GasCorrosion
TesterErthernet

KG 200
FactK,Inc
www.factk.jp

From our experience, of the four electric parts tests (defect analysis, standard test,deterioration accelerate test and gas corrosion test), we believe that the Gas Corrosion
Test is the most essential one.
As for the standard tests, various methods exist as listed below. In fact, the gas
corrosion test has complicated factors for reproducibility in comparison with common
environmental tests. Besides, it is said that the procedure for the tester itself is quite
troublesome.
However, as a result of our long years of development of our Gas Corrosion Tester bycarrying out actual numerous tests, we are confident that our Gas Corrosion Tester has
remained low-cost, easy to use, cost effective, safe, and some othercharacteristics. And we can carried out most of the following
standard tests listed below.
Example ofCorrosion ofGold SO2 10ppm
40℃ 75% 240h

Main Gas Corrosion Tests of Electric Parts currently being carried out are:
1.JIS H8502
SO2 25,or1000ppm,40℃,90%RH
13.Battel Saboratory
2.JIS H8620
H2S 3,or10ppm,40℃,90%RH
Class 2 H2S 10ppb+CL2 10ppb+NO2 200ppb,30℃,70%RH
3.JEIDA25,40 H2S 3ppm,40℃,85%RH
Class 3 H2S 100ppb+CL2 20ppb+NO2 200ppb,30℃,75%RH
4.JEIDA32,41 SO2 10ppm,40℃,85%RH
Class 4 H2S 200ppb+CL2 50ppb+NO2 200ppb,50℃,75%RH
5.EIAJ-RC-5608 H2S 1ppm,40or25℃,75%RH
14.ISO 10062-A SO2 0.5ppm,25℃,75%RH
6.EIAJ-CP-5102 SO2 25ppm,40℃,90%RH
B H2S 0.1ppm,25℃,75%RH
7.IEC 68-2-42 SO2 25ppm,25℃,75%RH
C SO2 0.5ppm+H2S 0.1ppm,25℃,75%RH
8.IEC 68-2-43 H2S 10ppm,25℃,75%RH
D H2S 0.1ppm+SO2 0.2ppm+CL2 0.02ppm
9.DIN 40046-36 SO2 10ppm,25℃,75%RH
25℃,75%RH
10.DIN 40046-37 H2S 1ppm,25℃,75%RH
15.Bellcore
Low
concentration 4 kinds
11.Corporate Standard H2S 3ppm+SO2 10ppm,40℃,75%RH
12.Corporate Standard H2S 0.5ppm+SO2 1ppm+NO2 1ppm,35℃,75%RH

KG200 Formation Diagram
KG200 Gas Corrosion Tester is<<Option>>designed to be carried out low

Gas Analyzer
Exhaust Pump

concentration mixed gas corrosion test
and other tests simply and specifically,Constant temperature chamber

which enables to control theconcentration below 0.01ppm.

Temperature
sensorHumidity sensor

Test chamber

Gas
Absorption
Equipment

Gas
Treatment
Tank

Also, this tester is designed to have aMeasurements
Dehumidifier

Teflon coated double layered chamberControl Value
Electronic Mass-FlowController

in order to cope with nitrogen dioxide

Humidifier

Gas Dilution
Equipment

Purge Systemand chlorine stably and it has a function

A gas concentration automatic
control system

Air Inhale Filter

Standard Gas

significant cause of corrosion.

Nitrogen Gas

to control the humidity which is the6000ppm～12000ppm

Hydrogen sulfide / Nitrogen sulfuric acid / Nitrogen dioxide / Chlorine / Ammonia, and so on. / OzonThis tester is designed to be negative pressure in the test

chamber with gas leakage main cock shut down equipment(OP)!
SafetyEasy setting of the temperature from 25℃65% up to 55℃98%!
Temp/Hum （OP85℃90％）

Gas
Concentration
Standard
Conformance
Cost
EffectivenessControl ofTest Record

Easy to set up the concentration by just turning the dial!
Corresponds to IEC StandardE ISO10062
Optional programs can be added based on your purpose.
Record and monitor the test results on Web-site byconnecting LAN(OP).

Defective Analysis

AC100V Operation(Certain value control type)

Standard Test

Not necessary to havea drainage and water supply
equipment

KG200
Back side

Corrosion-Resistance
Test
Deterioration Acceleration
Test

Easy to move
The back of the tester isvery plain.

