adapter

for DC Milliamp Clamp Logger

KEW 2510
Copyright
KYORITSU ELECTRICAL
INSTRUMENTS WORKS, LTD.
20157256
Output cord

KYORITSU ELECTRICAL INSTRUMENTS WORKS, LTD. Japan

www.kew-ltd.co.jp＊ Windows® is registered trademark ofMicrosoft in the United States.
＊ Bluetooth® is a registered trademark ofBluetooth SIG.

Analog output terminal for recorder connection

KEW Windows 9096
Carrying casefor 2510

● Selection Guide25002510

DC 20/100mAConductor size φ6mm max



Analog output

Digital Multimeter
DC10mV/DC1mA

Recorder or Logger
MODEL 7256
Output cord(Optinal accessory)
Note: The Auto power off can be disabled for long recordingMemoryData transmission (Bluetooth® )Power: BatteryPower: External supply

Please read the "Safety Warnings" in the instruction manual supplied with the instrument thoroughly and completelycorrect use. Failure to follow the safety rules can cause fire, trouble, electrical shock, etc. Therefore, make sure
Safety Warnings : forto operate the instrument on a correct power supply and voltage rating marked on each instrument.

For inquiries or orders :2-5-20, Nakane, Meguro-ku, Tokyo, 152-0031 Japan
Phone:+81-3-3723-0131
Fax:+81-3-3723-0152

This leaflet was produced in January 2024. The contents of this leaflet are subject to change without notice.

LC_2510_3_0_Jan24_AD_en