FreeStyle® Comfort®

Portable Oxygen Concentrator with autoSAT®, UltraSense® and autoDOSETM

User Manual (English)

Bruksanvisning (Norsk)

Portable Oxygen Concentrator with UltraSense® and autoDOSE™

Bærbar oksygenkonsentrator med UltraSense® og autoDOSE™

Concentrateur d’oxygène portable doté d’UltraSense® etd’autoDOSE™

Bärbar syrgaskoncentrator med UltraSense® och autoDOSE™

Manuel d’utilisation (Français)Bedienungsanleitung (Deutsch)

Tragbarer Sauerstoffkonzentrator mit UltraSense undautoDOSE™
®

Manual del usuario (Español)

Concentrador de oxígeno portátil con UltraSense® y autoDOSE™

Manuale dell’utente (Italiano)

Concentratore di ossigeno portatile con UltraSense® e
autoDOSE™

Manual do utilizador (Português)Concentrador de oxigénio portátil com UltraSense e autoDOSE™
®

Gebruikershandleiding (Nederlands)

Draagbare zuurstofconcentrator met UltraSense en autoDOSE™
®

Brugermanual (Dansk)

Bærbar iltkoncentrator med UltraSense og autoDOSE™
®

Bruksanvisning (Svenska)
Käyttöopas (Suomi)

Kannettava happirikastin UltraSense®- ja autoDOSE™-toiminnoillaΕγχειρίδιο χρήστη (Ελληνικά)Φορητός συμπυκνωτής οξυγόνου με UltraSense®
και autoDOSE™
Kullanıcı El Kitabı (Türkçe)
UltraSense® ve autoDOSE™ Özellikli Taşınabilir OksijenYoğunlaştırıcı

Uživatelská příručka (Čeština)Přenosný koncentrát kyslíku s UltraSense® a autoDOSE ™

Instrukcja obsługi (Polski)

Przenośny koncentrator tlenu z technologią UltraSense® i
autoDOSE ™

Felhasználói kézikönyv (Magyar)

Hordozható oxigénkoncentrátor UltraSense® és autoDOSE ™
segítségével

FreeStyle Comfort

User Controls & System Status Indicators
ISO 7000; Graphical symbols for use on
equipment—Index and synopsis

If the product unique device identifier(UDI) label has the CE#### symbolon it, the device complies with therequirements of Directive 93/42/EEC
concerning medical devices. The
CE#### symbol indicates notified
body number.

Read user’s manual before operation.
Reg. # 1641
Storage or operating temperature
range. Reg. # 0632
Storage humidity range.
Reg. # 2620

IEC 60417
Class II Equipment, Double
Insulated Reg. # 5172

Atmospheric pressure limitation.
Reg. # 2621
Name and address of manufacturer.
Reg. # 3082

Power switch. Reg. # 5009
ETL Certification

Date of manufacturer. Reg. # 2497

Safety agency for CAN/CSA C22.2
No. 601.1 M90 for medical electricalequipment

Catalog Number. Reg. # 2493
Serial Number. Reg. # 2498

Internal Symbols

This way up. Reg. # 0623

Keep away from flammablematerials, oil and grease.

Fragile, handle with care.
Reg. # 0621

Do not disassemble.

Keep away from rain, keep dry.
Reg. # 0626

Increase flow.

Stacking limit by number.
Reg. # 2403

Decrease flow.

ISO 7010: Graphical symbols—Safety colors andsafety signs—Registered safety signs

Menu button.

The instruction manual must be read.
Reg. # M002
Keep away from open flame, fire,
sparks. Open ignition source andsmoking prohibited. Reg. # P003

RTCA DO160 Section 21
Category M Compliant. FAA SFAR
106 requirement

Do not smoke near unit or while
operating unit. Reg. # P002

Class II Equipment, Double
Insulated

Type BF applied part (degree ofprotection against electric shock).
Reg. # 5333

FAA Approved – POC

Warning. Reg. # W001
Council Directive 93/42/EEC; concerning
medical devices
Authorized representative in theEuropean Community
2 - ENG

RTCA/DO160
Section 21
Category M
Compliant

PN MN232-C4 K | User Manual

21 CFR 801.15: Code of Federal Regulations Title
21
Federal law restricts this device tosale by or on the order of a physician.

FreeStyle Comfort
Council Directive 2012/19/EU: waste electrical andelectronic equipment (WEEE)
WEEE

IEC 60601-1: Medical electrical equipment
Part 1 General requirements for basic safety andessential performance
Drip Proof
This product may be covered by one or more
patents, US and international. Please visit our
website below for the listing of applicable patents.
Pat.: www.caireinc.com/corporate/patents/.

What is the Oxygen
Concentrator
The air we breathe contains approximately 21%
oxygen, 78% nitrogen, and 1% other gasses. In
the FreeStyle® Comfort®, room air is drawn into
the machine through the air intakes. It then passes
through an adsorbent material called molecularsieve. This material separates the oxygen fromthe nitrogen and allows only the oxygen to passthrough. The result is a flow of high-concentration
oxygen delivered to the user.

air we breathe, and our bodies depend on a steadysupply to function properly. Your physician
prescribed a flow or setting to address yourparticular respiratory condition.
Although oxygen is a non-addictive drug,
unauthorized oxygen therapy can be dangerous.
You must seek medical advice before you use thisoxygen concentrator. The Equipment Provider who
supplies your oxygen equipment will demonstrate
how to set the prescribed flow rate.

Operator Profile

The FreeStyle Comfort combines advanced oxygen
concentrator and oxygen conserving technologies
to create a portable device that allows for maximum
portability and ambulation. The advanced pulse
flow delivery quickly senses when the user breathes
in and delivers a pulse of oxygen at the beginningof each inhalation. Pulse flow delivery is ideal for
activities and time away from home. It conserves
oxygen and extends battery life as opposed
to continuous flow. The FreeStyle Comfort islightweight enough to be carried via shoulder strap.
It can be operated on AC (wall), DC (vehicle) orbattery power.

Concentrators are intended to supply supplemental
oxygen to users suffering from discomfort due toailments which effect the efficiency of one’s lungs
to transfer oxygen in the air to their bloodstream.
Portable oxygen concentrators (POCs) do not store
or contain oxygen. They do not need to be refilled,
and can recharge anywhere AC or DC power isavailable (except on an airplane). This makes theuser relatively self-sufficient in terms of in-homeuse, ambulation (both within and outside of thehome), mobility, and overall lifestyle. Oxygen
concentrator use requires a physician’s prescription
and is not intended for life support use.

Intended Use

Although oxygen therapy can be prescribed forusers of all ages, the typical oxygen therapy user
is older than 65 years of age and suffers from avariety of respiratory diseases, including Chronic
Obstructive Pulmonary Disease (COPD). Users
typically have good cognitive abilities and must
be able to communicate discomfort. If the user
is unable to communicate discomfort, or unable
to read and understand the concentrator labelingand instructions for use, then use is recommended
only under the supervision of one who can. If any
discomfort is felt while using the concentrator,
users are advised to contact their healthcare
provider. Users are also advised to have back-up
oxygen available (i.e. cylinder oxygen) in the eventof a power outage or concentrator failure. There areno other unique skills or user abilities required forconcentrator use.

The CAIRE FreeStyle Comfort Oxygen Concentrator is intended for the administration of supplemental oxygen. The device is not intended for life
support nor does it provide any patient monitoring
capabilities.

Why Your Physician
Prescribed Oxygen
Many people suffer from a variety of heart, lung,and other respiratory diseases. A significant numberof these people can benefit from supplemental
oxygen therapy at home, when traveling, or while
participating in daily activities away from home.
Oxygen is a gas that makes up 21% of the ambient

PN MN232-C4 K | User Manual

ENG - 3

FreeStyle Comfort

Unpacking Your FreeStyle Comfort
Verify that all of thecomponents listed are includedin the package. If any items are
missing, contact your oxygen
provider immediately.

Standard Accessories• Portable Oxygen Concentrator• Carrying Bag• Shoulder Strap• 8-Cell Battery Pack
• AC Power Supply• AC Power Cord
• DC Power Supply• User Manual

Optional Accessories• Additional 8-Cell Battery Pack• 16-Cell Battery Pack
• Desktop Charger• Carry-all Accessory Bag

Getting to Know Your FreeStyle Comfort
Become familiar with the key features of the FreeStyle Comfort and its user interface.

Front View
Strap Attachment

User Interface

OxygenOutletAir Outlet

Air Intake/Filters
Air Intake/
Filters
Power Port
Receptacle
Tab
Battery Pack
Battery Pack
Release Button
4 - ENGPN MN232-C4 K | User ManualFreeStyle Comfort
Top View
Display Screen
Increase Flow

Power Button

Menu Button

Decrease Flow

Display Screen On Battery
Flow Setting

Battery Display

autoSAT Indicator
Alarm Display

Battery
Percentage
Remaining

Bolus Delivery
Indicator

Display Screen On AC/DC Power
Flow Setting
Battery Display

Alarm Display
Space

Charge Indicator

PN MN232-C4 K | User Manual

ENG - 5

FreeStyle Comfort

Important!

Safety Instructions are defined as follows:
WARNING: IMPORTANT SAFETYINFORMATION FOR HAZARDS THATMIGHT CAUSE SERIOUS INJURY.
CAUTION: Important information forpreventing damage to the FreeStyle
Comfort.
Note: Information needing special attention.

Indications for Use
WARNING: IT IS VERY IMPORTANT
TO SELECT ONLY THE PRESCRIBEDLEVEL OF OXYGEN. DO NOT CHANGE
THE FLOW SELECTION UNLESS YOUHAVE BEEN DIRECTED TO DO SO BY A
LICENSED CLINICIAN.
WARNING: FEDERAL (USA) LAW RESTRICTS THIS
DEVICE TO SALE OR RENTAL BY ORDER OF APHYSICIAN OR OTHER LICENSED HEALTH CAREPROVIDER.
WARNING: THIS UNIT IS NOT TO BE USED FOR
LIFE SUPPORT. GERIATRIC, PEDIATRIC, OR
ANY OTHER USER UNABLE TO COMMUNICATEDISCOMFORT WHILE USING THIS DEVICE MAY
REQUIRE ADDITIONAL MONITORING. USERSWITH HEARING AND/OR SIGHT IMPAIRMENT(S)
MAY NEED ASSISTANCE WITH MONITORING
ALARMS.
WARNING: PULSE DOSE MODE SETTINGS MUSTBE DETERMINED FOR EACH USER INDIVIDUALLYFOR THEIR NEEDS AT REST, DURING EXERCISE,
AND WHEN TRAVELING.
WARNING: SOME RESPIRATORY EFFORTS
OF THE PATIENT MIGHT NOT TRIGGER THE
CONSERVING FUNCTION.
WARNING: THE FREESTYLE COMFORT IS NOTINTENDED FOR USE WITH A TRACHEOTOMISEDPATIENT.
WARNING: PREGNANT OR NURSINGWOMEN SHOULD NOT USE ACCESSORIESRECOMMENDED IN THIS MANUAL, THEY MAY
CONTAIN PHTHALATES.

6 - ENGPN MN232-C4 K | User ManualContraindications for Use
WARNING: IN CERTAIN
CIRCUMSTANCES, THE USE OF
NON-PRESCRIBED OXYGEN CAN BE
HAZARDOUS. THIS DEVICE SHOULD
ONLY BE USED WHEN PRESCRIBED BY
A PHYSICIAN.
WARNING: NOT FOR USE IN THE PRESENCE OF
FLAMMABLE ANESTHETICS.
WARNING: AS WITH ANY ELECTRICALLYPOWERED DEVICE, THE USER MAY EXPERIENCEPERIODS OF NON-OPERATION AS A RESULT OF
ELECTRICAL POWER INTERRUPTION, OR THE
NEED TO HAVE THE OXYGEN CONCENTRATORSERVICED BY A QUALIFIED TECHNICIAN. THE
OXYGEN CONCENTRATOR IS NOT APPROPRIATEFOR ANY USER WHO WOULD EXPERIENCEADVERSE HEALTH CONSEQUENCES AS THE
RESULT OF SUCH TEMPORARY INTERRUPTION.

Safety Guidelines
WARNING: NO MODIFICATION OF THISEQUIPMENT IS PERMITTEDWARNING: THIS DEVICE SUPPLIES HIGHCONCENTRATION OXYGEN THAT PROMOTESRAPID BURNING. DO NOT ALLOW SMOKING
OR OPEN FLAMES WITHIN THE SAME ROOMOF (1) THIS DEVICE, OR (2) ANY OXYGENCARRYING ACCESSORY. FAILURE TO OBSERVETHIS WARNING CAN RESULT IN SEVERE FIRE,
PROPERTY DAMAGE, AND/OR CAUSE PHYSICALINJURY OR DEATH.
WARNING: DO NOT OPERATE UNIT IN A
RESTRICTED OR CONFINED SPACE (I.E.,
A SMALL CASE OR HANDBAG) WHERE
VENTILATION CAN BE LIMITED. THIS CAN CAUSETHE OXYGEN CONCENTRATOR TO OVERHEATAND IMPAIR PERFORMANCE.
WARNING: THE CONCENTRATOR SHOULD BE
LOCATED AS TO AVOID SMOKE, POLLUTANTS
OR FUMES.
WARNING: IN THE EVENT THERE IS A SERIOUS
INCIDENT OCCURRING WITH THIS DEVICE,THE USER SHOULD IMMEDIATELY REPORTTHE INCIDENT TO THE PROVIDER AND/OR THE
MANUFACTURER. A SERIOUS INCIDENT IS
DEFINED AS AN INJURY, DEATH, OR POTENTIALTO CAUSE INJURY/DEATH SHOULD THERE BE
A REOCCURRENCE OF THE INCIDENT. THE
USER CAN ALSO REPORT THE INCIDENT TO
THE COMPETENT AUTHORITY IN THE COUNTRY
WHERE THE INCIDENT OCCURRED.

FreeStyle Comfort
WARNING: THE USE OF SOME OXYGENADMINISTRATING ACCESSORIESNOT SPECIFIED FOR THIS OXYGENCONCENTRATOR MAY IMPAIR ITSPERFORMANCE. RECOMMENDEDACCESSORIES ARE REFERENCED
WITHIN THIS MANUAL.
WARNING: IF THE OXYGEN CONCENTRATOR HAS
BEEN DROPPED, DAMAGED OR EXPOSED TO
WATER, PLEASE CONTACT YOUR EQUIPMENTPROVIDER FOR INSPECTION OR POSSIBLE
REPAIR OF THE DEVICE. DO NOT USE THE
OXYGEN CONCENTRATOR IF IT HAS A DAMAGEDPOWER CORD OR PLUG.
WARNING: PROPERLY SECURE, BELTOR OTHERWISE RESTRAIN THE OXYGENCONCENTRATOR WHEN IN A VEHICLE DURING
TRANSPORT TO PREVENT DAMAGE OR INJURY.
WARNING: THE US DEPARTMENT OF
TRANSPORTATION (DOT) AND UNITED NATIONS(UN) REGULATIONS REQUIRE THE REMOVALOF THE BATTERY FROM THE DEVICE FOR
ALL INTERNATIONAL AIRLINE TRAVEL WHEN
THE OXYGEN CONCENTRATOR IS CHECKED
AS LUGGAGE. WHEN SHIPPING THE OXYGENCONCENTRATOR, THE BATTERY MUST ALSO BE
REMOVED FROM THE DEVICE AND PACKAGED
PROPERLY.
WARNING: DO NOT OPERATE THISEQUIPMENT WITHOUT FIRST READING AND
UNDERSTANDING THIS MANUAL. IF YOU ARE
UNABLE TO UNDERSTAND THE WARNINGS AND
INSTRUCTIONS, CONTACT YOUR EQUIPMENTPROVIDER BEFORE ATTEMPTING TO USE THISEQUIPMENT; OTHERWISE, INJURY OR DAMAGE
MAY RESULT.
WARNING: IF YOU FEEL DISCOMFORT OR ARE
EXPERIENCING A MEDICAL EMERGENCY, SEEKMEDICAL ASSISTANCE IMMEDIATELY.
WARNING: OPERATING THE OXYGENCONCENTRATOR OUTSIDE OF THE OPERATIONALTEMPERATURE SPECIFICATIONS CAN LIMIT THE
CONCENTRATOR’S ABILITY TO MEET OXYGENCONCENTRATION SPECIFICATION. REFER TO THE
SPECIFICATION SECTION OF THIS MANUAL FORTEMPERATURE LIMITS.
WARNING: USE NO OIL, GREASE, OR
PETROLEUM-BASED OR OTHER FLAMMABLE
PRODUCTS WITH THE OXYGEN-CARRYINGACCESSORIES OR THE OXYGENCONCENTRATOR. ONLY WATER BASED,OXYGEN COMPATIBLE LOTIONS OR SALVESSHOULD BE USED. OXYGEN ACCELERATES THE
COMBUSTION OF FLAMMABLE SUBSTANCES.

WARNING: DO NOT ALLOW EITHER THE AIRINTAKE OR THE AIR OUTLET EXHAUST TO
BECOME BLOCKED. DO NOT DROP OR INSERTANY OBJECTS INTO ANY OPENINGS ON THE
DEVICE. THIS CAN CAUSE THE OXYGENCONCENTRATOR TO OVERHEAT AND IMPAIRPERFORMANCE.
WARNING: THE INCORRECT USE OF THE
BATTERY CAN CAUSE THE BATTERY TO GET HOT,IGNITE, AND MAY CAUSE SERIOUS INJURY. BE
SURE NOT TO PIERCE, STRIKE, STEP ON, DROP
THE BATTERY, OR OTHERWISE SUBJECT THE
BATTERY TO STRONG IMPACTS OR SHOCKS.
THE USE OF A DAMAGED BATTERY MAY CAUSEPERSONAL INJURY.
WARNING: DO NOT SHORT-CIRCUIT THE
BATTERY’S METAL CONTACTS WITH METALLICOBJECTS, SUCH AS KEYS OR COINS. THIS MAY
CAUSE SPARKS OR EXCESSIVE HEAT TO BE
GENERATED.
WARNING: WHILE USING THE PORTABLEOXYGEN CONCENTRATOR OUTDOORS WITH
THE AC POWER SUPPLY, CONNECT THE POWER
SUPPLY INTO A GROUND FAULT INTERRUPTED(GFI) OUTLET ONLY.
WARNING: THE OXYGEN CONCENTRATORSHOULD NOT BE USED ADJACENT TO OR
STACKED WITH OTHER EQUIPMENT. IF
ADJACENT OR STACKED USE IS UNAVOIDABLE,
THE DEVICE SHOULD BE OBSERVED TO VERIFYNORMAL OPERATION.
WARNING: USE OF CABLES AND ADAPTERS
OTHER THAN THOSE SPECIFIED, WITH THE
EXCEPTION OF CABLES AND ADAPTERS SOLDBY THE MANUFACTURER OF THE MEDICALELECTRICAL EQUIPMENT AS REPLACEMENTPARTS FOR INTERNAL COMPONENTS, MAY
RESULT IN INCREASED EMISSIONS OF
DECREASED IMMUNITY OF THE OXYGENCONCENTRATOR.
WARNING: USE ONLY ELECTRICAL VOLTAGE
SPECIFIED ON THE SPECIFICATION LABELAFFIXED TO THE DEVICE.
WARNING: DO NOT USE EXTENSION CORDS WITH
THIS UNIT OR CONNECT TOO MANY PLUGS INTO
THE SAME ELECTRICAL OUTLET. THE USE OF
EXTENSION CORDS COULD ADVERSELY AFFECT
THE PERFORMANCE OF THE DEVICE. TOO
MANY PLUGS INTO ONE OUTLET CAN RESULTIN AN OVERLOAD TO THE ELECTRICAL PANELCAUSING THE BREAKER/FUSE TO ACTIVATE
OR FIRE IF THE BREAKER OR FUSE FAILS TO
OPERATE.
PN MN232-C4 K | User Manual

ENG - 7

FreeStyle Comfort
CAUTION: In the event of an alarm or
you observe the Oxygen Concentrator
is not working properly; consult the
Troubleshooting section in this manual.
If you cannot resolve the problem,
consult your Equipment Provider.
CAUTION: To prevent a void warranty, follow all
manufacturers’ instructions.
WARNING: DO NOT ATTEMPT ANY
MAINTENANCE OTHER THAN THE
POSSIBLE SOLUTIONS LISTED WITHIN
THIS MANUAL. DO NOT REMOVE
COVERS, ONLY YOUR EQUIPMENTPROVIDER OR A QUALIFIED SERVICETECHNICIAN SHOULD REMOVE THE
COVERS OR SERVICE THIS DEVICE.
WARNING: ONLY ACCESSORIES RECOMMENDED
BY THE MANUFACTURER. USE OF ANY OTHER
MAY BE HAZARDOUS, CAUSE SERIOUS DAMAGE
TO YOUR OXYGEN CONCENTRATOR AND WILL
VOID THE WARRANTY .
WARNING: THE MANUFACTURER RECOMMENDS
AN ALTERNATE SOURCE OF SUPPLEMENTALOXYGEN IN THE EVENT OF A POWER OUTAGE,
ALARM CONDITION, OR MECHANICAL FAILURE.
CONSULT YOUR PHYSICIAN OR EQUIPMENTPROVIDER FOR THE TYPE OF RESERVE SYSTEMREQUIRED.
WARNING: DO NOT ALLOW EITHER THE AIRINTAKE OR THE AIR OUTLET VENTS TO BECOMEBLOCKED. DO NOT DROP OR INSERT ANYOBJECTS INTO ANY OPENINGS ON THE DEVICE.
THIS CAN CAUSE THE OXYGEN CONCENTRATORTO OVERHEAT AND IMPAIR PERFORMANCE.
WARNING: SMOKING WHILE USING OXYGEN IS
THE NUMBER ONE CAUSE OF FIRE INJURIES AND
RELATED DEATHS. YOU MUST FOLLOW THESE
SAFETY WARNINGS:WARNING: DO NOT ALLOW SMOKING, CANDLES,OR OPEN FLAMES IN THE SAME ROOM WITH
THE DEVICE OR THE OXYGEN-CARRYINGACCESSORIES.
WARNING: SMOKING WHILE WEARING ANOXYGEN CANNULA MAY CAUSE FACIAL BURNSAND POSSIBLY DEATH.
WARNING: IF YOU SMOKE, THESE 3 STEPSMAY SAVE YOUR LIFE: TURN OFF THE OXYGENCONCENTRATOR, TAKE OFF THE CANNULA,
AND LEAVE THE ROOM WHERE THIS DEVICE IS
LOCATED.

8 - ENGPN MN232-C4 K | User Manual

WARNING: “NO SMOKING – OXYGEN IN USE”
SIGNS MUST BE PROMINENTLY DISPLAYEDIN THE HOME, OR WHERE OXYGEN IS IN USE.
USERS AND THEIR CAREGIVERS MUST BE
INFORMED ABOUT THE DANGERS OF SMOKING
IN THE PRESENCE OF, OR WHILE USING,
MEDICAL OXYGEN.
WARNING: DO NOT USE YOUR OXYGENCONCENTRATOR IN THE PRESENCE OF
FLAMMABLE GASES. THIS CAN RESULT IN RAPID
BURNING CAUSING PROPERTY DAMAGE, BODILYINJURIES OR DEATH.
WARNING: REMOVING THE CANNULA AND
PUTTING IT ON CLOTHING, BEDDING, SOFAS,OR OTHER CUSHION MATERIAL WILL CAUSE A
FLASH FIRE WHEN EXPOSED TO A CIGARETTE,HEAT SOURCE, OR FLAME.
WARNING: DO NOT LEAVE A NASAL CANNULA
ON OR UNDER CLOTHING, BED COVERINGS OR
CHAIR CUSHIONS. IF THE UNIT IS TURNED ON
BUT NOT IN USE, THE OXYGEN WILL MAKE THE
MATERIAL FLAMMABLE. ENSURE THE POWER
IS OFF WHEN THE OXYGEN CONCENTRATOR IS
NOT IN USE.
WARNING: THE OXYGEN CONCENTRATOR MUSTBE OFF AND DISCONNECTED FROM EXTERNALPOWER BEFORE REMOVING THE BATTERY PACK.
WARNING: SMALL PARTS MAY
PRESENT A POTENTIAL CHOKING
HAZARD.
WARNING: AVOID CONTACT WITH EXHAUST AIR
VENT FOR LONGER THAN 10 SECONDS AS IT MAY
BE WARM.
WARNING: WIND OR STRONG DRAUGHTS CANADVERSELY AFFECT ACCURATE DELIVERY OF
OXYGEN THERAPY.
WARNING: THIS PRODUCT CAN EXPOSE YOU
TO CHEMICALS INCLUDING NICKEL, WHICH
IS KNOWN TO THE STATE OF CALIFORNIA TO
CAUSE CANCER. FOR MORE INFORMATION,
GO TO WWW.P65WARNINGS.CA.GOV.
CAUTION: Always disconnect AC
power supply from the wall before
disconnecting the AC power supply fromthe Oxygen Concentrator.
CAUTION: Do not position the Oxygen
Concentrator so that it is difficult to access thepower cord.

FreeStyle Comfort
CAUTION: Always place oxygen supply tubing andpower cords in a manner that prevents trip hazard
or possible accidental strangulation.
CAUTION: When using the Oxygen Concentrator in
an automobile, boat, or on other DC sources with
the DC power supply, make sure that the vehicle isstarted and running before connecting the Oxygen
Concentrator. Failure to follow these instructions
can result in the power supply not supplying power
to the Oxygen Concentrator.
CAUTION: When the automobile in which you
are using the Oxygen Concentrator unit is turnedoff, disconnect and remove the device from theautomobile. Do not store the Oxygen Concentrator
in a very hot or cold automobile or in other similar,
high- or low-temperature environments.
Note: Portable and mobile RF communications equipment can effect medical electrical equipment.
CAUTION: DO NOT leave the Oxygen
Concentrator or the Power Supply
plugged into the vehicle if the ignition isin the OFF position. Doing so may drain
the vehicle’s battery.
CAUTION: Operating or storing the Oxygen
Concentrator outside of its normal operating
temperature range and ambient pressure range(altitude) can affect performance and decrease
battery run time and/or increase battery charge
time. Refer to the Specifications section in thismanual for storage and operating temperature
limits.
CAUTION: For Oxygen Concentrators equippedwith batteries: Store in a cool and dry location to
help ensure the longevity of your battery. Storing
your Oxygen Concentrator for extended periods oftime at high temperatures or with a fully charged
or completely discharged battery can degrade
its overall battery life. Do not attempt to open thebattery; there are no serviceable parts inside thebattery. Keep batteries away from children.
CAUTION: ONLY USE the Manufacturer’s provided
batteries. For proper battery disposal, contact
your Equipment Provider or your local government
agency for disposal requirements.

Note: Ensure the cannula is fully inserted and secure.
This ensures that the Oxygen Concentrator canproperly detect inspiration for oxygen delivery. During
inhalation, you should hear or feel oxygen flow to theprongs of the nasal cannula.
Note: Lithium batteries may permanently lose capacity
when exposed to extremely hot temperatures with thebatteries fully charged or completely depleted. For
extended storage, it is recommended that batteries becharged 25 to 50% and remain within a temperature
range of -4°F – 68°F (-20°C – 20°C).
Note: The use of some oxygen administration
accessories not specified for use with thisoxygen concentrator may impair its performance.
Recommended accessories are referenced within thismanual.
Note: Do not operate the Oxygen Concentrator
without the air intake filter in place. If a second filter
is provided, insert the replacement filter before you
clean the dirty filter. Clean the dirty filter in a warm
soap and water solution, then dry thoroughly prior touse.
Note: The Manufacturer does not recommend thesterilization of this equipment.
Note: There is never a danger of depleting the oxygenin a room when you use your oxygen concentrator
unit.
WARNING: THE OXYGENCONCENTRATOR SHOULD NOT BEOPERATED OR PLUGGED IN TO
EXTERNAL (AC/DC) POWER WITHOUTA BATTERY INSTALLED. DO NOTATTEMPT TO POWER ON THE UNIT
OR CONNECT IT TO EXTERNALPOWER WHEN THE BATTERY PACK ISREMOVED.
ONLY REMOVE THE BATTERY PACK WHEN THE
UNIT IS POWERED OFF AND UNPLUGGED FROM
THE POWER SUPPLY. DO NOT ATTEMPT TOREMOVE THE BATTERY PACK WHILE THE UNIT
IS STILL ON OR PLUGGED INTO THE POWER
SUPPLY.
Note: If the oxygen concentrator has been stored
for an extended period of time outside of its normal
environmental operating conditions shown in thetable above, it should be allowed to return to normal
operating temperature before being powered on.

PN MN232-C4 K | User Manual

ENG - 9

FreeStyle Comfort

Specifications

FreeStyle Comfort
Dimensions (with Single Battery)

10.0 x 7.3 x 3.1 in (25.4 x 18.5 x 7.9 cm)

Dimensions (with Double Battery)

11.0 x 7.3 x 3.1 in (27.9 x 18.5 x 7.9 cm)

Weight (with Single Battery Pack)

5 lbs (2.3 kg)

Weight (with Double Battery Pack)

6 lbs (2.7 kg)

Nominal Sound Level***

39,93 dB(A) ± 0,42 dB(A) @ 2

Flow Settings

Pulse Setting 1, 2, 3, 4, 5

O2 Output Setting (@ 20bpm)

Setting 1 - 210 mL/min
Setting 2 - 420 mL/min
Setting 3 - 630 mL/min
Setting 4 - 840 mL/min
Setting 5 - 1050 mL/min

Oxygen Concentration*
Oxygen Monitor
Maximum Outlet Pressure
Dosing Sensitivity

90% (+5.5% / -3%)
Yes
<30 PSI> -0.1 cm H2O (adjustable)

AC Power

100–240 VAC, 50-60Hz, 120W Max

DC Power

11–18 VDC (10 max amp)

Battery Type

Lithium Ion

Battery Capacity

Single cell: 6700 mAh; Double cell: 13400 mAh

Battery Re-Charge Time

Single Battery: 3.5 hours
Double Battery: 6.0 hours

Battery Pack Durations

Single Battery Duration: Setting 2 - up to 4 hours
Double Battery Duration: Setting 2 - up to 8 hours

Operating Temperature**

41°F to 104°F (5°C to 40°C)

Transportation and Storage Temp. (Unit)-13°F to +158°F (-25°C to +70°C) up to 10000 ft (3048 m)

Transportation and Storage Temp.
(Battery)-13°F to +158°F (-25°C to +70°C) up to 10000 ft (3048 m)

Operating Altitude**-1250 to 10,000 ft (-381 to 3048 m) (tested to 700 – 1060 hPa)

Operating Humidity

15 - 95% relative humidity (non-condensing)

Storage Humidity

0 - 90% Non-Condensing

Limited Warranty

3 years unit
1 year battery & power supplies* Based on an atmospheric pressure of 14.7 psi (101 kPa) at 70°F (21°C)** The unit will function in all environments as long as the environment is with the specified temperature and humidity
ranges. Operating outside of these operational specifications can limit the concentrator’s ability to meet Oxygen Concentration specification at higher liter flow rates. Warm up screen may continue to display until the concentrator has reachedtarget concentration or up to 15 minutes. Continue to use unit during this time.
***Sound level measured per test method Nr. 14-1 10/2018 MDS-Hi.
The expected service life is a minimum of five years. The expected shelf life of the sieve beds is approximately one year
if concentrator is unused during the entire period.

10 - ENGPN MN232-C4 K | User Manual

Operating Instructions
1. Locate and position the FreeStyle Comfort in aproper location for use.
a. The concentrator should be well-ventilated
so that the air intake filters and air outlet are
not obstructed.
b. The concentrator should be positioned so that
all audible and visual indicators and alarms
can be easily seen and heard.
c. Be sure the air intake filters are in place
before operating the unit. If the unit ismissing a filter, contact your equipment
provider.
2. Install a Battery Pack in the concentrator.
a. Align the battery with the bottom housingfrom the center of the FreeStyle Comfort.

b. Slide the battery inward until you hear aclick to confirm that it has locked in place.

FreeStyle Comfort
power port receptacle on the front, bottom rightside of the unit.

4. Connect a nasal cannula to the oxygen outlet
barb. Ensure the connection is secure before
operation.
NOTE: Any cannula(s) used other than the recommended may affect performance.

Note: To Equipment Provider: The following oxygen
administration accessories are recommended for use
with the FreeStyle Comfort:• Nasal Cannula with 7 feet (2.1 m) of tubing (Maximum LPM: 6 LPM): CAIRE Part Number 5408-SEQ• Additional recommended accessories information isavailable online at www.caireinc.com.

3. Insert the AC or DC power supply into the
PN MN232-C4 K | User Manual

ENG - 11

FreeStyle Comfort
5. Turn on the FreeStyle Comfort by pressing thepower button and select your prescribed flow
setting utilizing the increase (+) or decrease (-)
flow buttons.

autoSAT Pulse Volume (mL)*Setting
Breath
Rate

1

2

3

4

5

15

10.5

21.0

31.5

42.0

52.5

20

10.5

21.0

31.5

42.0

52.5

25

10.5

21.0

31.5

42.0

42.0

30

10.5

21.0

31.5

35.0

35.0

35

10.5

21.0

30.0

30.0

30.0

40

10.5

21.0

26.3

26.3

26.3

*Tolerance is +/- 15%
non-autoSAT Pulse Volume (mL)* Setting

A brief audible alarm will sound to indicate thedevice has powered on.

Breath
Rate

1

2

3

4

5

Note: When the FreeStyle Comfort is initially turnedon the default flow setting is 2. If the setting ischanged, the unit will power on to the previous flow
setting.

15

14.0

28.0

42.0

56.0

70.0

20

10.5

21.0

31.5

42.0

52.5

25

8.4

16.8

25.2

33.6

42.0

30

7.0

14.0

21.0

28.0

35.0

6. Position the nasal cannula on your face per the
manufacturer’s instructions provided with thecannula and breathe normally. You should feela puff of oxygen delivered through the nasal
cannula each time you inhale.

35

6.0

12.0

18.0

24.0

30.0

40

5.3

10.5

15.8

21.0

26.3

7. To change the flow setting, press the increase (+)
or decrease (-) flow buttons to the desired flow
rate. It is normal to hear a difference in sound as
you change the flow settings.

8. Turn the FreeStyle Comfort off by pressing thepower button.
Note: The display screen will enter standby mode anddim after 20 seconds. Press any button to wake thescreen out of standby mode.

9. For Advanced Features see Menu Navigation
Instructions

SPECIAL PRODUCT FEATURES
autoSAT®
CAIRE’s proprietary autoSAT feature allows the
FreeStyle Comfort to deliver a consistent volumeof oxygen (bolus) per breath, up to a maximum
breath rate. As the user’s breath rate increases
during activities like walking, the autoSAT featureadjusts the device operation to maintain a set bolus
volume.
UltraSense®
This sensitive pulse dose trigger detects thepressure change during your breath, and releases
the pulse of oxygen when you inhale.
autoDOSE™
Should a “no breath alarm” occur, this feature willpulse oxygen at 20 breaths per minute at the unit’s
current flow setting until the alarm state is resolved.
12 - ENGPN MN232-C4 K | User ManualPower Supplies
The FreeStyle Comfort can be powered in threeways:• Rechargeable Battery Pack
• AC Power• DC Power
Information on the operation of each of thesecomponents is in the following sections.
WARNING: USE ONLY THE POWER
SUPPLIES OUTLINED IN THIS USER’SMANUAL. USE OF NON-CAIRE POWER
SUPPLIES AND BATTERY PACKS CANCAUSE DAMAGE TO THE UNIT AND
IMPROPER OPERATION.

FreeStyle Comfort
REMOVING THE BATTERY

Push the battery release button downward awayfrom the FreeStyle Comfort. While HOLDING thisbutton, slide the battery out by pulling it away fromthe concentrator.
WARNING: THE OXYGENCONCENTRATOR MUST BE OFF ANDDISCONNECTED FROM EXTERNALPOWER BEFORE REMOVING THE
BATTERY PACK.

BATTERY PACKThe FreeStyle Comfort can be operated by therechargeable battery pack which is supplied withthe device. There are no operational limitations of
the concentrator when running from the battery
pack. All flow rates are available on battery power.

CHECKING BATTERY PACK CHARGEINSTALLING THE BATTERY

Align the battery with the bottom housing from thecenter of the FreeStyle Comfort. Push the battery
inward until you hear a click to confirm that it has
locked in place.

To check the battery charge when it is installed in
the FreeStyle Comfort, reference the battery icon
and the estimated time remaining on the display.
80–100%60–80%

40–60%

20–40%

20%

PN MN232-C4 K | User Manual

ENG - 13

FreeStyle Comfort
Note: If the battery is in the process of charging, a plug
indicator will show instead of the charge percentage
remaining.
To check the battery charge when it is not installedin the FreeStyle Comfort, lay it flat and press the
TEST button. The battery gauge indicator lights(25-100%) will illuminate to the right of the TEST
button to indicate the level of the battery packcharge.

BATTERY OPERATING DURATION

When external AC or DC power is disconnected,
the concentrator will automatically switch over tousing battery power if a charged battery is installed.
The table below shows the typical durations fora new battery pack. Battery pack durations willremain constant regardless of your breath rate.
Pulse
Setting

Single (8-cell)
Battery Duration

Double (16-cell)
Battery Duration1

up to 8 hours

up to 16 hours

2

up to 4 hours

up to 8 hours

3

up to 3 hours

up to 6 hours

4

up to 2.25 hours

up to 4.5 hours

5

up to 2 hours

up to 4 hours

TYPICAL BATTERY RECHARGE TIME

The typical time to fully recharge your battery packfrom a fully discharged battery is 3.5 hours for asingle battery, and 6.0 hours for a double battery.

INITIAL CHARGINGThe new battery supplied with your FreeStyle
Comfort is only partially charged when it is shippedfrom the factory. Before using for the first time, you
should fully charge the battery pack using either
AC or DC power.

GENERAL CHARGING INFORMATIONThe FreeStyle Comfort will recharge the battery
pack anytime it is installed in the concentrator andthe concentrator is connected to an external AC or
DC power source (except on an airplane).
Battery is fully charged when it displays 100%.
Alternately, the battery pack may be charged using
the optional desktop charger.

14 - ENGPN MN232-C4 K | User Manual

Note: The battery may not initially start charging if thebattery pack has been fully depleted.

BATTERY SERVICE LIFE

The FreeStyle Comfort batteries are designed tolast 300 charge/discharge cycles.

AC & DC Power Supply

FreeStyle Comfort

The FreeStyle Comfort comes standard with AC
and DC power supplies.
The AC power supply includes the following:a. AC Power brick that operates between
100-240 volts, 50-60Hz.
b. AC Power cord to connect to the power
brick and corresponding AC outlet.

The DC power supply consists of a cord that plugs
directly into the unit and the DC outlet.

AC POWER

2. Connect the AC power cord to the AC power
brick.

AC power is for use at home or wherever standard
AC outlet power is available. The FreeStyle
Comfort is fully functional on AC power. The
FreeStyle Comfort will also recharge the battery
pack any time AC power is available (except on an
airplane). The battery pack will recharge whether
the FreeStyle Comfort is powered on or off.
WARNING: DO NOT CONNECT
EXTERNAL POWER WITHOUT A
BATTERY PACK INSTALLED IN THE
OXYGEN CONCENTRATOR.
To connect the FreeStyle Comfort to AC Power:
1. Connect the AC power brick to the power
connector inlet on the front, right bottom of the
FreeStyle Comfort. Be sure to push inward sothat the connection is secure.

3. Connect the opposite end of the AC power cord
to the wall or other appropriate electrical outlet.

PN MN232-C4 K | User Manual

ENG - 15

FreeStyle Comfort
4. Turn on the FreeStyle Comfort by pressing thepower button and select your prescribed flow
setting utilizing the increase (+) or decrease (-)
flow buttons.

Note: If the battery pack needs to be recharged, thebattery icon on the display will have a single orangebar.

DC POWER

DC power is for use in the accessory outlets ofautomobiles, boat, or other motor vehicles. The
FreeStyle Comfort is fully functional for use on DC
power. The FreeStyle Comfort will also recharge
the battery pack if needed any time DC power isavailable. The battery pack will recharge both if theunit is off, or on and in use.
WARNING: DO NOT CONNECT
EXTERNAL POWER WITHOUT A
BATTERY PACK INSTALLED IN THE
OXYGEN CONCENTRATOR.
WARNING: DO NOT OPERATE THE DEVICE ON
12 VDC CAR OUTLETS WHILE THE VEHICLE IS
TURNED OFF.
1. Turn on your motor vehicle.
2. Connect the DC power supply to the power
port receptacle front, bottom right side of the
FreeStyle Comfort. Be sure to push inward sothat the connection is secure.

16 - ENGPN MN232-C4 K | User Manual

3. Connect the opposite end of the DC power cord
to the automobile or other DC accessory outlet.

4. Turn on the FreeStyle Comfort by pressing thepower button and select your prescribed flow
setting utilizing the increase (+) or decrease (-)
flow buttons.
Note: If the battery pack needs to be recharged, thebattery icon on the display will have a single orangebar.

FreeStyle Comfort
Cleaning, Care & Routine
Maintenance
ROUTINE MAINTENANCE

Preventative MaintenanceThe device is maintenance free. Filters insidethe device are intended to last the lifetime of thedevice. If the device is not working properly,
the unit should be returned to provider for
repair. Servicing of internal components insideof the cabinet of the FreeStyle Comfort must be
performed by the oxygen provider.
The operator of the unit is responsible for thefollowing maintenance procedure.
• Weekly – Clean the Air Intake Filters
WARNING: DO NOT ATTEMPT TO OPEN
OR REMOVE THE PLASTIC COVERS OF
THE OXYGEN CONCENTRATOR. THE
UNIT SHOULD ONLY BE OPENED BY AQUALIFIED SERVICE TECHNICIAN.

AIR INTAKE FILTERS

Air enters the FreeStyle Comfort through air intake
filters located on the front, bottom of the oxygen
concentrator. These filters prevent dust and otherlarge particles in the air from entering the unit.
At least once perW week, the air intake filters
should be cleaned. Your oxygen provider may
advise you to clean them more frequently,
depending upon the operating conditions. Follow
these steps to properly clean the air intake filters.
1. Remove the two filters from the front of
the FreeStyle Comfort by pushing up on thecorresponding tab and pulling outward.

WARNING: DO NOT OPERATE THE
OXYGEN CONCENTRATOR WITHOUTTHE AIR INTAKE FILTER IN PLACE.

BATTERY PACK CARE

Do not allow the battery pack to be exposed toliquids.
Do not attempt to disassemble or open the battery
pack.
Do not expose the battery to open flames.
Do not drop the battery pack.
Keep the battery pack away from children.
If your battery pack appears damaged, contact youroxygen provide before use.
Do not place the battery in or near a microwave orother cooking appliances.

BATTERY PACK STORAGE

The oxygen concentrator battery does not need
to be fully discharged before recharging. It is
recommended to charge the oxygen concentrator
battery after each use.
Do not leave the battery in a location exposed toexcessive heat, such as in direct sunlight or in a car.
Doing so could cause the battery’s performanceand life to deteriorate. It may also cause thebattery to generate heat, smoke, catch fire, or
explode. The battery pack should be stored in astable environment characterized by low-humidity(less than 70% RH) and free of corrosive gasses.
To prevent rust, avoid conditions that can create
condensation such as rapid fluctuations in theambient temperature.
For storage of a month or more, a discharged or
partial charged state of charge between 50% and
75% (two to three LEDs) is recommended. Once
charged, remove the battery from the oxygen
concentrator.

2. Wash both filters in a warm solution of soap andwater.
3. Rinse the filters thoroughly and remove excesswater with a soft, lint-free cloth. Ensure the filters
are completely dry before replacing it.
4. Re-install the clean, dry filters.
PN MN232-C4 K | User Manual

ENG - 17

FreeStyle Comfort
CLEANING

Turn OFF the unit and disconnect from AC or DC
power before any cleaning. DO NOT spray theouter case directly. Use a damp (not wet) clothor sponge. Spray the cloth or sponge with a mildhousehold cleaner solution to clean the cabinet andpower supplies. Proceed as directed by the cleaner
manufacturer.
WARNING: TURN OFF THE UNIT AND
DISCONNECT THE POWER CORD FROMTHE ELECTRICAL OUTLET BEFOREYOU CLEAN THE UNIT TO PREVENT
ACCIDENTAL ELECTRICAL SHOCK AND
BURN HAZARD. ONLY YOUR EQUIPMENT PROVIDER OR A QUALIFIED SERVICE TECHNICIAN SHOULD REMOVETHE COVERS OR SERVICE THE UNIT.
WARNING: CARE SHOULD BE TAKEN TOPREVENT THE OXYGEN CONCENTRATOR AND
THE BATTERY PACK FROM GETTING WET
OR ALLOWING FLUIDS TO ENTER THE UNIT.
THIS CAN CAUSE A MALFUNCTION OR SHUT
DOWN, AND CAUSE AN INCREASED RISK FOR
ELECTRICAL SHOCK OR BURNS.
WARNING: DO NOT USE LIQUID DIRECTLY ON
THE UNIT. A LIST OF UNDESIRABLE CHEMICALAGENTS INCLUDES BUT IS NOT LIMITED TO THE
FOLLOWING: ALCOHOL AND ALCOHOL-BASEDPRODUCTS, CONCENTRATED CHLORINE-BASEDPRODUCTS (ETHYLENE CHLORIDE), AND OILBASED PRODUCTS. THESE ARE NOT TO BE
USED TO CLEAN THE PLASTIC HOUSING ON
THE CONCENTRATOR, AS THEY CAN DAMAGE
THE UNIT.
WARNING: CLEAN THE CABINET, CONTROLPANEL, AND POWER CORD ONLY WITH A MILD
HOUSEHOLD CLEANER APPLIED WITH A DAMP(NOT WET) CLOTH OR SPONGE, AND THEN
WIPE ALL SURFACES DRY. DO NOT ALLOW ANY
LIQUID TO GET INSIDE THE CONCENTRATOR. PAY
SPECIAL ATTENTION TO THE OXYGEN OUTLETFOR THE CANNULA CONNECTION TO MAKE
SURE IT REMAINS FREE OF DUST, WATER, AND
PARTICLES.

18 - ENGPN MN232-C4 K | User Manual

WARNING: USE A DAMP (NOT WET) CLOTH ORSPONGE TO CLEAN THE BATTERY PACK. FIRSTSPRAY THE CLOTH OR SPONGE WITH A MILD
HOUSEHOLD CLEANER AND CLEAN THE BATTERY PACK HOUSING.
WARNING: DO NOT CLEAN WITH AUTOMATED
WASHING SYSTEM.

CANNULA REPLACEMENT

Always follow the cannula manufacturer’s
instructions for proper use. Replace the nasal
cannula or oxygen tubing every 3 to 6 months(if daily use, 3 months) as per manufacturer’s
recommendation as recommended by the cannula
manufacturer or your oxygen provider. Yourphysician or oxygen provider will provide you withcleaning and replacement instructions.
Additional supplies for replacement are available
through your oxygen provider.

FreeStyle Comfort

Alarm Conditions
Priority #Alarm
Description

1

System
Malfunction
Battery High
Temperature

Display

AudibleSignalMore Details / Suggested Action

1 beepevery
20 secUnit will alarm when battery temperature isapproaching safety shutdown threshold.
If shutdown threshold is reached, unit will
shut down. If external power is connected,
alarm screen will display and audible alarm
will continue for 2 min.
Test with battery at room temperature. If
problem persists, contact provider.

2

System
MalfunctionHigh
Temperature

1 beepevery
20 secUnit will alarm when internal temperature isapproaching shutdown threshold.
If shutdown threshold is reached, unit will
shut down and alarm screen will display
and audible alarm will continue for 2 min.
Ensure there are no blockages on any
vents of the unit. If problem persists,
contact provider.

3

4

System
Malfunction
High
Pressure

1 beepevery
20 secNo Breath
Detected

1 beepevery
30 secIf shutdown threshold is reached, unit will
shut down and alarm screen will display
and audible alarm will continue for 2 min.
If problem persists, contact provider.
No breath detected for 15 seconds.
Activated after first breath or warm up
screen is off, whichever comes sooner.
Check cannula for kinks or leaks. Increase
the sensitivity setting. If problem persists,
contact provider.

5

6

Low Battery
Shut Down

Warm Up

1 beepevery
30 secAlarms until battery runs out of power andunit shuts down.

none

Unit will start by auto-dosing until a breathis detected or Warm up ends, whicheveris sooner.

Charge battery.

PN MN232-C4 K | User Manual

ENG - 19

FreeStyle Comfort
Priority #Alarm
Description

7

Display

AudibleSignalMore Details

Low
Concentration

1 beepevery
30 sec,for up to
5 min

If problem persists, contact provider.

8

System
Malfunction
Flow Rate(OCD Valve
Failure,Obstructionof GasPathway)

1 beepevery
30 secIf problem persists, contact provider.

9

Low Battery
Warning

1 beep,no repeat

Charge battery.

10

Breathing
Rate
Exceeded

1 beepevery
60 secBreathing Rate exceeds 40 bpm.

11

System
Malfunction
Fan Failurenone

If problem persists, contact provider.

12

System
Malfunction
Low Pressurenone

If problem persists, contact provider.

13

System
Malfunction
Low Temp

none

Allow unit to warm up to room temperature.
If problem persists, contact provider.

Info

Start up

1 infobeep

Screen displays at power-on and you
should hear one beep to confirm buzzer isfunctioning

If dose is being falsely triggered, decrease
the sensitivity setting. If problem persists,
contact provider.

Start up screen display current firmware
revision in bottom right corner.
20 - ENGPN MN232-C4 K | User ManualFreeStyle Comfort
Priority #Alarm
Description

Display

AudibleSignalInfo

Normal Run
Screen in
autoSAT
Mode (noDC Powerconnected)

none

Info

Normal Run
Screen with
autoSATdisabled (NoDC Powerconnected)

None

Info

SettingChanged

1 infobeep

InfoPlug inexternal
Power in theoff condition

none

Info

Plug inexternal
power in the
run condition

none

Info

Unplugexternal
power in theoff Condition

1 infobeep

InfoUnplugexternal
power in the
run Condition(with battery
present)

1 infobeepMore Details

Display CAIRE screen for 2 seconds,
followed by Battery & Plug status for 5 sec
before screen turns off

Display CAIRE screen for 2 seconds andthen turn display off.

PN MN232-C4 K | User Manual

ENG - 21

FreeStyle Comfort
Priority #Alarm
Description

Info

AudibleSignalMore Details

In Standby
Mode - whenup or down
button ispresseddisplay showsbattery andcharging
state

none

Screen will display for 5 sec then turn off

Info

Airplane
Mode on battery power

1 longbeep

Info

Airplane
Mode while
plugged in

1 longbeep

22 - ENGPN MN232-C4 K | User Manual

DisplayMenu Navigation Instructions

FreeStyle Comfort

You can enter the Menu in Standby or Run mode.
In Run Mode, if an alarm condition occurs themenu will automatically be exited. If no buttons
are pressed for 20 seconds, the Menu willautomatically be exited. In Standby, you may needto press the Menu button twice to wake unit, thenenter the Menu.
There are four Menu screens:
1. Status screen: Displays Firmware rev number,run time, battery discharge cycles

Airplane Mode

2. Airplane mode (ON/OFF)
3. Sensitivity: adjustable (1-5)
4. autoSAT (ON/OFF)

Sensitivity

Example Status Screen

SETTINGS MENU OPERATION
Settings menu is entered by pressing Menu/SelectbuttonautoSATin Run or Standby modes.

To scroll through the menus, push the.

Status Page

PN MN232-C4 K | User Manual

ENG - 23

FreeStyle Comfort
STATUS PAGE
The Status Page of the Freestyle Comfort shows
Firmware Revision, Run Hours, and number ofbattery cycles on the attached battery.

To cancel Airplane Mode on, press
CANCEL icon is visible.

while theAirplane Mode On Confirmed
Status Page

Run hours are shown in whole hours.
Battery Discharges is the number of cycles theattached battery has been drained and recharged.

AIRPLANE MODE
The FreeStyle Comfort ships with Airplane Modeturned off.

To turn Airplane Mode off, repeat above while
toggling to OFF.
Pressing
when the screen show the “Next”
symbol will move the menu page to the next screen
option.
Airplane Mode is intended for use in Run Mode.
Airplane Mode will be automatically disabled if theunit is left in Standby on battery power.
Note: See section Activate Airplane Mode for more
information about Airplane Mode.

SENSITIVITY MODE
Airplane Mode Main Screen

To turn the mode on, press
while the CHANGE
icon is visible. The value will toggle to the blue
word “ON”.

The FreeStyle Comfort ships with a default
Sensitivity Mode setting of 2 from the factoryand should not require adjustment under normal
conditions. Sensitivity Setting can be adjusted
between values 1-5, where 1 is the least sensitive,
5 is the most sensitive.

Sensitivity Mode Screen at Setting 2
Airplane Mode On Selected

To confirm Airplane Mode on, press
while theSAVE icon is visible. The value will toggle to theblack word “ON”.

24 - ENGPN MN232-C4 K | User Manual

To adjust the Sensitivity Mode, press
while theCHANGE icon is visible. The value will toggle tothe next blue number.

FreeStyle Comfort
AUTOSAT MODE
The FreeStyle Comfort ships with autoSAT Modeturned on.

Sensitivity Mode Setting 3 Selected

To confirm Setting 3 in Sensitivity Mode, press
while the SAVE icon is visible. The value will
toggle to the black “3”.
To cancel the Sensitivity Mode change, press
while the CANCEL icon is visible.

autoSAT Mode Main Screen

To turn the mode off, press
while theCHANGE icon is visible. The value will toggle tothe blue word “OFF”.

Sensitivity Mode Setting 3 Confirmed
autoSAT Mode Off Selected

To confirm autoSAT Mode off, press
while theSAVE icon is visible. The value will toggle to theblack word “OFF”.
To cancel autoSAT Mode on, press
CANCEL icon is visible.

while theSensitivity Mode Setting 5 (Max) Confirmed

Pressing
when the screen show the “Next”
symbol will move the menu page to the next screen
option.

autoSAT Mode Off Confirmed

To turn autoSAT Mode on, repeat above while
toggling to ON.
Pressing
when the screen show the “Next”
symbol will move the menu page to the next screen
option.

PN MN232-C4 K | User Manual

ENG - 25

FreeStyle Comfort

Troubleshooting
Problem

Probable Cause

Solution

FreeStyle Comfort displays
alarm and produces
intermittent beep.

Refer to Alarm Conditions
table.

Refer to Alarm Conditions table.

Oxygen concentrator does
not turn on.

Battery is discharged.

Power the unit through the AC or DC outlet.

Not connected to external
power.

Ensure that external connects are secure.

General malfunction.

Perform a Full Power Cycle by removing thebattery and disconnecting any AC or DC external
power source. Then reinstall battery, and thepower cycle is complete. Reconnect external
power source if desired.
Contact your Equipment Provider, and change toanother source of oxygen as necessary.

All other problems.

Perform a Full Power Cycle by removing thebattery and disconnecting any AC or DC external
power source. Then reinstall battery, and thepower cycle is complete. Reconnect external
power source if desired.
Change to another source of oxygen as
available, and contact your Equipment Provider.

Note: Some alarms will cause the unit to shut down.
Note: During a total power loss, the alarm system resets (clears any previous alarms) and will only display an
alarm if the condition still exists once power is restored.

26 - ENGPN MN232-C4 K | User ManualTravel Information

FreeStyle Comfort
1. Turn POC On using Battery Power ONLY, whenno external power source is connected.

The FreeStyle Comfort meets FAA guidelines foruse on-board commercial aircraft. Prior to any
travel (airline, train, cruise, etc.) you should:• Ensure the concentrator is in good operation
condition.
• Contact your carrier at least 48 hours prior to
departure to inform them you will be travelingwith a FreeStyle Comfort.
It is suggested to always carry extra batteries withyou on trips. For airline travel, please allow enoughbattery life to account for ground time (preflightcheck in, security, etc.) and layovers. Most airlines
require that you have an adequate number of fully
charged battery packs to power the device for atleast 150% of the expected maximum trip duration(flight times, ground time before and after theflight, connections, and unexpected delays).

2. Enable Airplane Mode using one of thefollowing methods:
a. Follow directions to enable Airplane Mode in
Menu Navigation Instructions section.

Note: If flying, it is recommended that extra batteries
are carried on in a secure carrying bag.

FAA REGULATION

Spare batteries must be individually protected toprevent short circuits by placement in the originalretail packaging or by otherwise insulating theterminals, e.g. by taping over exposed terminals
or placing each battery in a separate plastic bag orprotective pouch, and carried in carry-on baggage
only.

b. Press and hold both Up and Down buttons for
5 seconds until you hear a long beep and see theAirplane symbol appear.

ACTIVATE AIRPLANE MODE
Battery Power
POC can be used on an aircraft per normal
operating procedures using Battery Power ONLY,
when no external power source is connected
External PowerOn an airplane, if an electrical outlet is available,
operable, and its location is appropriate for use
as a POC power source, it may serve as a backup to POC batteries. In this case, the following
recommended procedure regarding the transition
from battery to aircraft electrical power must be
followed:

3. When the airplane symbol is displayed, plug in
the POC.

Note: Due to aircraft limitations, the batteries do not
charge in Airplane Mode.

PN MN232-C4 K | User Manual

ENG - 27

FreeStyle Comfort

Accessories

INSERTING UNIT INTO CARRYING BAG

For proper performance and safety, use only theselisted accessories supplied by CAIRE throughyour oxygen provider. Use of accessories not listed
below could adversely affect the performance and/or safety of the concentrator.

1. Insert unit into bottom of bag, properly aligningfilters and battery openings.

FreeStyle Comfort Standard Accessories
Carrying Bag

MI396-1

Backpack

MI459-1

Shoulder Strap

MI397-1

AC Power Supply (All Cords
Included)

PW036-1SDC Power Cord

CD041-1

External Gross Particle Filter (Right)

FI226-1

External Gross Particle Filter (Left)

FI227-1

2. Secure zipper around bottom of pouch.

Optional Accessories Available
Additional 8-Cell Battery Pack

BT034-1S

16-Cell Battery Pack

BT035-1SDesktop Battery Pack Charger

BT036-1

Carry-all Accessory Bag

MI372-2

Note: Additional options may be available for country-specific power cords where noted above. Contact
CAIRE or your oxygen provider if alternate options areneeded for order.

SHOULDER STRAP INSTALLATION
1. Shoulder strap can be attached to either the unit
directly or the carrying bag.
2. Clip strap onto the ring on the unit or bag.

Carrying Bag Instructions
To insert and secure your FreeStyle Comfort in the
carrying bag, utilize the following instructions.

28 - ENGPN MN232-C4 K | User ManualFreeStyle Comfort

Disposal and Reuse
Always return FreeStyle Comfort, including allcomponents, to your equipment provider for proper
disposal. You can also contact your local city or
town offices for instructions on proper disposal of
the battery. The unit can be reused between people
but must be returned to equipment provider for
reprocessing.

end of its life, per Waste Electrical and Electronic
Equipment (WEEE) Directive.
Our products will comply with the restriction ofHazardous Substances (RoHS) directive. They will
not contain more than trace amounts of lead orother hazardous material content.

WEEE and RoHS
This symbol is to remind the equipment
owners to return it to a recycling facility at the

CAUTION: For proper disposal,
contact your equipment provider or
local government agency for disposal
instructions

Materials in Direct Contact with the Operator
Concentrator Casing..........................................................................................................................PC/ABS
Concentrator Control Panel.............................................................................Polyester EBG7 or equivalent
Concentrator Handle................................................................................................................Polycarbonate
Oxygen Outlet, Nozzle.............................................................................................................Polycarbonate
Unit Label..............................................................................................................................................Lexan
Cord Connectors...............................................................................................Polycarbonate/Vinyl chloride
Power Cord.................................................................................................................................. PVC, Metal
Power Supply...................................................................................................... .Lexan 940(Polycarbonate)
Battery Pack......................................................................................................................................PC/ABS
Battery Pack, Power Supply Labels......................................................................................... Polyester film
Concentrator Carrying Case..................................................................... Cordura, Neoprene, Nylon Lining
Strap................................................................................ Polypropylene, Acetyl, Vinyl, Cordura, Neoprene

EMC Testing
Guidance and Manufacturer’s Declaration­—Electromagnetic Emissions
The FreeStyle Comfort is intended for use in the electromagnetic environment specified below. The customer or theuser of the FreeStyle Comfort should assure that it is used in such an environment.
Emissions Test
RF emissions
CISPR 11
RF emissions
CISPR 11
Harmonic emissions
IEC 61000-3-2
Voltage fluctuations/
flicker emissions
IEC 61000-3-3ComplianceElectromagnetic Environment - GuidanceGroup 1The FreeStyle Comfort uses RF energy only for its internalfunction. Therefore, its RF emissions are very low and are not
likely to cause any interference in nearby electronic equipment.

Class BClass A
Complies

The FreeStyle Comfort is suitable for use in all establishments,
including domestic establishments and those directly connectedto the public low-voltage power supply network that supplies
buildings used for domestic purposes.

PN MN232-C4 K | User Manual

ENG - 29

FreeStyle Comfort
Guidance and Manufacturer’s Declaration—Electromagnetic Immunity

The FreeStyle Comfort is intended for use in the electromagnetic environment specified below. The customer or theuser of the FreeStyle Comfort should assure that it is used in such an environment.
Immunity test
IEC 60601 test level
Compliance level
Electromagnetic environment – guidanceElectromagnetic
Floors should be wood, concrete orenvironment – guid- ±8 kV contact±8 kV contact
ceramic tile. If floors are covered withance
synthetic material, the relative±15 kV air discharge±15 kV air discharge
humidity should be at least 30%.
IEC 61000-4-2
Electrical fasttransient/burst
IEC 61000-4-4

Surge
IEC 61000-4-5±2 kV for power supplylines±1 kV for input/output
lines±1 kV line(s) to line(s)±2 kV for power supply
lines
N/A

±2kV common mode on ±1 kV line(s) to line(s)AC lines±1kV differential on
AC lines

Mains power quality should be that ofa typical professional healthcare facility
and home healthcare environments.

Mains power quality should be that ofa typical professional healthcare facility
and home healthcare environments.

±2kV common mode on
outdoor I/O lines
0% UT for 0.5 cycle (0°, 0% UT for 0.5 cycle (0°,45°, 90°, 135°, 180°,
45°, 90°, 135°, 180°,
Mains power quality should be that of
225°, 270°, 315°)
225°, 270°, 315°)
a typical professional healthcare facility
Voltage dips, shortand home healthcare environments.
00interruptions and
0% UT for 1 cycle (0 )0% UT for 1 cycle (0 )If the user of the FreeStyle Comfort
voltage variations
requires continued operation during
on power supply
power mains interruptions, it is
70%
U
70%
U
(30%
dip
in
U
)
(30%
dip
in
U
)
input lines
T
T
T
T
recommended that the FreeStyle
0
0
)
)
for
25/30
cycles(0for
25/30
cycles(0
IEC 61000-4-11
Comfort is powered from anuninterruptible power supply or a battery.
0% UT for
0% UT for00250/300 cycles (0 )250/300 cycles (0 )Power frequency magnetic fields
Power frequency
should be at levels characteristic of(50/60 Hz) magnetic 3 A / m
3A/ma typical location in a professional
field IEC 61000-4-8
healthcare facility and home healthcare
environments.
NOTE UT is the A.C. mains voltage prior to application of the test level.
NOTE In some cases, the Freestyle Comfort may reset after a significant Electrostatic Discharge, and may requirea Full Power Cycle to resume normal operation. See Troubleshooting section for more details.

30 - ENGPN MN232-C4 K | User ManualFreeStyle Comfort
Guidance and Manufacturer’s Declaration–Electromagnetic Immunity

The FreeStyle Comfort is intended for use in the electromagnetic environment specified below. The customer or theuser of the FreeStyle Comfort should assure that it is used in such an environment.
Immunity test
IEC 60601 test level
Compliance level
Electromagnetic environment – guidance
10 Vrms
Portable and mobile RF communications
10 Vrms
Conducted RF
equipment should be used no closer
150 kHz to 80 MHz
IEC 61000-4-6
to any part of the FreeStyle Comfort,
including cables, than the recommended
separation distance calculated from theequation applicable to the frequency of
10 V/m
10 V/mRadiated RF
the transmitter.
80 MHz to 2700 MHz
IEC 61000-4-3
Recommended separation distanced  1.2 P
d  1.2 P = 80 MHz to 800 MHz
d  1.2 P = 800 MHz to 2700 MHz
where P is the maximum output power
rating of the transmitter in watts (W)according to the transmitter manufacturer
and d is the recommended separation
distance in metres (m).
Field strengths from fixed RF
transmitters, as determined by anelectromagnetic site survey,a should beless than the compliance level in eachfrequency range. b
Interference may occur in the vicinityof equipment marked with the following
symbol:NOTE 1 At 80 MHz and 800 MHz, the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption andreflection from structures, objects and people.
aField strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones and land
mobile radios, amateur radio, AM and FM radio broadcast and TV broadcast cannot be predicted theoretically withaccuracy. To assess the electromagnetic environment due to fixed RF transmitters, an electromagnetic site survey
should be considered. If the measured field strength in the location in which the FreeStyle Comfort is used exceeds
the applicable RF compliance level above, the FreeStyle Comfort should be observed to verify normal operation. If
abnormal performance is observed, additional measures may be necessary, such as
re-orienting or relocating the FreeStyle Comfort.
bOver the frequency range 150 kHz to 80 MHz, field strengths should be less than 10 V/m.

PN MN232-C4 K | User Manual

ENG - 31

FreeStyle Comfort
Recommended separation distances between portable and mobile RFcommunications equipment and the FreeStyle Comfort

The FreeStyle Comfort is intended for use in an electromagnetic environment in which radiated RF disturbances
are controlled. The customer or the user of the FreeStyle Comfort can help prevent electromagnetic interference by
maintaining a minimum distance between portable and mobile RF communications equipment (transmitters) andthe FreeStyle Comfort as recommended below, according to the maximum output power of the communications
equipment.
Rated maximum
Separation distance according to frequency of transmitter
output power ofmtransmitter
150 kHz to 80 MHz
80 MHz to 800 MHz
800 MHz to 2,5 GHzd  1.2 P
d  1.2 P
d  1.2 P
W
0,01
0.12
0.12
0.23
0,1
0.38
0.38
0.73
1
1.2
1.2
2.3
10
3.8
3.8
7.3
100
12
12
23
For transmitters rated at a maximum output power not listed above, the recommended separation distance d inmetres (m) can be estimated using the equation applicable to the frequency of the transmitter, where P is the maximum output power rating of the transmitter in watts (W) according to the transmitter manufacturer.
NOTE 1 At 80 MHz and 800 MHz, the separation distance for the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption andreflection from structures, objects and people.

Classification

Type of protection against electric shock:Class II
Protection from electric shock is			achieved by double insulation.
Degree of protection against electric shock:Type BF Equipment providing a particular		 degree of protection against electric
		 shock regarding		 1) allowable leakage current;		 2) reliability of protective earth connection
		 (if present).
		 Not intended for direct cardiac application.
Independent testing for Medical Electrical
Equipment Standard:
Tested to be in compliance with
• IEC 60601-1 edition 3.1: Medical Electricalequipment - Part 1 General Requirements forsafety.
• CAN/CSA C22.2 No. 60601-1-8 edition
2.1, Medical Electrical Equipment – Part 1:
General Requirements for Safety
Protection against potential electromagnetic orother interference between the equipment and otherdevices.
• Tested to be in compliance with EN 60601-1-2/IEC 60601-1-2: 4th Edition.
32 - ENGPN MN232-C4 K | User Manual• Tested to be in compliance with RTCA/DO160
Section 21 Category M.
• CISPR 11 / EN 55011 Class B Group 1,“Industrial, Scientific, and Medical (ISM)
Equipment”• FCC Part 15, Subpart B – Class B
Unintentional Radiators
IP22 - Protection against ingress of fingers orsimilar objects greater than 12.5 mm diameter.
Vertically dripping water shall have no harmful
effect when the enclosure is tilted at an angle of
15° from its normal position.
Method of cleaning and infection control allowed:
Please refer to “Cleaning, Care, and Proper
Maintenance” section of this FreeStyle User
Manual.
Degree of safety of application in the presence offlammable anesthetic gases:
Equipment not suited for such application.
Mode of operation: Continuous duty.

FreeStyle Comfort
Commandes d’utilisation et indicateurs d’état du système
ISO 7000 : symboles graphiques à utiliser surl’équipement – index et résumé
Lisez le manuel d’utilisation avant lamise en marche. Nº enreg. 1641
Température de stockage ou defonctionnement. Nº enreg. 0632

Plage d’humidité de stockage.
Nº enreg. 2620

Directive du Conseil européen 93/42/CEE relative aux
dispositifs médicaux
Représentant agréé au sein de l’Union
européenne
Si l’étiquette de l’identifiant uniquedu produit (UDI) du produit porte
le symbole CE ####, l’appareil est
conforme aux exigences de la directive
93/42 / CEE concernant les dispositifs
médicaux. Le symbole CE #### indiquele numéro de l’organisme notifié.
CEI 60417
Équipement de classe II à double
isolation Reg. # 5172

Limite de température atmosphérique.
Nº enreg. 2621

Interrupteur. Nº enreg. 5009

Nom et adresse du fabricant.
Nº enreg. 3082
Date de fabrication. Nº enreg. 2497

Certification ETL
Agence de sécurité pour CAN/CSA
C22.2 no 601.1 M90 pour les appareilsélectriques médicaux

Référence catalogue. Nº enreg. 2493
Numéro de série. Nº enreg. 2498

Symboles internes

Haut. Nº enreg. 0623

Tenez l’unité éloignée des matériaux,des huiles et des graisses inflammables.

Fragile, à manipuler avec soin.
Nº enreg. 0621

Ne pas démonter.

Maintenez l’unité à l’abri de l’humidité.
Nº enreg. 0626

Augmenter le débit.

Limite de cumul en fonction du numéro.
Nº enreg. 2403

Diminuer le débit.

ISO 7010 : symboles graphiques – couleurs desécurité et étiquettes de sécurité – étiquettes desécurité enregistrées
Le manuel d’instruction doit être lu.
Nº enreg. M002
Tenez l’unité à l’écart des flammesnues, du feu, des étincelles. Sourced’incendie ouverte interdite etinterdiction de fumer. Nº enreg. P003
Ne fumez pas à proximité de l’appareilou pendant son fonctionnement.
Nº enreg. P002
Pièce appliquée de type BF (degré
de protection contre les déchargesélectriques). Nº enreg. 5333
Avertissement. Nº enreg. W001

Bouton Menu.
RTCA/
DO-160Section 21
Conforme
à lacatégorie M

Conforme à la catégorie M de lasection 21 de la norme RTCA DO-160.
Exigence SFAR 106 de la FAA
Équipement de classe II à double
isolation
Approuvé par la FAA – COP

21 CFR 801.15 : code de la réglementation fédérale,chapitre 21
La législation fédérale restreint la ventede cet appareil par un médecin ou sur
ordonnance de celui-ci.
PN MN232-C4 K | Manuel d’utilisation

FRE - 33

FreeStyle Comfort
Directive du Conseil européen 2012/19/UE : déchetsd’équipements électriques et électroniques (DEEE)
DEEE

CEI 60601-1 : Appareil électro-médical – Partie 1 –Règles générales de sécurité de base etperformances essentielles
Protection contre les gouttes d’eau
Ce produit peut être protégé par un ou plusieurs
brevets américains ou internationaux. Consultez notresite Web ci-dessous pour voir la liste des brevets.
Brevet : www.caireinc.com/corporate/patents/.

Qu’est-ce que le concentrateurd’oxygène ?
L’air que nous respirons contient environ 21 %d’oxygène, 78 % d’azote et 1 % d’autres gaz. Dansle FreeStyle® Comfort®, l’air ambiant de la pièce est
aspiré dans la machine par les prises d’air. Il passe
ensuite au travers d’un matériau absorbant appelé tamis
moléculaire. Ce matériau sépare l’oxygène de l’azote
et capture ce dernier pour laisser uniquement passerl’oxygène. Il en résulte un débit d’une forte concentrationd’oxygène transmis à l’utilisateur.
Le FreeStyle Comfort combine des technologies
avancées de concentration d’oxygène et de conservation
de l’oxygène pour créer un appareil portable qui permet
une portabilité et une déambulation maximales. Grâceà un débit pulsé optimal, l’appareil détecte rapidement
les respirations de l’utilisateur afin de lui fournir del’oxygène dès le début de chaque inhalation. Ce typede débit se révèle parfaitement adapté pour les activités
et les séjours à l’extérieur. Contrairement à un débit
continu, il conserve l’oxygène et augmente la durée devie de la batterie. Le FreeStyle Comfort est suffisamment
léger pour être porté en bandoulière. Il fonctionne sur
courant CA (mur), CC (véhicule) ou sur batterie.

Utilisation conforme
Le concentrateur d’oxygène CAIRE FreeStyle
Comfort est destiné à l’administration d’oxygène
supplémentaire. L’appareil n’est pas conçu pourle maintien en vie et ne fournit aucune capacité desurveillance des patients.

Pourquoi votre médecin vous prescritil de l’oxygène ?
Un grand nombre d’individus souffrent de différentes
maladies cardiaques, pulmonaires ou encore
respiratoires. La majorité d’entre eux peuvent bénéficier
d’une oxygénothérapie complémentaire à domicile,lorsqu’ils voyagent, ou même lorsqu’ils participent à desactivités à l’extérieur.
L’oxygène est un gaz qui constitue 21 % de l’air
ambiant que nous respirons et notre corps a besoin

34 - FREPN MN232-C4 K | Manuel d’utilisationd’un approvisionnement régulier pour fonctionnercorrectement. Votre médecin vous a recommandé undébit ou un réglage à définir afin de traiter votre affection
respiratoire.
Bien que l’oxygène ne constitue en aucun cas une drogue
provoquant un état de dépendance, une oxygénothérapie
non autorisée peut s’avérer dangereuse. N’oubliez pasde demander conseil à votre médecin avant d’utiliser ceconcentrateur d’oxygène. Le fournisseur de cet appareil
vous montrera comment régler le débit prescrit.

Profil de l’utilisateur
Les concentrateurs visent à fournir un apport d’oxygène
supplémentaire aux utilisateurs pris d’un malaise
découlant d’une maladie qui réduit la capacité despoumons à distribuer l’oxygène contenu dans l’air dansla circulation sanguine. Les concentrateurs d’oxygène
portables (COP) ne contiennent pas d’oxygène. Ils n’ont
pas besoin d’un nouveau remplissage et peuvent êtrerechargés partout où une alimentation CA ou CC est
disponible (sauf dans un avion). Ainsi, l’utilisateur peutfacilement l’utiliser à son domicile et le transporter (àl’intérieur comme à l’extérieur de son domicile), pour
un confort de vie amélioré. L’usage d’un concentrateur
requiert une prescription médicale et n’est pas destiné àdes fins de réanimation.
Bien qu’une oxygénothérapie puisse être prescrite aux
utilisateurs de tout âge, les patients sont généralementâgés de plus de 65 ans et souffrent de différentes
pathologies respiratoires, notamment la maladie
pulmonaire obstructive chronique (MPOC). En règle
générale, les utilisateurs sont dotés de capacités
intellectuelles suffisantes pour exprimer leur malaise.
Dans le cas contraire, ou lorsque l’utilisateur est
incapable de lire et de comprendre les étiquettes apposées
sur le concentrateur ou encore le mode d’emploi, leconcentrateur peut uniquement être utilisé sous la
supervision d’une personne compétente. Si une gêneest ressentie pendant l’utilisation du concentrateur, il est
conseillé aux utilisateurs de contacter leur prestatairede soins. Il est également recommandé aux utilisateursd’être équipés d’un apport d’oxygène de secours, telqu’une bouteille d’oxygène, dans le cas d’une coupureélectrique ou d’une panne du concentrateur. L’utilisation
du concentrateur n’exige aucune compétence ni
connaissance particulière.

Déballage de votre FreeStyle Comfort
Vérifiez que tous lescomposants listés sont inclus
dans l’emballage. En l’absenced’un composant, contactez
immédiatement votre fournisseurd’oxygène.

Accessoires standard• Concentrateur d’oxygène portable• Sac de transport• Bandoulière• Bloc-batterie avec 8 cellules• Alimentation CA• Cordon d’alimentation CA• Alimentation CC• Manuel d’utilisation

FreeStyle Comfort
Accessoires en option• Pack supplémentaire de batterie
8 cellules• Bloc-batterie avec 16 cellules• Chargeur de bureau• Sac de transport pour lesaccessoires

Présentation de votre FreeStyle Comfort
Familiarisez-vous avec les principales caractéristiques du FreeStyle Comfort et son interface utilisateur.
Fixation de labandoulière

Vue avantInterfaceutilisateur

Sortied’oxygène
Sortie d’air

Prise d’air/Filtres
Prise d’air/Filtres

Languettepour prisede portd’alimentation
Batterie
Bouton de déblocage
de la batterie
PN MN232-C4 K | Manuel d’utilisation

FRE - 35

FreeStyle Comfort
Vue de dessus
Écran d’affichage
Augmentationdu débit

Touche Marche/ArrêtBouton Menu

Diminutiondu débit

Écran d’affichage sur batterie
Réglage du débit

Indicateur debatterie

Indicateur autoSAT

Écran d’alarme

Pourcentage
de la durée devie restante dela batterie

Indicateurd’administration
du bolus

Écran d’affichage sur alimentation CA/CC

Réglage du débit
Indicateur debatterie

Zone du voyant
d’alarme

36 - FREPN MN232-C4 K | Manuel d’utilisation

Indicateur d’étatde charge

FreeStyle Comfort
Important !
Les instructions de sécurité se définissent
comme indiqué ci-dessous :AVERTISSEMENT : INFORMATIONSIMPORTANTES DE SÉCURITÉ RELATIVESAUX DANGERS POUVANT CAUSER DESBLESSURES GRAVES.
MISE EN GARDE : informations
importantes pour éviter d’endommager le
FreeStyle Comfort.
Remarque : informations exigeant une attention particulière.

Indications d’utilisation
AVERTISSEMENT : IL EST TRÈSIMPORTANT DE SÉLECTIONNER LE DÉBITD’OXYGÈNE QUI VOUS A ÉTÉ PRESCRIT.
NE MODIFIEZ PAS LE RÉGLAGE DE DÉBITSANS AVIS MÉDICAL.
AVERTISSEMENT : CONFORMÉMENT À LA LOIFÉDÉRALE DES ÉTATS-UNIS, LA VENTE ET LA
LOCATION DE CET APPAREIL DOIVENT SE FAIRESUR ORDONNANCE MÉDICALE OU PAR UN
PRESTATAIRE DE SOINS MUNI D’UNE LICENCE.
AVERTISSEMENT : CET APPAREIL NE DOIT PAS ÊTREUTILISÉ À DES FINS DE RÉANIMATION. IL PEUT ÊTRENÉCESSAIRE DE SURVEILLER LES UTILISATEURSDES SERVICES GÉRIATRIQUES, PÉDIATRIQUESOU TOUT AUTRE TYPE D’UTILISATEUR N’ÉTANTPAS EN MESURE DE COMMUNIQUER EN CAS DE
MALAISE LORS DE L’UTILISATION DE L’APPAREIL.
LES UTILISATEURS SOUFFRANT DE DÉFICIENCE
AUDITIVE ET/OU VISUELLE PEUVENT AVOIR BESOIN
D’UNE ASSISTANCE POUR SURVEILLER LESALARMES.
AVERTISSEMENT : LES RÉGLAGES DU MODE
D’ADMINISTRATION PULSÉ DOIVENT ÊTREDÉTERMINÉS EN FONCTION DES BESOINSINDIVIDUELS DE CHAQUE UTILISATEUR AU REPOS,
DURANT UN EFFORT ET PENDANT LES VOYAGES.
AVERTISSEMENT : CERTAINS EFFORTSRESPIRATOIRES DU PATIENT PEUVENT NE PAS
DÉCLENCHER LA FONCTION DE CONSERVATION.
AVERTISSEMENT : LE FREESTYLE COMFORT N’ESTPAS DESTINÉ À ÊTRE UTILISÉ AVEC UN PATIENTTRACHÉOTOMISÉ.
AVERTISSEMENT : LES FEMMES ENCEINTES OUQUI ALLAITENT NE DOIVENT PAS UTILISER LESACCESSOIRES RECOMMANDÉS DANS CE MANUEL,CEUX-CI POUVANT CONTENIR DES PHTALATES.

Contre-indications d’utilisation
AVERTISSEMENT : DANS CERTAINS CAS,L’UTILISATION D’OXYGÈNE SANS AVISMÉDICAL PEUT S’AVÉRER DANGEREUSE.
CET APPAREIL DOIT ÊTRE UTILISÉUNIQUEMENT SUR PRESCRIPTIONMÉDICALE.
AVERTISSEMENT : NE PAS UTILISER CET APPAREIL
EN PRÉSENCE D’ANESTHÉSIQUES INFLAMMABLES.
AVERTISSEMENT : COMME TOUT DISPOSITIF
ÉLECTRIQUE, CET APPAREIL PEUT NE PAS
FONCTIONNER EN RAISON D’UNE COUPURE DE
COURANT OU NÉCESSITER D’ÊTRE RÉPARÉ PAR
UN TECHNICIEN QUALIFIÉ. LE CONCENTRATEURD’OXYGÈNE NE PEUT ÊTRE UTILISÉ POUR LESUTILISATEURS QUI PRÉSENTERAIENT DES
RÉACTIONS INDÉSIRABLES DÉCOULANT D’UNE
INTERRUPTION TEMPORAIRE DE L’APPAREIL.

Consignes de sécurité
AVERTISSEMENT : AUCUNEMODIFICATION DE CET ÉQUIPEMENTN’EST AUTORISÉEAVERTISSEMENT : CET APPAREIL DISPENSE UN
OXYGÈNE EXTRÊMEMENT PUR, SUSCEPTIBLE
DE S’ENFLAMMER TRÈS VITE. NE FUMEZ PAS
À PROXIMITÉ ET ÉVITEZ TOUTE FLAMME NUEDANS LA MÊME PIÈCE QUE (1) CET APPAREIL,OU QUE (2) TOUT AUTRE ACCESSOIRE DE
TRANSPORT D’OXYGÈNE. LE NON-RESPECTDE CET AVERTISSEMENT PEUT AVOIR POURCONSÉQUENCES UN INCENDIE, DES DÉGÂTS
MATÉRIELS ET/OU DES BLESSURES GRAVES VOIREMORTELLES.
AVERTISSEMENT : N’UTILISEZ PAS CET APPAREIL
DANS UN ESPACE FERMÉ OU CONFINÉ (PAR
EXEMPLE DANS UNE PETITE VALISE OU UN SAC)
OFFRANT UNE VENTILATION POTENTIELLEMENTINSUFFISANTE. CECI POURRAIT PROVOQUER UNE
SURCHAUFFE DU CONCENTRATEUR D’OXYGÈNE ET
UNE DÉGRADATION DE SES PERFORMANCES.
AVERTISSEMENT : LE CONCENTRATEUR D’OXYGÈNE
DOIT ÊTRE PLACÉ À L’ÉCART DE TOUTE SOURCE DE
FUMÉE, DE POLLUTION OU DE VAPEURS.
AVERTISSEMENT: EN CAS D’INCIDENT GRAVESURVENANT AVEC CET APPAREIL, L’UTILISATEURDOIT IMMÉDIATEMENT SIGNALER L’INCIDENT AU
FOURNISSEUR ET / OU AU FABRICANT. UN INCIDENTGRAVE EST DÉFINI COMME UNE BLESSURE,
LA MORT OU LA POSSIBILITÉ DE CAUSER DESBLESSURES OU LA MORT EN CAS DE RÉAPPARITION
DE L’INCIDENT. L’UTILISATEUR PEUT ÉGALEMENTSIGNALER L’INCIDENT À L’AUTORITÉ COMPÉTENTEDU PAYS OÙ L’INCIDENT S’EST PRODUIT.

PN MN232-C4 K | Manuel d’utilisation

FRE - 37

FreeStyle Comfort
AVERTISSEMENT : LE RECOURS À DES
ACCESSOIRES D’ADMINISTRATIOND’OXYGÈNE NON PRÉVUS POUR UNE
UTILISATION AVEC CE CONCENTRATEURD’OXYGÈNE EST SUSCEPTIBLE DE NUIREAUX PERFORMANCES DE L’APPAREIL.
LES RÉFÉRENCES DES ACCESSOIRESRECOMMANDÉS SONT INDIQUÉES DANSCE MANUEL.
AVERTISSEMENT : EN CAS DE CHUTE,
D’ENDOMMAGEMENT OU D’EXPOSITION DU
CONCENTRATEUR D’OXYGÈNE À DE L’EAU,CONTACTEZ VOTRE ORGANISME DE ÉQUIPEMENT
AFIN DE PROCÉDER À L’INSPECTION OU AUXRÉPARATIONS POSSIBLES DE L’APPAREIL. VEILLEZ À
NE PAS UTILISER LE CONCENTRATEUR D’OXYGÈNE
SI SA FICHE OU SON CORDON D’ALIMENTATION
ÉLECTRIQUE EST ENDOMMAGÉ(E).
AVERTISSEMENT : VEILLEZ À ATTACHERCORRECTEMENT LE CONCENTRATEUR D’OXYGÈNE
À L’AIDE D’UNE CEINTURE OU PAR TOUT AUTRE
MOYEN APPROPRIÉ PENDANT LE TRANSPORT DANSUN VÉHICULE AFIN D’ÉVITER TOUTE BLESSURE OU
DÉTÉRIORATION.
AVERTISSEMENT : LES RÉGLEMENTATIONS DU
DÉPARTEMENT DES TRANSPORTS DES ÉTATSUNIS ET DES NATIONS UNIES (ONU) EXIGENT QUE
LA BATTERIE SOIT RETIRÉE DE L’APPAREIL SUR
TOUS LES VOLS INTERNATIONAUX SI CELUI-CI ESTENREGISTRÉ EN TANT QUE BAGAGE. LORS DE
L’EXPÉDITION DU CONCENTRATEUR D’OXYGÈNE,IL EST ÉGALEMENT IMPÉRATIF DE RETIRER LABATTERIE DE L’APPAREIL ET DE L’EMBALLER
CORRECTEMENT.
AVERTISSEMENT : LISEZ ET ASSUREZ-VOUS D’AVOIRCOMPRIS LES INFORMATIONS DE CE MANUEL AVANT
D’UTILISER CET APPAREIL. SI VOUS NE COMPRENEZ
PAS LES AVERTISSEMENTS ET LES INSTRUCTIONS,CONTACTEZ VOTRE FOURNISSEUR AGRÉÉ AVANT
D’ESSAYER D’UTILISER L’APPAREIL AFIN D’ÉVITER
TOUT RISQUE DE BLESSURE OU DE DÉGÂTS
MATÉRIELS.
AVERTISSEMENT : SI VOUS VOUS SENTEZ MAL
À L’AISE OU EN CAS D’URGENCE MÉDICALE,SOLLICITEZ IMMÉDIATEMENT UNE ASSISTANCEMÉDICALE.
AVERTISSEMENT : L’UTILISATION DU
CONCENTRATEUR D’OXYGÈNE À UNE TEMPÉRATURENON CONFORME AUX CARACTÉRISTIQUESTECHNIQUES PEUT LIMITER SA CAPACITÉ ÀDÉLIVRER LA CONCENTRATION EN OXYGÈNESPÉCIFIÉE. REPORTEZ-VOUS À LA SECTIONCARACTÉRISTIQUES TECHNIQUES DE CE MANUELPOUR CONSULTER LES LIMITES DE TEMPÉRATURE.
AVERTISSEMENT : VEILLEZ À NE PAS UTILISERD’HUILE, DE GRAISSE, DE PRODUITS À BASE
DE PÉTROLE OU D’AUTRES SUBSTANCESINFLAMMABLES AVEC LES ACCESSOIRESDE TRANSPORT D’OXYGÈNE OU AVEC LE
CONCENTRATEUR D’OXYGÈNE. SEULES DESLOTIONS ET DES CRÈMES À BASE D’EAU ET
COMPATIBLES AVEC LA PRÉSENCE D’OXYGÈNE
DOIVENT ÊTRE UTILISÉES. L’OXYGÈNE ACCÉLÈRE LACOMBUSTION DES SUBSTANCES INFLAMMABLES.

38 - FREPN MN232-C4 K | Manuel d’utilisation

AVERTISSEMENT : NE LAISSEZ PAS LES ORIFICES
D’ADMISSION ET D’ÉVACUATION D’AIR S’OBSTRUER.
NE FAITES PAS TOMBER L’APPAREIL OU N’INSÉREZAUCUN OBJET DANS SES OUVERTURES. CECIPOURRAIT PROVOQUER UNE SURCHAUFFE
DU CONCENTRATEUR D’OXYGÈNE ET UNE
DÉGRADATION DE SES PERFORMANCES.
AVERTISSEMENT : EN CAS D’UTILISATIONINCORRECTE DE LA BATTERIE, CELLE-CI PEUT
SURCHAUFFER, PRENDRE FEU ET PROVOQUER DESBLESSURES GRAVES. VEILLEZ À NE JAMAIS PERCER,
COGNER OU FAIRE TOMBER LA BATTERIE, MARCHERDESSUS OU L’EXPOSER À DES IMPACTS OU DES
CHOCS VIOLENTS. L’UTILISATION D’UNE BATTERIEENDOMMAGÉE PEUT PROVOQUER DES BLESSURES.
AVERTISSEMENT : VEILLEZ À NE PAS COURTCIRCUITER LES CONTACTS EN MÉTAL DE LA
BATTERIE AVEC DES OBJETS MÉTALLIQUES, TELSQU’UNE CLÉ OU UNE PIÈCE DE MONNAIE, CECIPOURRAIT PROVOQUER DES ÉTINCELLES OU LAGÉNÉRATION DE CHALEUR EXCESSIVE.
AVERTISSEMENT : SI VOUS UTILISEZ LE
CONCENTRATEUR D’OXYGÈNE PORTABLE À
L’EXTÉRIEUR À L’AIDE D’UNE ALIMENTATION CA,
BRANCHEZ L’ALIMENTATION DANS UNE PRISE AVECDISJONCTEUR DE FUITE DE TERRE UNIQUEMENT.
AVERTISSEMENT : LE CONCENTRATEUR D’OXYGÈNE
NE DOIT PAS ÊTRE UTILISÉ LORSQU’IL EST ACCOLÉÀ, OU EMPILÉ AVEC D’AUTRES ÉQUIPEMENTS. SI UN
TEL POSITIONNEMENT NE PEUT PAS ÊTRE ÉVITÉ,IL CONVIENT DE SURVEILLER L’APPAREIL POURS’ASSURER QU’IL FONCTIONNE CORRECTEMENT.
AVERTISSEMENT : L’UTILISATION DE CÂBLES ET
D’ADAPTATEURS AUTRES QUE CEUX SPÉCIFIÉS,
À L’EXCEPTION DES CÂBLES ET ADAPTATEURSCOMMERCIALISÉS PAR LE FABRICANT DE
L’ÉQUIPEMENT ÉLECTRO-MÉDICAL POUR LE
REMPLACEMENT DE COMPOSANTS INTERNES,
RISQUE D’ENTRAÎNER UNE AUGMENTATION DESÉMISSIONS ÉLECTROMAGNÉTIQUES OU UNEDIMINUTION DE L’IMMUNITÉ ÉLECTROMAGNÉTIQUEDU CONCENTRATEUR D’OXYGÈNE.
AVERTISSEMENT : UTILISEZ L’APPAREIL UNIQUEMENTÀ LA TENSION ÉLECTRIQUE INDIQUÉE SURL’ÉTIQUETTE DES CARACTÉRISTIQUES TECHNIQUESAPPOSÉE SUR L’APPAREIL.
AVERTISSEMENT : N’UTILISEZ PAS DE RALLONGE
ÉLECTRIQUE AVEC CET APPAREIL ET VEILLEZ
À NE PAS BRANCHER UN TROP GRANDNOMBRE D’APPAREILS SUR LA MÊME PRISESECTEUR. L’UTILISATION D’UNE RALLONGE
RISQUE D’AFFECTER LES PERFORMANCES DE
L’APPAREIL. DE PLUS, SI VOUS EFFECTUEZ TROP
DE BRANCHEMENTS SUR UNE MÊME PRISE, VOUS
RISQUEZ DE PROVOQUER UNE SURCHARGE DUTABLEAU ÉLECTRIQUE ET DE DÉCLENCHER LEDISJONCTEUR OU DE PROVOQUER UN INCENDIE SI
CELUI-CI NE FONCTIONNE PAS.

FreeStyle Comfort
MISE EN GARDE : si une alarme sedéclenche ou si vous constatez que leconcentrateur d’oxygène ne fonctionne
pas correctement, reportez-vous au
Tableau de dépannage figurant dansce manuel. Si vous ne parvenez pas àrésoudre le problème, consultez votre
fournisseur agréé.
MISE EN GARDE : suivez toutes les instructions dufabricant afin d’éviter l’annulation de la garantie.
AVERTISSEMENT : NE TENTEZ PAS
DE RÉALISER DES OPÉRATIONS DE
MAINTENANCE AUTRES QUE LESSOLUTIONS DE DÉPANNAGE DÉTAILLÉESDANS CE MANUEL. NE RETIREZ PAS LESCAPOTS. SEUL LE FOURNISSEUR DE
L’APPAREIL OU UN TECHNICIEN QUALIFIÉ
EST HABILITÉ À RETIRER LES CAPOTSOU RÉPARER L’APPAREIL.
AVERTISSEMENT : UTILISEZ EXCLUSIVEMENT LESACCESSOIRES RECOMMANDÉS PAR LE FABRICANT.
L’UTILISATION D’AUTRES ACCESSOIRES POURRAITS’AVÉRER DANGEREUSE ET ENDOMMAGERGRAVEMENT VOTRE CONCENTRATEUR D’OXYGÈNE,ET PROVOQUERAIT L’ANNULATION DE LAGARANTIE.
AVERTISSEMENT : LE FABRICANT RECOMMANDE DEDISPOSER D’UNE AUTRE SOURCE D’ALIMENTATIONEN OXYGÈNE EN CAS DE COUPURE DE COURANT,DE DÉCLENCHEMENT D’ALARME DE L’APPAREIL
OU DE PANNE MÉCANIQUE. CONSULTEZ VOTRE
MÉDECIN OU VOTRE FOURNISSEUR AGRÉÉ AUSUJET DU TYPE DE SYSTÈME DE RÉSERVE À
UTILISER.
AVERTISSEMENT : NE LAISSEZ PAS LESORIFICES D’ADMISSION ET D’ÉVACUATION DE
L’AIR S’OBSTRUER. NE FAITES PAS TOMBERL’APPAREIL OU N’INSÉREZ AUCUN OBJET DANS SESOUVERTURES. CECI POURRAIT PROVOQUER UNE
SURCHAUFFE DU CONCENTRATEUR D’OXYGÈNE ET
UNE DÉGRADATION DE SES PERFORMANCES.
AVERTISSEMENT : FUMER EN UTILISANT DEL’OXYGÈNE CONSTITUE LA PREMIÈRE CAUSE DE
BLESSURES ET DE DÉCÈS LIÉS AUX INCENDIES.
VOUS DEVEZ IMPÉRATIVEMENT RESPECTER LESAVERTISSEMENTS DE SÉCURITÉ SUIVANTS :AVERTISSEMENT : NE LAISSEZ PERSONNE FUMERET N’UTILISEZ PAS DE BOUGIES OU DE FLAMMESNUES DANS LA MÊME PIÈCE QUE L’APPAREILOU QUE LES ACCESSOIRES DE TRANSPORTD’OXYGÈNE.
AVERTISSEMENT : FUMER LORS DU PORT D’UNE
CANULE À OXYGÈNE PEUT PROVOQUER DESBRÛLURES AU VISAGE OU MÊME LA MORT.
AVERTISSEMENT : SI VOUS FUMEZ, CES 3 GESTESPEUVENT VOUS SAUVER LA VIE : ARRÊTEZ LE
CONCENTRATEUR D’OXYGÈNE, RETIREZ LA CANULEET QUITTEZ LA PIÈCE DANS LAQUELLE SE TROUVEL’APPAREIL.

AVERTISSEMENT : DES INDICATIONS« INTERDICTION DE FUMER - UTILISATIOND’OXYGÈNE » DOIVENT ÊTRE AFFICHÉES DEFAÇON BIEN VISIBLE À VOTRE DOMICILE OU DANSTOUT LIEU OÙ DE L’OXYGÈNE EST UTILISÉ. LESUTILISATEURS ET LE PERSONNEL SOIGNANT
DOIVENT ÊTRE INFORMÉS DES DANGERS LIÉSAU FAIT DE FUMER EN PRÉSENCE OU PENDANTL’UTILISATION D’OXYGÈNE MÉDICAL.
AVERTISSEMENT : N’UTILISEZ PAS VOTRE
CONCENTRATEUR D’OXYGÈNE EN PRÉSENCE DE
GAZ INFLAMMABLES. CELA PEUT PROVOQUERUN EMBRASEMENT RAPIDE SUSCEPTIBLE
D’ENTRAÎNER DES DÉGÂTS MATÉRIELS, DESBLESSURES OU LA MORT.
AVERTISSEMENT : LE FAIT DE RETIRER LA CANULEET DE LA PLACER SUR UN VÊTEMENT, UN DRAP,
UN SOFA OU TOUTE AUTRE MATIÈRE TEXTILE PEUTPROVOQUER UN EMBRASEMENT INSTANTANÉEN CAS D’EXPOSITION À UNE CIGARETTE, À UNE
SOURCE DE CHALEUR OU À UNE FLAMME.
AVERTISSEMENT : NE LAISSEZ JAMAIS UNE CANULE
NASALE SUR OU SOUS UN VÊTEMENT, UN DRAPOU UN COUSSIN. SI L’APPAREIL EST ALLUMÉSANS ÊTRE UTILISÉ, CES MATIÈRES RISQUENTDE S’ENFLAMMER AU CONTACT DE L’OXYGÈNE.
VEILLEZ À ÉTEINDRE LE CONCENTRATEURD’OXYGÈNE LORSQU’IL N’EST PAS UTILISÉ.
AVERTISSEMENT : LE CONCENTRATEURD’OXYGÈNE DOIT ÊTRE ÉTEINT ET DÉBRANCHÉ DEL’ALIMENTATION EXTÉRIEURE AVANT D’ENLEVER LE
MODULE DE LA BATTERIE.
AVERTISSEMENT : LES PETITES PIÈCESPEUVENT PRÉSENTER UN RISQUEPOTENTIEL D’ÉTOUFFEMENT.
AVERTISSEMENT : ÉVITEZ TOUT CONTACTAVEC L’ÉVACUATION D’AIR PENDANT PLUS DE
10 SECONDES, CAR CELA PEUT ÊTRE CHAUD.
AVERTISSEMENT : LE VENT OU LESFORTS COURANTS D’AIR PEUVENTNUIRE À L’ADMINISTRATION PRÉCISE DEL’OXYGÉNOTHÉRAPIE.
AVERTISSEMENT : CE PRODUIT PEUT VOUSEXPOSER À DES PRODUITS CHIMIQUES, Y COMPRISLE NICKEL, QUI EST RECONNU PAR L’ÉTAT DE
CALIFORNIE COMME ÉTANT CANCÉRIGÈNE. POUR
PLUS D’INFORMATIONS, VEUILLEZ VISITER LE SITE
WWW.P65WARNINGS.CA.GOV.
MISE EN GARDE : débranchez toujours
l’alimentation CA de la prise secteur murale
avant de déconnecter l’alimentation CA duconcentrateur d’oxygène.
MISE EN GARDE : ne placez pas le concentrateurd’oxygène de façon à ce qu’il soit difficile d’accéder au
cordon d’alimentation.