(If the tester is not equipped withcooling device)

Internal chamber

Saves electricity andstandard gas

It does'nt condense in complete
double layered chamber.

This Tester Can be Moved Even After Installation andit Can Be Operated by 100VAC for Home Use (Certain value control type)
KG120HT
Entrusted Gas Corrosion Guide
Various environmental tests exist to conduct duration test,

Cylinder Cabinet withGas Leakage
Detector.
危険なガスボンベ
は格納庫へ収納defect analysis of the functional parts, the contact mechanism
parts reliability of electronic parts; such as switches, connectors,etc. Of these tests,
Gas Corrosion Test for the environmental test is the most suitable

漏洩処理装置付
ボンベ格納庫試験装置本体one for the evaluation analysis of the electric parts.
This including International Standard (IEC) and individual Corporate
Standard, special four kinds of mixed gas test is currently being
carried out.
Since we have been mainly focusing on conducting the evaluation
analysis since our company was founded in 1977,
we can offer you the best service with our expertise and a know-how
we have for the Gas Corrosion Test.
We will be looking forward to serving you as we are able to offer
you quick and accurate work that you will be satisfied.

High Reliability Gas Corrosion Tester KG200ST

Bomb (Cylinder) Main Cock
Shut Down Equipment can be installed.

Test behavior and data
can be cllected through10BASE-T/Erthernet

WWW

Network can be connected so thatcollecting and controling data, suspending
the test, starting the operation andsuspending the operation for emergencies
can be done from remote area. Also, Bomb(Cylinder) main cock shut down can be controlled.

■ Specification

Name: Flow Gas Corrosion Tester Stainless and Alminum StructureModel: KG200ST (Possible to add up to four kinds mix gas)
Chamber Capacity : Approximately 200 Liter
Chamber Material : Heat-resistance Plyvinyl chloride Teflon seal treatment
Indicator: Digital Temperature and humidity indication,
Digital Indication Gas Flow, Concentration, Recorder
Temp Set-Up: 25 ~ 60℃Digital Set-Up (Cycle Test can be added as an
Optional test at 85℃)
Humidity: 60 ~ 95% RH Digital Set-Up (Cycle Test can be added as an
Optional test)
Gas Flow: 0 ~ 200CCM/max (Electronic mass flow controller with
Shut off function)
Gas Concentration : 10ppb ~ 500ppm (the type, concentration, indication andAutomatic concentration set-up functions are optional.)
Total Flow
: 300 liter/hChamber Illumination: Below 300 lux
Relative Speed: 0.2 ~ 0.3m/sec (adjustable)
Indication Accuracy : Temperature ±0.2℃, Humidity±3%RH
Deviation in: Temperature ±0.5℃, Humidity±5％RH (within 5cm Chamberfrom the inside wall of Chamber)
Recorder: Paperless 5.7inch TFT color liquid crystal
Communication: 10BASE-T Controlling and Monitoring of Erthernet Data is optional.
Electricity: AC100V/50 ~ 60Hz (Three-phase,200V)
Electricity Consumption : Maximum: 1300W , Normal: 800W (at AC100V)
Dimension: 1850H–950W–650D (Unit:cm)(This doesn't include bomb tank.)
Weight: 180Kg (without bomb tank)
Cooler: Contained a cooler for dehumidification (300W)
Contained a cooler for test chamber (700W)
Earthquake Sensor : 100 ~ 170 gal (1 gal/1cm/sec2) In case the earthquakefs horizontal vibration increases
at 0.3 ~ 0.7sec cycle, the operation will be stopped.
Fire: In case of fire, the controller of the tester will come to stop at the temperature
above 60 ~ 70℃
Others: Purge System is contained. Exhaustion Gas Treatment Equipment, Supply Warning , Drainage
Warning Apparatus, Earthquake Emergency Apparatus, Fire Emergency Bomb main cock shut- off Apparatus(OP), RemoteControl Emergency Suspension Apparatus(OP), Gas Leakage Treatment Apparatus with Bomb (Cylinder) Tank can be added
as optional.
■ As for further details on Options, Contuct us.
■ Please note that the specification of this tester is subject to change due to improve the performance.

Manufacturer: FactK Corporation
Address: 2-31-28 Minamikugahara ootaku
Tokyo(Zip Code: 146-0084)
Tel 03-3757-2881, Fax 03-3757-2861
Email Address: mailinfo@factk.jp
Homepage Address: http:www.factk.jp■Agency

Catalog No,10K0407