PN MN232-C4 K | Manuel d’utilisation

FRE - 39

FreeStyle Comfort
MISE EN GARDE : placez toujours les cordonsd’alimentation et la tubulure d’alimentation en oxygènede sorte à empêcher toute personne de trébucher ou
de s’étrangler accidentellement.
MISE EN GARDE : lorsque le concentrateurd’oxygène est utilisé dans une voiture, sur un bateau
ou sur d’autres sources de courant continu avec
l’alimentation CC, assurez-vous que le véhicule
est en marche avant de connecter l’appareil. Si ces
instructions ne sont pas respectées, le concentrateurd’oxygène risque de ne pas être alimenté.
MISE EN GARDE : lorsque le moteur du véhicule danslequel vous utilisez le concentrateur d’oxygène estarrêté, déconnectez l’appareil et retirez-le du véhicule.
Ne stockez pas le concentrateur d’oxygène dans unvéhicule dont l’habitacle peut devenir très chaudou très froid, ou dans tout environnement similaire
pouvant présenter des températures particulièrement
faibles ou élevées.
Remarque : les équipements portables et mobiles decommunication par radiofréquences peuvent affecter lefonctionnement des équipements médicaux électriques.
MISE EN GARDE : VEILLEZ À NE PAS
laisser le concentrateur d’oxygène ou
l’alimentation connecté(e) dans le véhicule
si le moteur est arrêté, afin de ne pas
décharger la batterie du véhicule.
MISE EN GARDE : l’utilisation ou le stockage duconcentrateur d’oxygène en dehors de sa plage detempérature de fonctionnement normale et de laplage de pression ambiante (altitude) peut affecter lesperformances et réduire le temps de fonctionnement
de la batterie et/ou augmenter sa durée de chargement.
Reportez-vous à la section Caractéristiques
techniques de ce manuel pour consulter leslimites de température relatives au stockage et au
fonctionnement de l’appareil.
MISE EN GARDE : pour les concentrateurs d’oxygèneéquipés de batteries : afin de préserver la longévité devotre batterie, stockez-la dans un endroit frais et sec.
Le stockage du concentrateur d’oxygène pendant delongues périodes à de hautes températures, avec une
batterie totalement chargée ou entièrement épuisée,peut écourter la durée de vie de celle-ci. Ne tentez pas
d’ouvrir la batterie. Elle ne contient aucune pièce à
entretenir ou à remplacer. Veillez à tenir la batterie horsde portée des enfants.
MISE EN GARDE : UTILISEZ UNIQUEMENT lesbatteries fournies par le fabricant. Contactez lefournisseur de l’équipement ou votre mairie pourconnaître les obligations en matière de recyclage desbatteries.

40 - FREPN MN232-C4 K | Manuel d’utilisation

Remarque : vérifiez que la canule est entièrement
insérée et convenablement positionnée. afin que leconcentrateur d’oxygène puisse correctement détecter
chaque inspiration pour délivrer l’oxygène requis. Durant
l’inhalation, vous devez entendre ou sentir l’oxygène
circuler dans les sondes de la canule nasale.
Remarque : Les batteries au lithium peuvent neplus fonctionner lorsqu’elles sont exposées à destempératures extrêmement élevées alors qu’elles sonttotalement chargées ou entièrement déchargées. Pourun stockage prolongé, il est recommandé de charger
les batteries entre 25 et 50 % et de les conserver à une
température de -20 °C à 20 °C (-4 °F à 68 °F).
Remarque : le recours à des accessoires d’administrationd’oxygène non prévus pour une utilisation avec ceconcentrateur d’oxygène est susceptible de nuire
aux performances de l’appareil. Les références desaccessoires recommandés sont indiquées dans ce
manuel.
Remarque : n’utilisez pas le concentrateur d’oxygène
lorsque le filtre d’admission d’air n’est pas en place. Si un
second filtre est fourni, insérez le filtre de remplacement
avant de nettoyer le filtre sale. Nettoyez le filtre saledans de l’eau chaude savonneuse, puis laissez-le sécher
totalement avant de l’utiliser.
Remarque : le fabricant ne recommande pas de stérilisercet équipement.
Remarque : l’utilisation du concentrateur d’oxygène nerisque en aucun cas d’épuiser tout l’oxygène de la pièce.
AVERTISSEMENT : LE CONCENTRATEURD’OXYGÈNE NE DOIT PAS ÊTRE UTILISÉOU BRANCHÉ DANS UNE ALIMENTATIONEXTERNE (AC/DC) SANS QU’AUCUNEBATTERIE NE SOIT INSTALLÉE. NE
TENTEZ PAS DE METTRE EN MARCHE
L’APPAREIL OU DE LE BRANCHER À UNE
ALIMENTATION EXTERNE LORSQUE LA
BATTERIE EST RETIRÉE.
RETIREZ UNIQUEMENT LA BATTERIE LORSQUEL’APPAREIL EST ÉTEINT ET DÉBRANCHÉ DEL’ALIMENTATION. NE TENTEZ PAS DE RETIRER LABATTERIE LORSQUE L’APPAREIL EST EN MARCHEOU CONNECTÉ À L’ALIMENTATION.
Remarque : si le concentrateur d’oxygène a été stocké
pendant une période prolongée dans un environnement
ne respectant pas les conditions de fonctionnement
normales décrites dans le tableau ci-dessous, il doit
tout d’abord être placé dans un endroit lui permettant
d’atteindre une température de fonctionnement normale
avant de pouvoir être allumé.

FreeStyle Comfort
Caractéristiques techniques
FreeStyle Comfort
Dimensions (avec une seule batterie)

25,4 x 18,5 x 7,9 cm (10.0 x 7.3 x 3.1 in)Dimensions (avec double batterie)27,9 x 18,5 x 7,9 cm (11.0 x 7.3 x 3.1 in)

Poids (avec batterie simple)

2,3 kg (5 lbs)

Poids (avec double batterie)

2,7 kg (6 lbs)

Niveau sonore nominal***39,93 dB (A) ± 0,42 dB (A) @ 2

Réglages du débit

Réglages pulsés 1, 2, 3, 4, 5

Paramètre de débit d’O2

Paramètre 1 - 210 mL/min
Paramètre 2 - 420 mL/min
Paramètre 3 - 630 mL/min
Paramètre 4 - 840 mL/min
Paramètre 5 - 1050 mL/min(à 20 bpm)

Concentration en oxygène*
Moniteur d’oxygène
Pression de sortie maximum
Sensibilité de dosage par défaut

90 % (+5,5 %/-3 %)Oui<30 PSI> -0,1 cm H2O (réglable)

Alimentation CA

100–240 VAC, 50-60Hz, 120W Max

Alimentation CC

11–18 VDC (10 amp max)

Temps de rechargement de la batterie

Une batterie : 3,5 heures
Batterie double : 6,0 heures

Autonomie de la batterie
Température de fonctionnement**
Type de batterie
Capacité de la batterie

Durée de la batterie simple : Paramètre 2 - jusqu’à 4 heures
Durée de la batterie double : Paramètre 2 - jusqu’à 8 heures5 à 40 °C (41 °F à 104 °F)
Lithium Ion
Cellule unique: 6700 mAh; Double cellule: 13400 mAh

Température de transport et destockage (Appareil)-25 °C à 70 °C (-13 °F à +158 °F) jusqu’à 3 048 m (10 000 pieds)

Température de transport et destockage (Batterie)-25 °C à 70 °C (-13 °F à +158 °F) jusqu’à 3 048 m (10 000 pieds)

Altitude de fonctionnement**-381 à 3 048 m (-1 250 à 10 000 pieds) (Testé de 700 à 1 060 hPa)

Humidité de fonctionnement

15 – 95 % d’humidité relative (sans condensation)

Humidité d’entreposage
Garantie limitéede 0% à 90% sans condensation
Appareil 3 ans
1 an pour la batterie et les alimentations* À une pression atmosphérique de 101 kPa (14,7 psi) et une température de 21 °C (70 °F).
** L’unité fonctionnera dans tous les environnements tant que l’environnement est avec les plages de température et d’humidité
spécifiées. Le fonctionnement en dehors de ces caractéristiques techniques peut limiter la capacité du concentrateur à fournir uneconcentration en oxygène adéquate à des débits plus élevés. L’écran de préchauffage peut continuer à s’afficher jusqu’à ce que leconcentrateur ait atteint la concentration cible ou jusqu’à 15 minutes. Continuez d’utiliser l’appareil pendant ce temps.
*** Niveau sonore mesuré par méthode d’essai Nr. 14-1 10/2018 MDS-Hi.
La durée de vie estimée est d’au moins cinq ans. La durée de conservation prévue des tamis moléculaires est d’environ un an si leconcentrateur n’est pas utilisé pendant toute la période.

PN MN232-C4 K | Manuel d’utilisation

FRE - 41

FreeStyle Comfort

Instructions d’utilisation
1. Localisez et placez le FreeStyle Comfort dans un
endroit approprié.
a. Le concentrateur doit être correctement
ventilé, de sorte que les orifices d’admission etd’évacuation d’air ne soient pas obstrués.
b. Le concentrateur doit être placé de façon à ceque tous les indicateurs et alarmes sonores etvisuels puissent être facilement vus et entendus.
c. Veillez à ce que les filtres d’admission d’air
soient placés avant d’utiliser l’appareil. Si un
filtre de l’appareil est manquant, contactez lefournisseur de votre équipement.
2. Placez la batterie dans le concentrateur.
a. Alignez la batterie sur le logement inférieur
depuis le centre du FreeStyle Comfort.

3. Insérez le bloc d’alimentation CA ou CC dans leréceptacle du port d’alimentation sur l’avant, en basà droite de l’appareil.

4. Connectez une canule nasale au raccord de sortie del’oxygène. Veillez à ce que le raccord soit solidement
fixé avant d’utiliser l’appareil.
REMARQUE : L’utilisation de toute canule autre que cellerecommandée peut affecter les performances.

b. Faites glisser la batterie à l’intérieur jusqu’à
entendre le clic confirmant sa bonne position.

42 - FREPN MN232-C4 K | Manuel d’utilisation

Remarque: Pour le fournisseur d’équipement: Lesaccessoires d’administration d’oxygène suivants sontrecommandés pour une utilisation avec le FreeStyle
Comfort:• Canule nasale avec 7 pieds (2,1 m) de tube (LPM
maximum: 6 LPM): référence CAIRE 5408-SEQ• Des informations supplémentaires sur les accessoires
recommandés sont disponibles en ligne sur www.caireinc.
com.

FreeStyle Comfort
5. Allumez le FreeStyle Comfort en appuyant sur lebouton d’alimentation et sélectionnez le réglage dedébit qui vous a été prescrit en utilisant les touches
d’augmentation (+) ou de diminution (-) du réglage
de débit.

Volume d’impulsions autoSAT(mL)*Réglages
Taux derespiration

1

2

3

4

5

15

10,5

21,0

31,5

42,0

52,5

20

10,5

21,0

31,5

42,0

52,5

25

10,5

21,0

31,5

42,0

42,0

30

10,5

21,0

31,5

35,0

35,0

35

10,5

21,0

30,0

30,0

30,0

40

10,5

21,0

26,3

26,3

26,3

*La tolérance est +/- 15 %Volume d’impulsions non autoSAT(mL)*Réglages

Une brève alarme sonore retentira pour indiquer que
l’appareil est sous tension.
Remarque : Lorsque le FreeStyle Comfort est mis en
marche, le paramètre de débit par défaut est 2. Si leréglage est modifié, l’appareil s’allume sur le réglage dedébit précédent.
6. Placez la canule nasale sur votre visage
conformément aux instructions du fabricant fournies
avec la canule, puis respirez normalement. Vous
devriez ressentir une bouffée d’oxygène provenantde la canule nasale à chaque fois que vous inspirez.
7. Pour modifier le réglage du débit, appuyez sur les
touches d’augmentation (+) ou de diminution (-) duréglage de débit pour obtenir le débit souhaité. Il est
normal qu’une différence de son se fasse entendre
lorsque vous modifiez les réglages du débit.

Taux derespiration

1

2

3

4

5

15

14,0

28,0

42,0

56,0

70,0

20

10,5

21,0

31,5

42,0

52,5

25

8,4

16,8

25,2

33,6

42,0

30

7,0

14,0

21,0

28,0

35,0

35

6,0

12,0

18,0

24,0

30,0

40

5,3

10,5

15,8

21,0

26,3

8. Éteignez le FreeStyle Comfort en appuyant sur lebouton d’alimentation.
Remarque : L’écran d’affichage passe en mode veille etdiminue après 20 secondes. Appuyez sur n’importe quel
bouton pour sortir l’écran du mode veille.
9. Pour découvrir les fonctionnalités avancées,consultez les instructions de navigation du menu

PARTICULARITÉS DU PRODUIT
autoSAT®
La fonction exclusive autoSAT de CAIRE permet
au FreeStyle Comfort de fournir un volume constantd’oxygène (bolus) par respiration, jusqu’à une
fréquence respiratoire maximale. À mesure quela fréquence respiratoire de l’utilisateur augmente
pendant des activités comme la marche, la fonction
autoSAT ajuste le fonctionnement de l’appareil pourmaintenir un volume de bolus défini.
UltraSense®
Ce déclencheur de dose pulsée sensible détecte lechangement de pression créé par votre respiration etlibère une dose d’oxygène lorsque vous inspirez.
autoDOSE™
En cas d’alarme de type « Aucun souffle détecté »,cette fonction entraînera la sortie d’oxygène à raison de
20 respirations par minute selon les réglages du débit
actuels jusqu’à ce que la cause de l’alarme soit réglée.
PN MN232-C4 K | Manuel d’utilisation

FRE - 43

FreeStyle Comfort

Alimentations

Le FreeStyle Comfort peut être alimenté de trois
façons :• Bloc-batterie rechargeable• Alimentation CA• Alimentation CC
Les informations relatives au fonctionnement deces trois composants sont reprises dans les sections
suivantes.
AVERTISSEMENT : UTILISEZ UNIQUEMENTLES ALIMENTATIONS DÉCRITES DANS CE
MANUEL D’UTILISATION. L’UTILISATION
DE BATTERIES ET D’ALIMENTATIONSAUTRES QUE CAIRE PEUT ENDOMMAGERL’APPAREIL ET PERTURBER SON
FONCTIONNEMENT.

RETRAIT DE LA BATTERIE

Poussez le bouton de dégagement de la batterie versle bas pour la sortir du FreeStyle Comfort. Tout en
MAINTENANT ce bouton, faites glisser la batteriepour la faire sortir du concentrateur.
AVERTISSEMENT : LE CONCENTRATEURD’OXYGÈNE DOIT ÊTRE ARRÊTÉ ET
DÉBRANCHÉ DE TOUTE SOURCE
D’ALIMENTATION EXTERNE AVANT
DE RETIRER LA BATTERIE.

BATTERIELe FreeStyle Comfort peut fonctionner avec la batterierechargeable fournie avec l’équipement. Aucune limite
de fonctionnement ne s’applique au concentrateur
lorsqu’il fonctionne sur batterie. Tous les débits sontégalement disponibles.

VÉRIFICATION DE LA CHARGE DE
LA BATTERIEINSTALLATION DE LA BATTERIE

Alignez la batterie sur le logement inférieur depuis
le centre du FreeStyle Comfort. Poussez la batterie à
l’intérieur jusqu’à entendre le clic confirmant sa bonne
position.

Pour vérifier la charge de la batterie lors de son
installation dans le FreeStyle Comfort, reportez-vousà l’icône de la batterie et à la durée estimée restanteaffichée sur l’écran.
80–100 %60-80 %

40-60 %

20-40 %

20 %44 - FREPN MN232-C4 K | Manuel d’utilisation

FreeStyle Comfort
Remarque : si la batterie est en cours de chargement, un
indicateur de prise s’affiche à la place du pourcentage decharge restant.
Pour vérifier la charge de la batterie lorsqu’elle n’est
pas installée dans l’appareil FreeStyle Comfort, posezla à plat et appuyez sur le bouton TEST. Situés à droite
du bouton TEST, les témoins lumineux de l’indicateurde la batterie (25-100 %) s’allument pour indiquer leniveau de charge de la batterie.

DURÉE D’AUTONOMIE DE LA BATTERIE

Lorsque l’alimentation CA ou CC externe est
débranchée, le concentrateur bascule automatiquement
sur la batterie si une batterie chargée est installée. Le
tableau ci-après présente l’autonomie généralement
affichée par une nouvelle batterie. L’autonomie d’une
batterie reste constante indépendamment de votre
fréquence respiratoire.
Réglagepulsé

Durée de la batterie
simple (8 cellules)

Durée de la batterie
double (16 cellules)

1jusqu’à 8 heuresjusqu’à 16 heures2

jusqu’à 4 heuresjusqu’à 8heures3

jusqu’à 3 heuresjusqu’à 6 heures4

jusqu’à 2,25 heuresjusqu’à 4,5 heures5

jusqu’à 2 heuresjusqu’à 4 heures

DURÉE TYPIQUE DE RECHARGEMENT DE
LA BATTERIEPREMIÈRE CHARGE

La nouvelle batterie fournie avec votre FreeStyle
Comfort n’est que partiellement chargée lorsqu’elle est
expédiée à sa sortie d’usine. Avant de l’utiliser pour lapremière fois, vous devez la charger complètement àl’aide d’une alimentation CA ou CC.

INFOIRMATIONS GÉNÉRALES DE
CHARGEMENTLe FreeStyle Comfort recharge la batterie à chaquefois qu’elle est installée dans le concentrateur et que leconcentrateur est connecté à une source d’alimentation
CA ou CC externe (sauf dans un avion).

La durée typique pour recharger complètement votre
batterie à partir d’une batterie complètement déchargée
est de 3,5 heures pour une batterie simple et de6 heures pour une batterie double.
Remarque : Il est possible que la batterie ne commencepas à se charger si la batterie est complètement
déchargée.

DURÉE DE VIE DE LA BATTERIE

Les batteries FreeStyle Comfort sont conçues pourdurer 300 cycles de charge/décharge.

La batterie est complètement chargée lorsqu’elle
affiche 100 %.
La batterie peut également être chargée à l’aide duchargeur de bureau fourni en option.

PN MN232-C4 K | Manuel d’utilisation

FRE - 45

FreeStyle Comfort

Alimentation CA et CC

Le FreeStyle Comfort est livré standard avec dessecteurs d’alimentation CA et CC.
L’alimentation CA inclut les composants suivants :a. Transformateur CA qui fonctionne entre
100-240 volts, 50-60Hz.
b. Cordon d’alimentation CA pour laconnexion au transformateur et à la prise
CA correspondante.

L’alimentation CC consiste en un cordon qui se
branche directement à l’appareil et à la prise CC.

ALIMENTATION CA

2. Branchez le cordon d’alimentation CA autransformateur.

L’alimentation CA est destinée à être utilisée au
domicile ou à tout endroit où une prise CA standard
est disponible. Le FreeStyle Comfort fonctionne
parfaitement sur une alimentation CA. Le FreeStyle
Comfort recharge également la batterie chaque fois quel’alimentation CA est disponible (sauf dans un avion).
La batterie se recharge, que l’appareil FreeStyle
Comfort soit allumé ou non.
AVERTISSEMENT : NE BRANCHEZ
PAS L’ALIMENTATION EXTERNE SI LABATTERIE N’EST PAS INSTALLÉE DANSLE CONCENTRATEUR D’OXYGÈNE.
Pour brancher l’appareil FreeStyle Comfort sur une
alimentation CA :1. Branchez le transformateur CA à l’entrée duconnecteur d’alimentation à l’avant, en bas à droite
du FreeStyle Comfort. Veillez à bien l’enfoncer.

46 - FREPN MN232-C4 K | Manuel d’utilisation

3. Raccordez l’extrémité opposée du cordon
d’alimentation CA à une prise murale ou à une priseélectrique appropriée.

FreeStyle Comfort
4. Allumez le FreeStyle Comfort en appuyant sur lebouton d’alimentation et sélectionnez le réglage dedébit qui vous a été prescrit en utilisant les touches
d’augmentation (+) ou de diminution (-) du réglage
de débit.

Remarque : si la batterie doit être rechargée, l’icône de labatterie sur l’écran aura une seule barre orange.

ALIMENTATION CC

L’alimentation CC est destinée à être utilisée dans lesprises pour accessoires installées dans les véhicules,
les bateaux et tout autre véhicule à moteur. Le
FreeStyle Comfort est entièrement fonctionnel pourune utilisation en courant continu (CC). Si nécessaire,
le FreeStyle Comfort permet également de recharger labatterie lorsqu’une alimentation CC est disponible. La
batterie se recharge si l’appareil est éteint, allumé et encours d’utilisation.

3. Raccordez l’extrémité opposée du cordon
d’alimentation CC à un allume-cigare ou à une
prise CC pour accessoire.

4. Allumez le FreeStyle Comfort en appuyant sur lebouton d’alimentation et sélectionnez le réglage dedébit qui vous a été prescrit en utilisant les touches
d’augmentation (+) ou de diminution (-) du réglage
de débit.
Remarque : si la batterie doit être rechargée, l’icône dela batterie sur l’écran aura une seule barre orange.

AVERTISSEMENT : NE BRANCHEZ
PAS L’ALIMENTATION EXTERNE SI LABATTERIE N’EST PAS INSTALLÉE DANSLE CONCENTRATEUR D’OXYGÈNE.
AVERTISSEMENT : NE FAITES PAS FONCTIONNERL’APPAREIL SUR DES PRISES DE COURANT 12 VDC
LORSQUE LE VÉHICULE EST À L’ARRÊT.
1. Démarrez votre véhicule.
2. Branchez l’alimentation CC à l’avant du réceptacled’alimentation, en bas à droite du FreeStyle Comfort.
Veillez à bien l’enfoncer.

PN MN232-C4 K | Manuel d’utilisation

FRE - 47

FreeStyle Comfort
Nettoyage, entretien et maintenancerégulière
MAINTENANCE RÉGULIÈRE

Entretien préventif
L’appareil ne nécessite aucun entretien. Les filtres àl’intérieur de l’appareil sont conçus pour durer toute ladurée de vie de l’appareil. Si l’appareil ne fonctionne
pas correctement, l’appareil doit être renvoyé
au fournisseur pour réparation.. L’entretien descomposants internes du boîtier de l’appareil FreeStyle
Comfort doit être réalisé par votre fournisseurd’oxygène.
L’utilisateur de l’appareil est responsable desprocédures de maintenance suivantes.
• Toutes les semaines – Nettoyer les filtres d’admission
d’air
AVERTISSEMENT : N’ESSAYEZ PAS
D’OUVRIR OU DE RETIRER LES CAPOTSEN PLASTIQUE DU CONCENTRATEURD’OXYGÈNE. L’APPAREIL PEUT
UNIQUEMENT ÊTRE OUVERT PAR UN
TECHNICIEN QUALIFIÉ.

FILTRES D’ADMISSION D’AIR

L’air entre dans l’appareil FreeStyle Comfort par lebiais de deux filtres d’admission d’air situés sur lapartie inférieure avant du concentrateur d’oxygène.
Ces filtres empêchent la poussière et d’autres grandes
particules contenues dans l’air d’entrer dans l’appareil.
Les filtres d’admission d’air doivent être nettoyés au
moins une fois par semaine. En fonction des conditions
d’utilisation, votre fournisseur d’oxygène peut vousconseiller de le nettoyer plus régulièrement. Respectez
ces étapes pour nettoyer correctement les filtresd’admission d’air.
1. Retirez les deux filtres situés à l’avant du FreeStyle
Comfort en appuyant sur l’onglet correspondant et en
tirant vers l’extérieur.

4. Réinstallez les filtres propres et secs.
AVERTISSEMENT : N’UTILISEZ PAS LE
CONCENTRATEUR D’OXYGÈNE LORSQUE
LE FILTRE D’ADMISSION D’AIR N’EST PAS
EN PLACE.

MAINTENANCE DE LA BATTERIE
N’exposez pas la batterie à des liquides.

Ne tentez pas de démonter ou d’ouvrir la batterie.
N’exposez pas la batterie aux flammes nues.
Ne laissez pas tomber la batterie.
Maintenez la batterie hors de portée des enfants.
Si votre batterie semble endommagée, contactez votre
fournisseur d’oxygène avant de l’utiliser.
Ne placez pas la batterie dans un four à micro-ondes
ou tout autre appareil de cuisson ni à proximité de cesderniers.

STOCKAGE DE LA BATTERIE

Il n’est pas nécessaire d’attendre le déchargement
complet de la batterie du concentrateur d’oxygène
avant de la recharger, Il est recommandé de charger
la batterie du concentrateur d’oxygène après chaqueutilisation.
Ne laissez pas la batterie dans un endroit exposé à unechaleur excessive, en plein soleil ou dans une voiture,
par exemple, au risque de dégrader les performances
et la durée de vie de la batterie. La batterie pourraitégalement générer de la chaleur et de la fumée, prendre
feu ou exploser. La batterie doit être conservée dansun environnement stable, caractérisé par une faible
humidité (humidité relative inférieure à 70 %) etl’absence de gaz corrosifs. Afin d’empêcher touteapparition de rouille, évitez les endroits susceptibles decréer de la condensation, notamment où la températureambiante peut rapidement varier.
Pour un stockage d’une période égale ou supérieureà un mois, la batterie doit être totalement déchargéeou chargée entre 50 et 75 % (deux des trois témoins
lumineux). Une fois chargée, retirez la batterie duconcentrateur d’oxygène.

2. Nettoyez les deux filtres dans une eau savonneusetiède.
3. Rincez soigneusement les filtres et essuyez l’excèsd’eau à l’aide d’une serviette douce et absorbante.
Veillez à ce que les filtres soient totalement secs
avant de les replacer.
48 - FREPN MN232-C4 K | Manuel d’utilisation

FreeStyle Comfort
NETTOYAGEArrêtez l’appareil et débranchez-le del’alimentation CA ou CC avant toute opération de
nettoyage ou de ré. nettoyant ménager. Ne vaporisez
PAS de liquides directement sur le boîtier extérieur.
Utilisez une éponge ou un chiffon légèrement humide.
Vaporisez un détergent doux sur l’éponge ou sur lechiffon afin de nettoyer le boîtier et les alimentations.
Respectez les instructions du fabricant du produit denettoyage.
AVERTISSEMENT : ÉTEIGNEZL’APPAREIL ET DÉBRANCHEZ LE
CORDON D’ALIMENTATION DE LA PRISE
ÉLECTRIQUE AVANT DE NETTOYERL’APPAREIL AFIN D’ÉVITER TOUT CHOC
ÉLECTRIQUE ACCIDENTEL OU RISQUE
DE BRÛLURE. SEUL LE FOURNISSEUR DE
L’APPAREIL OU UN TECHNICIEN QUALIFIÉ
EST HABILITÉ À RETIRER LES CAPOTSOU À RÉPARER L’APPAREIL.
AVERTISSEMENT : PRENEZ TOUTES LESPRÉCAUTIONS POUR ÉVITER DE MOUILLER LE
CONCENTRATEUR D’OXYGÈNE ET LA BATTERIE,ET ASSUREZ-VOUS QU’AUCUN LIQUIDE NE PEUTPÉNÉTRER À L’INTÉRIEUR DE L’APPAREIL AFIN
D’ÉVITER TOUT DYSFONCTIONNEMENT OU ARRÊT
DE CELUI-CI, AINSI QU’UN RISQUE ACCRU DE CHOC
ÉLECTRIQUE OU DE BRÛLURE.

AVERTISSEMENT : UTILISEZ UN CHIFFON OU UNEÉPONGE HUMIDE (PAS MOUILLÉE) POUR NETTOYER
LA BATTERIE. IMBIBEZ TOUT D’ABORD LE CHIFFON
OU L’ÉPONGE D’UN PRODUIT DE NETTOYAGE DOUXET NETTOYEZ LE BOÎTIER DE LA BATTERIE.
AVERTISSEMENT : NE PAS NETTOYER AVEC UN
SYSTÈME DE LAVAGE AUTOMATISÉ.

REMPLACEMENT DE LA CANULE

suivez toujours les instructions du fabricant de lacanule pour une utilisation correcte. Remplacer lacanule nasale ou la tubulure d’oxygène tous les 3 à
6 mois (si utilisation quotidienne, 3 mois) selon les
recommandations du fabricant tel que recommandépar le fabricant de la canule ou votre fournisseurd’oxygène. Votre médecin ou votre fournisseurd’oxygène vous transmettra des consignes concernant
son nettoyage et son remplacement.
Des pièces de rechange peuvent également êtreobtenues auprès de votre fournisseur d’oxygène.

AVERTISSEMENT : NE DÉVERSEZ PAS DE LIQUIDEDIRECTEMENT SUR L’APPAREIL. UNE LISTE
D’AGENTS CHIMIQUES INDÉSIRABLES COMPREND,
SANS TOUTEFOIS S’Y LIMITER, LES SUBSTANCESSUIVANTES : ALCOOL ET PRODUITS À BASE
D’ALCOOL, PRODUITS CONCENTRÉS À BASE DE
CHLORE (CHLORURE D’ÉTHYLÈNE) ET PRODUITSÀ BASE D’HUILE CES PRODUITS NE DOIVENT PAS
ÊTRE UTILISÉS POUR NETTOYER LE BOITIER EN
PLASTIQUE DU CONCENTRATEUR, AU RISQUE
D’ENDOMMAGER L’APPAREIL.
AVERTISSEMENT : NETTOYEZ LE BOÎTIER,
LE PANNEAU DE COMMANDE ET LE CORDON
D’ALIMENTATION UNIQUEMENT AVEC UN CHIFFONOU UNE ÉPONGE HUMIDE (PAS MOUILLÉE) IMBIBÉE
D’UN PRODUIT DE NETTOYAGE DOUX, PUISESSUYEZ TOUTES LES SURFACES. NE LAISSEZAUCUN LIQUIDE PÉNÉTRER À L’INTÉRIEUR DU
CONCENTRATEUR. FAITES PARTICULIÈREMENTATTENTION À LA SORTIE D’OXYGÈNE DESTINÉE
AU RACCORD DE LA CANULE, AFIN QU’ELLE NE
CONTIENNE NI POUSSIÈRE, NI EAU, NI PARTICULES.

PN MN232-C4 K | Manuel d’utilisation

FRE - 49

FreeStyle Comfort

Conditions d’alarme
N° deprioritéDescription del’alarme

1

Signal sonore

Plus d’informations/Action suggérée

Dysfonctionnement
du système Hautetempérature de labatterie

1 bip toutes les
20 secondes

L’alarme de la batterie se déclenche
lorsque la température de la batterie
approche du seuil de sécurité.
Si le seuil de coupure est atteint, l’appareils’éteint. Si l’alimentation externe est
connectée, l’écran d’alarme s’afficheet l’alarme sonore continue pendant
2 minutes.
Effectuez le test avec la batterie à
température ambiante. Si le problème
persiste, contactez le fournisseur.

2

Dysfonctionnement dusystème Températureélevée1 bip toutes les
20 secondes

L’appareil émet une alarme lorsque latempérature interne approche du seuilde coupure.
Si le seuil de coupure est atteint, l’unités’éteint et l’écran d’alarme s’affiche,
puis l’alarme sonore continue pendant
2 minutes.
Assurez-vous d’éviter toute obstruction
des orifices de ventilation de l’appareil.
Si le problème persiste, contactez lefournisseur.

3

Dysfonctionnement
du système Hautepression

1 bip toutes les
20 secondes

Si le seuil de coupure est atteint, l’unités’éteint et l’écran d’alarme s’affiche,
puis l’alarme sonore continue pendant
2 minutes.
Si le problème persiste, contactez lefournisseur.

4

Aucun souffle détecté

1 bip toutes les
30 secondes

Aucun souffle détecté pendant
15 secondes.
Activé après le premier souffle ou à l’arrêt
de l’écran de préchauffage, selon lapremière éventualité.
Vérifiez que la canule n’est pas tordue ou
qu’elle ne fuit pas. Augmentez le réglage
de la sensibilité. Si le problème persiste,contactez le fournisseur.

5Arrêt pour batteriefaible1 bip toutes les
30 secondes

Alarmes jusqu’à épuisement de la batterie
et l’arrêt de l’appareil.
Chargez la batterie.

6

Préchauffage

aucun

L’appareil commence par le dosage
automatique jusqu’à ce qu’un souffle soit
détecté ou que préchauffage soit terminé,selon la première éventualité.

50 - FREPN MN232-C4 K | Manuel d’utilisation

Écran

FreeStyle Comfort
N° deprioritéDescription del’alarme

7

Écran

Signal sonore

Détails supplémentaires

Faible concentration

1 bip toutes les
30 secondes,
pendant 5 min

Si le problème persiste, contactez lefournisseur.

8

Dysfonctionnement dusystème de débit (défaillance de la valve
OCD, obstruction ducircuit de gaz)1 bip toutes les
30 s

Si le problème persiste, contactez lefournisseur.

9

Alerte de batteriefaible1 bip, aucune
répétition

Chargez la batterie.

10

Dépassement du tauxd’inhalation

1 bip toutes les
60 s

Le taux d’inhalation dépasse 40 bpm.

11

Dysfonctionnement
du système Panne duventilateuraucun

Si le problème persiste, contactez lefournisseur.

12

Dysfonctionnement
du système Faible
pressionaucun

Si le problème persiste, contactez lefournisseur.

13

Dysfonctionnement
du système Bassetempérature

aucun

Laissez l’appareil se réchauffer à
température ambiante. Si le problème
persiste, contactez le fournisseur.

Informations

Démarrage

1 bip d’information

L’écran s’affiche à la mise sous tension etvous entendez un bip pour confirmer quele buzzer fonctionne

Si la dose est déclenchée de manière
incorrecte, diminuez le réglage desensibilité. Si le problème persiste,contactez le fournisseur.

Démarrez l’affichage de la révision
actuelle du firmware dans le coin inférieur
droit.

PN MN232-C4 K | Manuel d’utilisation

FRE - 51

FreeStyle Comfort
N° deprioritéDescription del’alarme

Informations

Écran d’exécution
normale en mode
autoSAT (aucune
alimentation CC
branchée)aucun

Informations

Écran d’exécution
normale avec
autoSAT désactivé(aucune alimentation
CC branchée)

Aucun

Informations

Le paramètre a étéchangé

1 bip d’information

Informations

Brancherl’alimentation externe
à l’arrêt

aucun

Informations

Brancherl’alimentation
externe à l’état defonctionnement

aucun

Informations

Débrancher
l’alimentation externe
lorsque l’appareil està l’arrêt.

1 bip d’information

Informations

Débrancher
l’alimentation externe
lorsque l’appareil
fonctionne (avecbatterie présente)

1 bip d’information

52 - FREPN MN232-C4 K | Manuel d’utilisation

Écran

Signal sonore

Détails supplémentaires

Affichage de l’écran CAIRE pendant 2
secondes, suivi de l’état de la batterie etde la prise pendant 5 secondes avant quel’écran s’éteigne

Affichage de l’écran CAIRE pendant2 secondes, puis l’écran s’éteint.

FreeStyle Comfort
N° deprioritéDescription del’alarme

Informations

Écran

Signal sonore

Détails supplémentaires

En mode veille lorsque le bouton hautou bas est enfoncé,l’affichage indiquel’état de la batterie etde charge

aucun

L’écran s’affiche pendant 5 secondes ets’éteint

Informations

Mode Avion lorsquel’appareil est sur labatterie

1 bip long

Informations

Mode Avion lorsquel’appareil est branché

1 bip long

PN MN232-C4 K | Manuel d’utilisation

FRE - 53

FreeStyle Comfort

Instructions de navigation du menu
Vous pouvez accéder au menu en mode veille ou
en mode d’exécution. En mode d’exécution, si
une condition d’alarme survient, vous quitterez
automatiquement le menu. Si vous n’appuyez sur
aucune touche pendant 20 secondes, vous quitterez
automatiquement le menu. En veille, vous devrez
peut-être appuyer deux fois sur le bouton Menu pourréactiver l’appareil, puis accéder au menu.
Il existe quatre écrans de menu :1. Écran d’état: Affiche le nombre de rév. dumicrologiciel, la durée d’exécution, les cycles dedécharge de la batterie

Mode avion

2. Mode avion (MARCHE/ARRÊT)3. Sensibilité: réglable (1-5)
4. autoSAT (MARCHE/ARRÊT)Sensibilité

Exemple d’écran d’état

FONCTIONNEMENT DU MENU RÉGLAGES
Vous pouvez accéder au menu Réglages en appuyant
sur le bouton Menu/Sélection
en modesd’exécution ou veille.
Pour faire défiler les menus, appuyez sur lebouton.

Page d’état

54 - FREPN MN232-C4 K | Manuel d’utilisation

autoSATFreeStyle Comfort
PAGE D’ÉTAT

Pour annuler le mode Marche du mode avion, appuyez
surpendant que l’icône ANNULER est visible.

La page d’état du Freestyle Comfort affiche la révision du micrologiciel, les heures de fonctionnement
et le nombre de cycles de batterie sur la batterieconnectée.

Mode Marche du mode avion confirmé

Pour activer le mode Arrêt du mode avion, répétez lesétapes ci-dessus tout en basculant sur ARRÊT.
Page d’étatLes heures de fonctionnement sont affichées enheures entières.
Les décharges de batterie correspondent au nombre
de cycles pendant lesquels la batterie connectée aété déchargée et rechargée.

MODE AVIONLe FreeStyle Comfort est livré avec le mode avion
désactivé.

Si vous appuyez surlorsque l’écran affiche lesymbole « Suivant », la page du menu passera àl’option d’écran suivante.
Le mode avion est destiné à être utilisé en moded’exécution. Le mode avion sera automatiquement
désactivé si l’appareil est laissé en veille lorsqu’il est
alimenté par la batterie.
Remarque : Consultez la section Activer le mode avion
pour en savoir plus sur le mode avion.

MODE DE SENSIBILITÉLe FreeStyle Comfort est livré de l’usine avec un modede sensibilité par défaut défini sur 2 et ne nécessite
aucun réglage supplémentaire dans des conditions
normales. Les paramètres de sensibilité peuvent êtreréglés à des valeurs situées entre 1 et 5, 1 étant lavaleur moins sensible et 5 la plus sensible.
Écran principal du mode avion

Pour activer le mode, appuyez sur
pendant quel’icône MODIFIER est visible. La valeur basculera surle mot bleu « MARCHE ».

Écran du mode de sensibilité défini sur 2

Pour régler le mode de sensibilité, appuyez sur
pendant que l’icône MODIFIER est visible. La valeurbasculera sur le prochain numéro bleu.
Mode Marche du mode avion sélectionné

Pour confirmer le mode Marche du mode avion,
appuyez sur
pendant que l’icône ENREGISTRER
est visible. La valeur basculera sur le mot noir
« MARCHE ».
PN MN232-C4 K | Manuel d’utilisation

FRE - 55

FreeStyle Comfort
MODE AUTOSATLe FreeStyle Comfort est livré avec le mode autoSAT
activé.

Réglage du mode de sensibilité défini sur 3 sélectionné

Pour confirmer le réglage du mode de sensibilité
défini sur 3, appuyez sur
pendant que l’icône
ENREGISTRER est visible. La valeur basculera sur lechiffre noir « 3 ».

Écran principal du mode autoSAT

Pour annuler la modification du mode de sensibilité,
appuyez sur
pendant que l’icône ANNULER est
visible.

Pour désactiver le mode, appuyez sur
pendant quel’icône MODIFIER est visible. La valeur basculera surle mot bleu « ARRÊT ».

Réglage du mode de sensibilité défini sur 3 confirmé

Mode Arrêt d’autoSAT sélectionné

Pour confirmer le mode Arrêt d’autoSAT, appuyez sur
pendant que l’icône ENREGISTRER est visible.
La valeur basculera sur le mot noir « ARRÊT ».
Pour annuler le mode Marche d’autoSAT, appuyez sur
pendant que l’icône ANNULER est visible.

Réglage du mode de sensibilité défini sur 5 (max.) confirmé

Si vous appuyez surlorsque l’écran affiche lesymbole « Suivant », la page du menu passera àl’option d’écran suivante.
Mode Arrêt d’autoSAT confirmé

Pour activer le mode Marche d’autoSAT, répétez lesopérations ci-dessus tout en basculant sur MARCHE.
Si vous appuyez surlorsque l’écran affiche lesymbole « Suivant », la page du menu passera àl’option d’écran suivante.

56 - FREPN MN232-C4 K | Manuel d’utilisation

FreeStyle Comfort
Dépannage
Problème

Cause probable

Solution

L’appareil FreeStyle Comfort
affiche une alarme et émet un
bip intermittent.

Consultez le tableau relatif àl’état de l’alarme.

Consultez le tableau relatif à l’état de l’alarme.

Le concentrateur d’oxygène nes’allume pas.

La batterie est déchargée.

Alimentez l’appareil à l’aide d’une prise CA ou CC.

Non branché à une sourced’alimentation externe.

Assurez-vous que les branchements externes
soient sécurisés.

Dysfonctionnement général.

Effectuez un cycle à pleine puissance en retirantla batterie et en déconnectant toute sourced’alimentation externe CA ou CC. Ensuite,réinstallez la batterie et le cycle d’alimentation est
terminé. Reconnectez la source d’alimentation
externe si vous le souhaitez.
Contactez le fournisseur de votre équipement
et passez à une autre source d’alimentation en
oxygène si nécessaire.

Tous les autres problèmes.

Effectuez un cycle à pleine puissance en retirantla batterie et en déconnectant toute sourced’alimentation externe CA ou CC. Ensuite,réinstallez la batterie et le cycle d’alimentation est
terminé. Reconnectez la source d’alimentation
externe si vous le souhaitez.
Passez à une autre source d’oxygène et contactez
le fournisseur de votre équipement.

Remarque : Certaines alarmes entrainent l’arrêt de l’appareil.
Remarque : pendant une perte d’alimentation totale, le système d’alarme se réinitialise (efface toutes les alarmesprécédentes) et n’affiche une alarme que si la condition persiste une fois l’alimentation rétablie.

PN MN232-C4 K | Manuel d’utilisation

FRE - 57

FreeStyle Comfort

Informations relatives au transport
Le FreeStyle Comfort est conforme aux directives
de la FAA pour une utilisation à bord des avions
commerciaux. Avant tout voyage (avion, train,bateau, etc.) :• veillez à ce que le concentrateur fonctionne
correctement.
• Contactez votre compagnie de transport au moins48 heures avant le départ afin de l’informer que vous
voyagerez avec un appareil FreeStyle Comfort.
Il est recommandé de toujours disposer de batteries
supplémentaires lors de vos déplacements. Pour unvoyage en avion, veuillez charger suffisamment labatterie, en tenant compte du temps passé au sol(vérification avant le vol, sécurité, etc.) et des escales.
La plupart des compagnies aériennes exigent quel’utilisateur dispose du nombre adéquat de batteries
pleinement chargées pour alimenter l’appareil durant
au moins 150 % de la durée maximum estimée duvoyage (temps de vol, temps au sol avant et après levol, correspondances et retards imprévus).
Remarque : Si vous prenez l’avion, il est recommandé deprendre des batteries supplémentaires dans un sac detransport sécurisé.

RÉGLEMENTATION FAA

la batterie UNIQUEMENT, lorsqu’aucune sourced’alimentation externe n’est connectée.

2. Activez le mode avion à l’aide de l’une desméthodes suivantes :a. Suivez les instructions pour activer le mode avion
dans la section des instructions de navigation dumenu.

b. Appuyez sur les boutons Haut et Bas etmaintenez-les enfoncés pendant 5 secondes jusqu’à
ce que vous entendiez un long bip et que le symbole
de l’avion apparaisse.

Les batteries de rechange doivent être protégées
individuellement pour éviter les courts-circuits en lesplaçant dans l’emballage d’origine ou en isolant lesbornes, par ex. en attachant du ruban adhésif sur lesbornes exposées ou en plaçant chaque batterie dans
un sac en plastique séparé ou dans une pochette deprotection, puis en les transportant uniquement dansles bagages à main.

ACTIVER LE MODE AVION
Alimentation par la batterie
Les systèmes d’alimentation POC peuvent êtreutilisés dans un avion suivant les procédures normales
d’utilisation en utilisant la batterie uniquement,
lorsqu’aucune source d’alimentation externe n’est
connectéePrise d’alimentation
Dans un avion, si une prise de courant est disponible,
opérationnelle et que son emplacement convient à une
utilisation comme source d’alimentation POC, elle peut
servir de batterie de secours aux batteries POC. Dansce cas, la procédure recommandée suivante concernant
la transition de la batterie vers l’alimentation électriquede l’avion doit être suivie :1. Allumez le système d’alimentation POC en utilisant

58 - FREPN MN232-C4 K | Manuel d’utilisation

3. Lorsque le symbole de l’avion est affiché, branchez
l’alimentation POC.

Remarque : En raison des limitations aériennes, lesbatteries ne se chargent pas en mode Avion.

FreeStyle Comfort
Accessoires

INSERTION DE L’APPAREIL DANS LE SAC
DE TRANSPORT

Afin de garantir des performances et une sécurité
optimales, utilisez uniquement les accessoires
ci-dessous fournis par CAIRE par le biais de votre
fournisseur d’oxygène. L’utilisation de produits
différents pourrait nuire aux performances et/ou à lasécurité du concentrateur.

1. Insérez l’appareil au fond du sac, en alignant les
ouvertures sur les filtres et la batterie.

Accessoires standard pour l’appareil FreeStyle
ComfortSac de transport

MI396-1

Sac à dos

MI459-1

Bandoulière

MI397-1

Alimentation CA (tous les cordons
sont inclus)

PW036-1S

Cordon d’alimentation CC

CD041-1

Filtre pour grosses particules externes(Droite)FI226-1

Filtre pour grosses particules externes(Gauche)FI227-1

2. Fermez les fermetures éclair situées au bas de lapoche.

Accessoires fournis en option
Pack supplémentaire de batterie 8
cellulesBT034-1S

Bloc-batterie avec 16 cellules

BT035-1S

Chargeur de bureau de la batterie

BT036-1

Sac de transport pour les accessoires

MI372-2

Remarque : si elles sont reprises ci-dessus, des options
supplémentaires peuvent être disponibles pour les cordonsd’alimentation spécifiques aux pays. Contactez CAIRE ou
votre fournisseur d’oxygène si vous souhaitez commander
d’autres accessoires.

FIXATION DE LA BANDOULIÈRE
1. La bandoulière peut être attachée directement àl’appareil ou au sac de transport.
2. Clipsez la bandoulière sur l’anneau de l’appareil ou
du sac.

Instructions relatives au sac detransport
Pour insérer et sécuriser votre appareil FreeStyle
Comfort dans le sac de transport, respectez les
instructions suivantes.

PN MN232-C4 K | Manuel d’utilisation

FRE - 59

FreeStyle Comfort

Élimination et réutilisation
Renvoyez toujours l’appareil FreeStyle Comfort, ycompris tous les composants, à votre fournisseur deéquipement pour une mise au rebut appropriée. Vous
pouvez également contacter vos services municipaux
locaux pour obtenir des instructions concernantl’élimination appropriée de la batterie. G. L’unité
peut être réutilisée entre les personnes mais doitêtre retournée au fournisseur de équipement pour unretraitement.

à un centre de recyclage à la fin de son cycle de vie,conformément à la directive relative aux déchetsd’équipements électriques et électroniques (DEEE).
Nos produits sont conformes à la Directive sur lalimitation des substances dangereuses (RoHS). Ils ne
contiendront pas plus qu’une quantité infime de plombou d’autres substances dangereuses.
MISE EN GARDE : concernant l’élimination
adéquate, contactez le fournisseur oul’organisme local pour en savoir plus.

DEEE et RoHS
Ce symbole a pour objet d’inciter lepropriétaire de l’équipement à le retourner

Matériaux en contact direct avec l’utilisateurBoîtier du concentrateur............................................................................................................................... PC/ABS
Panneau de commande du concentrateur.................................................................. Polyester EBG7 ou équivalent
Poignée du concentrateur....................................................................................................................Polycarbonate
Sortie d’oxygène, embout...................................................................................................................Polycarbonate
Plaque signalétique..........................................................................................................................................Lexan
Raccords du cordon..............................................................................................Polycarbonate/Chlorure de vinyle
Bouton de déroulement du cordon.......................................................................................................... PVC, métal
Alimentation....................................................................................................................Lexan 940 (polycarbonate)
Batterie......................................................................................................................................................... PC/ABS
Étiquettes sur l’alimentation et la batterie...........................................................................................Film polyester
Sac de transport du concentrateur................................................................ Cordura, Néoprène, doublure en nylon
Bandoulière...............................................................................Polypropylène, acétyle, vinyle, Cordura, Néoprène

Test CEM
Directives et déclaration du fabricant - Émissions électromagnétiques
L’appareil FreeStyle Comfort est destiné à être utilisé dans l’environnement électromagnétique spécifié ci-dessous. Le
client ou l’utilisateur de l’appareil FreeStyle Comfort doit s’assurer que ces derniers sont utilisés dans un environnement dece type.
Test d’émissions
Conformité
Environnement électromagnétique - Directives
Émissions RF
CISPR 11
Émissions RF
CISPR 11
Émissions de courant
harmonique
CEI 61000-3-2
Fluctuations de tension/
papillotement CEI 610003-3
60 - FREGroupe 1

L’appareil FreeStyle Comfort utilise uniquement l’énergie RF pour son
fonctionnement interne. Par conséquent, ses émissions RF sont très
faibles et ne sont pas susceptibles de provoquer des interférences avecdes équipements électroniques situés à proximité.

Classe B
Classe A
Conforme

PN MN232-C4 K | Manuel d’utilisation

Le FreeStyle Comfort est adapté à une utilisation dans tous lesétablissements, y compris les établissements domestiques et ceux
directement connectés au réseau public d’alimentation basse tension qui
alimentent les bâtiments utilisés à des fins domestiques.

FreeStyle Comfort
Directives et déclaration du fabricant - Immunité électromagnétique

L’appareil FreeStyle Comfort est destiné à être utilisé dans l’environnement électromagnétique spécifié ci-dessous. Le
client ou l’utilisateur de l’appareil FreeStyle Comfort doit s’assurer que ces derniers sont utilisés dans un environnement dece type.
Test d’immunité
Niveau de test
Niveau de conformité
Environnement électromagnétique CEI 60601
Directives
Environnement
Les sols doivent être en bois, en cimentContact ±8 kVélectromagnétique - Contact ±8 kV
ou carrelés. Si les sols sont revêtus d’unDirectives
matériau synthétique, le taux d’humidité
Décharge d’air ±15 kV
Décharge d’air ±15 kV
relative doit être d’au moins 30 %.
CEI 61000-4-2
Transitoiresélectriques rapides/
Salves
CEI 61000-4-4Surtension
CEI 61000-4-5±2 kV pour les lignes
d’alimentation±1 kV pour les lignes
d’entrée/sortie±1 kV entre fils de ligne
Mode commun ± 2kV surles lignes AC±2 kV pour les lignes
d’alimentation
S.O.

±1 kV entre fils de ligne

Différentiel ± 1kV sur leslignes AC
Mode commun ± 2kV
sur les lignes d’E/S
extérieures
UT de 0 % pour 0,5 cycle(0 °, 45 °, 90 °, 135 °,180 °, 225 °, 270 °,315 °)

Creux de tension,
coupures brèves
UT de 0 % pour 1 cycle
et variations de(00)
tension sur lescircuits d’alimentationélectrique CEI 61000- UT de 70 % (baisse de
30 % en UT) pendant
4-11
25/30 cycles (00)
UT de 0 % pour250/300 cycles (00)

UT de 0 % pour 0,5 cycle(0 °, 45 °, 90 °, 135 °,180 °, 225 °, 270 °,315 °)
UT de 0 % pour 1 cycle(00)
UT de 70 % (baisse de
30 % en UT) pendant
25/30 cycles (00)La qualité de l’alimentation secteur doitêtre celle d’un établissement de soinsde santé professionnel typique et desenvironnements de soins à domicile.

La qualité de l’alimentation secteur doitêtre celle d’un établissement de soinsde santé professionnel typique et desenvironnements de soins à domicile.

La qualité de l’alimentation secteur doitêtre celle d’un établissement de soinsde santé professionnel typique et desenvironnements de soins à domicile. Si
l’utilisateur du FreeStyle Comfort nécessite
un fonctionnement continu pendant les
coupures d’alimentation, il est recommandéque le FreeStyle Comfort soit alimenté parune source d’alimentation ininterruptible ou
une batterie.

UT de 0 % pour250/300 cycles (00)

Les champs magnétiques de la fréquenced’alimentation doivent être à des niveaux
caractéristiques d’un emplacement typique
3 A/m
3 A/m
dans un établissement de soins de santéprofessionnel et dans les environnements
de soins à domicile.
REMARQUE : UT correspond à la tension de secteur CA avant l’application du niveau de test.

Champ magnétiqueà la fréquence duréseau (50/60 Hz)
CEI 61000-4-8

REMARQUE Dans certains cas, le Freestyle Comfort peut être réinitialisé après une décharge électrostatique importante
et peut nécessiter un cycle à pleine puissance pour reprendre son fonctionnement normal. Voir la section Dépannage pourplus de détails.

PN MN232-C4 K | Manuel d’utilisation

FRE - 61FreeStyle Comfort
Directives et déclaration du fabricant - Immunité électromagnétique

L’appareil FreeStyle Comfort est destiné à être utilisé dans l’environnement électromagnétique spécifié ci-dessous. Le client
ou l’utilisateur de l’appareil FreeStyle Comfort doit s’assurer que ces derniers sont utilisés dans un environnement de ce
type.
Test d’immunité Niveau de test CEI 60601 Niveau de conformité
Environnement électromagnétique Directives
RF conduites
10 Vrms
10 Vrms
Les équipements de communication RF
portables et mobiles ne doivent pas êtreCEI 61000-4-6
150 kHz à 80 MHz
utilisés à proximité des composants del’appareil FreeStyle Comfort, y compris
les câbles, à une distance inférieure à
la distance de séparation recommandée
RF rayonnées
10 V/m
10 V/mcalculée sur la base de l’équation applicable
CEI 61000-4-3
80 MHz à 2700 MHz
à la fréquence de l’émetteur.
Distance de séparation recommandéed = 1,2√ P
d = 1,2√ P = 80 MHz à 800 MHz
d = 1,2√ P = 800 MHz à 2 700 MHz
où P correspond à la puissance nominalede sortie maximum de l’émetteur en watts(W), indiquée par le fabricant de l’émetteur,et d correspond à la distance de séparation
recommandée en mètres (m).
L’intensité des champs émis par desémetteurs RF fixes, déterminée par l’étude
électromagnétique d’un site a doit êtreinférieure au niveau de conformité pourchaque plage de fréquences. b
Des interférences peuvent se produireà proximité des équipements portant lesymbole suivant :REMARQUE 1 : à 80 MHz et 800 MHz, la plage de fréquences supérieure s’applique.
REMARQUE 2 : il est possible que ces directives ne s’appliquent pas à toutes les situations. La propagationélectromagnétique dépend des phénomènes d’absorption et de réflexion induits par les structures, les objets et lespersonnes.
aL’intensité des champs produits par des émetteurs fixes, tels que les stations de base pour les radiotéléphones(cellulaires/sans fil) et les radios mobiles, les radios amateurs, les radios AM et FM et la télévision ne peut pas être prédite
théoriquement avec précision. Pour évaluer l’environnement électromagnétique dû aux émetteurs RF fixes, une étude
électromagnétique du site doit être envisagée. Si l’intensité du champ, mesurée à l’emplacement où l’appareil FreeStyle
Comfort est utilisé, dépasse le niveau de conformité RF applicable indiqué ci-dessus, il conviendra de s’assurer du bon
fonctionnement de l’appareil FreeStyle Comfort dans un tel environnement. En cas d’anomalie, il pourra être nécessaire deprendre des mesures supplémentaires, par exemple en modifiant l’orientation ou la position de l’appareil FreeStyle Comfort.
b

Sur la plage de fréquences de 150 kHz à 80 MHz, l’intensité des champs doit être inférieure à 10 V/m.

62 - FREPN MN232-C4 K | Manuel d’utilisation

FreeStyle Comfort
Distances de séparation recommandées entre les équipements de communication RF
mobiles et portables et l’appareil FreeStyle Comfort

Le FreeStyle Comfort est prévu pour une utilisation dans un environnement dans lequel les perturbations dues aux
radiofréquences rayonnées sont contrôlées. Le client ou l’utilisateur de l’appareil FreeStyle Comfort peut contribuer
à prévenir les interférences électromagnétiques en maintenant une distance minimale entre les équipements decommunication RF portables (émetteurs) et le FreeStyle Comfort, comme indiqué ci-dessous, en fonction de la puissancede sortie maximale de l’équipement de communication.
Puissance nominale
Distance de séparation en fonction de la fréquence de l’émetteurde sortie maximum deml’émetteur
150 kHz à 80 MHz
80 MHz à 800 MHz
800 MHz à 2,5 GHz

d = 1,2√ P

d = 1,2√ P

d = 1,2√ P

W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Pour les émetteurs avec une puissance nominale de sortie maximum ne figurant pas dans le tableau ci-dessus, la distancede séparation recommandée d en mètres (m) peut être déterminée à l’aide de l’équation applicable à la fréquence del’émetteur, où P correspond à la puissance nominale de sortie maximum de l’émetteur en watts (W), indiquée par lefabricant de l’émetteur.
REMARQUE 1 À 80 MHz et 800 MHz, la distance de séparation correspondant à la plage de fréquences supérieures’applique.
REMARQUE 2 : il est possible que ces directives ne s’appliquent pas à toutes les situations. La propagationélectromagnétique dépend des phénomènes d’absorption et de réflexion induits par les structures, les objets et lespersonnes.

Classification

Type de protection contre les chocs électriques :Classe II	La protection contre les chocs électriques
est assurée par une double isolation.

section 21 de la norme RTCA DO-160.
	• CISPR 11/EN 55011 Classe B Groupe 1,« Appareils industriels, scientifiques et médicaux »	• Article 15 de la réglementation FCC, souspartie B – Radiations involontaires de classe B

Degré de protection contre les chocs électriques :Type BF Équipement assurant un degré particulier
de protection contre les chocs électriques
concernant
		 1) le courant de fuite autorisé ;		2) la fiabilité de la mise à la terre de protection (lecas échéant).
		 Non destiné aux utilisations cardiaques directes.

IP22 - Protection contre l’entrée de doigts ou d’objets
similaires d’un diamètre supérieur à 12,5 mm. L’eau
tombant verticalement ne doit pas avoir d’effet
néfaste lorsque l’habitacle est incliné à un angle de
15 degrés par rapport à sa position normale.

Tests indépendants pour la norme des équipementsélectromédicaux :Testé pour sa conformité• IEC 60601-1 édition 3.1 : Équipement électriquemédical - Partie 1 Exigences générales pour lasécurité.
• Norme CAN/CSA C22.2 no 60601-1-08
édition 2.1 Équipement électrique médical –Partie 1 : exigences générales de sécurité
Protection contre les interférences électromagnétiques
et autres entre cet appareil et d’autres dispositifs.
	• Testé pour être en conformité avec la norme
EN 60601-1-2/IEC 60601-1-2 : 4e édition.
	• Testé pour sa conformité à la catégorie M de laMode autorisé de nettoyage et de prévention
des infections : veuillez consulter la section« Nettoyage, entretien et maintenance régulière »de ce manuel d’utilisation.
Degré de sécurité d’utilisation en présence de gazanesthésiques inflammables :Appareil non adapté à ce type d’application.
Mode de fonctionnement : Fonctionnement en continu.

CAIRE et CAIRE Inc. sont des marques déposées de CAIRE Inc. Veuillez visiter notre site
Web ci-dessous pour obtenir une liste complète des marques de commerce.
Marques: www.careinc.com/corporate/trademarks.
Copyright © 2021 CAIRE Inc. CAIRE Inc. se réserve le droit d’interrompre lacommercialisation de ses produits ou de modifier les tarifs, les matériaux, les équipements,
les niveaux de qualité, les descriptions, les caractéristiques et/ou les processus de ses
produits, et ce à tout moment, sans avertissement préalable et sans aucune autre obligation
ni conséquence. Nous nous réservons tous les droits non expressément stipulés dans leprésent document, selon les conditions applicables.

PN MN232-C4 K | Manuel d’utilisation

FRE - 63

FreeStyle Comfort

Steuerungselemente und Systemstatusanzeigen
ISO 7000; graphische Symbole für die Verwendung auf
dem Gerät – Index und Übersicht
Lesen Sie vor der Verwendung dieseBedienungsanleitung. Reg.-Nr. 1641
Lager- oder Betriebstemperaturbereich.
Reg.-Nr. 0632

Lagerung luftfeuchtigkeit bereich.
Reg.-Nr. 2620

Richtlinie 93/42/EWG des Rates für MedizinprodukteAutorisierter Vertreter in derEuropäischen Gemeinschaft
Wenn auf dem UDI-Etikett (ProductUnique Device Identifier) das CE ####
-Symbol angebracht ist, entspricht das
Gerät den Anforderungen der Richtlinie
93/42 / EWG für Medizinprodukte. Das
CE #### -Symbol zeigt die Nummer desangemeldeten Körpers an.
IEC 60417
Gerät der Klasse II, doppelt isoliert
Reg.-Nr. 5172

Luftdruckbegrenzung. Reg.-Nr. 2621
Name und Adresse des Herstellers.
Reg.-Nr. 3082

Netzschalter. Reg.-Nr. 5009
ETL-Zertifizierung

Herstellungsdatum. Reg.-Nr. 2497

Sicherheitsstandard CAN/CSA C22.2
Nr. 601.1 M90 Medizinische elektrische
GeräteKatalognummer. Reg.-Nr. 2493
Seriennummer. Reg.-Nr. 2498

Interne Symbole

Diese Seite nach oben. Reg.-Nr. 0623

Von entzündlichen Materialien, Öl und
Fett fernhalten.

Zerbrechlich, Vorsicht bei derHandhabung. Reg.-Nr. 0621

Nicht zerlegen.

Vor Regen schützen, trocken
aufbewahren. Reg.-Nr. 0626

Durchfluss erhöhen.

Stapelgrenze (Zahl). Reg.-Nr. 2403
Durchfluss reduzieren.

ISO 7010: Graphische Symbole – Sicherheitsfarben undSicherheitszeichen – Registrierte Sicherheitszeichen
Die Betriebsanleitung ist aufmerksam
durchzulesen. Reg.-Nr. M002
Von Flammen, offenem Feuer undFunken fernhalten. Offene Zündquellen
und Rauchen verboten. Reg.-Nr. P003
Nicht in der Nähe des Geräts oderwährend dessen Betrieb rauchen.
Reg.-Nr. P002
Anwendungsteil des Typs BF(Schutzgrad gegen Stromschlag).
Reg.-Nr. 5333
Warnung. Reg.-Nr. W001

64 - GERPN MN232-C4 K | Bedienungsanleitung

Menü-Taste.
RTCA/
DO-160Abschnitt 21
Konform mit
Kategorie M

Entspricht RTCA DO160 Abschnitt 21
Kategorie M. Erfüllt Anforderung derRichtlinie FAA SFAR 106.
Gerät der Klasse II, doppelt isoliert

Zugelassen von FAA

FreeStyle Comfort
21 CFR 801.15: Code of Federal Regulations, Titel 21
Nach den in den USA geltenden Gesetzen
darf dieses Gerät nur von einem Arzt oderim Auftrag eines Arztes verkauft werden.
Richtlinie 2012/19/EU des Rates für: Elektro- undElektronik-Altgeräte (WEEE-Richtlinie)
WEEE-Richtlinie

Was genau ist der
Sauerstoffkonzentrator?

EN 60601-1: Medizinische elektrische Geräte – Teil
1: Allgemeine Festlegungen für die grundlegende
Sicherheit und die wesentlichen Leistungsmerkmale
Tropfwassergeschützt
Dieses Produkt kann von einem oder mehreren USamerikanischen oder internationalen Patenten abgedeckt
sein. Auf unserer Webseite finden Sie eine Auflistung dergeltenden Patente. Pat.: www.caireinc.com/corporate/
patents/

Einstellung gemäß Ihrem persönlichen respiratorischen
Zustand verschrieben.

Die Atemluft enthält ca. 21 % Sauerstoff, 78 % Stickstoff
und 1 % andere Gase. Die Raumluft wird durch den
Lufteinlass in den FreeStyle® Comfort® eingesaugt. Sie
fließt dann durch ein Absorbermaterial, das sogenannte
Molekularsieb. Dieses Material trennt den Sauerstoff
vom Stickstoff und lässt nur den Sauerstoff hindurch.
Das Ergebnis ist hochreiner Sauerstoff, der dem
Anwender zugeführt wird.

Obgleich Sauerstoff als Medikament nicht süchtig macht,
kann eine unbefugte Sauerstofftherapie gefährlich sein.
Vor der Verwendung dieses Sauerstoffkonzentrators
müssen Sie einen Arzt konsultieren. Der Geräteanbieter
Ihres Sauerstoffgeräts wird Ihnen zeigen, wie Sie dieverschriebene Flussrate einstellen.

Im FreeStyle Comfort werden fortschrittliche
Sauerstoffkonzentrator- und SauerstoffkonservierungsTechniken kombiniert. Das tragbare Gerät bietet
maximale Portabilität und Mobilität. Das fortschrittliche
Impulsfluss-Abgabesystem bemerkt schnell, wenn
der Benutzer einatmet, und gibt jeweils zu Beginn
der Einatemphase einen Sauerstoffimpuls ab. Das
Impulsfluss-Abgabesystem ist ideal für Aktivitäten
außerhalb des eigenen Hauses. Es konserviert
den Sauerstoff und verlängert im Gegensatz zum
kontinuierlichen Durchfluss die Lebensdauer des
Akkus. Der FreeStyle Comfort ist leicht und bequem
per Schultergurt zu tragen. Es kann mit Wechselstrom(Wandsteckdose), Gleichstrom (Fahrzeug) oder mit
dem Akku betrieben werden.

Konzentratoren sind für die zusätzliche Sauerstoffzufuhr
an den Anwender bestimmt, die unter Beschwerden
aufgrund eines beeinträchtigten Übergangs desSauerstoffs von der Lunge in den Blutkreislauf
leiden. Tragbare Sauerstoffkonzentratoren (POCs)
speichern und enthalten keinen Sauerstoff. Sie müssen
nicht wieder aufgefüllt werden und können überall
aufgeladen werden, wo Wechsel- oder Gleichstromvorhanden ist (außer im Flugzeug). Auf diese Weise
ist der Benutzer relativ unabhängig in Bezug auf
Flexibilität (sowohl zu Hause als auch unterwegs),
Mobilität und in seinem gesamten Lebensstil. Die
Verwendung eines Sauerstoffkonzentrators muss voneinem Arzt verschrieben werden. Das Gerät ist nicht fürlebenserhaltende Maßnahmen geeignet.

Verwendungszweck

Obgleich Sauerstofftherapien Menschen jeden
Alters verordnet werden können, handelt es sich inder Regel um Personen über 65 Jahre, die an einer
Atemwegserkrankung wie z. B. der chronischen
obstruktiven Lungenerkrankung (COPD) leiden.
Üblicherweise verfügen die Anwender noch über gutekognitive Fähigkeiten und müssen dazu in der Lage sein,
Unwohlsein mitzuteilen. Ist der Anwender nicht (mehr)in der Lage, Unwohlsein mitzuteilen oder die Aufkleber
auf dem Konzentrator bzw. die Gebrauchsanweisung zu
lesen und zu verstehen, darf das Gerät nur unter Aufsicht
einer Person verwendet werden, die hierzu in der Lage
ist. Treten während der Verwendung eines Konzentrators
Beschwerden auf, sollten sich die betroffenen
Anwender mit ihrem Arzt in Verbindung setzen. Es
wird Anwendern empfohlen, Reservesauerstoff (z. B.
in Form einer Sauerstoffflasche) bereitzuhalten, der im
Falle eines Stromausfalls oder eines sonstigen Ausfalls
des Konzentrators verwendet werden kann. Weiterebesondere Fertigkeiten oder Fähigkeiten des Anwenders
sind für den Gebrauch des Sauerstoffkonzentrators nicht
erforderlich.

WARNUNG: Der CAIRE FreeStyle Comfort
Sauerstoffkonzentrator ist für die Verabreichungvon zusätzlichem Sauerstoff vorgesehen. Das Gerät
ist weder für die Lebenserhaltung vorgesehen, nochbietet es Funktionen zur Patientenüberwachung.

Gründe für die Verschreibung vonSauerstoff durch den Arzt
Viele Menschen leiden unter verschiedenartigen Herz-,
Lungen- und anderen Atemwegserkrankungen. Einem
bedeutenden Teil dieser Menschen kann durch eine
zusätzliche Sauerstoffzufuhr zu Hause, auf Reisenoder bei Tagesaktivitäten geholfen werden.
Sauerstoff ist ein Gas, das 21 % der Umgebungsluft
ausmacht, die wir atmen. Unser Körper ist auf
eine konstante Zufuhr angewiesen, um richtig zu
funktionieren. Ihr Arzt hat Ihnen eine Flussrate oder eine

Profil des Anwenders

PN MN232-C4 K | Bedienungsanleitung

GER - 65

FreeStyle Comfort

Packen Sie Ihren FreeStyle Comfort ausÜberprüfen Sie, ob alleaufgeführten Komponenten im
Paket enthalten sind. Falls etwas
fehlen sollte, setzen Sie sich sofort
mit Ihrem Sauerstoffanbieter in
Verbindung.

Standardzubehör• Tragbarer Sauerstoffkonzentrator• Tragetasche• Schultergurt• 8-Zellen Akkupack• Wechselstrom-Netzteil• Wechselstromkabel• Gleichstrom-Netzteil• Bedienungsanleitung

Optionales Zubehör• Zusätzlicher 8-Zellen Akkupack• 16-Zellen Akkupack• Tischladegerät• Zubehörtasche für Reisegepäck

Einführende Informationen zu Ihrem FreeStyle Comfort
Machen Sie sich mit den wichtigsten Funktionen des FreeStyle Comfort und der Benutzeroberfläche vertraut.
Anbringen desSchultergurts

Vorderansicht
Benutzeroberfläche

Luftauslass
Sauerstoffauslass

Lufteinlass/Filter
Lufteinlass/Filter

Eingangfür den
Stromanschluss
Akkupack
AkkupackEntriegelungsknopf
66 - GERPN MN232-C4 K | Bedienungsanleitung

FreeStyle Comfort
Draufsicht
Bildschirm
Stufe erhöhen

Ein-/Aus-Schalter

Menü-Taste

Stufe reduzieren

Bildschirm bei Akkubetrieb
Einstellung derDurchflussrate

Akkuanzeige

autoSAT-Statusanzeige
Alarmanzeige
Restakkustandin Prozent
Anzeige
Bolusgabe

Display bei Gleich-/Wechselstrom
Einstellung derDurchflussrate
Akkuanzeige

Alarmanzeigefeld

Symbol fürden Anschluss
einer externen
Stromquelle

PN MN232-C4 K | Bedienungsanleitung

GER - 67

FreeStyle Comfort

Wichtig!

Warn- und Sicherheitshinweise sindfolgendermaßen gekennzeichnet:
WARNUNG: WEIST AUF GEFAHREN HIN,DIE SCHWERE VERLETZUNGEN ZUR
FOLGE HABEN KÖNNEN.
VORSICHT: Wichtige Informationen,um Schäden am FreeStyle Comfort zu
vermeiden.
Hinweis: Besonders zu beachtende Informationen.

Gebrauchshinweise
WARNUNG: DAS VORGESCHRIEBENESAUERSTOFFNIVEAU MUSS UNBEDINGTEINGEHALTEN WERDEN. EINSTELLUNGENDER FLUSSRATE DÜRFEN NUR
GEÄNDERT WERDEN, WENN DIES AUF
ANWEISUNG EINES APPROBIERTENARZTES ERFOLGT.
WARNUNG: LAUT US-GESETZEN DARFDIESES GERÄT NUR AUF ANWEISUNG EINESARZTES ODER EINES ANDEREN LIZENZIERTENGESUNDHEITSDIENSTLEISTERS VERKAUFT ODERVERLIEHEN WERDEN.
WARNUNG: DIESES GERÄT DARF NICHT FÜR
LEBENSERHALTENDE MASSNAHMEN EINGESETZTWERDEN. ÄLTERE, PÄDIATRISCHE ODER ANDEREANWENDER, DIE IHR UNWOHLSEIN WÄHREND DER
VERWENDUNG DES GERÄTS NICHT KOMMUNIZIERENKÖNNEN, MÜSSEN MÖGLICHERWEISE ZUSÄTZLICHÜBERWACHT WERDEN. ANWENDER MIT HÖRUND/ODER SEHBEHINDERUNG BENÖTIGEN
MÖGLICHERWEISE HILFE BEIM ÜBERWACHEN DER
ALARME.
WARNUNG: DIE EINSTELLUNGEN FÜR DENIMPULSDOSIS-MODUS MÜSSEN FÜR JEDEN NUTZERINDIVIDUELL ENTSPRECHEND DEM PERSÖNLICHENBEDARF IN RUHEPOSITION, BEI BEWEGUNG ODERAUF REISEN ANGEPASST WERDEN.
WARNUNG: EINIGE ATEMANSTRENGUNGEN DESPATIENTEN LÖSEN MÖGLICHERWEISE NICHT DIE
KONSERVIERUNGSFUNKTION AUS.
WARNUNG: DER FREESTYLE COMFORT IST
NICHT FÜR DIE VERWENDUNG MIT EINEM
TRACHEOTOMIERTEN PATIENTEN VORGESEHEN.
WARNUNG: SCHWANGERE ODER STILLENDEFRAUEN SOLLTEN KEINE IN DIESEM HANDBUCH
EMPFOHLENEN ZUBEHÖRTEILE VERWENDEN, DA
DIESE PHTHALATE ENTHALTEN KÖNNEN.

68 - GERPN MN232-C4 K | Bedienungsanleitung

Gegenanzeigen
WARNUNG: DIE VERWENDUNG VONSAUERSTOFF OHNE ÄRZTLICHEVERSCHREIBUNG IST UNTER
UMSTÄNDEN GEFÄHRLICH. DAS GERÄTDARF NUR AUF VERSCHREIBUNG EINESARZTES VERWENDET WERDEN.
WARNUNG: DAS GERÄT DARF NICHT VERWENDETWERDEN, WENN BRENNBARE ANÄSTHETIKA
VORHANDEN SIND.
WARNUNG: WIE ALLE ELEKTRISCH
BETRIEBENEN GERÄTE KANN AUCH DERSAUERSTOFFKONZENTRATOR, BEISPIELSWEISE
BEI EINEM STROMAUSFALL ODER WENN EINEWARTUNG DURCH EINEN QUALIFIZIERTENTECHNIKER ERFORDERLICH IST, AUSFALLEN.
DER SAUERSTOFFKONZENTRATOR IST NICHTFÜR ANWENDER GEEIGNET, DIE DURCH EINESOLCHE TEMPORÄRE UNTERBRECHUNGGESUNDHEITLICHEN SCHADEN NEHMEN WÜRDEN.

Sicherheitsrichtlinien
WARNUNG: AN DIESEM GERÄT DÜRFENKEINE MODIFIKATIONEN VORGENOMMEN
WERDEN.
WARNUNG: DIESES GERÄT LIEFERTHOCHKONZENTRIERTEN SAUERSTOFF, DER
SCHNELL ZU BRÄNDEN FÜHREN KANN. RAUCHENODER OFFENE FLAMMEN SIND IM RAUM, IN DEM
SICH (1) DAS GERÄT ODER (2) EIN SAUERSTOFF
TRANSPORTIERENDES ZUBEHÖRTEIL BEFINDEN,NICHT ERLAUBT. DIE MISSACHTUNG DIESERWARNUNG KANN SCHWERE BRÄNDE, SACH- BZW.
PERSONENSCHÄDEN ODER DEN TOD ZUR FOLGE
HABEN.
WARNUNG: BETREIBEN SIE DAS GERÄT NICHTIN EINEM EINGESCHRÄNKTEN ODER ENGEN
BEREICH (Z. B. EINEM KLEINEN KOFFER ODEREINER HANDTASCHE), IN DEM DIE BELÜFTUNG
UNZUREICHEND SEIN KÖNNTE. HIERDURCHKANN SICH DER SAUERSTOFFKONZENTRATORÜBERHITZEN UND DIE LEISTUNG BEEINTRÄCHTIGTWERDEN.
WARNUNG: DER KONZENTRATOR DARF NICHTMIT RAUCH, SCHADSTOFFEN ODER DÄMPFEN IN
KONTAKT KOMMEN.
WARNUNG: WENN BEI DIESEM GERÄT EIN
SCHWERWIEGENDER VORFALL AUFTRITT, SOLLTE
DER BENUTZER DEN VORFALL UNVERZÜGLICH
DEM ANBIETER UND / ODER DEM HERSTELLERMELDEN. EIN SCHWERWIEGENDER VORFALL IST
DEFINIERT ALS EINE VERLETZUNG, DER TOD ODERDIE MÖGLICHKEIT, EINE VERLETZUNG / DEN TODZU VERURSACHEN, FALLS DER VORFALL ERNEUTAUFTRITT. DER BENUTZER KANN DEN VORFALL
AUCH DER ZUSTÄNDIGEN BEHÖRDE DES LANDESMELDEN, IN DEM DER VORFALL AUFGETRETEN IST.

FreeStyle Comfort
WARNUNG: DER EINSATZ EINIGERSAUERSTOFF FÜHRENDER
ZUBEHÖRTEILE, DIE NICHT FÜR
DIE VERWENDUNG MIT DIESEMSAUERSTOFFKONZENTRATORANGEGEBEN SIND, KANN DESSEN
LEISTUNG BEEINTRÄCHTIGEN. AUF
EMPFOHLENES ZUBEHÖR WIRD IN
DIESER ANLEITUNG VERWIESEN.
WARNUNG: IST DER SAUERSTOFFKONZENTRATORHERUNTERGEFALLEN, WURDE ER BESCHÄDIGTODER WASSER AUSGESETZT, SOLLTEN SIE SICH MIT
IHREM AUSRÜSTUNGSANBIETER IN VERBINDUNGSETZEN, UM DAS GERÄT PRÜFEN ODER EVENTUELLREPARIEREN ZU LASSEN. VERWENDEN SIE DENSAUERSTOFFKONZENTRATOR NICHT, WENN
STROMKABEL ODER STECKER BESCHÄDIGT SIND.
WARNUNG: SICHERN SIE DENSAUERSTOFFKONZENTRATOR MIT EINEM GURTODER ÄHNLICHEN VORRICHTUNGEN, WENN ER
IN EINEM FAHRZEUG TRANSPORTIERT WIRD, UMSCHÄDEN BZW. VERLETZUNGEN ZU VERMEIDEN.
WARNUNG: DIE RICHTLINIEN DESVERKEHRSMINISTERIUMS DER USA (DOT) SOWIE
DER VEREINTEN NATIONEN (UN) SEHEN VOR, DASSDER AKKU AUS DEM GERÄT HERAUSGENOMMEN
WERDEN MUSS, WENN DIESES ALS GEPÄCK AUF
INTERNATIONALEN FLÜGEN AUFGEGEBEN WIRD.
BEI VERSAND DES SAUERSTOFFKONZENTRATORSMÜSSEN DER AKKU EBENFALLS AUS DEM GERÄT
ENTNOMMEN UND ORDNUNGSGEMÄSS VERPACKTWERDEN.
WARNUNG: DAS GERÄT NUR NACH EINGEHENDEMSTUDIUM UND VERSTÄNDNIS DIESER ANLEITUNGVERWENDEN. SOLLTEN SIE DIE WARNUNGEN UND
ANWEISUNGEN NICHT VERSTEHEN, WENDEN SIE
SICH AN DEN BEREITSTELLER DIESES GERÄTES,BEVOR SIE VERSUCHEN, ES ZU VERWENDEN;
ANDERNFALLS KANN ES ZU VERLETZUNGEN ODERSCHÄDEN KOMMEN.
WARNUNG: BEI UNWOHLSEIN ODER EINEM
MEDIZINISCHEN NOTFALL SOFORT ÄRZTLICHE HILFEZURATE ZIEHEN.
WARNUNG: WIRD DER SAUERSTOFFKONZENTRATORAUSSERHALB DES ANGEGEBENENTEMPERATURBEREICHES VERWENDET,KANN DIES DIE FÄHIGKEIT DES GERÄTES
BEEINTRÄCHTIGEN, DIE ANGEGEBENENSAUERSTOFFKONZENTRATIONEN ZU ERZEUGEN.
FÜR TEMPERATURGRENZWERTE SIEHE DEN
ABSCHNITT „TECHNISCHE DATEN“ IN DIESERANLEITUNG.
WARNUNG: KEINE ÖL-, FETT-, ODERPETROLEUMBASIERTEN ODER ANDERENENTZÜNDLICHEN PRODUKTE MIT DEM
SAUERSTOFFKONZENTRATOR ODER DESSENSAUERSTOFF BEINHALTENDEM ZUBEHÖR
VERWENDEN. ES SOLLTEN AUSSCHLIESSLICHWASSERBASIERTE, SAUERSTOFFVERTRÄGLICHELOTIONEN ODER SALBEN VERWENDET WERDEN.
SAUERSTOFF BESCHLEUNIGT DIE VERBRENNUNG
ENTZÜNDLICHER SUBSTANZEN.

WARNUNG: STELLEN SIE SICHER, DASSLUFTEINLASS UND -AUSLASS NICHT BLOCKIERTWERDEN. FÜHREN SIE KEINE GEGENSTÄNDE IN DIEGERÄTEÖFFNUNGEN EIN UND LASSEN SIE NICHTS IN
DIESE HINEINFALLEN. HIERDURCH KANN SICH DERSAUERSTOFFKONZENTRATOR ÜBERHITZEN UND DIE
LEISTUNG BEEINTRÄCHTIGT WERDEN.
WARNUNG: EIN NICHT ORDNUNGSGEMÄSSERGEBRAUCH DES AKKUS KANN ZU EINEM ERHITZENUND ENTZÜNDEN DES AKKUS FÜHREN, WAS
SCHWERE VERLETZUNGEN ZUR FOLGE HABEN
KANN. DURCHBOHREN SIE DEN AKKU NICHT UND
SCHLAGEN UND TRETEN SIE NICHT AUF DEN
AKKU. LASSEN SIE DEN AKKU NICHT FALLEN UND
SETZEN SIE DEN AKKU AUCH KEINEN SONSTIGENHEFTIGEN SCHLÄGEN ODER STÖSSEN AUS. DIE
VERWENDUNG EINES BESCHÄDIGTEN AKKUS KANNZU VERLETZUNGEN FÜHREN.
WARNUNG: DIE METALLKONTAKTE DES
AKKUS DÜRFEN NICHT MIT METALLISCHENGEGENSTÄNDEN, Z. B. SCHLÜSSELN ODER MÜNZEN,KURZGESCHLOSSEN WERDEN. DAS KANN ZU
FUNKEN ODER ÜBERHITZUNG FÜHREN.
WARNUNG: WIRD DER TRAGBARESAUERSTOFFKONZENTRATOR IM FREIEN ÜBERWECHSELSTROM BETRIEBEN, SCHLIESSEN SIE DASSTROMKABEL NUR AN EINE STROMQUELLE MIT
FEHLERSTROM-SCHUTZEINRICHTUNG (RCD) AN.
WARNUNG: DER SAUERSTOFFKONZENTRATORSOLLTE NICHT DIREKT NEBEN ANDEREN GERÄTENODER MIT ANDEREN GERÄTEN GESTAPELTVERWENDET WERDEN. IST EINE SOLCHE SITUATIONNICHT ZU VERMEIDEN, MUSS DAS GERÄT AUF
ORDNUNGSGEMÄSSEN BETRIEB HIN ÜBERWACHTWERDEN.
WARNUNG: DIE VERWENDUNG VON ANDEREN
KABELN UND ADAPTERN ALS DEN ANGEGEBENEN,
MIT AUSNAHME DER KABEL UND ADAPTER,
DIE VOM HERSTELLER DES MEDIZINISCHENELEKTRISCHEN GERÄTS ALS ERSATZTEILE FÜR
INTERNE KOMPONENTEN VERKAUFT WERDEN,
KANN ZU ERHÖHTEN EMISSIONEN ODEREINER REDUZIERTEN STÖRFESTIGKEIT DESSAUERSTOFFKONZENTRATORS FÜHREN.
WARNUNG: VERWENDEN SIE NUR DIE ELEKTRISCHESPANNUNG, DIE AUF DEM GERÄTEETIKETT MIT DEN
TECHNISCHEN DATEN ANGEGEBEN IST.
WARNUNG: KEINE VERLÄNGERUNGSKABELFÜR DIESES GERÄT VERWENDEN UND NICHT ZU
VIELE STECKER MIT DERSELBEN STECKDOSEVERBINDEN. DIE VERWENDUNG EINESVERLÄNGERUNGSKABELS KANN DIE LEISTUNG DESGERÄTES BEEINTRÄCHTIGEN. DIE VERBINDUNGZU VIELER STECKER MIT EINER STECKDOSE KANN
DEN SCHALTSCHRANK ÜBERLASTEN UND ZUMAUSLÖSEN VON TRENNSCHALTER/SICHERUNGODER BEI FEHLFUNKTION VON TRENNSCHALTER/
SICHERUNG ZUM AUSBRUCH EINES FEUERS
FÜHREN.

PN MN232-C4 K | Bedienungsanleitung

GER - 69

FreeStyle Comfort
VORSICHT: Sollte ein Alarm ausgegeben
werden oder der Sauerstoffkonzentrator
nicht ordnungsgemäß funktionieren, lesen
Sie im Abschnitt „Fehlerbehebung“ dieser
Anleitung nach. An den Geräteanbieter
wenden, wenn das Problem nicht gelöst
werden kann.
VORSICHT: Alle Anweisungen des Herstellers
befolgen, damit die Garantie ihre Gültigkeit behält.
WARNUNG: NICHT VERSUCHEN, DASGERÄT – ABGESEHEN VON DEN IN
DIESER ANLEITUNG AUFGEFÜHRTENSTÖRUNGSBEHEBUNGEN – ZU WARTEN.
ENTFERNEN SIE KEINE ABDECKUNG.
NUR DER BEREITSTELLER DESGERÄTES ODER EIN QUALIFIZIERTERWARTUNGSTECHNIKER DÜRFEN DIE
ABDECKUNGEN ENTFERNEN ODER DASGERÄT WARTEN.
WARNUNG: VERWENDEN SIE NUR VOM
HERSTELLER EMPFOHLENES ZUBEHÖR. DIE
VERWENDUNG ANDEREN ZUBEHÖRS KANNZU EINER SCHWEREN BESCHÄDIGUNG DESSAUERSTOFFKONZENTRATORS SOWIE ZUM
ERLÖSCHEN DER GARANTIE FÜHREN.
WARNUNG: FÜR DEN FALL EINES STROMAUSFALLS,ALARMS ODER MECHANISCHEN VERSAGENSEMPFIEHLT DER HERSTELLER EINE ALTERNATIVESAUERSTOFFZUFUHR. DEN ARZT ODERGERÄTEANBIETER FRAGEN, WELCHE ART VONRESERVESYSTEM BENÖTIGT WIRD.
WARNUNG: STELLEN SIE SICHER, DASSLUFTEINLASS UND -AUSLASS NICHT BLOCKIERTWERDEN. FÜHREN SIE KEINE GEGENSTÄNDE IN DIEGERÄTEÖFFNUNGEN EIN UND LASSEN SIE NICHTS IN
DIESE HINEINFALLEN. HIERDURCH KANN SICH DERSAUERSTOFFKONZENTRATOR ÜBERHITZEN UND DIE
LEISTUNG BEEINTRÄCHTIGT WERDEN.
WARNUNG: DAS RAUCHEN WÄHREND DER
SAUERSTOFFVERABREICHUNG IST DER
HAUPTGRUND FÜR BRANDVERLETZUNGEN UND-TODE. DIESE SICHERHEITSWARNUNGEN MÜSSEN
UNBEDINGT BEFOLGT WERDEN:
WARNUNG: RAUCHEN, KERZEN ODER OFFENE
FLAMMEN SIND IN DEM RAUM, IN DEM SICH DASGERÄT ODER EIN SAUERSTOFF BEFÖRDERNDESZUBEHÖRTEIL BEFINDEN, VERBOTEN.
WARNUNG: RAUCHEN WÄHREND DES TRAGENSEINER SAUERSTOFFKANÜLE KANN ZUGESICHTSVERBRENNUNGEN UND ZUM TODEFÜHREN.
WARNUNG: WENN SIE RAUCHEN, KÖNNEN
DIESE 3 SCHRITTE IHR LEBEN RETTEN: DEN
SAUERSTOFFKONZENTRATOR AUSSCHALTEN, DIEKANÜLE ABNEHMEN UND DEN RAUM, IN DEM DASGERÄT SICH BEFINDET, VERLASSEN.

70 - GERPN MN232-C4 K | Bedienungsanleitung

WARNUNG: SCHILDER MIT DER AUFSCHRIFT„RAUCHEN VERBOTEN – SAUERSTOFF INGEBRAUCH“ MÜSSEN IM HAUS ODER AN DEM
ORT, AN DEM DER SAUERSTOFF VERWENDETWIRD, DEUTLICH ERKENNBAR ANGEBRACHT SEIN.
NUTZER UND PFLEGEPERSONAL MÜSSEN ÜBERDIE GEFAHREN DES RAUCHENS IN DER NÄHE ODERWÄHREND DES GEBRAUCHS VON MEDIZINISCHEMSAUERSTOFF AUFGEKLÄRT WERDEN.
WARNUNG: DEN SAUERSTOFFKONZENTRATORNICHT BEI VORHANDENSEIN ENTZÜNDLICHER GASE
VERWENDEN. DIES KANN SCHNELL ZU BRÄNDEN
FÜHREN, DIE SACHSCHÄDEN, VERLETZUNGEN ODERDEN TOD NACH SICH ZIEHEN KÖNNEN.
WARNUNG: EIN ABNEHMEN DER KANÜLE UND
EINE LAGERUNG AUF KLEIDUNG, BETTWÄSCHE,SOFAS ODER ANDEREM POLSTERMATERIALFÜHREN BEI VORHANDENSEIN EINER ZIGARETTE,WÄRMEQUELLE ODER FLAMME ZU EINERVERPUFFUNG.
WARNUNG: LASSEN SIE EINE KANÜLE NICHT AUF
ODER UNTER KLEIDUNG, BETTWÄSCHE ODERSTUHLPOLSTERN LIEGEN. WENN DAS GERÄT
EINGESCHALTET IST, ABER NICHT BETRIEBENWIRD, KANN DER SAUERSTOFF ENTZÜNDLICHWIRKEN. STELLEN SIE SICHER, DASS DER
SAUERSTOFFKONZENTRATOR AUSGESCHALTET IST,WENN ER NICHT VERWENDET WIRD.
WARNUNG: DER SAUERSTOFFKONZENTRATORMUSS VOR DEM ENTFERNEN DES AKKUPACKSAUSGESCHALTEN UND VON DER EXTERNEN
STROMVERSORGUNG GETRENNT WERDEN.
WARNUNG: KLEINE TEILE KÖNNEN EINEPOTENZIELLE ERSTICKUNGSGEFAHRDARSTELLEN.
WARNUNG: VERMEIDEN SIE DEN KONTAKT MIT DER
ABLUFTÖFFNUNG FÜR LÄNGER ALS 10 SEKUNDEN,DA DIESE WARM SEIN KANN.
WARNUNG: WIND ODER STARKE ZUGLUFTKÖNNEN DIE PRÄZISE DURCHFÜHRUNG DERSAUERSTOFFTHERAPIE BEEINTRÄCHTIGEN.
WARNUNG: DURCH DIESES PRODUKT KÖNNEN
SIE MIT CHEMIKALIEN EINSCHLIESSLICH NICKELIN KONTAKT KOMMEN, WELCHES IM STAATKALIFORNIEN ALS KREBSVERURSACHEND GILT.
WEITERE INFORMATIONEN ERHALTEN SIE AUF WWW.
P65WARNINGS.CA.GOV.
VORSICHT: Ziehen Sie den WechselstromStecker immer zuerst aus der Steckdose,
bevor Sie das Wechselstrom-Kabel vom
Sauerstoffkonzentrator trennen.
VORSICHT: Platzieren Sie den Sauerstoffkonzentrator
immer so, dass das Stromkabel gut erreichbar ist.

FreeStyle Comfort
VORSICHT: Die Sauerstoffversorgungsleitung und das
Stromkabel immer so platzieren, dass niemand darüber
stolpern oder sich damit strangulieren kann.
VORSICHT: Verwenden Sie den Sauerstoffkonzentrator
in einem Automobil, Boot, oder bei anderen
Gleichstromquellen, stellen Sie sicher, dass der Motor
des Fahrzeuges angelassen ist und läuft, bevor Sie den
Sauerstoffkonzentrator anschließen. Nichtbeachtung
dieser Anweisungen kann dazu führen, dass der
Sauerstoffkonzentrator keinen Strom erhält.
VORSICHT: Wenn das Automobil, in dem Sie den
Sauerstoffkonzentrator verwenden, nicht läuft, trennen
und entfernen Sie das Gerät vom Automobil. Lagern
Sie den Sauerstoffkonzentrator nicht in einem sehr
heißen oder sehr kalten Fahrzeug oder in ähnlichen
Umgebungen, die extreme Temperaturen aufweisen.
Hinweis: Tragbare und mobile HFKommunikationsausrüstung kann medizinische elektrische
Geräte beeinträchtigen.
VORSICHT: Lassen Sie den
Sauerstoffkonzentrator oder das
Stromkabel NICHT eingesteckt, wenn die
Zündung AUSgeschaltet ist. Es kann dabei
zu einer Entladung der Fahrzeugbatteriekommen.
VORSICHT: Wird der Sauerstoffkonzentrator außerhalb
seines normalen Betriebstemperaturbereichs
und des Umgebungsdruckbereichs (Höhenlage)
betrieben oder gelagert, kann dies die Leistung
beeinträchtigen, die Akku-Laufzeit verringern und/oderdie Wiederaufladezeit des Akkus erhöhen. Lagerungsund Temperaturgrenzwerte finden Sie im Abschnitt„Technische Daten“ in dieser Anleitung.
VORSICHT: Für Sauerstoffkonzentratoren, die mit
Akkus ausgestattet sind: Lagern Sie die Akkus
kühl und trocken. Auf diese Weise kann die
Lebensdauer der Akkus verlängert werden. Wird
der Sauerstoffkonzentrator längere Zeit bei hoher
Temperatur bzw. mit einem voll geladenen/ganz leeren
Akku gelagert, kann dies die Haltbarkeit des Akkus
beeinträchtigen. Versuchen Sie nicht, den Akku zuöffnen. Er enthält keine wartbaren Teile. Halten Sie
Kinder vom Akku fern.
VORSICHT: VERWENDEN SIE NUR die vom Hersteller
bereitgestellten Akkus. Wenden Sie sich fürAnweisungen zur ordnungsgemäßen Entsorgung des
Akkus an den Bereitsteller Ihres Gerätes oder eineörtliche zuständige Stelle.

Hinweis: Sicherstellen, dass die Kanüle vollständig
eingeführt ist und fest sitzt. Damit wird sichergestellt, dass
der Sauerstoffkonzentrator ein Einatmen korrekt erkennen
und Sauerstoff liefern kann. Während des Einatmens sollten
Sie den Sauerstofffluss zu den Stutzen der Nasenkanüle
hören oder fühlen können.
Hinweis: Lithiumakkus können dauerhaft an Kapazität
verlieren, wenn sie im vollen oder völlig entleertem
Ladezustand extrem hohen Temperaturen ausgesetzt sind.
Bei längerer Lagerung wird empfohlen, dass die Akkus zu
25 bis 50 % geladen und bei Temperaturen von -20 °C – 20
°C (-4 °F – 68 °F) gelagert werden.
Hinweis: Der Einsatz einiger Sauerstoff führender
Zubehörteile, die nicht für die Verwendung mit diesem
Sauerstoffkonzentrator angegeben sind, kann dessen
Leistung beeinträchtigen. Auf empfohlenes Zubehör wird indieser Anleitung verwiesen.
Hinweis: Betreiben Sie den Sauerstoffkonzentrator nicht
ohne eingesetzten Lufteinlassfilter. Sollte ein zweiter Filter
mitgeliefert worden sein, setzen Sie den Ersatzfilter ein,bevor Sie den verschmutzten Filter reinigen. Reinigen Sie
den verschmutzten Filter in warmem Seifenwasser undtrocknen Sie ihn vor der Verwendung gründlich ab.
Hinweis: Der Hersteller empfiehlt, dieses Gerät nicht zu
sterilisieren.
Hinweis: Der Sauerstoff innerhalb eines Raumes kann nicht
aufgebraucht werden, wenn Sie den Sauerstoffkonzentrator
verwenden.
WARNUNG: DER
SAUERSTOFFKONZENTRATORSOLLTE NICHT BETRIEBEN ODERAN EINE EXTERNE STROMQUELLE(WECHSELSTROM/GLEICHSTROM)ANGESCHLOSSEN WERDEN, WENN DERAKKU NICHT EINGELEGT IST. VERSUCHENSIE NICHT, DAS GERÄT EINZUSCHALTENODER AN EINE EXTERNE STROMQUELLEANZUSCHLIESSEN, WENN DER AKKUPACKENTFERNT WURDE.
ENTFERNEN SIE DEN AKKUPACK NUR, WENNDAS GERÄT AUSGESCHALTET UND VON DER
STROMVERSORGUNG GETRENNT WURDE.
VERSUCHEN SIE NICHT, DEN AKKUPACK ZUENTFERNEN, WENN DAS GERÄT NOCH AN ODER AN
DIE STROMVERSORGUNG ANGESCHLOSSEN IST.
Hinweis: Wurde der Sauerstoffkonzentrator für einen
längeren Zeitraum außerhalb der üblichen und in derTabelle oben beschriebenen Betriebsbedingungen gelagert,
sollte dem Gerät vor dem Anschalten Zeit gelassen werden,
eine übliche Betriebstemperatur zu erreichen.

PN MN232-C4 K | Bedienungsanleitung

GER - 71

FreeStyle Comfort

Technische Daten

FreeStyle Comfort
Abmessungen (mit Einzelakku)

25,4 x 18,5 x 7,9 cm

Abmessungen (mit Doppelakku)

27,9 x 18,5 x 7,9 cm

Gewicht (mit Einzelakku)

2,3 kgGewicht (mit Doppelakku)

2,7 kgNenngeräuschpegel***

39,93 dB (A) ± 0,42 dB (A) bei 2

Volumenstrom-Einstellungen

Impulseinstellung 1, 2, 3, 4, 5

O2 Ausgabe-Einstellung

Einstellung 1 - 210 mL/min
Einstellung 2 - 420 mL/min
Einstellung 3 - 630 mL/min
Einstellung 4 - 840 mL/min
Einstellung 5 - 1050 mL/min(@ 20 Atemzüge pro Minute)

Sauerstoffkonzentration*
Sauerstoffmonitor
Maximaler AusgangsdruckStandard-Dosierempfindlichkeit

90 % (+5,5 %/-3 %)Ja< 2 bar (30 PSI)
> -0,1 cm H2O (einstellbar)

Wechselstrom

100–240 V, 50–60 Hz, 120W Max

Gleichstrom

11–18 V (max. 10 Ampere)

Wiederaufladezeit der Akkus

Einzelakku: 3,5 Stunden
Doppelakku: 6 Stunden

Laufzeiten der Akkupacks

Laufzeit des Einzelakku: Einstellung 2 - bis zu 4 Stunden
Laufzeit des Doppelakku: Einstellung 2 - bis zu 8 Stunden

Betriebstemperatur**

5 °C bis 40 °C

Batterietyp

Lithium-Ionen

Akku-Kapazitäten

Einzelakku: 6700 mAh; Doppelakku: 13400 mAh

Temperatur für Transport und Lagerung (Gerät)-25 °C bis +70 °C bis zu 3048 m

Temperatur für Transport und Lagerung (Akku)-25 °C bis +70 °C bis zu 3048 m

Einsatzhöhe**–381 bis 3.048 m (–1.250 bis 10.000 ft) (getestet bei 700–1.060 hPa)

Relative Luftfeuchtigkeit

15–95 % relative Luftfeuchtigkeit (nicht kondensierend)

Lagerungsfeuchtigkeit

0% - 90% nicht kondensierend

Beschränkte Garantie

3 Jahre auf das Gerät
1 Jahr auf Akku und Netzteile* Basierend auf einem atmosphärischen Druck von 101 kPa bei 21 °C** Das Gerät funktioniert in allen Umgebungen, solange die Umgebung den angegebenen Temperatur- und Luftfeuchtigkeitsbereichen
entspricht. Ein Betrieb außerhalb dieser Spezifikationen kann die Fähigkeit des Konzentrators einschränken, die Spezifikation der
Sauerstoffkonzentration bei höheren Literdurchflussmengen zu erfüllen. Der Aufwärmbildschirm kann weiterhin für bis zu 15 Minuten
angezeigt werden oder bis der Konzentrator die Zielkonzentration erreicht hat. Das Gerät kann während dieser Zeit weiterhin verwendet
werden.
*** Schallpegel gemessen nach Prüfmethode Nr. 14-1 10/2018 MDS-Hi.
Die voraussichtliche Nutzungsdauer beträgt mindestens fünf Jahre. Die voraussichtliche Nutzungsdauer der Siebböden beträgt ca. ein
Jahr, wenn der Konzentrator während des gesamten Zeitraums ungenutzt bleibt.

72 - GERPN MN232-C4 K | Bedienungsanleitung

FreeStyle Comfort
Gebrauchsanweisungen
1. Platzieren und positionieren Sie den FreeStyle
Comfort an einem für die Nutzung geeigneten
Standort.
a. Der Konzentrator sollte gut belüftet sein, damit
die Lufteinlässe und -auslässe nicht blockiert
werden.
b. Der Konzentrator sollte so positioniert werden,
dass alle akustischen und optischen Indikatoren
und Alarme leicht gehört und gesehen werden
können.
c. Vergewissern Sie sich, dass die Lufteinlassfilter
angebracht sind, bevor Sie das Gerät in Betrieb
nehmen. Wenn für das Gerät ein Filter fehlt,
wenden Sie sich bitte an Ihren Geräteanbieter.
2. Legen Sie einen Akkupack in den Konzentrator ein.
a. Richten Sie den Akku mittig auf die Unterseite
des Gehäuses Ihres FreeStyle Comfort aus.

b. Schieben Sie den Akku in das Gehäuse, bis er mit
einem Klicken einrastet.

3. Stecken Sie das Wechselstrom- oder GleichstromNetzteil in den Eingang für den Stromanschluss unten
rechts an der Vorderseite des Geräts.

4. Schließen Sie eine Nasenkanüle an die Spitze desSauerstoffauslasses an. Vergewissern Sie sich vor derInbetriebnahme, dass die Verbindung stabil ist.
HINWEIS: Das Verwenden von anderen als den
empfohlenen Kanülen kann die Leistung beeinträchtigen.

Hinweis: An den Geräteanbieter: Das folgende Zubehör für
die Sauerstoffverabreichung wird zur Verwendung mit dem
FreeStyle Comfort empfohlen:• Nasenkanüle mit 7 Fuß (2,1 m) Schlauch (maximale LPM:
6 LPM): CAIRE-Teilenummer 5408-SEQ• Weitere Informationen zu empfohlenem Zubehör finden
Sie online unter www.caireinc.com.

PN MN232-C4 K | Bedienungsanleitung

GER - 73

FreeStyle Comfort
5. Schalten Sie den FreeStyle Comfort ein, indem
Sie den Ein-/Aus-Schalter drücken. Wählen Sie die
Durchflusseinstellung, die Ihnen verschrieben wurde,
indem Sie auf Erhöhen (+) oder Reduzieren (-)
drücken.

autoSAT Impulsvolumeneinstellung (mL)*Atemfrequenz

1

2

3

4

5

15

10,5

21,0

31,5

42,0

52,5

20

10,5

21,0

31,5

42,0

52,5

25

10,5

21,0

31,5

42,0

42,0

30

10,5

21,0

31,5

35,0

35,0

35

10,5

21,0

30,0

30,0

30,0

10,5

21,0

26,3

26,3

26,3

40

*Die Toleranz beträgt +/- 15 %nicht-autoSAT Impulsvolumeneinstellung(mL)*
Ein kurzer akustischer Alarm ertönt, um anzuzeigen,dass das Gerät eingeschaltet ist.
Hinweis: Wenn der FreeStyle Comfort zum ersten Mal
eingeschaltet wird, ist standardmäßig die VolumenstromEinstellung 2 eingestellt. Wenn die Einstellung
geändert wird, schaltet sich das Gerät auf die vorherige
Durchflusseinstellung ein.
6. Positionieren Sie die Nasenkanüle entsprechend derAnleitung des Herstellers, die der Kanüle beiliegt,
in Ihrem Gesicht und atmen Sie normal. Bei jedem
Einatmen müssten Sie spüren, wie ein Sauerstoffstrom
durch die Nasenkanüle fließt.
7. Um den Durchfluss auf die gewünschte Einstellungzu bringen, drücken Sie auf Erhöhen (+) oder
Reduzieren (-). Es ist normal, dass sich das Geräusch
bei Änderung der Durchflusseinstellungen ändert.

Atemfrequenz

1

2

3

4

5

15

14,0

28,0

20

10,5

21,0

42,0

56,0

70,0

31,5

42,0

52,5
42,0

25

8,4

16,8

25,2

33,6

30

7,0

14,0

21,0

28,0

35,0

35

6,0

12,0

18,0

24,0

30,0

40

5,3

10,5

15,8

21,0

26,3

8. Schalten Sie den FreeStyle Comfort aus, indem Sie
den Netzschalter drücken.
Hinweis: Der Bildschirm wechselt in den Standby-Modus
und wird nach 20 Sekunden abgedunkelt. Drücken Sie eine
beliebige Taste, um den Bildschirm aus dem Standby-Modus zu wecken.
9. Weitere Funktionen finden Sie in den Anweisungenzur Menüführung

SPEZIELLE FUNKTIONEN DES PRODUKTS
autoSAT®
Mit der proprietären autoSAT-Funktion von CAIRE kann
der FreeStyle Comfort ein konstantes Sauerstoffvolumen(Bolus) pro Atemzug bis zu einer maximalen
Atemfrequenz liefern. Wenn sich die Atemfrequenz
des Benutzers bei Aktivitäten wie Gehen erhöht, passt
die autoSAT-Funktion den Gerätebetrieb an, um ein
eingestelltes Bolusvolumen beizubehalten.
UltraSense®
Dieser empfindliche Atemimpuls-Trigger erkennt
Druckveränderungen während des Atmens und führt
Sauerstoff beim Einatmen zu.
autoDOSE™
Falls ein Apnoe-Alarm (Atempausen-Alarm) auftritt,
führt diese Funktion bei 20 Atemzügen pro Minute
Sauerstoff in der am Gerät eingestellten Flussrate zu, bis
die Alarmbedingung behoben ist.

74 - GERPN MN232-C4 K | Bedienungsanleitung

FreeStyle Comfort
Stromversorgung
Der FreeStyle Comfort kann auf 3 Arten betrieben
werden:• Wiederaufladbarer Akkupack• Wechselstrom• Gleichstrom
Informationen zum Betrieb dieser Komponenten finden
Sie in den folgenden Abschnitten.
WARNUNG: VERWENDEN SIE NUR
DIE IN DER BEDIENUNGSANLEITUNG
AUFGEFÜHRTEN STROMQUELLEN. BEI
DER VERWENDUNG VON STROMQUELLEN
UND AKKUPACKS, DIE NICHT VON CAIREHERGESTELLT WURDEN, KANN DASGERÄT BESCHÄDIGT ODER DER BETRIEB
BEEINTRÄCHTIGT WERDEN.

ENTFERNEN DES AKKUS

Drücken Sie den Akku-Entriegelungsknopf nach
unten und weg vom FreeStyle Comfort. HALTEN Sie
diesen Knopf gedrückt, während Sie den Akku durch
Wegdrücken vom Konzentrator herausnehmen.
WARNUNG: DER
SAUERSTOFFKONZENTRATOR MUSSVOR DEM ENTFERNEN DES AKKUPACKSAUSGESCHALTET UND VON DER
EXTERNEN STROMVERSORGUNGGETRENNT WERDEN.

AKKUPACK

Der FreeStyle Comfort kann mit dem mitgelieferten
wiederaufladbaren Akkupack betrieben werden. Wird
der Konzentrator mit dem Akkupack verwendet, gibt es
keine betrieblichen Einschränkungen. Bei Akkubetrieb
sind alle Flussraten verfügbar.

ÜBERPRÜFEN DES AKKUPACKLADEZUSTANDSEINSETZEN DES AKKUS

Richten Sie den Akku mittig auf die Unterseite desGehäuses Ihres FreeStyle Comfort aus. Drücken Sie den
Akku hinein, bis er mit einem Klicken einrastet.

Wenn Sie den Ladestand des Akkus überprüfen möchten,
während dieser im FreeStyle Comfort eingelegt ist,sehen Sie sich das Batteriesymbol und die geschätzte
verbleibende Zeit auf dem Display an.
80 bis 100 %60 bis 80 %40 bis 60 %20 bis 40 %20 %PN MN232-C4 K | Bedienungsanleitung

GER - 75

FreeStyle Comfort
Hinweis: Wenn der Akku gerade aufgeladen wird, istanstelle des verbleibenden Ladeprozentsatzes eine
Steckeranzeige zu sehen.
Wenn Sie den Akku-Ladezustand überprüfen möchten,
während sich der Akku nicht im FreeStyle Comfort
befindet, legen Sie den Akku flach hin und drücken Sie
die TEST-Taste. Die Akku-Ladeleuchten (25–100 %)leuchten rechts neben der TEST-Taste auf und geben den
Ladezustand des Akkupacks an.

ANFÄNGLICHE LADUNG

Der neue Akku Ihres FreeStyle Comfort ist bei
Auslieferung nur teilweise geladen. Vor der ersten
Verwendung sollten Sie den Akkupack mit Wechseloder Gleichstrom vollständig aufladen.

ALLGEMEINE INFORMATIONEN ZUM
AUFLADEN

Der FreeStyle Comfort lädt den Akkupack immer
dann auf, wenn er sich im Konzentrator befindet
und der Konzentrator an eine externe Wechsel- oder
Gleichstromquelle angeschlossen ist (außer im
Flugzeug).
Der Akku ist vollständig geladen, wenn er 100 %anzeigt.
Alternativ kann der Akkupack mit der optionalen
Desktop-Aufladeeinheit geladen werden.

76 - GERPN MN232-C4 K | Bedienungsanleitung

AKKU-LAUFZEIT

Wird die externe Wechsel- oder Gleichstromversorgung,unterbrochen schaltet der Konzentrator automatischauf Akkubetrieb, wenn der Akkupack eingelegt undaufgeladen ist. Die folgende Tabelle zeigt die typischen
Laufzeiten eines neuen Akkupacks. Die Akkulaufzeit
bleibt unabhängig von der Atemfrequenz konstant.
Impulseinstellung

Laufzeit desEinzelakku(8-Zellen)

Laufzeit desDoppelakku(16-Zellen)

1

bis zu8 Stunden

bis zu16 Stunden2

bis zu4 Stunden

bis zu8 Stunden

3

bis zu3 Stunden

bis zu6 Stunden

4

bis zu2,25 Stunden

bis zu4,5 Stunden

5

bis zu2 Stunden

bis zu4 Stunden

TYPISCHE AKKU-LADEZEIT

Die benötigte Zeit zum vollständigen Aufladen eines
vollständig entladenen Akkus beträgt 3,5 Stunden füreinen Einzelakku und 6 Stunden für einen Doppelakku.
Hinweis: Wenn der Akku vollständig entladen ist, beginnt
der Ladevorgang möglicherweise nicht sofort.

AKKU-LEBENSDAUER

Der Akku des FreeStyle Comfort ist für 300 Lade- und
Entladezyklen ausgelegt.

FreeStyle Comfort
Wechselstrom- undGleichstromversorgung
Der FreeStyle Comfort wird standardmäßig mit einem
Wechsel- oder Gleichstromnetzteil geliefert.
Bestandteile der Wechselstromversorgung:a. Wechselstrom-Netzteil, das zwischen
100–240 Volt und 50–60 Hz, betrieben wird.
b. Wechselstromkabel zum Anschluss desNetzteils an die entsprechende Steckdose.

Beim Gleichstromnetzteil handelt es sich um ein Kabel,das direkt an das Gerät und an die Netzsteckdose
angeschlossen wird.

2. Verbinden Sie das Wechselstromkabel mit dem
Wechselstrom-Netzteil.

WECHSELSTROMVERSORGUNG

Das Wechselstrom-Netzteil ist für den Hausgebrauchoder ähnliche Umgebungen mit standardmäßigen
Wechselstrom-Steckdosen vorgesehen. Der FreeStyle
Comfort ist mit Wechselstrom voll funktionsfähig. Der
FreeStyle Comfort lädt den Akkupack zudem immer
dann auf, wenn Wechselstrom zur Verfügung steht (außer
im Flugzeug). Die Aufladung des Akkupacks erfolgt
unabhängig davon, ob der FreeStyle Comfort ein- oderausgeschaltet ist.
WARNUNG: SCHLIESSEN SIE KEINE
EXTERNE STROMVERSORGUNG AN,
WENN SICH DER AKKUPACK NICHT IMSAUERSTOFFKONZENTRATOR BEFINDET.

3. Schließen Sie das Steckerende desWechselstromkabels an eine Wandsteckdose oder eine
andere geeignete Steckdose an.

Um den FreeStyle Comfort an die
Wechselstromversorgung anzuschließen:
1. Verbinden Sie das Wechselstrom-Netzteil mit derNetzanschlussbuchse unten rechts an der Vorderseite
des FreeStyle Comfort. Führen Sie den Stecker bis
zum Anschlag ein, um eine sichere Stromverbindungzu gewährleisten.

PN MN232-C4 K | Bedienungsanleitung

GER - 77

FreeStyle Comfort
4. Schalten Sie den FreeStyle Comfort ein, indem
Sie den Ein-/Aus-Schalter drücken. Wählen Sie die
Durchflusseinstellung, die Ihnen verschrieben wurde,
indem Sie auf Erhöhen (+) oder Reduzieren (-)
drücken.

Hinweis: Wenn der Akkupack wieder aufgeladen werdenmuss, wird das Batteriesymbol auf dem Display mit einem
orangefarbenen Balken angezeigt.

GLEICHSTROM

Für den Betrieb über Steckdosen in Autos, Booten
und anderen Kraftfahrzeugen benötigen Sie das DCNetzteil. Der FreeStyle Comfort ist mit Gleichstromvoll funktionsfähig. Der FreeStyle Comfort lädt den
Akkupack bei Bedarf auch dann wieder auf, wenn
Gleichstrom zur Verfügung steht. Der Akkupack
wird sowohl bei ausgeschaltetem Gerät als auch bei
eingeschaltetem und in Betrieb befindlichem Gerät
aufgeladen.
WARNUNG: SCHLIESSEN SIE KEINE
EXTERNE STROMVERSORGUNG AN,
WENN SICH DER AKKUPACK NICHT IMSAUERSTOFFKONZENTRATOR BEFINDET.
WARNUNG: BETREIBEN SIE DAS GERÄT NICHTAN EINER 12 V AUTOSTECKDOSE, WÄHREND DASFAHRZEUG AUSGESCHALTET IST.
1. Starten Sie Ihr Kraftfahrzeug.
2. Verbinden Sie das Gleichstrom-Netzteil mit dem
Eingang für den Stromanschluss unten rechts an
der Vorderseite des FreeStyle Comfort. Führen Sie
den Stecker bis zum Anschlag ein, um eine sichere
Stromverbindung zu gewährleisten.

78 - GERPN MN232-C4 K | Bedienungsanleitung

3. Schließen Sie das Steckerende des GleichstromNetzkabels an die Autosteckdose oder an eine andere
geeignete Gleichstromquelle an.

4. Schalten Sie den FreeStyle Comfort ein, indem
Sie den Ein-/Aus-Schalter drücken. Wählen Sie die
Durchflusseinstellung, die Ihnen verschrieben wurde,
indem Sie auf Erhöhen (+) oder Reduzieren (-)
drücken.
Hinweis: Wenn der Akkupack wieder aufgeladen werdenmuss, wird das Batteriesymbol auf dem Display mit einem
orangefarbenen Balken angezeigt.

FreeStyle Comfort
Reinigung, Pflege und regelmäßige
Wartung
REGELMÄSSIGE WARTUNG

Vorbeugende Wartung
Das Gerät ist wartungsfrei. Filter im Gerät sollen die
Lebensdauer des Geräts halten. Wenn das Gerät nicht
ordnungsgemäß funktioniert, muss das Gerät zur
Reparatur an den Anbieter zurückgeschickt werden.
Wartungsarbeiten an den internen Komponenten des
FreeStyle Comfort müssen vom Sauerstoff-Versorger
durchgeführt werden.
Der Benutzer des Gerätes ist für die folgenden
Wartungsarbeiten zuständig.
• Wöchentlich – Reinigung der Luftansaugfilter
WARNUNG: VERSUCHEN SIE NICHT,DIE KUNSTSTOFFABDECKUNGENDES SAUERSTOFFKONZENTRATORSZU ÖFFNEN ODER ZU ENTFERNEN.
DAS GERÄT SOLLTE NUR VON EINEMQUALIFIZIERTEN SERVICETECHNIKER
GEÖFFNET WERDEN.

LUFTANSAUGFILTER

Der FreeStyle Comfort nimmt Luft über die
Luftansaugfilter auf, die sich unten auf der Vorderseite
des Sauerstoffkonzentrators befinden. Dieser Filter
verhindert das Eindringen von Staub und anderen
größeren Partikeln in das Gerät.
Die Luftansaugfilter des Geräts sollte mindestens einmal
pro Woche gereinigt werden. Ihr Sauerstoff-Versorgerkann Ihnen, je nach Einsatzbedingungen des Geräts,eine häufigere Filterreinigung empfehlen. Führen Sie
die folgenden Schritte aus, um den Luftansaugfilter
sachgemäß zu reinigen.
1. Entfernen Sie die beiden Filter von der Vorderseite
des FreeStyle Comfort, indem Sie die entsprechende
Lasche nach oben drücken und nach außen ziehen.

WARNUNG: BETREIBEN SIE DENSAUERSTOFFKONZENTRATORNICHT OHNE EINGESETZTENLUFTEINLASSFILTER.

RICHTIGER UMGANG MIT DEM AKKUPACK

Vermeiden Sie, dass der Akkupack mit Flüssigkeiten in
Berührung kommt.
Versuchen Sie nicht, den Akkupack zu öffnen oder zu
zerlegen.
Setzen Sie den Akkupack keinen offenen Flammen aus.
Lassen Sie den Akkupack nicht fallen.
Bewahren Sie den Akkupack außerhalb der Reichweite
von Kindern auf.
Wenn Ihr Akkupack beschädigt erscheint, kontaktieren
Sie Ihren Sauerstoff-Versorger vor einem erneuten
Gebrauch.
Halten Sie den Akkupack von Mikrowellen und anderen
Kochgeräten fern.

LAGERUNG DES AKKUPACKS

Der Akkupack des Sauerstoffkonzentrators muss vor
dem Aufladen nicht vollständig entladen werden. Es wird
empfohlen, den Akkupack des Sauerstoffkonzentrators
nach jedem Gebrauch aufzuladen.
Lagern Sie den Akkupack nicht an Orten, an denen
er starker Hitze ausgesetzt ist, wie z. B. in direktem
Sonnenlicht oder im Auto. Andernfalls können Leistungund Lebensdauer des Akkupacks stark beeinträchtigt
werden. Im schlimmsten Fall kann der Akkupack Hitze
und Rauch entwickeln und ggf. Feuer fangen oder
explodieren. Der Akkupack sollte in einer kontrollierten
Umgebung mit einer relativen Luftfeuchtigkeit unter
70 % und frei von korrosiven Gasen gelagert werden.
Um Rost vorzubeugen, vermeiden Sie Bedingungen, die
Kondensation verursachen können, wie zum Beispiel
große Temperaturschwankungen.
Für eine Lagerungszeit von einem Monat oder mehr wird
empfohlen, den Akkupack im vollständig entladenen
oder teilweise geladenen (50 % bis 75 % – zwei bis dreiLEDs) Zustand zu lagern. Entfernen Sie den Akkupack
aus dem Sauerstoffkonzentrator, sobald er vollständig
geladen ist.

2. Reinigen Sie beide Filter in einer Lösung aus warmem
Wasser und Seife.
3. Spülen Sie die Filter gründlich aus und entfernen Sieüberschüssiges Wasser mit einem weichen, fusselfreien
Tuch. Stellen Sie vor Wiedereinbau sicher, dass die
Filter vollständig trocken sind.
4. Setzen Sie die sauberen und trockenen Filter
wieder ein.

PN MN232-C4 K | Bedienungsanleitung

GER - 79

FreeStyle Comfort
REINIGUNG

Schalten Sie das Gerät ab und trennen Sie es von derStromversorgung (Gleich- oder Wechselstrom), bevor
Sie es reinigen. Sprühen Sie das Gehäuse NICHT
direkt ein. Verwenden Sie ein feuchtes (nicht nasses)
Tuch oder einen Schwamm. Besprühen Sie dasTuch bzw. den Schwamm zunächst mit einer milden
Desinfektionslösung und reinigen Sie dann das Gehäusesowie die Komponenten der Stromversorgung. Gehen
Sie gemäß den Herstelleranweisungen vor.
WARNUNG: ZIEHEN SIE DAS NETZKABELAUS DER STECKDOSE, BEVOR DASGERÄT GEREINIGT WIRD, UM DIE GEFAHREINES STROMSCHLAGS ODER EINERVERBRENNUNG AUSZUSCHLIESSEN.
NUR DER GERÄTEANBIETER ODER EINQUALIFIZIERTER SERVICETECHNIKER
DÜRFEN DIE ABDECKUNG ENTFERNENODER DAS GERÄT WARTEN.
WARNUNG: ES MUSS DARAUF GEACHTET WERDEN,DASS DER SAUERSTOFFKONZENTRATOR STETSTROCKEN BLEIBT UND DASS KEINE FLÜSSIGKEITIN DAS GERÄT GELANGT. ANDERNFALLS KANN
EINE FEHLFUNKTION AUFTRETEN ODER EINEABSCHALTUNG ERFOLGEN UND ES BESTEHT EIN
ERHÖHTES RISIKO FÜR EINEN ELEKTRISCHENSCHLAG UND FÜR VERBRENNUNGEN.
WARNUNG: SETZEN SIE DIREKT AUF DEM GERÄTKEINE FLÜSSIGKEITEN EIN. DIE LISTE DER
UNERWÜNSCHTEN CHEMISCHEN WIRKSTOFFE
BEINHALTET FOLGENDE FLÜSSIGKEITEN, IST
JEDOCH NICHT AUF DIESE BESCHRÄNKT: ALKOHOLUND ALKOHOLISCHE PRODUKTE, KONZENTRIERTECHLORHALTIGE PRODUKTE (ETHYLENCHLORID)
UND PRODUKTE AUF ÖLBASIS. EINE REINIGUNG MIT
DIESEN PRODUKTEN KANN DAS PLASTIKGEHÄUSE
NACHHALTIG BESCHÄDIGEN.
WARNUNG: DAS GEHÄUSE, BEDIENFELDUND NETZKABEL MIT EINEM SCHONENDEN
HAUSHALTSREINIGER REINIGEN. DAZU EIN
FEUCHTES (ABER KEIN NASSES) TUCH ODEREINEN FEUCHTEN SCHWAMM VERWENDEN.
ALLE BESTANDTEILE NACH DER REINIGUNGABTROCKNEN. KEINE FLÜSSIGKEIT IN DAS GERÄT
EINDRINGEN LASSEN. BESONDERS DARAUF
ACHTEN, DASS DER SAUERSTOFFAUSLASS FÜR
DEN ANSCHLUSS DER KANÜLE FREI VON STAUB,
WASSER UND ANDEREN PARTIKELN BLEIBT.

80 - GERPN MN232-C4 K | Bedienungsanleitung

WARNUNG: VERWENDEN SIE EIN FEUCHTES (ABER
KEIN NASSES) TUCH ODER EINEN FEUCHTENSCHWAMM, UM DEN AKKUPACK ZU REINIGEN.
SPRÜHEN SIE DAS TUCH BZW. DEN SCHWAMM ERSTMIT EINEM SCHONENDEN HAUSHALTSREINIGER
EIN UND REINIGEN SIE ANSCHLIESSEND DENAKKUPACK.
WARNUNG: REINIGEN SIE DAS GERÄT NICHT MIT
EINEM AUTOMATISCHEN REINIGUNGSSYSTEM.

ERSETZEN DER KANÜLE

Für einen ordnungsgemäßen Gebrauch stets den
Anleitungen des Kanülenherstellers folgen. Ersetzen
Sie die Nasenkanüle oder den Sauerstoffschlauch alle
3 bis 6 Monate (bei täglicher Anwendung alle 3 Monate)entsprechend den Empfehlungen des Kanülenherstellers
oder Ihres Sauerstoff-Versorgers. Informationen zur
Reinigung und zum Austausch erhalten Sie von Ihrem
Arzt oder Pflegedienst.
Weitere Ersatzteile erhalten Sie von Ihrem SauerstoffVersorger.

FreeStyle Comfort
Warnzustände
Prioritätsnummer
1

Warnbeschreibung
Systemstörung
Hohe AkkuTemperatur

Anzeige

Akustisches
Signal1 Signaltonalle
20 Sekunden

2

Systemstörung
hohe Temperatur

1 Signaltonalle
20 Sekunden

3

Systemstörung
hoher Druck

1 Signaltonalle
20 Sekunden

4

Keine Atmung
erkannt

1 Signaltonalle
30 Sekunden5

Abschaltung
bei schwachem
Akku

1 Signaltonalle
30 Sekunden

6

Aufwärmen

Keine

Weitere Informationen/Empfohlene
Maßnahme
Das Gerät warnt, wenn sich die Temperatur
des Akkus dem Grenzwert für die
Sicherheitsabschaltung nähert.
Wenn die Abschaltschwelle erreicht ist,schaltet das Gerät ab. Wenn eine externe
Stromversorgung angeschlossen ist, wird derWarnbildschirm angezeigt und das akustische
Warnsignal wird für 2 Minuten ausgegeben.
Testen Sie mit dem Akku bei Zimmertemperatur.
Wenn das Problem weiterbesteht, wenden Sie
sich an den Geräteanbieter.
Das Gerät warnt, wenn sich die interne
Temperatur dem Grenzwert für die
Sicherheitsabschaltung nähert.
Wenn die Abschaltschwelle erreicht ist, schaltet
sich das Gerät ab, der Warnbildschirm wird
angezeigt und das akustische Warnsignal wirdfür 2 Minuten ausgegeben.
Dafür sorgen, dass die Abluftöffnungen desGeräts nicht blockiert sind. Wenn das Problem
weiterbesteht, wenden Sie sich an den
Geräteanbieter.
Wenn die Abschaltschwelle erreicht ist, schaltet
sich das Gerät ab, der Warnbildschirm wird
angezeigt und das akustische Warnsignal wirdfür 2 Minuten ausgegeben.
Wenn das Problem weiterbesteht, wenden Sie
sich an den Geräteanbieter.
Es wurde 15 Sekunden lang kein Atemzug
festgestellt.
Wird nach dem ersten Atemzug oder nach derAbschaltung des Aufwärmbildschirms aktiviert,
je nachdem, was früher eintritt.
Überprüfen Sie die Kanüle auf Knickeoder Undichtigkeiten. Erhöhen Sie die
Empfindlichkeitseinstellungen. Wenn das
Problem weiterbesteht, wenden Sie sich an den
Geräteanbieter.
Warnmeldungen bis der Akku leer ist und das
Gerät heruntergefahren wird.
Laden Sie den Akku auf.

Das Gerät beginnt mit der automatischen
Dosierung, bis ein Atemzug erkannt wird oderdas Aufwärmen endet, je nachdem, was früher
eintritt.

PN MN232-C4 K | Bedienungsanleitung

GER - 81FreeStyle Comfort
Prioritätsnummer
7

Warnbeschreibung
Niedrige
Konzentration

Anzeige

Akustisches Weitere Einzelheiten
SignalWenn das Problem weiterbesteht, wenden Sie
1 Signaltonsich an den Geräteanbieter.
alle 30
Sekunden, fürbis zu 5 min

8Systemstörung
Durchflussmenge (Ausfall des
OCD-Ventils,
Blockierung desGasweges)

1 Signaltonalle
30 Sekunden

9

Warnung bei
schwachem
Akku

Laden Sie den Akku auf.
1 Signalton,
keine
Wiederholung

10

Atemfrequenzüberschritten

1 Signaltonalle
60 Sekunden

Atemfrequenz übersteigt 40 Atemzüge pro
Minute.

Wenn das Problem weiterbesteht, wenden Sie
sich an den Geräteanbieter.

Wird eine falsche Dosis abgegeben, reduzieren
Sie die Empfindlichkeitseinstellungen. Wenndas Problem weiterbesteht, wenden Sie sich an
den Geräteanbieter.

11

Systemstörung
Lüfterausfall

Keine

Wenn das Problem weiterbesteht, wenden Sie
sich an den Geräteanbieter.

12

Systemstörung
niedriger Druck

Keine

Wenn das Problem weiterbesteht, wenden Sie
sich an den Geräteanbieter.

13

Systemstörung
niedrige
Temperatur

Keine

Geben Sie dem Gerät Zeit, sich auf
Zimmertemperatur zu erwärmen. Wenn das
Problem weiterbesteht, wenden Sie sich an den
Geräteanbieter.

Info

Inbetriebnahme

1 InfoSignalton

Der Startbildschirm wird beim Einschaltenangezeigt und Sie sollten einen einzelnen
Signalton hören, der die Funktion desSignalgebers bestätigt.
Der Startbildschirm zeigt die aktuelle FirmwareVersion in der rechten unteren Ecke an.

82 - GERPN MN232-C4 K | Bedienungsanleitung

FreeStyle Comfort
Prioritätsnummer
Info

Warnbeschrei- AnzeigebungNormalanzeige
mit autoSATModus (nicht
an eine Gleichstromquelleangeschlossen)
Normalanzeige
mit autoSAT
deaktiviert (nicht
an eine Gleichstromquelleangeschlossen)

Akustisches
SignalKeine

Info

Einstellung
geändert

1 InfoSignalton

Info

Externe
Stromversorgung imausgeschalteten
Zustandeinstecken.

Keine

Info

Externe
Stromversorgung im
Betriebszustandeinstecken.

Keine

Info

Externe
Stromversorgung
im ausgeschalteten Zustandtrennen.

1 InfoSignalton

Info

Externe
Stromversorgung im
Betriebszustandtrennen (beivorhandenem
Akku).

1 InfoSignalton

Info

Weitere Einzelheiten

Keine

Anzeige des CAIRE-Bildschirms für 2
Sekunden, gefolgt von Akku- und Steckerstatus
für 5 Sekunden, bevor der Bildschirm
ausgeschaltet wird.

Anzeige des Caire-Bildschirms für 2 Sekunden
und anschließend Abschaltung der Anzeige.

PN MN232-C4 K | Bedienungsanleitung

GER - 83

FreeStyle Comfort
Prioritätsnummer
Info

Warnbeschrei- Anzeigebung
Im StandbyModus –wenn dieAufwärts- oder
Abwärtstaste
gedrücktwird, zeigt der
Bildschirm
den Akku- undLadezustand an.

Akustisches
SignalKeine

Info

Flugzeug-Modus
bei Akkubetrieb

1 langer
Signalton

Info

Flugzeug-Modus
bei Netzbetrieb

1 langer
Signalton

84 - GERPN MN232-C4 K | Bedienungsanleitung

Weitere Einzelheiten
Der Bildschirm wird 5 Sekunden lang angezeigt
und dann ausgeschaltet.

FreeStyle Comfort
Anweisungen zur Menüführung
Sie können auf das Menü im Standby- oder im
Betriebsmodus zugreifen. Tritt während desBetriebsmodus ein Alarmzustand auf, wird das Menü
automatisch verlassen. Wird innerhalb 20 Sekundenkeine Taste gedrückt, wird das Menü automatischverlassen. Um das Gerät aus dem Standby-Modus
aufzuwecken, müssen Sie die Menü-Taste eventuell
zweimal drücken. Dann können Sie auf das Menü
zugreifen.
Das Menü verfügt über vier Bildschirme:

Flugzeug-Modus

1. S
 tatus-Bildschirm: Zeigt die FirmwareVersionsnummer, Laufzeit, Akku-Entladezyklen an
2. Flugzeug-Modus (EIN/AUS)
3. Empfindlichkeit: einstellbar (1-5)
4. autoSAT (EIN/AUS)

Empfindlichkeit

Beispiel Status-Bildschirm

BETRIEBS-MENÜEINSTELLUNGEN
Drücken Sie im Betriebs- oder Standbymodus
auf die Taste Menü/Auswählen
, um auf die
Menüeinstellungen zuzugreifen.
Drücken Sie

autoSAT, um durch die Menüs zu scrollen.

Status-Seite

PN MN232-C4 K | Bedienungsanleitung

GER - 85

FreeStyle Comfort
STATUS SEITE
Auf der Statusseite des Freestyle Comfort werden
die Firmware-Version, die Betriebsstunden und die
Anzahl der Akkuzyklen des angeschlossenen Akkus
angezeigt.

Drücken Sie, während das Symbol SPEICHERN
angezeigt wird, um das Einschalten des Flugzeug-Moduszu bestätigen. Der Wert wechselt auf „EIN“ in schwarzer
Farbe.
Drücken Sie, während das Symbol LÖSCHEN
angezeigt wird, um das Einschalten des Flugzeug-Modusabzubrechen.

Status-Seite

Betriebsstunden werden in ganzen Stunden angezeigt.
Batterieentladung ist die Anzahl der Zyklen, in
denen die angeschlossene Batterie entladen undwieder aufgeladen wurde.

FLUGZEUG-MODUS
Der FreeStyle Comfort wird mit ausgeschaltetem
Flugzeug-Modus geliefert.

Flugzeug-Modus Ein Bestätigt

Wiederholen Sie diese Vorgänge und wechseln Sie auf
AUS, um den Flugzeug-Modus auszuschalten.
Indem Siedrücken, wenn auf dem Bildschirmdas Symbol „Weiter“ angezeigt wird, wechselt der
Bildschirm zur nächsten Menü-Option.
Der Flugzeug-Modus ist für den Gebrauch im
Betriebsmodus bestimmt. Sobald das Gerät sich imStandby-Modus oder Akkubetrieb befindet, wird der
Flugzeug-Modus automatisch deaktiviert.
Hinweis: Unter Flugzeug-Modus Aktivieren finden Sie
weitere Informationen zum Flugzeug-Modus.

EMPFINDLICHKEITS-MODUS

Drücken Sie, während das Symbol WECHSEL
angezeigt wird, um den Flugzeug-Modus zu aktivieren.
Der Wert wechselt auf „EIN“ in blauer Farbe.

Der FreeStyle Comfort wird vom Hersteller mit
einer standardmäßigen Empfindlichkeitseinstellung
von 2 geliefert und unter normalen Bedingungensollte keine andere Einstellung erforderlich sein. Die
Empfindlichkeitseinstellung reicht von den Werten 1-5,
wobei 1 für die geringste Empfindlichkeit steht und 5 fürdie höchste Empfindlichkeit.

Flugzeugmodus Ein Ausgewählt

Empfindlichkeitsbildschirm bei Einstellungswert 2

Hauptbildschirm Flugzeug-Modus

Drücken Sie, während das Symbol WECHSEL
angezeigt wird, um den Empfindlichkeits-Modus
einzustellen. Der Wert wechselt zur nächsten blauen
Zahl.
86 - GERPN MN232-C4 K | Bedienungsanleitung

FreeStyle Comfort
Der FreeStyle Comfort wird mit eingeschaltetem
autoSAT-Modus geliefert.

Empfindlichkeits-Modus Einstellungswert 3 Ausgewählt

Drücken Sie, während das Symbol SPEICHERN
angezeigt wird, um den Einstellungswert 3 zu bestätigen.
Der Wert wechselt auf die Zahl „3“ in schwarzer Farbe.
Drücken Sie, während das Symbol
LÖSCHEN angezeigt wird, um den Wechsel derEmpfindlichkeitseinstellung abzubrechen.

Empfindlichkeits-Modus Einstellungswert 3 Bestätigt

Hauptbildschirm autoSAT-Modus

Drücken Sie, während das Symbol WECHSEL
angezeigt wird, um den Modus zu deaktivieren. Der Wert
wechselt auf „AUS“ in blauer Farbe.

autoSAT-Modus Aus Ausgewählt

Drücken Sie, während das Symbol SPEICHERN
angezeigt wird, um das Ausschalten des autoSATModus zu bestätigen. Der Wert wechselt auf „AUS“ inschwarzer Farbe.
Drücken Sie, während das Symbol LÖSCHEN
angezeigt wird, um das Ausschalten des autoSAT-Modusabzubrechen.

Empfindlichkeits-Modus Einstellungswert 5 (max. Wert) Bestätigt

Indem Siedrücken, wenn auf dem Bildschirmdas Symbol „Weiter“ angezeigt wird, wechselt der
Bildschirm zur nächsten Menü-Option.

AUTOSAT-MODUS

autoSAT-Modus Aus Bestätigt

Wiederholen Sie diese Vorgänge und wechseln Sie auf
EIN, um den autoSAT-Modus einzuschalten.
drücken, wenn auf dem Bildschirm
Indem Siedas Symbol „Weiter“ angezeigt wird, wechselt der
Bildschirm zur nächsten Menü-Option.

PN MN232-C4 K | Bedienungsanleitung

GER - 87

FreeStyle Comfort

Fehlerbehebung
ProblemMögliche Ursache

LösungDer FreeStyle Comfort zeigt
die Warnung an und gibt einenintermittierenden Signalton aus.

Siehe Warnzustände-Tabelle.

Siehe Warnzustände-Tabelle.

Der Sauerstoffkonzentrator lässt
sich nicht einschalten.

Der Akku ist vollständig
entladen.

Das Gerät über die Netzsteckdose mit Strom versorgen.

Nicht an externer
Stromversorgung
angeschlossen.
Allgemeine Funktionsstörung.

Sicherstellen, dass die externen Anschlüsse sicher
verbunden sind.
Führen Sie einen vollständigen Leistungszyklus
durch, indem Sie den Akku entfernen und die externe
Wechselstrom- oder Gleichstromversorgung trennen.
Setzen Sie den Akku anschließend wieder ein und derLeistungszyklus ist abgeschlossen. Schließen Sie bei
Bedarf die externe Stromversorgung wieder an.
Kontaktieren Sie Ihren Geräteanbieter und steigen Siegegebenenfalls auf eine andere Sauerstoffquelle um.

Alle übrigen Probleme.

Führen Sie einen vollständigen Leistungszyklus
durch, indem Sie den Akku entfernen und die externe
Wechselstrom- oder Gleichstromversorgung trennen.
Setzen Sie den Akku anschließend wieder ein und derLeistungszyklus ist abgeschlossen. Schließen Sie bei
Bedarf die externe Stromversorgung wieder an.
Steigen Sie auf eine andere Sauerstoffquelle um undkontaktieren Sie Ihren Geräteanbieter.

Hinweis: Einige Warnungen führen dazu, dass das Gerät heruntergefahren wird.
Hinweis: Während eines totalen Stromausfalls wird das Warnsystem zurückgesetzt (sämtliche vorherigen Warnungen werden
gelöscht) und es wird nur dann eine Warnung angezeigt, wenn der Zustand nach der Wiederherstellung der Stromversorgung
noch immer besteht.

88 - GERPN MN232-C4 K | Bedienungsanleitung

FreeStyle Comfort
Reiseinformationen
Der FreeStyle Comfort erfüllt die Richtlinien der FAA(Bundesluftfahrtbehörde der Vereinigten Staaten) zur
Verwendung in Passagierflugzeugen. Im Vorfeld einer
jeden Reise (mit dem Flugzeug, dem Zug, Fähre usw.):• Vergewissern Sie sich, dass sich der Konzentrator ineinem betriebsbereiten Zustand befindet.
• Setzen Sie sich mindestens 48 Stunden vor dem Abflug
mit dem Transportunternehmen in Verbindung undteilen Sie diesem mit, dass Sie mit einem FreeStyle
Comfort reisen werden.
Es wird empfohlen, auf Reisen stets ReserveAkkupacks mitzuführen. Achten Sie bei Flugreisen
darauf, dass Sie über genügend Akkukapazität fürBodenzeiten (Check-In, Security usw.) und eventuelle
Zwischenlandungen verfügen. Bei den meisten
Fluggesellschaften ist es Pflicht, eine ausreichende
Zahl an vollständig geladenen Akkupacks zur
Stromversorgung des Geräts mitzuführen, welche für
150 % der erwarteten maximalen Reisezeit (Flugdauer,
gesamte Bodenzeit, d. h. Zeiten vor und nach dem
Flug sowie während des Umsteigens und unerwartete
Verspätungen) ausreicht.
Hinweis: Beim Fliegen empfiehlt es sich, die ReserveAkkupacks in einer sicheren Tragetasche mitzuführen.

BESTIMMUNGEN DER USLUFTFAHRTBEHÖRDE

1. Schalten Sie POC unter Verwendung des Akkus ein,
WENN keine externe Stromquelle angeschlossen ist.

2. Aktivieren Sie den Flugzeug-Modus nach einer derfolgenden Methode:a. Folgen Sie den Anweisungen zum Aktivieren desFlugzeug-Modus, die Sie unter dem entsprechenden
Abschnitt der Anweisungen zur Menüführung finden.

b. Halten Sie die Aufwärts- und Abwärtstaste
gleichzeitig 5 Sekunden lang gedrückt, bis ein langer
Signalton zu hören ist und das Flugzeug-Symbol
angezeigt wird.

Ersatzakkus müssen einzeln gegen Kurzschlüsse
geschützt werden. Hierzu sind sie entweder in die
ursprüngliche Verkaufsverpackung zu legen oder die
Kontaktflächen sind anderweitig zu isolieren, z. B. durch
ein Abkleben mit Klebeband, oder die Unterbringung der
Akkus in separaten Plastiktüte oder Schutztasche. Akkus
dürfen nur im Handgepäck mitgeführt werden.

FLUGZEUG-MODUS AKTIVIEREN

3. Wenn das Flugzeug-Symbol angezeigt wird, stecken
Sie den POC ein.

Akkubetrieb
POC kann in einem Flugzeug gemäß den normalen
Betriebsverfahren ausschließlich im Akkubetrieb
verwendet werden, wenn keine externe Stromquelleangeschlossen ist.
Externe Stromversorgung
Wenn in einem Flugzeug eine Steckdose zur Verfügung
steht, die betriebsbereit ist und deren Standort für die
Verwendung als POC-Stromquelle geeignet ist, kann sie
als Ersatz für POC-Akkus dienen. In diesem Fall ist das
folgende empfohlene Verfahren für den Übergang von
Akku- zu Flugzeugstrom zu befolgen:

Hinweis: Aufgrund der Beschränkungen von Flugzeugen
laden sich die Akkus im Flugzeug-Modus nicht auf.

PN MN232-C4 K | Bedienungsanleitung

GER - 89

FreeStyle Comfort

Zubehör

TRAGETASCHE

Um einen ordnungsgemäßen und sicheren Betrieb zu
gewährleisten, bitte ausschließlich folgende Zubehörteile
von CAIRE verwenden, die Ihnen von Ihrem SauerstoffVersorger bereitgestellt werden. Die Verwendung vonZubehörteilen, die hier nicht aufgeführt sind, kann die
Funktion und/oder die Sicherheit des Geräts nachhaltig
beeinträchtigen.

1. Stecken Sie das Gerät in den Boden der Tasche.
Achten Sie dabei darauf, dass die Öffnungen für Filter
und Akku korrekt zueinander ausgerichtet sind.

FreeStyle Comfort Standard Zubehör
Tragetasche

MI396-1

Rucksack

MI459-1

Schultergurt

MI397-1

Wechselstrom-Netzteil (sämtliche Kabel
enthalten)

PW036-1S

Gleichstrom-Netzkabel

CD041-1

Externer Grobpartikelfilter (rechts)

FI226-1

Externer Grobpartikelfilter (links)

FI227-1

2. Schließen und sichern Sie den Reißverschluss am
unteren Teil der Tasche.

Verfügbares zusätzliches Zubehör
Zusätzlicher 8-Zellen Akkupack

BT034-1S

16-Zellen Akkupack

BT035-1S

Tischladegerät

BT036-1

Zubehörtasche für Reisegepäck

MI372-2

Hinweis: Landesspezifische Stecker sind, sofern oben
angegeben, ebenfalls erhältlich. Falls weitere Ersatzteile
bestellt werden müssen, wenden Sie sich bitte an CAIRE
oder an Ihren Geräteanbieter.

Bedienungsanleitung Tragetasche
Um Ihren FreeStyle Comfort in der Tragetasche zu
verstauen und zu sichern, befolgen Sie die folgenden
Anweisungen.

VERSTAUEN DES GERÄTS IN DER

90 - GERPN MN232-C4 K | Bedienungsanleitung

ANBRINGEN DES TRAGEGURTES
1. Der Schultergurt kann entweder direkt am Gerät oderan der Tragetasche befestigt werden.
2. Befestigen Sie den Gurt am Ring des Geräts oder derTasche.

FreeStyle Comfort
Entsorgung und Wiederverwendung
Geben Sie den FreeStyle Comfort, einschließlich
aller Komponenten, stets zur Entsorgung an Ihren
Homecare-Anbieter zurück. Für Anweisungen zur
ordnungsgemäßen Entsorgung der Batterie können
Sie sich zudem an die zuständigen Stellen Ihres
Wohnortes wenden. G. Das Gerät kann zwischen
Personen wiederverwendet werden, muss jedoch
zur Wiederaufbereitung an den Homecare-Anbieter
zurückgesandt werden.

über Elektro- und Elektronik-Altgeräte am Ende seiner
Lebensdauer zu einer Recyclingstelle gebracht werden
muss.
Unsere Produkte entsprechen der Richtlinie zur
Beschränkung der Verwendung bestimmter gefährlicher
Stoffe (RoHS). Sie enthalten Blei oder andere gefährliche
Materialien höchstens in Spuren.
VORSICHT: Hinweise zur ordnungsgemäßen
Entsorgung Ihres Akkus erhalten Sie vondem Bereitsteller, der Ihnen das Gerät zur
Verfügung gestellt hat, und von den fürdie Entsorgung von Abfällen zuständigenÄmtern.

WEEE und RoHS
Das Symbol soll den Besitzer des Geräts
darauf hinweisen, dass das Gerät gemäß der Richtlinie

Materialien mit Direktkontakt zum Benutzer.
Gehäuse..................................................................................................................................................................PC/ABS
Konzentrator-Bedienfeld..............................................................................................Polyester EBG7 oder gleichwertig
Konzentrator-Haltegriff.................................................................................................................................. Polykarbonat
Sauerstoff-Ausgang, Düse............................................................................................................................. Polykarbonat
Geräteetikett................................................................................................................................................................Lexan
Schnurstecker........................................................................................................................... Polykarbonat/Vinylchlorid
Halter für.......................................................................................................................................................... PVC, Metall
Stromversorgung....................................................................................................................... Lexan 940 (Polykarbonat)
Akkupack................................................................................................................................................................PC/ABS
Netzteil- und Akkupack-Etiketten................................................................................................................. Polyesterfilm
Konzentrator-Tragetasche................................................................................................. Cordura, Neopren, Nylonfutter
Haltegurt..................................................................................................Polypropylen, Acetyl, Vinyl, Cordura, Neopren

EMV-Prüfung
Leitlinien und Herstellererklärung – elektromagnetische Aussendungen
Der FreeStyle Comfort ist für den Gebrauch in einer elektromagnetischen Umgebung bestimmt, wie sie nachfolgend
beschrieben ist. Der Kunde oder der Benutzer des FreeStyle Comfort sollte sicherstellen, dass das Gerät in einer
entsprechenden Umgebung verwendet wird.
StöraussendungsmessungenÜbereinstimmung
Elektromagnetische Umgebung – Leitlinien
HF-Aussendung
CISPR 11
HF-Aussendung
CISPR 11
Aussendungen vonOberschwingungennach IEC 61000-3-2
Aussendungen von
Spannungsschwankungen/
Flicker nach IEC 61000-3-3

Gruppe 1

Der FreeStyle Comfort nutzt HF-Energie nur für interne
Funktionen. Daher ist die HF-Aussendung sehr gering, und es ist
unwahrscheinlich, dass benachbarte elektronische Geräte gestört
werden.

Klasse B
Klasse A
stimmt überein

Der FreeStyle Comfort ist für den Gebrauch in allen Einrichtungeneinschließlich denen im Wohnbereich und solchen geeignet,
die unmittelbar an ein öffentliches Niederspannungsnetz
angeschlossen sind, das auch Gebäude versorgt, die zu
Wohnzwecken benutzt werden.

PN MN232-C4 K | Bedienungsanleitung

GER - 91

FreeStyle Comfort
Leitlinien und Herstellererklärung – elektromagnetische Störfestigkeit

DerFreeStyle Comfort ist für den Gebrauch in einer elektromagnetischen Umgebung bestimmt, wie sie nachfolgend beschrieben
ist. Der Kunde oder der Benutzer des FreeStyle Comfort sollte sicherstellen, dass das Gerät in einer entsprechenden Umgebung
verwendet wird.
Störfestigkeitsprüfung
IEC 60601 PrüfpegelÜbereinstimmungspegel
Elektromagnetische
Umgebung – Leitlinien
Fußböden sollten aus Holzoder Beton bestehen oder mit
Elektromagnetische
Keramikfliesen versehen sein. Wenn
±8 kV Kontakt±8 kV Kontakt
Umgebung – Leitlinien
der Fußboden mit synthetischem±15 kV Luftaustritt±15 kV Luftaustritt
Material versehen ist, muss die
IEC 61000-4-2
relative Luftfeuchte mindestens 30 %betragen.
Die Qualität des Stromnetzes sollte±2 kV für Netzleitungen
Schnelle transiente±2 kV für Netzleitungender eines typischen professionellen
elektrische Störgrößen/Gesundheitszentrums und der
Bursts
±1 kV für Eingangs- undhäuslichen Gesundheitsversorgung
N/A
IEC 61000-4-4
Ausgangsleitungenentsprechen.
±1 kV Gegentaktspannung

Stoßspannungen
IEC 61000-4-5±2 kV Gleichtakt auf
Wechselstromleitungen±1 kV Differential auf
Wechselstromleitungen±2 kV Gleichtakt
auf I/O-Leitungen im
Außenbereich
0 % UT für 0,5 Perioden(0°, 45°, 90°, 135°, 180°,
225°, 270°, 315°)

Spannungseinbrüche,Kurzzeitunterbrechungen
und Schwankungen der
Versorgungsspannung
gemäß IEC 61000-4-11

0 % UT für 1 Periode (00)
70 % UT (30 % Einbruch
der UT) für 25/30 Perioden(00)
0 % UT für 250/300
Perioden (00)Magnetfeld bei der
Netzfrequenz (50/60 Hz)
gemäß IEC 61000-4-8±1 kV Gegentaktspannung

3 A/m

0 % UT für 0,5 Perioden (0°,45°, 90°, 135°, 180°, 225°,
270°, 315°)
0 % UT für 1 Periode (00)
70 % UT (30 % Einbruch derUT) für 25/30 Perioden (00)
0 % UT für 250/300 Perioden
(00)

3 A/m

HINWEIS: UT ist die Wechselstrom-Netzspannung vor dem Anlegen des Prüfpegels.

Die Qualität des Stromnetzes sollte
der eines typischen professionellen
Gesundheitszentrums und derhäuslichen Gesundheitsversorgung
entsprechen.

Die Qualität des Stromnetzes sollte
der eines typischen professionellen
Gesundheitszentrums und derhäuslichen Gesundheitsversorgung
entsprechen. Wenn der Benutzer
des FreeStyle Comfort auch beim
Auftreten von Unterbrechungen derStromversorgung eine fortgesetzte
Funktion benötigt, wird empfohlen,den FreeStyle Comfort über eineunterbrechungsfreie Stromversorgung
oder einen Akku mit Strom zu
speisen.
Die Magnetfelder der Netzfrequenz
sollten auf einem Niveau
liegen, das charakteristisch füreinen gewöhnlichen Standort
in einer professionellen
Gesundheitseinrichtung und im
häuslichen Gesundheitswesen ist.

HINWEIS: In einigen Fällen kann es vorkommen, dass der Freestyle Comfort nach einer signifikanten elektrostatischen
Entladung zurückgesetzt wird und ein vollständiger Leistungszyklus erforderlich ist, um den normalen Betrieb wieder
aufzunehmen. Weitere Informationen hierzu finden Sie im Abschnitt Fehlerbehebung.

92 - GERPN MN232-C4 K | Bedienungsanleitung

FreeStyle Comfort
Leitlinien und Herstellererklärung – elektromagnetische Störfestigkeit

DerFreeStyle Comfort ist für den Gebrauch in einer elektromagnetischen Umgebung bestimmt, wie sie nachfolgend beschrieben
ist. Der Kunde oder der Benutzer des FreeStyle Comfort sollte sicherstellen, dass das Gerät in einer entsprechenden Umgebung
verwendet wird.
Störfestigkeitsprüfung
IEC 60601 Prüfpegel Übereinstimmungspegel Elektromagnetische
Umgebung – Leitlinien
Geleitete HF
10 Vrms
10 Vrms
Tragbare und mobile HFKommunikationsausrüstung sollten nicht in
IEC 61000-4-6
150 kHz bis 80 MHz
einem geringeren Abstand zum FreeStyle
Comfort verwendet werden, als derempfohlene Schutzabstand, der anhand
der Gleichung für die Sendefrequenz
Gestrahlte HF
10 V/m
10 V/mberechnet wird. Dieser Schutzabstand gilt
IEC 61000-4-3
80 MHz bis 2700 MHz
auch für die Geräteleitungen.
Empfohlener Schutzabstandd = 1,2√ P

d = 1,2√ P = 80 MHz bis 800 MHz
d = 1,2√ P = 800 MHz bis 2700 MHz
Mit P als der Nennleistung des Senders
in Watt (W) gemäß Angaben des
Senderherstellers und d als empfohlenem
Schutzabstand in Metern (m).
Die Feldstärke stationärer Funksendersollte bei allen Frequenzen gemäß einer
Untersuchung vor Ort a geringer als derÜbereinstimmungspegel sein. b
In der Umgebung von Geräten, diedas folgende Bildzeichen tragen, sind
Störungen möglich:

ANMERKUNG 1 Bei 80 MHz und 800 MHz gilt der höhere Frequenzbereich.
ANMERKUNG 2 Diese Leitlinien sind möglicherweise nicht in allen Fällen anwendbar. Die Ausbreitung elektromagnetischer
Wellen wird durch Absorptionen und Reflexionen von Gebäuden, Gegenständen und Menschen beeinflusst.
a
Die Feldstärke stationärer Sender, z. B. Basisstationen von Funktelefonen und mobilen Landfunkgeräten,Amateurfunkstationen, AM- und FM-Rundfunk- und Fernsehsendern, können theoretisch nicht genau vorherbestimmt
werden. Um die elektromagnetische Umgebung hinsichtlich der stationären Sender zu ermitteln, sollte eine Standortanalyse
erwogen werden. Wenn die gemessene Feldstärke an dem Standort, an dem der FreeStyle Comfort benutzt wird, die obigenÜbereinstimmungspegel überschreitet, sollte der FreeStyle Comfort beobachtet werden, um die bestimmungsgemäße Funktion
zu überprüfen. Wenn ungewöhnliche Leistungsmerkmale beobachtet werden, können zusätzliche Maßnahmen erforderlich sein,z. B. eine Neuausrichtung oder ein anderer Standort des FreeStyle Comfort.
bÜber den Frequenzbereich von 150 kHz bis 80 MHz muss die Feldstärke unter 10 V/m liegen.

PN MN232-C4 K | Bedienungsanleitung

GER - 93

FreeStyle Comfort
Empfohlene Schutzabstände zwischen tragbaren und mobilen HFKommunikationsausrüstungen und dem FreeStyle Comfort

Der FreeStyle Comfort ist für den Betrieb in einer elektromagnetischen Umgebung bestimmt, in der die HF-Störgrößen
kontrolliert werden. Der Kunde oder der Benutzer des FreeStyle Comfort kann helfen, elektromagnetische Störungen zu
vermeiden, indem er den Mindestabstand zwischen tragbaren und mobilen HF-Kommunikationsausrüstungen (Sendern) unddem FreeStyle Comfort – abhängig von der Ausgangsleistung des Kommunikationsgeräts, wie unten angegeben – einhält.
Nennleistung desTrennungsabstand je nach Frequenz des Senders
Sendersm
W

150 kHz bis 80 MHz
d = 1,2√ P

80 MHz bis 800 MHz
d = 1,2√ P

800 MHz bis 2,5 GHz
d = 1,2√ P

0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Für Sender, deren maximale Nennleistung in obiger Tabelle nicht angegeben ist, kann der empfohlene Schutzabstand d inMetern (m) unter Verwendung der Gleichung ermittelt werden, die für die jeweilige Sendefrequenz gilt. Dabei ist P die maximale
Nennleistung des Senders in Watt (W) gemäß Angabe des Senderherstellers.
ANMERKUNG 1 Bei 80 MHz und 800 MHz gilt der Schutzabstand für den höheren Frequenzbereich.
ANMERKUNG 2 Diese Leitlinien sind möglicherweise nicht in allen Fällen anwendbar. Die Ausbreitung elektromagnetischer
Wellen wird durch Absorptionen und Reflexionen von Gebäuden, Gegenständen und Menschen beeinflusst.

Klassifikation

Art des Schutzes vor elektrischem Schlag:
Klasse II	Der Schutz vor Stromschlag wird durcheine doppelte Isolierung erreicht.

	• CISPR 11/EN 55011, Klasse B Gruppe 1,„Industrielle, wissenschaftliche und medizinische
Geräte“	• FCC Part 15, Subpart B – Class B Unintentional
Radiators

Grad des Schutzes vor elektrischem Schlag:
Typ BF Das Gerät bietet einen bestimmten Grad		 von Schutz gegen Stromschlag hinsichtlich
		 1) zulässigem Verluststrom;		2) Zuverlässigkeit der Schutzerdung (sofern
vorhanden).
		Nicht für die Direktanwendung am Herzen
geeignet.

IP22 - Schutz gegen das Eindringen von Fingern oderähnlichen Gegenständen mit einem Durchmesser vonmehr als 12,5 mm. Vertikal tropfendes Wasser darf
keine schädliche Wirkung haben, wenn das Gehäusein einem Winkel von 15 Grad von seiner normalen
Position geneigt ist.

Unabhängige Prüfung gemäß den Normen fürmedizinische elektrische Geräte:
Getestet auf Konformität mit• IEC 60601-1 Auflage 3.1: Medizinischeelektrische Geräte - Teil 1 Allgemeine
Anforderungen an die Sicherheit.
• CAN/CSA C22.2 Nr. 60601-1-8 Auflage 2.1,
Medizinische elektrische Geräte – Teil 1:
Allgemeine SicherheitsanforderungenZulässige Reinigungs- und Infektionskontrollmethoden:
Siehe Abschnitt „Reinigung, Pflege undordnungsgemäße Wartung“ in dieser FreeStyleGebrauchsanleitung.

Schutz gegen potenzielle elektromagnetische oder andere
Störungen zwischen den Geräten und anderen Geräten.
	• Getestet auf Konformität mit EN 60601-1-2/
IEC 60601-1-2: 4. Auflage.
	• Getestet auf Konformität mit RTCA/DO160
Absatz 21 Kategorie M.

Betriebsmodus: Dauerbetrieb.

94 - GERPN MN232-C4 K | Bedienungsanleitung

Grad der Anwendungssicherheit bei Vorhandensein
entzündlicher anästhetischer Gase:	Die Geräte sind für solche Anwendungen nicht
geeignet.

CAIRE und CAIRE Inc. sind eingetragene Marken von CAIRE Inc. Eine vollständige Listeder Marken finden Sie auf unserer Website.
Warenzeichen: www.careinc.com/corporate/trademarks.
Copyright © 2021 CAIRE Inc. CAIRE Inc. behält sich das Recht vor, die Vermarktung seiner
Produkte einzustellen bzw. Preise, Werkstoffe, Ausrüstungsteile, Qualität, Beschreibungen,Spezifikationen und/oder Prozesse ohne Vorankündigung zu einem beliebigen Zeitpunktzu ändern, ohne dass hieraus irgendwelche Verpflichtungen oder Rechtsfolgen entstehen.
Alle hier nicht ausdrücklich genannten Rechte bleiben im gesetzlich zulässigen Rahmen
CAIRE vorbehalten.

FreeStyle Comfort
Controles del usuario e indicadores de estado del sistema
Norma ISO 7000: símbolos gráficos que deben usarse
en los equipos (índice y sinopsis)
Lea el manual del usuario antes de usarel equipo. Reg. n.º 1641
Rango de temperatura dealmacenamiento u operación.
Reg. n.º 0632
Rango de humedad de almacenamiento.
Reg. n.º 2620

Directiva del Consejo 93/42/CEE sobre dispositivos
médicos
Representante autorizado en la
Comunidad Europea
Si la etiqueta del identificador único dedispositivo (UDI) del producto tiene elsímbolo CE ####, el dispositivo cumple
con los requisitos de la Directiva 93/42/ CEE sobre dispositivos médicos. El
símbolo CE #### indica el número delcuerpo notificado.
IEC 60417
Equipo de Clase II, Doble aislamiento,
n.º de reg. 5172

Limitación de presión atmosférica.
Reg. n.º 2621

Interruptor de encendido. Reg. n.º 5009

Nombre y domicilio del fabricante.
Reg. n.º 3082
Fecha de fabricación. Reg. n.º 2497

Certificación ETL
Agencia de seguridad para la norma
CAN/CSA C22.2 n.º 601.1 M90 paraequipos eléctricos médicos.

Número de catálogo. Reg. n.º 2493
Número de serie. Reg. n.º 2498

Símbolos internos

Este lado hacia arriba. Reg. n.º 0623

Mantener la unidad lejos de materiales
inflamables, aceite y grasa.

Frágil. Manipular con cuidado.
Reg. n.º 0621

No desmontar.

Mantenerlo alejado de la lluvia.
Mantenerlo seco. Reg. n.º 0626

Aumentar flujo.

Límite de apilamiento por número.
Reg. n.º 2403

Disminuir flujo.

ISO 7010: Símbolos gráficos: colores y símbolosde seguridad; símbolos de seguridad registrados
Debe leer el manual de instrucciones.
Reg. n.º M002
Mantener alejado de las llamas, del fuego
y de las chispas. No acercar a fuentes
de ignición abiertas. Se prohíbe fumar.
Reg. n.º P003

Botón Menú.
Cumplecon RTCA/
DO-160
Sección 21
Categoría M

Cumple con RTCA DO160 Sección 21
Categoría M. Requisito FAA SFAR 106

No fumar cerca de la unidad ni mientras
la esté utilizando. Reg. n.º P002

Equipo de Clase II, Doble aislamiento

Parte aplicada tipo BF (grado deprotección frente a descargas eléctricas).
Reg. n.º 5333

Aprobado por la FAA, Concentradorde oxígeno móvil

Advertencia. Reg. n.º W001

PN MN234-C4 K | Manual del usuario

SPA - 95

FreeStyle Comfort
21 CFR 801.15: Código de Normas Federales, Título 21
La legislación federal limita la venta
de este dispositivo a médicos o por orden
médica.
Directiva del Consejo 2012/19/UE: Directiva
sobre residuos de aparatos eléctricos yelectrónicos (RAEE)RAEE

Qué es el concentrador de oxígeno
El aire que respiramos contiene aproximadamente un
21 % de oxígeno, un 78 % de nitrógeno y un 1 % deotros gases. En el FreeStyle® Comfort®, el aire ambiente
entra a la máquina a través de las entradas de aire.
A continuación, pasa a través de un material absorbente
llamado tamiz molecular. Este material separa eloxígeno del nitrógeno y permite que solo el oxígeno
pase a través de él. El resultado es el suministro alusuario de un flujo de oxígeno de alta concentración.

IEC 60601-1: Equipos médicos eléctricos, Parte 1:
requisitos generales para la seguridad básica y eldesempeño esencial.
A prueba de goteos.
Este producto puede estar cubierto por una o varias
patentes, de EE. UU. o internacionales. Visite nuestro
sitio web para obtener el listado de patentes aplicables.
Pat.: www.caireinc.com/corporate/patents/.

o una configuración para tratar su trastorno respiratorio
particular.
Aunque el oxígeno no se trata de una droga adictiva,
la oxigenoterapia sin autorización puede ser peligrosa.
Debe consultar con un médico antes de utilizar elconcentrador de oxígeno. El proveedor de equipos
que suministra el equipo de oxígeno le mostrará cómo
ajustar el caudal que se le ha prescrito.

Perfil del operador

El FreeStyle Comfort combina tecnologías avanzadas
de concentración de oxígeno y de conservación deoxígeno para crear un dispositivo portátil que permite
la mayor portabilidad y movilidad. La administraciónavanzada de flujo de pulso detecta rápidamente cuando
inspira el usuario y administra un pulso de oxígeno
al principio de cada inhalación. La administraciónde flujo de pulso es idónea para realizar actividades
y pasar tiempo fuera de casa. Conserva el oxígeno
y alarga la duración de la batería, a diferencia delflujo continuo. El FreeStyle Comfort es tan ligero quepuede transportarse mediante la correa para hombro.
Puede funcionar con alimentación de CA (pared),CC (vehículo) o de batería.

Los concentradores están destinados a suministrar
oxígeno suplementario a los usuarios con molestias
debidas a alteraciones que afectan a la eficiencia desus pulmones para transferir el oxígeno en el aire asu torrente sanguíneo. Los concentradores de oxígeno
móviles (POC) no almacenan ni contienen oxígeno.
No necesitan ser rellenados y pueden recargarse
en cualquier fuente de alimentación de CA o CC
disponible (salvo en un avión). Así, el usuario puede
ser relativamente autosuficiente en términos de uso
doméstico, en movimiento (tanto dentro como fuera
de casa), movilidad y estilo de vida en general. El uso
del concentrador de oxígeno requiere una prescripción
médica y no está destinado a soporte vital.

Uso previsto

Aunque puede prescribirse un tratamiento con oxígeno
a usuarios de todas las edades, el usuario habitual de laoxigenoterapia es mayor de 65 años y padece diversas
enfermedades respiratorias, incluida la enfermedad
pulmonar obstructiva crónica (EPOC). Por lo general,los usuarios presentan buenas capacidades cognitivas,
y deben ser capaces de comunicar su incomodidad.
Si el usuario es incapaz de comunicar su incomodidad,
o es incapaz de leer y comprender el etiquetado ylas instrucciones de uso del concentrador, el uso serecomienda solo bajo la supervisión de alguien que
pueda. Si se siente alguna molestia durante el uso
del concentrador, se recomienda a los usuarios quese pongan en contacto con su proveedor de atención
médica. Asimismo, se recomienda a los usuarios
que dispongan de una fuente de oxígeno de respaldo(por ejemplo, un tanque de oxígeno) para contar con
un suministro de oxígeno en caso de que se produzca
un corte eléctrico o una avería del concentrador. El uso
del concentrador de oxígeno no requiere de otras
habilidades o capacidades singulares del usuario.

El concentrador de oxígeno CAIRE FreeStyle Comfort
está diseñado para la administración de oxígeno
suplementario. El dispositivo no está destinado asoporte vital ni proporciona ninguna capacidad demonitoreo del paciente.

Por qué su médico le ha prescrito
oxígenoMuchas personas padecen multitud de enfermedades
cardíacas, pulmonares y otras enfermedades
respiratorias. Un número considerable de estas personas
puede beneficiarse del tratamiento con oxígeno
suplementario en casa, al viajar o mientras realiza
actividades cotidianas fuera de casa.
El oxígeno es un gas que constituye el 21 % del aireambiente que respiramos, y el cuerpo humano dependedel suministro constante de este para funcionar
correctamente. Su médico le ha prescrito un flujo
96 - SPAPN MN232-C4 K | Manual del usuario

Desembalar su FreeStyle® Comfort™
Compruebe que todos loscomponentes enumerados están
incluidos en el paquete. Si falta
algún elemento, póngase en
contacto con el proveedor desu concentrador de oxígeno
inmediatamente.

Accesorios estándares• Concentrador de oxígeno portátil• Bolso de transporte• Correa para hombro• Paquete de batería de 8 celdas• Fuente de alimentación de CA
o CC• Cable de alimentación de CA• Fuente de alimentación de CC• Manual del usuario

FreeStyle Comfort
Accesorios opcionales• Paquete de batería de 8 celdas
adicional• Paquete de batería de 16 celdas• Cargador de escritorio• Bolsa para llevar todos losaccesorios

Conocer su FreeStyle Comfort
Familiarícese con las nuevas funciones clave del FreeStyle Comfort y su interfaz de usuario.
Fijación dela correa

Vista frontalInterfaz de usuario

Salida deoxígenoSalidade aireEntrada de aire /Filtros
Entrada de aire /Filtros
Pestaña delreceptáculo
del puerto dealimentación
Batería
Botón de extracciónde la batería
PN MN234-C4 K | Manual del usuario

SPA - 97

FreeStyle Comfort
Vista superior
Pantalla
Aumentar flujo

Botón deencendido

Botón Menú

Disminuir flujo

Pantalla con información de la batería
Ajuste de flujo

Indicador debatería

Indicador autoSAT

Pantalla de alarma

Porcentaje
de batería
restante

Indicador desuministro bolo

Pantalla con información de alimentación CA/CC

Ajuste de flujo

Espacio delindicadorde alarma

98 - SPAPN MN232-C4 K | Manual del usuario

Indicadorde batería

Indicador de carga

FreeStyle Comfort¡Importante!
Las instrucciones de seguridad se definende la siguiente manera:
ADVERTENCIA: INFORMACIÓN DE
SEGURIDAD IMPORTANTE SOBREPELIGROS QUE PUEDEN CAUSARLESIONES GRAVES.
PRECAUCIÓN: Información importantesobre la prevención de daños en elsistema FreeStyle Comfort.
Aclaración: información que requiere atención especial.

Indicaciones de uso
ADVERTENCIA: ES MUY IMPORTANTESELECCIONAR SOLO EL NIVEL DEOXÍGENO PRESCRITO. NO CAMBIE LA
SELECCIÓN DE FLUJO A MENOS QUE UN
PROFESIONAL MÉDICO ACREDITADO ASÍ
SE LO HAYA INDICADO.
ADVERTENCIA: LAS LEYES FEDERALES (DE LOSESTADOS UNIDOS) LIMITAN LA VENTA O ALQUILER
DE ESTE DISPOSITIVO POR PARTE DE UN MÉDICOU OTRO PROVEEDOR DE ATENCIÓN MÉDICA
ACREDITADO.
ADVERTENCIA: LA UNIDAD NO DEBE USARSE COMOSOPORTE VITAL. EN LOS PACIENTES GERIÁTRICOSO PEDIÁTRICOS O CUALQUIER OTRO USUARIOINCAPAZ DE COMUNICAR SU INCOMODIDADMIENTRAS USA ESTE DISPOSITIVO, PODRÍA SER
NECESARIO UTILIZAR UN SISTEMA DE SUPERVISIÓN
O VIGILANCIA ADICIONAL. LOS USUARIOS CON
TRASTORNOS AUDITIVOS O VISUALES PUEDEN
NECESITAR ASISTENCIA PARA CONTROLAR LAS
ALARMAS.
ADVERTENCIA: LOS AJUSTES DEL MODO DE
DOSIS PULSADA DEBEN DETERMINARSE PARACADA USUARIO INDIVIDUALMENTE, SEGÚN
SUS NECESIDADES EN ESTADO DE REPOSO,
AL REALIZAR EJERCICIO Y AL VIAJAR.
ADVERTENCIA: ES POSIBLE QUE ALGUNOSESFUERZOS RESPIRATORIOS DEL PACIENTENO ACTIVEN LA FUNCIÓN DE CONSERVACIÓN.
ADVERTENCIA: EL FREESTYLE COMFORT NO
ESTÁ DESTINADO PARA SU USO EN PACIENTESLARINGECTOMIZADOS.
ADVERTENCIA: LAS MUJERES EMBARAZADASO LACTANTES NO DEBEN USAR LOS ACCESORIOSRECOMENDADOS EN ESTE MANUAL, YA QUEPUEDEN CONTENER FTALATOS.

Contraindicaciones de uso
ADVERTENCIA: EN ALGUNASCIRCUNSTANCIAS, EL USO DE OXÍGENO
NO PRESCRITO PUEDE SER PELIGROSO.
ESTE DISPOSITIVO SOLO DEBERÁ
UTILIZARSE CUANDO LO PRESCRIBA
UN MÉDICO.
ADVERTENCIA: NO UTILIZAR EN PRESENCIA DE
ANESTÉSICOS INFLAMABLES.
ADVERTENCIA: AL IGUAL QUE CON CUALQUIERDISPOSITIVO ELÉCTRICO, EL USUARIO PUEDE
EXPERIMENTAR PERÍODOS DE FALTA DEFUNCIONAMIENTO COMO RESULTADO DE
UN CORTE EN LA ENERGÍA ELÉCTRICA O LA
NECESIDAD DE TAREAS DE INSPECCIÓN EN EL
CONCENTRADOR DE OXÍGENO POR PARTE DE
UN TÉCNICO CUALIFICADO. EL CONCENTRADORDE OXÍGENO NO ES APROPIADO PARA USUARIOSQUE EXPERIMENTARÍAN CONSECUENCIASADVERSAS EN LA SALUD COMO RESULTADO DE
TAL INTERRUPCIÓN TEMPORAL.

Instrucciones de seguridad
ADVERTENCIA: NO SE PERMITEN
MODIFICACIONES A ESTE EQUIPO.
ADVERTENCIA: ESTE DISPOSITIVO SUMINISTRAOXÍGENO DE ALTA CONCENTRACIÓN, LO QUE LO
HACE MUY INFLAMABLE. NO PERMITA QUE SE
FUME NI LA PRESENCIA DE LLAMAS EXPUESTASDENTRO DE LA MISMA HABITACIÓN EN LA QUE SE
ENCUENTRE (1) ESTE DISPOSITIVO O (2) ALGÚNACCESORIO QUE CONTENGA OXÍGENO. SI NO SE
TIENE EN CUENTA ESTA ADVERTENCIA, PUEDE
PRODUCIRSE UN INCENDIO GRAVE, DAÑOS A LOSBIENES O HASTA LESIONES FÍSICAS O LA MUERTE.
ADVERTENCIA: NO OPERE LA UNIDAD EN UN
ESPACIO RESTRINGIDO O CONFINADO (P. EJ., UNA
CAJA O BOLSO PEQUEÑO) DONDE LA VENTILACIÓN
PUEDE SER LIMITADA. ESTO PUEDE OCASIONARQUE EL CONCENTRADOR DE OXÍGENO SE
SOBRECALIENTE Y QUE SU RENDIMIENTO SE VEAPERJUDICADO.
ADVERTENCIA: EL CONCENTRADOR DEBEUBICARSE DE FORMA TAL QUE SE PUEDA EVITAR
EL HUMO, LOS CONTAMINANTES O LOS GASES
TÓXICOS.
ADVERTENCIA: EN EL CASO DE QUE OCURRA UN
INCIDENTE GRAVE CON ESTE DISPOSITIVO, EL
USUARIO DEBE INFORMAR INMEDIATAMENTE EL
INCIDENTE AL PROVEEDOR Y / O AL FABRICANTE.
UN INCIDENTE GRAVE SE DEFINE COMO UNALESIÓN, MUERTE O EL POTENCIAL DE CAUSARLESIONES / MUERTE EN CASO DE QUE VUELVA A
OCURRIR EL INCIDENTE. EL USUARIO TAMBIÉN
PUEDE INFORMAR EL INCIDENTE A LA AUTORIDADCOMPETENTE EN EL PAÍS DONDE OCURRIÓ EL
INCIDENTE.
PN MN234-C4 K | Manual del usuario

SPA - 99

FreeStyle Comfort
ADVERTENCIA: EL USO DE ACCESORIOSDE ADMINISTRACIÓN DE OXÍGENO
NO ESPECIFICADOS PARA ESTECONCENTRADOR DE OXÍGENO PUEDEDISMINUIR SU RENDIMIENTO. LOSACCESORIOS RECOMENDADOS SE
MENCIONAN DENTRO DE ESTE MANUAL.

ADVERTENCIA: NO PERMITA QUE SE BLOQUEEN LAENTRADA DE AIRE NI LA SALIDA DE AIRE. NO DEJE
CAER NI INSERTE OBJETOS EN NINGUNA ABERTURA
DEL DISPOSITIVO. ESTO PUEDE OCASIONAR QUE EL
CONCENTRADOR DE OXÍGENO SE SOBRECALIENTEY QUE SU RENDIMIENTO SE VEA PERJUDICADO.

ADVERTENCIA: SI EL CONCENTRADOR DE OXÍGENO
SE HA CAÍDO, HA SUFRIDO DAÑOS O SE HAEXPUESTO AL AGUA, PÓNGASE EN CONTACTO CON
EL PROVEEDOR DE EQUIPOS SANITARIA PARA SUINSPECCIÓN O POSIBLE REPARACIÓN. NO USE EL
CONCENTRADOR DE OXÍGENO SI TIENE EL CABLE
DE ALIMENTACIÓN O EL ENCHUFE DAÑADOS.

ADVERTENCIA: EL USO INCORRECTO DE LA
BATERÍA PUEDE OCASIONAR QUE ESTA SE
CALIENTE O SE ENCIENDA, Y PUEDE CAUSARLESIONES SERIAS. ASEGÚRESE DE NO PERFORAR,GOLPEAR, PISAR, DEJAR CAER O SOMETER DEOTRA FORMA LA BATERÍA A IMPACTOS O GOLPESFUERTES. EL USO DE BATERÍAS QUE ESTÉNDAÑADAS PODRÍA CAUSAR LESIONES FÍSICAS.

ADVERTENCIA: ASEGURE FIRMEMENTE, CON EL
CINTURÓN DE SEGURIDAD O DE OTRA FORMA,
EL CONCENTRADOR DE OXÍGENO EN EL VEHÍCULO
DURANTE SU TRANSPORTE PARA PREVENIR DAÑOSO LESIONES.

ADVERTENCIA: NO PROVOQUE UN CORTOCIRCUITO
DE LOS CONTACTOS DE METAL DE LA BATERÍA
CON OTROS OBJETOS METÁLICOS, COMO LLAVESO MONEDAS. ESTO PUEDE HACER QUE SE
GENEREN CHISPAS O CALOR EXCESIVO.

ADVERTENCIA: LOS REGLAMENTOS DEL
DEPARTAMENTO DE TRANSPORTE (DOT, POR
SUS SIGLAS EN INGLÉS) ESTADOUNIDENSE Y
DE LAS NACIONES UNIDAS (ONU) REQUIEREN LAEXTRACCIÓN DE LA BATERÍA DEL DISPOSITIVOPARA TODOS LOS VIAJES EN AEROLÍNEAS
INTERNACIONALES CUANDO SE FACTURE EL
CONCENTRADOR DE OXÍGENO COMO EQUIPAJE.
CUANDO SE ENVÍA EL CONCENTRADOR DEOXÍGENO, LA BATERÍA TAMBIÉN DEBE EXTRAERSE
DEL DISPOSITIVO Y EMBALARSE ADECUADAMENTE.

ADVERTENCIA: MIENTRAS USA EL CONCENTRADORDE OXÍGENO PORTÁTIL AL AIRE LIBRE CON LA
FUENTE DE ALIMENTACIÓN DE CA, CONECTE
LA FUENTE DE ALIMENTACIÓN SOLAMENTE A
UNA TOMA CON INTERRUPTOR POR PÉRDIDA ATIERRA (GFI).

ADVERTENCIA: NO UTILICE ESTE EQUIPO SINANTES LEER Y COMPRENDER ESTE MANUAL.
SI NO COMPRENDE LAS ADVERTENCIAS E
INSTRUCCIONES, CONTACTE A SU PROVEEDOR
DEL EQUIPO ANTES DE INTENTAR USARLO;
DE LO CONTRARIO, PUEDEN PRODUCIRSELESIONES O DAÑOS.
ADVERTENCIA: SI SIENTE MALESTAR O TIENEUNA EMERGENCIA MÉDICA, BUSQUE ASISTENCIAMÉDICA DE INMEDIATO.
ADVERTENCIA: LA OPERACIÓN DELCONCENTRADOR DE OXÍGENO SIN TENEREN CUENTA LAS ESPECIFICACIONES DE LA
TEMPERATURA DE FUNCIONAMIENTO PUEDE
LIMITAR LA CAPACIDAD DEL CONCENTRADOR
DE CUMPLIR CON LAS ESPECIFICACIONES DE
CONCENTRACIÓN DE OXÍGENO. CONSULTE LASECCIÓN DE ESPECIFICACIONES DE ESTE MANUALPARA CONOCER LOS LÍMITES DE TEMPERATURA.
ADVERTENCIA: NO USE ACEITE, GRASA,PRODUCTOS A BASE DE PETRÓLEO NI OTROSPRODUCTOS INFLAMABLES CON LOS ACCESORIOSDE TRANSPORTE DE OXÍGENO O CON EL
CONCENTRADOR DE OXÍGENO. SOLO DEBENUSARSE LOCIONES O BÁLSAMOS COMPATIBLESCON EL OXÍGENO, A BASE DE AGUA. EL OXÍGENO
ACELERA LA COMBUSTIÓN DE LAS SUSTANCIASINFLAMABLES.

100 - SPAPN MN232-C4 K | Manual del usuario

ADVERTENCIA: EL CONCENTRADOR DE OXÍGENO
NO DEBE USARSE ADYACENTE A O APILADO CONOTROS EQUIPOS. SI EL USO ADYACENTE O APILADOES INEVITABLE, EL DISPOSITIVO DEBE OBSERVARSEPARA VERIFICAR SU FUNCIONAMIENTO NORMAL.
ADVERTENCIA: EL USO DE CABLES Y
ADAPTADORES DIFERENTES A LOSESPECIFICADOS, CON EXCEPCIÓN DE LOSCABLES Y DE LOS ADAPTADORES VENDIDOSPOR EL FABRICANTE DEL EQUIPO MÉDICOELÉCTRICO COMO PIEZAS DE REPUESTO PARACOMPONENTES INTERNOS, PUEDE DERIVAR EN
UNA MAYOR EMISIÓN DE MENOR INMUNIDAD DEL
CONCENTRADOR DE OXÍGENO.
ADVERTENCIA: USE ÚNICAMENTE EL VOLTAJE
ELÉCTRICO INDICADO EN LA ETIQUETA DEESPECIFICACIONES ADHERIDA AL DISPOSITIVO.
ADVERTENCIA: NO UTILICE CABLES DE EXTENSIÓN
EN ESTA UNIDAD NI CONECTE DEMASIADOSENCHUFES A LA MISMA TOMA ELÉCTRICA. EL USO
DE CABLES DE EXTENSIÓN PODRÍA AFECTARDE FORMA ADVERSA EL RENDIMIENTO DELDISPOSITIVO. DEMASIADOS ENCHUFES EN UNATOMA PUEDEN OCASIONAR UNA SOBRECARGA DEL
PANEL ELÉCTRICO Y HACER QUE EL DISYUNTOR/FUSIBLE SE ACTIVE O DISPARE SI EL DISYUNTOR OEL FUSIBLE NO FUNCIONAN.

FreeStyle Comfort
PRECAUCIÓN: en caso de alarma o si
observa que el concentrador de oxígeno
no funciona correctamente, consulte lasección de Resolución de problemasde este manual. Si no puede resolver el
problema, consulte con el proveedor desu equipo.

ADVERTENCIA: DEBEN EXHIBIRSE LETREROS DE"NO FUMAR: OXÍGENO EN USO" EN EL HOGAR
O EN EL SITIO DONDE SE USE EL OXÍGENO. LOS
USUARIOS Y LOS CUIDADORES SANITARIOS DEBEN
RECIBIR INFORMACIÓN ACERCA DE LOS PELIGROS
DE FUMAR AL USAR OXÍGENO MEDICINAL O ENPRESENCIA DE ESTE.

PRECAUCIÓN: para evitar la anulación de la garantía,
siga todas las instrucciones del fabricante.

ADVERTENCIA: NO USE SU CONCENTRADOR DEOXÍGENO EN PRESENCIA DE GASES INFLAMABLES.
ESTO PUEDE GENERAR UN FUEGO RÁPIDOQUE PUEDE PROVOCAR DAÑOS A LOS BIENES,LESIONES CORPORALES O LA MUERTE.

ADVERTENCIA: NO INTENTE REALIZARTAREAS DE MANTENIMIENTO APARTE
DE LAS SOLUCIONES POSIBLESENUMERADAS EN ESTE MANUAL. NO
RETIRE LAS CUBIERTAS: SOLO SUPROVEEDOR DEL EQUIPO O UN TÉCNICODE SERVICIO CUALIFICADO DEBENREVISAR O RETIRAR LAS CUBIERTASDE ESTE DISPOSITIVO.
ADVERTENCIA: USE SOLO ACCESORIOSRECOMENDADOS POR EL FABRICANTE. EL USO
DE OTRO ACCESORIO PUEDE SER PELIGROSO,CAUSARÁ DAÑOS GRAVES A SU CONCENTRADORDE OXÍGENO Y ANULARÁ LA GARANTÍA.
ADVERTENCIA: EL FABRICANTE RECOMIENDADISPONER DE UNA FUENTE ALTERNATIVA DEOXÍGENO COMPLEMENTARIO EN CASO DE
CORTE ELÉCTRICO, CONDICIÓN DE ALARMA O
FALLA MECÁNICA. CONSULTE A SU MÉDICO O AL
PROVEEDOR DEL EQUIPO PARA DETERMINAR EL
TIPO DE SISTEMA DE RESERVA QUE NECESITA.
ADVERTENCIA: NO PERMITA QUE SE BLOQUEEN LAENTRADA DE AIRE NI LAS ABERTURAS DE SALIDA
DE AIRE. NO DEJE CAER NI INSERTE OBJETOSEN NINGUNA ABERTURA DEL DISPOSITIVO. ESTO
PUEDE OCASIONAR QUE EL CONCENTRADORDE OXÍGENO SE SOBRECALIENTE Y QUE SU
RENDIMIENTO SE VEA PERJUDICADO.
ADVERTENCIA: FUMAR MIENTRAS SE USA OXÍGENO
ES LA CAUSA PRINCIPAL DE LESIONES POR
INCENDIOS Y MUERTES RELACIONADAS. DEBESEGUIR LAS ADVERTENCIAS DE SEGURIDAD:
ADVERTENCIA: NO PERMITA QUE HAYACIGARRILLOS ENCENDIDOS, VELAS O LLAMASEXPUESTAS EN LA MISMA HABITACIÓN EN QUE SE
ENCUENTRE EL DISPOSITIVO O LOS ACCESORIOSQUE TRANSPORTAN OXÍGENO.
ADVERTENCIA: FUMAR MIENTRAS SE USA
UNA CÁNULA DE OXÍGENO PUEDE CAUSARQUEMADURAS FACIALES Y POSIBLEMENTE LAMUERTE.
ADVERTENCIA: SI FUMA, ESTOS 3 PASOS PUEDENSALVAR SU VIDA: APAGUE EL CONCENTRADORDE OXÍGENO, RETIRE LA CÁNULA Y SALGA DE
LA HABITACIÓN DONDE SE ENCUENTRE ELDISPOSITIVO.

ADVERTENCIA: RETIRAR LA CÁNULA Y COLOCARLASOBRE LA ROPA, LA CAMA, EL SOFÁ U OTRO
MATERIAL ACOLCHADO CAUSARÁ UNA LLAMARADASI SE LA EXPONE A UN CIGARRILLO, UNA FUENTE
DE CALOR O UNA LLAMA.
ADVERTENCIA: NO DEJE UNA CÁNULA NASAL
SOBRE O DEBAJO DE LA ROPA, LA CAMA O LOSALMOHADONES DE UNA SILLA. SI LA UNIDAD
ESTÁ ENCENDIDA PERO NO EN USO, EL OXÍGENO
PODRÍA VOLVER INFLAMABLES ESOS MATERIALES.
ASEGÚRESE DE QUE LA ALIMENTACIÓN ESTÉAPAGADA CUANDO EL CONCENTRADOR DEOXÍGENO NO ESTÉ EN USO.
ADVERTENCIA: EL CONCENTRADOR DE OXÍGENO
DEBE ESTAR APAGADO Y DESCONECTADO DE UNAFUENTE DE ALIMENTACIÓN EXTERNA ANTES DE
PROCEDER A RETIRAR LA BATERÍA.
ADVERTENCIA: LAS PIEZAS PEQUEÑAS
PUEDEN CONSTITUIR UN RIESGOPOTENCIAL DE AHOGAMIENTO.
ADVERTENCIA: EVITE EL CONTACTO CON LA
SALIDA DE AIRE DE ESCAPE DURANTE MÁS DE
10 SEGUNDOS, YA QUE PUEDE ESTAR CALIENTE.
ADVERTENCIA: EL VIENTO O LAS CORRIENTES
FUERTES PUEDEN INFLUIR NEGATIVAMENTE EN
EL SUMINISTRO DE OXIGENOTERAPIA.
ADVERTENCIA: ESTE PRODUCTO PUEDE
EXPONERLO A SUSTANCIAS QUÍMICAS COMO
EL NÍQUEL, QUE, SEGÚN EL ESTADO DE
CALIFORNIA, CAUSA CÁNCER. PARA OBTENER MÁS
INFORMACIÓN, VISITE WWW.P65WARNINGS.CA.GOV.
PRECAUCIÓN: Siempre desconecte
la fuente de alimentación de CA de lapared antes de desconectar la fuente dealimentación de CA del concentrador deoxígeno.
PRECAUCIÓN: No coloque la unidad de manera que elacceso al cable de alimentación resulte complicado.
PRECAUCIÓN: Siempre coloque el tubo de suministrode oxígeno y los cables de alimentación de forma tal
que evite el riesgo de tropiezos o de estrangulación
accidental.

PN MN234-C4 K | Manual del usuario

SPA - 101

FreeStyle Comfort
PRECAUCIÓN: Al usar el concentrador de oxígeno en
un automóvil, embarcación o en otras fuentes de CC
con la fuente de alimentación de CC, asegúrese deque el vehículo esté funcionando antes de conectar
el concentrador de oxígeno. Si no se siguen estas
instrucciones, la fuente de alimentación podría nosuministrar energía al concentrador de oxígeno.
PRECAUCIÓN: Cuando el automóvil en el que se usa
la unidad del concentrador de oxígeno se apague,
desconecte y retire el dispositivo del automóvil. No
almacene el concentrador de oxígeno en un automóvil
muy caliente o muy frío o en otro ambiente similar dealta o baja temperatura.
Aclaración: los equipos de comunicaciones portátiles ymóviles de radiofrecuencia pueden afectar a los equipos
eléctricos médicos.
PRECAUCIÓN: NO deje el concentradorde oxígeno ni la fuente de alimentación
enchufados en el vehículo si se encuentra
apagado. De lo contrario, podría agotar labatería del vehículo.
PRECAUCIÓN: Si el concentrador de oxígeno se
opera fuera de su intervalo de temperatura defuncionamiento normal y rango de presión ambiente(altitud), se puede ver afectado el rendimiento y
reducirse el tiempo de funcionamiento de la batería y/oaumentar el tiempo de carga de la batería. Consultela sección de especificaciones de este manual para
conocer los límites de temperatura de funcionamientoy almacenamiento.
PRECAUCIÓN: Para concentradores de oxígeno
equipados con baterías: Para garantizar la duraciónde la vida de la batería, manténgala almacenada en
un lugar fresco y seco. Guardar el concentradorde oxígeno durante un lapso prolongado a altas
temperaturas o con una batería totalmente cargada/descargada puede reducir la vida útil general de labatería. NO intente abrir la batería, ya que no disponede piezas que puedan repararse en su interior.
Mantenga las baterías lejos de los niños.
PRECAUCIÓN: SOLO USE las baterías suministradas
por el fabricante. Para la eliminación correcta de labatería, contacte a su proveedor del equipo o a suorganismo gubernamental local a fin de conocer losrequisitos de eliminación.

102 - SPAPN MN232-C4 K | Manual del usuario

Aclaración: asegúrese de que la cánula esté bien insertaday quede firme. Esto asegura que el concentrador deoxígeno pueda detectar apropiadamente la inspiraciónpara la entrega de oxígeno. Durante la inhalación, debe
escuchar o sentir el flujo de oxígeno hacia el extremo de lacánula nasal.
Aclaración: las baterías de litio pueden perder su
capacidad para siempre si se exponen a temperaturas
extremadamente elevadas estando las baterías totalmente
cargadas o descargadas. Si se va a almacenar por un
periodo prolongado, se recomienda cargar las baterías a unnivel entre el 25 % y 50 % de su capacidad y guardarlas a
una temperatura de -20 °C - 20 °C (-4 °F - 68 °F).
Aclaración: el uso de accesorios de administraciónde oxígeno no especificados para su uso con este
concentrador de oxígeno puede hacer disminuir
su rendimiento. Los accesorios recomendados se
mencionan dentro de este manual.
Aclaración: no opere el concentrador de oxígeno sin
el filtro de entrada de aire colocado. Si dispone de
un segundo filtro, coloque este filtro de reemplazoantes de limpiar el filtro sucio. Limpie el filtro sucio con
una solución tibia de agua y jabón y luego seque porcompleto antes de usar.
Aclaración: el fabricante no recomienda la esterilización deeste equipo.
Aclaración: en ningún caso existe el peligro de reducir
el oxígeno presente en una habitación al usar elconcentrador de oxígeno.
ADVERTENCIA: EL CONCENTRADORDE OXÍGENO NO SE DEBE OPERARNI CONECTAR A UNA FUENTE DE
ALIMENTACIÓN EXTERNA (CA/CC)SIN HABER COLOCADO LA BATERÍAPREVIAMENTE. NO INTENTE ENCENDER
LA UNIDAD O CONECTARLA A UNAFUENTE DE ALIMENTACIÓN EXTERNASI SE HAN EXTRAÍDO LAS BATERÍAS.
RETIRE LA BATERÍA SOLO CUANDO LA UNIDADESTÉ APAGADA Y DESCONECTADA DE LA FUENTE
DE ALIMENTACIÓN. NO INTENTE RETIRAR LA
BATERÍA MIENTRAS LA UNIDAD ESTÉ ENCENDIDA
O CONECTADA A LA FUENTE DE ALIMENTACIÓN.
Aclaración: si el concentrador de oxígeno se ha
almacenado por un período prolongado fuera de su
estado de funcionamiento ambiente normal mostrado
en la tabla de arriba, permita que la unidad vuelva ala temperatura de funcionamiento normal antes deencenderla.

FreeStyle Comfort
Especificaciones
FreeStyle Comfort
Dimensiones (con una sola batería)

25,4 x 18,5 x 7,9 cm (10,0 x 7,3 x 3,1 pulgadas)

Dimensiones (con dos baterías)

27,9 x 18,5 x 7,9 cm (11,0 x 7,3 x 3,1 pulgadas)

Peso (con una sola batería)

2,3 kg (5 lb)

Peso (con batería doble)

2,7 kg (6 lb)

Nivel nominal del sonido***39,93 dB (A) ± 0,42 dB (A) @ 2

Ajustes de flujo
Ajuste de salida de O2(@ 20bpm)

Concentración de oxígeno*
Control de oxígeno
Presión máxima de salida
Sensibilidad de dosis predeterminada

Ajuste de pulso 1, 2, 3, 4, 5
Ajuste 1 - 210 mL/min
Ajuste 2 - 420 mL/min
Ajuste 3 - 630 mL/min
Ajuste 4 - 840 mL/min
Ajuste 5 - 1050 mL/min
90 % (+5.5 % / -3 %)Sí<30 PSI> -0,1 cm H2O (ajustable)

Energía de CA

100–240 V CA, 50–60 Hz, 120W Max

Energía de CC

11–18 V CC (10 A máx.)Tiempo de recarga de la batería:

Batería única: 3,5 horas
Batería doble: 6,0 horas

Duración de la batería
Temperatura de funcionamiento**
Tipo de batería
Capacidad de la batería

Duración de la batería única: Configuración 2: hasta 4 horas
Duración de la batería doble: Configuración 2: hasta 8 horas
5 °C a 40 °C (41 °F a 104 °F)
Ion de litio
Celda simple: 6700 mAh; Doble celda: 13400 mAh

Temperatura de transporte yalmacenamiento (Unidad)

De -25 °C a +70 °C (de -13 °F a +158 °F) hasta 3048 m (10000 ft)

Temperatura de transporte yalmacenamiento (Batería)

De -25 °C a +70 °C (de -13 °F a +158 °F) hasta 3048 m (10000 ft)

Funcionamiento en altitud**De -381 a 3048 m (de -1250 a 10.000 ft) (probado a 700 – 1060 hPa)

Humedad de funcionamiento

Humedad relativa de 15-95 % (sin condensación)

Humedad de almacenamiento

0 - 90% sin condensación

Garantía limitada

3 años unidad
1 año batería y fuentes de alimentación* Con base en el intervalo de presión atmosférica de 101 kPa (14,7 psi) a 21°C (70 °F).
** La unidad funcionará en todos los entornos siempre que el entorno esté con los rangos de temperatura y humedad especificados.
El funcionamiento fuera de estas especificaciones puede limitar la capacidad del concentrador de suministrar la especificación deconcentración de oxígeno a velocidades de flujo de litro más altas. Es posible que se siga visualizando la pantalla de calentamiento
hasta que se alcance el nivel de concentración objetivo o pasen 15 minutos. Continúe usando la unidad durante este tiempo.
*** Nivel sonoro medido por método de prueba Nr. 14-1 10/2018 MDS-Hola.
La vida útil esperada es de un mínimo de cinco años. La vida útil en almacenamiento de las capas de tamiz es de aproximadamente
un año si el concentrador no se usa durante ese tiempo.

PN MN234-C4 K | Manual del usuario

SPA - 103

FreeStyle Comfort

Instrucciones de funcionamiento
1. Localice y coloque el FreeStyle Comfort en una
posición adecuada para su uso.
a. El concentrador debe estar bien ventilado paraque los filtros de entrada y salida de aire no se
obstruyan.
b. Coloque el concentrador de modo que losindicadores visuales y sonoros y las alarmas
puedan verse y oírse fácilmente.
c. Asegúrese de que los filtros de entrada de aireestén bien colocados antes de utilizar la unidad.
Póngase en contacto con su proveedor delequipo si falta algún filtro de la unidad.
2. Instale la batería en el concentrador.
a. Alinee la batería con el alojamiento inferior delcentro del FreeStyle Comfort.

3. Inserte la fuente de alimentación de CA o CC en el
receptáculo del puerto de alimentación ubicado en laparte delantera inferior derecha de la unidad.

4. Conecte una cánula nasal al puerto de salida deoxígeno. Asegúrese de que la conexión sea segura
antes de poner el sistema en funcionamiento.
ACLARACIÓN: Usar una cánula diferente de lasrecomendadas puede influir negativamente en el
rendimiento.

b. Deslice la batería hacia dentro hasta que
escuche un clic para confirmar que se hayacolocado correctamente.

104 - SPAPN MN232-C4 K | Manual del usuario

Nota: Para el proveedor del equipo: Se recomiendan lossiguientes accesorios de administración de oxígeno para
usar con el FreeStyle Comfort:• Cánula nasal con 7 pies (2.1 m) de tubo (LPM máximo:
6 LPM): Número de parte CAIRE 5408-SEQ• Información adicional sobre accesorios recomendados
está disponible en línea en www.caireinc.com.

FreeStyle Comfort
5. Para encender el FreeStyle Comfort, pulse elbotón de encendido y seleccione el ajuste de flujo
recetado usando los botones «aumentar» (+) o (-)«disminuir» flujo.

Volumen del pulso autoSAT (mL)*Ajuste
Frecuencia
respiratoria

1

2

3

4

5

15

10,5

21,0

31,5

42,0

52,5

20

10,5

21,0

31,5

42,0

52,5

25

10,5

21,0

31,5

42,0

42,0

30

10,5

21,0

31,5

35,0

35,0

35

10,5

21,0

30,0

30,0

30,0

40

10,5

21,0

26,3

26,3

26,3

*La tolerancia es de +/- 15 %Se emitirá una breve alarma sonora para indicar que
el dispositivo se ha encendido.
Aclaración: Cuando se enciende el FreeStyle Comfort
por primera vez, el ajuste de flujo predeterminado es 2.
Si se cambia el ajuste, la unidad se encenderá con elajuste de flujo previo.
6. Coloque la cánula nasal en su cara siguiendo las
instrucciones del fabricante suministradas con la
cánula y respire con normalidad. Debe sentir unsoplo de oxígeno suministrado a través de la cánula
nasal cada vez que inhala.
7. Para cambiar el ajuste de flujo, pulse los botones«aumentar» (+) o «disminuir» (-) flujo hasta
conseguir la velocidad de flujo deseada. Es normal
escuchar un sonido diferente a medida que cambia
los ajustes de flujo.

Volumen del pulso no autoSAT (mL)*Ajuste
Frecuencia
respiratoria

1

2

3

4

5

15

14,0

28,0

42,0

56,0

70,0

20

10,5

21,0

31,5

42,0

52,5

25

8,4

16,8

25,2

33,6

42,0

30

7,0

14,0

21,0

28,0

35,0

35

6,0

12,0

18,0

24,0

30,0

40

5,3

10,5

15,8

21,0

26,3

8. Apague el FreeStyle Comfort pulsando el botón deencendido.
Aclaración: La pantalla entrará en modo de suspensión
y se atenuará al cabo de 20 segundos. Pulse cualquierbotón para que la pantalla salga del modo desuspensión.
9. Para las características avanzadas, consulte
Instrucciones del menú Navegación

CARACTERÍSTICAS ESPECIALES
DEL PRODUCTO:
autoSAT®
La característica autoSAT patentada de CAIRE permite
que FreeStyle Comfort entregue un volumen constantede oxígeno (bolo) por respiración, hasta un índicede respiración máximo. A medida que la frecuencia
respiratoria del usuario aumenta durante actividadescomo caminar, la función autoSAT ajusta la operacióndel dispositivo para mantener un volumen de bolo
establecido.
UltraSense®
Este sensible disparador de dosis en pulso detecta elcambio de presión durante la respiración y libera elpulso de oxígeno cuando inhala.
autoDOSE™
En caso de producirse una «alarma por falta derespiración», esta función emitirá un pulso de oxígeno
a 20 respiraciones por minuto, con el ajuste de flujo
actual de la unidad, hasta que se resuelva la alarma.
PN MN234-C4 K | Manual del usuario

SPA - 105

FreeStyle Comfort

Fuentes de alimentación
El FreeStyle Comfort se puede alimentar de tres
maneras:• Batería recargable• Energía de CA• Energía de CC
En las siguientes secciones tiene información de cómo
funciona cada uno de estos componentes.
ADVERTENCIA: USE SÓLO LAS FUENTES
DE ALIMENTACIÓN INDICADAS EN ESTEMANUAL DE USUARIO. EL USO DE
FUENTES DE ALIMENTACIÓN Y BATERÍASQUE NO SEAN DE CAIRE PUEDE
OCASIONAR DAÑOS EN LA UNIDAD Y UNFUNCIONAMIENTO INCORRECTO.

QUITAR LA BATERÍA

Presione el botón de extracción de la batería del
FreeStyle Comfort. Mientras MANTIENE PULSADO
este botón, deslice la batería hacia fuera retirándola delconcentrador.
ADVERTENCIA: EL CONCENTRADORDE OXÍGENO DEBE ESTAR APAGADO YDESCONECTADO DE UNA FUENTE DE
ALIMENTACIÓN EXTERNA ANTES DE
PROCEDER A RETIRAR LA BATERÍA.

BATERÍA

El sistema FreeStyle Comfort puede funcionar con labatería recargable que se suministra con el dispositivo.
No hay limitaciones de funcionamiento cuando elconcentrador funciona alimentado por la batería. Todaslas velocidades de flujo están disponibles cuando elconcentrador recibe la alimentación de la batería.

COMPROBAR LA CARGA DE LA BATERÍA

INSTALAR LA BATERÍA

Alinee la batería con el alojamiento inferior del centrodel FreeStyle Comfort. Deslice la batería hacia dentro
hasta que escuche un clic para confirmar que se hayacolocado correctamente.

Para comprobar la carga de la batería que está instalada
en el FreeStyle Comfort, consulte el icono de batería yel porcentaje de batería restante de la pantalla.
80–100 %60-80 %

40-60 %

20-40 %

20 %106 - SPAPN MN232-C4 K | Manual del usuario

FreeStyle Comfort
Aclaración: Si la batería está cargándose, se mostrará
un indicador de enchufe en lugar del porcentaje de carga
restante.
Para comprobar la carga de la batería cuando noestá instalada en el FreeStyle Comfort, colóquela enhorizontal y pulse el botón de prueba TEST. Las luces
del indicador del estado de batería (25-100 %)se iluminan a la derecha del botón TEST para indicar
el nivel de carga de la batería.

CARGA INICIAL

La batería nueva suministrada con el sistema FreeStyle
Comfort no está cargada completamente cuando se
envía de fábrica. Antes de usarlo por primera vez, debe
cargar completamente la batería usando una fuente dealimentación de CA o CC.

INFORMACIÓN GENERAL DE CARGAEl FreeStyle Comfort recargará la batería siempre que
esté instalada en el concentrador y el concentrador esté
conectado a una fuente de alimentación de CA o CC
(salvo que esté en un avión).

DURACIÓN DE FUNCIONAMIENTODE LA BATERÍA

Cuando se desconecta una fuente de alimentación
CA o CC externa, el concentrador pasará a usar
automáticamente la alimentación de batería si hay una
batería cargada instalada. La siguiente tabla muestra las
duraciones típicas de una batería nueva. Las duraciones
de la batería son constantes, independientemente de su
frecuencia respiratoria.
Ajustede pulso

Duración de labatería única
(8 celdas)

Duración de labatería doble(16 celdas)

1

hasta 8 horas

hasta 16 horas

2hasta 4 horas

hasta 8 horas

3hasta 3 horas

hasta 6 horas

4hasta 2,25 horas

hasta 4,5 horas

5hasta 2 horas

hasta 4 horas

TIEMPO TÍPICO DE RECARGA DE LA
BATERÍA

El tiempo típico de recarga de una bateríacompletamente descargada es de 3,5 horas para una
batería única y de 6,0 horas para una batería doble.
Aclaración: Es posible que la batería no empiece a cargar
inmediatamente si está completamente descargada.

VIDA ÚTIL DE LA BATERÍA

Las baterías del FreeStyle Comfort están diseñadas
para durar 300 ciclos de carga/descarga.

La batería está totalmente cargada cuando se muestra
un 100 %.
Si lo desea, puede cargar la batería usando el cargador
de escritorio opcional.

PN MN234-C4 K | Manual del usuario

SPA - 107

FreeStyle Comfort

Fuente de alimentación de CA y CC
El FreeStyle Comfort viene con una fuente dealimentación de CA y CC estándar.
La fuente de alimentación de CA incluye lo siguiente:a. El cargador de CA que funciona entre 100 y
240 voltios, 50-60Hz.
b. El cable de alimentación de CA para conectar
al cargador y salida de CA correspondiente.

2. Conecte el cable de alimentación de CA aladaptador de CA.
La fuente de alimentación de CC consta de un cableque se enchufa directamente en la unidad y en la salida
de CA.

ENERGÍA DE CA

La energía de CA universal se usa en el hogar o donde
esté disponible una salida de alimentación de CA.
El FreeStyle Comfort es totalmente funcional con laalimentación de CA. El FreeStyle Comfort también
carga la batería mientras recibe alimentación de CA(excepto en un avión). La batería se cargará tanto si el
FreeStyle Comfort está encendido como apagado.
ADVERTENCIA: NO CONECTE
ALIMENTACIÓN EXTERNA SIN HABER
COLOCADO PREVIAMENTE LA BATERÍA
EN EL CONCENTRADOR DE OXÍGENO.
Para conectar el FreeStyle Comfort a la alimentaciónde CA:
1. Conecte el adaptador de CA a la entrada delconector de alimentación ubicado en la partedelantera inferior derecha del FreeStyle Comfort.
Asegúrese de empujarla hacia dentro para que laconexión sea segura.

108 - SPAPN MN232-C4 K | Manual del usuario

3. Conecte el extremo opuesto del cablede alimentación de CA a la toma de pared o a otra
salida eléctrica adecuada.

FreeStyle Comfort
4. Para encender el FreeStyle Comfort, pulse elbotón de encendido y seleccione el ajuste de flujo
recetado usando los botones «aumentar» (+) o (-)«disminuir» flujo.

Aclaración: Si se necesita cargar la batería, el icono
de batería mostrará una única barra de color naranja.

ENERGÍA DE CC

La energía de CC se usa en salidas para accesoriosde vehículos, embarcaciones y otros vehículos a motor.
El FreeStyle Comfort es totalmente funcional con laalimentación de CA. La batería del FreeStyle Comfort
también se carga mediante la alimentación de CC.
La batería se carga tanto con la unidad apagada comoencendida y en uso.

3. Conecte el extremo opuesto del cable dealimentación de CC a la salida del vehículo o acualquier otra salida para accesorios de CC.

4. Para encender el FreeStyle Comfort, pulse elbotón de encendido y seleccione el ajuste de flujo
recetado usando los botones «aumentar» (+) o (-)«disminuir» flujo.
Aclaración: Si se necesita cargar la batería, el icono
de batería mostrará una única barra de color naranja.

ADVERTENCIA: NO CONECTE
ALIMENTACIÓN EXTERNA SIN HABER
COLOCADO PREVIAMENTE LA BATERÍA
EN EL CONCENTRADOR DE OXÍGENO.
ADVERTENCIA: NO HAGA FUNCIONAR ELDISPOSITIVO EN SALIDAS DE VEHÍCULO DE 12 V CC
MIENTRAS EL VEHÍCULO ESTÉ SIN ARRANCAR.
1. Arranque el motor del vehículo.
2. Conecte la fuente de alimentación de CC al
receptáculo del puerto de alimentación ubicado enla parte delantera inferior derecha del FreeStyle
Comfort. Asegúrese de empujarla hacia dentro paraque la conexión sea segura.

PN MN234-C4 K | Manual del usuario

SPA - 109

FreeStyle Comfort
Limpieza, cuidados y mantenimiento
periódico
MANTENIMIENTO PERIÓDICO

Mantenimiento preventivo
El dispositivo no necesita mantenimiento. Los filtros
dentro del dispositivo están destinados a durar toda
la vida útil del dispositivo. Si el dispositivo no
funciona correctamente, la unidad debe devolverse alproveedor para su reparación. El mantenimiento delos componentes internos del gabinete del FreeStyle
Comfort ha de ser llevado a cabo por el proveedor delconcentrador de oxígeno.
El operador de la unidad es responsable de lossiguientes procedimientos de mantenimiento.
• Semanalmente: Limpiar los filtros de entrada de aire
ADVERTENCIA: NO INTENTE ABRIR O
RETIRAR LAS CUBIERTAS DE PLÁSTICODEL CONCENTRADOR DE OXÍGENO.
LA UNIDAD SOLO DEBE SER ABIERTAPOR UN TÉCNICO DE MANTENIMIENTOCUALIFICADO.

FILTROS DE ENTRADA DE AIRE

El aire entra al FreeStyle Comfort a través de filtros
de entrada de aire ubicados en la parte inferior frontaldel concentrador de oxígeno. Los filtros evitan que elpolvo y otras partículas grandes existentes en el aire
entren a la unidad.
Debe limpiar los filtros de entrada de aire una vez
por semana como mínimo. Puede que su proveedordel concentrador de oxígeno le aconseje limpiarlo
con más frecuencia dependiendo de las condiciones
de funcionamiento. Siga estos pasos para limpiar
correctamente los filtros de entrada de aire.
1. Extraiga los dos filtros de la parte frontal del
FreeStyle Comfort. Para ello, presione la pestaña
correspondiente y tire de ella.

2. Lave ambos filtros con una solución templada deagua y jabón.
3. Enjuague bien los filtros y elimine el exceso deagua con un paño suave y absorbente. Asegúrese deque los filtros estén completamente secos antes devolver a colocarlos.
4. Vuelva a colocar los filtros limpiados y secos.
110 - SPAPN MN232-C4 K | Manual del usuario

ADVERTENCIA: NO OPERE EL
CONCENTRADOR DE OXÍGENO SIN
EL FILTRO DE ENTRADA DE AIRE
COLOCADO.

CUIDADO DE LA BATERÍA

No permita que la batería quede expuesta a líquidos.
No intente desmontar ni abrir la batería.
No exponga la batería a llamas abiertas.
No deje caer la batería.
Mantenga la batería fuera del alcance de los niños.
Si parece que la batería está dañada, póngase en
contacto con su proveedor del concentrador de oxígeno
antes de usarla.
No coloque la batería dentro ni cerca de un microondas
ni ningún otro aparato de cocina.

ALMACENAMIENTO DE LA BATERÍA

No es necesario que la batería del concentrador deoxígeno esté completamente descargada antes devolver a cargarla. Se recomienda cargar la bateríadel concentrador de oxígeno después de cada uso.
No deje la batería en un lugar expuesto a calor
excesivo, como luz directa solar o en un coche. De lo
contrario, el rendimiento y la vida útil de la batería
podrían deteriorarse. También puede hacer que labatería genere calor, humo, prenda fuego o explote.
La batería se debe almacenar en un entorno estable
caracterizado por una baja humedad (humedad relativa
inferior al 70 %) donde no haya gases corrosivos. Para
evitar la oxidación, evite condiciones que puedan crearcondensación tales como fluctuaciones rápidas detemperatura ambiente.
Para guardarla durante un mes o más, se recomienda
que la batería esté descargada o cargada entre el 50 %y 75 % (de dos a tres LED). Una vez cargada, extraiga
la batería del concentrador de oxígeno.

FreeStyle Comfort
LIMPIEZAAPAGUE la unidad y desconéctela de la toma de CAo de CC antes de limpiarla. NO pulverice la carcasa
exterior directamente. Utilice un paño húmedo (nomojado) o una esponja. Pulverice el paño o la esponja
con una solución limpiador doméstico suave para
limpiar el gabinete y las fuentes de alimentación. Sigalas instrucciones del fabricante del limpiador.
ADVERTENCIA: APAGUE LA UNIDADY DESCONECTE EL CABLE DE
ALIMENTACIÓN DE LA TOMA ELÉCTRICAANTES DE LIMPIAR LA UNIDAD PARAEVITAR UNA DESCARGA ELÉCTRICA
ACCIDENTAL O EL RIESGO DE
QUEMADURAS. ÚNICAMENTE ELPROVEEDOR DE SU EQUIPO O UNTÉCNICO DE SERVICIO CUALIFICADODEBEN QUITAR LAS TAPAS O REALIZARLA INSPECCIÓN DE LA UNIDAD.
ADVERTENCIA: HAY QUE TENER CUIDADO DE
EVITAR QUE EL CONCENTRADOR DE OXÍGENOY LA BATERÍA SE MOJEN Y NO DEBE PERMITIRSEQUE NINGÚN LÍQUIDO ENTRE EN LA UNIDAD. ESTO
PUEDE CAUSAR UN FALLO O APAGADO Y PRODUCIRUN MAYOR RIESGO DE DESCARGA ELÉCTRICAO QUEMADURAS.

ADVERTENCIA: UTILICE UN PAÑO HÚMEDO (NOMOJADO) O UNA ESPONJA PARA LIMPIAR LA
BATERÍA. PRIMERO, PULVERIZE EL PAÑO O LAESPONJA CON UN DETERGENTE DOMÉSTICO SUAVEY LIMPIE EL ALOJAMIENTO DE LA BATERÍA.
ADVERTENCIA: NO LIMPIE CON UN SISTEMA DE
LAVADO AUTOMATIZADO.

REEMPLAZO DE LA CÁNULA

Siempre siga las instrucciones del fabricante dela cánula para un uso correcto. Reemplace lacánula nasal o el tubo de oxígeno cada periodo deentre 3 y 6 meses (si se usa a diario, cada 3 meses)
atendiendo a la recomendación del fabricante de lacánula o de su proveedor del concentrador de oxígeno.
El médico o proveedor del concentrador de oxígeno lefacilitarán las instrucciones de limpieza y reemplazode las piezas.
El proveedor del concentrador de oxígeno dispone
de suministros adicionales de piezas de repuesto.

ADVERTENCIA: NO USE LÍQUIDO EN CONTACTODIRECTO CON LA UNIDAD. A CONTINUACIÓN,SE INCLUYE UNA LISTA DE ALGUNOS AGENTESQUÍMICOS NO DESEABLES, ENTRE OTROS:
ALCOHOL Y PRODUCTOS A BASE DE ALCOHOL,PRODUCTOS A BASE DE CLORO CONCENTRADO(CLORURO DE ETILENO) Y PRODUCTOS A BASE DE
ACEITE. ESTOS NO DEBEN USARSE PARA LIMPIAR
LA CARCASA DE PLÁSTICO DEL CONCENTRADORDE OXÍGENO, YA QUE PUEDEN DAÑAR LA UNIDAD.
ADVERTENCIA: LIMPIE EL GABINETE, EL PANEL DE
CONTROL Y EL CABLE DE ALIMENTACIÓN SOLOCON UN LIMPIADOR DOMÉSTICO APLICADO CON
UN PAÑO HÚMEDO (NO MOJADO) O UNA ESPONJAY LUEGO SEQUE TODAS LAS SUPERFICIES. NO
PERMITA QUE NINGÚN LÍQUIDO SE INTRODUZCA EN
EL CONCENTRADOR. PRESTE ESPECIAL ATENCIÓN
A LA SALIDA DE OXÍGENO PARA LA CONEXIÓN
DE LA CÁNULA PARA ASEGURARSE DE QUE SE
MANTENGA LIBRE DE POLVO, AGUA Y PARTÍCULAS.

PN MN234-C4 K | Manual del usuario

SPA - 111

FreeStyle Comfort

Estados de alarma
Prioridadn.ºDescripción dela alarma

1

Fallo delsistema por altatemperatura debatería

Pantalla

Señal
sonoraMás detalles / Acción sugerida

1 pitido
cada 20 s

La unidad emitirá una alarma cuando latemperatura de la alarma esté alcanzando elumbral de apagado de seguridad.
Si se alcanza el umbral de apagado, la unidad se
apagará. Si se conecta la alimentación externa,
se mostrará una pantalla de alarma y sonará una
alarma durante 2 minutos.
Probar con batería a temperatura ambiente. Si el
problema persiste, póngase en contacto con el
proveedor.

2

Fallo delsistema por altatemperatura

1 pitido
cada 20 s

La unidad emitirá una alarma cuando latemperatura interna esté alcanzando el umbral deapagado de seguridad.
Si se alcanza el umbral de apagado, se mostrará
una pantalla de alarma y sonará una alarma
durante 2 minutos.
Asegúrese de que no haya obstrucciones en
ninguna de las aberturas de la unidad. Si el
problema persiste, póngase en contacto conel proveedor.

3

Fallo delsistema por altapresión

1 pitido
cada 20 s

Si se alcanza el umbral de apagado, se mostrará
una pantalla de alarma y sonará una alarma
durante 2 minutos.
Si el problema persiste, póngase en contacto conel proveedor.

4

No se detecta larespiración

1 pitido
cada 30 s

No se detecta respiración durante 15 segundos.
Se activa cuando la pantalla de primera
respiración o calentamiento se apaga, lo que
ocurra primero.
Compruebe que no haya presiones o fugas en la
cánula. Aumente la configuración de sensibilidad.
Si el problema persiste, póngase en contacto conel proveedor.

5

Batería baja,
apagado1 pitido
cada 30 s

Emite una alarma hasta que la batería se agota yla unidad se apaga.
Cargue la batería.

6

Calentamiento

112 - SPAPN MN232-C4 K | Manual del usuario

ningunoLa unidad iniciará la dosificación automática
hasta que se detecte la respiración o finalice elcalentamiento, lo que ocurra primero.

FreeStyle Comfort
Prioridadn.ºDescripción de laalarma7

Pantalla

Señal
sonoraMás información

Baja concentración

1 pitido
cada 30 s,
hasta 5 min

Si el problema persiste, póngase en
contacto con el proveedor.

8

Fallo del sistema,velocidades de flujo(fallo de la válvula
OCD, obstrucción dela ruta del gas)

1 pitido
cada 30 s

Si el problema persiste, póngase en
contacto con el proveedor.

9

Advertencia debatería baja

1 pitido, sinrepetición

Cargue la batería.

10

Frecuencia
respiratoria excedida

1 pitido
cada 60 s

La frecuencia respiratoria excede los 40
bpm.
En caso de que la dosis no se suministre
correctamente, disminuya el ajuste desensibilidad. Si el problema persiste,póngase en contacto con el proveedor.

11

Fallo del sistema,
fallo del ventilador

ningunoSi el problema persiste, póngase en
contacto con el proveedor.

12

Fallo del sistema,
baja presión

ningunoSi el problema persiste, póngase en
contacto con el proveedor.

13

Fallo del sistema,
baja temperatura

ningunoDeje que la unidad alcance la temperatura
ambiente. Si el problema persiste,póngase en contacto con el proveedor.

Info

Arranque1 pitido
informativo

La pantalla se enciende durante elarranque y debe escuchar un pitido para
confirmar que el timbre funciona
La pantalla de arranque muestra la versiónde firmware en la parte inferior derecha dela esquina.
PN MN234-C4 K | Manual del usuario

SPA - 113

FreeStyle Comfort
Prioridadn.ºDescripción de laalarmaInfo

Pantalla defuncionamiento
normal en modo
autoSAT (no hay
una fuente dealimentación CC
conectada)ningunoInfo

Pantalla defuncionamiento
normal con
modo autoSAT
deshabilitado (nohay una fuente dealimentación CC
conectada)

NingunoInfo

Ajuste cambiado

1 pitido
informativo

Info

Alimentación externaconectada mientras
está apagado

ningunoInfo

Conecte laalimentación externa
mientras está en
funcionamiento

ningunoInfo

Desconecte laalimentación externa
mientras está
apagado1 pitido
informativo

Info

Desconecte laalimentación externa
mientras está en
funcionamiento (conbatería)1 pitido
informativo

114 - SPAPN MN232-C4 K | Manual del usuario

Pantalla

Señal
sonoraMás información

Se muestra la pantalla CAIRE durante
2 segundos, seguido del estado de labatería y del cable durante 5 s antes deque la pantalla se apague

Se muestra la pantalla CAIRE durante
2 segundos y luego la pantalla se apaga.

FreeStyle Comfort
Prioridadn.ºDescripción dela alarma

Info

Pantalla

Señal
sonoraMás información

En modo desuspensión:
cuando se
pulsa el botónde arriba oabajo la pantalla
muestra labatería y elestado de carga

ningunoLa pantalla se iluminará durante 5 s y después
se apagará

Info

Modo de avión
en alimentaciónde batería

1 pitido
largo

Info

Modo de avión
mientras estáenchufado

1 pitido
largo

PN MN234-C4 K | Manual del usuario

SPA - 115

FreeStyle Comfort

Instrucciones del menú Navegación
Es posible acceder al menú en modo de suspensióno de funcionamiento. En el modo de funcionamiento,
en caso de que se produjese una alarma, se saldrádel menú automáticamente. Si durante 20 segundos
no se presiona ningún botón, se saldrá del menú
automáticamente. En el modo de suspensión, es
posible que tenga que presionar el botón Menú dos
veces para que la unidad salga del modo de suspensión
y poder entrar en el menú.
Hay cuatro pantallas Menú:

Modo avión

1. Pantalla de estado: Muestra el número de versión defirmware, el tiempo de funcionamiento, los ciclos
de descarga
2. Modo avión (ENCENDIDO/APAGADO)
3. Sensibilidad: ajustable (1-5)
4. autoSAT (ENCENDIDO/APAGADO)

Sensibilidad

Ejemplo de pantalla de estado

FUNCIONAMIENTO DEL MENÚ AJUSTESEl menú Ajustes se introduce al presionar elen los modos debotón Menú/Seleccionar
funcionamiento o de suspensión.
Para navegar por los menús, pulse

Página de estado

116 - SPAPN MN232-C4 K | Manual del usuario.

autoSAT®

FreeStyle ComfortPÁGINA DE ESTADO
La página de estado de Freestyle Comfort muestra
la revisión del firmware, las horas de funcionamiento y el número de ciclos de batería en la batería
conectada.

Para cancelar el modo avión, presione
icono CANCEL esté visible.

mientras elModo avión encendido confirmado

Para apagar el modo avión, repita los pasos anteriores
mientras se coloca en «OFF».
Página de estado

Las horas de funcionamiento se muestran en horas
enteras.
La descarga de la batería es el número de ciclos quela batería conectada se ha drenado y recargado.

MODO AVIÓNEl FreeStyle Comfort se envía con el modo avión
apagado.

Al presionar
cuando la pantalla muestra el símbolo«Next», el menú pasará a la siguiente opción de pantalla.
El modo avión está diseñado para usarse en modo defuncionamiento. Se deshabilitará automáticamente si launidad se deja en suspensión con la batería.
Aclaración: Consulte la sección Activar modo avión paraobtener más información sobre el modo avión.

MODO DE SENSIBILIDADEl FreeStyle Comfort se envía desde fábrica con 2
ajustes del modo de sensibilidad predeterminados y no
debería necesitar de ajustes en condiciones normales.
El ajuste de sensibilidad se puede ajustar entre losvalores 1 y 5, donde 1 es el menos sensible y 5 el más
sensible.
Pantalla principal del modo avión

Para encender el modo, presione
mientras el
icono CHANGE esté visible. El valor pasará a «ON»,en azul.

Pantalla del modo de sensibilidad con el ajuste 2

Para ajustar el modo de sensibilidad, presione
mientras el icono CHANGE esté visible. El valor
pasará al siguiente número azul.
Modo avión encendido seleccionado

Para confirmar el modo avión, presione
mientrasel icono SAVE esté visible. El valor pasará a «ON»,en negro.

PN MN234-C4 K | Manual del usuario

SPA - 117

FreeStyle Comfort
MODO AUTOSATEl FreeStyle Comfort se envía con el modo autoSAT
encendido.

Ajuste 3 del modo de sensibilidad seleccionado

Para confirmar el ajuste 3 del modo de sensibilidad,
presione
mientras el icono SAVE esté visible.
El valor pasará a «3», en negro.
Para cancelar el modo de sensibilidad, presione
mientras el icono CANCEL esté visible.

Ajuste 3 del modo de sensibilidad confirmado

Pantalla principal del modo autoSAT

Para apagar el modo, presione
mientras el icono
CHANGE esté visible. El valor pasará a «OFF», en azul.

Modo autoSAT encendido seleccionado

Para confirmar el modo autoSAT, presione
mientrasel icono SAVE esté visible. El valor pasará a «OFF»,en negro.
Para cancelar el modo autoSAT, presione
el icono CANCEL esté visible.

mientras

Ajuste 5 del modo de sensibilidad (máximo) confirmado

Al presionar
cuando la pantalla muestra el símbolo«Next», el menú pasará a la siguiente opción de pantalla.

Modo autoSAT apagado confirmado

Para encender el modo autoSAT, repita los pasos
anteriores mientras se coloca en «ON».
Al presionar
cuando la pantalla muestra el símbolo«Next», el menú pasará a la siguiente opción de pantalla.

118 - SPAPN MN232-C4 K | Manual del usuario

FreeStyle Comfort
Resolución de problemas
Problema

Causa posible

Solución

El sistema FreeStyle Comfort
hace sonar una alarma y emite
un pitido intermitente.

Consulte la tabla de Estadosde alarma.

Consulte la tabla de Estados de alarma.

El concentrador de oxígeno
no se enciende.

La batería está descargada.

Conecte la unidad a una fuente de alimentación de
CA o CC.

No está conectado a una fuentede alimentación externa.
Falla general.

Asegúrese de que las conexiones externas son
seguras.
Haga un ciclo completo: retire la batería ydesconecte cualquier fuente de alimentación
externa CA o CC. A continuación, vuelva a
colocar la batería: el ciclo de alimentación se hacompletado. Si lo desea, vuelva a conectar lafuente de alimentación externa.
Comuníquese con el proveedor de su equipo
y cambie a otra fuente de oxígeno según sea
necesario.

Todos los demás problemas.

Haga un ciclo completo: retire la batería ydesconecte cualquier fuente de alimentación
externa CA o CC. A continuación, vuelva a
colocar la batería: el ciclo de alimentación se hacompletado. Si lo desea, vuelva a conectar lafuente de alimentación externa.
Cambie a otra fuente de oxígeno que hayadisponible y póngase en con su proveedor delequipo.

Aclaración: Algunas alarmas harán que la unidad se apague.
Aclaración: Cuando se pierde la alimentación, el sistema de alarma fuerza un reinicio (elimina cualquier alarma anterior
que pueda haber) y solo se mostrará la alarma si la incidencia persiste después de que se haya restablecido laalimentación.

PN MN234-C4 K | Manual del usuario

SPA - 119FreeStyle Comfort

Información de viaje

El FreeStyle Comfort cumple las directrices de la FAA
para usarse a bordo de aviones comerciales. Antes decualquier viaje (avión, tren, crucero, etc.), debe:• Asegurarse de que el concertador funcione
debidamente.
• Ponerse en contacto con su aerolínea al menos
48 horas antes de la salida del vuelo para informarlesde que viajará con un sistema FreeStyle Comfort.
Se recomienda viajar siempre con baterías adicionales.
Si va a viajar en avión, lleve la batería suficiente: tenga
en cuenta el tiempo en tierra (facturación, controles deseguridad, etc.) y escalas. La mayoría de las aerolíneas
requieren que cuente con un número adecuado debaterías cargadas completamente para alimentar eldispositivo por al menos 150 % del tiempo máximo deduración prevista del viaje (tiempos de vuelo, todo eltiempo en tierra, conexiones y demoras imprevistas).

1. Encienda el concentrador de oxígeno móvil SOLO
cuando no esté conectado a ninguna fuente dealimentación externa.

2. Habilite el modo avión mediante uno de estos
métodos:a. Siga las instrucciones para habilitar el modo avión
en la sección Instrucciones del menú Navegación.

Aclaración: Si va a volar, es recomendable que llevebaterías adicionales en una bolsa de transporte segura.

REGLAMENTO DE LA FAA

Las baterías de repuesto deben protegerse
individualmente para evitar cortocircuitos, colocándolasen su embalaje de venta original o mediante terminalesde aislamiento; por ejemplo, encintando los terminales
expuestos o colocando cada batería en una bolsade plástico o estuche protector por separado, ytransportadas en equipaje de mano solamente.

b. Presione y manténgalos presionados los botones dearriba y abajo durante 5 segundos hasta que oiga unpitido largo y vea que aparece al símbolo del avión.

ACTIVAR EL MODO AVIÓNAlimentación de batería
El concentrador de oxígeno móvil se puede usar
en una aeronave siguiendo los procedimientos defuncionamiento normales usando ÚNICAMENTE laalimentación de la batería, cuando no haya conectada
ninguna fuente de alimentación externa.

3. Cuando se muestre el símbolo del avión, conecte elconcentrador de oxígeno móvil.

Alimentación externa
En un avión, si dispone de una salida eléctrica que
funcione y esté cerca de usted para usarla como
fuente de alimentación del concentrador de oxígeno
móvil, puede servirle a modo de reserva de lasbaterías. En este caso, debe seguir el procedimiento
recomendado de transición de la batería a laalimentación eléctrica de la aeronave:
Aclaración: Debido a las limitaciones de las aeronaves,las baterías no se cargan durante el modo avión.

120 - SPAPN MN232-C4 K | Manual del usuario

FreeStyle Comfort
Accesorios

INSERTAR LA UNIDAD EN EL BOLSO DE
TRANSPORTE

Para obtener un rendimiento y seguridad adecuados,
utilice solo los accesorios indicados en la tabla que
CAIRE suministra a través del proveedor de su
concentrador de oxígeno. El uso de accesorios que
no se mencionan a continuación podría afectar demanera negativa al rendimiento o a la seguridad delconcentrador.

1. Coloque la unidad en el fondo del bolso, alineando
adecuadamente los filtros y las aberturas de labatería.

Accesorios estándares del FreeStyle Comfort
Bolso de transporte

MI396-1

MochilaMI459-1

Correa para hombro

MI397-1

Fuente de alimentación de CA (todos
los cables incluidos)

PW036-1S

Fuente de alimentación de CC

CD041-1

Filtro de partículas externas grandes(derecha)FI226-1

Filtro de partículas externas grandes(izquierda)

FI227-1

2. Asegure la cremallera alrededor de la parte inferiordel bolso.

Accesorios opcionales disponibles
Paquete de batería de 8 celdas
adicional

BT034-1S

Paquete de batería de 16 celdas

BT035-1S

Cargador de batería de escritorio

BT036-1

Bolsa para llevar todos los accesorios

MI372-2

Aclaración: puede haber opciones adicionales disponibles
para cables de alimentación específicos del país que no se
mencionan arriba. Póngase en contacto con CAIRE o conel proveedor de su concentrador de oxígeno si necesita
encargar opciones alternativas.

COLOCACIÓN DE LA CORREA PARAHOMBRO
1. La correa para hombro se puede acoplar a la unidad
directamente o al bolso de transporte.
2. Acople la correa al aro de la unidad o del bolso.

Instrucciones del bolso detransporte
Para insertar y fijar su FreeStyle Comfort en el bolso
de transporte, siga las siguientes instrucciones.

PN MN234-C4 K | Manual del usuario

SPA - 121

FreeStyle Comfort

Eliminación y reutilización
Devuelva siempre a su médico el FreeStyle Comfort
junto con todos sus componentes para su correcta
eliminación. También puede ponerse en contacto
con las oficinas pertinentes de su ciudad para obtener
instrucciones sobre la correcta eliminación de la batería.
La unidad puede reutilizarse entre personas, pero debe
devolverse al proveedor de atención domiciliaria parasu reprocesamiento.

de conformidad con la Directiva sobre residuos
de aparatos eléctricos y electrónicos (RAEE).
Nuestros productos cumplirán con la Restricción sobre
Sustancias Peligrosas (RoHS). No contendrán más que
trazas de plomo o de otros materiales peligrosos.

RAEE y RoHS

PRECAUCIÓN: Para un desecho adecuado,
comuníquese con su proveedor de equipos
o su agencia de gobierno local para
conocer las instrucciones para el desecho.

Este símbolo pretende recordar a lospropietarios de los equipos que deben llevarlos
a un centro de reciclado al final de su vida útil,Materiales en contacto directo con el operador
Carcasa del concentrador............................................................................................................................. PC/ABS
Panel de control del operador..................................................................................... Poliéster EBG7 o equivalente
Asa del concentrador............................................................................................................................Policarbonato
Salida de oxígeno, boquilla..................................................................................................................Policarbonato
Etiqueta de la unidad........................................................................................................................................Lexan
Conectores de cable................................................................................................Policarbonato/Cloruro de vinilo
Retenedor................................................................................................................................................ PVC, metal
Fuente de alimentación...................................................................................................Lexan 940 (Policarbonato)
Batería.......................................................................................................................................................... PC/ABS
Batería, etiquetas de suministro de alimentación...................................................................... Película de poliéster
Funda de transporte del concentrador................................................................ Forro de cordura, neopreno, náilon
Correa..............................................................................................Polipropileno, acetil, vinilo, cordura, neopreno

Prueba de EMC
Orientación y declaración del fabricante: emisiones electromagnéticas
El FreeStyle Comfort está destinado para su uso en el ambiente electromagnético que se especifica a continuación.
El cliente o el usuario del FreeStyle Comfort deben asegurar que se use en tal ambiente.
Prueba de emisiones

Cumplimiento

Ambiente electromagnético: orientación

Emisiones deradiofrecuencia

Grupo 1El sistema FreeStyle Comfort usa energía de radiofrecuencia solo parasu función interna. En consecuencia, sus emisiones de radiofrecuencia
son muy bajas y no es probable que causen interferencias en equipos
electrónicos cercanos.

CISPR 11
Emisiones deradiofrecuencia
CISPR 11
Emisiones armónicas
IEC 61000-3-2
Fluctuaciones de voltaje/emisiones de flicker
IEC 61000-3-3
122 - SPA

Clase B
Clase A
CumplePN MN232-C4 K | Manual del usuario

El sistema FreeStyle Comfort es adecuado para usar en todos losestablecimientos, incluidos los establecimientos domésticos y aquellos
conectados directamente a la red pública de energía eléctrica de bajo
voltaje que alimenta edificios usados para fines domésticos.

FreeStyle Comfort
Orientación y declaración del fabricante: inmunidad electromagnéticaEl FreeStyle Comfort está destinado para su uso en el ambiente electromagnético que se especifica a continuación.
El cliente o el usuario del FreeStyle Comfort deben asegurar que se use en tal ambiente.
Prueba de
Nivel de prueba IEC
Nivel de cumplimiento Ambiente electromagnético: orientación
inmunidad
60601
Ambiente
Los suelos deben ser de madera, hormigón
Contacto de ± 8 kV
Contacto de ± 8 kVelectromagnético:o cerámica. Si los suelos están cubiertos
Descarga de aire deDescarga de aire deorientación
con material sintético, la humedad relativa± 15 kV± 15 kVdebe ser de al menos un 30 %.
IEC 61000-4-2
Ráfagas eléctricas
de transitorios muy
rápidos
IEC 61000-4-4

Sobrecarga
IEC 61000-4-5

Caídas de voltaje,interrupciones
por cortocircuitos
y variaciones devoltaje en las
líneas de entrada
de las fuentes
de alimentación
IEC 61000-4-11± 2 kV para las líneas
de alimentación± 1 kV para líneas deentrada/salida± 1 kV línea(s) a línea(s)±2kV modo común en
líneas de CA± 2 kV para las líneas
de alimentación
N/A± 1 kV línea(s) a línea(s)±1kV diferencial en líneas
de CA±2kV modo comúnen líneas de E/S deexteriores
0 % UT para 0,5 ciclo (0°,45°, 90°, 135°, 180°,
225°, 270°, 315°)

0 % UT para 0,5 ciclo (0°,45°, 90°, 135°, 180°,
225°, 270°, 315°)

0 % UT forrara
250/300 ciclos (00)

0 % UT forrara
250/300 ciclos (00)

La calidad de la red eléctrica debe ser la deentornos de centros sanitarios y atención
sanitaria doméstica.

La calidad de la red eléctrica debe ser la deentornos de centros sanitarios y atención
sanitaria doméstica.

La calidad de la red eléctrica debe serla de entornos de centros sanitarios yatención sanitaria doméstica. Si el usuario
0 % UT para 1 ciclo (00)
0 % UT para 1 ciclo (00)del sistema FreeStyle Comfort requiere un
funcionamiento continuo durante cortes
70 % UT (30 % caída de
70 % UT (30 % caída de de la red eléctrica, se recomienda que eltensión en UT) para 25/30 tensión en UT) para 25/30 FreeStyle Comfort se alimente desde una
ciclos (00)ciclos (00)
fuente de alimentación sin interrupciones odesde una batería.

Los campos magnéticos a frecuencia
industrial deben tener un nivel
característico de una ubicación típica en un
3 A/m
3 A/m
entorno de centro sanitario o de asistencia
sanitaria doméstica.
NOTA: UT es el voltaje de CA de la red eléctrica antes de la aplicación del nivel de prueba.

Campo magnético
a frecuencia
industrial (50/60 Hz)
IEC 61000-4-8

NOTA: En algunos casos, el FreeStyle Comfort puede restablecerse después de una descarga electrostática notable, y es
posible que necesite hacer un ciclo completo de alimentación para reanudar el funcionamiento normal. Consulte la secciónde resolución de problemas para obtener más información.

PN MN234-C4 K | Manual del usuario

SPA - 123

FreeStyle Comfort
Orientación y declaración del fabricante: inmunidad electromagnéticaEl FreeStyle Comfort está destinado para su uso en el ambiente electromagnético que se especifica a continuación.
El cliente o el usuario del FreeStyle Comfortdeben asegurar que se use en tal ambiente.
Prueba de
Nivel de prueba IEC
Nivel de cumplimiento
Ambiente electromagnético: orientación
inmunidad
60601
Radiofrecuencia
10 Vrms
10 Vrms
Los equipos portátiles y móviles deconducidacomunicaciones por radiofrecuencia node 150 kHz a 80 MHz
deben usarse más cerca de las piezas delIEC 61000-4-6
FreeStyle Comfort, incluidos los cables, quela distancia de separación recomendadacalculada a partir de la ecuación aplicable a
10 V/m
10 V/mla frecuencia del transmisor.
Radiofrecuencia
de 80 MHz a 2700 MHz
irradiada
IEC 61000-4-3

Distancia recomendada de separaciónd = 1,2√ P

d = 1,2√ P = 80 MHz a 800 MHz
d = 1,2√ P = 800 MHz a 2700 MHz
donde P es la clasificación máxima depotencia de salida del transmisor en vatios(W) de acuerdo con el fabricante deltransmisor y d es la distancia de separación
recomendada en metros (m).
Las concentraciones de campo detransmisores fijos de RF, determinadas
por un estudio electromagnético delsitio, a deberían ser inferiores al nivel decumplimiento en cada rango de frecuencia. b
Puede producirse interferencia en lacercanía de los equipos marcados con elsímbolo siguiente:

NOTA 1: a 80 MHz y 800 MHz, se aplica el rango de mayor frecuencia.
NOTA 2: estas pautas pueden no ser de aplicación en todas las situaciones. La propagación electromagnética se ve
afectada por la absorción y la reflexión de estructuras, objetos y personas.
a
Las concentraciones de campo de los transmisores fijos, tales como las estaciones de base para teléfonos de radio(móviles/inalámbricos) y radios móviles terrestres, radios de aficionados y las transmisiones de radio de AM y FM y latransmisión de TV, no pueden predecirse teóricamente con precisión. Para evaluar el ambiente electromagnético debido
a transmisores fijos de RF, debe considerarse un estudio electromagnético del sitio. Si la concentración de campo medidaen la ubicación donde se usa el FreeStyle Comfort supera el nivel correspondiente de cumplimiento de RF anterior, el
FreeStyle Comfort debe observarse para verificar su funcionamiento normal. Si se observa un rendimiento anormal,
pueden necesitarse medidas adicionales, tales como la reorientación o la reubicación del sistema FreeStyle Comfort.
b

En el rango de frecuencia de 150 kHz a 80 MHz, las concentraciones de campo deben ser inferiores a 10 V/m.

124 - SPAPN MN232-C4 K | Manual del usuario

FreeStyle Comfort
Distancias recomendadas de separación entre equipos portátiles y móviles decomunicaciones de radiofrecuencia y el FreeStyle Comfort

El FreeStyle Comfort está destinado para su uso en un ambiente electromagnético en el que se controlan las interferencias
de radiofrecuencia irradiada. El cliente o el usuario del sistema FreeStyle Comfort pueden ayudar a prevenir lainterferencia electromagnética al mantener una distancia mínima entre los equipos portátiles y móviles de comunicación
por radiofrecuencia (transmisores) y el FreeStyle Comfort, tal como se recomienda a continuación, de acuerdo con lapotencia máxima de salida de los equipos de comunicaciones.
Potencia de salida
Distancia de separación de acuerdo con la frecuencia del transmisor
nominal máxima delmtransmisorde 150 kHz a 80 MHz
de 80 MHz a 800 MHz
de 800 MHz a 2,5 GHz

d = 1,2√ P

d = 1,2√ P

d = 1,2√ P

W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Para transmisores con una potencia nominal máxima de salida no mencionada anteriormente, la distancia de separación
recomendada d en metros (m) puede calcularse usando la ecuación aplicable a la frecuencia del transmisor, donde P es lapotencia nominal máxima de salida del transmisor en vatios (W) de acuerdo con el fabricante del transmisor.
NOTA 1: A 80 MHz y 800 MHz, se aplica la distancia de separación para el rango de mayor frecuencia.
NOTA 2: estas pautas pueden no ser de aplicación en todas las situaciones. La propagación electromagnética se ve afectada
por la absorción y la reflexión de estructuras, objetos y personas.

Clasificación

Sección 21 Categoría M.
	• CISPR 11 / EN 55011 Clase B Grupo 1, “Equipos
industriales, científicos y médicos”
	• FCC Parte 15, Subparte B – Clase B Radiadores
involuntarios

Grado de protección frente a descargas eléctricas:
Tipo BF Equipo que proporciona un grado
particular de protección frente a descargas eléctricas
en lo que respecta a		 1) corriente de fuga permitida;		2) fiabilidad de la puesta a tierra de seguridad(si está presente).
		 No está destinado para aplicación cardíaca directa.

IP22 - Protección contra la entrada de dedos u objetos
similares de diámetro superior a 12,5 mm. El goteo
vertical de agua no tendrá un impacto negativo si
la carcasa está inclinada en un ángulo de 15º de su
posición normal.

Tipo de protección frente a descargas eléctricas:
Clase II	La protección frente a descargas
eléctricas se consigue mediante un doble
aislamiento.

Pruebas independientes de estándares de equipos
eléctricos médicos:
Verificado para cumplir con el reglamento:• IEC 60601-1 edición 3.1: Equipos médicos
eléctricos, Parte 1: requisitos generales para laseguridad.
• CAN/CSA C22.2 N.º 60601-1-8 edición 2.1,
Equipos médicos eléctricos, Parte 1: requisitos
generales para la seguridad.
Protección contra interferencias electromagnéticas
potenciales o de otra índole entre el equipo y otros
dispositivos.
	• Probado para cumplir con el reglamento
EN 60601-1-2 /IEC 60601-1-2: 4ª edición.
	• Probado para cumplir con RTCA/DO160

Método permitido de limpieza y control de infección:
Consulte la sección “Limpieza, cuidados ymantenimiento adecuado” de este manual delusuario de FreeStyle.
Grado de seguridad de la aplicación en presencia degases anestésicos inflamables:
El equipo no está preparado para tal aplicación.
Modo de funcionamiento: Funcionamiento continuo.

CAIRE y CAIRE Inc. son marcas registradas de CAIRE Inc. Visite nuestro sitio web acontinuación para obtener una lista completa de marcas registradas.
Marcas registradas: www.careinc.com/corporate/trademarks.
Copyright © 2021 CAIRE Inc. CAIRE Inc. se reserva el derecho de suspender sus
productos o modificar los precios, los materiales, el equipo, la calidad, las descripciones,las especificaciones y/o los procesos de sus productos en cualquier momento, sin previo
aviso y sin ningún compromiso ni ninguna consecuencia. Queda reservado cualquierotro derecho que no se encuentre indicado explícitamente en este documento, según
corresponda.

PN MN234-C4 K | Manual del usuario

SPA - 125

FreeStyle Comfort

Controlli utente e indicatori di stato del sistema
ISO 7000; Simboli grafici per l’utilizzo
sull’apparecchiatura - Indice e sinossi
Leggere il Manuale dell’utente primadell’uso. N. reg. 1641
Intervallo di temperatura diconservazione o di esercizio.
N. reg. 0632
Intervallo di umidità di stoccaggio.
N. reg. 2620

Direttiva del Consiglio 93/42/CEE relativa ai dispositivi
mediciRappresentante autorizzato per la
Comunità Europea
Se sull’etichetta dell’identificatore
univoco del prodotto (UDI) del prodottoè riportato il simbolo CE ####, ildispositivo è conforme ai requisitidella Direttiva 93/42 / CEE relativa
ai dispositivi medici. Il simbolo CE#### indica il numero dell’organismo
notificato.
IEC 60417

Limite della pressione atmosferica.
N. reg. 2621

Apparecchio di Classe II, doppio
isolamento N. reg. 5172

Nome e indirizzo del produttore.
N. reg. 3082

Interruttore di alimentazione.
N. reg. 5009

Data di produzione. N. reg. 2497

Certificazione ETL
Agenzia per la sicurezza per
CAN/CSA C22.2 N. 601.1 M90 per
apparecchiature elettromedicali

Numero di catalogo. N. reg. 2493
Numero di serie. N. reg. 2498
Lato alto. N. reg. 0623

Simboli interni
Tenere lontano da materiali infiammabili,olio e grasso.

Fragile, maneggiare con cura.
N. reg. 0621

Non smontare.

Tenere lontano dalla pioggia e conservare
in luogo asciutto. N. reg. 0626

Incremento del flusso.

Limite di impilaggio per numero.
N. reg. 2403

Decremento del flusso.

ISO 7010: Simboli grafici – Colori di sicurezza e simbolidi sicurezza – Simboli di sicurezza registrati
Leggere il manuale di istruzioni.
N. reg. M002
Conservare lontano da fiamme libere,
fuoco e scintille. È vietato fumare eutilizzare fonti di accensione aperte.
N. reg. P003
Non fumare nei pressi dell’unità odurante il suo utilizzo. N. reg. P002
Parte applicata di tipo BF (gradodi protezione da scossa elettrica).
N. reg. 5333
Avvertenza. N. reg. W001

126 - ITLPN MN232-C4 K | Manuale dell’utente

Pulsante Menu.
Conforme a
RTCA/
DO-160
Sezione 21
Conforme
alla
Categoria M

Conforme a RTCA DO160 Sezione 21
Categoria M. FAA SFAR requisito 106

Apparecchio di Classe II, doppio
isolamento
Approvato FAA – POC

FreeStyle Comfort
21 CFR 801.15: Code of Federal Regulations, Title 21(Codice delle normative federali USA, Titolo 21)
La legge federale degli Stati Uniti limita
la vendita del presente dispositivo ai soli
medici o dietro prescrizione medica.
Direttiva del Consiglio 2012/19/UE: rifiuti diapparecchiature elettriche ed elettroniche (RAEE)RAEE

Cosa si intende per concentratore diossigeno
L’aria che respiriamo è composta all’incirca dal 21% diossigeno, 78% di azoto e 1% di altri gas. Nel sistema
FreeStyle® Comfort®, l’aria dell’ambiente viene
aspirata nella macchina attraverso gli ingressi dell’aria.
passando poi attraverso un materiale assorbente
denominato setaccio molecolare. Questo materiale
separa ossigeno e azoto, consentendo il passaggio solodell’ossigeno. Il risultato è l’erogazione all’utente di
un flusso di ossigeno ad alta concentrazione.
Il sistema FreeStyle Comfort combina le tecnologieavanzate dei concentratori di ossigeno e di conservazione
dell’ossigeno per creare un dispositivo portatile chepermette la massima portabilità e la deambulazione.
L’erogazione avanzata con flusso a impulsi rileva
rapidamente quando l’utente inspira ed eroga un
impulso di ossigeno all’inizio di ciascuna inalazione.
L’erogazione con flusso a impulsi è ideale per le attivitàe i periodi fuori casa, conserva l’ossigeno ed estende ladurata della batteria rispetto al flusso continuo. Il sistema
FreeStyle Comfort è abbastanza leggero da poter essere
trasportato con una tracolla. Può funzionare con CA (retedomestica), CC (veicolo) o a batteria.

Destinazione d’uso
Il concentratore di ossigeno Comfort FreeStyle
CAIRE è destinato alla somministrazione di ossigeno supplementare. Il dispositivo non è destinato al
supporto vitale né offre funzionalità di monitoraggio del paziente.

Perché il medico prescrivel’ossigenoterapia?
Molte persone soffrono di svariate patologie cardiache,polmonari e respiratorie di altro tipo. Un numero
significativo di queste persone può trarre beneficio daun’ossigenoterapia supplementare a casa, in viaggio odurante la partecipazione ad attività quotidiane lontanoda casa.
L’ossigeno è un gas che compone il 21% dell’aria che

IEC 60601-1: Apparecchi elettromedicali Parte 1
Prescrizioni generali per la sicurezza di base e leprestazioni essenziali
A prova di gocciolamento
Questo prodotto può essere coperto da uno o più
brevetti, validi negli Stati Uniti e all’estero. Visitare il
nostro sito Web, sotto indicato, per l’elenco dei brevetti
applicabili. Brev.: www.caireinc.com/corporate/
patents/

respiriamo; per funzionare correttamente, l’organismo
ne richiede una fornitura costante. Il medico prescrive
un flusso o un’impostazione per fronteggiare una
condizione respiratoria particolare.
Benché l’ossigeno sia un farmaco che non produce
dipendenza, l’ossigenoterapia non autorizzata puòessere pericolosa. È necessario consultare un medico
prima di utilizzare questo concentratore di ossigeno.
Il fornitore dell’apparecchio per l’ossigeno dimostreràcome impostare la portata prescritta.

Profilo operatore
I concentratori sono destinati a fornire ossigeno
supplementare agli utenti che soffrono di disagi
causati da disturbi che incidono sull’efficienza dei
polmoni nel trasferimento dell’ossigeno nell’aria alsangue. I concentratori di ossigeno portatili (POC)
non conservano né contengono ossigeno. Non devono
essere riforniti e possono essere ricaricati ovunquesia disponibile alimentazione CA o CC (tranne che abordo di un aeromobile). In questo modo l’utente èrelativamente autosufficiente in termini di utilizzo in
casa, deambulazione (sia all’interno sia all’esternodella casa), mobilità e stile di vita in generale.
L’utilizzo del concentratore di ossigeno richiede una
prescrizione del medico e non è destinato all’uso comesalvavita.
Sebbene l’ossigenoterapia possa essere
prescritta a utenti di tutte le età, il tipico utentedell’ossigenoterapia ha superato i 65 anni esoffre di svariate malattie respiratorie, tra cui la
broncopneumopatia cronica ostruttiva (BPCO).
Gli utenti in genere dispongono di buone capacità
cognitive e devono essere in grado di comunicare
il disagio. Se l’utente non è in grado di comunicare
il disagio o di leggere e comprendere le etichette e
le istruzioni per l’uso del concentratore, l’utilizzo è
consigliato solo sotto supervisione. In caso di disagio
durante l’uso del concentratore, gli utenti sono invitati
a rivolgersi al proprio fornitore di assistenza sanitaria.
Gli utenti sono inoltre invitati a disporre di una fonte diossigeno di riserva (ovvero, una bombola di ossigeno)
in caso di interruzione dell’alimentazione o guasto delconcentratore. Non sono previste altre abilità peculiario necessarie dell’utente per l’uso del concentratore.
PN MN232-C4 K | Manuale dell’utente

ITL - 127

FreeStyle Comfort

Estrazione del sistema FreeStyle Comfort dalla confezione
Verificare che la confezione
contenga tutti i componenti
elencati di seguito. In caso dielementi mancanti, contattare
immediatamente il fornitore diossigeno.

Accessori standard• Concentratore di ossigeno
portatile• Borsa per il trasporto• Tracolla• Gruppo batterie a 8 celle• Alimentatore CA• Cavo di alimentazione in CA• Alimentatore CC• Manuale dell’utente

Accessori opzionali• Gruppo batterie a 8 celle
aggiuntivo• Gruppo batterie a 16 celle• Caricatore desktop• Borsa per il trasporto degli
accessori

Cenni preliminari sul sistema FreeStyle Comfort
Acquisire familiarità con le funzionalità chiave del sistema FreeStyle Comfort e l’interfaccia utente

Attacco dellatracolla

Uscitadell’aria

Vista frontale
Interfaccia utente

Uscitadell’ossigeno

Ingresso dell’aria/
Filtri
Ingresso dell’aria/
Filtri
Linguettadella presadella porta dialimentazione
Gruppo batterie
Pulsante di sganciodel gruppo batterie
128 - ITLPN MN232-C4 K | Manuale dell’utenteFreeStyle Comfort
Vista dall’alto
Display
Incrementodel flusso

Pulsante diaccensione

Pulsante Menu

Decrementodel flusso

Display della batteria
Impostazione del flusso

Display dellabatteria

Indicatore autoSAT

Visualizzazione
allarme

Percentuale
di carica
rimanente

Indicatore
somministrazione
bolo

Display dell’alimentatore CA/CC

Impostazione delflussoArea divisualizzazione
degli allarmi

Display dellabatteria

Indicatore diricarica

PN MN232-C4 K | Manuale dell’utente

ITL - 129

FreeStyle Comfort

Importante!

Le istruzioni per la sicurezza sono definite
come indicato di seguito:
AVVERTENZA: IMPORTANTIINFORMAZIONI PER LA SICUREZZARELATIVE AI PERICOLI CHE POSSONOCAUSARE GRAVI LESIONI.
ATTENZIONE: informazioni importanti
per prevenire danni al sistema FreeStyle
Comfort.
Nota: informazioni che richiedono particolare attenzione.

Indicazioni per l’uso
AVVERTENZA: È MOLTO IMPORTANTESELEZIONARE SOLO IL LIVELLODI OSSIGENO PRESCRITTO. NON
MODIFICARE LA SELEZIONE DEL FLUSSOSE NON IN CASO DI PRECISE ISTRUZIONIDA PARTE DI UN MEDICO AUTORIZZATO.
AVVERTENZA: LA LEGGE FEDERALE (USA) LIMITALA VENDITA O IL NOLEGGIO DI QUESTO DISPOSITIVODIETRO PRESCRIZIONE DI UN MEDICO O DI UN
ALTRO PROFESSIONISTA SANITARIO QUALIFICATO.
AVVERTENZA: QUESTA UNITÀ NON DEVE ESSEREUTILIZZATA COME SUPPORTO SALVAVITA. I PAZIENTIGERIATRICI, PEDIATRICI O ALTRI UTENTI INCAPACIDI COMUNICARE I PROPRI DISTURBI DURANTEL’USO DI QUESTO DISPOSITIVO POTREBBERO
NECESSITARE DI UN MONITORAGGIO AGGIUNTIVO.
GLI UTENTI CON DANNI ALL’UDITO E/O ALLA VISTA
POTREBBERO NECESSITARE DI ASSISTENZA PERMONITORARE GLI ALLARMI.
AVVERTENZA: LE IMPOSTAZIONI DELLA MODALITÀDOSAGGIO IMPULSI DEVONO ESSERE DEFINITE PERCIASCUN UTENTE INDIVIDUALMENTE, IN FUNZIONEDELLE RELATIVE ESIGENZE A RIPOSO, DURANTE
L’ESERCIZIO FISICO E DURANTE I VIAGGI.
AVVERTENZA: ALCUNI SFORZI RESPIRATORIDEL PAZIENTE POTREBBERO NON ATTIVARE LAFUNZIONE DI CONSERVAZIONE.
AVVERTENZA: IL SISTEMA FREESTYLE COMFORTNON È ADATTO ALL’USO CON UN PAZIENTE
TRACHEOTOMIZZATO.
AVVERTENZA: LE DONNE INCINTE O CHE
ALLATTANO NON DOVRANNO USARE GLIACCESSORI RACCOMANDATI IN QUESTO MANUALE,
POICHÉ POSSONO CONTENERE FTALATI.

130 - ITLPN MN232-C4 K | Manuale dell’utente

Controindicazioni all’uso
AVVERTENZA: IN ALCUNE CIRCOSTANZE,L’USO DI OSSIGENO NON PRESCRITTO
PUÒ ESSERE PERICOLOSO. QUESTODISPOSITIVO VA UTILIZZATO SOLODIETRO PRESCRIZIONE MEDICA.
AVVERTENZA: NON UTILIZZARE IN PRESENZA DI
ANESTETICI INFIAMMABILI.
AVVERTENZA: ANALOGAMENTE A QUALSIASIALTRO DISPOSITIVO ELETTRICO, POTREBBERO
VERIFICARSI PERIODI DI INATTIVITÀ DOVUTIALL’INTERRUZIONE DELL’ALIMENTAZIONEELETTRICA, OPPURE ALLA NECESSITÀ DI FARRIPARARE IL CONCENTRATORE DI OSSIGENO DA
UN TECNICO QUALIFICATO. IL CONCENTRATOREDI OSSIGENO NON È INDICATO PER UTENTI CHE
A SEGUITO DELL’INTERRUZIONE TEMPORANEAPOTREBBERO SUBIRE CONSEGUENZE NEGATIVEPER LA PROPRIA SALUTE.

Linee guida per la sicurezza
AVVERTENZA: NON È CONSENTITOAPPORTARE ALCUNA MODIFICA ALDISPOSITIVO
AVVERTENZA: QUESTO DISPOSITIVO EROGAOSSIGENO A ELEVATA CONCENTRAZIONE CHE
FAVORISCE IL RAPIDO SVILUPPO DI USTIONI. NON
CONSENTIRE AL PAZIENTE DI FUMARE O DI USARE
FIAMME LIBERE ALL’INTERNO DELLA STANZA IN CUISI TROVA (1) QUESTO DISPOSITIVO O (2) QUALSIASIALTRO ACCESSORIO PER IL TRASPORTO
DELL’OSSIGENO. IL MANCATO RISPETTO DI QUESTA
AVVERTENZA PUÒ COMPORTARE GRAVI INCENDI EDANNI MATERIALI E/O PROVOCARE LESIONI FISICHEO IL DECESSO.
AVVERTENZA: NON UTILIZZARE L’UNITÀ IN SPAZICONFINATI O ANGUSTI (AD ESEMPIO UNA BORSA OUNA CUSTODIA DI PICCOLE DIMENSIONI) IN CUI LAVENTILAZIONE POSSA ESSERE LIMITATA. IN QUESTO
MODO IL CONCENTRATORE DI OSSIGENO
POTREBBE SURRISCALDARSI PREGIUDICANDO LE
PRESTAZIONI.
AVVERTENZA: IL CONCENTRATORE VA POSIZIONATO
IN MODO DA EVITARE FUMO, AGENTI INQUINANTI O
VAPORI.
AVVERTENZA: NEL CASO IN CUI SI VERIFICHI UN
GRAVE INCIDENTE CON QUESTO DISPOSITIVO,
L’UTENTE DEVE SEGNALARE IMMEDIATAMENTEL’INCIDENTE AL FORNITORE E / O AL PRODUTTORE.
UN INCIDENTE GRAVE È DEFINITO COME UNALESIONE, MORTE O POSSIBILITÀ DI CAUSARELESIONI / MORTE IN CASO DI RIPETIZIONE
DELL’INCIDENTE. L’UTENTE PUÒ ANCHESEGNALARE L’INCIDENTE ALL’AUTORITÀ
COMPETENTE NEL PAESE IN CUI SI È VERIFICATO
L’INCIDENTE.

FreeStyle Comfort
AVVERTENZA: L’USO DI ALCUNIACCESSORI PER LA SOMMINISTRAZIONEDI OSSIGENO NON ESPRESSAMENTE
INDICATI PER L’USO CON QUESTOCONCENTRATORE DI OSSIGENO PUÒ
PREGIUDICARNE LE PRESTAZIONI.
GLI ACCESSORI RACCOMANDATI SONORIPORTATI ALL’INTERNO DEL PRESENTEMANUALE.
AVVERTENZA: SE IL CONCENTRATORE DI OSSIGENO
VIENE FATTO CADERE, DANNEGGIATO O ESPOSTO
ALL’ACQUA, RIVOLGERSI AL FORNITORE DIASSISTENZA ATTREZZATURA PER UN’ISPEZIONEO UN’EVENTUALE RIPARAZIONE DEL DISPOSITIVO.
NON UTILIZZARE IL CONCENTRATORE DI OSSIGENOSE IL CAVO O LA SPINA DI ALIMENTAZIONE SONODANNEGGIATI.
AVVERTENZA: DURANTE IL TRASPORTO
ALL’INTERNO DI UN VEICOLO, FISSARE IL
CONCENTRATORE DI OSSIGENO IN MANIERAADEGUATA, MEDIANTE UNA CINTURA O UN FERMO DI
ALTRO TIPO, PER EVITARE DANNI O LESIONI.
AVVERTENZA: IL DEPARTMENT OF TRANSPORTATION(DOT) DEGLI STATI UNITI E I REGOLAMENTI DELLE
NAZIONI UNITE RICHIEDONO LA RIMOZIONEDELLA BATTERIA DAL DISPOSITIVO PER L’INTERO
VIAGGIO AEREO INTERNAZIONALE QUANDO IL
CONCENTRATORE DI OSSIGENO VIENE REGISTRATO
COME BAGAGLIO. DURANTE LA SPEDIZIONE DEL
CONCENTRATORE DI OSSIGENO, È NECESSARIORIMUOVERE LA BATTERIA DAL DISPOSITIVO E
IMBALLARLO IN MANIERA ADEGUATA.
AVVERTENZA: NON UTILIZZARE QUESTODISPOSITIVO SENZA PRIMA AVER LETTO ECOMPRESO IL PRESENTE MANUALE. IN CASODI DIFFICOLTÀ A COMPRENDERE LE AVVERTENZE
E LE ISTRUZIONI, RIVOLGERSI AL FORNITORE
DELL’APPARECCHIATURA PRIMA DI TENTAREDI UTILIZZARLA; IN CASO CONTRARIO, POSSONOVERIFICARSI DANNI O LESIONI.
AVVERTENZA: SE SI MANIFESTANO DISTURBI O
IN CASO DI EMERGENZA MEDICA, RIVOLGERSIIMMEDIATAMENTE A UN MEDICO.
AVVERTENZA: L’UTILIZZO DEL CONCENTRATOREDI OSSIGENO AL DI FUORI DELLE SPECIFICHE DITEMPERATURA DI ESERCIZIO PUÒ LIMITARE LACAPACITÀ DEL CONCENTRATORE DI SODDISFAREI VALORI SPECIFICATI PER LA CONCENTRAZIONE
DELL’OSSIGENO. PER I LIMITI DI TEMPERATURA,
FARE RIFERIMENTO ALLA SEZIONE DELLE
SPECIFICHE RIPORTATA ALL’INTERNO DEL PRESENTEMANUALE.

AVVERTENZA: NON FAR BLOCCARE LO SCARICOIN INGRESSO O IN USCITA DELL’ARIA. NON FARCADERE O INSERIRE ALCUN OGGETTO NELLE
APERTURE PRESENTI SUL DISPOSITIVO. IN QUESTO
MODO IL CONCENTRATORE DI OSSIGENO POTREBBESURRISCALDARSI PREGIUDICANDO LE PRESTAZIONI.
AVVERTENZA: L’USO NON CORRETTO DELLABATTERIA PUÒ CAUSARNE IL SURRISCALDAMENTO
E L’ACCENSIONE, PROVOCANDO GRAVI LESIONI.
PRESTARE ATTENZIONE A NON BUCARE, COLPIRE,SALIRE SOPRA O FAR CADERE LA BATTERIA, E NONESPORLA A IMPATTI O SOLLECITAZIONI FORTI. L’USO
DI UNA BATTERIA DANNEGGIATA PUÒ CAUSARELESIONI PERSONALI.
AVVERTENZA: NON CORTOCIRCUITARE I CONTATTI IN
METALLO DELLA BATTERIA CON OGGETTI METALLICI,COME CHIAVI O MONETE; POTREBBERO PRODURSISCINTILLE O UN CALORE ECCESSIVO.
AVVERTENZA: DURANTE L’UTILIZZO DELCONCENTRATORE DI OSSIGENO PORTATILE
ALL’APERTO CON L’ALIMENTATORE CA, COLLEGAREL’ALIMENTATORE SOLO A UNA PRESA CON
INTERRUTTORE DIFFERENZIALE (GFI).
AVVERTENZA: NON UTILIZZARE IL CONCENTRATOREDI OSSIGENO ACCANTO O IMPILATO SU ALTRE
APPARECCHIATURE. SE NON È POSSIBILE EVITARETALE MODALITÀ DI UTILIZZO, OSSERVARE ILDISPOSITIVO PER VERIFICARNE IL NORMALEFUNZIONAMENTO.
AVVERTENZA: L’USO DI CAVI E ADATTATORI DIVERSIDA QUELLI SPECIFICATI, A ECCEZIONE DI QUELLIVENDUTI DAL PRODUTTORE DEL DISPOSITIVOELETTROMEDICALE COME PARTI DI RICAMBIOPER I COMPONENTI INTERNI, PUÒ PROVOCAREUN AUMENTO DELLE EMISSIONI E UNA RIDUZIONE
DELL’IMMUNITÀ DEL CONCENTRATORE DI OSSIGENO.
AVVERTENZA: UTILIZZARE SOLO LA TENSIONEELETTRICA SPECIFICATA SULL’ETICHETTA DELLE
SPECIFICHE APPLICATA SUL DISPOSITIVO.
AVVERTENZA: NON UTILIZZARE PROLUNGHECON QUESTA UNITÀ O EVITARE DI COLLEGARETROPPE SPINE ALLA STESSA PRESA ELETTRICA.
L’USO DI PROLUNGHE PUÒ AVERE EFFETTI
AVVERSI SULLE PRESTAZIONI DEL DISPOSITIVO.
IL COLLEGAMENTO DI TROPPE SPINE A UN’UNICAPRESA PUÒ PROVOCARE UN SOVRACCARICO DELQUADRO ELETTRICO, CAUSANDO L’ATTIVAZIONE
DELL’INTERRUTTORE/FUSIBILE O UN INCENDIOIN CASO DI MANCATO FUNZIONAMENTO
DELL’INTERRUTTORE/FUSIBILE.

AVVERTENZA: NON UTILIZZARE PRODOTTI A BASE
DI OLIO, GRASSO O PETROLIO O ALTRI PRODOTTIINFIAMMABILI CON GLI ACCESSORI PER IL
TRASPORTO DELL’OSSIGENO O IL CONCENTRATOREDI OSSIGENO. UTILIZZARE ESCLUSIVAMENTELOZIONI O UNGUENTI A BASE D’ACQUA COMPATIBILICON L’OSSIGENO. L’OSSIGENO ACCELERA LACOMBUSTIONE DELLE SOSTANZE INFIAMMABILI.
PN MN232-C4 K | Manuale dell’utente

ITL - 131

FreeStyle Comfort
ATTENZIONE: in caso di allarme o qualora
si osservi un funzionamento non correttodel concentratore di ossigeno, consultare
la sezione Risoluzione dei problemi delpresente manuale. Se non si riesce arisolvere il problema, rivolgersi al fornitore
dell’apparecchiatura.
ATTENZIONE: per evitare l’annullamento della garanzia,
attenersi a tutte le istruzioni del produttore.
AVVERTENZA: NON TENTAREDI ESEGUIRE INTERVENTI DI
MANUTENZIONE, A ECCEZIONE DELLE
SOLUZIONI POSSIBILI ELENCATE
ALL’INTERNO DEL PRESENTE MANUALE.
NON RIMUOVERE LE COPERTURE, TALEOPERAZIONE O GLI INTERVENTI DI
ASSISTENZA DEVONO ESSERE ESEGUITIESCLUSIVAMENTE DAL FORNITORE
DELL’APPARECCHIATURA O DA UNTECNICO DI ASSISTENZA QUALIFICATO.
AVVERTENZA: IMPIEGARE ESCLUSIVAMENTEACCESSORI RACCOMANDATI DAL PRODUTTORE.
L’USO DI ALTRI ACCESSORI PUÒ ESSEREPERICOLOSO, CAUSARE GRAVI DANNI AL
CONCENTRATORE DI OSSIGENO E RENDERE NULLALA GARANZIA.
AVVERTENZA: IL PRODUTTORE CONSIGLIA DI
TENERE A DISPOSIZIONE UN’ALTRA SORGENTE
DI OSSIGENO SUPPLEMENTARE IN CASO DI
INTERRUZIONI DI CORRENTE, CONDIZIONI DI
ALLARME O GUASTI MECCANICI. PER CONOSCEREIL TIPO DI SISTEMA DI RISERVA NECESSARIO,RIVOLGERSI AL PROPRIO MEDICO O AL FORNITORE
DELL’APPARECCHIATURA.
AVVERTENZA: NON FAR BLOCCARE L’INGRESSOO GLI SFIATI DI USCITA DELL’ARIA. NON FARCADERE O INSERIRE ALCUN OGGETTO NELLE
APERTURE PRESENTI SUL DISPOSITIVO. IN
QUESTO MODO IL CONCENTRATORE DI OSSIGENO
POTREBBE SURRISCALDARSI PREGIUDICANDO LE
PRESTAZIONI.
AVVERTENZA: FUMARE DURANTE L’USO
DELL’OSSIGENO COSTITUISCE LA CAUSA NUMERO
UNO DI USTIONI E MORTI CORRELATE AGLI INCENDI.
È NECESSARIO ATTENERSI ALLE SEGUENTIAVVERTENZE DI SICUREZZA:
AVVERTENZA: NON CONSENTIRE AL PAZIENTEDI FUMARE O DI USARE CANDELE O FIAMME
LIBERE ALL’INTERNO DELLA STANZA IN CUI SI
TROVA IL DISPOSITIVO OPPURE QUALSIASI ALTROACCESSORIO PER L’EROGAZIONE DELL’OSSIGENO.
AVVERTENZA: FUMARE MENTRE SI INDOSSA UNACANNULA PER L’OSSIGENO PUÒ PROVOCAREUSTIONI AL VISO E PROBABILMENTE LA MORTE.
AVVERTENZA: SE SI FUMA, QUESTI 3 PASSAGGIPOSSONO SALVARE LA VITA: SPEGNERE IL
CONCENTRATORE DI OSSIGENO, RIMUOVERE LACANNULA E LASCIARE LA STANZA IN CUI SI TROVAIL DISPOSITIVO.
132 - ITLPN MN232-C4 K | Manuale dell’utente

AVVERTENZA: IN CASA O NEL LUOGO IN CUI ÈUTILIZZATO L’OSSIGENO OCCORRE AFFIGGERECARTELLI "NON FUMARE – OSSIGENO IN USO"
IN MODO CHE SIANO BEN VISIBILI. GLI UTENTI ECOLORO CHE PRESTANO ASSISTENZA DEVONOESSERE INFORMATI DEI PERICOLI DEL FUMO INPRESENZA O DURANTE L’UTILIZZO DI OSSIGENO
MEDICO.
AVVERTENZA: NON UTILIZZARE IL CONCENTRATOREDI OSSIGENO IN PRESENZA DI GAS INFIAMMABILI,POICHÉ POSSONO VERIFICARSI FENOMENIDI COMBUSTIONE RAPIDA, CHE POSSONOCOMPORTARE DANNI MATERIALI, LESIONICORPOREE O IL DECESSO.
AVVERTENZA: RIMUOVENDO LA CANNULA E
POSIZIONANDOLA SU INDUMENTI, BIANCHERIA,DIVANI O ALTRI MATERIALI IMBOTTITI SI POSSONOVERIFICARE LAMPI DI FUOCO IN CASO DIESPOSIZIONE A SIGARETTE, FONTI DI CALOREO FIAMME.
AVVERTENZA: NON LASCIARE LA CANNULA NASALE
SOPRA O SOTTO INDUMENTI, COPERTE O CUSCINIDI SEDIE. SE L’UNITÀ È ACCESA MA NON IN USO,L’OSSIGENO PUÒ RENDERE INFIAMMABILE IL
MATERIALE. ASSICURARSI CHE L’ALIMENTAZIONESIA SPENTA QUANDO IL CONCENTRATORE DIOSSIGENO NON È IN USO.
AVVERTENZA: IL CONCENTRATORE DI OSSIGENO
DEVE ESSERE SPENTO E SCOLLEGATO
DALL’ALIMENTAZIONE ESTERNA PRIMA DI
RIMUOVERE IL GRUPPO BATTERIE.
AVVERTENZA: LE PICCOLE PARTIPOSSONO PRESENTARE UN PERICOLODI SOFFOCAMENTO.
AVVERTENZA: EVITARE IL CONTATTO CON LA
BOCCHETTA DI SCARICO PER PIÙ DI 10 S PERCHÉPOTREBBE ESSERE CALDA.
AVVERTENZA: VENTO O CORRENTI D’ARIA
FORTI POSSONO INFLUIRE NEGATIVAMENTE
SULL’EROGAZIONE DELL’OSSIGENOTERAPIA.
AVVERTENZA: IL PRESENTE PRODOTTO PUÒESPORRE L’UTENTE A SOSTANZE CHIMICHE,TRA CUI IL NICHEL, CHE, SECONDO LO STATODELLA CALIFORNIA, PUÒ CAUSARE IL CANCRO.
PER MAGGIORI INFORMAZIONI, VISITARE
WWW.P65WARNINGS.CA.GOV.
ATTENZIONE: scollegare sempre
l’alimentatore CA dalla presa di rete
prima di scollegarlo dal concentratore diossigeno.
ATTENZIONE: non posizionare il concentratore diossigeno in modo che il cavo di alimentazione sia
difficile da raggiungere.

FreeStyle Comfort
ATTENZIONE: sistemare sempre i tubi di mandata
dell’ossigeno e i cavi di alimentazione in modo daprevenire pericoli di inciampo o lo strangolamento
accidentale.
ATTENZIONE: quando si utilizza il concentratore
di ossigeno in auto, in barca o con altre sorgenti
CC tramite l’alimentatore CC, accertarsi di avviare
e mettere in funzione il veicolo prima di collegare il
concentratore di ossigeno. La mancata osservanzadi tali istruzioni può comportare la mancata
alimentazione del concentratore di ossigeno da partedell’alimentatore.
ATTENZIONE: scollegare e rimuovere il concentratore
di ossigeno quando si spegne l’automobile in cui si
utilizza l’unità. Non conservare il concentratore diossigeno in un’auto molto calda o molto fredda o in
altri ambienti simili a temperatura molto alta o bassa.
Nota: le apparecchiature mobili di comunicazione a RF
possono avere effetti sui dispositivi elettromedicali.
ATTENZIONE: NON lasciare il
concentratore di ossigeno e l’alimentatore
collegati al veicolo se il sistema diaccensione è spento. Si potrebbe
scaricare la batteria del veicolo.
ATTENZIONE: l’utilizzo o la conservazione delconcentratore di ossigeno al di fuori del normale
intervallo di temperatura operativa o di pressioneambientale (altitudine) può influire sulle prestazioni
e ridurre il tempo di funzionamento della batteria
e/o aumentarne il tempo di ricarica. Per i limiti ditemperatura di funzionamento e conservazione, fareriferimento alla sezione delle specifiche riportata
all’interno del presente manuale.
ATTENZIONE: per i concentratori di ossigeno dotati
di batterie: conservare in un luogo fresco e asciutto
per contribuire a garantire la longevità della batteria.
La conservazione del concentratore di ossigeno
per periodi prolungati a temperature elevate o conla batteria completamente carica o completamente
scarica può ridurre la durata complessiva dellabatteria. Non tentare di aprire la batteria, che noncontiene componenti riparabili. Tenere le batterie
lontano dai bambini.
ATTENZIONE: UTILIZZARE ESCLUSIVAMENTE
le batterie fornite dal produttore. Per il corretto
smaltimento della batteria, rivolgersi al fornitore
dell’apparecchio o all’ente governativo preposto per i
requisiti di smaltimento.

Nota: accertarsi che la cannula sia completamente
inserita e salda. In questo modo il concentratore diossigeno è in grado di rilevare correttamente l’inspirazione
per l’erogazione dell’ossigeno. Durante l’inalazione, si
deve udire o avvertire il flusso di ossigeno attraverso labiforcazione della cannula nasale.
Nota: le batterie al litio possono perdere in modo
permanente la capacità se esposte a temperature
estremamente elevate con le batterie completamente
cariche o completamente esaurite. Per la conservazione
prolungata, si raccomanda di caricare le batterie tra il 25
e il 50% e di conservarle a una temperatura di -20 °C –20 °C (-4 °F – 68 °F).
Nota: l’uso di alcuni accessori per la somministrazione
di ossigeno non espressamente indicati per l’uso conquesto concentratore di ossigeno può pregiudicarne leprestazioni. Gli accessori raccomandati sono riportati
all’interno del presente manuale.
Nota: non utilizzare il concentratore di ossigeno senza ilfiltro di ingresso dell’aria in posizione. Se è disponibile unsecondo filtro, inserire il filtro di ricambio prima di pulire
quello sporco. Pulire il filtro sporco con una soluzione
tiepida di acqua e sapone e asciugare a fondo primadell’uso.
Nota: il produttore non raccomanda la sterilizzazione diquesto dispositivo.
Nota: non c’è mai il rischio di esaurire l’ossigeno in una
stanza quando si utilizza l’unità del concentratore diossigeno.
AVVERTENZA: IL CONCENTRATOREDI OSSIGENO NON VA UTILIZZATO OCOLLEGATO ALL’ALIMENTAZIONEESTERNA (CA/CC) SENZA LA BATTERIAINSTALLATA. NON TENTARE DI
ACCENDERE L’UNITÀ O COLLEGARLAALL’ALIMENTAZIONE ESTERNA QUANDOIL GRUPPO BATTERIE È RIMOSSO.
RIMUOVERE IL GRUPPO BATTERIE SOLO QUANDOL’APPARECCHIO È SPENTO E SCOLLEGATO
DALL’ALIMENTAZIONE. NON TENTARE DIRIMUOVERE IL GRUPPO BATTERIE MENTRE
L’UNITÀ È ANCORA ACCESA O COLLEGATA
ALL’ALIMENTAZIONE.
Nota: se il concentratore di ossigeno è stato conservato
per un periodo prolungato al di fuori delle normali
condizioni operative ambientali indicate nella tabella
precedente, prima dell’accensione attendere che torni
alla normale temperatura di funzionamento.

PN MN232-C4 K | Manuale dell’utente

ITL - 133

FreeStyle Comfort

Specifiche

FreeStyle Comfort
Dimensioni (con batteria singola)

25,4 x 18,5 x 7,9 cm (10,0 x 7,3 x 3,1 poll.)

Dimensioni (con batteria doppia)

27,9 x 18,5 x 7,9 cm (11,0 x 7,3 x 3,1 poll.)

Peso (con gruppo batterie singolo)

2,3 kg (5 lb)

Peso (con gruppo batterie doppio)

2,7 kg (6 lb)

Rumorosità nominale***39,93 dB (A) ± 0,42 dB (A) @ 2

Impostazioni di flusso

Impostazione di flusso 1, 2, 3, 4, 5

Impostazione di uscita O2(@ 20 bpm)

Concentrazione di ossigeno*
Monitor dell’ossigeno
Pressione di uscita massima
Sensibilità di dosaggio predefinita

Impostazione 1 - 210 ml/min
Impostazione 2 - 420 ml/min
Impostazione 3 - 630 ml/min
Impostazione 4 - 840 ml/min
Impostazione 5 - 1050 ml/min
90% (+5,5%/-3%)
Sì<30 PSI> -0,1 cm H2O (regolabile)

Alimentazione CA

100-240 VCA, 50-60 Hz, 120W Max

Alimentazione CC

11–18 VCC (10 A max)

Tempo di ricarica della batteria

Batteria singola: 3,5 ore
Batteria doppia: 6,0 ore

Durata dei gruppi batterie
Temperatura di esercizio**
Tipo di batteria
Capacità della batteria

Durata batteria singola: Impostazione 2 - fino a 4 ore
Durata batteria doppia: Impostazione 2 - fino a 8 ore
5 °C - 40 °C (41 °F - 104 °F)
Loni di litio
Singola cella: 6700 mAh; Doppia cella: 13400 mAh

Temp. di trasporto e conservazione(unità)Da -25 a +70 °C (da -13 a +158 °F), fino a 3048 m (10000 ft)

Temp. di trasporto e conservazione(batteria)

Da -25 a +70 °C (da -13 a +158 °F), fino a 3048 m (10000 ft)

Altitudine di esercizio**
Umidità di esercizio
Umidità di conservazione
Garanzia limitata

Da -381 a 3048 m (da -1250 a 10.000 ft) (testata a 700 – 1060 hPa)
15 - 95% di umidità relativa (senza condensa)
0 - 90% senza condensa
Unità: 3 anni
1 anno su batteria e alimentatori* Sulla base di una pressione atmosferica di 101 kPa (14,7 psi) a 21 °C (70 °F)** L’unità funzionerà in tutti gli ambienti purché l’ambiente si trovi con le gamme di temperatura e umidità specificate. Il
funzionamento al di fuori di queste specifiche operative può limitare la capacità del concentratore di soddisfare le specifiche diconcentrazione dell’ossigeno alle portate più elevate in litri. La schermata del riscaldamento può continuare a comparire finché
il concentratore non ha raggiunto la concentrazione desiderata o per un periodo massimo di 15 minuti. Continuare a usare l’unità
durante questo periodo.
*** Livello sonoro misurato per metodo di prova n. 14-1 10/2018 MDS-Ciao.
La durata prevista è di almeno cinque anni. La durata di conservazione dei setacci molecolari è di circa un anno se il concentratore
non viene utilizzato durante l’intero periodo.
134 - ITLPN MN232-C4 K | Manuale dell’utenteFreeStyle Comfort
Istruzioni per l’uso
1. Collocare e sistemare il sistema FreeStyle Comfort
in una posizione appropriata per l’uso.
a. Il concentratore deve essere ben ventilato inmodo che i filtri di ingresso dell’aria e l’uscita
dell’aria non siano ostruiti.
b. Il concentratore deve essere posizionato inmodo che tutti gli indicatori acustici e visivi
e gli allarmi possano essere facilmente visti esentiti.
c. Prima di utilizzare l’unità, accertarsi che i filtri
di ingresso dell’aria siano in posizione. Se il
filtro del filtro dell’unità manca, contattare il
distributore dell’apparecchio.

3. Inserire l’alimentazione CA o CC nella presa dellaporta di alimentazione sulla parte anteriore, in basso
a destra dell’unità.

2. Installare un gruppo batterie nel concentratore.
a. Allineare la batteria con l’alloggiamento inferiore
dal centro del sistema FreeStyle Comfort.

4. Collegare una cannula nasale o il raccordo di uscita
dell’ossigeno. Assicurarsi che sia ben collegata primadell’utilizzo.
NOTA: l’uso di cannule diverse da quelle consigliate può
incidere sulle prestazioni.

b. Faccia scorrere la batteria verso l’interno fino asentire un clic a conferma dell’avvenuto blocco
in sede.

Nota: al fornitore dell’apparecchiatura: si consiglia diutilizzare i seguenti accessori per la somministrazione diossigeno con il FreeStyle Comfort:• Cannula nasale con tubo da 2,1 m (7 piedi) (LPM
massimo: 6 LPM): numero parte CAIRE 5408-SEQ• Ulteriori informazioni sugli accessori consigliati sono
disponibili online all’indirizzo www.caireinc.com.

PN MN232-C4 K | Manuale dell’utente

ITL - 135

FreeStyle Comfort
5. Accendere il sistema FreeStyle Comfort premendo
il pulsante di accensione e selezionare l’impostazionedel flusso prescritta utilizzando i pulsanti diincremento (+) e decremento (-).

Impostazione volume di impulso autoSAT(ml)*
Frequenza
respiratoria

1

2

3

4

5

15

10,5

21,0

31,5

42,0

52,5

20

10,5

21,0

31,5

42,0

52,5

25

10,5

21,0

31,5

42,0

42,0

30

10,5

21,0

31,5

35,0

35,0

35

10,5

21,0

30,0

30,0

30,0

40

10,5

21,0

26,3

26,3

26,3

*La tolleranza è pari a +/- 15%

Impostazione volume di impulso non
autoSAT (ml)*Si udirà un allarme acustico indicante che ildispositivo è acceso.
Nota: all’accensione iniziale del sistema FreeStyle
Comfort, l’impostazione del flusso predefinita è la n. 2.
Se l’impostazione viene modificata, l’unità si accenderà
all’impostazione del flusso precedente.
6. Posizionare la cannula nasale sul viso secondo leistruzioni del produttore fornite con la cannula e
respirare normalmente. Si dovrebbe percepire unsoffio di ossigeno erogato attraverso la cannula
nasale ogni volta che si inala.
7. Per modificare l’impostazione del flusso, prema
il pulsante di incremento (+) o decremento (-) finoa ottenere la portata desiderata. È normale sentire
una differenza di suono quando si modificano leimpostazioni del flusso.

Frequenza
respiratoria

1

2

3

4

5

15

14,0

28,0

42,0

56,0

70,0

20

10,5

21,0

31,5

42,0

52,5

25

8,4

16,8

25,2

33,6

42,0

30

7,0

14,0

21,0

28,0

35,0

35

6,0

12,0

18,0

24,0

30,0

40

5,3

10,5

15,8

21,0

26,3

8. Spegnere il sistema FreeStyle Comfort premendo il
pulsante di accensione/spegnimento.
Nota: lo schermo del display entrerà nella modalità distandby e si spegnerà dopo 20 s. Premere qualsiasi
pulsante per fare uscire lo schermo dalla modalità distandby.
9. Per le funzionalità avanzate consultare le istruzionidel menu di navigazione

FUNZIONALITÀ SPECIALI DEL PRODOTTO
autoSAT®
La funzione autoSAT proprietaria di CAIRE consentea FreeStyle Comfort di erogare un volume costante
di ossigeno (bolo) per respiro, fino a una frequenza
respiratoria massima. Man mano che la frequenza
respiratoria dell’utente aumenta durante attività
come la deambulazione, la funzione autoSAT regola
il funzionamento del dispositivo per mantenere unvolume di bolo impostato.
UltraSense®
Questo sensibile erogatore del dosaggio a impulsi
rileva la variazione di pressione nella respirazione ederoga l’impulso di ossigeno durante l’inalazione.
autoDOSE™
In presenza di un “allarme respiro assente”, questa
funzionalità pulsa ossigeno a 20 respiri al minuto, inbase all’impostazione del flusso attuale dell’unità, fino
al termine dello stato di allarme.
136 - ITLPN MN232-C4 K | Manuale dell’utenteFreeStyle Comfort
AlimentatoriÈ possibile alimentare il sistema FreeStyle Comfort intre modi:• Gruppo batterie ricaricabile• Alimentazione CA• Alimentazione CC
Le informazioni sul funzionamento di ciascuno di questi
componenti sono riportate nelle sezioni seguenti.
AVVERTENZA: UTILIZZARE SOLO GLIALIMENTATORI DESCRITTI IN QUESTO
MANUALE DELL’UTENTE. L’UTILIZZO DIALIMENTATORI E GRUPPI BATTERIE NONCAIRE PUÒ CAUSARE DANNI ALL’UNITÀ E
UN FUNZIONAMENTO NON CORRETTO.

RIMOZIONE DELLA BATTERIA

Tirare il pulsante di rilascio della batteria verso ilbasso allontanandolo dal sistema FreeStyle Comfort.
TENENDO PREMUTO questo pulsante, far scorrere labatteria verso l’esterno estraendola dal concentratore.
AVVERTENZA: IL CONCENTRATORE DIOSSIGENO DEVE ESSERE SPENTO ESCOLLEGATO DALL’ALIMENTAZIONEESTERNA PRIMA DI RIMUOVERE IL
GRUPPO BATTERIE.

GRUPPO BATTERIE

Il sistema FreeStyle Comfort può essere alimentato
dal gruppo batterie ricaricabile che viene fornito con
il dispositivo. Non sono previste limitazioni operative
al concentratore quando è alimentato tramite il
gruppo batterie; sono disponibili tutte le portate con
l’alimentazione a batteria.

CONTROLLO DELLA RICARICA DEL
GRUPPO BATTERIEINSTALLAZIONE DELLA BATTERIA

Allineare la batteria con l’alloggiamento inferiore
dal centro del sistema FreeStyle Comfort. Premere
la batteria verso l’interno fino a sentire un clic per
confermare l’avvenuto blocco in sede.

Per verificare il livello di carica della batteria quando
questa è installata nel sistema FreeStyle Comfort,
fare riferimento alla relativa icona e alla percentualerimanente presenti sul display.
80–100%60-80%

40-60%

20-40%

20%

PN MN232-C4 K | Manuale dell’utente

ITL - 137

FreeStyle Comfort
Nota: se la batteria si sta caricando, anziché la percentuale
residua, comparirà un indicatore della spina.
Per controllare la carica della batteria quando questa nonè installata nel sistema FreeStyle Comfort, metterla inpiano e premere il pulsante TEST. Le spie dell’indicatoredella batteria (25-100%) si accendono a destra delpulsante TEST per indicare il livello di carica.

DURATA DEL FUNZIONAMENTO A BATTERIA
Quando l’alimentazione CA o CC esterna èscollegata, il concentratore passerà automaticamente
al funzionamento a batteria se il gruppo batterie è
installato ed è carico. La tabella seguente mostra
le durate tipiche di un gruppo batterie nuovo.
Le durate del gruppo batterie rimarranno costantiindipendentemente dalla frequenza respiratoria.
Impostazione
degli impulsi

Durata batteria
singola (8 celle)

Durata batteria
doppia (16 celle)1fino a 8 ore

fino a 16 ore

2fino a 4 ore

fino a 8 ore

3fino a 3 ore

fino a 6 ore

4fino a 2,25 ore

fino a 4,5 ore

5fino a 2 orefino a 4 ore

TEMPO DI RICARICA TIPICO DELLABATTERIACARICA INIZIALE

La nuova batteria in dotazione con il sistema FreeStyle
Comfort è carica solo parzialmente quando è spedita
dalla fabbrica. Prima di utilizzarla per la prima volta, ènecessario ricaricare completamente il gruppo batterieutilizzando l’alimentazione CA o CC.

INFORMAZIONI GENERALI DI RICARICA

A meno che non si trovi in modalità aereo, il sistema
FreeStyle Comfort ricarica il gruppo batterie ogni
volta che è installato nel concentratore e quest’ultimo
è collegato a una sorgente di alimentazione CA o CC
esterna.
La batteria è completamente carica quando viene
visualizzato 100%.
In alternativa, è possibile ricaricare il gruppo batterieutilizzando il caricabatterie desktop opzionale.

138 - ITLPN MN232-C4 K | Manuale dell’utente

Normalmente il gruppo batterie impiega 3,5 ore per
una batteria singola e 6,0 ore per una batteria doppia
per ricaricarsi completamente.
Nota: la batteria inizialmente potrebbe non avviare laricarica se il gruppo batterie è completamente scarico.

DURATA DELLA BATTERIA

Le batterie del sistema FreeStyle Comfort sono ideate
per durare per 300 cicli di carica/scarica.

FreeStyle Comfort
Alimentatore CA e CC
Il sistema Freestyle Comfort è dotato di alimentatori
CA e CC.
L’alimentatore CA include gli elementi seguenti:
a. Alimentatore CA che funziona tra i 100 e i
240 volt, 50-60 Hz.
b. Un cavo di alimentazione CA per collegare
l’alimentatore alla presa CA corrispondente.

L’alimentatore CC è composto da un cavo che sicollega direttamente all’unità a un’estremità, e alla
presa CC all’altra estremità.

ALIMENTAZIONE CA

2. Collegare il cavo di alimentazione CA
all’alimentatore CA.

L’alimentazione CA è per uso domestico o ovunquesia disponibile una presa di alimentazione standard
CA. Il sistema FreeStyle Comfort è completamentefunzionante con alimentazione CA. Il sistema
FreeStyle Comfort caricherà il gruppo batterieogni volta che è disponibile un’alimentazione CA(a eccezione di quando si trova nella modalità aereo).
Il gruppo batterie si ricarica se il sistema FreeStyle
Comfort è acceso o spento.
AVVERTENZA: NON COLLEGAREL’ALIMENTAZIONE ESTERNA SENZA UNGRUPPO BATTERIE INSTALLATO NEL
CONCENTRATORE DI OSSIGENO.
Per collegare il sistema FreeStyle Comfort
all’alimentazione CA:
1. Collegare l’alimentatore CA alla presa delconnettore di alimentazione sulla parte anteriore,in basso a destra del sistema FreeStyle Comfort.
Assicurarsi di spingere verso l’interno in modo chela connessione sia sicura.

3. Collegare l’estremità opposta del cavo dialimentazione CA a una presa a parete o appropriata
di altro tipo.

PN MN232-C4 K | Manuale dell’utente

ITL - 139

FreeStyle Comfort
4. Accendere il sistema FreeStyle Comfort premendo
il pulsante di accensione e selezionare l’impostazionedel flusso prescritta utilizzando i pulsanti diincremento (+) e decremento (-).

Nota: se il gruppo batterie deve essere ricaricato,l’icona della batteria sul display conterrà solo una barra
arancione.

ALIMENTAZIONE CC

L’alimentazione CC è destinata all’uso nelle prese
accessorie delle automobili, delle barche o di altriveicoli a motore. Il sistema FreeStyle Comfort ècompletamente funzionante con alimentazione CC.
Il sistema FreeStyle Comfort ricarica anche il gruppo
batterie se necessario ogniqualvolta sia disponibile
l’alimentazione CC. La batteria può essere ricaricata
sia se l’apparecchio è spento sia mentre è in uso.
AVVERTENZA: NON COLLEGAREL’ALIMENTAZIONE ESTERNA SENZA UNGRUPPO BATTERIE INSTALLATO NEL
CONCENTRATORE DI OSSIGENO.
AVVERTENZA: NON UTILIZZARE IL DISPOSITIVO CON
LE PRESE A 12 VCC PRESENTI NELLE AUTOMOBILISE IL VEICOLO È SPENTO.
1. Accendere il veicolo a motore.
2. Collegare l’alimentatore CC alla presa delconnettore di alimentazione sulla parte anteriore,in basso a destra del sistema FreeStyle Comfort.
Assicurarsi di spingere verso l’interno in modo chela connessione sia sicura.

140 - ITLPN MN232-C4 K | Manuale dell’utente

3. Collegare l’estremità opposta del cavo dialimentazione CC alla presa dell’automobile oaccessoria CC di altro tipo.

4. Accendere il sistema FreeStyle Comfort premendo
il pulsante di accensione e selezionare l’impostazionedel flusso prescritta utilizzando i pulsanti diincremento (+) e decremento (-).
Nota: se il gruppo batterie deve essere ricaricato,l’icona della batteria sul display conterrà solo una barra
arancione.

FreeStyle Comfort
Pulizia, cura e manutenzione
ordinaria
MANUTENZIONE ORDINARIA

Manutenzione preventiva
Il dispositivo non richiede manutenzione. I filtri
all’interno del dispositivo hanno lo scopo di durare
per tutta la vita del dispositivo. Se il dispositivo non
funziona correttamente, l’unità deve essere restituita
al fornitore per la riparazione. La manutenzione deicomponenti interni all’interno dell’alloggiamento delsistema FreeStyle Comfort deve essere effettuata dal
fornitore di ossigeno.
L’operatore dell’unità è responsabile delle seguenti
procedure di manutenzione:• Settimanale – Pulire i filtri di ingresso dell’aria
AVVERTENZA: NON TENTARE DI APRIREO RIMUOVERE I COPERCHI DI PLASTICA
DEL CONCENTRATORE DI OSSIGENO.
L’UNITÀ DEVE ESSERE APERTA SOLODA UN TECNICO DI ASSISTENZAQUALIFICATO.

FILTRI DI INGRESSO DELL’ARIA

L’aria entra nel sistema FreeStyle Comfort attraverso ifiltri di ingresso dell’aria, in basso nella parte anterioredel concentratore di ossigeno. Questi filtri impediscono
alla polvere e ad altre particelle di grandi dimensioni
presenti nell’aria di entrare nell’unità.
I filtri di ingresso dell’aria vanno puliti almeno una
volta alla settimana. Il fornitore di ossigeno può
consigliare di pulirli più frequentemente, a secondadelle condizioni operative. Attenersi a questa procedura
per pulire correttamente i filtri di ingresso dell’aria.
1. Rimuovere i due filtri dalla parte anteriore delsistema FreeStyle Comfort spingendo verso l’alto la
linguetta corrispondente e tirando verso l’esterno.

AVVERTENZA: NON UTILIZZARE IL
CONCENTRATORE DI OSSIGENO SENZA
IL FILTRO DI INGRESSO DELL’ARIA IN
POSIZIONE.

CURA DEL GRUPPO BATTERIE

Non esporre il gruppo batterie ai liquidi.
Non tentare di smontare o aprire il gruppo batterie.
Non esporre il gruppo batterie a fiamme libere.
Non far cadere il gruppo batterie.
Tenere il gruppo batterie lontano dai bambini.
Se il gruppo batterie appare danneggiato, rivolgersi al
fornitore di ossigeno prima dell’uso.
Non lasciare mai la batteria all’interno o vicino a un
forno a microonde o ad altri apparecchi di cottura.

CONSERVAZIONE DEL GRUPPO BATTERIE
Non è necessario che la batteria del concentratore
di ossigeno sia completamente scarica prima dellaricarica. Si consiglia di ricaricare la batteria delconcentratore di ossigeno dopo ciascun utilizzo.

Non lasciare la batteria in un luogo esposto a
calore eccessivo, come la luce solare diretta o in
auto. Ciò potrebbe incidere negativamente sulleprestazioni e sulla durata della batteria. Può anche
causare l’emissione di calore, fumo, l’accensione ol’esplosione della batteria. Il gruppo batterie deveessere conservato in un ambiente stabile, caratterizzato
da bassa umidità (umidità relativa inferiore al 70%) e
privo di gas corrosivi. Per prevenire la ruggine, evitarecondizioni che possono creare condensa come rapide
fluttuazioni della temperatura ambiente.
Per la conservazione per periodi superiori o pari a unmese, si raccomanda una carica parziale tra 50% e 75%(due o tre LED). Una volta ricaricata, rimuovere labatteria dal concentratore di ossigeno.

2. Lavare i filtri in una soluzione di acqua calda esapone.
3. Sciacquare bene i filtri e rimuovere l’acqua ineccesso con un panno morbido privo di pelucchi.
Assicurarsi che i filtri siano completamente asciutti
prima di riposizionarli.
4. Reinserire i filtri asciutti e puliti.
PN MN232-C4 K | Manuale dell’utente

ITL - 141

FreeStyle Comfort
PULIZIA

Spegnere l’unità e scollegarla dall’alimentazione
CA o CC prima di qualsiasi attività di pulizia. NON
spruzzare direttamente alcun prodotto sul telaio
esterno. Utilizzare una spugna o un panno umido (nonbagnato). Per pulire l’alloggiamento e gli alimentatori,
spruzzare una soluzione detergente per la casa delicata
sul panno o sulla spugna. Procedere come indicato dal
produttore del detergente.
AVVERTENZA: SPEGNERE L’UNITÀ E
SCOLLEGARE IL CAVO DI ALIMENTAZIONEDALLA PRESA ELETTRICA PRIMA DI
PULIRE L’UNITÀ PER PREVENIRE IL
PERICOLO DI SCOSSE ELETTRICHE E
USTIONI ACCIDENTALI. GLI INTERVENTIDI RIPARAZIONE E LA RIMOZIONE DELLE
COPERTURE DEVONO ESSERE ESEGUITIESCLUSIVAMENTE DAL FORNITORE
DELL’APPARECCHIO O DA UN TECNICO DI
ASSISTENZA QUALIFICATO.
AVVERTENZA: PRESTARE ATTENZIONE A EVITARE
CHE IL CONCENTRATORE DI OSSIGENO E IL
GRUPPO BATTERIE SI BAGNINO O CHE FLUIDIENTRINO NELL’UNITÀ. CIÒ PUÒ PROVOCARE
UN MALFUNZIONAMENTO O UN ARRESTO E UN
AUMENTO DEL RISCHIO DI USTIONI O SCOSSE
ELETTRICHE.
AVVERTENZA: NON UTILIZZARE LIQUIDIDIRETTAMENTE SULL’UNITÀ. SEGUE UN
ELENCO NON ESAUSTIVO DI AGENTI CHIMICINON CONSIGLIATI: ALCOL E PRODOTTI A BASE
ALCOLICA, PRODOTTI CONCENTRATI A BASE DI
CLORO (CLORURO DI ETILENE) E PRODOTTI A BASE
OLEOSA. QUESTI NON DEVONO ESSERE UTILIZZATIPER PULIRE L’ALLOGGIAMENTO IN PLASTICA
SUL CONCENTRATORE, POICHÉ POSSONODANNEGGIARE L’UNITÀ.
AVVERTENZA: PULIRE IL CONTENITORE,IL PANNELLO DI CONTROLLO E IL CAVO DI
ALIMENTAZIONE ESCLUSIVAMENTE CON UN
DETERGENTE DELICATO PER USO DOMESTICOAPPLICATO CON UNA SPUGNA O UN PANNOINUMIDITI (NON BAGNATI), QUINDI ASCIUGARETUTTE LE SUPERFICI. EVITARE L’INFILTRAZIONEDI LIQUIDI ALL’INTERNO DEL CONCENTRATORE.
PRESTARE PARTICOLARE ATTENZIONE ALLAPRESA DELL’OSSIGENO PER ASSICURARSI CHE IL
COLLEGAMENTO DELLA CANNULA RESTI LIBERO
DA POLVERE, ACQUA E PARTICELLE.

142 - ITLPN MN232-C4 K | Manuale dell’utente

AVVERTENZA: UTILIZZARE UNA SPUGNA O UN
PANNO INUMIDITI (NON BAGNATI) PER PULIRE IL
GRUPPO BATTERIE. SPRUZZARE IL DETERGENTE
DELICATO SUL PANNO O SULLA SPUGNA, QUINDI
PULIRE L’ALLOGGIAMENTO DEL GRUPPO BATTERIE.
AVVERTENZA: NON PULIRE CON UN SISTEMA DI
LAVAGGIO AUTOMATIZZATO.

SOSTITUZIONE DELLA CANNULA

Per l’utilizzo adeguato, attenersi sempre alle istruzionidel produttore della cannula. Sostituire la cannula nasaleo il tubo dell’ossigeno ogni 3-6 mesi (ogni 3 mesi in caso
di utilizzo quotidiano) secondo i consigli del produttore
come consigliato dal produttore della cannula o dal
fornitore di ossigeno. Il medico o il fornitore di ossigeno
forniranno le informazioni riguardanti la pulizia e lasostituzione.
Le forniture aggiuntive per la sostituzione sono
disponibili presso il fornitore di ossigeno.

FreeStyle Comfort
Condizioni di allarmeN.
priorità

Descrizione allarme

1

Malfunzionamento
sistema, altatemperatura dellabatteria

Display

SegnaleacusticoPiù dettagli / azione suggerita

1 bip ogni
20 s

L’unità entra in allarme quando la batteria si sta
avvicinando alla soglia di arresto di sicurezza.
Se viene raggiunta la soglia di arresto, l’unità si
spegne. Se l’unità è collegata all’alimentazione
esterna, compare una schermata di allarme el’allarme acustico si protrae per 2 minuti.
Test con la batteria a temperatura ambiente. Se
il problema persiste, contattare il fornitore.

2

Malfunzionamento
sistema, altatemperatura

1 bip ogni
20 s

L’unità entra in allarme quando la temperaturainterna si sta avvicinando alla soglia di arresto.
Se viene raggiunta la soglia di arresto, l’unitàsi spegne, compare una schermata di allarmee l’allarme acustico si protrae per 2 minuti.
Accertarsi che le ventole dell’unità non siano
ostruite. Se il problema persiste, contattare il
fornitore.

3

Malfunzionamento
sistema, alta pressione

1 bip ogni
20 s

Se viene raggiunta la soglia di arresto, l’unitàsi spegne, compare una schermata di allarmee l’allarme acustico si protrae per 2 minuti.
Se il problema persiste, contattare il fornitore.

4

Nessuna respirazione
rilevata

1 bip ogni
30 s

Nessuna respirazione rilevata per 15 s.
Si attiva dopo la prima respirazione o quando
la schermata del riscaldamento è spenta, aseconda della condizione che si verifichi per
prima.
Controllare che la cannula non presentipieghe o perdite. Aumentare l’impostazione disensibilità. Se il problema persiste, contattare
il fornitore.

5

Arresto batteria scarica

1 bip ogni
30 sL’allarme suona finché la batteria non si
scarica e l’unità si spegne.
Caricare la batteria.

6

Riscaldamento

nessunoL’unità inizia l’auto-dosaggio fino alrilevamento di una respirazione o fino alla finedel riscaldamento, a seconda della condizione
che si verifichi per prima.

PN MN232-C4 K | Manuale dell’utente

ITL - 143

FreeStyle Comfort
N.
priorità

Descrizione allarme

7

Display

Segnaleacustico

Ulteriori dettagli

Concentrazione ridotta

1 bipogni 30 s,
per unmassimodi 5 minSe il problema persiste, contattare il fornitore.

8

Malfunzionamento
sistema, portata ridotta(valvola OCD guasta,
ostruzione del percorsodel gas)1 bip ogni
30 s

Se il problema persiste, contattare il fornitore.

9

Avvertenza batteria
scarica

1 bip,
senza
ripetizione

Caricare la batteria.

10Frequenza respiratoria
superata

1 bip ogni
60 s

La frequenza respiratoria supera i 40 bpm.

11

Malfunzionamento
sistema, errore ventola

nessunoSe il problema persiste, contattare il fornitore.

12

Malfunzionamento
sistema, bassapressione

nessunoSe il problema persiste, contattare il fornitore.

13

Malfunzionamento
sistema, bassa
temperatura

nessunoLasciare riscaldare l’unità fino a temperatura
ambiente. Se il problema persiste, contattare
il fornitore.

Info

Avvio

1 bip info

All’accensione compare la schermata e si
dovrebbe sentire un bip che conferma ilfunzionamento del segnalatore acustico

Se il dosaggio viene innescato erroneamente,
diminuire l’impostazione di sensibilità. Se il
problema persiste, contattare il fornitore.

La schermata di avvio visualizza la versione
software attuale nell’angolo inferiore destro.

144 - ITLPN MN232-C4 K | Manuale dell’utenteFreeStyle Comfort
N.
priorità

Descrizione allarme

Display

Segnaleacustico

Info

Schermata difunzionamento normale
in modalità autoSAT(nessun’alimentazione
CC collegata)nessunoInfo

Schermata difunzionamento
normale con
autoSAT disabilitato(nessun’alimentazione
CC collegata)

NessunoInfo

Impostazione
modificata

1 bip info

Info

Collegare
all’alimentazione
esterna nella
condizione spenta

nessunoInfo

Collegare
all’alimentazione
esterna nella
condizione difunzionamento

nessunoInfo

Scollegare
l’alimentazione esterna
nella condizione spenta

1 bip info

Info

Scollegare
l’alimentazione esterna
nella condizione difunzionamento (conbatteria inserita)

1 bip info

Ulteriori dettagliCompare la schermata CAIRE per 2 secondi,
seguita dallo stato Batteria e alimentazione per
5 s prima che lo schermo si spenga.

Compare la schermata CAIRE per 2 secondi,
quindi spegnere il display.

PN MN232-C4 K | Manuale dell’utente

ITL - 145

FreeStyle Comfort
N.
priorità

Descrizione allarmeInfo

Display

Segnaleacustico

Ulteriori dettagli

Nella modalità distandby: quando viene
premuto il pulsante suo giù, il display mostra
lo stato della batteria e
di carica

nessunoCompare la schermata per 5 s, quindi si
spegne.

Info

Modalità aereo con
funzionamento abatteria

1 bip
lungo

Info

Modalità aereo con
alimentazione di rete

1 bip
lungo

146 - ITLPN MN232-C4 K | Manuale dell’utenteFreeStyle Comfort
Istruzioni per la navigazione del menu
Si può accedere al Menu in modalità Standby
o Funzionamento. In modalità Funzionamento,
se si verifica un allarme si uscirà automaticamente
dal menu. Se non vengono premuti pulsanti per
20 secondi, si uscirà automaticamente dal menu.
In modalità Standby potrebbe essere necessariopremere il pulsante Menu due volte per riattivare
l’unità, quindi sarà possibile entrare nel Menu.
Sono presenti quattro schermate del menu:
1. Schermata di stato: Mostra il codice di rev. delfirmware, il tempo di funzionamento e i cicli diconsumo della batteria

Modalità aereo

2. Modalità aereo (ON/OFF)
3. Sensibilità: regolabile (1-5)
4. autoSAT (ON/OFF)

Sensibilità

Esempio di schermata di stato

FUNZIONAMENTO DEL MENUIMPOSTAZIONI

autoSATSi accede al menu impostazioni premendo il pulsanteMenu/Seleziona
o Standby.

nelle modalità Funzionamento

Per scorrere fra i menu, premere.

Pagina di stato

PN MN232-C4 K | Manuale dell’utente

ITL - 147

FreeStyle Comfort
PAGINA DI STATO

è visibile l’icona ANNULLA.

La pagina di stato del Freestyle Comfort mostra
la revisione del firmware, le ore di funzionamento
e il numero di cicli della batteria sulla batteria
collegata.

Confermata attivazione della Modalità aereo

Per disattivare la Modalità aereo, ripetere i passi
illustrati sopra portando il selettore su OFF.
Pagina di stato

Le ore di funzionamento sono mostrate in ore
intere.
La scarica della batteria è il numero di cicli in cui labatteria collegata è stata scaricata e ricaricata.

MODALITÀ AEREO
Il sistema FreeStyle Comfort viene fornito in modalitàaereo disattivata.

Premerementre la schermata mostra il simbolo“Avanti” farà avanzare la pagina del menu alla
schermata di opzioni successiva.
La Modalità aereo è prevista per l’utilizzo in modalitàdi funzionamento. La Modalità aereo verrà disattivata
automaticamente se l’unità viene lasciata in modalitàStandby o con la sola alimentazione a batteria.
Nota: Consultare la sezione “Attivare la Modalità aereo”
per maggiori informazioni sulla modalità aereo.

MODALITÀ DI SENSIBILITÀ
Il sistema FreeStyle Comfort viene fornito con
un’impostazione di fabbrica della modalità disensibilità con valore 2, e in condizioni normali
non dovrebbe richiedere alcuna regolazione.
L’impostazione di sensibilità può essere regolata fra
i valori 1-5, dove 1 è il valore di sensibilità più basso,e 5 il più alto.
Schermata principale della modalità aereo

Per attivare la modalità, premere
quando l’icona
MODIFICA è visibile. Il valore si attiverà mostrando
la parola “ON” in blu.

Schermata della modalità di sensibilità con impostazione valore 2

Per regolare la modalità di sensibilità, premere
mentre è visibile l’icona MODIFICA. Il valore passeràal numero in blu successivo.
Selezionata l’attivazione della Modalità aereo

Per confermare l’attivazione della modalità aereo,
premerementre l’icona SALVA è visibile.
Il valore si attiverà mostrando “ON” in nero.
Per annullare la modalità aereo, premere
148 - ITLPN MN232-C4 K | Manuale dell’utente

mentreFreeStyle Comfort
MODALITÀ AUTOSAT
Il sistema FreeStyle Comfort viene fornito con lamodalità autoSAT attivata.

Selezionata modalità di sensibilità sul valore 3

Per confermare l’impostazione 3 in modalità disensibilità, premere
mentre è visibile l’icona
SALVA. Il valore cambierà mostrando un “3” in nero.
Per annullare la modifica della modalità di sensibilità,
premerementre l’icona ANNULLA è visibile.

Schermata principale della modalità autoSAT

Per disattivare la modalità, premere
quandol’icona MODIFICA è visibile. Il valore si attiverà
mostrando la parola “OFF” in blu.

Confermata la modalità di sensibilità sul valore 3
Selezionata attivazione della modalità autoSAT

Per confermare la disattivazione della modalità
autoSAT, premere
mentre l’icona SALVA è
visibile. Il valore si attiverà mostrando “OFF” in nero.
Per annullare la modalità autoSAT, premere
mentre è visibile l’icona ANNULLA.

Confermata la modalità di sensibilità sul valore 5 (massimo)

Premerementre la schermata mostra il simbolo“Avanti” farà avanzare la pagina del menu alla
schermata di opzioni successiva.

Confermata disattivazione della modalità autoSAT

Per attivare la modalità autoSAT, ripetere i passi
illustrati sopra portando il selettore su ON.
Premerementre la schermata mostra il simbolo“Avanti” farà avanzare la pagina del menu alla
schermata di opzioni successiva.

PN MN232-C4 K | Manuale dell’utente

ITL - 149

FreeStyle Comfort

Risoluzione dei problemi
Problema

Probabile causa

Soluzione

FreeStyle Comfort visualizza
un allarme e produce un bip
intermittente.

Fare riferimento alla tabella
Condizioni di allarme.

Fare riferimento alla tabella Condizioni di allarme.

Il concentratore di ossigeno
non si accende.

La batteria è scarica.

Alimentare l’unità mediante una presa CC o CA.

Non è collegato
all’alimentazione esterna.

Assicurarsi che i collegamenti esterni siano saldi.

Malfunzionamento generale.

Eseguire un ciclo di accensione e spegnimento
completo rimuovendo la batteria e scollegando
dalla fonte di alimentazione esterna CA o CC.
Quindi reinserire la batteria; il ciclo di accensione e
spegnimento è completo. Ricollegare alla fonte dialimentazione esterna, se necessario.
Contattare il fornitore dell’apparecchiatura epassare a un’altra fonte di ossigeno secondo
necessità.

Tutti gli altri problemi.

Eseguire un ciclo di accensione e spegnimento
completo rimuovendo la batteria e scollegando
dalla fonte di alimentazione esterna CA o CC.
Quindi reinserire la batteria; il ciclo di accensione e
spegnimento è completo. Ricollegare alla fonte dialimentazione esterna, se necessario.
Passare a un’altra fonte di ossigeno disponibile e
contattare il fornitore dell’apparecchio.

Nota: alcuni allarmi causano lo spegnimento dell’unità.
Nota: durante un’interruzione di corrente generale, il sistema di allarme si azzera (cancella qualsiasi allarme precedente) evisualizzerà un allarme solo se la condizione sussiste dopo il ripristino dell’alimentazione.

150 - ITLPN MN232-C4 K | Manuale dell’utenteFreeStyle Comfort
Informazioni di viaggio
Il sistema FreeStyle Comfort è conforme alle linee
guida FAA per l’utilizzo su aeromobili commerciali.
Prima del viaggio (aereo, in treno, crociera ecc.) sarànecessario:• Assicurarsi che il concentratore sia in buone
condizioni di funzionamento.
• Contattare la compagnia aerea almeno 48 ore primadella partenza per informarla che si intende viaggiarecon FreeStyle Comfort.
Si consiglia di portare sempre con sé batterie
aggiuntive durante i viaggi. Per i viaggi in aereo, farein modo di avere sufficiente batteria considerando il
tempo a terra (check-in pre-volo, controlli di sicurezzaecc.) e le soste. La maggior parte delle compagnie
aeree richiede di disporre di un numero adeguato di
gruppi batterie completamente carichi per alimentare
il dispositivo per almeno il 150% della durata massima
del viaggio prevista (ore di volo, tempo complessivo
a terra prima e dopo il volo e durante i collegamenti,
nonché per ritardi imprevisti).
Nota: per i voli si consiglia di trasportare le batterie diriserva in una borsa per il trasporto sicura.

NORMATIVA FAA

1. Accendere il COP usando SOLO l’alimentazione abatteria, senza collegarlo ad alimentazioni esterne.

2. Abilitare la modalità aereo utilizzando uno dei
seguenti metodi:a. Seguire le istruzioni per attivare la Modalità aereo
nella sezione Istruzioni menu di navigazione.

b. Premere e mantenere entrambi i pulsanti Sue Giù per 5 secondi finché non si sente un “bip”
prolungato, e si vede apparire il simbolo dell’aereo.

Per evitare cortocircuiti, le batterie di riserva devono
essere protette una a una inserendole nella confezione
originale o isolando in altro modo i terminali,
ad esempio applicando del nastro adesivo sui terminali
esposti o inserendo ogni batteria in un sacchetto diplastica o in una custodia protettiva; inoltre, le batterie
devono essere trasportate esclusivamente nel bagaglio
a mano.

ATTIVARE LA MODALITÀ AEREO
Alimentazione a batteria
Il COP può essere utilizzato su un aeromobile secondo
le procedure normali SOLO con alimentazione abatteria, quando non è collegato ad alcuna fonte dialimentazione esterna
Alimentazione esterna
Sugli aeromobili, se è disponibile e la sua posizione
è idonea all’uso come fonte di alimentazione COP,
è possibile utilizzare una presa elettrica funzionante
come supporto per le batterie COP. In tal caso,
è necessario seguire questa procedura consigliata per ilpassaggio da alimentazione a batteria ad alimentazione
elettrica dell’aeromobile:

3. Quando compare il simbolo dell’aereo, collegare
il COP.

Nota: a causa delle limitazioni sugli aeromobili, le batterie
non si caricano nella modalità aereo.

PN MN232-C4 K | Manuale dell’utente

ITL - 151

FreeStyle Comfort

Accessori

1. Inserire l’unità sul fondo di una borsa, allineando
correttamente le aperture di batteria e filtri.

Per prestazioni e sicurezza corrette, utilizzare
esclusivamente gli accessori in elenco forniti da
CAIRE mediante il fornitore di ossigeno. L’utilizzo
di accessori non presenti nel seguente elenco può
influenzare negativamente sulle prestazioni e/o sulla
sicurezza del concentratore.
Accessori standard del sistema FreeStyle Comfort
Borsa per il trasporto

MI396-1

Zaino

MI459-1

Tracolla

MI397-1

Alimentatore CA (tutti i cavi inclusi)

PW036-1S

Cavo di alimentazione CC

CD041-1

Filtro a particelle grosse esterno(destro)FI226-1

Filtro a particelle grosse esterno(sinistro)

FI227-1

2. Chiudere la cerniera nella parte inferiore della borsa.

Accessori opzionali disponibili
Gruppo batterie a 8 celle aggiuntivo

BT034-1S

Gruppo batterie a 16 celle

BT035-1S

Caricabatterie desktop del gruppo
batterie

BT036-1

Borsa per il trasporto degli accessori

MI372-2

Nota: potrebbero essere disponibili opzioni aggiuntive percavi specifici del Paese dove indicato. Contattare CAIRE
o il fornitore di ossigeno se è necessario ordinare opzioni
alternative.

Istruzioni della borsa per il trasporto
Per inserire e fissare bene l’unità FreeStyle Comfort
nella borsa per il trasporto, utilizzare le seguenti
istruzioni.

INSERIMENTO DELL’UNITÀ IN UNA BORSAPER IL TRASPORTO
152 - ITLPN MN232-C4 K | Manuale dell’utente

INSTALLAZIONE DELLA TRACOLLA
1. La tracolla può essere fissata sia direttamente
all’unità sia alla borsa per il trasporto.
2. Attaccare la tracolla all’anello sull’unità o sulla
borsa.

FreeStyle Comfort
Smaltimento e riutilizzo
Restituire sempre FreeStyle Comfort, inclusi tutti icomponenti, al fornitore di attrezzatura per il corretto
smaltimento. È inoltre possibile rivolgersi agli uffici
locali per istruzioni sul corretto smaltimento dellabatteria. L’unità può essere riutilizzata tra le persone
ma deve essere restituita al fornitore di attrezzatura peril ritrattamento.

elettriche ed elettroniche (RAEE).
I nostri prodotti sono conformi alle restrizioni dettate
dalla Direttiva sulle sostanze pericolose (RoHS). È
possibile che essi contengano tracce, ma non quantità
rilevanti, di piombo o altri materiali pericolosi.
ATTENZIONE: per il corretto smaltimento,
rivolgersi al fornitore dell’apparecchioo all’ente governativo preposto per leistruzioni di smaltimento

RAEE e RoHS
Questo simbolo ha lo scopo di ricordare ai
proprietari dell’apparecchio di consegnarlo presso uncentro per il riciclaggio al termine della durata utile, inconformità alla Direttiva sui rifiuti di apparecchiature

Materiali a contatto diretto con l’operatore
Telaio del concentratore............................................................................................................................... PC/ABS
Pannello di controllo del concentratore....................................................................Poliestere EBG7 o equivalente
Maniglia del concentratore...................................................................................................................Policarbonato
Uscita dell’ossigeno, ugello.................................................................................................................Policarbonato
Etichetta dell’unità...........................................................................................................................................Lexan
Connettori dei cavi....................................................................................................Policarbonato/cloruro di vinile
Cavo di alimentazione.......................................................................................................................... PVC, metallo
Alimentatore................................................................................................................... .Lexan 940 (policarbonato)
Gruppo batterie............................................................................................................................................ PC/ABS
Etichette di gruppo batterie e alimentatore....................................................................................Film di poliestere
Custodia per il trasporto del concentratore........................................................Cordura, neoprene, fodera in nylon
Tracolla...........................................................................................Polipropilene, acetile, vinile, cordura, neoprene

Test CEM
Linee guida e dichiarazione del produttore - Emissioni elettromagnetiche
Il sistema FreeStyle Comfort è destinato all’uso nell’ambiente elettromagnetico specificato di seguito. Il cliente ol’utilizzatore del sistema FreeStyle Comfort deve garantirne l’uso all’interno di un ambiente di questo tipo.
Test delle emissioni
Emissioni RF
CISPR 11
Emissioni RF
CISPR 11
Emissioni armoniche
IEC 61000-3-2
Fluttuazioni di tensione/
emissioni di sfarfallii
IEC 61000-3-3

Conformità

Ambiente elettromagnetico - Linee guida

Gruppo 1

Il sistema FreeStyle Comfort impiega energia a radiofrequenza solo
per il funzionamento interno. Pertanto, le emissioni RF sono molto
basse ed è improbabile che possano causare interferenze negli
apparecchi elettronici nelle vicinanze.

Classe B
Classe A
Conforme

Il sistema FreeStyle Comfort è idoneo all’uso in tutti gli edifici, compresi
quelli ad uso abitativo e quelli collegati direttamente alla rete pubblica
di alimentazione a bassa tensione utilizzata per rifornire gli edifici ad
uso abitativo.

PN MN232-C4 K | Manuale dell’utente

ITL - 153

FreeStyle Comfort
Linee guida e dichiarazione del produttore - Immunità elettromagnetica

Il sistema FreeStyle Comfort è destinato all’uso nell’ambiente elettromagnetico specificato di seguito. Il cliente o l’utilizzatore delsistema FreeStyle Comfort deve garantirne l’uso all’interno di un ambiente di questo tipo.
Test dell’immunità Livello di test IEC 60601
Livello di conformità
Ambiente elettromagnetico Linee guida
Il pavimento deve essere di legno,
Ambiente
cemento o piastrelle in ceramica. Se±8 kV per contatto±8 kV per contatto
elettromagnetico i pavimenti sono ricoperti di materiale
Linee guida±15 kV per aria
±15 kV per aria
sintetico, l’umidità relativa deve essere
IEC 61000-4-2
almeno del 30%.
±2 kV per linee di±2 kV per linee diLa qualità dell’alimentazione di rete
alimentazione
Transitori elettrici
alimentazione
deve corrispondere a quella di una tipica
veloci/burst
struttura sanitaria professionale e degli
IEC 61000-4-4±1 kV per linee di ingresso/
ambienti sanitari domestici.
N/D
uscita±1 kV da linea a linea
Sovratensionetransitoria
IEC 61000-4-5

Cali di tensione,
brevi interruzioni
e variazionidella tensione dialimentazione sulle
linee di ingresso
dell’alimentazione
IEC 61000-4-11Campo magneticoalla frequenza direte (50/60 Hz)
IEC 61000-4-8±2 kV modalità comune
sulle linee CA±1 kV da linea a linea±1 kV differenziale sulle
linee CALa qualità dell’alimentazione di retedeve corrispondere a quella di una tipica
struttura sanitaria professionale e degli
ambienti sanitari domestici.

±2 kV modalità comune sulle
linee I/O esterne
0% UT per 0,5 cicli (0°, 45°, 0% UT per 0,5 cicli (0°, 45°,90°, 135°, 180°, 225°, 270°, 90°, 135°, 180°, 225°, 270°,
La qualità dell’alimentazione di rete
315°)
315°)
deve corrispondere a quella di una tipica
struttura sanitaria professionale e degli
ambienti sanitari domestici. Se l’utilizzatore
0% UT per 1 ciclo (00)
0% UT per 1 ciclo (00)di FreeStyle Comfort richiede continuità
di funzionamento durante le interruzioni
70% UT (30% di calo in UT) 70% UT (30% di calo in UT) dell’alimentazione elettrica, è consigliabile00
per 25/30 cicli (0 )per 25/30 cicli (0 )alimentare FreeStyle Comfort con un
gruppo di continuità o una batteria.
0% UT per 250/300 cicli (00)

0% UT per 250/300 cicli (00)

3 A/m

3 A/m

NOTA: UT indica la tensione di rete CA prima dell’applicazione del livello di test.

I campi magnetici alla frequenza dialimentazione devono essere ai livelli diun luogo tipico in una struttura sanitaria
professionale e degli ambienti sanitari
domestici.

NOTA: in alcuni casi, il sistema Freestyle Comfort può azzerarsi dopo una scarica elettrostatica importante e potrebbe richiedere
un ciclo di accensione e spegnimento completo per riprendere il normale funzionamento. Per ulteriori dettagli vedere la sezione
Risoluzione dei problemi.

154 - ITLPN MN232-C4 K | Manuale dell’utenteFreeStyle Comfort
Linee guida e dichiarazione del produttore - Immunità elettromagnetica

Il sistema FreeStyle Comfort è destinato all’uso nell’ambiente elettromagnetico specificato di seguito. Il cliente ol’utilizzatore del sistema FreeStyle Comfort deve garantirne l’uso all’interno di un ambiente di questo tipo.
Test
Livello di test IEC 60601 Livello di conformità
Ambiente elettromagnetico - Linee guidadell’immunità
RF condotta
10 Vrms
10 Vrms
Gli apparecchi di comunicazione a RF
mobili e portatili devono essere utilizzati
IEC 61000-4-6
Da 150 kHz a 80 MHz
a una distanza da qualsiasi componente
di FreeStyle Comfort, compresi i cavi,
superiore a quella calcolata in baseall’equazione applicabile alla frequenza delRF irradiata
10 V/m
10 V/mtrasmettitore.
IEC 61000-4-3
Da 80 MHz a 2700 MHz
Distanza di separazione consigliata

d = 1,2√ P

d = 1,2√ P = Da 80 MHz a 800 MHz
d = 1,2√ P = Da 800 MHz a 2700 MHz
dove P è la potenza di uscita nominalemassima del trasmettitore in watt (W)secondo le indicazioni del produttore e d èla distanza di separazione consigliata in
metri (m).
Le intensità di campo dai trasmettitori RF
fissi, determinate mediante un’indagine
elettromagnetica nella sede,a devono
essere inferiori al livello di conformità in ogni
intervallo di frequenza.b
Le interferenze possono verificarsi nei pressidelle apparecchiature contrassegnate daquesto simbolo:

NOTA 1: a 80 MHz e 800 MHz, si applica l’intervallo di frequenza superiore.
NOTA 2: queste linee guida potrebbero non essere applicabili in tutte le situazioni. La propagazione elettromagnetica è
influenzata dall’assorbimento e dalla riflessione di strutture, oggetti e persone.
a
Le intensità di campo da trasmettitori fissi, quali stazioni base per telefoni via radio (cellulari e cordless) e radio mobili
terrestri, radio amatoriali, trasmissioni radiofoniche AM/FM e trasmissioni televisive, non possono essere previste con
accuratezza a livello teorico. Per valutare l’ambiente elettromagnetico prodotto da trasmettitori RF fissi, è opportuno
eseguire un’indagine elettromagnetica della sede. Se l’intensità di campo misurata nel punto in cui è utilizzato FreeStyle
Comfort supera il livello applicabile di conformità RF sopra indicato, è opportuno appurare che FreeStyle Comfort funzioni
correttamente. Qualora fosse riscontrato un funzionamento fuori dalla norma, potrebbe essere necessario adottare ulteriori
provvedimenti, ad esempio cambiando l’orientamento o la posizione di FreeStyle Comfort.
b

Oltre l’intervallo di frequenza da 150 kHz a 80 MHz, le intensità di campo devono essere inferiori a 10 V/m.

PN MN232-C4 K | Manuale dell’utente

ITL - 155

FreeStyle Comfort
Distanze di separazione consigliate tra apparecchi per la comunicazione a RF portatili emobili e FreeStyle Comfort

FreeStyle Comfort è destinato all’uso in un ambiente elettromagnetico in cui i disturbi da RF irradiata sono tenuti sotto controllo.
Il cliente o l’utilizzatore di FreeStyle Comfort può prevenire le interferenze elettromagnetiche mantenendo una distanza minima
tra le apparecchiature di comunicazione RF portatili e mobili (trasmettitori) e FreeStyle Comfort come indicato di seguito,
calcolando tale distanza in base alla potenza massima di uscita dell’apparecchiatura di comunicazione.
Potenza di uscita
Distanza di separazione in base alla frequenza del trasmettitore
nominale massima delm
trasmettitore
Da 150 kHz a 80 MHz
Da 80 MHz a 800 MHz
Da 800 MHz a 2,5 GHz

d = 1,2√ P

d = 1,2√ P

d = 1,2√ P

W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Per i trasmettitori con una potenza di uscita massima non elencata sopra, la distanza di separazione consigliata d in metri(m) può essere stimata utilizzando l’equazione applicabile alla frequenza del trasmettitore, dove P è la potenza di uscita
massima del trasmettitore in watt (W) ottenuta in base alle informazioni fornite dal produttore.
NOTA 1: a 80 MHz e 800 MHz, si applica la distanza di separazione per l’intervallo di frequenza superiore.
NOTA 2: queste linee guida potrebbero non essere applicabili in tutte le situazioni. La propagazione elettromagnetica è
influenzata dall’assorbimento e dalla riflessione di strutture, oggetti e persone.

Classificazione

	• CISPR 11 / EN 55011 Classe B Gruppo 1,“Apparecchi industriali, scientifici e medici (ISM)”
	• FCC Parte 15, Sottoparte B – Radiatori non
intenzionali di Classe B

Grado di protezione dalle scosse elettriche:
Tipo BF	Apparecchio che fornisce un particolare
grado di protezione dalle scosse
elettriche per quanto riguarda
		 1) corrente di dispersione ammissibile;		2) affidabilità del collegamento di messa a terra
protettivo (se presente).
		 Non utilizzare per applicazione cardiaca diretta.

IP22 - Protezione contro l’ingresso di dita o oggetti
simili di diametro superiore a 12,5 mm. L’acquaa caduta verticale non avrà effetto nocivo sel’involucro è inclinato con un’angolazione di 15°
dalla posizione normale.

Tipo di protezione dalle scosse elettriche:
Classe II	Protezione dalle scosse elettriche grazie
al doppio isolamento.

Test indipendente per apparecchio elettromedicale
standard:
Testato per la compatibilità alle norme• IEC 60601-1 edizione 3.1: Apparecchia
elettromedicali - Parte 1 Requisiti generali disicurezza.
• CAN/CSA C22.2 N. 60601-1-8 edizione 2.1
Apparecchi elettromedicali – Parte 1: Requisiti
generali di sicurezza
Protezione contro i possibili campi elettromagnetici oaltre interferenze tra l’apparecchio e altri dispositivi.
	• Testato per la compatibilità a EN 60601-1-2/
IEC 60601-1-2: 4a edizione.
	• Testato per la compatibilità a RTCA/DO160
Sezione 21 Categoria M.

156 - ITLPN MN232-C4 K | Manuale dell’utente

Metodo di pulizia e controllo delle infezioni consentiti:
consultare la sezione "Pulizia, cura e manutenzione
corretta" di questo Manuale dell’utente di FreeStyle.
Grado di sicurezza dell’applicazione in presenza di gas
anestetici infiammabili:
apparecchio non adatto per questo tipo diapplicazione.
Modalità operativa: funzionamento continuo.

CAIRE e CAIRE Inc. sono marchi registrati di CAIRE Inc. Visitare il nostro sito Web diseguito per un elenco completo dei marchi.
Marchi: www.careinc.com/corporate/trademarks.
Copyright © 2021 CAIRE Inc. CAIRE Inc. si riserva il diritto di interrompere la produzione
o di modificare prezzi, materiali, apparecchiature, qualità, descrizioni, specifiche e/oprocessi dei propri prodotti in qualsiasi momento, senza preavviso e senza ulteriori obblighio conseguenze. Tutti i diritti non espressamente indicati nel presente documento sono
riservati, laddove applicabili.

FreeStyle Comfort
Controlos do utilizador e Indicadores de estado do sistema
ISO 7000; Símbolos gráficos para utilização noequipamento—Índice e resumo
Ler o manual do utilizador antes deutilizar. Reg. # 1641
Faixa de temperatura dearmazenamento ou operação.
Reg. # 0632
Faixa de umidade de armazenamento.
Reg. # 2620

Diretiva do Conselho 93/42/EEC; respeitante a
dispositivos médicos
Representante autorizado na União
Europeia
Se a etiqueta UDI (identificador
exclusivo de dispositivo) do produto
contiver o símbolo CE ####, odispositivo cumprirá os requisitos
da Diretiva 93/42 / CEE relativa adispositivos médicos. O símbolo CE#### indica o número do organismo
notificado.
IEC 60417

Limites de pressão atmosférica.
Reg. # 2621

Equipamento de Classe II, com
Isolamento Duplo Reg. # 5172

Nome e morada do fabricante.
Reg. # 3082

Interruptor de alimentação.
Reg. # 5009

Data de fabrico. Reg. # 2497

Certificação ETLAgência de segurança da CAN/CSA
C22.2 Nº 601.1 M90 respeitante aequipamentos médicos elétricos

Número de catálogo. Reg. # 2493
Número de série. Reg. # 2498
Este lado para cima. Reg. # 0623

Símbolos internos
Manter afastado de materiais
inflamáveis, óleo e gorduras.

Frágil, manusear com cuidado.
Reg. # 0621
Manter afastado da chuva, manter seco.
Reg. # 0626

Não desmontar.

Limite de empilhamento por número.
Reg. # 2403

Aumentar o fluxo.

ISO 7010: Símbolos gráficos—Cores e sinalização desegurança—Sinalização de segurança registada

Reduzir o fluxo.

O manual de instruções tem de ser lido.
Reg. n.º M002
Manter afastado de chamas e faíscas.
É proibido fumar e são proibidas fontes
de ignição abertas. Reg. n.º P003
Não fume perto da unidade ou quando
a estiver a utilizar. Reg. n.º P002Botão de menu.
Em conformidade
com RTCA/
DO-160Secção 21
Categoria M

Em conformidade com RTCA DO160
Secção 21 Categoria M. Requisitos da
FAA SFAR 106

Equipamento aplicado tipo BF (graude proteção contra choques elétricos).
Reg. # 5333

Equipamento de Classe II, com
Isolamento Duplo

Aviso. Reg. n.º W001

Aprovado pela FAA – POC

PN MN232-C4 K | Manual do utilizador

POR - 157

FreeStyle Comfort
21 CFR 801.15: Código dos regulamentos federais
Título 21
A lei federal restringe a venda deste
dispositivo a ou por ordem de um médico.
Diretiva do conselho 2012/19/EU: resíduos deequipamentos elétricos e eletrónicos (REEE)WEEE

IEC 60601-1: Equipamento médico elétrico
Parte 1 Requisitos gerais de segurança básica e
desempenho essencial
À prova de gotas
Este produto pode estar abrangido por uma ou mais
patentes dos EUA e internacionais. Visite o nosso
website abaixo para obter uma lista das patentes
aplicáveis. Pat.: www.caireinc.com/corporate/patents/.

Descrição do Concentrador deOxigénio

adequadamente. O seu médico prescreveu um fluxo ou
configuração para atender à sua condição respiratória
específica.

O ar que respiramos contém cerca de 21% de oxigénio,78% de azoto, e 1% de outros gases. No FreeStyle®
Comfort®, o ar ambiental é conduzido para o aparelho
através das entradas de ar. Em seguida, passa atravésde um material absorvente chamado peneira molecular.
Este material separa o oxigénio do azoto e permite
apenas a passagem do oxigénio. O resultado é um fluxode oxigénio de elevada concentração administrado ao
utilizador.

Embora o oxigénio não cause dependência, as terapias
com oxigénio não autorizadas podem ser perigosas.
Deve procurar aconselhamento médico antes de usar
este concentrador de oxigénio. O Fornecedor deequipamentos que lhe forneceu o equipamento deoxigénio demonstrará como definir a taxa de fluxo
prescrita.

O FreeStyle Comfort combina tecnologias avançadas
de concentração e conservação de oxigénio para
criar um dispositivo portátil que permite a máxima
portabilidade e mobilidade. A administração por fluxo
pulsátil avançada deteta rapidamente a inspiração doutilizador e administra oxigénio no início de cadainalação. A administração por fluxo pulsátil é ideal para
atividades e períodos passados no exterior. Este tipo deadministração conserva oxigénio e prolonga a vida útilda bateria, ao contrário do fluxo contínuo. O FreeStyle
Comfort é suficientemente leve para ser transportado
com uma alça de tiracolo. Pode ser alimentado por CA(parede), CC (veículo), ou bateria.

Uso pretendido
O concentrador de oxigênio CAIRE FreeStyle
Comfort destina-se à administração de oxigênio
suplementar. O dispositivo não se destina ao
suporte de vida nem fornece recursos de monitoramento do paciente.

Por que motivo o seu médico lhe
prescreveu oxigénio
Muitas pessoas sofrem de uma variedade de doenças
cardíacas, pulmonares e outras patologias respiratórias.
Um número significativo dessas pessoas pode
beneficiar de terapias de oxigénio suplementar em
casa, quando viajam, ou enquanto participam em
atividade diárias fora de casa.
O oxigénio é um gás que compõem 21% do arambiente que respiramos e os nossos corpos dependem
de um fornecimento estável para que funcionem
158 - PORPN MN232-C4 K | Manual do utilizador

Perfil do operador
Os concentradores destinam-se a fornecer oxigénio
suplementar a utilizadores que sofrem de desconforto
devido a patologias que afetam a capacidade dospulmões de transferir o oxigénio no ar para a corrente
sanguínea. Os concentradores de oxigénio portáteis
(POCs) não armazenam nem contêm oxigénio.
Não necessitam de reabastecimento e podem serrecarregados em qualquer lado onde exista alimentação
CA ou CC disponível (exceto numa aeronave). Isto
torna o utilizador relativamente autossuficiente em
termos de utilização doméstica, em ambulatório (nointerior e exterior da casa) e facilita a mobilidade eestilo de vida geral. A utilização do concentrador deoxigénio requer a prescrição de um médico e não sedestina ao suporte de vida.
Embora as terapias com oxigénio possam ser prescritas
a utilizadores de todas as idades, o utilizador típico
de terapias com oxigénio tem uma idade superiora 65 anos e sofre de uma variedade de doenças
respiratórias, incluindo a Doença Pulmonar Obstrutiva
Crónica (DPOC). Geralmente, os utilizadores têm
boas capacidades cognitivas e têm de ser capazes decomunicar o seu desconforto. Se o utilizador não forcapaz de comunicar o seu desconforto, ou não forcapaz de ler e compreender os rótulos e instruções
de utilização do concentrador, a utilização do mesmosó é recomendada sob a supervisão de alguém com
capacidade para tal. Em caso de desconforto ao
utilizar o concentrador, o utilizador deve contactar
o seu prestador de cuidados de saúde. Recomendase também aos utilizadores que disponham de uma
fonte de oxigénio de reserva (ou seja, um cilindro deoxigénio) para a eventualidade de uma falha de energia
ou falha do concentrador. A utilização do concentradornão requer outras aptidões ou capacidades especiais.

FreeStyle Comfort

Desembalar o FreeStyle Comfort
Verifique se todos oscomponentes listados estão
incluídos na embalagem. Se
faltar algum artigo, contacte o
seu fornecedor de oxigénio deimediato.

Acessórios padrão
• Concentrador de oxigénio portátil• Saco de transporte• Alça a tiracolo• Unidade da bateria de 8 células• Transformador de CA• Cabo de alimentação CA• Transformador de CC• Manual do utilizador

Acessórios opcionais• Unidade da bateria de 8 células
adicional• Unidade da bateria de 16 células• Carregador de bancada• Bolsa porta-tudo acessória

Conhecer o FreeStyle Comfort
Familiarize-se com as principais características do FreeStyle Comfort e da sua interface do utilizador.

Vista frontal
Fixação da alçaInterface doutilizador

Saída deOxigénio
Saída de ar

Admissão de ar/filtros
Admissão de ar/filtros
Patilha dorecetáculo
da porta dealimentação
Unidade da bateriaBotão de libertação
da unidade da bateria
PN MN232-C4 K | Manual do utilizador

POR - 159

FreeStyle Comfort
Vista superiorEcrã do visor
Aumentar o fluxoBotão de energiaBotão de menu

Reduzir o fluxoEcrã do visor alimentado por bateria
Configuração de Fluxo

Visualização dabateria

Indicador autoSAT

Visor de alarmes

Percentagem
da bateria
restante

Indicador deadministração
do boloEcrã do visor na alimentação CA/CC

Configuração de
Fluxo

Espaço do visor dealarmes

160 - PORPN MN232-C4 K | Manual do utilizador

Visualização dabateria

Indicador de carga

FreeStyle Comfort
Importante!
As instruções de segurança estão definidas
da seguinte forma:
AVISO: INFORMAÇÕES DE SEGURANÇA
IMPORTANTES RELATIVAS A PERIGOSQUE PODEM CAUSAR LESÕES GRAVES.
ATENÇÃO: Informações importantes paraevitar danos no FreeStyle Comfort.
Nota: Informações que necessitam de atenção especial.

Indicações de UtilizaçãoAVISO: É MUITO IMPORTANTESELECIONAR APENAS O NÍVEL DEOXIGÉNIO PRESCRITO. NÃO ALTERE
A SELEÇÃO DO FLUXO, EXCETO SE
TIVER RECEBIDO INDICAÇÕES NESSESENTIDO POR PARTE DE UM MÉDICOQUALIFICADO.
AVISO: A LEI FEDERAL (EUA) RESTRINGE A VENDAOU ALUGUER DESTE DISPOSITIVO À CLASSEMÉDICA OU A OUTROS PROFISSIONAIS DE SAÚDEQUALIFICADOS.
AVISO: ESTA UNIDADE NÃO DEVE SER USADA COMOSUPORTE DE VIDA. OS PACIENTES GERIÁTRICOS,PEDIÁTRICOS OU OUTROS UTILIZADORESINCAPAZES DE COMUNICAR DESCONFORTO
DURANTE A UTILIZAÇÃO DESTE DISPOSITIVOPODEM NECESSITAR DE MONITORIZAÇÃOADICIONAL. OS UTILIZADORES COM INSUFICIÊNCIA
AUDITIVA E/OU VISUAL PODEM NECESSITAR DEAJUDA COM OS ALARMES DE MONITORIZAÇÃO.
AVISO: AS CONFIGURAÇÕES DO MODO DE
DOSAGEM PULSÁTIL DEVEM SER DETERMINADAS
INDIVIDUALMENTE PARA CADA UTILIZADOR,DE ACORDO COM AS SUAS NECESSIDADES EMREPOUSO, DURANTE O EXERCÍCIO E DURANTEVIAGENS.
AVISO: ALGUNS ESFORÇOS RESPIRATÓRIOS DO
DOENTE PODEM NÃO ACIONAR A FUNÇÃO DE
CONSERVAÇÃO.
AVISO: O FREESTYLE COMFORT NÃO SE DESTINA AUTILIZAÇÃO NUM DOENTE TRAQUEOSTOMIZADO.
AVISO: AS MULHERES GRÁVIDAS OU A AMAMENTAR
NÃO DEVEM UTILIZAR OS ACESSÓRIOS
RECOMENDADOS NESTE MANUAL, UMA VEZ QUECONTÊM FTALATOS.

Contraindicações de utilizaçãoAVISO: EM DETERMINADAS
CIRCUNSTÂNCIAS, A UTILIZAÇÃO DEOXIGÉNIO SEM PRESCRIÇÃO MÉDICA
PODE SER PERIGOSA. ESTE APARELHODEVE SER UTILIZADO APENAS QUANDOPRESCRITO POR UM MÉDICO.
AVISO: NÃO DEVE SER UTILIZADO NA PRESENÇA DE
ANESTÉSICOS INFLAMÁVEIS.
AVISO: TAL COMO ACONTECE COM QUALQUEROUTRO APARELHO ALIMENTADO A ENERGIAELÉTRICA, O UTILIZADOR PODE EXPERIENCIAR
PERÍODOS DE NÃO FUNCIONAMENTO COMORESULTADO DE INTERRUPÇÕES DA ENERGIAELÉTRICA, OU NECESSITAR QUE O CONCENTRADORDE OXIGÉNIO SEJA REPARADO POR UM TÉCNICOQUALIFICADO. O CONCENTRADOR DE OXIGÉNIONÃO É ADEQUADO PARA UTILIZADORES QUEPOSSAM SOFRER CONSEQUÊNCIAS ADVERSAS
PARA A SAÚDE COMO RESULTADO DE TAIS
INTERRUPÇÕES TEMPORÁRIAS.

Diretrizes de Segurança
AVISO: É PROIBIDA QUALQUER
MODIFICAÇÃO A ESTE EQUIPAMENTO.
AVISO: O DISPOSITIVO FORNECE OXIGÉNIO DE
ELEVADA CONCENTRAÇÃO QUE PROMOVE UMACOMBUSTÃO RÁPIDA. NÃO AUTORIZE QUE SE
FUME OU QUE SE FAÇAM CHAMAS DENTRO DAMESMA DIVISÃO (1) DESTE DISPOSITIVO OU (2) DEQUALQUER ACESSÓRIO PORTADOR DE OXIGÉNIO. O
NÃO CUMPRIMENTO DESTE AVISO PODE ORIGINAR
UM INCÊNDIO GRAVE, DANOS MATERIAIS E/OU
PROVOCAR LESÕES FÍSICAS OU MORTE.
AVISO: NÃO UTILIZE A UNIDADE NUM ESPAÇO
RESTRITO OU CONFINADO (OU SEJA, NUMACAIXA PEQUENA OU NUMA BOLSA) ONDE A
VENTILAÇÃO POSSA FICAR LIMITADA. ESTA AÇÃOPODE PROVOCAR O SOBREAQUECIMENTO DO
CONCENTRADOR DE OXIGÉNIO E PREJUDICAR OSEU DESEMPENHO.
AVISO: O CONCENTRADOR DEVE SER COLOCADONUM LOCAL ONDE SEJA EVITADA A ENTRADA DE
FUMO, POLUENTES OU VAPORES.
AVISO: NO CASO DE OCORRER UM INCIDENTESÉRIO COM ESTE DISPOSITIVO, O USUÁRIO
DEVE RELATAR IMEDIATAMENTE O INCIDENTEAO FORNECEDOR E / OU AO FABRICANTE. UM
INCIDENTE GRAVE É DEFINIDO COMO UMA LESÃO,MORTE OU POTENCIAL DE CAUSAR LESÃO /MORTE, CASO OCORRA UMA REINCIDÊNCIA. O
USUÁRIO TAMBÉM PODE RELATAR O INCIDENTE
À AUTORIDADE COMPETENTE DO PAÍS EM QUE OINCIDENTE OCORREU.

PN MN232-C4 K | Manual do utilizador

POR - 161

FreeStyle Comfort
AVISO: A UTILIZAÇÃO DE ALGUNSACESSÓRIOS DE ADMINISTRAÇÃODE OXIGÉNIO, NÃO ESPECIFICADOSPARA ESTE CONCENTRADOR, PODEPREJUDICAR O SEU DESEMPENHO. OS
ACESSÓRIOS RECOMENDADOS ESTÃO
MENCIONADOS NESTE MANUAL.
AVISO: SE O CONCENTRADOR DE OXIGÉNIO TIVER
SOFRIDO UMA QUEDA, ESTIVER DANIFICADO OU
TIVER SIDO EXPOSTO À ÁGUA, CONTACTE O SEUPRESTADOR DE CUIDADOS DOMICILIÁRIOS PARAAPPARATUUR UMA INSPEÇÃO OU EVENTUALREPARAÇÃO DO APARELHO. NÃO USE O
CONCENTRADOR DE OXIGÉNIO CASO ESTEAPRESENTE O CABO DE ALIMENTAÇÃO OU FICHADANIFICADOS.
AVISO: PRENDA DEVIDAMENTE O CONCENTRADORDE OXIGÉNIO QUANDO O TRANSPORTAR NUM
VEÍCULO, COM O CINTO OU DE OUTRO MODO, PARAPREVENIR DANOS OU FERIMENTOS.
AVISO: OS REGULAMENTOS DO DEPARTAMENTONORTE-AMERICANO DOS TRANSPORTES (DOT) E
DAS NAÇÕES UNIDAS (NU) EXIGEM A REMOÇÃODA BATERIA DO APARELHO EM TODAS AS
VIAGENS AÉREAS INTERNACIONAIS QUANDO OCONCENTRADOR DE OXIGÉNIO É REGISTADO
COMO BAGAGEM. PARA TRANSPORTAR OCONCENTRADOR DE OXIGÉNIO, A BATERIATAMBÉM TEM DE SER REMOVIDA DO DISPOSITIVO E
EMBALADA ADEQUADAMENTE.
AVISO: NÃO UTILIZE ESTE EQUIPAMENTO SEM
PRIMEIRO LER E COMPREENDER ESTE MANUAL.
CASO NÃO PERCEBA OS AVISOS E AS INSTRUÇÕES,CONTACTE O SEU FORNECEDOR DO EQUIPAMENTOANTES DE TENTAR USÁ-LO; CASO CONTRÁRIO,PODEM OCORRER LESÕES OU DANOS.
AVISO: SE SENTIR ALGUM DESCONFORTO OU SEENCONTRAR NUMA SITUAÇÃO DE EMERGÊNCIA
MÉDICA, PROCURE ASSISTÊNCIA MÉDICA DE
IMEDIATO.
AVISO: UTILIZAR O CONCENTRADOR DE OXIGÉNIOFORA DOS LIMITES DE TEMPERATURA DEFUNCIONAMENTO PODE LIMITAR A CAPACIDADE
DO CONCENTRADOR DE CUMPRIR COM A
ESPECIFICAÇÃO DA CONCENTRAÇÃO DE OXIGÉNIO.
CONSULTE A SECÇÃO RELATIVA A ESPECIFICAÇÕES
DESTE MANUAL PARA CONSULTAR OS LIMITES DETEMPERATURA.
AVISO: NÃO UTILIZE LUBRIFICANTES, GORDURAS,
PRODUTOS DERIVADOS DO PETRÓLEO OUINFLAMÁVEIS COM OS ACESSÓRIOS PORTADORESDE OXIGÉNIO, OU COM O CONCENTRADOR DEOXIGÉNIO. DEVEM SER USADAS APENAS LOÇÕESOU UNGUENTOS À BASE DE ÁGUA, COMPATÍVEISCOM OXIGÉNIO. O OXIGÉNIO ACELERA ACOMBUSTÃO DE SUBSTÂNCIAS INFLAMÁVEIS.

162 - PORPN MN232-C4 K | Manual do utilizador

AVISO: NÃO PERMITA QUE A ENTRADA E A SAÍDA DE
AR FIQUEM BLOQUEADAS. NÃO DEIXE CAIR NEM
INSIRA QUAISQUER OBJETOS NAS ABERTURAS
DO DISPOSITIVO. ESTA AÇÃO PODE PROVOCARO SOBREAQUECIMENTO DO CONCENTRADOR DEOXIGÉNIO E PREJUDICAR O SEU DESEMPENHO.
AVISO: A UTILIZAÇÃO INCORRETA DA BATERIA PODEPROVOCAR O AQUECIMENTO E INFLAMAÇÃO DA
MESMA E CAUSAR LESÕES GRAVES. CERTIFIQUESE DE NÃO PERFURAR, ATINGIR, PISAR OU
DEIXAR CAIR A BATERIA, OU DE A SUBMETER A
IMPACTOS OU CHOQUES FORTES. A UTILIZAÇÃO
DE UMA BATERIA DANIFICADA PODE PROVOCAR
FERIMENTOS.
AVISO: NÃO PROVOQUE UM CURTO-CIRCUITO NOSCONTACTOS DE METAL DA BATERIA COM OBJETOSMETÁLICOS, COMO CHAVES OU MOEDAS. ISTO
PODE RESULTAR EM FAÍSCAS OU NA GERAÇÃO DE
UM CALOR EXCESSIVO.
AVISO: DURANTE A UTILIZAÇÃO DO
CONCENTRADOR DE OXIGÉNIO PORTÁTIL NO
EXTERIOR COM A FONTE DE ALIMENTAÇÃO
CA, LIGUE A FONTE DE ALIMENTAÇÃO
EXCLUSIVAMENTE A UMA TOMADA PROTEGIDA COM
UM DISJUNTOR DIFERENCIAL RESIDUAL (DR).
AVISO: O CONCENTRADOR DE OXIGÉNIO NÃO DEVESER ENCOSTADO OU EMPILHADO COM OUTROSDISPOSITIVOS. SE NÃO FOR POSSÍVEL EVITARENCOSTAR OU EMPILHAR O DISPOSITIVO, ESTEDEVE SER SUPERVISIONADO A FIM DE GARANTIR OSEU NORMAL FUNCIONAMENTO.
AVISO: A UTILIZAÇÃO DE CABOS E ADAPTADORESQUE NÃO OS ESPECIFICADOS, À EXCEÇÃODOS CABOS E ADAPTADORES VENDIDOS PELOFABRICANTE DO EQUIPAMENTO ENQUANTO
PEÇAS DE SUBSTITUIÇÃO PARA COMPONENTESINTERNOS, PODE RESULTAR NO AUMENTO DASEMISSÕES OU NUMA DIMINUIÇÃO DA IMUNIDADE DO
CONCENTRADOR DE OXIGÉNIO.
AVISO: UTILIZE APENAS A TENSÃO ELÉTRICAESPECIFICADA NA ETIQUETA DE ESPECIFICAÇÕESAFIXADA NO DISPOSITIVO.
AVISO: NÃO UTILIZE EXTENSÕES COM ESTAUNIDADE OU LIGUE VÁRIAS FICHAS À MESMATOMADA ELÉTRICA. A UTILIZAÇÃO DE EXTENSÕES
PODE AFETAR NEGATIVAMENTE O DESEMPENHODO DISPOSITIVO. A EXISTÊNCIA DE DEMASIADASFICHAS NUMA TOMADA PODE RESULTAR NUMASOBRECARGA DO QUADRO ELÉTRICO, LEVANDOÀ ATIVAÇÃO DO DISJUNTOR/FUSÍVEL OU A UM
INCÊNDIO CASO O DISJUNTOR OU O FUSÍVEL
DEIXEM DE FUNCIONAR.

FreeStyle ComfortATENÇÃO: Na eventualidade de alarmeou se observar que o Concentradorde Oxigénio não está a trabalharadequadamente, consulte a secção
de Resolução de Problemas deste
manual. Se não conseguir resolver o
problema, contacte o seu Fornecedor do
Equipamento.

AVISO: OS DÍSTICOS “PROIBIDO FUMAR – OXIGÉNIOEM UTILIZAÇÃO” DEVEM ESTAR BEM VISÍVEIS NA
CASA OU NO LOCAL ONDE O OXIGÉNIO ESTÁ A
SER UTILIZADO. OS UTILIZADORES E OS SEUS
PRESTADORES DE CUIDADOS TÊM DE SERINFORMADOS ACERCA DOS PERIGOS DE FUMAR
NA PRESENÇA DE OU DURANTE A UTILIZAÇÃO DEOXIGÉNIO MEDICINAL.

ATENÇÃO: Para evitar a anulação da garantia, siga
todas as instruções dos fabricantes.

AVISO: NÃO UTILIZE O CONCENTRADOR DEOXIGÉNIO NA PRESENÇA DE GASES INFLAMÁVEIS,SOB RISCO DE UMA COMBUSTÃO RÁPIDA
PODER PROVOCAR DANOS MATERIAIS, LESÕESCORPORAIS OU MORTE.

AVISO: NÃO PROCEDA A QUALQUER
TIPO DE MANUTENÇÃO PARA ALÉM DASSOLUÇÕES POSSÍVEIS APRESENTADASNESTE MANUAL. NÃO RETIRE AS
TAMPAS. APENAS O SEU FORNECEDORDO EQUIPAMENTO OU UM TÉCNICO DE
MANUTENÇÃO QUALIFICADO PODERÃOREMOVER AS TAMPAS OU PROCEDER À
MANUTENÇÃO DA UNIDADE.
AVISO: UTILIZE APENAS ACESSÓRIOS
RECOMENDADOS PELO FABRICANTE. A UTILIZAÇÃODE OUTROS ACESSÓRIOS PODE SER PERIGOSA,CAUSAR GRAVES DANOS NO SEU CONCENTRADORDE OXIGÉNIO E INVALIDAR A GARANTIA.
AVISO: O FABRICANTE RECOMENDA UMA FONTE
ALTERNATIVA DE OXIGÉNIO SUPLEMENTAR NA
EVENTUALIDADE DE UMA FALHA DE ENERGIA,CONDIÇÃO DE ALARME OU AVARIA MECÂNICA.
CONSULTE O SEU MÉDICO OU FORNECEDOR DO
EQUIPAMENTO PARA INFORMAÇÕES SOBRE O TIPO
DE SISTEMA DE RESERVA REQUERIDO.
AVISO: NÃO PERMITA QUE A ENTRADA E A SAÍDA DE
AR FIQUEM BLOQUEADAS. NÃO DEIXE CAIR NEM
INSIRA QUAISQUER OBJETOS NAS ABERTURAS
DO DISPOSITIVO. ESTA AÇÃO PODE PROVOCARO SOBREAQUECIMENTO DO CONCENTRADOR DEOXIGÉNIO E PREJUDICAR O SEU DESEMPENHO.
AVISO: FUMAR DURANTE A UTILIZAÇÃO DEOXIGÉNIO É A PRIMEIRA CAUSA DE LESÕES
PROVOCADAS POR FOGO E MORTESRELACIONADAS. DEVERÁ CUMPRIR OS SEGUINTESAVISOS DE SEGURANÇA:AVISO: NÃO AUTORIZE QUE SE FUME, QUE
SE ACENDAM VELAS, OU SE FAÇAM CHAMASNA MESMA DIVISÃO ONDE SE ENCONTRA ODISPOSITIVO OU OS ACESSÓRIOS PORTADORES DEOXIGÉNIO.
AVISO: FUMAR DURANTE A UTILIZAÇÃO DE
UMA CÂNULA DE OXIGÉNIO PODE PROVOCARQUEIMADURAS FACIAIS E POSSIVELMENTE A
MORTE.
AVISO: CASO SEJA FUMADOR, OS 3 PASSOSSEGUINTES PODEM SALVAR-LHE A VIDA: DESLIGUEO CONCENTRADOR DE OXIGÉNIO, RETIRE ACÂNULA E SAIA DA DIVISÃO NA QUAL SE ENCONTRAO DISPOSITIVO.

AVISO: REMOVER A CÂNULA E COLOCÁ-LA NO
VESTUÁRIO, ROUPA DE CAMA, SOFÁS OU OUTRO
MATERIAL ALMOFADADO IRÁ DEFLAGRAR UMINCÊNDIO QUANDO EXPOSTOS A UM CIGARRO,FONTE DE CALOR OU CHAMA.
AVISO: NUNCA DEIXE A CÂNULA NASAL EM CIMA/
BAIXO DE PEÇAS DE ROUPA, COLCHAS OU
ALMOFADAS DE CADEIRAS. SE A UNIDADE ESTIVER
LIGADA MAS NÃO ESTIVER EM UTILIZAÇÃO, O
OXIGÉNIO TORNARÁ O MATERIAL INFLAMÁVEL.
CERTIFIQUE-SE DE QUE A ALIMENTAÇÃO ESTÁDESLIGADA (OFF) QUANDO O CONCENTRADOR DEOXIGÉNIO NÃO ESTIVER A SER UTILIZADO.
AVISO: O CONCENTRADOR DE OXIGÉNIODEVE ESTAR DESATIVADO E DESLIGADO DA
ALIMENTAÇÃO EXTERNA ANTES DE RETIRAR A
UNIDADE DA BATERIA.
AVISO: AS PEÇAS PEQUENAS PODEM
REPRESENTAR UM POTENCIAL PERIGODE ASFIXIA.
AVISO: EVITE O CONTACTO COM A ABERTURA DESAÍDA DO AR DURANTE MAIS DE 10 SEGUNDOS UMA
VEZ QUE PODE ESTAR QUENTE.
AVISO: O VENTO OU CORRENTES DE AR FORTESPODEM AFETAR ADVERSAMENTE A ADMINISTRAÇÃOPRECISA DA TERAPIA DE OXIGÉNIO.
AVISO: ESTE PRODUTO PODE EXPÔ-LO A
SUBSTÂNCIAS QUÍMICAS COMO O NÍQUEL, QUE
É CONSIDERADO UM AGENTE CANCERÍGENO
PELO ESTADO DA CALIFÓRNIA. PARA MAIS
INFORMAÇÕES, VISITE WWW.P65WARNINGS.
CA.GOV.
ATENÇÃO: Desligue sempre a fontede alimentação CA da parede antes dedesligar a fonte de alimentação CA doconcentrador de oxigénio.
ATENÇÃO: Não posicione o Concentrador de oxigénio
de um modo que dificulte o acesso ao cabo dealimentação.

PN MN232-C4 K | Manual do utilizador

POR - 163

FreeStyle ComfortATENÇÃO: Posicione sempre os tubos deadministração de oxigénio e os cabos dealimentação de forma a evitar o perigo de quedas ou
estrangulamentos acidentais.
ATENÇÃO: Quando usar o Concentrador de oxigénio
num automóvel, barco, ou noutras fontes de CC com
a fonte de alimentação CC, certifique-se de que oveículo é posto em funcionamento antes de ligar oConcentrador de oxigénio. O não seguimento destas
instruções pode resultar na falta de fornecimento deenergia ao Concentrador de oxigénio.
ATENÇÃO: Quando o automóvel onde está a usaro Concentrador de oxigénio for desligado, desligue
e remova o aparelho do automóvel. Não guarde oConcentrador de oxigénio num automóvel muito
quente ou frio ou noutros ambientes similares de altas
ou baixas temperaturas.
Nota: O equipamento de comunicação RF portátil ou
móvel pode afetar o equipamento elétrico médico.
ATENÇÃO: NÃO deixe o Concentrador deoxigénio ou o transformador ligados ao
veículo se a ignição estiver na posição de
DESLIGADO. Fazê-lo pode descarregar abateria do veículo.
ATENÇÃO: Utilizar ou armazenar o Concentradorde oxigénio fora do seu intervalo de temperaturade funcionamento normal e intervalo de pressão
ambiente (altitude) pode afetar o desempenho ereduzir a autonomia da bateria e/ou aumentar o tempo
de carregamento da bateria. Consulte os limites detemperatura de funcionamento e armazenamento nasecção Especificações deste manual.
ATENÇÃO: Para os concentradores de oxigénio
equipados com baterias: Armazene em local frescoe seco para ajudar a garantir uma maior vida útilda sua bateria. O armazenamento do Concentradorde oxigénio por períodos de tempo prolongados aaltas temperaturas ou com uma bateria totalmentecarregada ou descarregada pode degradar a vidaútil da bateria. Não tente abrir a bateria: não existem
peças reparáveis no seu interior. Mantenha as baterias
afastadas do alcance das crianças.
ATENÇÃO: UTILIZE APENAS as baterias fornecidas
pelo Fabricante. Para a eliminação adequada dabateria, entre em contacto com o seu Fornecedor do
Equipamento ou a sua agência governamental local
para se informar sobre os requisitos de eliminação.

164 - PORPN MN232-C4 K | Manual do utilizador

Nota: Certifique-se de que a cânula está totalmente
inserida e fixa. Isso garante que o Concentradorde oxigénio deteta corretamente a inspiração paraadministração do oxigénio. Durante a inalação, deverá
ouvir ou sentir o fluxo de oxigénio pelos tubos da cânula
nasal.
Nota: As baterias de lítio podem perder a capacidade
permanentemente quando expostas a temperaturas
extremamente quentes com as baterias totalmentecarregadas ou completamente gastas. Para um
armazenamento prolongado, recomendamos que asbaterias sejam carregadas a um nível de 25 a 50%
e permaneçam num intervalo de temperatura de -20°C –20°C (-4°F – 68°F).
Nota: A utilização de alguns acessórios de administração
de oxigénio, não especificados para utilização com este
Concentrador, pode prejudicar o seu desempenho. Os
acessórios recomendados estão mencionados neste
manual.
Nota: Não opere o Concentrador de Oxigénio sem que ofiltro de admissão de ar esteja corretamente posicionado.
Se for fornecido um segundo filtro, introduza o filtro desubstituição antes de limpar o filtro sujo. Limpe o filtro
sujo numa solução tépida de água e sabão e depois
seque muito bem antes de voltar a usar.
Nota: O fabricante não recomenda a esterilização deste
equipamento.
Nota: Nunca existe o perigo de esgotar o oxigénio de
uma divisão ao usar o Concentrador de oxigénio.
AVISO: O CONCENTRADOR DE OXIGÉNIONUNCA DEVE SER UTILIZADO OU
LIGADO A UMA FONTE DE ALIMENTAÇÃOEXTERNA (CA/CC) SEM UMA BATERIAINSTALADA. NÃO TENTE ATIVAR A
UNIDADE NEM LIGÁ-LA À ALIMENTAÇÃOEXTERNA QUANDO A UNIDADE DA
BATERIA TIVER SIDO RETIRADA.
RETIRE A UNIDADE DA BATERIA APENAS QUANDOA UNIDADE ESTIVER DESATIVADA (OFF) E
DESLIGADA DA ALIMENTAÇÃO ELÉTRICA. NÃO
TENTE RETIRAR A UNIDADE DA BATERIA COM A
UNIDADE AINDA ATIVA (ON) OU LIGADA À FONTE DE
ALIMENTAÇÃO.
Nota: Se o Concentrador de oxigénio tiver sidoarmazenado durante um período de tempo prolongado
fora das suas condições de funcionamento ambientais
normais (apresentadas no quadro acima), deve
permitir que o equipamento volte à temperatura defuncionamento normal antes de ligá-lo.

FreeStyle Comfort
Especificações
FreeStyle Comfort
Dimensões (com uma bateria
individual)

10,0 x 7,3 x 3,1 in. (25,4 x 18,5 x 7,9 cm)

Dimensões (com uma bateria dupla)

11,0 x 7,3 x 3,1 in. (27,9 x 18,5 x 7,9 cm)

Peso (com uma bateria individual)

5 lbs (2,3 kg)

Peso (com uma bateria dupla)

6 lbs (2,7 kg)

Nível sonoro nominal***39,93 dB (A) ± 0,42 dB (A) @ 2

Definições de fluxo

Definição pulsátil 1, 2, 3, 4, 5

Definição de saída de O2

Definição 1 - 210 mL/min
Definição 2 - 420 mL/min
Definição 3 - 630 mL/min
Definição 4 - 840 mL/min
Definição 5 - 1050 mL/min(a 20bpm)

Concentração de oxigénio*Monitor de oxigénio
Pressão máxima de saída
Sensibilidade de dosagem padrão

90% (+5,5% / -3%)
Sim
<30 PSI> -0,1 cm H2O (ajustável)Alimentação CA

100–240 V CA, 50-60 Hz, 120W Max

Alimentação CC

11–18 V CC (10 Amp. máx.)Tempo de recarga da bateria

Bateria individual: 3,5 horas
Bateria dupla: 6 horas

Durações da unidade da bateria

Duração da bateria individual: Definição 2 - até 4 horas
Duração da bateria dupla: Definição 2 - até 8 horas

Temperatura de funcionamento**

De 5°C a 40°C (41°F a 104°F)

Tipo de Bateria
Capacidade da bateriaÍon de Lítio
Célula única: 6700 mAh; Célula dupla: 13400 mAh

Temperatura de transporte earmazenamento (unidade)-13°F a +158°F (-25°C a +70°C) até 10000 pés (3048 m)

Temperatura de transporte earmazenamento (bateria)-13°F a +158°F (-25°C a +70°C) até 10000 pés (3048 m)

Altitude de funcionamento**-1250 a 10 000 pés (-381 a 3048 m) (testado a 700 – 1060 hPa)

Humidade de funcionamento

Humidade relativa de 15 - 95% (sem condensação)Umidade de armazenamento

0 - 90% sem condensação

Garantia limitada

3 anos para a unidade
1 ano para a bateria e fontes de alimentação

* Com base nos limites de pressão atmosférica de 14,7 psi (101 kPa) a 70°F (21°C)** A unidade funcionará em todos os ambientes, desde que o ambiente esteja com as faixas de temperatura e umidade
especificadas. A utilização fora destas especificações de funcionamento pode limitar a capacidade do concentrador de satisfazera especificação de Concentração de oxigénio a taxas de fluxo por litro mais altas. O ecrã de aquecimento poderá continuar a serapresentado até que o concentrador alcance a concentração alvo ou até 15 minutos. Continue a utilizar a unidade durante este
período.
*** Nível sonoro medido pelo método de teste 14-1 10/2018 MDS-Hi.
A vida útil esperada do dispositivo é de, no mínimo, cinco anos. A vida útil esperada dos leitos da peneira é de aproximadamente
um ano se o concentrador não for utilizado durante todo esse período.
PN MN232-C4 K | Manual do utilizador

POR - 165

FreeStyle Comfort

Instruções de utilização
1. Coloque e posicione o FreeStyle Comfort num local
adequado para a utilização.
a. O concentrador deve ser bem ventilado para que
todos os filtros de entrada de ar e saída de ar não
fiquem obstruídos.
b. O Concentrador deve ser posicionado de tal
modo que todos os indicadores sonoros e visuais
sejam facilmente vistos e ouvidos.
c. Certifique-se de que os filtros de entrada doar estão colocados antes de utilizar a unidade.
Se faltar um filtro na unidade, contacte ofornecedor do seu equipamento.
2. Instale uma unidade da bateria no concentrador.
a. Alinhe a bateria com a estrutura inferior a partir
do centro do FreeStyle Comfort.

3. Introduza a fonte de alimentação CA ou CC norecetáculo da porta de alimentação no canto inferiordireito frontal da unidade.

4. Ligue uma cânula nasal ao encaixe de saída deoxigénio. Certifique-se de que a ligação está segura
antes de utilizar.
NOTA: Quaisquer cânulas utilizadas que não sejam asrecomendadas podem afetar o desempenho.

b. Deslize a bateria para dentro até ouvir um
estalido para confirmar que ficou encaixada no
local.

166 - PORPN MN232-C4 K | Manual do utilizador

Nota: Para o fornecedor do equipamento: Os
seguintes acessórios de administração de oxigênio são
recomendados para uso com o FreeStyle Comfort:• Cânula nasal com 7 pés (2,1 m) de tubulação (LPM
máximo: 6 LPM): Número de peça CAIRE 5408-SEQ• Informações adicionais sobre acessórios recomendadosestão disponíveis online em www.caireinc.com.

FreeStyle Comfort
5. Ligue o FreeStyle Comfort premindo o botão deenergia e selecione a sua definição de fluxo de arprescrita, utilizando os botões para aumentar (+) ou
reduzir (-) o fluxo.

Volume pulsátil com autoSAT(mL)*Configuração
Frequência
respiratória

1

2

3

4

5

15

10,5

21,0

31,5

42,0

52,5

20

10,5

21,0

31,5

42,0

52,5

25

10,5

21,0

31,5

42,0

42,0

30

10,5

21,0

31,5

35,0

35,0

35

10,5

21,0

30,0

30,0

30,0

40

10,5

21,0

26,3

26,3

26,3

*A tolerância é de +/- 15%

É emitido um breve sinal sonoro para indicar queo dispositivo está ligado.

Volume pulsátil sem autoSAT(mL)* Configuração

Nota: Quando o FreeStyle Comfort é ligado inicialmente
a definição de fluxo predefinida é 2. Se a definição foralterada, a unidade liga na definição de fluxo anterior.

Frequência
respiratória

1

2

3

4

5

15

14,0

28,0

42,0

56,0

70,0

6. Coloque a cânula nasal no rosto, seguindo as
instruções do fabricante fornecidas com a cânula
e respire normalmente. Deverá sentir um sopro deoxigénio administrado pela cânula nasal de cada vez
que inala.

20

10,5

21,0

31,5

42,0

52,5

25

8,4

16,8

25,2

33,6

42,0

30

7,0

14,0

21,0

28,0

35,0

35

6,0

12,0

18,0

24,0

30,0

7. Para alterar a definição do fluxo, prima os botões
para aumentar (+) ou reduzir (-) o fluxo até obtera taxa de fluxo pretendida. É normal ouvir uma
diferença no som ao alterar as definições de fluxo.

40

5,3

10,5

15,8

21,0

26,3

8. Desligue o FreeStyle Comfort premindo o botão deenergia.
Nota: O ecrã do visor entra no modo de espera eescurece após 20 segundos. Prima qualquer botão parareativar o ecrã do modo de espera.
9. Para as funcionalidades avançadas, consulte as
instruções de navegação do menu

CARACTERÍSTICAS ESPECIAIS DO
PRODUTO
autoSAT®
O recurso autoSAT proprietário do CAIRE permite queo FreeStyle Comfort forneça um volume consistente deoxigênio (bolus) por respiração, até uma taxa máxima
de respiração. À medida que a taxa de respiração dousuário aumenta durante atividades como caminhar, orecurso autoSAT ajusta a operação do dispositivo paramanter um volume de bolus definido.
UltraSense®
Este dispositivo sensível de dosagem por pulsação
deteta as alterações de pressão durante a sua respiração
e administra a pulsação de oxigénio quando inspira.
autoDOSE™
Em caso de ocorrência de um alarme de “sem inspiração”,
esta funcionalidade efetua pulsações de oxigénio a 20
inspirações por minuto na atual configuração de fluxo da
unidade até o estado de alarme ser resolvido.
PN MN232-C4 K | Manual do utilizador

POR - 167

FreeStyle Comfort

Fontes de alimentação

O FreeStyle Comfort pode ser alimentado de três formas:• Unidade da bateria recarregável• Alimentação CA• Alimentação CC
As informações sobre a operação de um destes
componentes estão disponíveis nas secções que seseguem.

REMOVER A BATERIA

Empurre o botão de libertação da bateria para baixo,
afastando-o do FreeStyle Comfort. MANTENDO estebotão premido, deslize a bateria para fora afastando-a
do concentrador.
AVISO: O CONCENTRADOR DE OXIGÉNIODEVE ESTAR DESATIVADO E DESLIGADODA ALIMENTAÇÃO EXTERNA ANTES DE
RETIRAR A UNIDADE DA BATERIA.

AVISO: UTILIZE APENAS AS FONTES
DE ALIMENTAÇÃO DESCRITAS NESTEMANUAL DO UTILIZADOR. A UTILIZAÇÃO
DE FONTES DE ALIMENTAÇÃO E
UNIDADES DA BATERIA QUE NÃO SEJAMDA CAIRE PODE PROVOCAR DANOSNA UNIDADE E UM FUNCIONAMENTOINCORRETO.

UNIDADE DA BATERIAO FreeStyle Comfort pode ser alimentado pela unidadeda bateria recarregável fornecida com o aparelho. Não
existem limitações operacionais do concentrador ao
funcionar com a unidade da bateria. Todas as taxas
de fluxo estão disponíveis na alimentação através dabateria.

VERIFICAR A CARGA DA UNIDADE DA
BATERIA

Para verificar a carga da unidade da bateria instalada
no FreeStyle Comfort, consulte o ícone da bateria e otempo restante previsto no visor.

INSTALAR A BATERIA

Alinhe a bateria com a estrutura inferior a partir docentro do FreeStyle Comfort. Empurre a bateria para
dentro até ouvir um estalido para confirmar que ficou
encaixada no local.

80–100%60-80%

40-60%

20-40%

20%

168 - PORPN MN232-C4 K | Manual do utilizador

FreeStyle Comfort
Nota: Se a bateria estiver a carregar, é apresentado um
indicador de ficha em vez da percentagem de carga
restante.
Para verificar a carga da bateria quando não está
instalada no FreeStyle Comfort, coloque-a plana e
prima o botão TEST. As luzes indicadoras do medidor
da bateria (25-100%) acendem-se à direita do botãoTEST para indicarem o nível de carga da unidade dabateria.

TEMPOS DE FUNCIONAMENTO DA BATERIA
Quando a alimentação externa CA e CC é desligada,o concentrador muda automaticamente para autilização da energia da bateria se estiver instalada uma
bateria carregada. O quadro abaixo mostra os tempos
habituais para uma unidade da bateria nova. Os tempos
da unidade da bateria mantêm-se constantes seja qual
for a velocidade da sua respiração.
Configuração
PulsátilDuração da bateria
individual (8 células)Duração da bateria
dupla (16 células)1

até 8 horas

até 16 horas2

até 4 horas

até 8 horas3

até 3 horas

até 6 horas4

até 2,25 horas

até 4,5 horas5

até 2 horas

até 4 horasPERÍODO TÍPICO DE RECARGA DA BATERIACARREGAMENTO INICIAL

A bateria nova fornecida com o seu FreeStyle Comfort
só está parcialmente carregada quando é enviada defábrica. Antes da primeira utilização, deve carregar
completamente a unidade da bateria utilizando energia
CA ou CC.

INFORMAÇÕES GERAIS DE
CARREGAMENTO

O período típico para recarregar completamente a
sua unidade da bateria totalmente descarregada é de
3,5 horas para uma bateria individual e 6 horas parauma bateria dupla.

Nota: A bateria pode não começar a carregar inicialmente
se a unidade da bateria estiver completamente gasta.

VIDA ÚTIL DA BATERIA

As baterias FreeStyle Comfort foram concebidas para
durarem 300 ciclos de carga/descarga.

O FreeStyle Comfort recarrega a unidade sempre que
esteja instalada no concentrador e o concentrador
esteja ligado a uma fonte de alimentação CA ou CC
externa (exceto numa aeronave).
A bateria está totalmente carregada quando
apresentar 100%.
Alternativamente, a unidade da bateria pode sercarregada utilizando o carregador de bancada opcional.

PN MN232-C4 K | Manual do utilizador

POR - 169

FreeStyle Comfort

Fonte de alimentação CA e CC
O FreeStyle Comfort inclui uma fonte de alimentação
CA e CC de origem.
A fonte de alimentação CA inclui o seguinte:a. Unidade de alimentação CA que funciona
entre 100-240 V, 50-60 Hz.
b. Cabo de alimentação CA para estabelecer
ligação à unidade de alimentação e à tomada de
CA correspondente.

O transformador CC consiste num cabo que é ligado
diretamente à unidade e à tomada de CC.

ALIMENTAÇÃO CA

2. Ligue o cabo de alimentação CA à unidade dealimentação CA.

A alimentação CA destina-se à utilização em casa
ou onde se encontre alimentação de energia CApadrão disponível. O FreeStyle Comfort é totalmente
funcional com energia CA. O FreeStyle Comfort
também recarrega a unidade da bateria sempre que
exista energia CA disponível (exceto numa aeronave).
A unidade da bateria recarrega quer o FreeStyle
Comfort esteja ligado ou desligado.
AVISO: NÃO LIGUE A ALIMENTAÇÃOEXTERNA SEM UMA UNIDADEDA BATERIA INSTALADA NO
CONCENTRADOR DE OXIGÉNIO.
Para ligar o FreeStyle Comfort à alimentação CA:
1. Ligue a unidade de alimentação CA á entrada doconetor no canto inferior direito frontal do FreeStyle
Comfort. Certifique-se de que empurra para dentro
de modo a que a ligação fique segura.

170 - PORPN MN232-C4 K | Manual do utilizador

3. Ligue a extremidade oposta do cabo de alimentação
CA à parede ou a outra tomada elétrica adequada.

FreeStyle Comfort
4. Ligue o FreeStyle Comfort premindo o botão deenergia e selecione a sua definição de fluxo de arprescrita, utilizando os botões para aumentar (+) ou
reduzir (-) o fluxo.

Nota: Se a unidade da bateria tiver de ser recarregada,o ícone da bateria no vidro apresenta uma única barra
cor de laranja.

ALIMENTAÇÃO CC

A alimentação CC destina-se à utilização em tomadaspara acessórios de automóveis, embarcações ou outros
veículos motorizados. O FreeStyle Comfort é totalmente
funcional com energia CC. O FreeStyle Comfort
também recarrega a unidade da bateria se necessário
sempre que exista alimentação CC disponível. A
unidade da bateria recarrega se a unidade estiver ligada
ou desligada, caso esteja em utilização ou não.

3. Ligue a extremidade oposta do cabo de alimentaçãoCC à saída de alimentação CC do automóvel ou deoutro veículo.

4. Ligue o FreeStyle Comfort premindo o botão deenergia e selecione a sua definição de fluxo de arprescrita, utilizando os botões para aumentar (+) ou
reduzir (-) o fluxo.
Nota: Se a unidade da bateria tiver de ser recarregada,o ícone da bateria no vidro apresenta uma única barra
cor de laranja.

AVISO: NÃO LIGUE A ALIMENTAÇÃOEXTERNA SEM UMA UNIDADEDA BATERIA INSTALADA NO
CONCENTRADOR DE OXIGÉNIO.
AVISO: NÃO UTILIZE O APARELHO EM TOMADAS DE
AUTOMÓVEIS DE 12 V CC ENQUANTO O VEÍCULO
ESTIVER DESLIGADO.
1. Ligue o veículo motorizado.
2. Ligue a fonte de alimentação CC ao recetáculo da
porta de alimentação no canto inferior direito frontal
do FreeStyle Comfort. Certifique-se de que empurrapara dentro de modo a que a ligação fique segura.

PN MN232-C4 K | Manual do utilizador

POR - 171

FreeStyle Comfort
Limpeza, Conservação e
Manutenção de Rotina
MANUTENÇÃO DE ROTINAManutenção preventiva
O dispositivo não requer manutenção. Os filtros dentro
do dispositivo destinam-se a durar a vida útil dodispositivo. Se o dispositivo não estiver funcionando
corretamente, a unidade deverá ser devolvida ao
fornecedor para reparo. A reparação dos componentes
internos no interior da caixa do FreeStyle Comfort
deve ser realizada pelo fornecedor do oxigénio.
O operador da unidade és responsável pelos seguintes
procedimentos de manutenção.
• Semanalmente – Limpar os filtros de entrada do ar
AVISO: NÃO TENTE ABRIR NEM
REMOVER AS TAMPAS DE PLÁSTICODO CONCENTRADOR DE OXIGÉNIO. A
UNIDADE SÓ DEVE SER ABERTA POR UM
TÉCNICO DE ASSISTÊNCIA QUALIFICADO.

FILTROS DE ENTRADA DO AR

O ar entra no FreeStyle Comfort através de filtros
de entrada do ar situados na parte inferior do painel
frontal do concentrador de oxigénio. Estes filtros
impedem a entrada de pó e de outras partículas
grandes no interior da unidade.
Os filtros de entrada do ar devem ser limpos, nomínimo, uma vez por semana. O seu fornecedor
de oxigénio poderá aconselhá-lo a limpá-los com
maior frequência, dependendo das condições defuncionamento. Siga estes passos para limpar
corretamente os filtros de entrada de ar.
1. Retire os dois filtros do painel frontal do FreeStyle
Comfort empurrando a patilha correspondente para
cima e puxando para fora.

AVISO: NÃO OPERE O CONCENTRADORDE OXIGÉNIO SEM QUE O FILTRO
DE ADMISSÃO DE AR ESTEJACORRETAMENTE POSICIONADO.

CUIDADOS DA UNIDADE DA BATERIA

Não permita que a unidade da bateria seja exposta alíquidos.
Não tente desmontar ou abrir a unidade da bateria.
Não exponha a bateria a chamas.
Não deixe cair a unidade da bateria.
Mantenha a unidade da bateria afastada do alcance dascrianças.
Se a sua unidade da bateria parecer danificada, contacte
o seu fornecedor de oxigénio antes de utilizar.
Não coloque a bateria no interior ou perto de um forno
micro-ondas ou outros eletrodomésticos de cozinha.

ARMAZENAMENTO DA UNIDADE DA
BATERIAA bateria do concentrador de oxigénio não tem de estar
completamente descarregada antes de recarregar. É
recomendada a recarga da bateria do concentrador deoxigénio após cada utilização.
Não deixe a bateria num local exposto a calor
excessivo, tal como exposto á luz solar direta ou nointerior de um automóvel. Fazê-lo poderá provocar
a deterioração do desempenho e da vida útil dabateria. Também poderá fazer com que a bateria gerecalor, fumo, incendiar-se ou explodir. A unidade dabateria deve ser armazenada num ambiente estável,caracterizado por um nível reduzido de humidade(inferior a 70% de HR) e livre de gases corrosivos.
Para evitar a formação de ferrugem, evite condições
que possam criar condensação, tais como flutuações
rápidas na temperatura ambiente.
Para um armazenamento durante um mês ou mais, érecomendado um estado descarregado ou parcialmentecarregado entre 50% e 75% (dois a três LEDs). Depois
de carregada, retire a bateria do concentrador deoxigénio.

2. Lave ambos os filtros numa solução de água quente
e detergente.
3. Enxague cuidadosamente os filtros e remova oexcesso de água com uma toalha suave e sem cotão.
Certifique-se de que os filtros estão completamente
secos antes de voltar a instalá-los.
4. Instale os filtros limpos e secos.
172 - PORPN MN232-C4 K | Manual do utilizador

FreeStyle Comfort
LIMPEZADESLIGUE a unidade e desconecte-a da alimentação
CA ou CC antes de qualquer ação de limpeza. NÃO
pulverize diretamente a caixa exterior. Utilize um panoou esponja húmidos (não molhados). Pulverize o panoou a esponja com uma solução de . limpador domésticosuave para limpar a caixa e as fontes de alimentação.
Proceda como indicado pelo fabricante da solução delimpeza.
AVISO: DESLIGUE A UNIDADE E O
CABO DE ALIMENTAÇÃO DA TOMADAELÉTRICA ANTES DE LIMPAR A
UNIDADE PARA EVITAR CHOQUESELÉTRICOS ACIDENTAIS E O PERIGO DE
QUEIMADURAS. APENAS O FORNECEDORDO SEU EQUIPAMENTO OU UM TÉCNICO
DE ASSISTÊNCIA QUALIFICADO
DEVERÃO REMOVER AS TAMPAS OUREALIZAR TAREFAS DE MANUTENÇÃO
NA UNIDADE.
AVISO: DEVE SER EXERCIDO CUIDADO PARAEVITAR QUE O CONCENTRADOR DE OXIGÉNIO OU AUNIDADE DA BATERIA SE MOLHEM OU A ENTRADA
DE SUBSTÂNCIAS LÍQUIDAS NO INTERIOR DA
UNIDADE. ISTO PODE PROVOCAR UMA AVARIA OU
PARALISAÇÃO E RESULTAR NUM MAIOR RISCO DE
CHOQUES ELÉTRICOS OU QUEIMADURAS.

AVISO: USE UM PANO HÚMIDO (NÃO MOLHADO) OUESPONJA PARA LIMPAR A UNIDADE DA BATERIA.
PRIMEIRO, PULVERIZE O PANO OU ESPONJA COM
UM PRODUTO DE LIMPEZA DOMÉSTICA SUAVE E
LIMPE A CAIXA DA UNIDADE DA BATERIA.
AVISO: NÃO LIMPE O SISTEMA DE LAVAGEM
AUTOMATIZADO.

SUBSTITUIÇÃO DA CÂNULA

Siga sempre as instruções do fabricante da cânula
para uma utilização correta. Substitua a cânula
nasal ou os tubos de oxigénio a cada 3 a 6 meses(se utilizar diariamente, 3 meses) de acordo com asrecomendações do fabricante Tal como recomendado
pelo fabricante da cânula ou pelo fornecedor deoxigénio. O seu médico ou fornecedor de oxigénio
fornecer-lhe-á as instruções de limpeza e substituição.
Pode adquirir artigos de substituição adicionais através
do seu fornecedor de oxigénio.

AVISO: NÃO UTILIZE LÍQUIDOS DIRETAMENTE NA
UNIDADE. UMA LISTA DE AGENTES QUÍMICOSINDESEJÁVEIS INCLUI, MAS NÃO SE LIMITA AOSSEGUINTES: ÁLCOOL E PRODUTOS À BASE
DE ÁLCOOL, PRODUTOS À BASE DE CLORO
CONCENTRADOS (CLORETO DE ETILENO) E
PRODUTOS À BASE DE ÓLEO. ESTES PRODUTOS
NÃO DEVEM SER UTILIZADOS PARA LIMPAR A CAIXA
DE PLÁSTICO NO CONCENTRADOR, UMA VEZ QUE
PODEM DANIFICAR A UNIDADE.
AVISO: LIMPE A CAIXA, O PAINEL DE CONTROLO EO CABO DE ALIMENTAÇÃO COM UM DETERGENTEDOMÉSTICO SUAVE APLICADO COM UM PANO OUESPONJA HÚMIDOS (NÃO MOLHADOS) E DEPOISSEQUE BEM TODAS AS SUPERFÍCIES. NÃO PERMITAA ENTRADA DE LÍQUIDOS NO INTERIOR DO
CONCENTRADOR. PRESTE ESPECIAL ATENÇÃO ÀSAÍDA DE OXIGÉNIO PARA A LIGAÇÃO DA CÂNULA
A FIM DE GARANTIR QUE SE MANTÉM LIVRE DE PÓ,ÁGUA E PARTÍCULAS.

PN MN232-C4 K | Manual do utilizador

POR - 173

FreeStyle Comfort

Condições de Alarme
N.º daprioridade

Descrição
do alarme

1

Avaria dosistema,Temperatura
elevada dabateria

Visor

Sinal
sonoroMais detalhes / Ação recomendada

1 bip a cada
20 segundos

A Unidade emite um alarme quando atemperatura atingir o limiar de desativação
de segurança.
Se o limiar de desativação for atingido, aunidade desativa-se. Se a alimentação
externa estiver ligada, é apresentado oecrã de alarme e um sinal sonoro é emitido
durante 2 min.
Efetue um teste com a bateria à temperatura
ambiente. Se o problema persistir, contacte
o fornecedor.

2

Avaria dosistema, Alta
temperatura

1 bip a cada
20 segundos

A Unidade emite um alarme quando atemperatura interna se aproximar do limiar
de desativação.
Se o limiar de desativação for atingido, aunidade desativa-se e é apresentado o ecrãde alarme e o sinal sonoro é emitido durante
2 min.
Certifique-se de que nenhuma saída deventilação da unidade está obstruída. Se o
problema persistir, contacte o fornecedor.

3

Avaria dosistema, Altapressão

1 bip a cada
20 segundos

Se o limiar de desativação for atingido, aunidade desativa-se e é apresentado o ecrãde alarme e o sinal sonoro é emitido durante
2 min.
Se o problema persistir, contacte ofornecedor.

4

Não é
detetada
respiração

1 bip a cada
30 segundosNão é detetada respiração durante
15 segundos.
Ativado após a primeira respiração ou
quando o ecrã de aquecimento é desativado,o que ocorrer primeiro.
Verifique se a cânula apresenta dobras
ou fugas. Aumente a configuração desensibilidade. Se o problema persistir,contacte o fornecedor.

5

Bateria fraca,
Desativação

1 bip a cada
30 segundos

Emite um alarme até a bateria ficar sem
energia e a unidade se desativar.
Carregue a bateria.

6

Aquecimento

174 - PORPN MN232-C4 K | Manual do utilizador

nenhum

A unidade inicia a dosagem automática atéque seja detetada uma respiração ou atéque o Aquecimento termine, o que ocorrer
primeiro.

FreeStyle Comfort
N.º daprioridade

Descrição
do alarme

7

Visor

Sinal
sonoroMais detalhes

Concentração de O2

1 bip a cada
30 segundos
durante5 min.

Se o problema persistir, contacte ofornecedor.

8

Avaria dosistema,
Taxa de fluxo(Falha da
válvula OCD,
Obstrução
da trajetória
do gás)1 bip a cada
30 segundos

Se o problema persistir, contacte ofornecedor.

9

Aviso debateria fraca

1 bip, semrepetição

Carregue a bateria.

10Frequência
respiratória
excedida

1 bip a cada
60 segundos

A frequência respiratória excede os 40 bpm.

11

Avaria dosistema,Falha daventoinha

nenhumSe o problema persistir, contacte ofornecedor.

12

Avaria dosistema,Baixa
pressão

nenhumSe o problema persistir, contacte ofornecedor.

13

Avaria dosistema,Baixa
temperatura

nenhumDeixe a unidade aquecer até à temperatura
ambiente. Se o problema persistir, contacte
o fornecedor.

Informação

Arranque

1 bip deinformação

O ecrã é apresentado no arranque e deverá
ouvir um bip para confirmar que o sinal
sonoro está a funcionar

Se a dosagem for acionada incorretamente,
reduza a configuração da sensibilidade. Se o
problema persistir, contacte o fornecedor.

O ecrã de arranque apresenta a versão dofirmware no canto inferior direito.

PN MN232-C4 K | Manual do utilizador

POR - 175

FreeStyle Comfort
N.º daprioridade

Descrição
do alarme

Informação

Ecrã defuncionamento normal no
Modo autoSAT (semalimentação
CC conetada)nenhumInformação

Ecrã defuncionamento normal
com autoSAT
desativado(sem alimentação CC
conetada)

NenhumInformação

Definição
alterada

1 bip deinformação

Informação

Liga aalimentação
externa na
condição deequipamento
desligado

nenhumInformação

Liga aalimentação
externa na
condição defuncionamento

nenhumInformação

Desliga aalimentação
externa na
condição deequipamento
desligado

1 bip deinformação

Informação

Desliga aalimentação
externa na
condição defuncionamento (coma bateriapresente)

1 bip deinformação

176 - POR

Visor

PN MN232-C4 K | Manual do utilizador

SinalsonoroMais detalhes

Apresenta o ecrã CAIRE durante
2 segundos, seguido pelo estado Bateria
e ligação durante 5 segundos antes de se
desligar

Apresenta o ecrã CAIRE durante 2 segundose, em seguida, desliga-se.

FreeStyle Comfort
N.º daprioridade

Descrição
do alarme

Informação

Visor

Sinal
sonoroMais detalhes

No modode espera quando obotão para
cima oupara baixo
é premido,
apresentao estado dabateria e decarregamento

nenhumO ecrã é apresentado durante 5 segundos e,em seguida, desliga-se

Informação

Modo deavião comalimentação
da bateria

1 bip longo

Informação

Modode avião
enquanto
está ligado

1 bip longo

PN MN232-C4 K | Manual do utilizador

POR - 177

FreeStyle Comfort

Instruções de navegação do menu
É possível entrar no Menu nos modos de Espera e deFuncionamento. Se ocorrer uma condição de alarme
no modo de funcionamento, a unidade sai do Menude forma automática. Se nenhum botão for premido
durante 20 segundos, a unidade sai do menu de forma
automática. No modo de Espera, poderá ter de premiro botão de Menu duas vezes para ativar a unidade e,em seguida, entrar no Menu.
Existem quatro ecrãs de Menu:
1. Ecrã de estado: Exibe o número de rev. do firmware,
tempo de funcionamento, ciclos de descarga dabateria

Modo de voo

2. Modo de voo (LIGADO/DESLIGADO)
3. Sensibilidade: regulável (1-5)
4. autoSAT (LIGADO/DESLIGADO)

Sensibilidade

Exemplo de ecrã de estado

UTILIZAÇÃO DO MENU DE DEFINIÇÕES
Para entrar no menu de Definições, prima o botão
Menu/Seleção
nos modos de Funcionamento ou
de Espera.
Para navegar através dos menus, prima

Página de estado

178 - PORPN MN232-C4 K | Manual do utilizador.

autoSATFreeStyle ComfortPÁGINA DE STATUS
A página de status do Freestyle Comfort mostra
a revisão do firmware, as horas de execução e onúmero de ciclos da bateria na bateria conectada.

Para cancelar o Modo de voo, primaquando oícone CANCEL (CANCELAR) estiver visível.

Modo de voo ligado confirmado

Página de estado

As horas de execução são mostradas em horas
inteiras.
Descargas da bateria é o número de ciclos em que abateria conectada foi descarregada e recarregada.

MODO DE VOOO FreeStyle Comfort é enviado com o Modo de voo
desligado.

Para desligar o Modo de voo, repita o procedimento
anterior mudando para OFF (Desligado).
Premirquando o ecrã apresenta o símbolo “Next”(Seguinte) fará a página do menu avançar para a opção
de página seguinte.
O Modo de voo destina-se a ser usado no Modo defuncionamento. O modo de voo fica automaticamente
inativo se a unidade for deixada em Modo de espera
quando alimentada através da bateria.
Nota: Consulte a secção Ativar o modo de voo para
obter mais informações sobre o Modo de voo.

MODO DE SENSIBILIDADEEcrã principal do modo de voo

O FreeStyle Comfort é enviado da fábrica com uma
configuração padrão do Modo de sensibilidade de 2 e,em condições normais, não serão necessários ajustes.
A configuração de sensibilidade pode ser ajustada entre
os valores 1-5, sendo 1 a menos sensível e 5 a maissensível.

Para ligar o modo, primaquando o ícone
CHANGE (ALTERAR) estiver visível. O valor
mudará para a palavra azul “ON” (Ligado).

Ecrã do modo de sensibilidade com a Configuração 2

Modo de voo ligado selecionado

Para confirmar o Modo de voo, primaquandoo ícone SAVE (GUARDAR) estiver visível. O valor
mudará para a palavra preta “ON” (Ligado).

Para ajustar o Modo de sensibilidade, primaquando o ícone CHANGE (ALTERAR) estiver visível.
O valor mudará para o número azul seguinte.

PN MN232-C4 K | Manual do utilizador

POR - 179

FreeStyle Comfort
MODO AUTOSATO FreeStyle Comfort é fornecido com o Modo
autoSAT ligado.

Configuração 3 do modo de sensibilidade selecionada

Para confirmar a Configuração 3 no Modo dequando o ícone SAVE
sensibilidade, prima(GUARDAR) estiver visível. O valor mudará para o“3” preto.
Para cancelar a alteração do Modo de sensibilidade,quando o ícone CANCEL (CANCELAR)
prima
estiver visível.

Configuração 3 do modo de sensibilidade confirmada

Ecrã principal do modo autoSAT

Para desligar o modo, primaquando o ícone
CHANGE (ALTERAR) estiver visível. O valor mudará
para a palavra azul “OFF” (Desligado).

Modo autoSAT desligado selecionado

Para confirmar o Modo autoSAT desligado, primaquando o ícone SAVE (GUARDAR) estiver visível. O
valor mudará para a palavra preta “OFF” (Desligado).
Para cancelar o Modo autoSAT ligado, primaquando o ícone CANCEL (CANCELAR) estivervisível.

Configuração 5 (Máx) do modo de sensibilidade confirmada

Premirquando o ecrã apresenta o símbolo “Next”(Seguinte) fará a página do menu avançar para a opção
de página seguinte.
Modo autoSAT desligado confirmado

Para ligar o Modo autoSAT, repita o procedimento
anterior mudando para ON (Ligado).
Premirquando o ecrã apresenta o símbolo “Next”(Seguinte) fará a página do menu avançar para a opção
de página seguinte.

180 - PORPN MN232-C4 K | Manual do utilizador

FreeStyle Comfort
Resolução de Problemas
Problema

Causa provável

Solução

O FreeStyle Comfort apresentaum alarme e emite um bip
intermitente.

Consulte o quadro deCondições de alarme.

Consulte o quadro de Condições de alarme.

O Concentrador de oxigénio
não liga.

A bateria está descarregada.

Alimente a unidade através da tomada de
CA ou CC.

O concentrador não está ligado
à alimentação externa.
Avaria geral.

Assegure-se de que as conexões externas estão
bem efetuadas.
Execute um Ciclo de alimentação completoretirando a bateria e desligando qualquer fontede alimentação externa CA ou CC. Em seguida,
instale novamente a bateria e conclua o ciclode alimentação. Ligue novamente a fonte dealimentação externa, conforme necessário.
Contacte o fornecedor do equipamento e use outra
fonte de oxigénio, conforme necessário.

Restantes problemas.

Execute um Ciclo de alimentação completoretirando a bateria e desligando qualquer fontede alimentação externa CA ou CC. Em seguida,
instale novamente a bateria e conclua o ciclode alimentação. Ligue novamente a fonte dealimentação externa, conforme necessário.
Mude para outra fonte de oxigénio, conforme
disponível, e contacte o fornecedor do
equipamento.

Nota: Alguns alarmes fazem com que a unidade se desligue.
Nota: Durante uma perda de energia total, o sistema de alarmes é reiniciado (elimina todos os alarmes anteriores) e só
apresenta um alarme se a condição persistir depois de restaurada a alimentação.

PN MN232-C4 K | Manual do utilizador

POR - 181

FreeStyle Comfort

Informações de transporte
O FreeStyle Comfort cumpre todas as diretrizes
da FAA referentes à utilização a bordo de aviões
comerciais. Antes de qualquer viagem (de avião,comboio, cruzeiro, etc.), deverá:• Garantir que o concentrador se encontra em boas
condições de funcionamento.
• Contacte a empresa de transporte pelo menos
48 horas antes da partida para informar que irá
viajar com um FreeStyle Comfort.
É recomendado transportar sempre baterias extra
nas suas viagens. Para viagens aéreas, providencie
a autonomia da bateria suficiente para o tempo
em terra (check-in antes do voo, segurança, etc.) etransbordos. A maioria das companhias aéreas exige
que tenha um número adequado de unidades de bateria
completamente carregadas para alimentar o dispositivo
durante, pelo menos, 150% da duração máxima da
viagem prevista (tempos de voo, tempo em terra antes
e após o voo, ligações e atrasos inesperados).

1. Ligue o POC utilizando APENAS a alimentação dabateria quando não estiver ligada qualquer fonte dealimentação externa.

2. Ative o Modo de voo usando um dos métodos
seguintes:
a. Consulte a secção de Instruções de navegação domenu para ativar o Modo de voo.

Nota: Nos voos, recomendamos que as baterias extra
sejam transportadas num saco de transporte protegido.

REGULAMENTO DA FAA

As baterias sobresselentes devem ser protegidas
individualmente para evitar curtos-circuitos através dacolocação na embalagem de comercialização originalou através de outra forma de isolamento dos terminais,
por exemplo, ao aplicar fita adesiva sobre os terminais
expostos ou colocar cada bateria num saco plásticoseparado ou numa bolsa protetora, e transportadas
apenas em bagagens de mão.

b. Mantenha os botões Cima e Baixo premidos
durante 5 segundos até ouvir um longo “bip” esurgir o símbolo do Avião.

ATIVAR O MODO DE AVIÃOAlimentação da bateria
O POC pode ser utilizado numa aeronave de acordo
com os procedimentos de funcionamento normais
utilizando APENAS a alimentação da bateria quandonão estiver qualquer fonte de alimentação externa
ligadaAlimentação Externa
Numa aeronave, se existir uma tomada elétrica
disponível, funcional e que a sua localização sejaadequada para utilizar como fonte de alimentação
do POC, poderá servir de reserva para as baterias do
POC. Neste caso, deve ser seguido o procedimento
recomendado relativamente à transição da bateria paraa alimentação elétrica da aeronave:

182 - PORPN MN232-C4 K | Manual do utilizador

3. Quando o símbolo do avião for apresentado,
ligue o POC.

Nota: Devido às limitações das aeronaves, as baterias
não carregam no Modo de avião.

FreeStyle Comfort
Acessórios

INSERIR A UNIDADE NO SACO DE
TRANSPORTE

Para um desempenho e segurança apropriados, use
apenas estes acessórios listados fornecidos pela CAIRE
através do seu fornecedor de oxigénio. A utilização
de acessórios não incluídos abaixo pode prejudicar odesempenho e/ou segurança do concentrador.

1. Introduza a unidade no fundo do saco, alinhandocorretamente os filtros e as aberturas da bateria.

Acessórios padrão do FreeStyle Comfort
Saco de transporte

MI396-1

MochilaMI459-1

Alça a tiracolo

MI397-1

Fonte de alimentação CA (todos oscabos incluídos)PW036-1SCabo de alimentação CC

CD041-1

Filtro de partículas grossas externo(direita)

FI226-1

Filtro de partículas grossas externo(esquerda)

FI227-1

2. Feche o fecho de correr em torno da bolsa.

Acessórios opcionais disponíveis
Unidade da bateria de 8 células
adicional

BT034-1S

Unidade da bateria de 16 células

BT035-1S

Carregador da unidade da bateria debancadaBT036-1

Bolsa porta-tudo acessória

MI372-2

Nota: Podem estar disponíveis opções adicionais paracabos de alimentação específicos de cada país, tal comomencionado anteriormente. Contacte a CAIRE ou o seu
fornecedor de oxigénio, caso seja necessário encomendaropções alternativas.

INSTALAÇÃO DA ALÇA A TIRACOLO
1. A alça a tiracolo pode ser afixada diretamente àunidade ou ao saco de transporte.
2. Prenda a alça no anel que se encontra na unidade ou
no saco.

Instruções do saco de transporte
Para inserir e fixar o seu FreeStyle Comfort no saco detransporte, siga as seguintes instruções.

PN MN232-C4 K | Manual do utilizador

POR - 183

FreeStyle Comfort

Eliminação e Reutilização
Devolva sempre o FreeStyle Comfort, incluindo todos
os componentes, ao seu prestador de equipamentos
para garantir uma eliminação adequada. Pode também
entrar em contracto com as entidades locais para obter
instruções acerca da correta eliminação da bateria.
A unidade pode ser reutilizada entre as pessoas, masdeve ser devolvida ao provedor de equipamentos para
reprocessamento.

a uma instalação de reciclagem no final da sua vidaútil, de acordo com a Diretiva relativa aos Resíduos deEquipamentos Elétricos e Eletrónicos (REEE).
Os nossos produtos estão em conformidade com adiretiva de Restrição de Substâncias Perigosas (RoHS).
Não contêm mais do que quantidades residuais dechumbo ou de outros materiais perigosos.

WEEE (REEE) e RoHS
Este símbolo destina-se a recordar os
proprietários do equipamento de que devem entregá-loATENÇÃO: Para uma eliminação adequada,
entre em contacto com o fornecedor do seu
equipamento ou agência governamental
local para obter as instruções deeliminação

Materiais em contacto direto com o operador
Invólucro do concentrador........................................................................................................................... PC/ABS
Painel de controlo do concentrador.......................................................................... Poliéster EBG7 ou equivalente
Pega do concentrador...........................................................................................................................Policarbonato
Saída de oxigénio, bocal......................................................................................................................Policarbonato
Etiqueta da unidade..........................................................................................................................................Lexan
Fichas do cabo.......................................................................................................... Policarbonato/cloreto de vinilo
Cabo de Alimentação..............................................................................................................................PVC, Metal
Fonte de alimentação......................................................................................................Lexan 940 (Policarbonato)
Unidade da bateria....................................................................................................................................... PC/ABS
Unidade da bateria, etiquetas da fonte de alimentação............................................................. Película de poliéster
Estojo de transporte do concentrador.................................................................Cordura, neopreno, forro em nylon
Alça................................................................................................ Polipropileno, acetil, vinilo, Cordura, neopreno

Testes de compatibilidade eletromagnética (EMC)
Orientações e Declaração do Fabricante­—Emissões eletromagnéticas
O FreeStyle Comfort destina-se a ser usado no ambiente eletromagnético especificado abaixo. O cliente ou utilizador do
FreeStyle Comfort deve certificar-se de que este é usado no ambiente descrito.
Teste de Emissões
Emissões RF
CISPR 11
Emissões RF
CISPR 11
Emissões harmónicas
IEC 61000-3-2
Flutuações de tensão/emissões inconstantes
IEC 61000-3-3

184 - POR

Conformidade

Ambiente eletromagnético - Orientações

Grupo 1O FreeStyle Comfort utiliza energia de RF apenas para o seu
funcionamento interno. Por conseguinte, as suas emissões de RF
são muito baixas e não é provável que causem interferências nos
equipamentos eletrónicos próximos.

Classe B
Classe A
Em conformidade

PN MN232-C4 K | Manual do utilizador

A utilização do FreeStyle Comfort é adequada em todos osestabelecimentos, incluindo os estabelecimentos domésticos e os que
se encontram ligados à rede pública de fornecimento de energia debaixa voltagem que abastece os edifícios usados para fins domésticos.

FreeStyle Comfort
Orientações e Declaração do Fabricante—Imunidade eletromagnética

O FreeStyle Comfort destina-se a ser usado no ambiente eletromagnético especificado abaixo. O cliente ou utilizador do
FreeStyle Comfort deve certificar-se de que este é usado no ambiente descrito.
Teste de imunidade Nível de teste IEC 60601 Nível de conformidade Ambiente eletromagnético – orientações
Ambiente
O piso deve ser de madeira, cimento
eletromagnético –Contacto ±8 kV
Contacto ±8 kV
ou azulejo. Se o piso estiver coberto deorientações
material sintético, a humidade relativa deve
Descarga de ar ±15 kV
Descarga de ar ±15 kV
ser de 30%, no mínimo.
IEC 61000-4-2
Transiente/disparo
rápido elétrico±2 kV para linhas defornecimento energético

IEC 61000-4-4±1 kV para linhas deentrada/saída±1 kV linha(s) paralinha(s)Sobretensão

±2 kV modo comum em
linhas de CA

IEC 61000-4-5±1 kV diferencial em
linhas de CA±2 kV para linhas defornecimento energético
N/A±1 kV linha(s) paralinha(s)A qualidade da alimentação elétrica
deve ser equivalente à de instalações decuidados de saúde profissionais típicas
e de ambientes de cuidados de saúde
domésticos.

A qualidade da alimentação elétrica
deve ser equivalente à de instalações decuidados de saúde profissionais típicas
e de ambientes de cuidados de saúde
domésticos.

±2 kV modo comum em
linhas de E/S no exterior
0% UT para 0,5 ciclos
0% UT para 0,5 ciclos(0°, 45°, 90°, 135°, 180°, (0°, 45°, 90°, 135°, 180°,
225°, 270°, 315°)
225°, 270°, 315°)
A qualidade da alimentação elétrica
deve ser equivalente à de instalações deQuedas de tensão,cuidados de saúde profissionais típicas
00interrupções curtas e 0% U par 1 ciclo (0 )0% UT para 1 ciclo (0 )Te de ambientes de cuidados de saúde
variações de tensão
domésticos. Se o utilizador do FreeStylenas linhas de entrada
Comfort necessitar de um funcionamento
70% UT (queda de 30%
70% UT (queda de 30%
de fornecimento
na UT) para 25/30 ciclos
na UT) para 25/30 ciclos continuado durante interrupções dede energia
energia, recomendamos que o FreeStyle
(00)
(00)
IEC 61000-4-11
Comfort seja alimentado por uma fonte dealimentação ininterrupta ou bateria.
0% UT para
0% UT para
250/300 ciclos (00)
250/300 ciclos (00)
Os campos magnéticos de frequência
Campo magnético
de energia devem estar a um nível
de frequência decaracterístico de um local típico em
3A/m
3A/m
energia (50/60 Hz)
ambientes de cuidados de saúde
IEC 61000-4-8
profissionais e domésticos.
NOTA UT é a tensão de CA antes da aplicação do nível de teste.
NOTA Em alguns casos, o Freestyle Comfort poderá reiniciar após uma descarga eletrostática significativa e poderá
necessitar de um Ciclo de alimentação completo para retomar o funcionamento normal. Consulte a secção Resolução deproblemas para obter mais detalhes.

PN MN232-C4 K | Manual do utilizador

POR - 185

FreeStyle Comfort
Orientações e Declaração do Fabricante–Imunidade eletromagnética

O FreeStyle Comfort destina-se a ser usado no ambiente eletromagnético especificado abaixo. O cliente ou utilizador do
FreeStyle Comfort deve certificar-se de que este é usado no ambiente descrito.
Teste deNível de teste IEC 60601 Nível de conformidade
Ambiente eletromagnético – orientações
imunidade
RF Conduzida
10 Vrms
10 Vrms
O equipamento portátil e móvel decomunicações RF não deve ser utilizado
IEC 61000-4-6
150 kHz a 80 MHz
mais perto de qualquer peça do FreeStyle
Comfort, incluindo cabos, do que a distância
de separação recomendada calculada apartir da equação aplicável à frequência do
10 V/m
RF irradiada
10 V/mtransmissor.
IEC 61000-4-3
80 MHz a 2700 MHz
Distância de separação recomendada

d = 1,2√ P
d = 1,2√ P = 80 MHz a 800 MHz
d = 1,2√ P = 800 MHz a 2700 MHz
onde P é a potência nominal máxima desaída do transmissor em Watts (W), de
acordo com o fabricante do transmissor ed é a distância de separação recomendada
em metros (m).
As forças de campo provenientes detransmissores de RF fixos, conforme
determinado por um levantamento
eletromagnético do local a devem serinferiores ao nível de conformidade em cadaintervalo de frequência. b
Poderão ocorrer interferências nas
imediações de equipamentos assinalados
com o seguinte símbolo:

NOTA 1 A 80 MHz e 800 MHz, aplica-se o intervalo de frequência superior.
NOTA 2 Estas diretrizes podem não se aplicar em todas as situações. A propagação eletromagnética é afetada pela
absorção e reflexão de estruturas, objetos e pessoas.
a
As forças de campo de transmissores fixos, tais como estações de base para telefones de rádio (telemóveis/sem fios)e rádios móveis terrestres, rádio amador, transmissão de rádio AM e FM e transmissão de televisão não se podem
prever teoricamente com precisão. Para avaliar o ambiente eletromagnético para transmissores de RF fixos, deverá
ser considerado um estudo eletromagnético do local. Se a força do campo medido no local onde o FreeStyle Comfort éutilizado exceder o nível de conformidade RF aplicável indicado acima, o FreeStyle Comfort deve ser observado para
determinar o seu funcionamento normal. Se for observado um desempenho anormal, poderão ser necessárias medidas
adicionais, tais como reorientar ou relocalizar o FreeStyle Comfort.
b

No intervalo de frequências de 150 kHz a 80 MHz, as forças de campo devem ser inferiores a 10 V/m.

186 - PORPN MN232-C4 K | Manual do utilizador

FreeStyle Comfort
Distâncias de separação recomendadas entre os equipamentos de comunicação de RF
portáteis e móveis e o FreeStyle Comfort

O FreeStyle Comfort destina-se a ser utilizado num ambiente eletromagnético no qual as perturbações de RF irradiadas
sejam controladas. O cliente ou o utilizador do FreeStyle Comfort pode ajudar a evitar interferências eletromagnéticas
mantendo uma distância mínima entre os equipamentos de comunicação por RF portáteis e móveis (transmissores) e o
FreeStyle Comfort, tal como se recomenda em baixo, de acordo com a potência máxima de saída do equipamento decomunicações.
Distância de separação de acordo com a frequência do transmissorPotência nominal
máxima de saída domtransmissor
150 kHz a 80 MHz
80 MHz a 800 MHz
800 MHz a 2,5 GHz

d = 1,2√ P

d = 1,2√ P

d = 1,2√ P

W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Para transmissores com uma potência nominal de saída máxima não indicada acima, a distância de separação
recomendada d em metros (m) pode ser estimada utilizando a equação aplicável à frequência do transmissor, em que P é
a potência nominal máxima de saída do transmissor em Watts (W) segundo o fabricante do transmissor.
NOTA 1 A 80 MHz e 800 MHz, aplica-se a distância de separação para o intervalo de frequência superior.
NOTA 2 Estas diretrizes podem não se aplicar em todas as situações. A propagação eletromagnética é afetada pela
absorção e reflexão de estruturas, objetos e pessoas.

Classificação

Tipo de proteção contra choques elétricos:
Classe II
A proteção contra choques elétricos é
obtida através de um isolamento duplo.
Grau de proteção contra choques elétricos:
Tipo BF Equipamento que fornece um grau
particular de proteção contra choques elétricos no que
respeita a		 1) corrente de fuga admissível;		2) fiabilidade da ligação à terra de proteção
(se presente).
		 Não se destina a aplicação cardíaca direta.

Secção 21 Categoria M.
• CISPR 11 / EN 55011 Classe B Grupo 1,“Equipamento industrial, científico e médico
(ISM)”
• Parte 15 das FCC, subparte B – Classe B
Radiadores involuntários
IP22 - Proteção da entrada de dedos ou objetos
semelhantes com mais de 12,5 mm de diâmetro.
A água de gotejamento vertical não deve ter qualquer
efeito nocivo quando a caixa estiver inclinada a umângulo de 15° em relação à sua posição normal.

Testes independentes para a Norma para equipamentos
médicos elétricos:
Testado para cumprir a• IEC 60601-1 edição 3.1: Equipamento médico
elétrico - Parte 1 Requisitos gerais de segurança.
• CAN/CSA C22.2 N.º 60601-1-8 edição
2.1, Equipamento médico elétrico – Parte 1:
Requisitos gerais de segurança

Método de limpeza e controlo de infeções permitido:
Consulte a secção "Limpeza, Conservação e
Manutenção Adequada" do Manual do utilizador
deste FreeStyle.

Proteção de potenciais interferências eletromagnéticas
ou outras entre o equipamento e outros dispositivos.
• Testado para cumprir a EN 60601-1-2/IEC 606011-2: 4.ª edição.
• Testado para cumprir a norma RTCA/DO160

Modo de funcionamento: Funcionamento contínuo.

Grau de segurança da aplicação na presença de gases
anestésicos inflamáveis:
Equipamento não apropriado para tal aplicação.

CAIRE e CAIRE Inc. são marcas registradas da CAIRE Inc. Visite nosso site abaixo para
obter uma lista completa de marcas comerciais.
Marcas comerciais: www.careinc.com/corporate/trademarks.
Copyright © 2021 CAIRE Inc. A CAIRE Inc. reserva-se o direito de descontinuar os
seus produtos ou alterar os preços, materiais, equipamento, qualidade, descrições,especificações e/ou processos dos seus produtos a qualquer momento e sem aviso
prévio, sem quaisquer obrigações ou consequências. Reservamos todos os direitos não
expressamente aqui declarados, conforme aplicável.

PN MN232-C4 K | Manual do utilizador

POR - 187

FreeStyle Comfort

Gebruikersbediening en indicatoren van de systeemstatus
ISO 7000; Grafische symbolen voor gebruik op deapparatuur - Index en samenvatting
Lees de gebruikershandleiding voordat
u het systeem gebruikt. Reg.-nr. 1641
Opslag- of bedrijfstemperatuurbereik.
Reg.-nr. 0632

Luchtvochtigheid bij opslag. Reg.-nr. 2620

Richtlijn 93/42/EEG van de Raad inzake medische
instrumenten
Gevolmachtigde vertegenwoordiger in deEuropese Gemeenschap
Als op het etiket van het product uniquedevice identifier (UDI) het CE ####
symbool staat, voldoet het apparaat aande vereisten van Richtlijn 93/42 / EEG
betreffende medische hulpmiddelen. Het
CE #### symbool geeft het nummer van
de aangemelde instantie aan.
IEC 60417
Apparatuur van klasse II, dubbelgeïsoleerd Reg.-nr. 5172

Luchtdrukbereik. Reg.-nr. 2621

Stroomschakelaar. Reg.-nr. 5009

Naam en adres van de fabrikant.
Reg.-nr. 3082
Productiedatum. Reg.-nr. 2497

ETL-certificering
Agentschap voor veiligheid voor
CAN/CSA C22.2 nr. 601.1 M90 voor
elektrische medische apparatuur

Catalogusnummer. Reg.-nr. 2493
Serienummer. Reg.-nr. 2498

Interne symbolen

Deze kant omhoog. Reg.-nr. 0623

Uit de buurt bewaren van ontvlambare
materialen, olie en vet.

Breekbaar, voorzichtig. Reg.-nr. 0621
Niet demonteren.

Droog en uit de buurt van regen bewaren.
Reg.-nr. 0626

Debiet verhogen.

Stapellimiet in aantal. Reg.-nr. 2403

Debiet verlagen.

ISO 7010: Grafische symbolen - Veiligheidskleuren
en veiligheidssymbolen - Geregistreerde
veiligheidssymbolen
Het lezen van de instructiehandleiding is
verplicht. Reg.-nr. M002
Uit de buurt van open vuur,
vlammen en vonken bewaren. Open
ontstekingsbronnen en roken verboden.
Reg.-nr. P003
Rook niet in de buurt van het apparaat oftijdens de bediening ervan. Reg.-nr. P002
Toegepast onderdeel van type BF(mate van bescherming tegen elektrischeschokken). Reg.-nr. 5333
Waarschuwing. Reg.-nr. W001

188 - DUTPN MN232-C4 K | Gebruikershandleiding

Menuknop.
RTCA/DO160 Deel 21
Conform aan
Categorie M

RTCA DO160 Deel 21 conform aan
Categorie M. Vereiste FAA SFAR 106
Apparatuur van klasse II, dubbelgeïsoleerd
Door de FAA goedgekeurd - POC

FreeStyle Comfort
21 CFR 801.15: Code van federale regelgeving, titel 21
Dit product mag op grond van de federale
wetgeving uitsluitend worden verkocht
door of namens een arts.
Richtlijn 2012/19/EU van de Raad: afgedankteelektrische en elektronische apparatuur (AEEA)
AEEA

Wat is de zuurstofconcentrator?
De lucht die we inademen bevat ongeveer 21% zuurstof,
78% stikstof en 1% overige gassen. Bij de FreeStyle®
Comfort® wordt omgevingslucht via de luchtinlaten in
het apparaat gezogen. Deze lucht stroomt vervolgens
langs een adsorberend materiaal, een 'moleculaire zeef'
genaamd. Dit materiaal scheidt de zuurstof van destikstof en laat alleen de zuurstof door. Bijgevolg wordt
hoogzuivere zuurstof naar de gebruiker gevoerd.
De FreeStyle Comfort combineert een geavanceerde
zuurstofconcentrator en zuurstofbesparende
technologieën om een draagbaar apparaat te ontwikkelen
met het oog op optimale draagbaarheid en mobilisatie.
Dankzij de geavanceerde toediening met pulsdebiet
merkt het apparaat al snel wanneer de gebruiker inademt
en geeft het aan het begin van elke inademing een
zuurstofpuls vrij. De toediening met pulsdebiet is ideaal
voor activiteiten en tijd weg van huis. In tegenstelling tot
een continu debiet wordt er zuurstof bespaard en wordt
de accuduur verlengd. De FreeStyle Comfort is licht
genoeg om met een schouderriem te worden gedragen.
Het apparaat werkt op wisselstroom (muur), gelijkstroom(voertuig) of een accu.

Beoogd gebruik
De CAIRE FreeStyle Comfort zuurstofconcentrator
is bedoeld voor het toedienen van extra zuurstof. Het
apparaat is niet bedoeld voor levensondersteuning enbiedt ook geen mogelijkheden voor patiëntbewaking.

Waarom heeft uw arts zuurstof
voorgeschreven?
Heel wat mensen lijden aan verschillende hart-, long- en
andere luchtwegaandoeningen. Een groot deel hiervan
heeft baat bij de toediening van extra zuurstof thuis,tijdens het reizen of bij dagelijkse activiteiten weg van
huis.
Zuurstof is een gas dat 21% van de omgevingslucht
vormt die we inademen en onze lichamen zijn
afhankelijk van een constante toevoer om goed te kunnen
functioneren. Uw arts heeft u een debiet of instellingvoorgeschreven om uw specifieke luchtwegaandoening
te behandelen.

IEC 60601-1: Medische elektrische apparatuur, deel
1 - Algemene basale veiligheidseisen en essentiële
werking
Druppelbestendig
Op dit product kunnen een of meer Amerikaanse eninternationale octrooien van toepassing zijn. Bezoek
onze website voor een lijst met toepasselijke octrooien.
Pat.: www.caireinc.com/corporate/patents/.

Hoewel zuurstof een niet-verslavend geneesmiddel
is, kan ongeoorloofde zuurstoftoediening gevaarlijk
zijn. U moet medisch advies inwinnen voordat u dezezuurstofconcentrator gebruikt. De leverancier die u
de zuurstofapparatuur verstrekt, zal u tonen hoe u het
voorgeschreven debiet kunt instellen.

Gebruikersprofiel
Concentrators zijn bedoeld voor de toediening van extra
zuurstof aan gebruikers die lijden aan aandoeningen
die hun longen verhinderen op een efficiënte manier
zuurstof vanuit de lucht naar hun bloedbaan te voeren.
Draagbare zuurstofconcentrators (POC's) bewarenof bevatten geen zuurstof. Ze hoeven niet te wordenbijgevuld en kunnen op om het even welke plaats met
wisselstroom of gelijkstroom worden opgeladen (behalvein een vliegtuig). Dit biedt de gebruiker een relatieve
zelfstandigheid bij thuisgebruik, mobilisatie (zowel inals van huis), mobiliteit en algemene levensstijl. Voor het
gebruik van een zuurstofconcentrator is een voorschrift
van een arts vereist en het apparaat doet geen dienst als
een levensondersteunend apparaat.
Hoewel er aan gebruikers van alle leeftijden extra
zuurstof kan worden voorgeschreven, is de typische
gebruiker ouder dan 65 jaar en lijdt hij/zij aanverschillende luchtwegaandoeningen, zoals chronischeobstructieve longaandoening (COPD). Gebruikers
hebben doorgaans goede cognitieve capaciteiten en
moeten hun ongemak kenbaar kunnen maken. Als degebruiker zijn ongemak niet kenbaar kan maken of deetikettering en gebruiksinstructies van de concentrator
niet kan lezen of begrijpen, dan is het gebruik ervanslechts aanbevolen onder toezicht van iemand diehier wel toe in staat is. Als er tijdens het gebruik van
de concentrator ongemak wordt ervaren, worden
gebruikers aanbevolen contact op te nemen met hun
zorgverlener. Ook worden gebruikers aangeraden overreservezuurstof te beschikken (d.w.z. zuurstofflessen)in het geval van een stroomonderbreking of een defectaan de concentrator. Er zijn geen andere specifieke
vaardigheden of gebruikerscapaciteiten vereist voor het
gebruik van de concentrator.

PN MN232-C4 K | Gebruikershandleidin

DUT - 189

FreeStyle Comfort

Uw FreeStyle Comfort uitpakken
Controleer of alle hieronder
opgesomde onderdelen inde verpakking zitten. Neem
onmiddellijk contact op
met de leverancier van uw
zuurstofapparatuur als er
onderdelen ontbreken.

Standaardaccessoires• Draagbare zuurstofconcentrator• Draagtas• Schouderriem• 8-cellige accu
• Wisselstroomvoeding• Wisselstroomsnoer• Gelijkstroomvoeding• Gebruikershandleiding

Optionele accessoires• Extra 8-cellige accu
• 16-cellige accu
• Oplaadstation• Ruime accessoiretas

Uw FreeStyle Comfort leren kennen
Maak u vertrouwd met de belangrijkste functies van de FreeStyle Comfort en de gebruikersinterface.

Vooraanzicht
Riembevestiging

Gebruikersinterface

Luchtuitlaat
Zuurstofuitlaat

Luchtinlaat/-filters
Luchtinlaat/-filters

Houderlipje
voedingspoort

Accu
Ontgrendelingsknop
accu
190 - DUTPN MN232-C4 K | Gebruikershandleiding

FreeStyle Comfort
Bovenaanzicht
Weergavescherm
Debiet verhogen

Aan-/uitknop

Menuknop

Debiet verlagen

Weergavescherm op accu
Debietinstelling

Accuweergave

autoSAT-indicator
Alarmweergave
Resterend
accupercentage

Bolustoedieningsindicator

Weergavescherm op wissel-/gelijkstroom

Debietinstelling
Accuweergave

Alarmweergaveruimte

Laadindicator

PN MN232-C4 K | Gebruikershandleidin

DUT - 191

FreeStyle Comfort

Belangrijk!

Veiligheidsinstructies zijn als volgt vermeld:
WAARSCHUWING: BELANGRIJKE
VEILIGHEIDSINFORMATIE VOORGEVAREN DIE ERNSTIG LETSEL KUNNEN
VEROORZAKEN.
LET OP: Belangrijke informatie om schadeaan de FreeStyle Comfort te voorkomen.
Opmerking: deze informatie vereist extra aandacht.

Indicaties voor gebruik
WAARSCHUWING: HET IS ERG
BELANGRIJK DAT ALLEEN HETVOORGESCHREVEN ZUURSTOFNIVEAUWORDT GEKOZEN. WIJZIG DE
DEBIETSELECTIE NIET TENZIJ DIT WERDAANGEGEVEN DOOR EEN BEVOEGDECLINICUS.
WAARSCHUWING: VOLGENS DE AMERIKAANSEFEDERALE WETGEVING MAG DIT APPARAATALLEEN WORDEN VERKOCHT OF VERHUURD OP
VOORSCHRIFT VAN EEN ARTS OF EEN ANDEREERKENDE PROFESSIONELE ZORGVERLENER.
WAARSCHUWING: DIT APPARAAT MAG NIET ALSLEVENSINSTANDHOUDINGSSYSTEEM WORDENGEBRUIKT. BIJ GERIATRISCHE, PEDIATRISCHE OFENIGE ANDERE GEBRUIKERS DIE NIET IN STAATZIJN EEN ONGEMAK KENBAAR TE MAKEN, KAN
BIJKOMEND TOEZICHT VEREIST ZIJN. GEBRUIKERSMET AUDITIEVE EN/OF VISUELE BEPERKING(EN)VEREISEN MOGELIJK HULP OP HET VLAK VAN
ALARMBEWAKING.

Contra-indicaties voor gebruik
WAARSCHUWING: IN BEPAALDEOMSTANDIGHEDEN KAN GEBRUIK VAN
NIET-VOORGESCHREVEN ZUURSTOFGEVAARLIJK ZIJN. DIT APPARAAT IS
ALLEEN BESTEMD VOOR GEBRUIK OPVOORSCHRIFT VAN EEN ARTS.
WAARSCHUWING: NIET BESTEMD VOOR GEBRUIK IN
DE NABIJHEID VAN ONTVLAMBARE ANESTHETICA.
WAARSCHUWING: ZOALS BIJ ELK ELEKTRISCH
APPARAAT KRIJGT DE GEBRUIKER MOGELIJK TE
MAKEN MET ONDERBREKINGEN VAN DE WERKINGTEN GEVOLGE VAN EEN STROOMSTORING OF DENOODZAAK OM DE ZUURSTOFCONCENTRATORDOOR EEN ERKENDE TECHNICUS TE LATENNAZIEN. DE ZUURSTOFCONCENTRATOR IS NIET
AANGEWEZEN VOOR GEBRUIKERS VAN WIE DE
GEZONDHEID NADELIG ZOU WORDEN BEÏNVLOED INGEVAL VAN EEN TIJDELIJKE ONDERBREKING.

Veiligheidsrichtlijnen
WAARSCHUWING: ER MOGEN GEENWIJZIGINGEN WORDEN AANGEBRACHTAAN DEZE APPARATUUR.
WAARSCHUWING: DIT APPARAAT LEVERTHOOGZUIVERE ZUURSTOF DIE EEN SNELLEVERBRANDING BEVORDERT. ZORG ERVOORDAT MEN NIET ROOKT EN DAT ER ZICH GEEN
OPEN VLAMMEN BEVINDEN IN DEZELFDERUIMTE ALS: (1) DIT APPARAAT OF (2) ALLEZUURSTOFBEVATTENDE ACCESSOIRES. NIETNALEVING VAN DEZE WAARSCHUWING KAN LEIDENTOT ERNSTIGE BRAND, MATERIËLE SCHADE EN/OF
FYSIEK LETSEL OF DE DOOD.

WAARSCHUWING: VOOR ELKE GEBRUIKER DIENENINDIVIDUELE INSTELLINGEN VOOR PULSDOSERINGTE WORDEN BEPAALD OVEREENKOMSTIG DE
BEHOEFTE IN RUST, IN BEWEGING EN TIJDENSREIZEN.

WAARSCHUWING: GEBRUIK HET APPARAAT NIET IN
EEN BEPERKTE OF BESLOTEN RUIMTE (BIJV. EEN
KLEINE KOFFER OF HANDTAS) MET ONVOLDOENDE
VENTILATIE. DIT KAN OVERVERHITTING VAN DEZUURSTOFCONCENTRATOR VEROORZAKEN EN DE
GOEDE WERKING ERVAN BEÏNVLOEDEN.

WAARSCHUWING: SOMMIGE INSPANNINGEN VAN
DE LUCHTWEGEN VAN DE PATIËNT KUNNEN DE
CONSERVINGSFUNCTIE NIET AANSTUREN.

WAARSCHUWING: DE CONCENTRATOR MOETWORDEN GEPLAATST OP EEN LOCATIE ZONDER
ROOK, VERVUILENDE STOFFEN OF DAMPEN.

WAARSCHUWING: DE FREESTYLE COMFORT IS NIETBEDOELD VOOR GEBRUIK BIJ EEN PATIËNT DIE EEN
TRACHEOTOMIE HEEFT ONDERGAAN.

WAARSCHUWING: IN HET GEVAL ER EEN ERNSTIGINCIDENT MET DIT APPARAAT OPTREEDT, MOET DEGEBRUIKER HET INCIDENT ONMIDDELLIJK MELDENAAN DE AANBIEDER EN / OF DE FABRIKANT. EEN
ERNSTIG INCIDENT WORDT GEDEFINIEERD ALS
EEN LETSEL, OVERLIJDEN OF MOGELIJK LETSEL/ OVERLIJDEN VEROORZAKEN ALS HET INCIDENTZICH OPNIEUW VOORDOET. DE GEBRUIKER KANHET INCIDENT OOK MELDEN AAN DE BEVOEGDEAUTORITEIT IN HET LAND WAAR HET INCIDENT
PLAATSVOND.

WAARSCHUWING: ZWANGERE OF BORSTVOEDENDEVROUWEN MOGEN GEEN ACCESSOIRES GEBRUIKENDIE WORDEN AANBEVOLEN IN DEZE HANDLEIDING,ZE KUNNEN FTALATEN BEVATTEN.

192 - DUTPN MN232-C4 K | Gebruikershandleiding

FreeStyle Comfort
WAARSCHUWING: BEPAALDEZUURSTOFTOEDIENINGSACCESSOIRESDIE NIET ZIJN GENOEMD VOORGEBRUIK IN COMBINATIE MET DEZEZUURSTOFCONCENTRATOR, KUNNEN DE
WERKING VAN HET APPARAAT NADELIG
BEÏNVLOEDEN. DE AANBEVOLEN
ACCESSOIRES STAAN VERMELD IN DEZE
HANDLEIDING.
WAARSCHUWING: ALS DEZUURSTOFCONCENTRATOR GEVALLEN,BESCHADIGD OF BLOOTGESTELD IS AAN WATER,NEEM DAN CONTACT OP MET UW LEVERANCIER VAN
APPARATUUR VOOR CONTROLE EN EVENTUEELHERSTEL VAN HET APPARAAT. GEBRUIK DEZUURSTOFCONCENTRATOR NIET ALS HET SNOER
OF DE STEKKER IS BESCHADIGD.
WAARSCHUWING: VOORKOM SCHADE OF LETSELTIJDENS VERVOER IN EEN VOERTUIG DOOR DEZUURSTOFCONCENTRATOR STEVIG TE BEVESTIGEN
MET EEN GORDEL OF ANDERSZINS.
WAARSCHUWING: DE REGELGEVING VAN HETAMERIKAANSE DEPARTMENT OF TRANSPORTATION(DOT - MINISTERIE VAN VERVOER) EN DE VERENIGDE
NATIES (VN) EISEN DAT DE ACCU UIT HET APPARAAT
WORDT VERWIJDERD TIJDENS INTERNATIONALE
VLUCHTEN ALS DE ZUURSTOFCONCENTRATORWORDT INGECHECKT ALS BAGAGE. WANNEER DEZUURSTOFCONCENTRATOR WORDT VERVOERD,
MOET DE ACCU OOK VERWIJDERD EN CORRECTVERPAKT WORDEN.
WAARSCHUWING: LEES DEZE HANDLEIDING EN
GA NA OF U DE INHOUD BEGRIJPT VOORDATU DE APPARATUUR IN GEBRUIK NEEMT. ALS U
DE WAARSCHUWINGEN EN INSTRUCTIES NIETBEGRIJPT, DIENT U CONTACT OP TE NEMEN MET DE
LEVERANCIER VAN UW APPARATUUR ALVORENSU DEZE IN GEBRUIK NEEMT; DIT OM LETSEL OFSCHADE TE VERMIJDEN.
WAARSCHUWING: VRAAG ONMIDDELLIJK MEDISCHEHULP ALS U ONGEMAK OF EEN MEDISCHENOODTOESTAND ERVAART.
WAARSCHUWING: GEBRUIK DEZUURSTOFCONCENTRATOR NIET
BIJ TEMPERATUREN DIE BUITEN DE
BEDRIJFSSPECIFICATIES LIGGEN, DIT KAN HETVERMOGEN VAN HET APPARAAT BEÏNVLOEDEN OM
DE GESPECIFICEERDE ZUURSTOFCONCENTRATIE TE
BEREIKEN. RAADPLEEG HET DEEL 'SPECIFICATIES'IN DEZE HANDLEIDING VOOR MEER INFORMATIE
OVER HET TEMPERATUURBEREIK.
WAARSCHUWING: GEBRUIK GEEN OLIE, SMEER,
PRODUCTEN OP BASIS VAN PETROLEUM OF ANDEREONTVLAMBARE PRODUCTEN IN COMBINATIEMET ZUURSTOFHOUDENDE ACCESSOIRES OF DEZUURSTOFCONCENTRATOR. GEBRUIK ALLEEN
LOTIONS OF ZALVEN OP BASIS VAN WATER DIECOMPATIBEL ZIJN MET ZUURSTOF. ZUURSTOF
VERSNELT DE VERBRANDING VAN ONTVLAMBARE
STOFFEN.

WAARSCHUWING: BLOKKEER NOOIT DE OPENINGEN
VAN DE ROOSTERS VOOR LUCHTINLAAT/-UITLAAT.
STEEK GEEN VOORWERPEN IN DE OPENINGEN VAN
HET APPARAAT EN VOORKOM DAT ER VOORWERPENIN VALLEN. DIT KAN OVERVERHITTING VAN DEZUURSTOFCONCENTRATOR VEROORZAKEN EN DE
GOEDE WERKING ERVAN BEÏNVLOEDEN.
WAARSCHUWING: DOOR INCORRECT GEBRUIKVAN DE ACCU KAN DEZE VERHITTEN, VLAM VATTENEN ERNSTIG LETSEL VEROORZAKEN. DOORBOOR
DE ACCU NIET, KLOP ER NIET OP, LAAT HEM NIET
VALLEN, STAP ER NIET OP EN STEL DE ACCU NIET
BLOOT AAN STERKE STOTEN OF SCHOKKEN.
HET GEBRUIK VAN EEN BESCHADIGDE ACCU KANLEIDEN TOT PERSOONLIJK LETSEL.
WAARSCHUWING: VEROORZAAK GEEN
KORTSLUITING VAN DE METALENCONTACTKLEMMEN VAN DE ACCU MET METALENVOORWERPEN, ZOALS SLEUTELS OF MUNTEN.
DIT KAN VONKEN OF OVERMATIGE HITTEVEROORZAKEN.
WAARSCHUWING: ALS U DE DRAAGBAREZUURSTOFCONCENTRATOR BUITEN GEBRUIKTMET WISSELSTROOMVOEDING, MAG DE VOEDINGALLEEN AANGESLOTEN WORDEN OP EEN
AARDLEKSCHAKELAAR.
WAARSCHUWING: DE ZUURSTOFCONCENTRATORMAG NIET WORDEN GEBRUIKT NAAST OFGESTAPELD OP ANDERE APPARATUUR.
ALS AANPALEND OF GESTAPELD GEBRUIKONVERMIJDELIJK IS, MOET WORDENGECONTROLEERD OF HET APPARAAT NORMAALWERKT.
WAARSCHUWING: GEBRUIK VAN NIETGESPECIFICEERDE KABELS EN ADAPTERS, METUITZONDERING VAN KABELS EN ADAPTERS DIEWORDEN VERKOCHT DOOR DE FABRIKANT VAN
DE MEDISCHE ELEKTRISCHE APPARATUUR ALSVERVANGENDE ONDERDELEN VOOR INTERNECOMPONENTEN, KAN RESULTEREN IN EEN
VERHOOGDE UITSTOOT OF VERMINDERDEIMMUNITEIT VAN DE ZUURSTOFCONCENTRATOR.
WAARSCHUWING: GEBRUIK ALLEEN DE SPANNING
DIE WORDT VERMELD OP HET TYPEPLAATJE VAN
HET APPARAAT.
WAARSCHUWING: GEBRUIK GEEN
VERLENGSNOEREN IN COMBINATIE METDIT APPARAAT EN BEPERK HET AANTALAPPARATEN DAT IS AANGESLOTEN OP
HETZELFDE STOPCONTACT. HET GEBRUIK VAN
VERLENGSNOEREN KAN DE PRESTATIES VAN HETAPPARAAT NEGATIEF BEÏNVLOEDEN. INDIEN ER
TOCH TE VEEL APPARATEN OP EEN STOPCONTACTWORDEN AANGESLOTEN, DAN BESTAAT HET RISICOOP OVERBELASTING WAARDOOR DE ZEKERINGEN
KUNNEN SPRINGEN. ALS DE ZEKERINGEN
NIET GOED WERKEN, KAN DIT ZELFS BRAND
VEROORZAKEN.
PN MN232-C4 K | Gebruikershandleidin

DUT - 193

FreeStyle Comfort
LET OP: raadpleeg deprobleemoplossingstabel in deze
handleiding in geval van een alarm of als uvaststelt dat de zuurstofconcentrator niet
naar behoren werkt. Neem contact op met
de leverancier van uw apparatuur als u er
niet in slaagt om het probleem op te lossen.

LET OP: om te voorkomen dat de garantie zijn
geldigheid verliest, dient u alle aanwijzingen van defabrikant in acht te nemen.
WAARSCHUWING: VOER GEENONDERHOUD UIT DAT GEEN DEELUITMAAKT VAN DE MOGELIJKEOPLOSSINGEN IN DEZE HANDLEIDING.
VERWIJDER GEEN DEKSELS.
ALLEEN DE LEVERANCIER VAN UWAPPARATUUR OF EEN BEVOEGDEONDERHOUDSTECHNICUS MAG DEKSELSVERWIJDEREN OF ONDERHOUDUITVOEREN AAN DIT APPARAAT.
WAARSCHUWING: GEBRUIK ALLEEN ACCESSOIRESDIE DOOR DE FABRIKANT ZIJN AANBEVOLEN.
GEBRUIK VAN ANDERE ACCESSOIRES KANGEVAARLIJK ZIJN, ERNSTIGE SCHADE TOEBRENGEN
AAN UW ZUURSTOFCONCENTRATOR EN DEGARANTIE ANNULEREN.
WAARSCHUWING: DE FABRIKANT RAADT AAN OM
EEN ALTERNATIEVE BRON VAN EXTRA ZUURSTOFTE GEBRUIKEN IN GEVAL VAN EEN STROOMUITVAL,
ALARMSITUATIES OF MECHANISCHE STORINGEN.
RAADPLEEG UW ARTS OF DE LEVERANCIER VAN
UW APPARATUUR VOOR HET VEREISTE SOORTRESERVESYSTEEM.
WAARSCHUWING: BLOKKEER NOOIT DE OPENINGEN
VAN DE LUCHTINLAAT/-UITLAAT. STEEK GEEN
VOORWERPEN IN DE OPENINGEN VAN HETAPPARAAT EN VOORKOM DAT ER VOORWERPENIN VALLEN. DIT KAN OVERVERHITTING VAN DEZUURSTOFCONCENTRATOR VEROORZAKEN EN DE
GOEDE WERKING ERVAN BEÏNVLOEDEN.
WAARSCHUWING: ROKEN TIJDENS GEBRUIKVAN ZUURSTOF IS DE VOORNAAMSTEOORZAAK VAN BRANDWONDEN EN DAARMEEGEPAARD GAAND OVERLIJDEN. VOLG DEZE
VEILIGHEIDSWAARSCHUWINGEN:WAARSCHUWING: STA NIET TOE DAT ER WORDTGEROOKT, DAT ER KAARSEN BRANDEN OF OPEN
VUUR IS IN DE KAMER WAAR HET APPARAAT OF
DE ZUURSTOFHOUDENDE ACCESSOIRES ZICH
BEVINDEN.
WAARSCHUWING: ROKEN TERWIJL U EENZUURSTOFCANULE DRAAGT, KAN LEIDEN TOT
BRANDWONDEN IN HET GELAAT OF ZELFS DE DOOD.
WAARSCHUWING: ALS U ROOKT, KUNNEN DE
VOLGENDE 3 STAPPEN UW LEVEN REDDEN:
SCHAKEL DE ZUURSTOFCONCENTRATOR UIT,VERWIJDER DE CANULE EN VERLAAT DE KAMERWAAR HET APPARAAT ZICH BEVINDT.
194 - DUTPN MN232-C4 K | Gebruikershandleiding

WAARSCHUWING: PLAATS DUIDELIJKE BORDEN MET'ROKEN VERBODEN - GEBRUIK VAN ZUURSTOF' IN
DE WONING OF ELDERS WAAR ZUURSTOF WORDTGEBRUIKT. GEBRUIKERS EN HUN ZORGVERLENERS
MOETEN INGELICHT WORDEN OVER DE GEVARENDIE ROKEN IN DE AANWEZIGHEID VAN OF BIJGEBRUIK VAN MEDISCHE ZUURSTOF INHOUDT.
WAARSCHUWING: GEBRUIK UWZUURSTOFCONCENTRATOR NIET IN DE
AANWEZIGHEID VAN ONTVLAMBARE GASSEN. DITKAN SNEL BRAND VEROORZAKEN EN LEIDEN TOT
MATERIËLE SCHADE, LICHAMELIJK LETSEL OFOVERLIJDEN.
WAARSCHUWING: ALS U DE CANULE AFNEEMTEN PLAATST OP KLEDING, BEDLINNEN, EEN SOFAOF KUSSENS ONTSTAAT ER EEN STEEKVLAM BIJ
BLOOTSTELLING AAN EEN SIGARET, WARMTEBRONOF VLAM.
WAARSCHUWING: PLAATS GEEN NEUSCANULE OP
OF ONDER KLEDING, BEDLINNEN OF KUSSENS VAN
EEN STOEL. ALS HET APPARAAT IS INGESCHAKELD
MAAR NIET WORDT GEBRUIKT, ZAL DE ZUURSTOF
HET MATERIAAL ONTVLAMBAAR MAKEN.
CONTROLEER OF DE ZUURSTOFCONCENTRATORIS UITGESCHAKELD WANNEER HIJ NIET WORDTGEBRUIKT.
WAARSCHUWING: DE ZUURSTOFCONCENTRATOR
MOET WORDEN UITGESCHAKELD EN VAN DE
EXTERNE VOEDING WORDEN LOSGEKOPPELDALVORENS DE ACCU TE VERWIJDEREN.
WAARSCHUWING: KLEINEONDERDELEN KUNNEN MOGELIJKVERSTIKKINGSGEVAAR OPLEVEREN.
WAARSCHUWING: VERMIJD CONTACT MET DEONTLUCHTINGSOPENING GEDURENDE LANGER DAN
10 SECONDEN, WANT DEZE KAN WARM ZIJN.
WAARSCHUWING: WIND OF STERKELUCHTSTROMEN KUNNEN EEN NEGATIEVEINVLOED HEBBEN OP DE JUISTE TOEDIENING VAN
ZUURSTOFTHERAPIE.
WAARSCHUWING: DIT PRODUCT KAN UBLOOTSTELLEN AAN CHEMISCHE STOFFEN ZOALSNIKKEL, WAARVAN DE STAAT CALIFORNIË WEETDAT HET KANKERVERWEKKEND IS. GA VOOR MEER
INFORMATIE NAAR WWW.P65WARNINGS.CA.GOV.
LET OP: koppel de wisselstroomvoeding
altijd los van het contact voor u dewisselstroomvoeding loskoppelt van dezuurstofconcentrator.
LET OP: plaats de zuurstofconcentrator nooit zodanig
dat het snoer moeilijk toegankelijk is.

FreeStyle Comfort
LET OP: plaats de zuurstoftoevoerslangen en
netsnoeren altijd zodanig dat struikelgevaar en het
gevaar van onopzettelijke wurging worden voorkomen.
LET OP: bij gebruik van de zuurstofconcentrator in
een auto, boot of met andere gelijkstroombronnen
met de gelijkstroomvoeding, moet het voertuig zijn
opgestart en draaien voordat u de zuurstofconcentrator
aansluit. Als deze instructies niet nageleefd worden, is
het mogelijk dat de voeding geen stroom levert aan dezuurstofconcentrator.
LET OP: als de auto waarin u de zuurstofconcentrator
gebruikt uitgeschakeld is, moet het apparaat worden
losgekoppeld en uit de auto worden verwijderd. Bergde zuurstofconcentrator niet op in een heel warme of
koude auto of andere vergelijkbare omgevingen met
een hoge of lage temperatuur.
Opmerking: draagbare en mobiele RF-communicatieapparatuur kan elektrische medische apparatuurbeïnvloeden.
LET OP: laat de zuurstofconcentrator ofde voedingsbron NIET aangesloten op het
voertuig als de ontsteking in de uit-standstaat. Dit kan de batterij van het voertuig
ontladen.
LET OP: als de zuurstofconcentrator buiten het normale
bedrijfstemperatuurbereik en omgevingsdrukbereik(hoogte) werkt of wordt opgeslagen, kan dit deprestaties beïnvloeden en de zelfstandige gebruiksduur
van de accu verminderen en/of de oplaadduur vande accu verlengen. Raadpleeg het deel Specificaties
in deze handleiding voor meer informatie over het
temperatuurbereik voor opslag en gebruik.
LET OP: voor zuurstofconcentrators met accu's:
Bewaar de accu op een koele en droge plaats om delevensduur ervan te verzekeren. Langdurige opberging
van de zuurstofconcentrator bij hoge temperaturenof met een volledig geladen of ontladen accu, kan delevensduur van de accu beperken. Probeer niet om deaccu te openen. De accu bevat geen onderdelen die
door de gebruiker kunnen worden onderhouden. Houdaccu's buiten het bereik van kinderen.
LET OP: GEBRUIK ALLEEN batterijen die afkomstig zijn
van de fabrikant. Neem contact op met de leverancier
van uw apparatuur of met de bevoegde lokale
autoriteiten voor de afdankingsrichtlijnen voor een
correcte afdanking van accu's.

Opmerking: controleer of de canule correct is geplaatst en
goed vastzit. Zo kan de zuurstofconcentrator correct deademhaling detecteren voor de toediening van zuurstof.
Tijdens het inademen, moet u de zuurstoftoediening via detakken van de neuscanule horen of voelen.
Opmerking: lithiumbatterijen dreigen permanent hun
capaciteit te verliezen als ze worden blootgesteld aan
zeer hete temperaturen en volledig opgeladen of volledig
ontladen zijn. Voor langdurige opslag is het aanbevolen
de batterijen 25 tot 50% op te laden en binnen een
temperatuurbereik van -20 °C - 20 °C (-4 °F - 68 °F) te
bewaren.
Opmerking: bepaalde zuurstoftoedieningsaccessoires dieniet zijn gespecificeerd voor gebruik in combinatie met dezezuurstofconcentrator, kunnen de werking ervan nadeligbeïnvloeden. De aanbevolen accessoires staan vermeld indeze handleiding.
Opmerking: gebruik de zuurstofconcentrator niet als het
luchtinlaatfilter niet is bevestigd. Als een tweede filter ismeegeleverd, plaatst u het vervangingsfilter voordat u het
vervuilde filter reinigt. Maak het vuile filter schoon in een
warme oplossing van zeep en water en laat het volledig
drogen vóór gebruik.
Opmerking: de fabrikant raadt niet aan om deze apparatuur
te steriliseren.
Opmerking: er is nooit risico op uitputting van zuurstof ineen ruimte waar u uw zuurstofconcentrator gebruikt.
WAARSCHUWING: DEZUURSTOFCONCENTRATOR MAG NIETWORDEN GEBRUIKT OF OP EEN EXTERNE(GELIJKSTROOM-/WISSELSTROOM-)VOEDING WORDEN AANGESLOTEN
ALVORENS DE ACCU IS GEPLAATST.
TRACHT HET APPARAAT NIET IN TESCHAKELEN OF OP EEN EXTERNEVOEDING AAN TE SLUITEN ZONDER DAT
DE ACCU WERD GEPLAATST.
VERWIJDER DE ACCU PAS WANNEER HETAPPARAAT IS UITGESCHAKELD EN VAN DE
VOEDING IS LOSGEKOPPELD. TRACHT DE ACCUNIET TE VERWIJDEREN ZOLANG HET APPARAATNOG IS INGESCHAKELD OF AAN DE VOEDING ISGEKOPPELD.
Opmerking: als de zuurstofconcentrator voor lange
tijd buiten de normale omgevingsomstandigheden uit
bovenstaande tabel werd opgeslagen, moet het apparaat
eerst terug de normale bedrijfstemperatuur bereiken voordat
het wordt ingeschakeld.

PN MN232-C4 K | Gebruikershandleidin

DUT - 195

FreeStyle Comfort

Specificaties

FreeStyle Comfort
Afmetingen (met enkele accu)

25,4 x 18,5 x 7,9 cm (10,0 x 7,3 x 3,1 in)

Afmetingen (met dubbele accu)

27,9 x 18,5 x 7,9 cm (11,0 x 7,3 x 3,1 in)Gewicht (met enkele accu)

2,3 kg (5 lbs)

Gewicht (met dubbele accu)

2,7 kg (6 lbs)

Nominaal geluidsniveau***

39,93 dB (A) ± 0,42 dB (A) @ 2

Debietinstellingen

Pulsinstelling 1, 2, 3, 4, 5O2-uitvoerinstelling

Instelling 1 - 210 ml/min
Instelling 2 - 420 ml/min
Instelling 3 - 630 ml/min
Instelling 4 - 840 ml/min
Instelling 5 - 1050 ml/min(bij 20 bpm)

Zuurstofconcentratie*
Zuurstofmonitor
Maximale uitvoerdruk
Standaard doseergevoeligheid

90% (+ 5,5% / - 3%)Ja< 30 PSI> -0,1 cm H2O (aanpasbaar)

Wisselstroom

100 - 240 VAC, 50 - 60 Hz, 120W Max

Gelijkstroom

11 – 18 VDC (10 A max)

Oplaadtijd accu

Enkele accu: 3,5 uur
Dubbele accu: 6,0 uur

Accuduur

Duur enkele accu: instelling 2 - max. 4 uur
Duur dubbele accu: Instelling 2 - max. 8 uur

Bedrijfstemperatuur**
Batterijtype
Batterijcapaciteit

5 °C tot 40 °C (41 °F tot 104 °F )Lithiumion
Enkele cel: 6700 mAh; Dubbele cel: 13400 mAh

Vervoers- en opslagtemperatuur (apparaat)-25 °C tot +70 °C (-13 °F tot +158 °F ) tot 3048 m (10000 ft)

Vervoers- en opslagtemp. (accu)-25 °C tot +70 °C (-13 °F tot +158 °F ) tot 3048 m (10000 ft)

Hoogte bij gebruik**-381 tot 3048 m (1250 tot 10.000 ft) (getest bij 700-1060 hPa)

Bedrijfsvochtigheid

15 - 95% relatieve vochtigheid (niet-condenserend)

Opslagvochtigheid

0 - 90% Niet-condenserend

Beperkte garantie

3 jaar voor apparaat
1 jaar voor accu en voedingen* Op basis van een luchtdruk van 101 kPa (14,7 psi) bij 21 °C (70 °F)** Het apparaat werkt in alle omgevingen, zolang de omgeving maar binnen de opgegeven temperatuur- en vochtigheidsbereiken is.
Bij gebruik buiten deze bedrijfsspecificaties dreigt het vermogen van de concentrator om aan de zuurstofconcentratiespecificaties tevoldoen te worden beperkt bij een hoger debiet. Het opwarmscherm kan blijven weergeven totdat de concentrator de doelconcentratie
of maximaal 15 minuten heeft bereikt. Blijf het apparaat gedurende deze tijd gebruiken.
*** Geluidsniveau gemeten per testmethode Nr. 14-1 10/2018 MDS-Hi.
De verwachte levensduur van dit apparaat bedraagt minimaal vijf jaar. De verwachte houdbaarheid van de zeefbodems is ongeveer één
jaar als de concentrator gedurende de gehele periode niet wordt gebruikt.

196 - DUTPN MN232-C4 K | Gebruikershandleiding

FreeStyle Comfort
Gebruiksaanwijzing
1. Plaats en installeer de FreeStyle Comfort op een
gepaste gebruikslocatie.
a. De concentrator moet voldoende geventileerdzijn, zodat de luchtinlaatfilters en luchtuitlaat niet
worden geblokkeerd.
b. De concentrator moet zodanig worden geplaatst
dat alle hoorbare en zichtbare controlelampjes en
alarmen snel kunnen worden gezien en gehoord.
c. Controleer of de luchtinlaatfilters zijn geplaatst
alvorens het apparaat te gebruiken. Als er een
filter ontbreekt op het apparaat moet u contact
opnemen met de leverancier van uw apparatuur.
2. Plaats een accu in de concentrator.
a. Lijn de accu uit met de onderste behuizing vanuit
het midden van de FreeStyle Comfort.

3. Steek de wissel- of gelijkstroomvoeding in devoedingspoort aan de voorkant, rechts onderaan op het
apparaat.

4. Sluit een neuscanule aan op dezuurstofuitlaataansluiting. Controleer de aansluiting
vóór gebruik.
OPMERKING: ander dan het aanbevolen gebruik van decanule(s) kan de prestaties negatief beïnvloeden.

b. Schuif de accu naar binnen totdat u een klik hoort
om de vergrendeling aan te geven.

Opmerking: aan leverancier van apparatuur: De volgende
accessoires voor zuurstoftoediening worden aanbevolen
voor gebruik met de FreeStyle Comfort:• Neuscanule met 2,1 m (7 voet) slang (maximale LPM: 6
LPM): CAIRE-onderdeelnummer 5408-SEQ• Aanvullende informatie over aanbevolen accessoires isonline beschikbaar op www.caireinc.com.

PN MN232-C4 K | Gebruikershandleidin

DUT - 197

FreeStyle Comfort
5. Schakel de FreeStyle Comfort in door op de aan-/uitknop te drukken en selecteer uw voorgeschrevendebietinstelling met behulp van de debietknoppen voor
verhogen (+) of verlagen (-).

Pulsvolume autoSAT (ml)*Instelling
Ademhalingsfrequentie

1

2

3

4

5

15

10,5

21,0

31,5

42,0

52,5

20

10,5

21,0

31,5

42,0

52,5

25

10,5

21,0

31,5

42,0

42,0

30

10,5

21,0

31,5

35,0

35,0

35

10,5

21,0

30,0

30,0

30,0

40

10,5

21,0

26,3

26,3

26,3

*Tolerantie is +/- 15%

Er klinkt een kort hoorbaar alarm om aan te geven dat
het apparaat is ingeschakeld.
Opmerking: als de FreeStyle Comfort voor het eerst wordt
ingeschakeld, is de standaard debietinstelling 2. Als deinstelling wordt gewijzigd, wordt het apparaat ingeschakeld
met de vorige debietinstelling.
6. Plaats de neuscanule op uw gezicht inovereenstemming met de bij de canule geleverde
fabrikantsinstructies en adem normaal. Telkens als u
inademt, dient u via de neuscanule zuurstof toegediend
te krijgen.
7. Om de debietinstelling te wijzigen, drukt u op dedebietknoppen voor verhogen (+) of verlagen (-) tot het
gewenste debiet. Het is normaal dat u een ander geluidhoort wanneer u de debietinstellingen aanpast.

Pulsvolume zonder autoSAT (ml)*Instelling
Ademhalingsfrequentie

1

2

3

4

5

15

14,0

28,0

42,0

56,0

70,0

20

10,5

21,0

31,5

42,0

52,5

25

8,4

16,8

25,2

33,6

42,0

30

7,0

14,0

21,0

28,0

35,0

35

6,0

12,0

18,0

24,0

30,0

40

5,3

10,5

15,8

21,0

26,3

8. Schakel de FreeStyle Comfort uit door op de aan-/uitknop te drukken.
Opmerking: het weergavescherm gaat naar de stand-bymodus en wordt na 20 seconden gedimd. Druk op een
willekeurige knop om het scherm uit de stand-by-modus
te halen.
9. Raadpleeg voor geavanceerde functies het menu
Navigatie-instructies

BIJZONDERE PRODUCTEIGENSCHAPPEN
autoSAT®
Met de gepatenteerde autoSAT-functie van CAIRE kan
de FreeStyle Comfort een consistent zuurstofvolume(bolus) per ademhaling toedienen, tot een maximale
ademfrequentie. Naarmate de ademhaling van degebruiker toeneemt tijdens activiteiten zoals wandelen,
past de autoSAT-functie de werking van het apparaat aanom een ingesteld bolusvolume te behouden.
UltraSense®
Deze gevoelige pulsdosistrigger detecteert dedrukverandering tijdens het ademhalen en geeft dezuurstofpuls af wanneer u inademt.
autoDOSE™
Als er een 'geen adem'-alarm optreedt, geeft deze functie
zuurstofpulsen af met 20 ademhalingen per minuut bij
de huidige debietinstelling van het apparaat totdat dealarmtoestand is verholpen.
198 - DUTPN MN232-C4 K | Gebruikershandleiding

FreeStyle Comfort
Voedingen
De FreeStyle Comfort kan op drie manieren worden
aangedreven:• Herlaadbare accu
• Wisselstroom• Gelijkstroom
In de volgende rubrieken vindt u informatie over dewerking van al deze systemen.
WAARSCHUWING: GEBRUIKSLECHTS DE VOEDINGEN DIE IN DEZEGEBRUIKERSHANDLEIDING WORDENOMSCHREVEN. HET GEBRUIK VAN
VOEDINGEN EN ACCU'S DIE NIET DOORCAIRE WORDEN VERSTREKT, DREIGTHET APPARAAT TE BESCHADIGEN EN DE
WERKING ERVAN TE HINDEREN.

DE ACCU VERWIJDEREN

Trek de accu-ontgrendelingsknop weg van de FreeStyle
Comfort. HOUD deze knop INGEDRUKT en schuif deaccu intussen uit de concentrator door hem ervan weg tetrekken.
WAARSCHUWING: DEZUURSTOFCONCENTRATOR MOETWORDEN UITGESCHAKELD EN VAN
DE EXTERNE VOEDING WORDENLOSGEKOPPELD ALVORENS DE ACCU TEVERWIJDEREN.

ACCUDe FreeStyle Comfort kan worden aangedreven door deherlaadbare accu die bij het apparaat wordt geleverd. Er
gelden geen gebruiksbeperkingen voor de concentrator
wanneer deze op de accu werkt. Alle debieten zijnbeschikbaar bij aandrijving via de accu.

DE ACCULADING CONTROLEREN
DE ACCU PLAATSEN

Lijn de accu uit met de onderste behuizing vanuit het
midden van de FreeStyle Comfort. Duw de accu naarbinnen totdat u een klik hoort om de vergrendeling aan
te geven.

Om de acculading te controleren wanneer deze inde FreeStyle Comfort is geplaatst, raadpleegt u het
accupictogram en de geschatte resterende tijd op de
weergave.
80–100%60–80%

40–60%

20–40%

20%

PN MN232-C4 K | Gebruikershandleidin

DUT - 199

FreeStyle Comfort
Opmerking: als de accu bezig is met opladen, wordt een
stekkerindicator weergegeven in plaats van het resterende
laadpercentage.
Om de acculading te controleren wanneer deze niet in de
FreeStyle Comfort is geplaatst, legt u de accu op zijn rug
en drukt u op de knop 'TEST'. De controlelampjes voorde acculading (25-100%) aan de rechterkant van de knop
'TEST' gaan branden om de acculading aan te geven.

ACCUDUUR IN WERKING

Wanneer de externe wisselstroom- ofgelijkstroomvoeding wordt losgekoppeld, schakelt deconcentrator automatisch over op het accuvermogen als
de accu is geplaatst en is opgeladen. In de onderstaande
tabel wordt de gebruikelijke werkingsduur van een
nieuwe accu getoond. De accuduur wordt niet beïnvloeddoor uw ademhalingsfrequentie.
Pulsinstelling

Duur van enkele(8-cellige) accu

Duur van dubbele(16-cellige) accu

1

tot 8 uur

tot 16 uur2

tot 4 uur

tot 8 uur3

tot 3 uur

tot 6 uur4

tot 2,25 uur

tot 4,5 uur

5

tot 2 uur

tot 4 uurGEBRUIKELIJKE OPLAADTIJD ACCU
EERSTE KEER OPLADEN

De nieuwe accu die bij uw FreeStyle Comfort wordt
geleverd, is slechts gedeeltelijk opgeladen wanneer deze
uit de fabriek komt. Vóór het eerste gebruik moet u deaccu volledig opladen met wisselstroom of gelijkstroom.

ALGEMENE LAADINFORMATIE

De FreeStyle Comfort zal de accu altijd opladen
wanneer deze in de concentrator is geplaatst en wanneer
de concentrator op een externe wisselstroom- ofgelijkstroomvoeding is aangesloten (behalve in een
vliegtuig).
De accu is volledig opgeladen wanneer 100% wordt
weergegeven.
Ook kan de accu met het optionele laadstation worden
opgeladen.

200 - DUTPN MN232-C4 K | Gebruikershandleiding

De gebruikelijke tijd om uw accu volledig op te laden
vanaf een volledig lege accu is 3,5 uur voor een enkele
accu en 6,0 uur voor een dubbele accu.
Opmerking: de accu begint mogelijk niet meteen met
opladen als de accu volledig leeg is.

LEVENSDUUR ACCUDe FreeStyle Comfort-accu's zijn ontworpen om
300 laad-/ontlaadcycli mee te gaan.

FreeStyle Comfort
Wisselstroom- en gelijkstroomvoeding
De FreeStyle Comfort wordt standaard geleverd met een
wisselstroomvoeding en een gelijkstroomvoeding.
De wisselstroomvoeding omvat de volgende zaken:
a. Wisselstroomvoedingsblok dat werkt tussen
100 - 240 V, 50 - 60 Hz.
b. Wisselstroomsnoer om aan te sluiten
op het voedingsblok en de bijbehorende
wisselstroomuitgang.

2. Sluit het wisselstroomsnoer aan op het
wisselstroomvoedingsblok.

De gelijkstroomvoeding bestaat uit een snoer dat
rechtstreeks op het apparaat en de gelijkstroombron kan
worden aangesloten.

WISSELSTROOM

Wisselstroom is bedoeld voor gebruik in huis of waar
er standaard wisselstroom beschikbaar is. De FreeStyle
Comfort functioneert volledig op wisselstroom.
De FreeStyle Comfort laadt de accu ook op als er
wisselstroom beschikbaar is (behalve in een vliegtuig).
De accu wordt opgeladen, ongeacht of de FreeStyle
Comfort is in- of uitgeschakeld.

3. Steek het andere uiteinde van het wisselstroomsnoer
in de muur of in een ander geschikt stopcontact.

WAARSCHUWING: SLUIT GEEN EXTERNEVOEDING AAN ZONDER DAT ER EEN ACCU
IN DE ZUURSTOFCONCENTRATOR ISGEPLAATST.
De FreeStyle Comfort aansluiten op wisselstroom:
1. Sluit het wisselstroomvoedingsblok aan op de ingang
van de stroomaansluiting aan de voorkant, rechts
onderaan de FreeStyle Comfort. Duw de voeding
stevig in de aansluiting zodat hij goed vastzit.

PN MN232-C4 K | Gebruikershandleidin

DUT - 201

FreeStyle Comfort
4. Schakel de FreeStyle Comfort in door op de aan-/uitknop te drukken en selecteer uw voorgeschrevendebietinstelling met behulp van de debietknoppen voor
verhogen (+) of verlagen (-).

Opmerking: als de accu moet worden opgeladen, heeft het
accupictogram op de weergave één oranje balk.

GELIJKSTROOM

Gelijkstroom is bedoeld voor gebruik op
voedingsaansluitingen van auto's, boten of andere
motorvoertuigen. De FreeStyle Comfort is volledig
functioneel voor gebruik op gelijkstroom. Ook laadt
de FreeStyle Comfort de accu zo nodig op zodra er
gelijkstroom beschikbaar is. De accu wordt opgeladen,
ongeacht of het apparaat is in- of uitgeschakeld of al danniet wordt gebruikt.
WAARSCHUWING: SLUIT GEEN EXTERNEVOEDING AAN ZONDER DAT ER EEN ACCU
IN DE ZUURSTOFCONCENTRATOR ISGEPLAATST.
WAARSCHUWING: GEBRUIK HET APPARAAT NIET
OP 12V-GELIJKSTROOMAANSLUITINGEN IN DE AUTOTERWIJL HET VOERTUIG IS UITGESCHAKELD.
1. Schakel uw motorvoertuig in.
2. Sluit de gelijkstroomvoeding aan op de voorkant
van de voedingspoort, rechts onderaan de FreeStyle
Comfort. Duw de voeding stevig in de aansluiting
zodat hij goed vastzit.

202 - DUTPN MN232-C4 K | Gebruikershandleiding

3. Steek het andere uiteinde van het gelijkstroomsnoer
in de voedingsaansluiting van de auto of in een andere
gelijkstroomaansluiting.

4. Schakel de FreeStyle Comfort in door op de aan-/uitknop te drukken en selecteer uw voorgeschrevendebietinstelling met behulp van de debietknoppen voor
verhogen (+) of verlagen (-).
Opmerking: als de accu moet worden opgeladen, heeft het
accupictogram op de weergave één oranje balk.

FreeStyle Comfort
Reiniging, zorg en routineonderhoud
ROUTINEONDERHOUDWAARSCHUWING: GEBRUIK DEZUURSTOFCONCENTRATOR NIET ALS HETLUCHTINLAATFILTER NIET IS BEVESTIGD.

Preventief onderhoud
Het apparaat is onderhoudsvrij. Filters in het apparaatzijn bedoeld om de levensduur van het apparaat teverlengen. Als het apparaat niet goed werkt, moet het
apparaat voor reparatie naar de leverancier worden
teruggestuurd. Onderhoud aan interne onderdelen binnen
in de behuizing van de FreeStyle Comfort moet door dezuurstofleverancier worden uitgevoerd.

ONDERHOUD ACCU

De gebruiker van het apparaat is verantwoordelijk voor
de onderstaande onderhoudsprocedures.
• Wekelijks: reiniging van de luchtinlaatfilters

Houd de accu buiten het bereik van kinderen.

WAARSCHUWING: PROBEER
DE PLASTIC DEKSELS VAN DEZUURSTOFCONCENTRATOR NIET TEOPENEN OF TE VERWIJDEREN. HET
APPARAAT MAG SLECHTS DOOR EEN
BEVOEGDE ONDERHOUDSTECHNICUS
WORDEN GEOPEND.

LUCHTINLAATFILTERS

Lucht komt de FreeStyle Comfort binnen via
luchtinlaatfilters aan de voorkant, onderkant van dezuurstofconcentrator. Deze filters voorkomen dat stof
en andere grote deeltjes uit de lucht in het apparaat
terechtkomen.
De luchtinlaatfilters moeten minstens eenmaal per
week worden gereinigd. De leverancier van uw
zuurstofapparatuur raadt u mogelijk aan het filter vaker
te reinigen, naargelang de gebruiksomstandigheden. Volgdeze stappen om de luchtinlaatfilters correct te reinigen:
1. Verwijder de twee filters aan de voorkant van de
FreeStyle Comfort door het bijbehorende lipje omhoog
te duwen en naar buiten te trekken.

Stel de accu niet bloot aan vloeistoffen.
Tracht de accu niet te demonteren of te openen.
Stel de accu niet bloot aan open vlammen.
Laat de accu niet vallen.

Als uw accu beschadigd lijkt, neem dan contact op met
de leverancier van uw zuurstofapparatuur vóór gebruik.
Plaats de accu niet in of in de buurt van een magnetron ofandere kookapparaten.

OPSLAG ACCU

De accu van de zuurstofconcentrator hoeft niet volledig
leeg te zijn alvorens deze weer op te laden. Het isaanbevolen de accu van de zuurstofconcentrator na elk
gebruik op te laden.
Stel de accu niet bloot aan overmatige hitte, zoals in
rechtstreeks zonlicht of in een auto. Hierdoor dreigen
de prestaties en levensduur van de accu erop achteruit
te gaan. Ook dreigt de accu hierdoor hitte af te geven,te roken, vuur te vatten of te ontploffen. De accu moetin een stabiele omgeving worden bewaard, met eenlage vochtigheidsgraad (relatieve vochtigheid vanminder dan 70%) en zonder corrosieve gassen. Om
roest te voorkomen, moet u omstandigheden waarincondensvorming kan optreden vermijden, zoals snelle
schommelingen van de omgevingstemperatuur.
Voor opslag gedurende een maand of langer wordt een
ontladen accu of een deels geladen accu tussen 50% en
75% (twee tot drie leds) aanbevolen. Haal de accu uit dezuurstofconcentrator zodra deze is opgeladen.

2. Was beide filters in een warme oplossing van zeep
en water.
3. Spoel de filters grondig en verwijder overtollig water
met een zachte, stofvrije doek. Controleer of de filtersvolledig droog zijn alvorens ze terug te plaatsen.
4. Plaats de schone, droge filters terug.

PN MN232-C4 K | Gebruikershandleidin

DUT - 203

FreeStyle Comfort
REINIGING

Schakel het apparaat UIT en koppel het los van wisselof gelijkstroomvoeding voordat u het schoonmaakt.
Spuit NIET rechtstreeks op de behuizing. Gebruik een
vochtige (geen natte) doek of spons. Besproei de doek ofspons met een milde huishoudreiniger om de behuizing
en voedingsbronnen schoon te maken. Ga te werk
overeenkomstig de aanbevelingen van de fabrikant van
het schoonmaakmiddel.
WAARSCHUWING: SCHAKEL HETAPPARAAT UIT EN TREK HET SNOERUIT HET STOPCONTACT ALVORENSHET APPARAAT TE REINIGEN OM
ONOPZETTELIJKE ELEKTRISCHESCHOKKEN EN BRANDGEVAAR TEVOORKOMEN. ALLEEN DE LEVERANCIER
VAN UW APPARATUUR OF EEN
BEVOEGDE SERVICEMONTEUR MOGEN
DE DEKSELS VERWIJDEREN OF HETAPPARAAT ONDERHOUDEN.
WAARSCHUWING: U MOET EROVER WAKEN DAT DEZUURSTOFCONCENTRATOR EN DE ACCU NIET NAT
WORDEN EN DAT ER GEEN VLOEISTOFFEN IN HETAPPARAAT TERECHTKOMEN. HIERDOOR DREIGTHET APPARAAT EEN STORING TE VERTONEN OF
UIT TE VALLEN. BOVENDIEN NEEMT HET RISICO OP
ELEKTRISCHE SCHOKKEN OF BRANDWONDEN TOE.
WAARSCHUWING: GEBRUIK GEEN VLOEISTOF
RECHTSTREEKS OP HET APPARAAT. HIERONDER
VINDT U EEN NIET-ALLESOMVATTENDE LIJSTVAN ONGEWENSTE CHEMISCHE STOFFEN:
ALCOHOL EN ALCOHOLHOUDENDE PRODUCTEN,GECONCENTREERDE CHLOORHOUDENDE
PRODUCTEN (ETHYLEENCHLORIDE) EN
OLIEHOUDENDE PRODUCTEN. DEZE MOGEN NIETWORDEN GEBRUIKT OM DE PLASTIC BEHUIZING VAN
DE CONCENTRATOR TE REINIGEN, AANGEZIEN DEZEHET APPARAAT KUNNEN BESCHADIGEN.
WAARSCHUWING: REINIG DE BEHUIZING,HET BEDIENINGSPANEEL EN HET NETSNOERUITSLUITEND MET EEN MILDE HUISHOUDREINIGERDIE U AANBRENGT MET EEN VOCHTIGE (GEEN
NATTE) DOEK OF SPONS. DROOG NADIEN HETVOLLEDIGE OPPERVLAK. ZORG DAT ER GEENVLOEISTOF IN DE CONCENTRATOR KOMT.
BESTEED BIJZONDERE AANDACHT AAN DEZUURSTOFUITGANG VOOR DE CANULEAANSLUITINGEN ZORG ERVOOR DAT DEZE ALTIJD VRIJ IS VAN
STOF, WATER EN DEELTJES.

204 - DUTPN MN232-C4 K | Gebruikershandleiding

WAARSCHUWING: GEBRUIK EEN VOCHTIGE(GEEN NATTE) DOEK OF SPONS OM DE ACCUTE REINIGEN. SPUIT ALLEREERST EEN MILDEHUISHOUDREINIGER OP DE DOEK OF SPONS ENREINIG DE ACCUBEHUIZING.
WAARSCHUWING: NIET REINIGEN MET EENGEAUTOMATISEERD WASSYSTEEM.

DE CANULE VERVANGEN

Volg altijd de instructies voor correct gebruik van defabrikant van de canule. Vervang de neuscanule ofzuurstofslang elke 3 tot 6 maanden (bij dagelijks gebruik,
3 maanden) volgens de aanbevelingen van de fabrikant,zoals aanbevolen door de fabrikant van de canule ofde leverancier van uw zuurstofapparatuur. Uw arts ofzuurstofleverancier zal u informatie geven over reiniging
en vervanging.
Extra vervangonderdelen zijn verkrijgbaar bij deleverancier van uw zuurstofapparatuur.

FreeStyle Comfort
Alarmsituaties
Prioriteit
nr.
1

Alarmbeschrijving
Systeemstoring
hoge accutemperatuur

Weergave

Hoorbaarsignaal1 pieptoon
elke 20 sec.

2

Systeemstoring
hoge
temperatuur

1 pieptoon
elke 20 sec.

3

Systeemstoring
hoge druk

1 pieptoon
elke 20 sec.

4

Geen
ademhaling
gedetecteerd

1 pieptoon
elke 30 sec.

5

Uitschakelingdoor bijna legeaccu

1 pieptoon
elke 30 sec.

6

Opwarmen

geen

Meer details/aanbevolen handeling
Het apparaat slaat alarm
als de accutemperatuur deveiligheidsuitschakelgrens nadert.
Als de uitschakelgrens is bereikt, wordt
het apparaat uitgeschakeld. Als de externe
voeding is aangesloten, verschijnt het
alarmscherm en is het hoorbare alarm
gedurende 2 minuten te horen.
Test met een accu op kamertemperatuur.
Neem contact op met de leverancier als het
probleem blijft aanhouden.
Het apparaat slaat alarm als de interne
temperatuur de uitschakelgrens nadert.
Als de uitschakelgrens is bereikt, wordt
het apparaat uitgeschakeld en verschijnt
het alarmscherm en is het hoorbare alarm
gedurende 2 minuten te horen.
Controleer of er geen openingen geblokkeerd
zijn. Neem contact op met de leverancier als
het probleem blijft aanhouden.
Als de uitschakelgrens is bereikt, wordt
het apparaat uitgeschakeld en verschijnt
het alarmscherm en is het hoorbare alarm
gedurende 2 minuten te horen.
Neem contact op met de leverancier als het
probleem blijft aanhouden.
Geen ademhaling gedetecteerd gedurende
15 seconden.
Geactiveerd na de eerste ademhaling ofhet opwarmscherm wordt uitgeschakeld,
afhankelijk van wat eerder komt.
Controleer of de canule knikken of lekken
vertoont. Verhoog de gevoeligheidsinstelling.
Neem contact op met de leverancier als het
probleem blijft aanhouden.
Het alarm is hoorbaar totdat de accu leeg isen het apparaat wordt uitgeschakeld.
Laad de accu op.

Het apparaat begint met automatisch doseren
totdat een ademhaling wordt gedetecteerdof het opwarmen eindigt, afhankelijk van wat
eerder komt.

PN MN232-C4 K | Gebruikershandleidin

DUT - 205

FreeStyle Comfort
Prioriteit
nr.
7

Alarmbeschrijving
Lageconcentratie

8

Weergave

Hoorbaarsignaal1 pieptoonelke
30 seconden,
gedurendemax.
5 minuten

Meer details

Systeemstoring
debiet (OCDklepstoring,obstructie vangasbaan)1 pieptoon
elke 30 sec.

Neem contact op met de leverancier als het
probleem blijft aanhouden.

9

Waarschuwing
batterij bijna leeg

1 pieptoon,
Laad de accu op.
geen herhaling

10

Ademhalingsfrequentie
overschreden

1 pieptoon
elke 60 sec.

11

Systeemstoring
ventilatorstoring

geen

Neem contact op met de leverancier als het
probleem blijft aanhouden.

12

Systeemstoringlage druk

geen

Neem contact op met de leverancier als het
probleem blijft aanhouden.

13

Systeemstoringlage temp.

geenLaat het apparaat opwarmen tot op
kamertemperatuur. Neem contact op met deleverancier als het probleem blijft aanhouden.

Info

Opstarten

1 infopieptoon

Het scherm verschijnt bij het inschakelen enu hoort één pieptoon om te bevestigen dat dezoemer functioneert

Neem contact op met de leverancier als het
probleem blijft aanhouden.

De ademhalingsfrequentie overschrijdt
40 bpm.
Als de dosis ten onrechte wordt getriggerd,
stel de gevoeligheidsinstelling dan lager in.
Neem contact op met de leverancier als het
probleem blijft aanhouden.

Het startscherm toont de huidige
firmwarerevisie in de rechteronderhoek.

206 - DUTPN MN232-C4 K | Gebruikershandleiding

FreeStyle Comfort
Prioriteit
nr.
Info

Alarmbeschrijving
Normaalbedrijfsschermin autoSATmodus (geengelijkstroom
aangesloten)

Info

Normaalbedrijfsscherm
met autoSAT
gedeactiveerd(geen
gelijkstroom
aangesloten)
Instellinggewijzigd

Geen

Info

Sluit de externe
voeding aan inde uit-stand

geen

Info

Sluit de externe
voeding aan inde bedrijfsstand

geen

Info

Koppel deexterne voeding
los in de uitstand

1 infopieptoon

Info

Koppel deexterne voeding
los in de
bedrijfsstand(met accu
geplaatst)

1 infopieptoon

Info

Weergave

Hoorbaarsignaalgeen

Meer details

1 infopieptoon

Het CAIRE-scherm verschijnt gedurende
2 sec., gevolgd door de accu- en
stekkerstatus gedurende 5 sec. voordat
het scherm wordt uitgeschakeld

Laat het CAIRE-scherm 2 seconden
verschijnen en schakel daarna het scherm uit.

PN MN232-C4 K | Gebruikershandleidin

DUT - 207

FreeStyle Comfort
Prioriteit
nr.
Info

Alarmbeschrij- Weergaveving
In de stand-bymodus: wanneer
op de omhoogof omlaag-knop
wordt gedrukt,
toont het schermde accu- enlaadstatus

Hoorbaarsignaalgeen

Info

Vliegtuigmodus
op accuvoeding

1 langepieptoon

Info

Vliegtuigmodus
bij aansluiting op
externe voeding

1 langepieptoon

208 - DUTPN MN232-C4 K | Gebruikershandleiding

Meer details
Het scherm verschijnt gedurende 5 sec. en
wordt vervolgens uitgeschakeld

FreeStyle Comfort
Menu Navigatie-instructies
U kunt het menu raadplegen in de stand-by-modus ofde bedrijfsmodus. Als er in de bedrijfsmodus een alarm
optreedt, wordt het menu automatisch afgesloten. Als
er 20 seconden geen knop wordt ingedrukt, wordt het
menu automatisch afgesloten. In stand-by-modus moetu mogelijk tweemaal de menuknop indrukken om het
apparaat uit de slaapstand te halen, waarna u het menu
kunt raadplegen.
Er zijn vier menuschermen:
1. Statusscherm: hier worden het
firmwarerevisienummer, de bedrijfstijd en de accuontladingscycli weergegeven

Vliegtuigmodus

2. Vliegtuigmodus (AAN/UIT)
3. Gevoeligheid: aanpasbaar (1-5)
4. autoSAT (AAN/UIT)

Gevoeligheid

Statusscherm ter illustratie

BEDIENING INSTELLINGENMENU
U kunt het Instellingenmenu raadplegen door in debedrijfs- of stand-by-modus op de knop Menu/Selecteren
te drukken.
Duw opautoSATom door de menu's te scrollen.

Statuspagina

PN MN232-C4 K | Gebruikershandleidin

DUT - 209

FreeStyle Comfort
STATUSPAGINA
De statuspagina van de Freestyle Comfort toont
firmware-revisie, draaiuren en het aantal batterijcycli op de aangesloten batterij.

Vliegtuigmodus aan bevestigd

Om de vliegtuigmodus uit te schakelen, herhaalt u debovenstaande stappen terwijl u 'UIT' selecteert.

Statuspagina

Bedrijfsuren worden weergegeven in hele uren.
Batterijontlading is het aantal cycli dat de aangesloten batterij is leeggemaakt en opgeladen.

VLIEGTUIGMODUS
Om de FreeStyle Comfort te vervoeren, wordt devliegtuigmodus uitgeschakeld.

Als u opdrukt wanneer op het scherm het symbool'Volgende' verschijnt, schakelt de menupagina naar devolgende schermoptie.
De vliegtuigmodus is bedoeld om te worden gebruikt inde bedrijfsmodus. De vliegtuigmodus wordt automatischuitgeschakeld wanneer het apparaat stand-by blijft op deaccuvoeding.
Opmerking: raadpleeg de rubriek 'De vliegtuigmodus
activeren' voor meer informatie over de vliegtuigmodus.

GEVOELIGHEIDSMODUS
De FreeStyle Comfort wordt vervoerd met
een standaardfabrieksinstelling van 2 voor degevoeligheidsmodus en dit hoeft niet te worden
aangepast in normale omstandigheden. De
gevoeligheidsinstelling kan worden ingesteld van 1 tot 5,
waarbij 1 het minst gevoelig en 5 het meest gevoelig is.
Hoofdscherm vliegtuigmodus

Om de modus te activeren, drukt u op
terwijl het
pictogram WIJZIGEN zichtbaar is. Het blauwe woord'AAN' verschijnt.

Gevoeligheidsmodusscherm bij instelling 2

Om de gevoeligheidsmodus aan te passen, drukt u op
terwijl het pictogram WIJZIGEN zichtbaar is. De
waarde verspringt naar het volgende blauwe nummer.
Vliegtuigmodus aan geselecteerd

Om vliegtuigmodus aan te bevestigen, drukt u op
terwijl het pictogram OPSLAAN zichtbaar is. Het zwarte
woord 'AAN' verschijnt.
Om de vliegtuigmodus te annuleren, drukt u op
terwijl het pictogram ANNULEREN zichtbaar is.
Gevoeligheidsmodusinstelling 3 geselecteerd
210 - DUTPN MN232-C4 K | Gebruikershandleiding

FreeStyle Comfort
Om de instelling 3 in de gevoeligheidsmodus
te bevestigen, drukt u op
terwijl het pictogram
OPSLAAN zichtbaar is. De waarde '3' wordt in het zwart
weergegeven.
Om de wijziging van de gevoeligheidsmodus teannuleren, drukt u op
terwijl het pictogram
ANNULEREN zichtbaar is.
autoSAT-modus uit geselecteerd

Om de uitschakeling van de autoSAT-modus tebevestigen, drukt u op
terwijl het pictogram
OPSLAAN zichtbaar is. Het zwarte woord 'UIT'
verschijnt.

Gevoeligheidsmodusinstelling 3 bevestigd

Gevoeligheidsinstelling 5 (max.) bevestigd

Als u opdrukt wanneer op het scherm het symbool'Volgende' verschijnt, schakelt de menupagina naar devolgende schermoptie.

AUTOSAT-MODUS

Om de inschakeling van de autoSAT-modus te annuleren,
drukt u op
terwijl het pictogram ANNULEREN
zichtbaar is.

autoSAT-modus uit bevestigd

Om de autoSAT-modus in te schakelen, herhaalt u debovenstaande stappen terwijl u 'AAN' selecteert.
Als u opdrukt wanneer op het scherm het symbool'Volgende' verschijnt, schakelt de menupagina naar devolgende schermoptie.

De FreeStyle Comfort wordt vervoerd met de autoSATmodus ingeschakeld.

Hoofdscherm autoSAT-modus

Om de modus uit te schakelen, drukt u op
terwijl het
pictogram WIJZIGEN zichtbaar is. Het blauwe woord'UIT' verschijnt.

PN MN232-C4 K | Gebruikershandleidin

DUT - 211

FreeStyle Comfort

Probleemoplossing
Probleem

Mogelijke oorzaak

OplossingDe FreeStyle Comfort toont een
alarmbericht en laat een nietcontinue pieptoon horen.

Raadpleeg de tabel met
alarmsituaties.

Raadpleeg de tabel met alarmsituaties.

De zuurstofconcentrator kan niet
worden aangezet.

De accu is ontladen.

Sluit het apparaat aan op een gelijkstroom- ofwisselstroombron.

Niet verbonden met een
stroombron.
Algemene storing.

Controleer of de uitwendige aansluitingen goed zijnbevestigd.
Voer een volledige opstartcyclus uit door de accu
te verwijderen en een eventuele externe wissel- of
gelijkstroomvoedingsbron los te koppelen. Plaats deaccu vervolgens opnieuw, waarna de opstartcyclus
voltooid is. Sluit indien gewenst de externe
voedingsbron opnieuw aan.
Neem contact op met de leverancier van uw apparaat
en schakel zo nodig over op een alternatieve
zuurstofbron.

Alle overige problemen.

Voer een volledige opstartcyclus uit door de accu
te verwijderen en een eventuele externe wissel- of
gelijkstroomvoedingsbron los te koppelen. Plaats deaccu vervolgens opnieuw, waarna de opstartcyclus
voltooid is. Sluit indien gewenst de externe
voedingsbron opnieuw aan.
Schakel over op een alternatieve zuurstofbronen neem contact op met de leverancier van uw
apparatuur.

Opmerking: bepaalde alarmen zorgen ervoor dat het apparaat wordt uitgeschakeld.
Opmerking: tijdens een volledige stroomuitval wordt het alarmsysteem opnieuw ingesteld (eerdere alarmen worden daarbij
gewist) en verschijnt een alarm alleen als er nog steeds sprake is van de alarmsituatie zodra de voeding is hersteld.

212 - DUTPN MN232-C4 K | Gebruikershandleiding

FreeStyle Comfort
Reisinformatie
De FreeStyle Comfort voldoet aan de FAA-richtlijnen
voor gebruik aan boord van commerciële vliegtuigen.
Vóór elke reis (vliegtuig, trein, cruiseschip, enz.) moet u:• De goede werking van de concentrator controleren.
• Minstens 48 uur voor vertrek contact opnemen met
de reismaatschappij om te laten weten dat u met een
FreeStyle Comfort zult reizen.
Het is aanbevolen om altijd extra accu's bij u te hebben
wanneer u reist. Bij vliegreizen moet de accu voldoende
opgeladen zijn om de tijd op de grond (inchecken,
veiligheidscontroles, enz.) en tussenlandingen te dekken.
De meeste luchtvaartmaatschappijen vereisen dat u overeen geschikt aantal volledig opgeladen accu's beschiktom het toestel van stroom te voorzien gedurende
minstens 150% van de verwachte maximale duur van dereis (vluchttijden, tijd op de grond voor en na de vlucht,
tijdens verbindingen en voor onverwachte vertragingen).

1. Schakel de POC in met behulp van ALLEEN
accuvoeding als er geen externe voedingsbron isaangesloten.

2. Schakel de vliegtuigmodus in met behulp van een vande volgende methoden:
a. Volg de instructies om de vliegtuigmodus in teschakelen in het menu Navigatie-instructies.

Opmerking: als u vliegt, is het raadzaam extra accu's in een
veilige draagtas mee te nemen.

FAA-REGELGEVING

Reserveaccu's moeten, om kortsluiting tevoorkomen, individueel worden beschermd door ze
in de oorspronkelijke verpakking te plaatsen of deaansluitingen op een andere manier te isoleren, bijv. door
blootliggende aansluitingen te bedekken met tape ofdoor elke accu in een aparte plastic zak of beschermhoes
te doen, en mogen alleen in de handbagage worden
vervoerd.

b. Houd de omhoog- en omlaag-knop 5 seconden
ingedrukt totdat u een lange pieptoon hoort en het
vliegtuigsymbool ziet.

DE VLIEGTUIGMODUS ACTIVEREN
Accuvoeding
De POC kan in een vliegtuig ALLEEN met behulp vande accuvoeding worden gebruikt volgens de normale
bedieningsprocedures als er geen externe voedingsbron
is aangesloten

3. Wanneer het vliegtuigsymbool verschijnt, sluit u de
POC aan.

Externe voeding
Als in een vliegtuig een elektrisch stopcontactbeschikbaar is dat werkt en de locatie daarvan geschikt
is voor gebruik als een POC-voedingsbron, kan dit
dienen als back-up voor POC-accu's. In dit geval moet
de volgende aanbevolen procedure met betrekking tot
de overgang van de accuvoeding naar de elektrischevoeding van het vliegtuig worden gevolgd:
Opmerking: vanwege vliegtuigbeperkingen kunnen deaccu's niet worden opgeladen in de vliegtuigmodus.

PN MN232-C4 K | Gebruikershandleidin

DUT - 213

FreeStyle Comfort

Accessoires

HET APPARAAT IN DE DRAAGTAS PLAATSEN

Met het oog op correcte prestaties en veiligheid mag u
alleen de opgesomde accessoires gebruiken die via deleverancier van uw zuurstofapparatuur door CAIRE
worden verstrekt. Het gebruik van accessoires die niet
op de onderstaande lijst staan, kan de prestaties en/of
veiligheid van de concentrator negatief beïnvloeden.

1. Plaats het apparaat tot onderin de tas en stem de filteren accuopeningen juist af.

Standaardaccessoires van de FreeStyle Comfort
Draagtas

MI396-1

RugzakMI459-1

Schouderriem

MI397-1

Wisselstroomvoeding (alle snoeren
inbegrepen)

PW036-1S

Gelijkstroomsnoer

CD041-1

Externe groffilter (rechts)

FI226-1

Externe groffilter (links)

FI227-1

2. Doe de rits onderaan de tas dicht.

Optionele accessoires beschikbaar
Extra 8-cellige accu

BT034-1S

16-cellige accu

BT035-1S

Acculaadstation

BT036-1

Ruime accessoiretas

MI372-2

Opmerking: er zijn mogelijk bijkomende opties beschikbaar
voor landspecifieke snoeren, daar waar dit hierboven
is aangegeven. Neem contact op met CAIRE of met deleverancier van uw zuurstofapparatuur als u alternatieve
opties nodig hebt voor een bestelling.

Instructies draagtas
Volg de onderstaande instructies om uw FreeStyle
Comfort veilig in de draagtas te plaatsen.

214 - DUTPN MN232-C4 K | Gebruikershandleiding

DE SCHOUDERRIEM BEVESTIGEN
1. De schouderriem kan direct aan het apparaat of aan
de draagtas worden bevestigd.
2. Maak de riem vast aan de ring op het apparaat ofde tas.

FreeStyle Comfort
Afvoer en hergebruik
Lever de FreeStyle Comfort, inclusief alle onderdelen,altijd in bij uw apparatuuraanbieder voor de juiste afvoer
ervan. U kunt ook contact opnemen met de lokale
autoriteiten voor inlichtingen over de correcte afvoer
van de batterij. De eenheid kan worden hergebruikt
tussen mensen, maar moet worden teruggestuurd naar
apparatuurverlener voor opwerking.

AEEA en RoHS
Dit symbool herinnert gebruikers eraan om
de apparatuur aan het einde van de levensduur naar een

recyclingbedrijf te brengen, conform de AEEA-richtlijn(afgedankte elektrische en elektronische apparatuur).
Onze producten voldoen aan de beperkingen van derichtlijn inzake gevaarlijke stoffen (RoHS). Ze bevatten
geen sporen van lood of andere gevaarlijke stoffen.
LET OP: neem contact op met deleverancier van uw apparatuur of met uw
gemeente voor afvalverwerkingsinstructies
om het apparaat op een correcte manier af
te voeren.

Materialen in rechtstreeks contact met de gebruiker
Behuizing concentrator..........................................................................................................................................PC/ABS
Bedieningspaneel concentrator......................................................................................Polyester EBG7 of gelijkwaardig
Hendel concentrator......................................................................................................................................Polycarbonaat
Zuurstofuitlaat, mondstuk.............................................................................................................................Polycarbonaat
Etiket apparaat............................................................................................................................................................Lexan
Netsnoeraansluitingen............................................................................................................Polycarbonaat/vinylchloride
Netsnoer........................................................................................................................................................... PVC, metaal
Voeding...................................................................................................................................Lexaan 940 (polycarbonaat)
Accu........................................................................................................................................................................PC/ABS
Etiketten accu, voeding..................................................................................................................................Polyesterfolie
Draagkoffer concentrator..............................................................................................Cordura, neopreen, nylon voering
Riem....................................................................................................... Polypropyleen, acetyl, vinyl, Cordura, neopreen

EMC-testen
Leidraad en Verklaring van de fabrikant - Elektromagnetische emissies
De FreeStyle Comfort is bedoeld voor gebruik in de elektromagnetische omgeving die hieronder wordt gespecificeerd. De klant
of de gebruiker van de FreeStyle Comfort dient te verzekeren dat ze in een dergelijke omgeving worden gebruikt.
Emissietest
RF-emissies
CISPR 11
RF-emissies
CISPR 11
Harmonische emissies
IEC 61000-3-2
Spanningsfluctuaties/
flikkeremissies
IEC 61000-3-3

Naleving

Elektromagnetische omgeving - Leidraad

Groep 1De FreeStyle Comfort gebruikt alleen RF-energie voor zijn interne
werking. De RF-emissies zijn daarom erg laag en het is onwaarschijnlijk
dat deze enige storing zouden veroorzaken in elektronische apparatuur
die zich dichtbij bevindt.

Klasse B
Klasse A
Conform

De FreeStyle Comfort is geschikt voor gebruik in alle instellingen,waaronder instellingen met een woonfunctie en instellingen die
rechtstreeks zijn aangesloten op een laagspanningsnetwerk dat
gebouwen met een woonfunctie van stroom voorziet.

PN MN232-C4 K | Gebruikershandleidin

DUT - 215

FreeStyle Comfort
Leidraad en Verklaring van de fabrikant - Elektromagnetische immuniteit

De FreeStyle Comfort is bedoeld voor gebruik in de elektromagnetische omgeving die hieronder wordt gespecificeerd. De klant
of de gebruiker van de FreeStyle Comfort dient te verzekeren dat ze in een dergelijke omgeving worden gebruikt.
Immuniteitstest
IEC 60601-testniveau
Nalevingsniveau
Elektromagnetische omgeving - Leidraad
Houten, betonnen of keramische tegelvloer
Elektromagnetische±8 kV contact±8 kV contact
vereist. Indien de vloerbedekking bestaat
omgeving - Leidraad
uit synthetisch materiaal, dient de relatieve± 15 kV luchtafvoer± 15 kV luchtafvoer
IEC 61000-4-2
luchtvochtigheid minimaal 30% te zijn.
±2 kV voor voedingskabels ±2 kV voor
Snelle elektrische
De kwaliteit van de netspanning moet die zijn
voedingskabelstransiënten/burst
van een typische professionele zorginstelling±1 kV voor ingangs-/
en thuiszorgomgevingen.
IEC 61000-4-4
uitgangsleidingen
N.v.t.
± 1 kV lijn(en) naar lijn(en)Stroomstoten
IEC 61000-4-5

Spanningsdalingen,korte
onderbrekingen en
spanningsfluctuaties
bij ingangslijnen van
de stroomvoorziening
IEC 61000-4-11± 2 kV
gemeenschappelijke
modus opwisselstroomlijnen± 1 kV differentieel op
wisselstroomlijnen± 2 kV
gemeenschappelijke
modus op I/O-buitenlijnen
0% UT gedurende0,5 cyclus (0°, 45°, 90°,135°, 180°, 225°, 270°,315°)
0% UT gedurende
1 cyclus (00)
70% UT (30% daling in UT)
gedurende 25/30 cycli (00)

± 1 kV lijn(en) naar lijn(en) De kwaliteit van de netspanning moet die zijn
van een typische professionele zorginstellingen thuiszorgomgevingen.

0% UT gedurende0,5 cyclus (0°, 45°, 90°,135°, 180°, 225°, 270°,315°)

De kwaliteit van de netspanning moet die zijn
van een typische professionele zorginstellingen thuiszorgomgevingen. Indien de
0% UT gedurende 1 cyclus gebruiker van de FreeStyle Comfort een(00)
constante werking vereist gedurendestroomonderbrekingen, wordt het aanbevolen
70% UT (30% daling in UT) de FreeStyle Comfort aan te sluiten op een
gedurende 25/30 cycli (00) niet-onderbreekbare stroomtoevoer of een
batterij.

0% UT gedurende 250/300 0% UT gedurende 250/300
cycli (00)cycli (00)
Netfrequentie
De magnetische velden van de netfrequentie(50/60 Hz) magnetisch 3 A / m
3A/mdienen die van een typische professionele
veld IEC 61000-4-8
zorginstelling en thuiszorgomgevingen te zijn.
OPMERKING UT is de spanning van de wisselstroomnetvoeding voorafgaand aan de toepassing van het testniveau.

OPMERKING In sommige gevallen is het mogelijk dat de FreeStyle Comfort opnieuw moet worden ingesteld na een aanzienlijke
elektrostatische ontlading en kan het apparaat een volledige opstartcyclus vereisen om de normale werking te kunnen hervatten.
Zie het deel Probleemoplossing voor meer informatie.

216 - DUTPN MN232-C4 K | Gebruikershandleiding

FreeStyle Comfort
Leidraad en Verklaring van de fabrikant - Elektromagnetische immuniteit

De FreeStyle Comfort is bedoeld voor gebruik in de elektromagnetische omgeving die hieronder wordt gespecificeerd. De klant
of de gebruiker van de FreeStyle Comfort dient te verzekeren dat ze in een dergelijke omgeving worden gebruikt.
Immuniteitstest
IEC 60601-testniveau
Nalevingsniveau
Elektromagnetische omgeving - Leidraad
Geleide RF
10 Vrms
10 Vrms
Draagbare en mobiele RFcommunicatieapparatuur mag niet dichterIEC 61000-4-6
150 kHz tot 80 MHz
bij onderdelen van de FreeStyle Comfort,
inclusief de kabels, worden gebruikt dan
de aanbevolen scheidingsafstand die
werd berekend a.d.h.v. de vergelijking die
Uitgestraalde RF
10 V/m
10 V/mtoepasselijk is op de frequentie van de zender.
IEC 61000-4-3
80 MHz tot 2700 MHz
Aanbevolen scheidingsafstandd = 1.,2 P

d = 1.,2 P = 80 MHz tot 800 MHz
d = 1.,2 P = 800 MHz tot 2700 MHz
waarbij P staat voor het maximale
uitgangsvermogen van de zender in watt(W) volgens de fabrikant van de zender end voor de aanbevolen scheidingsafstand inmeter (m).
Veldsterktes van stationaire RF-zenders,zoals bepaald door een overzicht van het
elektromagnetisch veld,a dienen lager
te zijn dan het nalevingsniveau in elk
frequentiebereik. b
Interferentie kan voorkomen in de nabijheid
van apparatuur waarop het volgende symbool
staat:OPMERKING 1 Bij 80 MHz en 800 MHz geldt het hogere frequentiebereik.
OPMERKING 2 Het kan zijn dat deze richtlijnen niet op alle situaties van toepassing zijn. Elektromagnetische propagatie wordtbeïnvloed door absorptie en reflectie van gebouwen, voorwerpen en mensen.
aVeldsterktes van stationaire zenders zoals basisstations voor (mobiele/draadloze) radiotelefoons en landmobiele radio's,
amateurradio-, AM- en FM-radio-uitzendingen en televisie-uitzendingen, kunnen niet nauwkeurig theoretisch worden
voorspeld. Om de elektromagnetische omgeving van stationaire RF-zenders te evalueren, kan men een elektromagnetischterreinonderzoek uitvoeren. Wanneer de gemeten veldsterkte op de plek waar de FreeStyle Comfort gebruikt wordt boven
het hierboven weergegeven RF-nalevingsniveau uitkomt, moet de FreeStyle Comfort in de gaten worden gehouden om tecontroleren of het apparaat normaal werkt. Als u constateert dat het apparaat niet normaal werkt, kan het nodig zijn om extra
maatregelen te treffen, zoals het anders richten of het verplaatsen van de FreeStyle Comfort.
bBinnen het frequentiebereik van 150 kHz tot 80 MHz dienen de veldsterktes minder dan 10 V/m te zijn.

PN MN232-C4 K | Gebruikershandleidin

DUT - 217

FreeStyle Comfort
Aanbevolen scheidingsafstand tussen draagbare en mobiele RF-communicatieapparatuur en
de FreeStyle Comfort

De FreeStyle Comfort is bedoeld voor gebruik in een elektromagnetische omgeving waarin uitgestraalde RF-storingen worden
beheerst. De klant of de gebruiker van de FreeStyle Comfort kan elektromagnetische storing helpen voorkomen door een
minimale scheidingsafstand tussen de draagbare en mobiele RF-communicatieapparatuur (zenders) en de FreeStyle Comfort in
acht te nemen zoals hieronder aanbevolen, overeenkomstig het maximale uitgangsvermogen van de communicatieapparatuur.
Nominaal maximaal
Scheidingsafstand volgens frequentie van de zenderuitgangsvermogen vanmde zender
150 kHz tot 80 MHz
80 MHz tot 800 MHz
800 MHz tot 2,5 GHzd  1.2 P

d  1.2 P

d  1.2 P

W
= ,
= ,
= ,
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Voor zenders met een ander maximaal uitgangsvermogen dan hierboven vermeld, kan de aanbevolen scheidingsafstand d inmeter (m) worden geschat met behulp van de vergelijking die van toepassing is op de frequentie van de zender, waarbij P staat
voor het maximale uitgangsvermogen van de zender in watt (W) volgens de fabrikant van de zender.
OPMERKING 1 Bij 80 MHz en 800 MHz geldt de scheidingsafstand voor het hogere frequentiebereik.
OPMERKING 2 Het kan zijn dat deze richtlijnen niet op alle situaties van toepassing zijn. Elektromagnetische propagatie wordtbeïnvloed door absorptie en reflectie van gebouwen, voorwerpen en mensen.

ClassificatieType bescherming tegen elektrische schokken:
Klasse II	De bescherming tegen elektrischeschokken wordt gewaarborgd door
dubbele isolatie.
Mate van bescherming tegen elektrische schokken:
Type BF Apparatuur die een bepaalde mate van		bescherming tegen elektrische schokken biedt op
het gebied van		 1) toegestane lekstroom;		2) betrouwbaarheid van aardaansluiting (indien
aanwezig).
		 Niet bestemd voor rechtstreekse harttoepassingen.
Onafhankelijke testen voor de norm voor medischeelektrische apparatuur:
Conform bevonden aan• IEC 60601-1 uitgave 3.1: Medische elektrische
apparatuur - Deel 1 Algemene veiligheidseisen.
• CAN/CSA C22.2 nr. 60601-1-8 uitgave 2.1,
Medische elektrische apparatuur - Deel 1:
Algemene veiligheidseisen
Bescherming tegen potentiële elektromagnetische ofandere interferentie tussen de apparatuur en andere
toestellen.
	• Conform bevonden aan EN 60601-1-2 /IEC 60601-1-2: 4e uitgave.
	• Conform bevonden aan RTCA/DO160 Deel 21
Categorie M.
	• CISPR 11 / EN 55011 Klasse B Groep 1,'Industriële, wetenschappelijke en medische
apparatuur'
218 - DUTPN MN232-C4 K | Gebruikershandleiding	• FCC Deel 15, Subdeel B - Klasse B Onopzettelijke
stralers
IP22 - Bescherming tegen het binnendringen van vingers
of soortgelijke voorwerpen met een diameter vanmeer dan 12,5 mm. Verticaal druipend water heeftgeen schadelijke gevolgen wanneer de behuizing
onder een hoek van 15° wordt gekanteld ten opzichte
van de normale positie ervan.
Toegestane methode voor reiniging en
infectiebeheersing: raadpleeg het deel
'Reiniging, zorg en routineonderhoud' van dezegebruikershandleiding van de FreeStyle.
Mate van veiligheid van de toepassing in deaanwezigheid van brandbare anesthesiegassen:	De apparatuur is niet geschikt voor een dergelijke
toepassing.
Bedrijfswijze: continubedrijf.

CAIRE en CAIRE Inc. zijn gedeponeerde handelsmerken van CAIRE
Inc. Bezoek onze onderstaande website voor een volledige lijst vanhandelsmerken. Handelsmerken: www.careinc.com/corporate/trademarks.
Copyright © 2021 CAIRE Inc. CAIRE Inc. behoudt zich het recht voorom te allen tijde zonder voorafgaande kennisgeving en zonder verdereverplichtingen of gevolgen producten uit het assortiment te halen en om
prijzen, materialen, uitrusting, kwaliteit, beschrijvingen, specificaties en/of
processen van producten te wijzigen. Alle hierin niet uitdrukkelijk genoemde
rechten zijn, waar van toepassing, voorbehouden.

FreeStyle Comfort
Brugerkontrolelementer og indikatorer for systemstatus
ISO 7000; grafiske symboler til brug på udstyr –indeks og synopsisRådets direktiv 93/42/EØF om medicinske
anordninger
Autoriseret repræsentant i EU
Hvis den produkt-unikke
enhedsidentifikator (UDI)-mærkning har CE #### -symbolet
på, overholder enheden kravene i
direktiv 93/42 / EØF vedrørende
medicinsk udstyr. CE #### symbolet
angiver det bemyndigede organ
nummer.

Læs brugervejledningen inden
betjening. Reg.nr. 1641
Opbevaring eller
driftstemperaturområde. Reg.nr. 0632

Opbevaring fugtighedsområde.
Reg.nr. 2620

IEC 60417
Klasse II-udstyr, dobbeltisoleret.
Reg.nr. 5172

Atmosfærisk trykbegrænsning.
Reg.nr. 2621
Producentens navn og adresse.
Reg.nr. 3082
Dato for producenten. Reg.nr. 2497

Afbryder. Reg.nr. 5009
ETL-certificering
Sikkerhedsagentur for
CAN/CSA C22.2 nr. 601.1 M90 for
elektromedicinsk udstyr.

Katalognummer. Reg.nr. 2493
Serienummer. Reg.nr. 2498

Interne symboler

Denne side op. Reg.nr. 0623

Må ikke anbringes i nærheden afbrændbare materialer, olie og fedt.

Skrøbelig – skal håndteres forsigtigt.
Reg.nr. 0621

Må ikke adskilles.

Må ikke udsættes for regn. Opbevares
tørt. Reg.nr. 0626

Øg hastigheden.

Stablingsgrænse efter nummer.
Reg.nr. 2403

Reducer hastigheden.

ISO 7010: Grafiske symboler – sikkerhedsfarverog -skiltning – registreret sikkerhedsskiltning
Brugervejledningen skal læses.
Reg.nr. M002
Må ikke udsættes for åben ild, brandeller gnister. Åben antændelseskildeog rygning er ikke tilladt.
Reg.nr. P003
Der må ikke ryges i nærheden af
enheden eller under drift.
Reg.nr. P002
Type BF-del (beskyttelsesgrad modelektrisk stød). Reg.nr. 5333

Knappen Menu.
RTCA/DO-160,
afsnit 21
I overensstemmelse med
kategori M

RTCA DO160, afsnit 21,
i overensstemmelse med kategori M.
FAA SFAR 106-krav
Klasse II-udstyr, dobbeltisoleret

FAA-godkendt – POC

Advarsel! Reg.nr. W001

PN MN232-C4 K | Brugermanual

DAN - 219

FreeStyle Comfort
21 CFR 801.15: Code of Federal Regulations Title 21
Ifølge føderal amerikansk lovgivningmå dette apparat kun sælges af enlæge eller efter ordination af en læge.
Europa-Parlamentets og Rådets
direktiv 2012/19/EU: om affald af elektrisk
og elektronisk udstyr (WEEE)
WEEE

Hvad er en iltkoncentrator?
Den luft, vi indånder, indeholder ca. 21 % ilt, 78 %kvælstof og 1 % andre luftarter. I FreeStyle® Comfort®
suges almindelig luft ind i maskinen igennem
luftindtagene. Derefter strømmer den igennem etsugende materiale, der kaldes en molekylesi. Dettemateriale adskiller ilten fra kvælstoffet og lader kun
ilten trænge igennem. Resultatet er en strøm af meget
ren ilt, der leveres til brugeren.
FreeStyle Comfort kombinerer avanceret
iltkoncentratorteknologi med iltbevarende teknologi ien bærbar enhed, der sikrer den bedst mulige mobilitet
og ambulante brug. Den avancerede pulserende
flowlevering registrerer hurtigt, når brugeren ånder
ind, og leverer en iltpuls i starten af hver indånding.
Pulserende flowlevering er ideelt til aktiviteter og
brug uden for hjemmet. I modsætning til et uafbrudt
flow bevarer den ilt og forlænger batteriets levetid.
FreeStyle Comfort er let nok til at kunne bæres meden skulderrem. Produktet kan køre på vekselstrøm(stikkontakt), jævnstrøm (i et køretøj) eller på batteri.

Anvendelsesformål
CAIRE FreeStyle Comfort Oxygen Concentrator er
beregnet til administration af supplerende ilt. Enheden
er ikke beregnet til livsstøtte og giver heller ikke nogenpatientovervågningsfunktioner.

Grunden til, at din læge har ordineret
ilt
Mange mennesker lider af en række forskellige hjerte-,lunge- og andre luftvejssygdomme. Et væsentligt
antal af disse personer kan have fordel af supplerende
iltbehandling i hjemmet, på rejser eller ved deltagelsei daglige aktiviteter uden for hjemmet.
Ilt er en luftart, der udgør 21 % af den luft, vi indånder,og kroppen er afhængig af en konstant iltforsyning forat kunne fungere ordentligt. Din læge har ordinereten iltstrøm eller en indstilling, der passer til netop din
tilstand.
220 -DANPN MN232-C4 K | Brugermanual

IEC 60601-1: Elektromedicinsk udstyr
Del 1 Generelle sikkerhedskrav og væsentlige
funktionskrav
Dryptæt
Dette produkt er muligvis omfattet af ét eller flereamerikanske eller internationale patenter. Gå til vores
websted nedenfor for at se en liste over gældende
patenter. Pat.: www.caireinc.com/corporate/patents/.

Selvom ilt ikke er et vanedannende lægemiddel,
kan uautoriseret iltbehandling være farlig. Du skalrådføre dig med en læge, inden du anvender denne
iltkoncentrator. Leverandøren, der leverer dit iltudstyr,vil vise dig, hvordan du skal indstille den ordinerede
flowhastighed.

Brugerprofil
Koncentratorer er beregnet til at levere supplerende ilt
til brugere, der oplever ubehag som følge af lidelser,
der nedsætter lungernes evne til at transportere
ilt fra luften til blodet. Bærbare iltkoncentratorer(POC’er) hverken indeholder eller lagrer ilt. De skal
ikke genopfyldes, og de kan genoplades, hvor som
helst du har adgang til jævnstrøm eller vekselstrøm(undtagen ombord på fly). Det gør brugeren relativt
selvforsynende med hensyn til hjemmebrug, ambulant
brug (både i og uden for hjemmet), mobilitet og
generel livsstil. Brug af iltkoncentratoren kræver
ordination fra lægen, og den er ikke beregnet som
respirator.
Selvom iltbehandling kan ordineres til brugere ialle aldre, er den typiske bruger ældre end 65 år og
lider af en række forskellige luftvejssygdomme, bl.a.
kronisk obstruktiv lungesygdom (KOL). Brugerne
har typisk gode kognitive evner og skal være istand til at kommunikere ubehag. Hvis brugeren
ikke er i stand til at kommunikere ubehag ellerlæse og forstå mærkaterne på iltkoncentratorenog brugervejledningen, anbefales brug kun underovervågning af en person, som kan dette. Hvisder opleves ubehag under brug af koncentratoren,
anbefales det, at brugerne kontakter hjemmeplejen.
Det anbefales også, at brugerne har ekstra ilttilgængelig (dvs. iltflaske) i tilfælde af strømsvigt
eller fejl i koncentratoren. Der kræves ingensærlige færdigheder eller brugerevner for atbruge koncentratoren.

Udpakning af din FreeStyle Comfort
Kontrollér, at alle dele, der er
anført, er i pakken. Kontaktomgående iltudstyrsleverandøren,hvis noget mangler.

Standardtilbehør• Bærbar iltkoncentrator• Bæretaske• Skulderrem• 8-cellet batteripakke• Vekselstrømforsyning• Vekselstrømskabel• Jævnstrømsforsyning• Brugermanual

FreeStyle Comfort
Valgfrit ekstraudstyr• Ekstra 8-cellet batteripakke• 16-cellet batteripakke• Ekstern oplader• TilbehørstaskeLær din FreeStyle Comfort at kendeSæt dig ind i nøglefunktionerne i FreeStyle Comfort og dens brugergrænseflade.

Set forfra
Remfastgørelse

Brugergrænseflade

IltudtagLuftudtag

Luftindtag/filtre
Luftindtag/filtre

Tap på
strømstik

Batteripakke
Udløserknap til
batteripakke
PN MN232-C4 K | Brugermanual

DAN - 221

FreeStyle Comfort
Set ovenfra
DisplayØg hastighedenTænd/sluk-knap

Knappen Menu

Reducerhastigheden

Display på batteristrøm
Flowindstilling

Batterivisning

Indikatoren autoSAT

Alarmdisplay
Resterendebatteriprocent

Indikator forlevering af bolus

Display på vekselstrøm/jævnstrøm

Flowindstilling
BatterivisningOmråde til
alarmvisning

222 -DANPN MN232-C4 K | Brugermanual

Opladningsindikator

FreeStyle Comfort

Kontraindikationer for anvendelse

Vigtigt!
Sikkerhedsinstruktionerne er defineret som
følger:ADVARSEL: VIGTIGESIKKERHEDSOPLYSNINGER IFORBINDELSE MED RISIKO FOR
ALVORLIG PERSONSKADE.
FORSIGTIG: Vigtige oplysninger om
forebyggelse af skader på FreeStyle
Comfort.
Bemærk: Oplysninger, der kræver særlig opmærksomhed.

Indikationer for anvendelse
ADVARSEL: DET ER MEGET VIGTIGT AT
VÆLGE DET ORDINEREDE ILTNIVEAU.
UNDGÅ AT ÆNDRE FLOWINDSTILLINGEN,
MEDMINDRE EN AUTORISERET KLINISKARBEJDENDE LÆGE HAR BEDT DIG OM
AT GØRE DET.
ADVARSEL: IFØLGE NATIONAL LOVGIVNING I USA
MÅ DENNE ENHED KUN SÆLGES ELLER UDLEJESEFTER HENVISNING FRA EN LÆGE ELLER ANDET
AUTORISERET SUNDHEDSPERSONALE.
ADVARSEL: DENNE ENHED MÅ IKKE ANVENDESSOM RESPIRATOR. GERIATRISKE, PÆDIATRISKE
ELLER ANDRE BRUGERE, SOM IKKE ER I STAND
TIL AT KOMMUNIKERE UBEHAG UNDER BRUG
AF DENNE ENHED, KAN KRÆVE YDERLIGEREOVERVÅGNING. BRUGERE MED HØRE- OG/ELLERSYNSNEDSÆTTELSE SKAL HAVE ASSISTANCE MEDOVERVÅGNING AF ALARMER.
ADVARSEL: INDSTILLINGER FOR
PULSDOSISTILSTAND (PULSE DOSE MODE) SKALFASTSÆTTES FOR HVER ENKELT BRUGER IHENHOLD TIL VEDKOMMENDES BEHOV UNDERHVILE, TRÆNING OG PÅ REJSER.
ADVARSEL: NOGLE AF PATIENTENSVEJRTRÆKNINGER UDLØSER MULIGVIS IKKE DEN
ILTBESPARENDE FUNKTION.
ADVARSEL: FREESTYLE COMFORT ER
IKKE BEREGNET TIL BRUG SAMMEN MED
TRAKEOTOMEREDE PATIENTER.
ADVARSEL: GRAVIDE OG AMMENDE KVINDER BØR
IKKE BRUGE TILBEHØR, DER ANBEFALES I DENNE
VEJLEDNING, DA DISSE KAN INDEHOLDE FTALATER.

ADVARSEL: UNDER VISSE FORHOLDKAN DET VÆRE FARLIGT AT ANVENDE
IKKE-ORDINERET ILT. ENHEDEN BØR KUN
ANVENDES EFTER ORDINATION FRA ENLÆGE.
ADVARSEL: MÅ IKKE ANVENDES I NÆRHEDEN AF
BRÆNDBARE ANÆSTESIMIDLER.
ADVARSEL: SOM MED ALLE ELEKTRISKE ENHEDER
KAN BRUGEREN OPLEVE PERIODER UDEN
DRIFT SOM FØLGE AF STRØMAFBRYDELSER
ELLER BEHOV FOR, AT ILTKONCENTRATORENSERVICERES AF EN KVALIFICERET TEKNIKER.
ILTKONCENTRATOREN ER IKKE VELEGNETTIL BRUGERE, DER VILLE FÅ ALVORLIGE
SUNDHEDSMÆSSIGE FØLGER AF EN SÅDAN
MIDLERTIDIG STRØMAFBRYDELSE.

Sikkerhedsretningslinjer
ADVARSEL: DET ER IKKE TILLADT AT
MODIFICERE UDSTYRET.
ADVARSEL: DENNE ENHED LEVERER EN HØJ
ILTKONCENTRATION, DER FREMMER HURTIGFORBRÆNDING. RYGNING ELLER BRUG AF ÅBENILD ER IKKE TILLADT I SAMME RUM SOM (1)DENNE ENHED ELLER (2) ILTBÆRENDE UDSTYR.
MANGLENDE OVERHOLDELSE AF DENNE ADVARSEL
KAN MEDFØRE BRAND, MATERIEL SKADE OG/ELLERFYSISKE SKADER ELLER DØDSFALD.
ADVARSEL: ANVEND IKKE APPARATET I ET
AFLUKKET ELLER BEGRÆNSET RUM (DVS. EN
LILLE TASKE ELLER TASKE), HVOR VENTILATIONEN
KAN VÆRE DÅRLIG. DET KAN MEDFØREOVEROPHEDNING AF ILTKONCENTRATOREN,HVILKET FORRINGER YDEEVNEN.
ADVARSEL: KONCENTRATOREN SKAL PLACERESPÅ GOD AFSTAND AF RØG, FORURENENDESTOFFER ELLER GASSER.
ADVARSEL: HVIS DER OPSTÅR EN ALVORLIGHÆNDELSE MED DENNE ENHED, SKAL BRUGERENSTRAKS RAPPORTERE HÆNDELSEN TIL
UDBYDEREN OG / ELLER PRODUCENTEN. EN
ALVORLIG HÆNDELSE DEFINERES SOM EN SKADE,DØD ELLER POTENTIALE TIL AT FORÅRSAGESKADE / DØD, HVIS DER ER EN GENTAGELSE AF
HÆNDELSEN. BRUGEREN KAN OGSÅ RAPPORTEREHÆNDELSEN TIL DEN KOMPETENTE MYNDIGHED I
DET LAND, HVOR HÆNDELSEN FANDT STED.

PN MN232-C4 K | Brugermanual

DAN - 223

FreeStyle Comfort
ADVARSEL: BRUGEN AF VISSE TYPERILTADMINISTRATIONSUDSTYR, DER IKKEER SPECIFICERET TIL BRUG MED DENNE
ILTKONCENTRATOR, KAN PÅVIRKEYDEEVNEN. ANBEFALET TILBEHØR ER
ANGIVET I NÆRVÆRENDE VEJLEDNING.

ADVARSEL: LUFTINDTAGET ELLERLUFTUDTAGSHULLET MÅ IKKE BLOKERES. UNDGÅAT TABE ELLER INDSÆTTE GENSTANDE I ENHEDENSÅBNINGER. DET KAN MEDFØRE OVEROPHEDNING
AF ILTKONCENTRATOREN, HVILKET FORRINGERYDEEVNEN.

ADVARSEL: HVIS ILTKONCENTRATOREN BLIVERTABT, BESKADIGET ELLER UDSÆTTES FOR VAND,
SKAL DU KONTAKTE HJEMMEPLEJEN FOR AT FÅ
DEN UNDERSØGT OG EVENTUELT REPARERET.
UNDGÅ AT BRUGE ILTKONCENTRATOREN, HVISLEDNINGEN ELLER STIKKET ER BESKADIGET.

ADVARSEL: FORKERT BRUG AF BATTERIET KAN FÅBATTERIET TIL AT BLIVE VARMT ELLER ANTÆNDE
OG KAN FORÅRSAGE ALVORLIG PERSONSKADE.
UNDGÅ AT PRIKKE HUL I, SLÅ PÅ, TRÆDE PÅ ELLERTABE BATTERIET ELLER PÅ ANDEN MÅDE UDSÆTTEBATTERIET FOR STØD ELLER SLAG. BRUG AF ET
BESKADIGET BATTERI KAN FORÅRSAGE ALVORLIGPERSONSKADE.

ADVARSEL: SØRG FOR AT SIKRE, FASTSPÆNDE OG
PÅ ANDEN MÅDE FASTGØRE ILTKONCENTRATORENUNDER KØRSEL FOR AT FORHINDRE
BESKADIGELSE ELLER ULYKKER.
ADVARSEL: BESTEMMELSER FRA DETAMERIKANSKE DEPARTMENT OF TRANSPORTATION(DOT) OG DE FORENEDE NATIONER (FN) KRÆVER,AT BATTERIET FJERNES FRA ENHEDEN I
FORBINDELSE MED INTERNATIONALE FLYVNINGER,
HVOR ILTKONCENTRATOREN ER CHECKETIND SOM BAGGAGE. VED TRANSPORT AF
ILTKONCENTRATOREN SKAL BATTERIET OGSÅFJERNES FRA ENHEDEN OG EMBALLERES SEPARAT.
ADVARSEL: UNDGÅ AT BRUGE UDSTYRET
UDEN FØRST AT HAVE LÆST OG FORSTÅET
DENNE VEJLEDNING. HVIS DU IKKE FORSTÅRADVARSLERNE OG INSTRUKTIONERNE, SKAL DU
KONTAKTE LEVERANDØREN AF UDSTYRET, FØR DU
FORSØGER AT BRUGE DET. I MODSAT FALD KAN
DET MEDFØRE SKADER.
ADVARSEL: HVIS DU FØLER DIG UTILPAS ELLER
BEFINDER DIG I EN MEDICINSK NØDSITUATION,SKAL DU STRAKS SØGE LÆGE.
ADVARSEL: HVIS ILTKONCENTRATOREN BENYTTESUDEN FOR DE ANGIVNE DRIFTSTEMPERATURER,
KAN DET BEGRÆNSE KONCENTRATORENSEVNE TIL AT OPFYLDE SPECIFIKATIONERNEFOR ILTKONCENTRATION. DU KAN SETEMPERATURGRÆNSERNE I AFSNITTET OM
SPECIFIKATIONER I DENNE VEJLEDNING.
ADVARSEL: UNDGÅ AT BENYTTE OLIE,
FEDTSTOF ELLER PETROLEUMSBASEREDE
ELLER ANDRE BRÆNDBARE PRODUKTERSAMMEN MED ILTBÆRENDE TILBEHØR ELLER
ILTKONCENTRATOREN. KUN VANDBASEREDEPRODUKTER, ILTKOMPATIBLE LOTIONER ELLERSALVER BØR ANVENDES. ILT FREMSKYNDERFORBRÆNDING AF BRÆNDBARE STOFFER.

224 -DANPN MN232-C4 K | Brugermanual

ADVARSEL: BATTERIETS METALKONTAKTER MÅ
IKKE KORTSLUTTES MED METALGENSTANDE,SOM F.EKS. NØGLER ELLER MØNTER. DETTE KANMEDFØRE, AT DER GENERERES GNISTER ELLEROVERDREVEN VARME.
ADVARSEL: HVIS DEN BÆRBARE
ILTKONCENTRATOR BENYTTES UDEN DØRESAMMEN MED JÆVNSTRØMSFORSYNINGEN, MÅSTRØMFORSYNINGEN KUN TILSLUTTES ET GFIUDTAG (GROUND FAULT INTERRUPTED).
ADVARSEL: ILTKONCENTRATOREN BØR IKKEBRUGES VED SIDEN AF ELLER STABLET MED ANDETUDSTYR. HVIS DET ER NØDVENDIGT AT ANVENDE
DEN PLACERET VED SIDEN AF ELLER STABLET,SKAL ENHEDEN OVERVÅGES FOR AT SIKRE
KORREKT DRIFT.
ADVARSEL: BRUG AF ANDRE KABLER OG
ADAPTERE END DEM, DER ER ANGIVET, MED
UNDTAGELSE AF KABLER OG ADAPTERE, SOM
PRODUCENTEN AF DEN ELEKTRISKE MEDICINSKEANORDNING SÆLGER SOM RESERVEDELE TIL
INTERNE KOMPONENTER, KAN MEDFØRE FORØGET
EMISSION AF REDUCERET IMMUNITET FRAILTKONCENTRATOREN.
ADVARSEL: BRUG KUN ELEKTRISK SPÆNDING SOM
ANGIVET PÅ ENHEDENS MÆRKAT.
ADVARSEL: BRUG IKKE FORLÆNGERLEDNINGER
TIL DENNE ENHED, OG TILSLUT IKKE FOR MANGESTIK I DEN SAMME STIKKONTAKT. BRUGEN
AF ​FORLÆNGERLEDNINGER KAN PÅVIRKEENHEDENS YDEEVNE. ALT FOR MANGE STIK I
DEN SAMME STIKKONTAKT KAN RESULTERE
I EN OVERBELASTNING PÅ EL-PANELET, SÅ
AFBRYDEREN/SIKRINGEN AKTIVERES ELLER GÅR I
BRAND, HVIS AFBRYDEREN ELLER SIKRINGEN IKKEFUNGERER.

FreeStyle Comfort
FORSIGTIG: I tilfælde af en alarm, ellerhvis det konstateres, at iltkoncentratoren
ikke fungerer korrekt, henvises der til
afsnittet Fejlfinding i denne vejledning.
Kontakt udstyrsleverandøren, hvis
problemet ikke kan løses.

ADVARSEL: DER SKAL VÆRE TYDELIG SKILTNING
MED ”RYGNING FORBUDT – ILTAPPARAT I BRUG”
I HJEM, HVOR DER ANVENDES ILTAPPARATER.
BRUGERE OG PLEJERE SKAL INFORMERES OM
FARERNE VED AT RYGE I NÆRHEDEN AF, ELLER
UNDER BRUG AF, MEDICINSK ORDINERET ILT.

FORSIGTIG: Følg alle producentens anvisninger for
ikke at ophæve garantien.

ADVARSEL: UNDGÅ AT BENYTTE
ILTKONCENTRATOREN I NÆRHEDEN AF
BRÆNDBARE GASSER. DET KAN MEDFØRE BRAND
OG MEDFØLGENDE RISIKO FOR TINGSSKADE,PERSONSKADE OG DØDSFALD.

ADVARSEL: UNDGÅ AT UDFØRE ANDEN
VEDLIGEHOLDELSE END ANGIVET I
DENNE VEJLEDNING. UNDGÅ AT FJERNEDÆKSLER ELLER UDFØRE SERVICEPÅ ENHEDEN. DET BØR KUN GØRES
AF DIN HJEMMEPLEJER ELLER ENKVALIFICERET SERVICETEKNIKER.
ADVARSEL: KUN TILBEHØR, DER ER ANBEFALET
AF PRODUCENTEN. ENHVER ANDEN BRUG KAN
VÆRE FARLIG, MEDFØRE ALVORLIGE SKADER PÅ
ILTKONCENTRATOREN OG OPHÆVE GARANTIENSDÆKNING.
ADVARSEL: PRODUCENTEN ANBEFALER, AT DU
HAR EN ALTERNATIV KILDE TIL SUPPLERENDE ILTI TILFÆLDE AF STRØMUDFALD, ALARMTILSTAND
ELLER MEKANISK FEJL. SPØRG DIN LÆGEELLER UDSTYRSLEVERANDØR, HVILKEN TYPERESERVESYSTEM DER ER PÅKRÆVET.
ADVARSEL: SØRG FOR, AT LUFTINDTAGET OG
LUFTDYSERNE IKKE BLIVER BLOKERET. UNDGÅ AT
TABE ELLER INDSÆTTE GENSTANDE I ENHEDENSÅBNINGER. DET KAN MEDFØRE OVEROPHEDNING
AF ILTKONCENTRATOREN, HVILKET FORRINGERYDEEVNEN.
ADVARSEL: RYGNING UNDER BRUG AF ILT ER
DEN STØRSTE ÅRSAG TIL BRANDULYKKER OG
RELATEREDE DØDSFALD. DU SKAL OVERHOLDEFØLGENDE SIKKERHEDSADVARSLER:ADVARSEL: UNDGÅ RYGNING, LEVENDE LYS ELLERÅBEN ILD I SAMME RUM SOM ENHEDEN ELLERILTBÆRENDE TILBEHØR.
ADVARSEL: RYGNING UNDER BRUG AFNÆSEKANYLE KAN UDLØSE FORBRÆNDING IANSIGTET OG RISIKO FOR DØDSFALD.
ADVARSEL: HVIS DU RYGER, KAN FØLGENDE
3 TRIN MULIGVIS REDDE DIT LIV: SLUKILTKONCENTRATOREN, TAG KANYLEN AF, OG
FORLAD DET LOKALE, HVOR ENHEDEN ER
PLACERET.

ADVARSEL: HVIS DU TAGER NÆSEKANYLEN AF
OG ANBRINGER DEN PÅ TØJ, SENGETØJ, MØBLERELLER ANDET POLSTRET MATERIALE, UDLØSES ENEKSPLOSIONSAGTIG BRAND, HVIS DEN KOMMER
I FORBINDELSE MED CIGARETTER, VARMEKILDER
ELLER FLAMMER.
ADVARSEL: UNDGÅ AT EFTERLADE NÆSEKANYLEN
PÅ ELLER UNDER TØJ, SENGETØJ ELLER PUDER.
HVIS ENHEDEN TÆNDES UDEN AT VÆRE I BRUG,KAN ILTEN GØRE MATERIALET BRÆNDBART. SØRGFOR, AT DER ER SLUKKET FOR STRØMMEN, NÅR
ILTKONCENTRATOREN IKKE ER I BRUG.
ADVARSEL: ILTKONCENTRATOREN SKAL VÆRESLUKKET OG KOBLET FRA DEN EKSTERNE
STRØMFORSYNING, INDEN BATTERIPAKKENTAGES UD.
ADVARSEL: SMÅ DELE KAN UDGØRE
KVÆLNINGSFARE.
ADVARSEL: UNDGÅ KONTAKT MED
UDLUFTNINGSHULLERNE I MERE END
10 SEKUNDER, DA DE KAN VÆRE VARME.
ADVARSEL: VIND OG KRAFTIG TRÆK KAN
HAVE NEGATIV INDFLYDELSE PÅ NØJAGTIGILTBEHANDLING.
ADVARSEL: DETTE PRODUKT KAN UDSÆTTE
DIG FOR KEMIKALIER, HERUNDER NIKKEL,
SOM I STATEN CALIFORNIEN ER KENDT FOR AT
FORÅRSAGE KRÆFT. FÅ FLERE OPLYSNINGER PÅ
WWW.P65WARNINGS.CA.GOV.
FORSIGTIG: Tag altidvekselstrømsforsyningen ud
af stikkontakten, før du kobler
vekselstrømsforsyningen fra
iltkoncentratoren.
FORSIGTIG: Iltkoncentratoren må ikke placeres, så det
er vanskeligt at komme til strømkablet.

PN MN232-C4 K | Brugermanual

DAN - 225

FreeStyle Comfort
FORSIGTIG: Anbring altid iltslanger og ledninger på
en sådan måde, at man ikke kan falde i dem eller blive
kvalt af dem.
FORSIGTIG: Hvis iltkoncentratoren benyttes ien bil, båd eller på andre jævnstrømkilder medjævnstrømforsyningen, skal du sørge for, at kilden
kører, før du tilslutter iltkoncentratoren. Hvis disse
instruktioner ikke overholdes, kan det medføre, atstrømforsyningen ikke kan tilføre iltkoncentratoren
strøm.
FORSIGTIG: Hvis den bil, hvori du bruger
iltkoncentratoren, slukkes, skal du frakoble og fjerne
enheden fra bilen. Undgå at opbevare iltkoncentratoren
i en meget varm eller kold bil eller i andre lignende
miljøer med høje eller lave temperaturer.
Bemærk: Bærbart og mobilt RF-kommunikationsudstyr
kan påvirke elektrisk medicinsk udstyr.
FORSIGTIG: UNDGÅ at ladeiltkoncentratoren eller strømforsyningenvære tilsluttet i bilen, hvis tændingen er
slukket. Det kan opbruge bilens batteri.
FORSIGTIG: Hvis iltkoncentratoren anvendes
eller opbevares uden for dens normale
driftstemperaturområde og omgivelsestrykområde
(højde), kan det påvirke ydeevnen og reducere
batteriets driftstid og/eller øge batteriets opladningstid.
Se afsnittet Specifikationer i denne manual for at få
oplysninger om temperaturgrænser for opbevaringog drift.
FORSIGTIG: Iltkoncentratorer med batterier:Opbevares et køligt og tørt sted for at sikre langbatterilevetid. Hvis du opbevarer iltkoncentratoren
over længere tid ved høje temperaturer eller med
et fuldt opladet eller et helt afladet batteri, kan det
forringe batteriets generelle holdbarhed. Forsøg ikke
at åbne batteriet. Der er ingen dele inde i batteriet,
som kan serviceres. Opbevar batterier uden for børnsrækkevidde.
FORSIGTIG: BRUG KUN batterier fra producenten.
Kontakt udstyrsleverandøren eller kommunen med
henblik på korrekt bortskaffelse af batterier.

226 -DANPN MN232-C4 K | BrugermanualBemærk: Sørg for, at kanylen er helt indsat og sidder fast.
Det sikrer, at iltkoncentratoren kan detektere inspiration
korrekt med henblik på iltlevering. Under inhalering skal
du kunne høre og føle iltflow på nasalkanylen.
Bemærk: Lithiumbatterier kan permanent miste kapacitet,
hvis de udsættes for kraftig varme, når de er enten
helt opladet eller helt afladet. Ved længere opbevaring
anbefales det at oplade batterierne 25-50 % og opbevaredem i et temperaturområde på -20 °C - 20 °C (-4 °F - 68°F).
Bemærk: Brug af iltadministrationstilbehør, der ikke
er specificeret til brug med denne iltkoncentrator,
kan påvirke ydeevnen. Anbefalet tilbehør er angiveti nærværende vejledning.
Bemærk: Undgå at bruge iltkoncentratoren, uden at
luftindtagsfilteret er isat. Hvis der er medleveret et ekstra
filter, skal reservefilteret isættes, før det snavsede filter
rengøres. Rengør det snavsede filter med varmt vand ogsæbe, og tør det grundigt før brug.
Bemærk: Producenten anbefaler ikke sterilisering
af udstyret.
Bemærk: Der er aldrig fare for at løbe tør for ilt i et lokale,når du bruger iltkoncentratoren.
ADVARSEL: ILTKONCENTRATOREN MÅ
IKKE ANVENDES ELLER TILSLUTTESEKSTERNE STRØMFORSYNINGER(VEKSELSTRØM/JÆVNSTRØM),HVIS DER IKKE ER ISAT ET BATTERI.
GØR IKKE FORSØG PÅ AT TÆNDE
ENHEDEN ELLER SLUTTE DEN TIL EN
EKSTERN STRØMFORSYNING, HVISBATTERIPAKKEN ER TAGET UD.
BATTERIPAKKEN MÅ KUN TAGES UD, NÅR
ENHEDEN ER SLUKKET OG KOBLET FRA DENEKSTERNE STRØMFORSYNING. GØR IKKE FORSØG
PÅ AT TAGE BATTERIPAKKEN UD, MENS ENHEDENER TÆNDT ELLER TILSLUTTET EN EKSTERNSTRØMFORSYNING.
Bemærk: Hvis iltkoncentratoren har været
opbevaret i længere tid uden for dens normale
driftstemperaturområde, som er anført i tabellen ovenfor,skal enheden først opnå normal driftstemperatur, før dentændes.

FreeStyle Comfort
Specifikationer
FreeStyle Comfort
Mål (med enkelt batteri)25,4 x 18,5 x 7,9 cm (10,0 x 7,3 x 3,1")

Mål (med dobbelt batteri)27,9 x 18,5 x 7,9 cm (11,0 x 7,3 x 3,1")

Vægt (med enkelt batteripakke)

2,3 kg (5 lbs)

Vægt (med dobbelt batteripakke)

2,7 kg (6 lbs)

Nominelt støjniveau***

39,93 dB (A) ± 0,42 dB (A) @ 2

Flowindstillinger

Pulseringsindstilling 1, 2, 3, 4, 5

O2-udgangsindstilling(ved 20 vejrtrækninger pr. minut)

Iltkoncentration*
Iltmonitor
Maks. udgangstryk
Standardindstilling for doseringsfølsomhed

Indstilling 1 – 210 ml/min
Indstilling 2 – 420 ml/min
Indstilling 3 – 630 ml/min
Indstilling 4 – 840 ml/min
Indstilling 5 – 1050 ml/min
90 % (+5,5 %/-3 %)Ja< 30 PSI> -0,1 cm H2O (justerbar)

Vekselstrøm

100-240 VAC, 50-60 Hz, 120W Max

Jævnstrøm

11-18 VDC (maks. 10 ampere)

Batterigenopladningstid

Enkelt batteri: 3,5 timer
Dobbelt batteri: 6,0 timer

Batteripakkens driftstid

Varighed af enkelt batteri: Indstilling 2 – op til 4 timer
Varighed af dobbelt batteri: Indstilling 2 – op til 8 timer

Driftstemperatur**
Batteritype
Batterikapacitet

5 °C til 40 °C (41 °F til 104 °F)
Litiumion
Enkeltcelle: 6700 mAh; Dobbelt celle: 13400 mAh

Transport- og opbevaringstemp. (enhed)-25 °C til +70 °C (-13 °F til +158 °F) op til 3048 m (10.000 ft)

Transport- og opbevaringstemp. (batteri)-25 °C til +70 °C (-13 °F til +158 °F) op til 3048 m (10.000 ft)

Driftshøjde**-381 til 3048 m (-1250 til 10.000 ft) (testet til 700-1060 hPa)

Driftsluftfugtighed

15-95 % relativ fugtighed (ikke-kondenserende)

Opbevaring fugtighed

0 - 90% ikke-kondenserendeBegrænset garanti

3 år på enheden
1 år på batteri og strømforsyninger* Baseret på et atmosfærisk tryk på 14,7 psi (101 kPa) ved 21 °C (70 °F)** Enheden fungerer i alle miljøer, så længe miljøet er med de specificerede temperatur- og fugtighedsområder. Brug uden fordisse driftsspecifikationer kan begrænse koncentratorens evne til at opfylde specifikationerne for iltkoncentration ved højere
flowhastigheder. Opvarmningsskærmbilledet vises muligvis, indtil koncentratoren har nået målkoncentrationen eller i op til 15
minutter. Bliv ved med at bruge enheden i denne periode.
*** Lydniveau målt pr. Testmetode Nr. 14-1 10/2018 MDS-Hej.
Den forventede levetid er minimum fem år. Molekylsiernes forventede holdbarhed er ca. ét år, hvis koncentratoren ikke anvendes
i hele perioden.

PN MN232-C4 K | Brugermanual

DAN - 227

FreeStyle Comfort

Brugsanvisning

1. Anbring og opstil FreeStyle Comfort et sted, der er
egnet til anvendelsen.
a. Koncentratoren skal være godt ventileret,således at luftindtagsfiltrene og luftudtaget ikke
blokeres.
b. Koncentratoren skal opstilles, så det er nemt at
høre og se alle hørbare og synlige indikatorer
og alarmer.
c. Sørg for, at luftindtagsfiltrene er monteret, indenenheden tages i brug. Hvis der mangler et filter
i enheden, skal du kontakte leverandøren af
udstyret.

3. Sæt veksel- eller jævnstrømsforsyningen istrømstikket nederst til højre foran på enheden.

2. Sæt en batteripakke i koncentratoren.
a. Ret batteriet ind efter bunden af huset fra midten
af FreeStyle Comfort.

4. Slut en næsekanyle til iltudtagsstudsen. Sørg for, atden er tilsluttet korrekt inden brug.

b. Før batteriet indad, indtil du hører et klik, sombekræfter, at det er låst fast.

228 -DANPN MN232-C4 K | BrugermanualBEMÆRK: Brug af andre kanyler end den anbefalede kanpåvirke ydeevnen.

Bemærk: Til udstyrsudbyder: Følgende tilbehør til
iltadministration anbefales til brug sammen med
FreeStyle Comfort:• Nasal kanyle med 7 fod (2,1 m) af rør (Maksimal LPM: 6
LPM): CAIRE Delnummer 5408-SEQ• Yderligere information om anbefalet tilbehør ertilgængelig online på www.caireinc.com.

FreeStyle Comfort
5. Tænd for FreeStyle Comfort ved at trykke påtænd/sluk-knappen, og vælg den foreskrevne
flowindstilling ved hjælp af knappen til øgning (+)
eller reduktion (-) af flow.

Indstilling for *autoSAT-pulsvolumen (ml)Respirationsfrekvens

1

2

3

4

5

15

10,5

21,0

31,5

42,0

52,5

20

10,5

21,0

31,5

42,0

52,5

25

10,5

21,0

31,5

42,0

42,0

30

10,5

21,0

31,5

35,0

35,0

35

10,5

21,0

30,0

30,0

30,0

40

10,5

21,0

26,3

26,3

26,3

*Tolerancen er +/- 15 %Der udsendes et kortvarigt lydsignal, som angiver, atenheden er tændt.
Bemærk: Når FreeStyle Comfort tændes første gang, er
standardflowindstillingen 2. Hvis indstillingen er blevetændret, tændes enheden ved den forrige flowindstilling.

Indstilling for ikke-autoSATpulsvolumen (ml)*Respirationsfrekvens

1

2

3

4

15

14,0

20

10,5

5

28,0

42,0

56,0

70,0

21,0

31,5

42,0

52,5

6. Sæt næsekanylen på ansigtet i henhold til
producentens instruktioner, som følger med kanylen,og træk vejret helt normalt. Du bør kunne mærke, at
et pust af ilt leveres igennem næsekanylen, hver gangdu ånder ind.

25

8,4

16,8

25,2

33,6

42,0

30

7,0

14,0

21,0

28,0

35,0

35

6,0

12,0

18,0

24,0

30,0

7. Hvis du vil ændre flowindstillingen, skal du trykke
på knappen til øgning (+) eller reduktion (-) for atindstille den ønskede flowhastighed. Det er normalt
at høre forskellige lyde, når flowindstillingerneændres.

40

5,3

10,5

15,8

21,0

26,3

8. Sluk for FreeStyle Comfort ved at trykke på tænd/sluk-knappen.
Bemærk: Displayet skifter til standbytilstand og dæmpes
efter 20 sekunder. Tryk på en vilkårlig knap for at vække
skærmen fra standbytilstand.
9. Der findes oplysninger om avancerede funktioner
i instruktionerne til menunavigation

SPECIELLE PRODUKTFUNKTIONER
autoSAT®
CAIREs proprietære autoSAT-funktion giver FreeStyle
Comfort mulighed for at levere et konstant volumen
af ilt (bolus) pr. Åndedræt, op til en maksimalåndedrætsfrekvens. Når brugerens åndedrætsfrekvensøges under aktiviteter som at gå, justerer autoSATfunktionen enhedsdriften for at opretholde en indstillet
bolusvolumen.
UltraSense®
Denne sensitive pulsdosisudløser registrerertrykændring i dit åndedræt, og frigør iltindtag pr.
hjerteslag ved indånding.
autoDOSE™
Skulle der forekomme en ”ingen vejrtrækning”-alarm,vil denne funktion puste ilt ind med 20 vejrtrækninger
pr. minut med enhedens aktuelle flowindstilling, indtil
alarmtilstanden er afviklet.
PN MN232-C4 K | Brugermanual

DAN - 229

FreeStyle Comfort

Strømforsyninger

FreeStyle Comfort kan forsynes med strøm på tre måder:• Genopladelig batteripakke• Vekselstrøm• Jævnstrøm
Oplysninger om anvendelsen af hver af disse
komponenter findes i de efterfølgende afsnit.
ADVARSEL: BRUG KUN DESTRØMFORSYNINGER, DER ER ANFØRT
I DENNE BRUGERVEJLEDNING.
BRUG AF STRØMFORSYNINGER OG
BATTERIPAKKER, DER IKKE ER LEVERETAF CAIRE, KAN FORÅRSAGE SKADE PÅ
ENHEDEN OG UKORREKT DRIFT.

UDTAGNING AF BATTERIETTræk batteriudløserknappen nedad og væk fra
FreeStyle Comfort. Mens denne knap HOLDES
NEDE, skal du føre batteriet ud ved at trække det væk
fra koncentratoren.
ADVARSEL: ILTKONCENTRATOREN SKAL
VÆRE SLUKKET OG KOBLET FRA DENEKSTERNE STRØMFORSYNING, INDENBATTERIPAKKEN TAGES UD.

BATTERIPAKKE

FreeStyle Comfort kan køre på den genopladelige
batteripakke, som følger med enheden. Der er ikke
nogen driftsmæssige begrænsninger af koncentratoren,når den kører på batteripakken. Alle flowhastigheder
kan anvendes, når enheden kører på batteri.

KONTROL AF BATTERIPAKKENSLADETILSTAND
ISÆTNING AF BATTERIET

Ret batteriet ind efter bunden af huset fra midten af
FreeStyle Comfort. Tryk batteriet indad, indtil du hører
et klik, som bekræfter, at det er låst fast.

Hvis du vil kontrollere batteriets ladetilstand, når det er
sat i FreeStyle Comfort, skal du se på batteriikonet og
den anslåede resterende tid på displayet.
80-100 %60-80 %

40-60 %

20-40 %

20 %

230 -DANPN MN232-C4 K | BrugermanualFreeStyle ComfortBemærk: Hvis batteriet er ved at blive ladet op, vises etstikikon i stedet for den resterende ladeprocent.
Hvis du vil kontrollere batteriets ladetilstand, når det
ikke er sat i Comfort, skal du lægge batteriet fladt ned
og trykke på knappen TEST. Batterimålerindikatorerne
til højre for knappen TEST tændes for at angive
batteripakkens ladetilstand (25-100 %).

BATTERIETS DRIFTSTID

Når den eksterne veksel- eller jævnstrømsforsyning
frakobles, skifter koncentratoren automatisk til
at køre på batteriet, hvis et opladet batteri er isat.
Nedenstående tabel viser de normale driftstider foren ny batteripakke. Batteripakkens driftstid forbliver
konstant, uafhængigt af hvor hurtigt du trækker vejret.
Pulseringsindstilling

Varighed af enkelt(8-cellet) batteri

Varighed af dobbelt(16-cellet) batteri

1

op til 8 timer

op til 16 timer

2

op til 4 timer

op til 8 timer

3

op til 3 timer

op til 6 timer

4

op til 2,25 timer

op til 4,5 timer

5

op til 2 timer

op til 4 timer

NORMAL BATTERIGENOPLADNINGSTID
DEN FØRSTE OPLADNING

Det nye batteri, som følger med din FreeStyle Comfort,
er kun delvist opladet, når det leveres fra fabrikken. Duskal lade batteripakken helt op med enten veksel- ellerjævnstrøm, inden du tager det i brug første gang.

GENERELLE OPLYSNINGER OM
OPLADNING

Den typiske genopladningstid for batteripakken,
hvis den er helt afladet, er 3,5 for et enkelt batteri og
6,0 timer for et dobbelt batteri.
Bemærk: Batteriet starter muligvis ikke genopladningen
med det samme, hvis batteripakken er helt afladet.

BATTERILEVETIDFreeStyle Comfort-batterierne er udviklet til at holde
til 300 opladnings-/afladningscyklusser.

FreeStyle Comfort genoplader altid batteripakken,
når denne er sat i koncentratoren, og koncentratoren
tilsluttes en ekstern veksel- eller jævnstrømsforsyning(undtagen ombord på et fly).
Batteriet er helt opladet, når det viser 100 %.
Batteripakken kan også oplades med den eksterne
oplader (ekstraudstyr).

PN MN232-C4 K | Brugermanual

DAN - 231

FreeStyle Comfort

Veksel- og jævnstrømsforsyning
FreeStyle Comfort leveres med vekselstrøms- ogjævnstrømsforsyning som standard.
Vekselstrømsforsyningen omfatter følgende:a. Vekselstrømsblokken, der fungerer mellem
100-240 volt, 50-60 Hz
b. Vekselstrømskablet, der forbinder
strømblokken med stikkontakten.

2. Slut vekselstrømskablet til vekselstrømsblokken.

Jævnstrømsforsyningen består af en ledning, dertilsluttes direkte enheden og jævnstrømskilden.

VEKSELSTRØM

Vekselstrøm bruges derhjemme eller andre steder, hvor
en almindelig stikkontakt står til rådighed. FreeStyle
Comfort er fuldt funktionsdygtig, når den kører påvekselstrøm. FreeStyle Comfort genoplader ogsåbatteripakken, hver gang den tilsluttes vekselstrøm(undtagen ombord på fly). Batteripakken genoplades,
uanset om FreeStyle Comfort er tændt eller slukket.
ADVARSEL: UNDGÅ AT TILSLUTTE ENEKSTERN STRØMFORSYNING, HVISDER IKKE ER SAT EN BATTERIPAKKE I
ILTKONCENTRATOREN.
Tilslutning af Comfort til vekselstrøm:
1. Sæt vekselstrømsblokken i strømindgangen nederst
til højre foran på FreeStyle Comfort. Sørg for at
skubbe den indad, indtil den sidder forsvarligt fast.

232 -DANPN MN232-C4 K | Brugermanual

3. Slut den modsatte ende af vekselstrømskablet
til en stikkontakt i væggen eller en anden egnet
stikkontakt.

FreeStyle Comfort
4. Tænd for FreeStyle Comfort ved at trykke påtænd/sluk-knappen, og vælg den foreskrevne
flowindstilling ved hjælp af knappen til øgning (+)
eller reduktion (-) af flow.

Bemærk: Hvis batteripakken skal genoplades, harbatteriikonet på displayet en enkelt orange bjælke.

JÆVNSTRØM

Jævnstrøm bruges i tilbehørsstik i biler, både eller
andre motorkøretøjer. FreeStyle Comfort er fuldt
funktionsdygtig til brug, når den kører på jævnstrøm.
FreeStyle Comfort genoplader også batteripakken, hvis
det er nødvendigt, hver gang den tilsluttes jævnstrøm.
Batteripakken genoplades, uanset om enheden ertændt, slukket eller i brug.

3. Slut den modsatte ende af jævnstrømskablet
til en bils tilbehørsstik eller til et andet egnet
jævnstrømsudtag.

4. Tænd for FreeStyle Comfort ved at trykke påtænd/sluk-knappen, og vælg den foreskrevne
flowindstilling ved hjælp af knappen til øgning (+)
eller reduktion (-) af flow.
Bemærk: Hvis batteripakken skal genoplades, harbatteriikonet på displayet en enkelt orange bjælke.

ADVARSEL: UNDGÅ AT TILSLUTTE ENEKSTERN STRØMFORSYNING, HVISDER IKKE ER SAT EN BATTERIPAKKE I
ILTKONCENTRATOREN.
ADVARSEL: HVIS ENHEDEN ER TILSLUTTET EN
BILS 12 VDC-STIK, MÅ DEN IKKE ANVENDES, MENSKØRETØJET ER SLUKKET.
1. Start køretøjets motor.
2. Sæt jævnstrømsforsyningen i strømindgangennederst til højre foran på FreeStyle Comfort. Sørg forat skubbe den indad, indtil den sidder forsvarligt fast.

PN MN232-C4 K | Brugermanual

DAN - 233

FreeStyle Comfort
Rengøring, pasning og
rutinevedligeholdelse

PLEJE AF BATTERIPAKKEN

RUTINEVEDLIGEHOLDELSE

Batteriet må ikke udsættes for åben ild.

Forebyggende vedligeholdelse
Enheden er vedligeholdelsesfri. Filtre inde i enheden
er beregnet til at vare enhedens levetid. Hvis enheden
ikke fungerer korrekt, skal enheden returneres til
udbyderen for reparation. Service af komponenter
inde i kabinettet på FreeStyle Comfort skal udføres af
iltleverandøren.
Brugeren af denne enhed bærer ansvaret fornedenstående vedligeholdelsesprocedurer.
• Ugentligt – Rengør luftindtagsfiltrene
ADVARSEL: GØR IKKE FORSØG
PÅ AT ÅBNE ELLER FJERNE
PLASTIKDÆKSLERNE PÅ
ILTKONCENTRATOREN. ENHEDENMÅ KUN ÅBNES AF EN KVALIFICERETSERVICETEKNIKER.

LUFTINDTAGSFILTRE

Luften kommer ind i FreeStyle Comfort gennem
luftindtagsfiltre, der er placeret nederst foran på
iltkoncentratoren. Disse filtre forhindrer, at der trænger
støv eller andre store partikler ind i systemet.
Luftindtagsfiltrene skal rengøres mindst én gangom ugen. Din iltleverandør kan eventuelt rådedig til at rengøre de hyppigere afhængigt af
driftsbetingelserne. Udfør trinnene herunder for atrengøre luftindtagsfiltrene korrekt.
1. Fjern de to filtre fra forsiden af FreeStyle Comfort
ved at trykke den tilsvarende tap opad og trække denudad.

2. Vask begge filtre i en varm opløsning af sæbe og
vand.
3. Skyl filtrene grundigt, og fjern overskydende vandmed en blød, fnugfri klud. Sørg for, at filterene erhelt tørre, inden de sættes på plads.
4. Isæt de rene, tørre filtre igen.
ADVARSEL: UNDGÅ AT BRUGE
ILTKONCENTRATOREN, UDEN AT
LUFTINDTAGSFILTERET ER ISAT.

234 -DANPN MN232-C4 K | Brugermanual

Batteripakken må ikke udsættes for væsker.
Gør ikke forsøg på at adskille eller åbne batteripakken.

Undgå at tabe batteripakken.
Opbevar batteripakken uden for børns rækkevidde.
Hvis der er tegn på beskadigelse af batteripakken,
skal du kontakte din iltleverandør, inden du tager den
i brug.
Batteriet må ikke anbringes i eller i nærheden af en
mikrobølgeovn eller andet husholdningsudstyr.

OPBEVARING AF BATTERIPAKKEN

Det er ikke nødvendigt at aflade batteriet iiltkoncentratoren helt, før det genoplades. Det
anbefales at oplade batteriet i iltkoncentratoren efter
hver brug.
Batteriet må ikke ligge et sted, hvor det er udsat for
overdreven varme, som f.eks. i direkte sollys eller i
en bil. Hvis det sker, kan det forringe batteriets ydelseog forkorte dets levetid. Det kan også få batteriet til at
generere varme og røg, bryde i brand eller eksplodere.
Batteripakken skal opbevares i stabile omgivelser med
lav luftfugtighed (under 70 % relativ luftfugtighed) og
fri for ætsende gasser. For at forebygge rustdannelse
er det vigtigt at undgå omgivelser, som kan forårsage
kondensering på grund af eksempelvis hurtigeændringer af den omgivende temperatur.
Ved opbevaring i en måned eller derover anbefales
afladning eller delvis afladning til et niveau mellem
50 % og 75 % (to til tre lysdioder). Tag batteriet ud af
iltkoncentratoren, når det er opladet.

FreeStyle ComfortRENGØRING

Sluk for enheden, og frakobl veksel- ellerjævnstrømforsyningen, før al rengøring. Du MÅ IKKE
sprøjte direkte på kabinettet. Brug en fugtig (ikke våd)klud eller svamp. Spray kluden eller svampen med enmild husholdningsrenser for at rengøre kabinettet og
strømforsyningerne. Forsæt efter anvisningerne fra
producenten af rengøringsmidlet.
ADVARSEL: SLUK FOR ENHEDEN,
OG TRÆK STRØMKABLET UD AFSTIKKONTAKTEN, FØR DU RENGØR
ENHEDEN, FOR AT FORHINDREUTILSIGTET ELEKTRISK STØD OG
FORBRÆNDINGER. FJERNELSE AF
DÆKSLER ELLER SERVICE PÅ ENHEDENMÅ KUN UDFØRES AF LEVERANDØREN
AF UDSTYRET ELLER EN KVALIFICERETSERVICETEKNIKER.
ADVARSEL: VÆR OMHYGGELIG MED AT UNDGÅ,AT ILKONCENTRATOREN OG BATTERIPAKKENBLIVER VÅDE, ELLER AT DER KOMMER VÆSKE IND
I ENHEDEN. DETTE KAN MEDFØRE FUNKTIONSFEJL
ELLER NEDLUKNING SAMT ØGET RISIKO FORELEKTRISK STØD ELLER FORBRÆNDINGER.

ADVARSEL: BRUG EN FUGTIG (IKKE VÅD) KLUDELLER SVAMP TIL AT RENGØRE BATTERIPAKKEN.
SPRAY FØRST KLUDEN ELLER SVAMPEN MED
ET MILDT RENGØRINGSMIDDEL, OG RENGØRBATTERIPAKKENS HUS.
ADVARSEL: UNDGÅ AT RENGØRE VED HJÆLP AF ET
AUTOMATISK VASKESYSTEM.

UDSKIFTNING AF KANYLEN

Følg altid kanyleproducentens vejledning for korrekt
brug. Udskift næsekanylen eller iltslangerne hver 3. til
6. måned (hver 3. måned ved daglig brug) i henhold til
kanyleproducentens eller iltleverandørens anbefaling.
Din læge eller iltleverandør informerer dig om
rengøring, desinfektion og udskiftning.
Du kan få yderligere udskiftningsdele hos diniltleverandør.

ADVARSEL: UNDGÅ AT BRUGE VÆSKE DIREKTEPÅ ENHEDEN. LISTEN OVER UØNSKEDE KEMISKESTOFFER OMFATTER, MEN ER IKKE BEGRÆNSET
TIL, FØLGENDE: ALKOHOL OG ALKOHOLBASEREDEPRODUKTER, KONCENTREREDE KLORINBASEREDEPRODUKTER (ÆTYLENKLORID) SAMTOLIEBASEREDE PRODUKTER. DISSEMÅ IKKE ANVENDES TIL RENGØRING AF
PLASTIKKABINETTET PÅ KONCENTRATOREN, DA DE
KAN BESKADIGE ENHEDEN.
ADVARSEL: RENGØR KUN KABINETTET,BETJENINGSPANELET OG STRØMKABLET MED
ET MILDT RENGØRINGSMIDDEL PÅ EN FUGTIG(IKKE VÅD) KLUD ELLER SVAMP, OG AFTØR ALLEOVERFLADER. UNDGÅ, AT DER KOMMER VÆSKE INDI KONCENTRATOREN. VÆR SÆRLIG OPMÆRKSOMPÅ ILTUDTAGET TIL KANYLETILSLUTNINGEN FOR
AT SIKRE, AT DEN ER FRI FOR STØV, VAND OG
PARTIKLER.

PN MN232-C4 K | Brugermanual

DAN - 235

FreeStyle Comfort

Alarmtilstande
Prioritetnr.

Alarmbeskrivelse

1

Systemfejl,
batteri høj
temperatur

Display

Lydsignal

Flere detaljer/anbefalet handling

1 bip hvert
20. sek.

Enheden udløser en alarm, når batteriets
temperatur nærmer sig tærsklen for
sikkerhedsnedlukning.
Enheden lukker ned, hvis nedlukningstærsklen
overskrides. Hvis en ekstern strømkilde er
tilsluttet, vises alarmskærmbilledet, og derudsendes en lydalarm, som fortsætter i 2 min.
Test med batteri ved stuetemperatur. Kontakt
leverandøren, hvis problemet vedvarer.

2

Systemfejl, høj
temperatur

1 bip hvert
20. sek.

Enheden udløser en alarm, når den indvendige
temperatur nærmer sig nedlukningstærsklen.
Hvis nedlukningstærsklen overskrides, lukker
enheden ned, og alarmskærmbilledet vises,
mens der udsendes en lydalarm, som fortsætter
i 2 min.
Sørg for, at enhedens ventilationssprækker ikke
blokeres af noget. Kontakt leverandøren, hvis
problemet vedvarer.

3

Systemfejl, højt
tryk

1 bip hvert
20. sek.

Hvis nedlukningstærsklen overskrides,
lukker enheden ned, og alarmskærmbilledet
vises, mens der udsendes en lydalarm, som
fortsætter i 2 min.
Kontakt leverandøren, hvis problemetvedvarer.

4

Ingenvejrtrækning
detekteret1 bip hvert
30. sek.

Ingen vejrtrækning detekteret i 15 sekunder.
Aktiveres efter første vejrtrækning, eller når
opvarmningsskærmbilledet er slukket, alt efter
hvad der kommer først.
Kontroller kanylen for knæk eller lækager. Forøg
følsomhedsindstillingen. Kontakt leverandøren,hvis problemet vedvarer.

5

Nedlukningpga. lavt
batteriniveau

1 bip hvert
30. sek.

6

Opvarmning

Intet

Enheden starter med at dosere automatisk,
indtil der detekteres en vejrtrækning, eller
opvarmningen slutter, alt efter hvad derkommer først.

7

Lav
koncentration

1 bip hvert
30. sek. i
op til 5 min.

Kontakt leverandøren, hvis problemetvedvarer.

236 -DANPN MN232-C4 K | Brugermanual

Alarmen fortsætter, indtil batteriet løber tør for
strøm, og enheden lukkes ned.
Genoplad batteriet.

FreeStyle Comfort
Prioritetnr.

Alarmbeskrivelse

8

Display

Lydsignal

Flere oplysninger

Systemfejl,
flowhastighed(ventilsvigt iiltkoncentrator,
blokering af
gasvej)

1 bip hvert
30. sek.

Kontakt leverandøren, hvis problemetvedvarer.

9

Advarsel for lavt
batteriniveau

1 bip, ingengentagelse

Genoplad batteriet.

10

Respirationsfrekvens
overskredet

1 bip hvert
60. sek.

Respirationsfrekvensen overskrider
40 vejrtrækninger pr. minut.

11

Systemfejl,
ventilatorsvigt

Intet

Kontakt leverandøren, hvis problemetvedvarer.

12

Systemfejl, lavt
tryk

Intet

Kontakt leverandøren, hvis problemetvedvarer.

13

Systemfejl, lav
temp.

Intet

Lad enheden varme op til stuetemperatur.
Kontakt leverandøren, hvis problemetvedvarer.

Info

Opstart1 infobip

Dette skærmbillede vises under opstarten,og du bør høre et bip, som bekræfter, atbrummeren fungerer.

Hvis der udløses dosis utilsigtet, skal
følsomhedsindstillingen forøges. Kontakt
leverandøren, hvis problemet vedvarer.

Opstartsskærmbilledet viser den aktuelle
firmwarerevision i nederste højre hjørne.
InfoSkærmbillede
for normal drift
i autoSATtilstand (ingenjævnstrøm
tilsluttet)

Intet

PN MN232-C4 K | Brugermanual

DAN - 237

FreeStyle Comfort
Prioritetnr.

Alarmbeskrivelse

Display

Lydsignal

InfoSkærmbillede
for normal drift
med autoSATtilstanddeaktiveret(ingenjævnstrøm
tilsluttet)

Intet

Info

Indstillingændret1 infobip

Info

Tilslut ekstern
strømkilde i
fra-tilstand

Intet

Info

Tilslut ekstern
strømkilde ikørselstilstand

Intet

Info

Frakobl ekstern
strømkilde i
fra-tilstand

1 infobip

Info

Frakobl ekstern
strømkilde ikørselstilstand(med batteri
isat)

1 infobip

238 -DANPN MN232-C4 K | Brugermanual

Flere oplysninger

CAIRE-skærmbilledet vises i 2 sekunder,
efterfulgt af batteri- og stikstatus i 5 sek.,før displayet slukkes.

CAIRE-skærmbilledet vises i 2 sekunder,
hvorefter displayet slukkes.

FreeStyle Comfort
Prioritetnr.

Alarmbeskrivelse

Info

Display

Lydsignal

Flere oplysninger

I standbytilstand – når
der trykkes
på op- eller
nedknappen,
viser displayet
batteri- og
ladetilstandIntetSkærmbilledet vises i 5 sek., hvorefter
displayet slukkes.

Info

Flytilstand med
batteristrøm

1 langt bip

Info

Flytilstand med
tilslutning til
stikkontakt

1 langt bip

PN MN232-C4 K | Brugermanual

DAN - 239

FreeStyle Comfort

Instruktioner vedr. menunavigation
Du kan åbne menuen i tilstanden Standby eller
kørselstilstand. Hvis der udløses en alarmtilstand ikørselstilstand, afsluttes menuen automatisk. Hvis derikke trykkes på nogen knapper i 20 sekunder, afsluttes
menuen automatisk. I standby skal du muligvis trykke
to gange på knappen Menu for at vække enheden og
derefter åbne menuen.
Der er fire menuskærmbilleder:
1. Statusskærmbillede: viser firmwarens
revisionsnummer, kørselstid,batteriafladningscyklusser

Flytilstand2. Flytilstand (TIL/FRA)
3. Følsomhed: justerbar (1-5)
4. autoSAT (TIL/FRA)

FølsomhedEksempel på statusskærmbillede

BETJENING AF INDSTILLINGSMENUEN
Indstillingsmenuen åbnes ved at trykke på knappen
Menu/Vælg
i tilstanden Kør eller Standby.
Løb igennem menuerne ved at trykke på

Statusside

240 -DANPN MN232-C4 K | Brugermanual.

autoSATFreeStyle Comfort
STATUS SIDE
Statussiden for Freestyle Comfort viser firmwareversion, kørselstider og antal batterikredsløb på det
tilsluttede batteri.

Flytilstand aktiveret bekræftet

Deaktiver flytilstand ved at gentage ovenstående, mens
du skifter til FRA.
Statusside

Kørselstimer vises i hele timer.
Batteriopladning er antallet af cyklusser, som det
tilsluttede batteri er blevet tømt og genopladet.

FLYTILSTAND
FreeStyle Comfort leveres med flytilstand deaktiveret.

Tryk på
. Når skærmbilledet viser symbolet”Næste”, overgår menusiden til det næste skærmvalg.
Flytilstand er beregnet til brug i kørselstilstand.
Flytilstand deaktiveres automatisk, hvis enheden
efterlades i standby på batteristrøm.
Bemærk: Se afsnittet Aktiver flytilstand for at få yderligere
oplysninger om flytilstand.

FØLSOMHEDSTILSTAND
FreeStyle Comfort leveres med enstandardfølsomhedstilstand på 2 fra fabrikkens sideog bør ikke kræve justering under normale forhold.
Følsomhedsindstillingen kan justeres i værdiintervallet
1-5, hvor 1 er det mindst følsomme, og 5 er det mest
følsomme.
Hovedskærmbillede for flytilstand

Aktiver tilstanden ved at trykke på, mens ikonet
SKIFT er synligt. Værdien skifter til det blå ord”TÆNDT”.

Skærmbilledet Følsomhedstilstand ved indstillingen 2

Juster følsomhedstilstanden ved at trykke på,mens ikonet SKIFT er synligt. Værdien skifter til detnæste blå tal.

Flytilstand aktiveret valgtBekræft flytilstand ved at trykke på, mens ikonet
GEM er synligt. Værdien skifter tilbage til ordet”TÆND”.
Annuller flytilstand ved at trykke på
ANNULLER er synligt.

, mens ikonet
Følsomhedstilstand indstilling 3 valgt

PN MN232-C4 K | Brugermanual

DAN - 241

FreeStyle ComfortBekræft indstilling 3 i følsomhedstilstand ved at trykkepå, mens ikonet GEM er synligt. Værdien skifter
tilbage til ”3”.
Annuller Følsomhedstilstanden ved at trykke påmens ikonet ANNULLER er synligt.

,

TILSTANDEN AUTOSATFreeStyle Comfort leveres med tilstanden autoSAT
aktiveret.

Hovedskærmbillede for autoSAT
Følsomhedstilstand indstilling 3 bekræftet

Deaktiver tilstanden ved at trykke på, mensikonet SKIFT er synligt. Værdien skifter til det blå ord”SLUKKET”.

Følsomhedstilstand indstilling 5 (maks.) bekræftet

Tryk på
. Når skærmbilledet viser symbolet”Næste”, overgår menusiden til det næste skærmvalg.

Tilstanden autoSAT aktiveret er valgtBekræft tilstanden autoSAT ved at trykke på,mens ikonet GEM er synligt. Værdien skifter tilbage til
ordet ”SLUK”.
Annuller tilstanden autoSAT ved at trykke påmens ikonet ANNULLER er synligt.

,

Tilstanden autoSAT deaktiveret bekræftet

Aktiver flytilstand ved at gentage ovenstående, mens
du skifter til TIL.
Tryk på
. Når skærmbilledet viser symbolet”Næste”, overgår menusiden til det næste skærmvalg.

242 -DANPN MN232-C4 K | BrugermanualFreeStyle Comfort
Fejlfinding
ProblemMulig årsag

Løsning

FreeStyle Comfort viser alarmog udsender periodiske bip.

Se tabellen med alarmtilstande.

Se tabellen med alarmtilstande.

Iltkoncentratoren tændes ikke.

Batteriet er afladet.

Tilslut enheden til en jævnstrøm- eller
vekselstrømforsyning.

Den er ikke sluttet til ekstern
strømforsyning.
Generel fejlfunktion.

Kontrollér, at de eksterne tilslutninger er sikre.
Udfør en komplet sluk/tænd-cyklus ved at tagebatteriet ud og frakoble en eventuel ekstern vekseleller jævnstrømskilde. Sæt derefter batteriet i igen,hvorefter sluk/tænd-cyklussen er fuldført. Tilslut
eventuelt den eksterne strømkilde igen.
Kontakt udstyrsleverandøren, og skift om
nødvendigt til en anden iltkilde.

Ved alle andre problemer.

Udfør en komplet sluk/tænd-cyklus ved at tagebatteriet ud og frakoble en eventuel ekstern vekseleller jævnstrømskilde. Sæt derefter batteriet i igen,hvorefter sluk/tænd-cyklussen er fuldført. Tilslut
eventuelt den eksterne strømkilde igen.
Skift til en anden iltkilde, og kontakt
udstyrsleverandøren.

Bemærk: Nogle alarmer får enheden til at lukke ned.
Bemærk: Under en total strømafbrydelse nulstilles alarmsystemet (rydder alle tidligere alarmer), og der vises kun en
alarm, hvis tilstanden stadig er aktuel, når strømmen genoprettes.

PN MN232-C4 K | Brugermanual

DAN - 243

FreeStyle Comfort

Oplysninger om rejser

1. Tænd KUN POC med batteristrøm, når der ikke er
tilsluttet en ekstern strømkilde.

FreeStyle Comfort imødekommer FAA’s retningslinjer
vedr. brug på ruteflyvninger. Forud for enhver rejse(med fly, tog, skib osv.) skal du:• sørge for, at koncentratoren er i god stand og
driftsklar• kontakte rejseselskabet mindst 48 timer før afrejsen
for at informere dem om, at du rejser med en
FreeStyle Comfort.
Det anbefales, at du altid medbringer ekstra batterier pårejser. I forbindelse med flyrejser skal du sørge for athave tilstrækkeligt batteri til hele forløbet (check-in før
flyvning, sikkerhedstjek osv.) samt mellemlandinger.
De fleste flyselskaber kræver, at du har et tilstrækkeligt
antal fuldt opladede batteripakker til drift af enheden imindst 150 % af turens forventede maksimale varighed(flyvetider, al tid på jorden, før og efter flyvningen,under forbindelser samt til uventede forsinkelser).

2. Aktiver flytilstand ved brug af én af følgende
metoder:a. Følg instruktionerne for at aktivere flytilstand i
afsnittet Instruktioner vedr. menunavigation.

Bemærk: På flyrejser anbefales det at medbringe de ekstra
batterier i en sikker taske.

FAA-REGULERING

For at forhindre kortslutninger skal de ekstra
batterier beskyttes individuelt ved at blive anbragt
i den oprindelige emballage eller ved anden
terminalisolering, f.eks. ved at sætte tape hen overudsatte terminaler eller ved at anbringe hvert batteri
i en separat plastikpose eller et beskyttelsesetui, ogde må kun transporteres i håndbagagen.

b. Tryk på op- og nedknapperne, og hold dem nede i
5 sekunder, indtil du hører et langt bip, og symbolet
for flytilstand vises.

AKTIVÉR FLYTILSTAND
Batteristrøm
POC kan bruges ombord på fly i henhold til de normale
betjeningsprocedurer, KUN med batteristrøm, hvis derikke er tilsluttet en ekstern strømkilde.

3. Når flysymbolet vises, skal du slutte POC til
stikkontakten.

Eksternt strømHvis der er en tilgængelig stikkontakt ombord på etfly, og den er placeret, så den er praktisk at brugesom POC-strømkilde, kan den fungere som reserve
for POC-batterierne. I dette tilfælde skal følgende
anbefalede procedure for skift fra batteri til strøm fra
flyet følges:Bemærk: Ombord på fly er det ikke muligt at opladebatterierne i flytilstand.

244 -DANPN MN232-C4 K | BrugermanualFreeStyle Comfort
Tilbehør

PLACERING AF SYSTEMET I BÆRETASKEN

Anvend kun tilbehør på listen, som er leveret af
CAIRE via din iltleverandør, af hensyn til korrekt
ydeevne og sikkerhed. Brug af tilbehør, der ikke
er angivet nedenfor, kan påvirke ydeevnen og/eller
sikkerheden af koncentratoren.

1. Anbring systemet i bunden af tasken, og sørg for atplacere filtrene og batteriåbningerne korrekt.

Standardtilbehør til FreeStyle ComfortBæretaske

MI396-1RygsækMI459-1

Skulderrem

MI397-1

Vekselstrømsforsyning (alle ledninger
medfølger)

PW036-1SJævnstrømskabel

CD041-1

Eksternt grovpartikelfilter (højre)FI226-1

Eksternt grovpartikelfilter (venstre)

FI227-1

2. Luk lynlåsen rundt om bunden af tasken.

Valgfrit ekstraudstyr
Ekstra 8-cellet batteripakke

BT034-1S

16-cellet batteripakke

BT035-1S

Ekstern oplader til batteripakker

BT036-1

Tilbehørstaske

MI372-2Bemærk: Der kan være flere valgmuligheder med henblik
på landespecifikke strømkabler, som anført ovenfor.
Kontakt CAIRE eller din iltleverandør, hvis du har brug for
flere valgmuligheder i forbindelse med en bestilling.

Instruktioner til bæretasken

MONTERING AF SKULDERREM
1. Skulderremmen kan fastgøres direkte på enheden
eller på bæretasken.
2. Sæt remmens clips fast på enhedens eller taskens
ring.

Følg nedenstående instruktioner for at placere og
fastgøre dit FreeStyle Comfort-system i bæretasken.

PN MN232-C4 K | Brugermanual

DAN - 245

FreeStyle Comfort

Bortskaffelse og genbrug
Aflever altid FreeStyle Comfort, med allekomponenter, til din udstyr med henblik påbortskaffelse. Du kan også kontakte den lokale
myndighed angående instruktioner om korrektbortskaffelse af batteriet. Enheden kan genbruges
mellem mennesker, men skal returneres til udstyr
udbyderen til oparbejdning.

elektrisk og elektronisk udstyr (WEEE).
Vores produkter overholder begrænsningerne i RoHSdirektivet. De indeholder ikke mere end spor af bly
eller andre farlige stoffer.
FORSIGTIG: Kontakt leverandøren
af udstyret eller din lokale offentlige
institution for at høre om kravene tilbortskaffelse.

WEEE og RoHS
Dette symbol har til formål at minde ejere
af udstyr om at returnere det til et genvindingsanlæg,når det er udtjent, i henhold til direktivet om affald af

Materialer, der kommer i direkte berøring med brugeren
Koncentratorens kabinet.............................................................................................................................. PC/ABS
Koncentratorens betjeningspanel.......................................................................... Polyester EBG7 eller tilsvarende
Koncentratorens håndtag.......................................................................................................................Polycarbonat
Iltudtag, dyse.........................................................................................................................................Polycarbonat
Systemets mærkat............................................................................................................................................Lexan
Ledningsstik.......................................................................................................................Polycarbonat/vinylklorid
Ledning................................................................................................................................................... PVC, metal
Strømforsyning................................................................................................................. Lexan 940 (polycarbonat)
Batteripakke................................................................................................................................................. PC/ABS
Batteripakke, strømforsyningernes mærkater...................................................................................... Polyesterfilm
Koncentratorens bæretaske............................................................................................ Cordura, neopren, nylonfor
Rem...................................................................................................Polypropylen, acetyl, vinyl, Cordura, neopren

EMC-test
Vejledning og producentens erklæring – elektromagnetiske emissioner
FreeStyle Comfort er beregnet til brug i det elektromagnetiske miljø, der er angivet nedenfor. Kunden eller brugeren af
FreeStyle Comfort er ansvarlig for at sikre, at det anvendes i et sådant miljø.
Emissionstest
RF-emissioner
CISPR 11
RF-emissioner
CISPR 11
Harmoniske emissioner
IEC 61000-3-2Spændingsudsving/flickeremissioner
IEC 61000-3-3

246 -DAN

Overensstemmelse

Elektromagnetisk miljø – vejledning

Gruppe 1

FreeStyle Comfort anvender kun RF-energi til interne funktioner.
Derfor er RF-emissionerne meget lave og vil sandsynligvis ikkeforårsage interferens i elektronisk udstyr i nærheden.

Klasse B
Klasse A
Overholder

PN MN232-C4 K | Brugermanual

FreeStyle Comfort er egnet til brug i alle bygninger, såvel
privatboliger som bygninger, der er direkte forbundet til det offentlige
lavspændingsnet, som forsyner beboelsesejendomme.

FreeStyle Comfort
Vejledning og fabrikantens erklæring – elektromagnetisk immunitet

FreeStyle Comfort er beregnet til brug i det elektromagnetiske miljø, der er angivet nedenfor. Kunden eller brugeren af
FreeStyle Comfort er ansvarlig for at sikre, at det anvendes i et sådant miljø.
Immunitetstest
IEC 60601-testniveau
Overensstemmelsesniveau
Elektromagnetisk miljø –vejledning
Gulvene bør være af træ, beton eller
Elektromagnetisk miljø
keramiske fliser. Hvis gulvene er±8 kV kontakt±8 kV kontakt– vejledningbeklædt med syntetisk materiale, skal
±15 kV luftudladning±15 kV luftudladning
den relative fugtighed være mindst
IEC 61000-4-230 %.
±2 kV for strømledninger
±2
kV
for
strømledninger
Elektrisk hurtig
Strømkvaliteten skal være dentransient/burst
samme som i typiske hospitals±1 kV for input/og hjemmeplejemiljøer.
IEC 61000-4-4
outputledninger±1 kV linje til linje

Strømstød
IEC 61000-4-5±2kV normal tilstand på
vekselstrømsledninger±1kV differensstrøm på
vekselstrømsledninger±2kV normal tilstand på
udendørs I/O-ledninger
0 % UT i 0,5 cyklus (0°,45°, 90°, 135°, 180°,
225°, 270°, 315°)Spændingsfald,korte afbrydelser ogspændingsvariationer
på strømforsyningens
indgangsledninger
IEC 61000-4-11

Driftsfrekvens(50/60 Hz) magnetfelt
IEC 61000-4-8±1 kV linje til linje

0 % UT i 1 cyklus (00)
70 % UT (30 % fald i UT) i
25/30 cyklusser (00)Strømkvaliteten skal være densamme som i typiske hospitalsog hjemmeplejemiljøer.

0 % UT i 0,5 cyklus (0°, 45°,90°, 135°, 180°, 225°, 270°,
315°)

Strømkvaliteten skal være densamme som i typiske hospitals- og
hjemmeplejemiljøer. Hvis brugeren
0 % UT i 1 cyklus (00)af FreeStyle Comfort kræver
uafbrudt drift under strømafbrydelser,
70 % UT (30 % fald i UT) i 25/30 anbefales det, at FreeStyle
Comfort forsynes med strøm fra en
cyklusser (00)
nødstrømsforsyning eller et batteri.

0 % UT i 250/300
cyklusser (00)

0 % UT i 250/300 cyklusser (00)

3 A/m

3 A/m

Netfrekvensmagnetfelterne skal værepå et niveau svarende til niveauet påen typisk placering i et hospitals- eller
hjemmeplejemiljø.

BEMÆRK UT er netspændingen før anvendelse af testniveauet.

BEMÆRK I nogle tilfælde kan en betydelig elektrostatisk udladning medføre, at Freestyle Comfort nulstilles og kræver en
komplet sluk/tænd-cyklus, før normal drift kan genoptages. Se afsnittet Fejlfinding for at få flere oplysninger.

PN MN232-C4 K | Brugermanual

DAN - 247

FreeStyle Comfort
Vejledning og fabrikantens erklæring – elektromagnetisk immunitet

FreeStyle Comfort er beregnet til brug i det elektromagnetiske miljø, der er angivet nedenfor. Kunden eller brugeren af
FreeStyle Comfort er ansvarlig for at sikre, at det anvendes i et sådant miljø.
Immunitetstest
IEC 60601-testniveau
Overensstemmelsesniveau
Elektromagnetisk miljø – vejledning
Ledet RF
10 Vrms
10 VrmsBærbart og mobilt RFkommunikationsudstyr må ikke
IEC 61000-4-6
150 kHz til 80 MHz
anvendes tættere på nogen del af
FreeStyle Comfort, herunder kabler,end den anbefalede afstand beregnet
ud fra den ligning, som gælder for
Udstrålet RF
10 V/m
10 V/msenderfrekvensen.
IEC 61000-4-3
80 MHz til 2700 MHz
Anbefalet sikkerhedsafstandd = 1,.2 P

d = 1,.2 P = 80 MHz til 800 MHz
d = 1,.2 P = 800 MHz til 2.700 MHz
hvor P er senderens maksimale
udgangseffekt i watt (W) i henhold til
producenten af senderen, og d er den
anbefalede afstand i meter (m).
Feltstyrker fra faste RF-sendere,som fastsat af en elektromagnetisk
undersøgelse, a, skal være mindre end
overensstemmelsesniveauet i hvert
frekvensområde. b
Interferens kan forekomme i nærheden
af​udstyr mærket med følgende symbol:

NOTE 1 Ved 80 MHz og 800 MHz gælder det højere frekvensområde.
NOTE 2 Disse retningslinjer gælder muligvis ikke i alle situationer. Elektromagnetisk interferens påvirkes af absorption og
refleksion fra bygninger, genstande og mennesker.
aFeltstyrker fra faste sendere, f.eks. basestationer til radiotelefoner (mobile/trådløse) og landmobile radioer, amatørradio,
AM- og FM-radioudsendelser og tv-udsendelser kan ikke forudsiges teoretisk med nøjagtighed. Det bør overvejes atforetage en elektromagnetisk undersøgelse for at vurdere det elektromagnetiske miljø, som skyldes faste RF-sendere. Hvis
den målte feltstyrke på det sted, hvor FreeStyle Comfort anvendes, overstiger ovenstående gældende RF-niveau, bør det
kontrolleres, at FreeStyle Comfort fungerer normalt. Hvis der konstateres unormal ydeevne, kan yderligere forholdsreglervære nødvendige, såsom at vende eller flytte FreeStyle Comfort.
bOver frekvensområdet 150 kHz til 80 MHz skal feltstyrken være mindre end 10 V/m.

248 -DANPN MN232-C4 K | BrugermanualFreeStyle Comfort
Anbefalet separationsafstand mellem bærbart og mobilt RF-kommunikationsudstyr og
FreeStyle Comfort

FreeStyle Comfort er beregnet til brug i et elektromagnetisk miljø, hvori udstrålede RF-forstyrrelser styres. Kundeneller brugeren af FreeStyle Comfort kan hjælpe til med at forhindre elektromagnetisk interferens ved at bevare en
minimumsafstand mellem bærbart og mobilt RF-kommunikationsudstyr (sendere) og FreeStyle Comfort, som anbefalet
nedenfor, i henhold til kommunikationsudstyrets maksimale udgangseffekt.
Senderens maksimale
Sikkerhedsafstand i henhold til senderens frekvens
udgangseffektmW

150 kHz til 80 MHz

80 MHz til 800 MHz

d = 1,.2 P

d = 1.,2 P

800 MHz til 2,5 GHz

d = 1,.2 P
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
For sendere med en maksimal udgangseffekt, der ikke er angivet ovenfor, kan den anbefalede sikkerhedsafstand d imeter (m) anslås ved hjælp af den ligning, der gælder for den pågældende senders frekvens, hvor P er den maksimale
udgangseffekt for senderen i watt (W) i henhold til producenten af senderen.
NOTE 1 Ved 80 MHz og 800 MHz gælder sikkerhedsafstanden for det højere frekvensområde.
NOTE 2 Disse retningslinjer gælder muligvis ikke i alle situationer. Elektromagnetisk interferens påvirkes af absorption og
refleksion fra bygninger, genstande og mennesker.

Klassifikation

Beskyttelsestype mod elektrisk stød:
Klasse II	Beskyttelse mod elektrisk stød opnås ved
dobbelt isolering.
Grad af beskyttelse mod elektrisk stød:
Type BF Udstyret yder en bestemt grad af
beskyttelse mod elektrisk stød med hensyn til		 1) tilladt lækstrøm		2) pålidelighed af jordforbindelsen (hvis til
stede).
		 Ikke beregnet til direkte hjerteanvendelse.
Uafhængig prøvning iht. Standard for
Elektromedicinsk udstyr:
Testet til at være i overensstemmelse med• IEC 60601-1 udgave 3.1: Elektromedicinsk
udstyr – Del 1 Generelle krav til sikkerhed.
• CAN/CSA C22.2 No. 60601-1-8 udgave 2.1,
Medicinsk elektrisk udstyr – Del 1: Generelle
krav til sikkerhed
Beskyttelse mod potentiel elektromagnetisk eller anden
interferens mellem udstyret og andre enheder.
	• Testet til at være i overensstemmelse med
EN 60601-1-2/IEC 60601-1-2: 4. udgave.

	• Testet til at være i overensstemmelse med RTCA/
DO160, afsnit 21, kategori M.
	• CISPR 11/EN 55011, klasse B, gruppe 1,”Industrielt, videnskabeligt og medicinsk (ISM)
udstyr"	• FCC-del 15, underdel B – klasse B, hændeligestrålingskilder
IP22 – Beskyttelse mod indtrængning af fingre og
lignende genstande, der har en diameter på over
12,5 mm. Vertikale dråber må ikke have skadeligevirkninger, når udstyret har en hældningsvinkel op
til 15° i forhold til vertikallinjen.
Tilladt metode til rengøring og infektionskontrol:
Se afsnittet ”Rengøring, pasning og korrekt
vedligeholdelse” i denne FreeStyle-brugermanual.
Sikkerhedsgrad ved anvendelse under tilstedeværelse
af brændbare anæstesigasser:
Udstyret egner sig ikke til en sådan anvendelse.
Funktionsmetode: Kontinuerlig anvendelse.

CAIRE og CAIRE Inc. er registrerede varemærker tilhørende CAIRE Inc. Besøg vores
websted herunder for en fuld liste over varemærker.
Varemærker: www.careinc.com/corporate/trademarks.
Copyright © 2021 CAIRE Inc. CAIRE Inc. forbeholder sig retten til at lade deres
produkter udgå eller ændre priserne, materialerne, udstyret, kvaliteten, beskrivelserne,
specifikationerne og/eller processerne for deres produkter til enhver tid uden forudgående
varsel og uden yderligere forpligtelser eller konsekvens. Alle rettigheder, der ikke
udtrykkeligt er angivet heri, er forbeholdt os, hvor det er relevant.

PN MN232-C4 K | Brugermanual

DAN - 249

FreeStyle Comfort

Brukerkontroller og indikatorer for systemstatus.
ISO 7000; Tegnesymboler til bruk på spesielt utstyr –indeks og sammendragRådsdirektiv 93/42/EØF vedrørende medisinsk utstyr
Autorisert representant i EU
Hvis produktets unikeenhetsidentifikator (UDI) -etikett
har CE #### -symbolet på, oppfyller
enheten kravene i direktiv 93/42 /EØF om medisinsk utstyr. CE ####
symbolet indikerer varslet organ
nummer.

Les bruksanvisningen før bruk.
Reg. nr. 1641
Lagring eller
driftstemperaturområde. Reg. nr.
0632
Oppbevaring fuktighetsområde. Reg.
nr. 2620

IEC 60417
Klasse II utstyr, dobbeltisolertReg. nr. 5172

Atmosfærisk trykkbegrensning.
Reg. nr. 2621
Produsentens navn og adresse.
Reg. nr. 3082

Strømbryter. Reg. nr. 5009
ETL-sertifisering

Produksjonsdato. Reg. nr. 2497

Sikkerhetsbyrå for
CAN/CSA C22.2 nr. 601.1 M90
for medisinsk elektrisk utstyr

Katalognummer. Reg. nr. 2493
Serienummer. Reg. nr. 2498

Interne symbolerMå holdes unna brennbare
materialer, olje og smørefett.

Denne siden opp. Reg. nr. 0623
Glass – håndteres med forsiktighet.
Reg. nr. 0621

Må ikke demonteres.

Må holdes unna regn – oppbevares
tørt. Reg. nr. 0626

Øk oksygentilførsel.

Stablingsgrense etter nummer.
Reg. nr. 2403

Reduser oksygentilførsel.

ISO 7010: Grafiske symboler – Sikkerhetsfarger og
sikkerhetsskilter – Registrerte sikkerhetsskilt
Instruksjonsmanualen må leses.
Reg. nr. M002
Holdes unna åpen flamme, brann,
gnister. Åpne tennkilder og røyking
forbudt. Reg. nr. P003
Røyking forbudt nær enheten eller
mens enheten brukes. Reg. nr. P002
Anvendt del type BF (grad av
beskyttelse mot elektrisk støt).
Reg. nr. 5333
Advarsel. Reg.nr. W001

250 - NORPN MN232-C4 K | Bruksanvisning

Menyknapp.
RTCA/DO-160
avsnitt 21
Samsvarskategori M

RTCA DO160 avsnitt 21
Samsvarskategori M. FAA SFAR
106 krav
Utstyr av klasse II, dobbeltisolert

FAA-godkjent – POC
21 CFR 801.15: Samling av offentlige forskrifter
seksjon 21
Amerikansk føderal lovgivningbegrenser denne enheten til salg fra
en lege eller foreskrevet av en lege.

FreeStyle ComfortRådsdirektiv 2012/19/EF: avfall fra elektrisk og
elektronisk utstyr (WEEE)
WEEE

IEC 60601-1: Medisinsk elektrisk utstyr Del 1
Generelle krav for grunnleggende sikkerhet og
nødvendig ytelseDråpetettDette produktet kan være dekket av ett eller flereamerikanske eller internasjonale patenter. Gå til vår
hjemmeside nedenfor for en liste med gjeldende
patenter. Pat.: www.caireinc.com/corporate/patents/.

Hva er oksygenkonsentratoren

din bestemte respiratoriske tilstand.

Luften vi puster inn inneholder omtrent 21 % oksygen,
78 % nitrogen og 1 % andre gasser. I FreeStyle®
Comfort® trekkes luften i rommet inn i maskinen
via et luftinntak. Deretter passerer det gjennom etabsorberende materiale som heter molekylsikt. Dettematerialet skiller oksygenet fra nitrogenet og lar bareoksygenet slippe gjennom. Resultatet er en strøm av
høykonsentrert oksygen som leveres til brukeren.

Selv om oksygen ikke er et vanedannende legemiddel,
kan uautorisert oksygenbehandling være farlig.
Du må rådføre deg med lege før du bruker denne
oksygenkonsentratoren. Utstyrsleverandøren som
leverer oksygenutstyret ditt vil demonstrere hvordan
det innstilles til den forordnede strømningshastigheten.

FreeStyle Comfort kombinerer
avansert oksygenkonsentrator- med
oksygenkonserveringsteknologier, for å skape enbærbar enhet som muliggjør maksimal bærbarhet og
mobilitet. Den avanserte pulsstrømningsleveringen
føler raskt når brukeren puster inn, og leverer en puls
med oksygen ved begynnelsen av hver inhalasjon.
Levering med pulsstrømning er ideelt for aktiviteter og
tid hjemmefra. Den konserverer oksygenet og forlenger
batteriets brukstid, i motsetning til ved kontinuerlig
strømning. FreeStyle Comfort er lett nok til å bli båret
med skulderrem. Den kan brukes med vekselstrøm(vegg), likestrøm (bil) eller batteristrøm.

Konsentratorene er ment for levering av supplerende
oksygen til brukere som opplever ubehag på grunn
av lidelser som påvirker effektiviteten til lungene ioverføringen av oksygen i luften til blodet. Bærbare
oksygenkonsentratorer (POC-er) lagrer ikke eller
inneholder ikke oksygen. De trenger ikke å fylles
opp på nytt og kan lades opp hvor som helst hvor
vekselstrøm eller likestrøm er tilgjengelig (unntatt på
et fly). Dette gjør brukeren relativt selvhjulpen når det
gjelder hjemmebruk, bevegelighet (både innenfor og
utenfor hjemmet), mobilitet og generell livsstil. Brukav oksygenkonsentratoren krever en forordning fralege og er ikke ment brukt til livredning.

Tiltenkt bruk
CAIRE FreeStyle Comfort Oxygen Concentrator er
beregnet på administrering av tilleggsstoff. Enheten er
ikke beregnet på livsstøtte, og gir heller ikke noen evne
til å overvåke pasienten.

Hvorfor legen din har foreskrevet
oksygenMange personer lider av forskjellige hjerte-, lunge- og
andre respiratoriske sykdommer. Et betydelig antall
av disse personene kan dra nytte av supplerende
oksygenbehandling hjemme, på reise eller når de deltar
i daglige aktiviteter hjemmefra.
Oksygen er en gass som utgjør 21 % av luften irommet som vi puster inn, og kroppen vår er avhengig
av en jevn forsyning for å fungere ordentlig. Legen dinforeskrev en strømning eller innstilling for å behandle

Operatørprofil

Selv om oksygenbehandling kan foreskrives til
brukere i alle aldre, er den typiske brukeren av
oksygenbehandling over 65 år og lider av forskjellige
respiratoriske sykdommer, inkludert kronisk
obstruktiv lungesykdom (KOLS). Brukerne harvanligvis gode kognitive evner og må være i stand tilå kommunisere ubehag. Hvis brukeren ikke er i standtil å kommunisere ubehag, eller ikke kan lese eller
forstå etiketter på konsentratoren og bruksanvisningen,anbefales det at den bare brukes under oppsyn av
noen som kan. Hvis brukere føler noe slags ubehag
under bruk av konsentratoren, skal de ta kontakt
med helsepersonell. Brukerne rådes til å ha ekstra
oksygen tilgjengelig (dvs. sylinderoksygen) i tilfelle
strømbrudd eller konsentratorsvikt. Det kreves ingen
andre unike egenskaper eller brukerevner for å bruke
konsentratoren.

PN MN232-C4 K | Bruksanvisning

NOR - 251

FreeStyle Comfort

Utpakking av FreeStyle Comfort
Kontroller at alle komponentene
i listen medfølger i pakningen.
Dersom noen elementer mangler,ta umiddelbart kontakt med
oksygenleverandøren din.

Standard tilbehør• Bærbar oksygenkonsentrator• Bæreveske• Skulderstropp• Batteripakke med 8 celler• Vekselstrømforsyning• Strømledning til vekselstrøm• Likestrømtilførsel
• Bruksanvisning

Valgfritt tilbehør• Ytterligere batteripakke med 8celler• Batteripakke med 16 celler• Bordlader• Tilbehørsveske

Gjør deg kjent med FreeStyle Comfort
Gjør deg kjent med de viktigste funksjonene og brukergrensesnittet til FreeStyle Comfort.

Sett forfra
Stroppfeste

Brukergrensesnitt

Luftutslipp
Oksygenutløp

Luftinntak/filtre
Luftinntak/filtre

Knapp forstrømporttaket

Batteripakke
Utløserknapp forbatteripakke
252 - NORPN MN232-C4 K | Bruksanvisning

FreeStyle Comfort
Sett ovenfra
SkjermØk oksygentilførsel

På/av-knapp

Meny-knapp

Reduseroksygentilførsel

Skjerm på batteri
Innstillingen for
strømning

Batterivisning

autoSAT-indikator

Alarmvisning

Gjenværendebatteriprosent

Bolusleveringsindikator

Skjerm på veksel-/likestrøm

Innstillingen for
strømning

Visningsområde
for alarm

Batterivisning

Ladeindikator

PN MN232-C4 K | Bruksanvisning

NOR - 253

FreeStyle Comfort

Viktig!

Sikkerhetsinstruksjoner defineres slik:
ADVARSEL: VIKTIGSIKKERHETSINFORMASJON OM FARER
SOM KAN FORÅRSAKE ALVORLIGSKADE.
FORSIKTIG: Viktig informasjon for å
forebygge skade på FreeStyle Comfort.
Merk: Informasjon som trenger spesiell oppmerksomhet.

Indikasjoner for bruk
ADVARSEL: DET ER VELDIG VIKTIG Å KUN
VELGE DET FORESKREVNE NIVÅET AV
OKSYGEN. VALG AV OKSYGENTILFØRSELSKAL IKKE ENDRES MED MINDRE DU
HAR FÅTT BESKJED AV AUTORISERTHELSEPERSONELL Å GJØRE DET.
ADVARSEL: FØDERAL LOV (I USA) BEGRENSER
DENNE ENHETEN TIL Å SELGES TIL ELLER LEIESUT TIL EN LEGE ELLER ANNEN AUTORISERTLEVERANDØR INNEN HELSESEKTOREN.
ADVARSEL: DENNE ENHETEN SKAL IKKE BRUKES
SOM LIVSOPPRETTHOLDENDE. GERIATRISKE,PEDIATRISKE, ELLER ANDRE BRUKERE SOM ERUTE AV STAND TIL Å KOMMUNISERE UBEHAG VEDBRUK AV DENNE ENHETEN KAN HA BEHOV FOREKSTRA TILSYN. BRUKERE MED NEDSATT HØRSEL
OG/ELLER SYN KAN HA BEHOV FOR HJELP MED ÅOVERVÅKE ALARMENE.

Kontraindikasjoner for bruk
ADVARSEL: I NOEN TILFELLER KANBRUK AV OKSYGEN SOM IKKE ER
FORESKREVET VÆRE FARLIG. DENNE
ENHETEN SKAL KUN BRUKES NÅR DET
ER FORESKREVET AV LEGE.
ADVARSEL: MÅ IKKE BRUKES I NÆRHETEN AV
BRANNFARLIGE ANESTESIMIDLER.
ADVARSEL: I LIKHET MED ENHVER ELEKTRISKDREVET ENHET, KAN BRUKEREN OPPLEVEPERIODER MED NEDETID SOM FØLGE AV
STRØMBRUDD ELLER BEHOV FOR VEDLIKEHOLD AV
OKSYGENKONSENTRATOREN VED EN KVALIFISERTTEKNIKER. OKSYGENKONSENTRATOREN PASSERIKKE FOR ENHVER BRUKER SOM KAN OPPLEVEHELSEMESSIGE BIVIRKNINGER SOM FØLGE AV
SLIKE MIDLERTIDIGE AVBRYTELSER.

Retningslinjer for sikkerhet
ADVARSEL: INGEN FORM FOR
MODIFISERING AV DETTE UTSTYRETER TILLATTADVARSEL: DENNE ENHETEN LEVERER OKSYGEN
AV HØY KONSENTRASJON SOM FREMMER RASKBRENNING. TILLAT ALDRI RØYKING ELLER BRUK AVÅPEN ILD I SAMME ROM SOM (1) DENNE ENHETEN,
ELLER (2) NOE TILBEHØR MED OKSYGEN. DERSOMDENNE ADVARSELEN IKKE FØLGES KAN DET FØRETIL ALVORLIG BRANN, MATERIELL SKADE OG/ELLERPERSONSKADE ELLER DØD.

ADVARSEL: INNSTILLINGER FOR PULSDOSEMODUS
MÅ BESTEMMES INDIVIDUELT FOR HVER BRUKERETTER DERES BEHOV VED HVILE, VED FYSISKANSTRENGELSE OG UNDER REISE.

ADVARSEL: IKKE BRUK ENHETEN I ET AVGRENSET
ELLER TRANGT ROM (F.EKS. EN LITEN KOFFERT,ELLER EN VESKE) DER VENTILASJONEN KAN
VÆRE BEGRENSET. DETTE KAN FØRE TIL AT
OKSYGENKONSENTRATOREN BLIR OVEROPPHETETOG YTELSEN SVEKKES.

ADVARSEL: ENKELTE PUSTEFORSØK FRA
PASIENTENS SIDE VIL IKKE NØDVENDIGVIS UTLØSEKONSERVERINGSFUNKSJONEN.

ADVARSEL: KONSENTRATOREN BØR PLASSERESSLIK AT DEN IKKE BLIR UTSATT FOR RØYK,FORURENSNING ELLER GASSER.

ADVARSEL: FREESTYLE COMFORT SKAL IKKEBRUKES PÅ EN TRAKETOMISERT PASIENT.

ADVARSEL: I TILFELLE DET OPPSTÅR EN ALVORLIGHENDELSE MED DENNE ENHETEN, SKAL BRUKEREN
UMIDDELBART RAPPORTERE HENDELSEN TIL
LEVERANDØREN OG / ELLER PRODUSENTEN. EN
ALVORLIG HENDELSE ER DEFINERT SOM EN SKADE,DØD ELLER POTENSIELL FOR Å FORÅRSAKESKADE / DØD DERSOM DET SKULLE OPPSTÅ EN NY
HENDELSE. BRUKEREN KAN OGSÅ RAPPORTERE
HENDELSEN TIL DEN KOMPETENTE MYNDIGHETEN I
LANDET DER HENDELSEN SKJEDDE.

ADVARSEL: GRAVIDE ELLER AMMENDE KVINNERSKAL IKKE BRUKE TILBEHØRENE I DENNEHÅNDBOKEN. DE KAN INNEHOLDE FTALATER.

254 - NORPN MN232-C4 K | Bruksanvisning

FreeStyle Comfort
ADVARSEL: BRUKEN AV NOEN TYPER
TILBEHØR FOR ADMINISTRERING AV
OKSYGEN, SOM IKKE ER SPESIFISERTFOR DENNE KONSENTRATOREN, KAN
SVEKKE DENS YTELSE. ANBEFALTTILBEHØR ER VIST TIL I DENNE
BRUKERVEILEDNINGEN.
ADVARSEL: HVIS OKSYGENKONSENTRATORENHAR FALT NED, ER BLITT SKADET ELLER UTSATT
FOR VANN, KONTAKT HJEMMETJENESTEN FOR
INSPEKSJON ELLER EVENTUELL REPARASJON AV
ENHETEN. IKKE BRUK OKSYGENKONSENTRATOREN
HVIS STRØMLEDNINGEN ELLER STØPSELET ERSKADET.
ADVARSEL: NÅR OKSYGENKONSENTRATOREN
TRANSPORTERES I BIL MÅ DEN SPENNESFORSVARLIG FAST MED BELTE ELLER ANNEN
FESTEANORDNING, FOR Å UNNGÅ SKADE PÅ
ENHETEN ELLER PERSON.
ADVARSEL: FORSKRIFTER FRA US DEPARTMENT
OF TRANSPORTATION (DOT) (TRANSPORTETATEN I
USA) OG DE FORENTE NASJONER (FN), KREVER AT
BATTERIET TIL DENNE ENHETEN FJERNES UNDER
ALLE INTERNASJONALE FLYVNINGER MED RUTEFLYNÅR OKSYGENKONSENTRATOREN SJEKKES INN
SOM BAGASJE. BATTERIET MÅ OGSÅ FJERNESFRA OKSYGENKONSENTRATOREN OG PAKKESFORSKRIFTSMESSIG NÅR ENHETEN SKAL SENDES.
ADVARSEL: IKKE BRUK DETTE UTSTYRETUTEN FØRST Å HA LEST OG FORSTÅTT DENNE
VEILEDNINGEN. DERSOM DU IKKE FORSTÅR ALLEADVARSLENE OG INSTRUKSJONENE, KONTAKTUTSTYRSLEVERANDØREN FØR DU FORSØKER Å
BRUKE DETTE UTSTYRET. HVIS DU IGNORERER
DETTE KAN DET FØRE TIL PERSONSKADE ELLER
MATERIELL SKADE.
ADVARSEL: OPPSØK LEGEHJELP ØYEBLIKKELIG
HVIS DU FØLER UBEHAG ELLER OPPLEVER ENMEDISINSK NØDSITUASJON.
ADVARSEL: BRUK AV OKSYGENKONSENTRATORENVED TEMPERATURER UTENFOR DET SPESIFISERTETEMPERATUROMRÅDET KAN BEGRENSE
KONSENTRATORENS EVNE TIL Å AVGI DENSPESIFISERTE OKSYGENKONSENTRASJONEN. SE
ETTER TEMPERATURGRENSER I AVSNITTET FOR
SPESIFIKASJONER I DENNE BRUKERVEILEDNINGEN.
ADVARSEL: IKKE BRUK OLJE, FETT ELLERPETROLEUM-BASERTE, ELLER ANDRE
BRENNBARE PRODUKTER SAMMEN MEDOKSYGENKONSENTRATOREN ELLER TILBEHØRMED OKSYGEN. BARE VANNBASERTE,OKSYGEN-KOMPATIBLE KREMER ELLER SALVER
SKAL BRUKES. OKSYGEN AKSELERERER
FORBRENNINGEN AV BRENNBARE STOFFER.

ADVARSEL: PASS PÅ AT HVERKEN LUFTINNTAKET
ELLER LUFTUTTAKET BLIR BLOKKERT. IKKESLIPP ELLER FØR NOEN GJENSTANDER INN I
ÅPNINGENE PÅ UTSTYRET. DETTE KAN FØRE TIL AT
OKSYGENKONSENTRATOREN BLIR OVEROPPHETETOG YTELSEN SVEKKES.
ADVARSEL: FEIL BRUK AV BATTERIET KAN FØRETIL AT BATTERIET BLIR VARMT, TAR FYR, OG KAN
FØRE TIL ALVORLIG PERSONSKADE. FORSIKRE DEG
OM AT BATTERIET IKKE STIKKES HULL PÅ, SLÅSELLER MISTES I BAKKEN, ELLER PÅ ANNEN MÅTEBLIR UTSATT FOR KRAFTIGE SLAG ELLER STØT.
BRUKEN AV ET SKADET BATTERI KAN FØRE TILPERSONSKADE.
ADVARSEL: DU MÅ IKKE KORTSLUTTE BATTERIETS
METALLKONTAKTER MED METALLOBJEKTER, SOMNØKLER ELLER MYNTER. DETTE KAN FORÅRSAKEGNISTER ELLER STERK VARME.
ADVARSEL: NÅR OKSYGENKONSENTRATORENBRUKES UTENDØRS MED VEKSELSTRØMTILFØRSEL
MÅ DEN KUN KOBLES TIL EN JORDET STIKKONTAKTPÅ EN KURS MED JORDFEILBRYTER.
ADVARSEL: OKSYGENKONSENTRATOREN BØR
IKKE BRUKES RETT VED SIDEN AV ELLER OPPÅANNET UTSTYR. HVIS DET IKKE ER MULIG Å UNNGÅ
DETTE, BØR UTSTYRET OBSERVERES FOR Å SIKRENORMAL DRIFT.
ADVARSEL: BRUK AV ANDRE KABLER OG
ADAPTERE ENN DE SOM ER SPESIFISERT,MED UNNTAK AV KABLER OG ADAPTERE SOM
SELGES AV PRODUSENTEN TIL DET MEDISINSKELEKTRISKE UTSTYRET SOM RESERVEDELER
FOR INTERNE KOMPONENTER, KAN FØRE TILØKTE UTSLIPP AV REDUSERT IMMUNITET FOROKSYGENKONSENTRATOREN.
ADVARSEL: BRUK KUN DEN SPENNINGEN SOM ERSPESIFISERT PÅ MERKEETIKETTEN SOM ER FESTETTIL ENHETEN.
ADVARSEL: IKKE BRUK SKJØTELEDNING TIL DENNE
ENHETEN. KOBLE HELLER IKKE FOR MANGESTØPSLER I DEN SAMME STIKKONTAKTEN. BRUK AVSKJØTELEDNING KAN HA EN NEGATIV EFFEKT PÅYTELSEN TIL ENHETEN. FOR MANGE STØPSLER I ÉNSTIKKONTAKT KAN FØRE TIL OVERBELASTNING PÅ
PANELET SOM KAN FØRE TIL AT SIKRINGEN GÅR,ELLER BRANN HVIS SIKRINGEN IKKE FUNGERERSOM DEN SKAL.

PN MN232-C4 K | Bruksanvisning

NOR - 255

FreeStyle Comfort
FORSIKTIG: I tilfeller der du observerer en
alarm eller at oksygenkonsentratoren ikke
fungerer som den skal, se feilsøkingsdeleni denne brukerveiledningen. Rådfør deg
med utstyrsleverandøren hvis du ikkeklarer å løse problemet.
FORSIKTIG: Følg alle produsentens anvisninger for atgarantien ikke skal bli ugyldig.
ADVARSEL: FORSØK IKKE NOE
ANNET VEDLIKEHOLD ENN DE
MULIGE LØSNINGENE SOM ER ANGITTI DENNE BRUKERVEILEDNINGEN.
FJERN ALDRI DEKSLER. BAREUTSTYRSLEVERANDØREN ELLER ENKVALIFISERT SERVICETEKNIKER SKALFJERNE DEKSLENE ELLER REPARERE
DENNE ENHETEN.
ADVARSEL: KUN TILBEHØR ANBEFALT AV
PRODUSENTEN. BRUK AV NOE ANNET KAN
VÆRE FARLIG, FØRE TIL ALVORLIG SKADE PÅ
OKSYGENKONSENTRATOREN OG VIL GJØREGARANTIEN UGYLDIG.
ADVARSEL: PRODUSENTEN ANBEFALER EN
ALTERNATIV OKSYGENKILDE I TILFELLER SOM FOR
EKSEMPEL STRØMUTFALL, ALARMTILSTAND ELLER
MEKANISK FEIL. RÅDFØR DEG MED DIN LEGE ELLERUTSTYRSLEVERANDØR VEDRØRENDE NØDVENDIG
RESERVEUTSTYR.
ADVARSEL: PASS PÅ AT HVERKEN LUFTINNTAKET
ELLER LUFTUTTAKENE BLOKKERES. IKKESLIPP ELLER FØR NOEN GJENSTANDER INN I
ÅPNINGENE PÅ UTSTYRET. DETTE KAN FØRE TIL AT
OKSYGENKONSENTRATOREN BLIR OVEROPPHETETOG YTELSEN SVEKKES.
ADVARSEL: RØYKING SAMTIDIG SOM BRUK AV
OKSYGEN ER DEN MEST VANLIGE ÅRSAKEN TIL
BRANNSKADER OG RELATERTE DØDSFALL. DU MÅFØLGE DISSE SIKKERHETSADVARSLENE:ADVARSEL: TILLAT ALDRI RØYKING, STEARINLYS
ELLER ÅPEN ILD I SAMME ROM SOM DENNE
ENHETEN, ELLER NOE TILBEHØR MED OKSYGEN.
ADVARSEL: Å RØYKE MENS MAN BRUKER ENOKSYGENKANYLE KAN FØRE TIL BRANNSKADE IANSIKTET OG MULIG DØD.
ADVARSEL: HVIS DU RØYKER KAN DISSE
3 TRINNENE REDDE LIVET DITT: SKRU AV
OKSYGENKONSENTRATOREN, TA UT KANYLEN,
OG GÅ UT AV ROMMET DER DENNE ENHETEN ER
PLASSERT.

256 - NORPN MN232-C4 K | Bruksanvisning

ADVARSEL: SKILT MED TEKSTEN «RØYKINGFORBUDT – OKSYGEN I BRUK» MÅ VÆRE TYDELIG
OG SYNLIG I HJEMMET, ELLER DER OKSYGEN ER IBRUK. PASIENTER OG DERES PLEIERE MÅ VÆREINFORMERT OM FARENE VED Å RØYKE I NÆRHETEN
AV MEDISINSK OKSYGEN, ELLER NÅR MEDISINSKOKSYGEN ER I BRUK.
ADVARSEL: IKKE BRUKOKSYGENKONSENTRATOREN I NÆRHETEN AV
BRANNFARLIGE GASSER. DETTE KAN RESULTERE IEKSPLOSIV BRANN SOM KAN FØRE TIL MATERIELL
SKADE, PERSONSKADE ELLER DØD.
ADVARSEL: HVIS KANYLEN BLIR TATT UT OG
LAGT PÅ KLÆR, SOFAER ELLER ANDRE MYKE
MATERIALER NÅR DEN TAS UT, VIL DET KUNNE
FORÅRSAKE ØYEBLIKKELIG BRANN HVIS DETUTSETTES FOR EN SIGARETT, VARMEKILDE ELLERÅPEN ILD.
ADVARSEL: LEGG ALDRI FRA DEG EN NESEKANYLE
PÅ ELLER UNDER KLÆR, SENGETØY ELLER
STOLPUTER. HVIS ENHETEN ER SLÅTT PÅ, MEN
IKKE BRUKES, VIL OKSYGENET GJØRE MATERIALET
BRENNBART. SØRG FOR AT STRØMMEN ER AV NÅR
OKSYGENKONSENTRATOREN IKKE ER I BRUK.
ADVARSEL: OKSYGENKONSENTRATOREN MÅVÆRE AV OG KOBLET FRA EKSTERN STRØM FØR
FJERNING AV BATTERIPAKKEN.
ADVARSEL: SMÅ DELER KAN UTGJØRE
EN KVELNINGSFARE.
ADVARSEL: UNNGÅ KONTAKT MED UTLUFTINGEN I
MER ENN 10 SEKUNDER, DA DET KAN VÆRE VARMT.
ADVARSEL: VIND ELLER STERK TREKKKAN PÅVIRKE NØYAKTIG LEVERING AV
OKSYGENBEHANDLING.
ADVARSEL: DETTE PRODUKTET KAN EKSPONEREDEG FOR KJEMIKALIER, INKLUDERT NIKKEL, SOMIFØLGE STATEN CALIFORNIA KAN FORÅRSAKEKREFT. FOR MER INFORMASJON, GÅ TIL
WWW.P65WARNINGS.CA.GOV.
FORSIKTIG: Koble alltid strømtilførselen fra
stikkontakten i veggen fra før kontakten på
oksygenkonsentratoren kobles fra.
FORSIKTIG: Ikke plasser oksygenkonsentratoren slik
at det er vanskelig å få tilgang til strømledningen.

FreeStyle Comfort
FORSIKTIG: Plasser alltid oksygenforsyningsslangenog strømledningene på en måte som forebygger
snublefare eller mulig utilsiktet kvelning.
FORSIKTIG: Når oksygenkonsentratoren brukes i bil,båt eller et annet sted med likestrømtilførsel, sørg forat kjøretøyet startes opp før oksygenkonsentratoren
kobles til. Dersom det unnlates å følge disse
instruksjonene kan det resultere i at strømtilførselen
ikke gir strøm til oksygenkonsentratoren.
FORSIKTIG: Når bilen som du bruker
oksygenkonsentratoren i er slått av, kobler dufra og fjerner enheten fra bilen. Oppbevar aldri
oksygenkonsentratoren i en veldig varm eller kaldbil, eller i andre lignende miljøer med høy eller lav
temperatur.
Merk: Bærbart og mobilt RF-kommunikasjonsutstyr kanpåvirke medisinsk elektrisk utstyr.
FORSIKTIG: La IKKE
oksygenkonsentratoren eller
strømtilførselen være tilkoblet kjøretøyet
hvis tenningen er slått AV. Det kan tømme
bilens batteri for strøm.
FORSIKTIG: Drift eller oppbevaring av
oksygenkonsentratoren utenfor dens normale
temperaturområde og omgivelsestrykkområde (høyde)kan påvirke ytelsen og senke driftstiden for batteriet,
og/eller øke ladetiden. Se avsnittet for Spesifikasjoner
i denne bruksanvisningen for temperaturgrenser for
lagring og drift.
FORSIKTIG: For oksygenkonsentratorer som er utstyrt
med batterier: Lagres kjølig og tørt for å sikre lang
levetid for batteriet. Lagring av oksygenkonsentratoren
over lengre tid ved høy temperatur eller med etfulladet, eller fullstendig tomt batteri kan redusere
batteriets samlede levetid. Ikke prøv å åpne batteriet.
Det er ingen innvendige deler som kan repareres. Holdbatteriene unna barn.
FORSIKTIG: BRUK KUN batterier levert av
produsenten. For riktig avhending av batteriene,kontakt utstyrsleverandøren eller lokale myndigheter
for avhendingskrav.

Merk: Påse at kanylen sitter helt inne og godt fast.
Dette sikrer at oksygenkonsentratoren kan utføre
korrekt registrering av innånding for oksygentilførsel.
Ved innånding bør du høre eller føle oksygen flyte til
utstikkerne på nesekanylen.
Merk: Litiumbatterier kan permanent miste sin kapasitet
når de eksponeres for ekstremt varme temperaturer med
batteriene fullt oppladet eller helt utladet. Ved langvarig
lagring anbefales det at batteriene er ladet 25 til 50 % og
oppbevares i et temperaturområde på -20 °C – 20 °C (-4°F – 68 °F).
Merk: Bruken av noen typer tilbehør foroksygenadministrering som ikke er spesifisert for denne
oksygenkonsentratoren kan svekke dens ytelse. Anbefalt
tilbehør er vist til i denne brukerveiledningen.
Merk: Ikke bruk oksygenkonsentratoren uten at
luftinntaksfilteret er på plass. Hvis enheten er levert
med et ekstra filter, sett inn dette «reservefilteret» før
du rengjør det skitne filteret. Rengjør det skitne filteret i
varmt vann med såpe, tørk deretter grundig før bruk.
Merk: Produsenten anbefaler ikke sterilisering av dette
utstyret.
Merk: Det er aldri fare for å utarme oksygenet i et rom
hvor du bruker oksygenkonsentratorenheten.
ADVARSEL: OKSYGENKONSENTRATORENSKAL IKKE BRUKES ELLER KOBLES TIL
EKSTERNSTRØM (AC/DC) UTEN AT ET
BATTERI ER SATT INN. IKKE FORSØK ÅSLÅ PÅ ENHETEN ELLER KOBLE DEN TIL
EKSTERN STRØM NÅR BATTERIPAKKENER FJERNET.
BATTERIPAKKEN SKAL BARE FJERNES NÅRENHETEN ER SLÅTT AV OG ER KOBLET FRASTRØMFORSYNINGEN. IKKE FORSØK Å FJERNEBATTERIPAKKEN MENS ENHETEN FORTSATT ER PÅ
ELLER KOBLET TIL STRØMFORSYNINGEN.
Merk: Hvis oksygenkonsentratoren har vært lagret
i en lengre periode utenfor området for normal
driftstemperatur som er vist i tabellen ovenfor, skal dula enheten komme tilbake til normal driftstemperatur før
den slås på.

PN MN232-C4 K | Bruksanvisning

NOR - 257

FreeStyle Comfort

Spesifikasjoner

FreeStyle Comfort
Modeller (med ett batteri)

10,0 x 7,3 x 3,1 in. (25,4 x 18,5 x 7,9 cm)

Mål (med dobbelt batteri)

11,0 x 7,3 x 3,1 in. (27,9 x 18,5 x 7,9 cm)

Vekt (med én batteripakke)

5 lbs (2,3 kg)

Vekt med (dobbel batteripakke)

6 lbs (2,7 kg)

Nominelt lydnivå***39,93 dB (A) ± 0,42 dB (A) @ 2

Innstillinger for strømning

Pulsinnstillinger 1, 2, 3, 4, 5

O2-effektinnstilling

Innstilling 1 – 210 mL/min
Innstilling 2 – 420 mL/min
Innstilling 3 – 630 mL/min
Innstilling 4 – 840 mL/min
Innstilling 5 – 1050 mL/min(@ 20bpm)

Oksygenkonsentrasjon*

90 % (+5,5 %; -3 %)Oksygenovervåkning

Ja

Maksimalt utløpstrykk<30 PSI

Standard doseringsfølsomhet> -0,1 cm H2O (justerbar)

Vekselstrøm

100–240 VAC, 50-60 Hz, 120W Max

Likestrøm

11–18 VDC (10 maks ampere)

Batteriladetid

Enkeltbatteri: 3,5 timer
Dobbelt batteri: 6,0 timer

Batteripakkens varighet

Varighet av enkeltbatteri: Innstilling 2 – inntil 4 timer
Varighet av dobbelt batteri: Innstilling 2 – inntil 8 timer

Driftstemperatur**
Batteritype
Batterikapasitet

5 °C til 40 °C (41 °F til 104 °F)
Litiumion
Enkelcelle: 6700 mAh; Dobbeltcelle: 13400 mAh

Transports- og oppbevaringstemp. (enhet)-25 °C til +70 °C opp til 10000 fot (3048 m)

Transports- og oppbevaringstemp. (batteri)-25 °C til +70 °C opp til 10000 fot (3048 m)

Høyde ved drift**
Luftfuktighet ved drift-381 til 3048 m (-1250 til 10 000 fot) (testet til 700 – 1060 hPa)
15 - 95 % relativ luftfuktighet (ikke-kondenserende)

Lagringsfuktighet

0 - 90% ikke-kondenserendeBegrenset garanti

3 års enhet
1 års batteri og strømforsyning* Basert på et atmosfærisk trykk på 14,7 psi (101 kPa) ved 21 °C (70 °F)** Enheten vil fungere i alle miljøer så lenge miljøet har de angitte temperatur- og fuktighetsområdene. Drift utenfor disse
driftsspesifikasjonene kan begrense konsentratorens evne til å oppfylle oksygenkonsentrasjonsspesifikasjonen ved høyere liters
strømningshastigheter. Oppvarmingsskjermen kan bli vist vedvarende til konsentratoren har nådd målkonsentrasjonen eller opptil
15 minutter. Fortsett å bruke enheten i løpet av denne tiden.
*** Lydnivå målt per testmetode Nr. 14-1 10/2018 MDS-Hei.
Forventet levetid er minst fem år. Den forventede holdbarhetstiden til filtrene er ca. ett år dersom konsentratoren ikke blir brukt ihele denne perioden.

258 - NORPN MN232-C4 K | Bruksanvisning

FreeStyle Comfort
Instruksjoner for bruk
1. Finn og plasser FreeStyle Comfort på et passende
sted for bruk.
a. Konsentratoren skal være godt ventilert slik at
luftinntaks- og luftuttaksfiltrene ikke er blokkert.
b. Konsentratoren skal posisjoneres slik at allehørbare og visuelle indikatorer og alarmer er
lette å se og høre.
c. Sørg for at luftinntaksfiltrene er på plass
før du bruker enheten. Ta kontakt med
utstyrsleverandøren din hvis enheten mangler
et filter.
2. Installer en batteripakke i konsentratoren.
a. Juster batteriet med bunnhuset fra midten av
FreeStyle Comfort.

b. Skyv batteriet innover til du hører et klikk for å
bekrefte at det er låst på plass.

3. Plugg inn strømledningen for veksel- eller likestrøm
i strømkontakten til høyre på forsiden av enheten.

4. Koble en nesekanyle til oksygenuttakets hake. Sørg
for at koblingen sitter godt fast før drift.
MERK: Bruk av andre kanyler enn de som er anbefalt kanpåvirke ytelsen.

Merk: Til utstyrsleverandør: Følgende tilbehør foroksygenadministrasjon anbefales for bruk sammen med
FreeStyle Comfort:• Nasal kanyle med 2,1 m (2 fot) rør (Maksimal LPM: 6
LPM): CAIRE-delenummer 5408-SEQ (• Ytterligere informasjon om anbefalt tilbehør er
tilgjengelig online på www.caireinc.com.

PN MN232-C4 K | Bruksanvisning

NOR - 259

FreeStyle Comfort
5. Slå på FreeStyle Comfort ved å trykke på på/avknappen og velg foreskrevet strømningsinnstillingved å bruke knappene for økning (+) eller reduksjon(-) av oksygentilførsel.

Innstilling for autoSATpulsvolum (mL)*
Pustefrekvens

1

2

3

4

5

15

10,5

21,0

20

10,5

21,0

31,5

42,0

52,5

31,5

42,0

52,5
42,0

25

10,5

21,0

31,5

42,0

30

10,5

21,0

31,5

35,0

35,0

35

10,5

21,0

30,0

30,0

30,0

10,5

21,0

26,3

26,3

26,3

40

*Toleranse er +/- 15 %Innstilling for ikke-autoSAT
pulsvolum (mL)*

En alarm vil lyde kort for å indikere at enheten erslått på.
Merk: Når FreeStyle Comfort slås på for først gang, er
standard oksygeninnstilling 2. Hvis innstillingen endres,aktiveres enheten i oksygeninnstillingen som ble bruktsist.
6. Posisjoner nesekanylen over ansiktet i henhold til
produsentens instruksjoner levert med kanylen, og
pust normalt. Du skal føle et pust av oksygen levert
gjennom nesekanylen hver gang du puster inn.
7. For å endre oksygeninnstillingen, trykk på knappene
for økning (+) eller reduksjon av (-) oksygentilførsel
for å velge ønsket nivå. Det er normalt å høre en
forskjell i lyd når du endrer strømningsinnstillinger.

Pustefrekvens

1

2

3

4

5

15

14,0

28,0

42,0

56,0

70,0

20

10,5

21,0

31,5

42,0

52,5
42,0

25

8,4

16,8

25,2

33,6

30

7,0

14,0

21,0

28,0

35,0

35

6,0

12,0

18.0

24,0

30,0

40

5.3

10,5

15,8

21,0

26,3

8. Slå av FreeStyle Comfort ved å trykke på på/avknappen.
Merk: Skjermen vil gå inn i standbymodus og dimmes
etter 20 sekunder. Trykk på enhver knapp for å vekke
skjermen fra standbymodus.
9. For avanserte funksjoner, se instruksjoner ommenynavigering

SPESIELLE PRODUKTEGENSKAPER
autoSAT®
CAIREs egenutviklede autoSAT-funksjon lar FreeStyle
Comfort levere et konstant volum oksygen (bolus) per
pust, opp til en maksimal pustefrekvens. Når brukerens
pustefrekvens øker under aktiviteter som turgåing,justerer autoSAT-funksjonen enhetsdriften for å
opprettholde et innstilt bolusvolum.
UltraSense®
Denne sensitive pulsdoseutløseren oppdager
trykkendringen når du puster, og frigjør oksygen når duinhalerer.
autoDOSE™
Hvis en «ingen pust-alarm» oppstår, vil denne
funksjonen pulsere oksygen ved 20 pust per minutt
ved enhetens gjeldende strømningsinnstilling til
alarmtilstanden er løst.
260 - NORPN MN232-C4 K | Bruksanvisning

FreeStyle Comfort
Strømforsyninger
FreeStyle Comfort kan motta strøm på tre måter:• Oppladbar batteripakke• Vekselstrøm• Likestrøm
Informasjon om driften av hver av disse komponentene
er i følgende avsnitt.
ADVARSEL: BRUK KUNSTRØMFORSYNINGENE SOM
ER BESKREVET I DENNE
BRUKSANVISNINGEN. BRUKAV STRØMFORSYNINGER OG
BATTERIPAKKER SOM IKKE ER FRACAIRE KAN FØRE TIL SKADE PÅ
ENHETEN OG FEILAKTIG DRIFT.

FJERNE BATTERIET

Trykk batteriutløserknappen nedover, vekk fra
FreeStyle Comfort. Mens du HOLDER knappen,
skyver du batteriet ut ved å dra det vekk fra
konsentratoren.
ADVARSEL: OKSYGENKONSENTRATORENMÅ VÆRE AV OG KOBLET FRAEKSTERN STRØM FØR FJERNING AV
BATTERIPAKKEN.

BATTERIPAKKE

FreeStyle Comfort kan drives av den oppladbare
batteripakken som følger med enheten. Det er ingen
driftsbegrensninger for konsentratoren når den brukes
med batteripakken. Alle strømningshastigheter ertilgjengelige på batteristrøm.

KONTROLLERE LADINGEN AV
BATTERIPAKKENINSTALLERE BATTERIET

Juster batteriet med bunnhuset fra midten av FreeStyle
Comfort. Skyv batteriet innover til du hører et klikk forå bekrefte at det er låst på plass.

For å kontrollere batteriladningen når batteriet er
installert i FreeStyle Comfort, se batteriikonet og den
gjenværende tiden på displayet.
80–100 %60–80 %

40–60 %

20–40 %20 %PN MN232-C4 K | Bruksanvisning

NOR - 261

FreeStyle Comfort
Merk: Hvis batteriet er i ferd med å lades, vises enpluggindikator i stedet for gjenværende ladeprosent.
For å kontrollere batteriladingen når den ikke er
installert i FreeStyle Comfort, legg den flat og trykk
på TEST-knappen. Indikatorlyset for batterimåler
(25–100 %) vil tennes til høyre for TEST-knappen forå indikere nivået på lading av batteripakken.

VARIGHET BATTERIDRIFT

Når ekstern vekselstrøm eller likestrøm kobles fra,vil konsentratoren automatisk skifte til å bruke
batteristrøm hvis et ladet batteri er installert. Tabellen
nedenfor viser typisk varighet for en ny batteripakke.
Varigheten til batteripakken holder seg konstant,uavhengig av pustehastigheten din.
Innstilling
av puls

Varighet av
enkeltbatteri(8 celler)

Varighet av dobbelt
batteri (16 celler)1

inntil 8 timer

inntil 16 timer

2

inntil 4 timer

inntil 8 timer

3

inntil 3 timer

inntil 6 timer

4

inntil 2,25 timer

inntil 4,5 timer

5

inntil 2 timer

inntil 4 timer

TYPISK BATTERILADNINGSTID
FØRSTE LADING

Det nye batteriet som følger med din FreeStyle
Comfort er bare delvis ladet når det sendes fra
fabrikken. Før første gangs bruk skal du ladebatteripakken helt opp med bruk av enten vekselstrøm
eller likestrøm.

GENERELL INFORMASJON LADINGFreeStyle Comfort vil lade opp batteripakken hvergang den installeres i konsentratoren og konsentratoren
er koblet til en ekstern vekselstrøm- eller
likestrømkilde (unntatt på et fly).
Batteriet er fulladet når det viser 100 %.
Alternativt kan batteripakken lades ved bruk av den
valgfrie bordladeren.

262 - NORPN MN232-C4 K | Bruksanvisning

Fullstendig oppladning av en batteripakke med
et fullstendig utladet batteri er 3,5 timer for et
enkeltbatteri og 6,0 timer for et dobbeltbatteri.
Merk: Batteriet vil kanskje ikke bli ladet med en gang hvis
batteriet er fullstendig utladet.

BATTERILEVETIDFreeStyle Comfort-batterier er utformet for å varei minst 300 oppladnings-/utladningssykluser.

FreeStyle Comfort
Veksel- og likestrømtilførsel
FreeStyle Comfort leveres med vekselstrøms- og
likestrømstilførsel som standard.
Vekselstrømtilførselen inkluderer følgende:a. Vekselstrømkloss som opererer mellom
100‑240 volt, 50-60 Hz.
b. Vekselstrømledning for å koble strømklossen
til tilsvarende strømuttak.

Likestrømtilførselen består av en ledning som plugges
direkte inn i enheten og likestrømuttaket.

VEKSELSTRØM

2. Koble vekselstrømledningen for vekselstrømklossen.

Vekselstrøm er til bruk hjemme eller alle stederhvor standard stikkontakter med vekselstrøm er
tilgjengelig. FreeStyle Comfort er fullt ut funksjonell
med vekselstrøm. FreeStyle Comfort vil også lade opp
batteripakken hver gang vekselstrøm er tilgjengelig(unntatt på et fly). Batteripakken vil lades opp igjen
uansett om FreeStyle Comfort er slått på eller av.
ADVARSEL: IKKE KOBLE TIL
EKSTERN STRØM UTEN AT EN
BATTERIPAKKE ER INSTALLERT IOKSYGENKONSENTRATOREN.
For å koble FreeStyle Comfort til vekselstrøm:
1. Koble vekselstrømklossen til strømkontakten
nederst på forsiden av FreeStyle Comfort. Sørg for å
trykke innover, slik at koblingen er godt festet.

3. Koble den motsatte enden av strømledningen forvekselstrøm til stikkontakten i veggen eller en annen
passende stikkontakt.

PN MN232-C4 K | Bruksanvisning

NOR - 263

FreeStyle Comfort
4. Slå på FreeStyle Comfort ved å trykke på på/avknappen og velg foreskrevet strømningsinnstillingved å bruke knappene for økning (+) eller reduksjon(-) av oksygentilførsel.

Merk: Hvis batteripakken må lades opp, vil batteriikonet
på displayet vise en eneste oransje stolpe.

LIKESTRØM

Likestrøm brukes i utstyrskontakter i biler, båter eller
andre motorkjøretøy. FreeStyle Comfort er fullt ut
funksjonell med likestrøm. FreeStyle Comfort vil ogsålade opp igjen batteripakken, hvis nødvendig, hvergang likestrøm er tilgjengelig. Batteripakken lades opp,både når enheten er av eller på og i bruk.
ADVARSEL: IKKE KOBLE TIL
EKSTERN STRØM UTEN AT EN
BATTERIPAKKE ER INSTALLERT IOKSYGENKONSENTRATOREN.
ADVARSEL: IKKE BRUK ENHETEN PÅ 12 VDC
BILUTTAK MENS KJØRETØYET ER SLÅTT AV.
1. Start motorkjøretøyet.
2. Koble likestrømforsyningen til strømkontakten
nederst til høyre på forsiden av FreeStyle Comfort.
Sørg for å trykke innover, slik at koblingen er godt
festet.

264 - NORPN MN232-C4 K | Bruksanvisning

3. Koble den motsatte enden av likestrømsledningen
til kontakten i bilen eller en annen utstyrskontakt for
likestrøm.

4. Slå på FreeStyle Comfort ved å trykke på på/avknappen og velg foreskrevet strømningsinnstillingved å bruke knappene for økning (+) eller reduksjon(-) av oksygentilførsel.
Merk: Hvis batteripakken må lades opp, vil batteriikonet
på displayet vise en eneste oransje stolpe.

FreeStyle Comfort
Rengjøring, pleie og rutinemessig
vedlikehold

VEDLIKEHOLD AV BATTERIPAKKE

RUTINEMESSIG VEDLIKEHOLD

Ikke eksponer batteriet for åpne flammer.

Forebyggende vedlikehold
Enheten er vedlikeholdsfri. Filtre inne i enheten er
ment å vare levetiden til enheten. Hvis enheten ikke
fungerer som den skal, skal enheten returneres tilleverandøren for reparasjon. Vedlikehold av indre
komponenter inne i kabinettet på FreeStyle Comfort
må utføres av oksygenleverandøren.
Operatøren er ansvarlig for følgende
vedlikeholdsprosedyrer.
• Ukentlig – Rengjøre luftinntaksfiltere
ADVARSEL: IKKE FORSØK Å ÅPNEELLER FJERNE PLASTDEKSLENE TIL
OKSYGENKONSENTRATOREN. ENHETENSKAL KUN ÅPNES AV EN KVALIFISERTSERVICETEKNIKER.

LUFTINNTAKSFILTERE

Luft går inn i FreeStyle Comfort gjennom etluftinntaksfilter plassert nederst på forsiden av
oksygenkonsentratoren. Dette filteret hindrer støv og
andre store partikler i luften fra å komme inn i enheten.
Luftinntaksfiltrene skal rengjøres minst én gang iuken. Oksygenleverandøren vil kanskje råde deg tilå rengjøre dem oftere, avhengig av driftsforholdene.
Følg de følgende trinnene for riktig rengjøring av
luftinntaksfiltrene.
1. Fjern de to filtrene fra forsiden av FreeStyle
Comfort ved å trykke opp på den tilsvarende tappen
og trekke den utover.

Ikke la batteripakken bli eksponert for væsker.
Ikke forsøk å demontere eller åpne batteripakken.

Ikke mist batteripakken.
Oppbevar batteripakken utenfor barns rekkevidde.
Hvis du ser at batteripakken virker skadet, ta kontakt
med oksygenleverandøren før bruk.
Ikke plasser batteriet i eller i nærheten av en
mikrobølgeovn eller andre apparater for matlaging.

LAGRING AV BATTERIPAKKEN

Batteriet til oksygenkonsentratoren trenger ikke å lades
helt ut før den lades opp igjen. Det anbefales å ladeopp batteriet til oksygenkonsentratoren etter hver bruk.
Ikke legg batteriet på et sted som eksponeres for sterk
varme, som i direkte sollys eller i en bil. Dersom dette
gjøres kan det svekke batteriets ytelse og brukstid. Det
kan også føre til at batteriet genererer varme, røyk,
tar fyr eller eksploderer. Batteripakken skal lagres
under stabile forhold, kjennetegnet av lav fuktighet(mindre enn 70 % RF) og fritt for etsende gasser. For
å forhindre rust skal du unngå forhold som kan skape
kondensering, som for eksempel raske svingninger iomgivelsestemperaturen.
Dersom batteriet skal lagres i en måned eller
mer, anbefales det at batteriet er helt utladet eller
delvis oppladet mellom 50 % og 75 % (to av tre
LED-er). Når det er oppladet fjerner du batteriet fra
oksygenkonsentratoren.

2. Vask begge filtrene i en varm løsning av såpe og
vann.
3. Skyll filtrene nøye og fjern overflødig vann med
en myk, lofri klut. Forsikre deg om at filtrene er helt
tørre før du setter det på plass igjen.
4. Installer de rene, tørre filtrene på nytt.
ADVARSEL: IKKE BRUKOKSYGENKONSENTRATOREN UTEN AT
LUFTINNTAKSFILTERET ER PÅ PLASS.

PN MN232-C4 K | Bruksanvisning

NOR - 265

FreeStyle Comfort
RENGJØRING

Slå AV enheten og koble fra vekselstrøm eller
likestrøm før enhver rengjøring. IKKE spray direktepå det ytre dekselet. Bruk en fuktig (ikke våt) kluteller svamp. Spray på kluten eller svampen med mildt
husholdningsrenser til rengjøring av kabinettet og
strømforsyningene. Fortsett som angitt av produsenten
av rengjøringsmiddelet.
ADVARSEL: FOR Å FOREBYGGEUTILSIKTET ELEKTRISK STØT OG
BRANNFARE, SLÅ AV ENHETEN
OG KOBLE STRØMLEDNINGENFRA STIKKONTAKTEN FØR DU
RENGJØR ENHETEN. DET ER BAREUTSTYRSLEVERANDØREN ELLER ENKVALIFISERT TEKNIKER SOM SKALFJERNE DEKSLENE ELLER REPARERE
ENHETEN.
ADVARSEL: UTVIS FORSIKTIGHET FOR Å
FORHINDRE AT OKSYGENKONSENTRATOREN OG
BATTERIPAKKEN BLIR VÅT. VÆSKER MÅ IKKE
KOMME INN I ENHETEN. DETTE KAN FØRE TIL EN
FUNKSJONSFEIL ELLER AT ENHETEN SLÅR SEG AV,
OG FØRER TIL ØKT RISIKO FOR ELEKTRISK STØTELLER BRANNSKADER.
ADVARSEL: IKKE BRUK VÆSKE DIREKTE PÅENHETEN. EN LISTE OVER UØNSKEDE KJEMISKE
MIDLER INKLUDERER, MEN ER IKKE BEGRENSET
TIL, FØLGENDE: ALKOHOL OG ALKOHOLBASERTEPRODUKTER, KONSENTRERTE KLORBASERTEPRODUKTER (ETYLENKLORID), OG OLJEBASERTEPRODUKTER. DISSE SKAL IKKE BRUKES TIL Å
RENGJØRE PLASTHUSET PÅ KONSENTRATOREN,
DA DE KAN SKADE ENHETEN.
ADVARSEL: KABINETTET, KONTROLLPANELET OGSTRØMLEDNINGEN SKAL BARE RENGJØRES MED
ET MILDT VASKEMIDDEL PÅFØRT MED EN FUKTIG(IKKE VÅT) KLUT ELLER SVAMP, OG DERETTERSKAL ALLE OVERFLATES TØRKES MED EN TØRRKLUT. IKKE LA NOE SLAGS VÆSKE KOMME INN I
KONSENTRATOREN. VÆR SPESIELT NØYE MEDOKSYGENUTLØPET FOR KANYLETILKOBLINGENSLIK AT DETTE HOLDES FRI FOR STØV, VANN OG
PARTIKLER.

266 - NORPN MN232-C4 K | Bruksanvisning

ADVARSEL: BRUK EN FUKTIG (IKKE VÅT) KLUT
ELLER SVAMP TIL Å RENGJØRE BATTERIPAKKEN.
SPRAY FØRST ET MILDT VASKEMIDDEL PÅ KLUTEN
ELLER SVAMPEN, OG RENGJØR DERETTER UTSIDEN
AV BATTERIPAKKEN.
ADVARSEL: IKKE RENGJØR MED AUTOMATISERTVASKESYSTEM.

UTSKIFTING AV KANYLE

Følg alltid kanyleprodusentens anvisninger for riktig
bruk. Bytt nesekanylen eller oksygenslangen hver
3. til 6. måned (hver 3. måned ved daglig bruk) i
henhold til produsentens anbefaling som anbefalt av
kanyleprodusenten eller oksygenleverandøren. Dinlege eller oksygenleverandør vil gi deg informasjon om
rengjøring og utskifting.
Ytterligere materiell for utskifting er tilgjengelig
gjennom din oksygenleverandør.

FreeStyle Comfort
Alarmtilstander
Prioritetsnr.

Alarmbeskrivelse

1

Systemfeil, høy
batteritemperatur

Display

Lydsignal

Flere detaljer / Foreslått tiltak

1 pipetone
hvert
20 sek.

Enheten vil avgi en alarm når
batteritemperaturen nærmer seg
terskelen for sikkerhetsavbrudd.
Enheten slår seg av når terskelen er
nådd. Hvis ekstern strøm er tilkoblet,
vil alarmskjermen vises og lydalarmen
fortsette i 2 minutter.
Test med batteri ved romtemperatur.
Kontakt leverandør hvis problemetvedvarer.

2

Systemfeil, høy
temperatur

1 pipetone
hvert
20 sek.

Enheten vil avgi en alarm når deninterne temperaturen nærmer seg
terskelen for sikkerhetsavbrudd.
Hvis terskelen er nådd, vil enhetenslå seg av, alarmskjermen vises og
lydalarmen fortsette i 2 minutter.
Kontroller at det ikke er noen
blokkeringer på enhetens luftehull.
Kontakt leverandør hvis problemetvedvarer.

3

Systemfeil, høyt
trykk

1 pipetone
hvert
20 sek.

Hvis terskelen er nådd, vil enhetenslå seg av, alarmskjermen vises og
lydalarmen fortsette i 2 minutter.
Kontakt leverandør hvis problemetvedvarer.

4

Ingen pust oppdaget

1 pipetone
hvert
30 sek.

Ingen pust oppdaget på 15 sekunder.
Aktivert etter første pust eller når
oppvarmingsskjerm slås av, avhengig
av hva som skjer før.
Sjekk kanylen for knekker eller
lekkasjer. Øk innstillingen forfølsomhet. Kontakt leverandør hvis
problemet vedvarer.

5

6

Driftstans ved lavt
batterinivå

Oppvarming

1 pipetone
hvert
30 sek.

Alarmen lyder til batteriet går tomt for
strøm og enheten slår seg av.

ingen

Enheten starter med automatisk
dosering til et pust oppdages eller
oppvarming slutter, avhengig av hva
som skjer først.

Lad batteriet.

PN MN232-C4 K | Bruksanvisning

NOR - 267

FreeStyle Comfort
Prioritetsnr.

Alarmbeskrivelse

Lydsignal

Flere detaljer

7

Lav konsentrasjon

Display

1 pipetone
hvert
30 sek.,i opp til
5 minutter

Kontakt leverandør hvis problemetvedvarer.

8

Systemfeil,
oksygentilførsel
(sviktende OCDventil, blokkering av
gassbane)

1 pipetone
hvert
30 sek.

Kontakt leverandør hvis problemetvedvarer.

9

Varsel ved lavt
batterinivå

1 pipetone,
gjentasikke

Lad batteriet.

10

Pustefrekvens
overskredet

1 pipetone
hvert
60 sek.

Pustefrekvens overskrider 40 bpm.

11

Systemfeil, viftefeil

ingen

Kontakt leverandør hvis problemetvedvarer.

12

Systemfeil, lavt trykk

ingen

Kontakt leverandør hvis problemetvedvarer.

13

Systemfeil, lav
temp.

ingen

La enhet varmes opp til
romtemperatur. Kontakt leverandør
hvis problemet vedvarer.

Info

Oppstart

1 infopip

Skjermen vises når enheten slås på,
og vil ett avgi pip for å bekrefte atsummeren fungerer

Hvis dosen utløses ved en feil, mådu redusere følsomhetsinnstillingen.
Kontakt leverandør hvis problemetvedvarer.

Oppstartsskjermen viser gjeldende
fastvarerevidering nederst til høyre.

268 - NORPN MN232-C4 K | Bruksanvisning

FreeStyle Comfort
Prioritetsnr.

Alarmbeskrivelse

Info

Normal kjøreskjerm
i autoSAT-modus(ingen likestrøm
tilkoblet)

Display

Lydsignal
ingen

Info

Normal kjøreskjerm
med autoSAT
deaktivert (ingenlikestrøm tilkoblet)

Ingen

Info

Innstilling endret

1 infopip

Info

Plugg inn ekstern
strøm i avslått
tilstand

ingen

Info

Plugg inn ekstern
strøm i driftstilstand

ingen

Info

Koble fra ekstern
strøm i avslått
tilstand1 infopip

Info

Koble fra ekstern
strøm i driftstilstand(med batteri tilstede)

1 infopip

Flere detaljer

Viser CAIRE-skjermen i 2 sekunder,
etterfulgt av batteri- og støpselstatus i
5 sekunder før skjermen slås av

Vis CAIRE-skjermen i 2 sekunder og
slå deretter skjermen av.

PN MN232-C4 K | Bruksanvisning

NOR - 269

FreeStyle Comfort
Prioritetsnr.

Alarmbeskrivelse

Lydsignal

Flere detaljer

Info

I standbymodus– når opp- eller nedknappen trykkes,
viser displayet
batteri- eller
ladetilstanden

ingen

Skjermen vises 5 sek. før den slukkes

Info

Flymodus på
batteristrøm

1 langvarig
pip

Info

Flymodus med
strømledning
tilkoblet

1 langvarig
pip

270 - NORPN MN232-C4 K | Bruksanvisning

DisplayFreeStyle Comfort
Instruksjoner for menynavigering
Du kan gå inn i menyen i standby- eller kjøremodus.
I kjøremodus lukkes menyen automatisk hvis en
alarmtilstand oppstår. Hvis ingen knapper trykkes i 20
sekunder, avsluttes menyen automatisk. I ventemodus
må du kanskje trykke på menyknappen to ganger for å
vekke enheten, og deretter gå inn i menyen.
Der er fire menyskjermer:
1. Statusskjerm: Viser revisjonsnummer, kjøretid,
batteriutladningssykluser for fastvare
2. Flymodus (PÅ/AV)

Flymodus

3. Følsomhet: justerbar (1–5)
4. autoSAT (PÅ/AV)

Følsomhet

Eksempel på statusskjerm

BRUK AV INNSTILLINGSMENYDu går inn i innstillingsmenyen ved å trykke på Meny/
Select-knappen
i kjøre- eller standbymodus.
For å rulle gjennom menyene, trykk på.

autoSAT

StatussidePN MN232-C4 K | Bruksanvisning

NOR - 271

FreeStyle Comfort
STATUS SIDE
Statussiden for Freestyle Comfort viser fastvareversjon, kjøretid og antall batterisykluser på detvedlagte batteriet.

Flymodus på bekreftetFor å slå av flymodus, gjenta skrittene ovenfor mens
du veksler til AV.
Trykk pånår skjermen viser «Neste»-symbolet forå flytte menysiden til neste skjermalternativ.

Statusside

Løpetimer vises i hele timer.
Batteriutladninger er antall sykluser det vedlagtebatteriet har blitt tappet og ladet opp igjen.

Flymodus er egnet for bruk i kjøremodus. Flymodus
deaktiveres automatisk hvis enheten blir værende i
Ventemodus på-batteristrøm.

FLYMODUS

Merk: Se avsnittet Aktiver flymodus hvis du vil vite merom flymodus.

FreeStyle Comfort leveres med flymodus slått av.

FØLSOMHETSMODUS
FreeStyle Comfort leveres med en standard
følsomhetsmodusinnstilling på 2 fra fabrikken,
og vil ikke kreve justering under normale forhold.
Følsomhetsinnstilling kan justeres mellom verdiene
1–5, der 1 er minst følsom og 5 er mest følsom.

Hovedskjerm i flymodus

For å slå modusen på, trykk påmens ENDREikonet er synlig. Verdien vil skifte til det blå
ordet «PÅ».

Følsomhetsmodus-skjerm i innstilling 2

For å justere følsomhetsmodus, trykk på
mens
ENDRE-ikonet er synlig. Verdien vil skifte til nesteblåe nummer.

Flymodus på valgt

For å bekrefte Flymodus på, trykk på
mens
LAGRE-ikonet er synlig. Verdien vil skifte til det
svarte ordet «PÅ».
For å avbryte Flymodus på, trykk på
AVBRYT-ikonet er synlig.

272 - NORPN MN232-C4 K | BruksanvisningmensFreeStyle Comfort
AUTOSAT-MODUS
FreeStyle Comfort leveres med autoSAT-modusslått på.

Følsomhetsmodus, innstilling 3 valgt

For å bekrefte innstilling 3 i følsomhetsmodus, trykk
påmens Lagre-ikonet er synlig. Verdien vil veksle
til svart «3».
For å avbryte følsomhetsmodusendringen, trykk påmens AVBRYT-ikonet er synlig.

Hovedskjerm i autoSAT-modus

For å slå modusen av, trykk påmens ENDREikonet er synlig. Verdien vil skifte til det blå
ordet «AV».

Følsomhetsmodus, innstilling 3, bekreftet
autoSAT-modus av valgt

For å bekrefte autoSAT-modus av, trykk på
mens
LAGRE-ikonet er synlig. Verdien vil skifte til det
svarte ordet «AV».
For å avbryte autoSAT-modus på, trykk på
AVBRYT-ikonet er synlig.

mens

Følsomhetsmodus, innstilling 5, (maks) bekreftet

Trykk pånår skjermen viser «Neste»-symbolet forå flytte menysiden til neste skjermalternativ.

autoSAT-modus av bekreftetFor å slå på autoSAT-modus, gjenta skrittene ovenfor
mens du veksler til PÅ.
Trykk pånår skjermen viser «Neste»-symbolet forå flytte menysiden til neste skjermalternativ.

PN MN232-C4 K | Bruksanvisning

NOR - 273

FreeStyle Comfort

Feilsøking
ProblemMulig årsak

Løsning

FreeStyle Comfort viser alarmog avgir intermitterende pip.

Se Alarmtilstandstabellen.

Se Alarmtilstandstabellen.

Oksygenkonsentratoren slårseg ikke på.

Batteriet er utladet.

Koble enheten til et uttak fra vekselstrøm eller
likestrøm.

Ikke tilkoblet ekstern strøm.
Generell feil.

Kontroller at eksterne kontakter er sikre.
Utfør en full strømsyklus ved å fjerne batteriet og
koble fra en hvilken som helst ekstern veksel- eller
likestrømkilde. Sett deretter batteriet inn igjen for å
fullføre strømsyklusen. Koble til ekstern strømkilde
ved behov.
Ta kontakt med utstyrsleverandøren din og bytt tilen annen oksygenkilde som nødvendig.

Alle andre problemer.

Utfør en full strømsyklus ved å fjerne batteriet og
koble fra en hvilken som helst ekstern veksel- eller
likestrømkilde. Sett deretter batteriet inn igjen for å
fullføre strømsyklusen. Koble til ekstern strømkilde
ved behov.
Bytt til en annen oksygenkilde, som tilgjengelig, og
ta kontakt med utstyrsleverandøren din.

Merk: Noen alarmer vil føre til at enheten slår seg av.
Merk: Under et totalt strømbrudd tilbakestilles alarmsystemet (tidligere alarmer slettes), og vil bare vise en alarm hvis
tilstanden fortsatt eksisterer når strømmen gjenopprettes.

274 - NORPN MN232-C4 K | Bruksanvisning

FreeStyle Comfort
Reiseinformasjon

1. Slå POC på BARE ved bruk av batteristrøm når
ingen ekstern strømkilde er tilkoblet.

FreeStyle Comfort oppfyller FAAs-retningslinjer forbruk om bord kommersielle fly. Før enhver reise (fly,tog, båt osv.) skal du:• Sikre at konsentratoren er i god driftstilstand.
• Kontakte flyselskapet ditt minst 48 timer før
avgang for å informere dem om at du reiser med en
FreeStyle Comfort.
Vi forslår at du alltid har med deg ekstra batterier påreiser. Ved flyreise må du sørge for nok batteri til å
dekke bakketiden (før innsjekking, sikkerhetskontrollosv.) og mellomlandinger. De fleste flyselskaper
krever at du har et tilstrekkelig antall fulladede
batterier til å drive enheten i minst 150 % av den
forventet maksimale reisetiden (flytider, bakketid før
og etter flyturen, videre flyvninger og for uventede
forsinkelser).

2. Aktiver flymodus med én av følgende metoder:a. Følg instruksjonene for å aktivere flymodus
i avsnittet Instruksjoner for menynavigering.

Merk: Hvis du flyr, anbefales det å bære ekstra batterier i
en sikker bærepose.

FAA-REGULERING

Ekstra batterier må være beskyttet hver for seg for å
hindre kortslutning, ved å plasseres i den originale
emballasjen eller ved at polene isoleres på annen måte,f.eks. ved å tape over de eksponerte polene eller plasser
hver batteri i en egen plastpose eller beskyttende pose,og skal bare transporteres i håndbagasjen.

b. Trykk og hold inne både opp- og ned-knappene
i 5 sekunder til du hører et langt pip og ser atflysymbolet vises.

AKTIVER FLYMODUS
Batterispenning
POC kan bare brukes på et fly under normale
driftsprosedyrer, BARE ved bruk av batteristrøm når
ingen ekstern strømkilde er tilkoblet.

3. Plugg inn POC når flysymbolet vises.

Ekstern strøm
Dersom det finnes en stikkontakt på et fly, kan den
fungere som en reserveløsning for POC-batterierhvis den er brukbar stand på et egnet sted til å brukes
som en POC-strømkilde. I dette tilfellet må følgende
anbefalte fremgangsmåte for overgang fra batteri til
elektrisk strøm i fly følges:
Merk: På grunn av flybegrensninger lades ikke batteriene
opp i flymodus.

PN MN232-C4 K | Bruksanvisning

NOR - 275

FreeStyle Comfort

Tilbehør

PLASSERE ENHETEN I BÆREVESKEN

For riktig ytelse og sikkerhet skal du bare bruke
tilbehøret i listen som er levert av CAIRE gjennom
din oksygenleverandør. Bruk av tilbehør som du ikke
finner i listen nedenfor, kan svekke ytelsen og/eller
sikkerheten til konsentratoren.

1. Sett enheten i bunnen av vesken, med filtre og
batteriåpninger riktig innrettet.

Standard tilbehør for FreeStyle ComfortBæreveske

MI396-1

Ryggsekk

MI459-1

Skulderstropp

MI397-1

Strømforsyning for likestrøm (alleledninger inkludert)

PW036-1S

Strømledning til likestrøm

CD041-1

Eksternt grovpartikkelfilter (høyre)FI226-1

Eksternt grovpartikkelfilter (venstre)

FI227-1

2. Fest glidelåsen rundt bunnen av posen.

Tilgjengelig valgfritt tilbehør
Ytterligere batteripakke med 8 celler

BT034-1S

Batteripakke med 16 celler

BT035-1S

Bordlader for batteripakke

BT036-1

Tilbehørsveske

MI372-2

Merk: Flere alternativer kan være tilgjengelige forlandspesifikke strømledninger som notert ovenfor. Ta
kontakt med CAIRE eller din oksygenleverandør hvis
alternative muligheter er nødvendige for bestillingen.

Instruksjoner for bæreveske
Bruk følgende instruksjoner for å plassere og feste fastdin FreeStyle Comfort-enhet i bærevesken.

276 - NORPN MN232-C4 K | Bruksanvisning

MONTERING AV SKULDERSTROPP
1. Skulderstropp kan festes enten til enheten direkteeller til bæreposen.
2. Klem stroppen på ringen på enheten eller vesken.

FreeStyle Comfort
Avhending og gjenbruk
Returner alltid FreeStyle Comfort, med allekomponenter, til udstyrsleverandør din for riktig
avhending. Du kan også kontakte lokale myndigheter
for anvisning om korrekt deponering av batteriet.
Enheten kan gjenbrukes mellom mennesker, men må
returneres til udstyrsleverandør for opparbeidelse.

WEEE og RoHS

til direktiv om avfall fra elektrisk og elektronisk utstyr(WEEE-direktivet).
Produktene våre skal være i samsvar med begrensningav farlige stoffer (RoHS-direktivet). De vil ikke
inneholde mer enn spormengder av bly eller andre
farlige materialer.

Dette symbolet skal påminne eierne
av utstyret om at utstyret skal leveres til etgjenvinningsanlegg når brukstiden er omme, i henhold

FORSIKTIG: For riktig deponering
av batteriene, ta kontakt med
utstyrsleverandøren eller lokale
myndigheter for krav til deponering.

Materialer i direkte kontakt med operatøren
Konsentratorens deksel................................................................................................................................ PC/ABS
Konsentratorens kontrollpanel.............................................................................. Polyester EBG7 eller tilsvarende
Konsentratorens håndtak...................................................................................................................... Polykarbonat
Oksygenutløp, dyse.............................................................................................................................. Polykarbonat
Enhetens etikett................................................................................................................................................Lexan
Ledningskontakter..............................................................................................................Polykarbonat/vinylklorid
Strømkabel............................................................................................................................................. PVC, metall
Strømforsyning................................................................................................................ .Lexan 940 (polykarbonat)
Batteripakke................................................................................................................................................. PC/ABS
Batteripakke, strømforsyningsetiketter................................................................................................ PolyesterfilmBæreveske til konsentratoren....................................................................................Cordura, neopren, nylonforing
Stropp................................................................................................Polypropylen, acetyl, vinyl, Cordura, neopren

EMC-testing
Veiledning og produsentens erklæring – elektromagnetisk stråling
FreeStyle Comfort er ment brukt i det elektromagnetiske miljøet som er spesifisert nedenfor. Kunden eller brukeren av
FreeStyle Comfort skal påse at det brukes i et slikt miljø.
Utslippstest
RF-utslipp
CISPR 11
RF-utslipp
CISPR 11
Harmoniske emisjoner
IEC 61000-3-2
Spenningssvingninger/
flimmerutslipp IEC 610003-3

SamsvarElektromagnetisk miljø – veiledning

Gruppe 1

FreeStyle Comfort bruker bare RF-energi til sin indre funksjon. Derfor
er RF-utslippet meget lavt, og det er ikke sannsynlig at det fører til
forstyrrelser for elektronisk utstyr i nærheten.

Klasse B
Klasse A
Overensstemmelser

FreeStyle Comfort passer for bruk i alle typer bygninger, inklusive
bosteder og de som er direkte koblet til offentlig lavspent
forsyningsnettverk, som gir strøm til bygninger som brukes
for boformål.

PN MN232-C4 K | Bruksanvisning

NOR - 277

FreeStyle Comfort
Veiledning og produsentens erklæring – elektromagnetisk immunitet

FreeStyle Comfort er ment brukt i det elektromagnetiske miljøet som er spesifisert nedenfor. Kunden eller brukeren av
FreeStyle Comfort skal påse at det brukes i et slikt miljø.
Immunitetstest
IEC 60601 testnivå
Samsvarsnivå
Elektromagnetisk miljø – veiledning
Gulvetbør være av tre, betong eller
Elektromagnetisk miljø± 8 kV-kontakt± 8 kV-kontaktkeramiske fliser. Hvis gulvene er dekket– veiledning
med syntetiske materialer, skal den± 15 kV luftutslipp± 15 kV luftutslipp
IEC 61000-4-2
relative fuktigheten være minst 30 %.
±2 kV for±2 kV forstrømforsyningslinjer
Elektrisk rask
Kvaliteten på nettstrømmen skal tilsvare
strømforsyningslinjer
forbigående/burst
kvaliteten i en/et typisk helseinstitusjon
eller hjemmepleiemiljø.
IEC 61000-4-4±1 kV for inndata/utdataN/A
linjer
±1 kV linje(r) til linje(r)Overspenning
IEC 61000-4-5

Spenningsfall,korte avbrudd og
spenningsvariasjoner
i strømforsyningen
IEC 61000-4-11

Strømfrekvens(50/60 Hz) magnetfelt
IEC 61000-4-8±2 kV fellesmodus påvekselstrømlinjer±1 kV linje(r) til linje(r)±1 kV differensial påvekselstrømlinjer±2 kV fellesmodus på
utendørs I/O-linjer
0 % UT for 0,5 syklus (0°,45°, 90°, 135°, 180°,
225°, 270°, 315°)

0 % UT for 0,5 syklus (0°,45°, 90°, 135°, 180°,
225°, 270°, 315°)

00 % UT for 1 syklus (0 )0 % UT for 1 syklus (0 )70 % UT (30 % fall i UT)
for 25/30 sykluser (00)

70 % UT (30 % fall i UT)
for 25/30 sykluser (00)

0 % UT for 250/300
sykluser (00)

0 % UT for 250/
300 sykluser (00)

3 A/m

3 A/m

0

MERK, UT er vekselstrøms nettstrømspenningen før anvendelse av testnivået.

Kvaliteten på nettstrømmen skal tilsvare
kvaliteten i en/et typisk helseinstitusjon
eller hjemmepleiemiljø.

Kvaliteten på nettstrømmen skal tilsvare
kvaliteten i en/et typisk helseinstitusjon
eller hjemmepleiemiljø. Hvis brukeren av
FreeStyle Comfort trenger kontinuerlig
drift under et strømbrudd, anbefales
det at FreeStyle Comfort drives av en
avbruddsfri strømforsyning eller et batteri.

Strømfrekvensmagnetfeltene børvære på nivå med en typisk plassering
i en/et typisk helseinstitusjon eller
hjemmepleiemiljø.

MERK I noen tilfeller kan Freestyle Comfort bli tilbakestilt etter en betydelig elektrostatisk utladning, og kan kreve en fullstrømsyklus for å gjenoppta normal drift. Se avsnittet Feilsøking for flere detaljer.

278 - NORPN MN232-C4 K | Bruksanvisning

FreeStyle Comfort
Veiledning og produsentens erklæring – elektromagnetisk immunitet

FreeStyle Comfort er ment brukt i det elektromagnetiske miljøet som er spesifisert nedenfor. Kunden eller brukeren av
FreeStyle Comfort skal påse at det brukes i et slikt miljø.
Immunitetstest
IEC 60601 testnivå
Samsvarsnivå
Elektromagnetisk miljø – veiledning
Ledet RF
10 Vrms
10 VrmsBærbart og mobilt RF-kommunikasjonsutstyr
skal ikke brukes nærmere noen del av
IEC 61000-4-6
150 kHz til 80 MHz
FreeStyle Comfort, inkludert kabler, enn
den anbefalte avstanden beregnet ut fra
ligningen som gjelder for senderfrekvensen.
Utstrålt RF

10 V/mIEC 61000-4-3

80 MHz til 2700 MHz

10 V/mAnbefalt avstandd  1.2 P=,

d  1.2 P = 80 MHz til 800 MHz=,

d  1.2 P = 800 MHz til 2700 MHz=,

når P er den maksimale
utgangseffektgraden til senderen i watt (W) ihenhold til senderprodusenten, og d er den
anbefalte adskillelsesavstanden i meter (m).
Feltstyrker fra faste RF-sendere,som bestemt av en elektromagnetisk
undersøkelse,a skal være mindre enn
samsvarsnivået i hvert frekvensområde. b
Det kan forekomme forstyrrelser i nærheten
av utstyr merket med følgende symbol:

MERKNAD 1 Ved 80 MHz og 800 MHz gjelder høyere frekvensområdet.
MERKNAD 2 Retningslinjene gjelder kanskje ikke i alle situasjoner. Elektromagnetisk spredning påvirkes av absorpsjon og
refleksjon fra strukturer, gjenstander og mennesker.
aFeltstyrker fra faste sendere, så som basestasjoner for radiotelefoner (mobil/trådløs) og walkie-talkies, amatørradioer,
AM og FM radio- og TV-sending kan ikke forutsies teoretisk med nøyaktighet. For å vurdere det elektromagnetiske miljøet
ved faste RF-sendere, bør en elektromagnetisk stedsundersøkelse vurderes. Hvis den målte feltstyrken på stedet hvor
FreeStyle Comfort brukes overskrider gjeldende RF-samsvarsnivå ovenfor, bør FreeStyle Comfort observeres for å bekreftenormal drift. Hvis unormal ytelse observeres, kan ytterligere tiltak være nødvendig, slik som reorientering eller omplassering
av FreeStyle Comfort.
bOver frekvensområdet mellom 150 kHz og 80 MHz skal feltstyrkene være lavere enn 10 V/m.

PN MN232-C4 K | Bruksanvisning

NOR - 279

FreeStyle Comfort
Anbefalt avstand mellom bærbart og mobilt RF-kommunikasjonsutstyr og
FreeStyle Comfort

FreeStyle Comfort er egnet for bruk i et elektromagnetisk miljø hvor utstrålte RF-forstyrrelser er kontrollert. Kundeneller brukeren av FreeStyle Comfort kan bidra til å forhindre elektromagnetisk forstyrrelse ved å opprettholde en
minimumsavstand mellom bærbart og mobilt RF-kommunikasjonsutstyr (sendere) og FreeStyle Comfort, som anbefalt
nedenfor, i henhold til maksimal utgangseffekt for kommunikasjonsutstyr.
Maksimal nominell
Avstand i henhold til senderens frekvens
utgangseffekt formsenderen
150 kHz til 80 MHz
80 MHz til 800 MHz
800 MHz til 2,5 GHz

d = 1,.2 P
d = 1.,2 Pd = 1,.2 P
W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
For sendere med en maksimal utgangseffekt ikke oppført ovenfor, kan den anbefalte avstanden d i meter (m) beregnes ved
hjelp av ligningen som gjelder for senderfrekvensen, der P er maksimal utgangseffekt for senderen i watt (W) i henhold til
produsenten av senderen.
MERKNAD 1 Ved 80 MHz og 800 MHz gjelder avstanden for høyere frekvensområder.
MERKNAD 2 Retningslinjene gjelder kanskje ikke i alle situasjoner. Elektromagnetisk spredning påvirkes av absorpsjon og
refleksjon fra strukturer, gjenstander og mennesker.

Klassifisering

Type beskyttelse mot elektrisk støt:
Klasse II	Beskyttelse mot elektrisk støt oppnås
med dobbelt isolering.
Grad av beskyttelse mot elektrisk støt:
Type BF Utstyr som gir en bestemt grad av
beskyttelse mot elektrisk støt angående
		 1) tillatt lekkasjestrøm,		2) pålitelighet av beskyttende jordforbindelse(hvis til stede).
		 Ikke tiltenkt direkte kardial anvendelse.
Uavhengig testing for medisinsk elektrisk
utstyrsstandard:
Testet til å overholde• IEC 60601-1 utgave 3.1: Medisinsk elektrisk
utstyr – Del 1 Generelle krav for sikkerhet.
• CAN/CSA C22.2 nr. 60601-1-8 utgave 2,1,
Medisinsk elektrisk utstyr – del 1: Generelle
sikkerhetskrav
Beskyttelse mot potensiell elektromagnetisk eller
annen interferens mellom utstyret og andre enheter.
	• Testet til å overholde EN 60601-1-2/
IEC 60601-1-2: 4. utgave.

	• Testet til å overholde RTCA/DO160 del 21,
kategori M.
	• CISPR 11 / EN 55011 Klasse B, gruppe 1,«Industrielt, vitenskapelig og medisinsk (ISM)
utstyr»	• FCC del 15, underdel B – utilsiktede radiatorer av
klasse B
IP22 – Beskyttelse mot inntrengning av fingre eller
lignende gjenstander større enn 12,5 mm diameter.
Vertikalt dryppende vann skal ikke ha skadelig
effekt når kabinettet er tiltet i en vinkel på 15° fra
sin normale posisjon.
Tillat metode for rengjøring og infeksjonskontroll: Se
avsnittet «Rengjøring, stell og vedlikehold» i denne
bruksanvisningen for FreeStyle.
Grad av sikkerhet ved bruk i nærvær av brennbare
anestesigasser:
Utstyr ikke tilpasset slik bruk.
Driftsmodus: Kontinuerlig drift.

CAIRE og CAIRE Inc. er registrerte varemerker for CAIRE Inc. Vennligst besøk vår
hjemmeside nedenfor for en fullstendig oversikt over varemerker.
Varemerker: www.careinc.com/corporate/trademarks.

280 - NORPN MN232-C4 K | Bruksanvisning

Copyright © 2021 CAIRE Inc. CAIRE Inc. forbeholder seg retten til å stanse sine produkter,eller endre priser, materialer, utstyr, kvalitet, spesifikasjoner og/eller prosesser for sine
produkter når som helst, uten forvarsel og uten videre forpliktelser eller konsekvenser. Allerettigheter som ikke er uttrykkelig nevnt her, er forbeholdt oss, som anvendelig.

FreeStyle Comfort
Användarkontroller och indikatorer för systemstatus
ISO 7000: Grafiska symboler för användningpå maskiner och utrustningar – register och
sammanställning
Läs bruksanvisningen före användning.
Reg.nr 1641
Lagring eller driftstemperaturområde.
Reg.nr 0632

Lagringsfuktighetsområde. Reg.nr 2620

Rådets direktiv 93/42/EEG om medicintekniska
produkter
Auktoriserad representant i den
Europeiska gemenskapen
Om den produkt unika
enhetsidentifieraren (UDI) -etiketten
har CE-nummer #### på, uppfyller
enheten kraven i direktiv 93/42 / EEG
om medicinsk utrustning. CE ####
symbolen indikerar anmält organ
nummer.
IEC 60417
Klass II-utrustning, dubbelisoleradReg.nr 5172

Begränsning av atmosfärtryck.
Reg.nr 2621
Tillverkarens namn och adress.
Reg.nr 3082

Strömbrytare. Reg.nr 5009
ETL-certifiering

Tillverkningsdatum. Reg.nr 2497

Säkerhetsbyrå för CAN/CSA C22.2
nr 601.1 M90 gällande elektrisk
utrustning för medicinskt bruk

Katalognummer. Reg.nr 2493
Serienummer. Reg.nr 2498

Interna symboler

Denna sida upp. Reg.nr 0623

Håll på avstånd från lättantändliga
material, olja och fett.

Ömtåligt, hanteras varsamt.
Reg.nr 0621

Får inte demonteras.

Skydda den mot regn, håll den torr.
Reg.nr 0626

Ökar flödet.

Staplingsgräns efter antal. Reg.nr 2403

Minskar flödet.

ISO 7010: Grafiska symboler – säkerhetsfärger och
säkerhetssymboler – registrerade säkerhetssymboler
Bruksanvisningen måste läsas.
Reg.nr M002Håll på avstånd från öppna lågor, eld
och gnistor. Öppna antändningskällor
och rökning förbjuden. Reg.nr P003

Menyknapp.
Anpassadtill RTCA/
DO-160avsnitt 21
Anpassad tillkategori M

Anpassad till RTCA DO160 avsnitt
21 kategori M. FAA SFAR 106-krav

Rök inte nära enheten eller när duanvänder den. Reg.nr P002

Klass II-utrustning, dubbelisolerad

Patientansluten del av typ BF (grad av
skydd mot elektrisk stöt). Reg.nr 5333

FAA-godkänd – POC

Varning. Reg.nr W001

PN MN232-C4 K | Bruksanvisning

SWE - 281

FreeStyle Comfort
21 CFR 801.15: Code of Federal Regulations (CFR)
titel 21
Enligt amerikansk federal lag får denna
anordning endast säljas av läkare ellerpå läkares anmodan.
Rådets direktiv 2012/19/EU: avfall som utgörs av
eller innehåller elektrisk och elektronisk utrustning(WEEE)WEEE

Vad är en syrgaskoncentrator?
Luften vi andas innehåller cirka 21 % syre, 78 %kväve och 1 % andra gaser. I FreeStyle® Comfort®
sugs rumsluft in i maskinen genom luftintagen. Luften
passerar därefter genom ett adsorberande material som
kallas molekylsikt. Detta material separerar syret frånkvävet och släpper endast igenom syret. Resultat är
ett flöde av högkoncentrerad syrgas som levereras till
användaren.
I FreeStyle Comfort kombineras avancerade tekniker
för koncentration och konservering av syrgas för att
skapa en portabel enhet som ger maximal bärbarhet
och rörelsefrihet. Den avancerade pulsflödestillförseln
känner snabbt av när användaren andas in och
ger en syrgaspuls i början av varje inandning.
Pulsflödestillförsel är idealiskt vid aktiviteter och
vistelser utanför hemmet. Det sparar syrgas och gör
att batteriet räcker längre än vid ett kontinuerligt flöde.
FreeStyle Comfort är tillräckligt lätt för att bäras i enaxelrem. Den kan drivas med växelström (vägguttag),likström (i fordon) eller batteri.

Avsedd användning
CAIRE FreeStyle Comfort Oxygen Concentrator
är avsedd för administrering av extra syre. Enheten
är inte avsedd för livsstöd och ger inte heller någon
patientövervakningsfunktion.

Varför har din läkare ordinerat
syrgas?
Många människor lider av olika hjärt- ochlungsjukdomar samt andra andningssjukdomar.
Ett betydande antal av dessa kan dra nytta av
kompletterande syrgasbehandling hemma, på resor
eller när de deltar i dagliga aktiviteter utanför hemmet.
Syrgas är en gas som utgör 21 % av luften vi andas
och våra kroppar är beroende av en konstant tillförsel
för att fungera ordentligt. Din läkare har ordinerat ett
282 - SWEPN MN232-C4 K | Bruksanvisning

IEC 60601-1: Elektrisk utrustning för medicinskt bruk
– Del 1: Allmänna fordringar beträffande säkerhet ochväsentliga prestanda
Droppskyddad
Den här produkten kan omfattas av ett eller flera
amerikanska eller internationella patent. Besök vår
webbplats nedan för att se en lista över tillämpliga
patent. Pat.: www.caireinc.com/corporate/patents/.

flöde eller en inställning som behandling för just dinaandningsbesvär.
Även om syrgas inte är en beroendeframkallandedrog kan en ej godkänd syrgasbehandling varafarlig. Du måste rådgöra med en läkare innan
du använder syrgaskoncentratorn. Leverantören
av syrgasutrustningen visar hur den ordinerade
flödesfrekvensen ställs in.

Användarprofil
Koncentratorer används för att ge kompletterande
syrgas till användare som upplever obehag på grund
av sjukdomar som påverkar lungornas effektivitet när
det gäller att överföra syre i luften till blodomloppet.
Bärbara syrgaskoncentratorer används inte för att
lagra eller förvara syre. De behöver inte fyllas påoch kan laddas överallt där växelström eller likström
är tillgänglig (förutom på ett flygplan). Det gör att
användaren blir förhållandevis oberoende och kan
använda utrustningen i hemmet, på rörlig fot (både ioch utanför hemmet), på resor och vid olika aktiviteter.
Syrgaskoncentratorn får endast användas på ordination
av läkare och är inte avsedd för livsuppehållande
behandling.
Även om syrgasbehandling kan ordineras till
användare i alla åldrar är den typiska syrgasanvändarenäldre än 65 år och lider av olika luftvägssjukdomar,t.ex. kronisk obstruktiv lungsjukdom (KOL).
Användarna har vanligtvis god kognitiv förmåga
och måste kunna meddela om obehag uppstår. Om
användaren inte kan meddela upplevt obehag, eller
inte kan läsa och förstå koncentratorns märkning eller
bruksanvisning, bör användning endast ske underöverinseende av någon som kan göra detta. Vid obehagi samband med användning av koncentratorn bör
användaren kontakta vårdgivaren. Användaren böräven ha tillgång till reservsyrgas (dvs. en syrgastub)om det blir strömavbrott eller koncentratorn slutar att
fungera. Användaren behöver ingen särskild kunskap
eller förmåga för att använda koncentratorn.

FreeStyle Comfort

Packa upp FreeStyle Comfort
Kontrollera att alla komponentersom listas finns med i paketet.
Om något saknas ska duomedelbart kontakta leverantören
av syrgasutrustningen.

Standardtillbehör• Bärbar syrgaskoncentrator• Transportväska• Axelrem• 8-cellsbatteri• Växelströmsadapter• Nätsladd• Likströmsadapter• Användarhandbok

Extra tillbehör• Extra 8-cellsbatteri• 16-cellsbatteri• Bordsladdare• Tillbehörsväska

Bekanta dig med FreeStyle Comfort
Bekanta dig med de viktigaste FreeStyle Comfort-funktionerna och dess användargränssnitt.

Framsida
RemfästeAnvändargränssnitt

Syrgasutlopp
Luftutlopp

Luftintag/filter
Luftintag/filter

Strömuttagsflik

Batteripaket
Frigöringsknapp
för batteri
PN MN232-C4 K | Bruksanvisning

SWE - 283

FreeStyle Comfort
Ovansida
DisplayÖkar flödet

Strömknapp

Menyknapp

Minskar flödet

Display för batteri
Inställning av flöde

Batteridisplay

autoSAT-indikator

LarmdisplayÅterstående
batteriprocent

Indikator för
bolusdosering

Display för växelström/likström

Inställning av flöde
Batteridisplay

Displayplats
för larm

284 - SWEPN MN232-C4 K | Bruksanvisning

Laddningsindikator

FreeStyle Comfort
Viktigt!

Kontraindikationer vid användning

Säkerhetsinstruktioner definieras enligt
följande:
VARNING: VIKTIGSÄKERHETSINFORMATION FÖRFAROR SOM KAN ORSAKA ALLVARLIGPERSONSKADA.
VAR FÖRSIKTIG: Viktig information för att
förhindra skador på FreeStyle Comfort.
Obs! Särskild uppmärksamhet bör ägnas åt informationen.

Indikationer för användning
VARNING: DET ÄR MYCKET VIKTIGTATT ENBART STÄLLA IN DEN
ORDINERADE NIVÅN AV SYRE. ÄNDRA
INTE FLÖDESVALET OM DU INTE HARBLIVIT UPPMANAD ATT GÖRA DET AV EN
KVALIFICERAD KLINIKER.
VARNING: ENLIGT FEDERAL LAGSTIFTNING I
USA FÅR DENNA ENHET ENDAST SÄLJAS ELLERHYRAS UT PÅ LÄKARES ELLER ANNAN UTBILDADSJUKVÅRDSPERSONALS ORDINATION.
VARNING: ENHETEN ÄR INTE AVSEDD FÖRLIVSUPPEHÅLLANDE BRUK. SMÅ BARN, ÄLDREELLER ANDRA ANVÄNDARE SOM INTE KAN
KOMMUNICERA OBEHAG MEDAN DE ANVÄNDERAPPARATEN KAN BEHÖVA EXTRA ÖVERVAKNING.
ANVÄNDARE MED HÖRSEL- OCH/ELLERSYNNEDSÄTTNINGAR KAN BEHÖVA HJÄLP MED ATTÖVERVAKA LARM.
VARNING: INSTÄLLNINGAR FÖR PULSDOSLÄGE
MÅSTE BESTÄMMAS INDIVIDUELLT BASERATPÅ VARJE ANVÄNDARES BEHOV I VILA, UNDERTRÄNING ELLER VID RESA.
VARNING: VISSA ANDNINGSANSTRÄNGNINGARFRÅN PATIENTEN KANSKE INTE UTLÖSER
KONSERVERINGSFUNKTIONEN.
VARNING: FREESTYLE COMFORT ÄR INTE AVSEDD
ATT ANVÄNDAS AV EN TRAKEOSTOMERAD PATIENT.
VARNING: GRAVIDA ELLER AMMANDE KVINNORSKA INTE ANVÄNDA TILLBEHÖREN SOM
REKOMMENDERAS I DENNA HANDBOK EFTERSOMDE KAN INNEHÅLLA FTALATER.

VARNING: UNDER VISSA
OMSTÄNDIGHETER KAN ANVÄNDNINGEN
AV ICKE-ORDINERAT SYRE VARA FARLIG.
DEN HÄR ENHETEN BÖR ENDASTANVÄNDAS VID ORDINATION FRÅN ENLÄKARE.
VARNING: FÅR INTE ANVÄNDAS I NÄRHETEN AVLÄTTANTÄNDLIGA ANESTETIKA.
VARNING: I LIKHET MED ALLA ELDRIVNA ENHETER
KAN DRIFTEN TILLFÄLLIGT AVBRYTAS TILL FÖLJDAV STRÖMAVBROTT ELLER PÅ GRUND AV ATTSYRGASKONCENTRATORN BEHÖVER SERVAS AV
EN BEHÖRIG TEKNIKER. SYRGASKONCENTRATORNÄR INTE LÄMPLIG FÖR ANVÄNDARE VARS HÄLSADRABBAS AV ALLVARLIGA KONSEKVENSER VID
SÅDANA TILLFÄLLIGA AVBROTT.

Säkerhetsanvisningar
VARNING: DET ÄR INTE TILLÅTET ATT
MODIFIERA DENNA UTRUSTNINGVARNING: DENNA PRODUKT LEVERERAR SYREAV HÖG KONCENTRATION SOM UNDERLÄTTARSNABB FÖRBRÄNNING. TILLÅT INTE RÖKNING
ELLER ÖPPEN ELD I SAMMA RUM SOM 1) DENNAENHET ELLER 2) SYREFYLLDA TILLBEHÖR. OM MAN
INTE TAR HÄNSYN TILL DENNA VARNING KAN DET
LEDA TILL ALLVARLIG BRAND, EGENDOMSSKADA,KROPPSSKADA OCH/ELLER DÖD.
VARNING: ANVÄND INTE ENHETEN I ETTBEGRÄNSAT UTRYMME (T.EX. ETT LITET SKÅPELLER EN VÄSKA) DÄR VENTILATIONEN ÄRBEGRÄNSAD. DETTA KAN LEDA TILL ATTSYRGASKONCENTRATORN ÖVERHETTAS OCH ATTPRESTANDAN FÖRSÄMRAS.
VARNING: KONCENTRATORN BÖR PLACERAS SÅ
ATT DEN INTE UTSÄTTS FÖR RÖK, FÖRORENINGAR
ELLER ÅNGOR.
VARNING: OM DET INTRÄFFAR EN ALLVARLIG
HÄNDELSE MED DEN HÄR ENHETEN BÖRANVÄNDAREN OMEDELBART RAPPORTERA
HÄNDELSEN TILL LEVERANTÖREN OCH / ELLERTILLVERKAREN. EN ALLVARLIG HÄNDELSE
DEFINIERAS SOM EN SKADA, DÖDSFALL ELLERPOTENTIAL ATT ORSAKA SKADA / DÖDSFALL OM
HÄNDELSEN SKULLE ÅTERKOMMA. ANVÄNDARENKAN OCKSÅ RAPPORTERA HÄNDELSEN TILL
DEN BEHÖRIGA MYNDIGHETEN I DET LAND DÄRHÄNDELSEN INTRÄFFADE.

PN MN232-C4 K | Bruksanvisning

SWE - 285

FreeStyle Comfort
VARNING: ANVÄNDNING AV VISSATILLBEHÖR FÖR SYRGASHANTERING
SOM INTE ÄR DIREKT AVSEDDAATT ANVÄNDAS MED DENNASYRGASKONCENTRATOR KAN PÅVERKA
DESS PRESTANDA NEGATIVT. DENNA
BRUKSANVISNING BESKRIVER VILKATILLBEHÖR SOM REKOMMENDERAS.
VARNING: VAR VÄNLIG KONTAKTA DIN LEVERANTÖRAV UTRUSTNING FÖR INSPEKTION OCH MÖJLIGREPARATION AV SYRGASKONCENTRATORNOM ENHETEN HAR TAPPATS, SKADATS ELLEREXPONERATS FÖR VATTEN. ANVÄND INTESYRGASKONCENTRATORN OM NÄTSLADDEN ELLERKONTAKTEN ÄR SKADAD.
VARNING: FÄST SYRGASKONCENTRATORN
ORDENTLIGT, MED BÄLTE ELLER PÅ ANNAT SÄTT,NÄR DEN TRANSPORTERAS I ETT FORDON FÖR ATTUNDVIKA SKADOR.
VARNING: U.S. DEPARTMENT OF
TRANSPORTATIONS (DOT) OCH FÖRENTA
NATIONERNAS (FN) FÖRORDNINGAR KRÄVER
ATT BATTERIET AVLÄGSNAS FRÅN ENHETENFÖR ALLA INTERNATIONELLA FLYGRESOR DÄRSYRGASKONCENTRATORN CHECKAS IN SOMBAGAGE. BATTERIET MÅSTE OCKSÅ AVLÄGSNASFRÅN ENHETEN OCH PAKETERAS ORDENTLIGT VIDFÖRSÄNDELSE AV SYRGASKONCENTRATORN.
VARNING: ANVÄND INTE DENNA UTRUSTNINGUTAN ATT FÖRST LÄSA OCH FÖRSTÅ DENNA
BRUKSANVISNING. KONTAKTA ÅTERFÖRSÄLJAREN
INNAN DU FÖRSÖKER ANVÄNDA DEN OM DU INTEFÖRSTÅR VARNINGARNA OCH INSTRUKTIONERNA,ANNARS KAN FÖLJDEN BLI PERSON- ELLER
EGENDOMSSKADOR.
VARNING: OM DU KÄNNER OBEHAG ELLER
UPPLEVER EN MEDICINSK NÖDSITUATION SKA DUSÖKA LÄKARHJÄLP OMEDELBART.
VARNING: ANVÄNDNING AVSYRGASKONCENTRATORN UTANFÖR DET
SPECIFICERADE DRIFTSTEMPERATURINTERVALLET
KAN BEGRÄNSA KONCENTRATORNSMÖJLIGHET ATT UPPFYLLA SPECIFIKATIONENFÖR SYRGASKONCENTRATION. SE
SPECIFIKATIONSAVSNITTET I DENNA
BRUKSANVISNING FÖR INFORMATION OM
TEMPERATURGRÄNSERNA.
VARNING: ANVÄND INTE OLJE-, FETT-,PETROLEUMBASERADE ELLER ANDRALÄTTANTÄNDLIGA PRODUKTER TILLSAMMANS
MED SYREFYLLDA TILLBEHÖR ELLERSYRGASKONCENTRATORN. ANVÄND ENDASTVATTENBASERADE, SYREKOMPATIBLA LOTIONEROCH SALVOR. SYRE PÅSKYNDAR FÖRBRÄNNINGEN
AV LÄTTANTÄNDLIGA ÄMNEN.

286 - SWEPN MN232-C4 K | Bruksanvisning

VARNING: SE TILL ATT VENTILERNA FÖR LUFTINTAG
OCH LUFTUTLOPP INTE RISKERAR ATT BLOCKERAS.
TAPPA INTE OCH STICK INTE IN NÅGRA OBJEKTI NÅGRA AV ENHETENS ÖPPNINGAR. DETTA
KAN LEDA TILL ATT SYRGASKONCENTRATORNÖVERHETTAS OCH ATT PRESTANDAN FÖRSÄMRAS.
VARNING: FELAKTIG ANVÄNDNING AV BATTERIETKAN LEDA TILL ATT BATTERIET BLIR VARMT ELLERFATTAR ELD, VILKET KAN ORSAKA ALLVARLIGSKADA. SE TILL ATT INTE GÖRA HÅL I, SLÅ PÅ,
TRAMPA PÅ ELLER TAPPA BATTERIET, ELLER PÅ
ANNAT SÄTT UTSÄTTA BATTERIET FÖR KRAFTIGASTÖTAR ELLER SLAG. ANVÄNDNING AV ETT SKADATBATTERI KAN ORSAKA PERSONSKADOR.
VARNING: KORTSLUT INTE BATTERIETS
METALLKONTAKTER MED METALLOBJEKT, T.EX.
NYCKLAR ELLER MYNT. DETTA KAN SKAPA GNISTOR
ELLER ÖVERDRIVEN VÄRME.
VARNING: VID ANVÄNDNING AV DEN BÄRBARASYRGASKONCENTRATORN UTOMHUS MED
VÄXELSTRÖMSADAPTERN MÅSTE ADAPTERNANSLUTAS TILL ETT JORDFELSBRYTARSKYDDAT(GFI) UTTAG.
VARNING: SYRGASKONCENTRATORN BÖR INTEANVÄNDAS DIREKT INTILL ELLER OVANPÅ ANNAN
UTRUSTNING. OM DETTA INTE GÅR ATT UNDVIKABÖR ENHETEN BEVAKAS FÖR ATT SÄKERSTÄLLANORMAL DRIFT.
VARNING: ANVÄNDNING AV ANDRA KABLAR OCH
ADAPTRAR ÄN DE SPECIFICERADE, FÖRUTOMSÅDANA SOM TILLVERKAREN AV DEN MEDICINSKAUTRUSTNINGEN SÄLJER SOM ERSÄTTNINGSDELARFÖR INGÅENDE KOMPONENTER, KAN ORSAKAÖKAD EMISSION ELLER FÖRSÄMRAD IMMUNITETHOS SYRGASKONCENTRATORN.
VARNING: ANVÄND ENDAST DEN
ELEKTRISKA SPÄNNINGEN SOM ANGES PÅ
SPECIFIKATIONSETIKETTEN PÅ ENHETEN.
VARNING: ANVÄND INTE FÖRLÄNGNINGSSLADDARMED DENNA ENHET OCH UNDVIK ATT ANSLUTAFÖR MÅNGA STICKPROPPAR I SAMMA ELUTTAG.
ANVÄNDNING AV FÖRLÄNGNINGSSLADDAR KAN
PÅVERKA PRODUKTENS PRESTANDA NEGATIVT.
FÖR MÅNGA STICKPROPPAR I ETT ELUTTAG KAN
LEDA TILL EN ÖVERBELASTNING I ELCENTRALENVILKET GÖR ATT PROPPEN/SÄKRINGEN LÖSES UTELLER BRAND OM PROPPEN ELLER SÄKRINGEN
INTE FUNGERAR.

FreeStyle Comfort
VAR FÖRSIKTIG: Se felsökningsavsnittet
i denna bruksanvisning om du
uppmärksammar ett alarm ellersyrgaskoncentratorn inte fungerar som
den ska. Kontakta återförsäljaren om duinte kan lösa problemet.
VAR FÖRSIKTIG: Följ alla anvisningar från tillverkaren
för att undvika att garantin upphävs.
VARNING: UTFÖR INGETUNDERHÅLLSARBETE ANNAT ÄN
DE MÖJLIGA LÖSNINGAR SOM TAS
UPP I DENNA BRUKSANVISNING. TA
INTE AV NÅGRA HÖLJEN. ENDASTÅTERFÖRSÄLJAREN ELLER ENKVALIFICERAD SERVICETEKNIKER FÅR
TA AV HÖLJEN ELLER UTFÖRA SERVICEPÅ DENNA ENHET.

VARNING: SKYLTAR MED ”RÖKNING EJ TILLÅTEN –SYRGAS ANVÄNDS” MÅSTE VARA VÄL SYNLIGAI HEMMET ELLER DÄR SYRGAS ANVÄNDS.
ANVÄNDARE OCH DERAS VÅRDGIVARE MÅSTE
INFORMERAS OM FARORNA MED RÖKNING INÄRHETEN AV ELLER UNDER ANVÄNDNING AV
MEDICINSK SYRGAS.
VARNING: ANVÄND INTE SYRGASKONCENTRATORNI NÄRHETEN AV LÄTTANTÄNDLIGA GASER. DET
KAN LEDA TILL SNABB FÖRBRÄNNING OCH
ORSAKA EGENDOMSSKADA, KROPPSSKADA ELLERDÖDSFALL.
VARNING: OM MAN TAR AV NÄSKANYLEN OCH
LÄGGER DEN PÅ KLÄDER, SÄNGKLÄDER, SOFFOR
ELLER ANDRA MJUKA MATERIAL KAN BRAND
PLÖTSLIGT UPPSTÅ OM NÄSKANYLEN UTSÄTTSFÖR CIGARETTGLÖD, EN VÄRMEKÄLLA ELLERLÅGA.

VARNING: ANVÄND ENDAST TILLBEHÖR
SOM REKOMMENDERAS AV TILLVERKAREN.
ANVÄNDNING AV ANDRA TILLBEHÖR KAN VARAFARLIGT, DE KAN ORSAKA ALLVARLIG SKADA PÅSYRGASKONCENTRATORN OCH DET UPPHÄVERGARANTIN.

VARNING: LÄMNA INTE NÄSKANYLEN PÅ
ELLER UNDER KLÄDER, SÄNGKLÄDER ELLERSTOLSDYNOR. OM ENHETEN ÄR PÅSLAGEN MEN
INTE ANVÄNDS KAN SYRET GÖRA MATERIALET
BRANDFARLIGT. SE TILL ATT STRÖMMEN ÄR AV NÄRSYRGASKONCENTRATORN INTE ANVÄNDS.

VARNING: TILLVERKAREN REKOMMENDERAR EN
ALTERNATIV RESERVSYREKÄLLA I HÄNDELSE
AV STRÖMAVBROTT, LARMTILLSTÅND ELLER
MEKANISKT FEL. RÅDFRÅGA DIN LÄKAREELLER ÅTERFÖRSÄLJARE MED AVSEENDE PÅNÖDVÄNDIGT RESERVSYSTEM.

VARNING: SYRGASKONCENTRATORN MÅSTE VARA
AVSTÄNGD OCH FRÅNKOPPLAD FRÅN EXTERNSTRÖM INNAN BATTERIPAKETET TAS UT.

VARNING: SE TILL ATT VENTILERNA FÖR LUFTINTAG
OCH LUFTUTTAG INTE RISKERAR ATT BLOCKERAS.
TAPPA INTE OCH STICK INTE IN NÅGRA OBJEKTI NÅGRA AV ENHETENS ÖPPNINGAR. DETTA
KAN LEDA TILL ATT SYRGASKONCENTRATORNÖVERHETTAS OCH ATT PRESTANDAN FÖRSÄMRAS.
VARNING: RÖKNING I SAMBAND MEDSYRGASANVÄNDNING ÄR DEN FRÄMSTAORSAKEN TILL BRANDSKADOR OCH DÄRTILL
RELATERADE DÖDSFALL. DU MÅSTE FÖLJA DESSA
SÄKERHETSVARNINGAR:VARNING: TILLÅT INTE RÖKNING, LJUSLÅGORELLER ÖPPEN ELD I SAMMA RUM SOM DENNAENHET ELLER SYREFYLLDA TILLBEHÖR.
VARNING: RÖKNING I SAMBAND MED ANVÄNDNING
AV NÄSKANYL KAN ORSAKA BRÄNNSKADOR IANSIKTET OCH EVENTUELLT DÖDSFALL.
VARNING: DESSA 3 STEG KAN RÄDDA DITT LIV OM
DU RÖKER: STÄNG AV SYRGASKONCENTRATORN,TA AV NÄSKANYLEN OCH LÄMNA RUMMET DÄR
ENHETEN STÅR.

VARNING: SMÅDELAR KAN UTGÖRA EN
POTENTIELL KVÄVNINGSRISK.
VARNING: UNDVIK KONTAKT MED
FRÅNLUFTSVENTILEN I MER ÄN 10 SEKUNDER,EFTERSOM DEN KAN VARA VARM.
VARNING: VIND ELLER KRAFTIGT LUFTDRAG
KAN HA NEGATIV INVERKAN PÅ EN KORREKTSYRGASBEHANDLING.
VARNING: DENNA PRODUKT KAN UTSÄTTA DIG
FÖR KEMIKALIER, BLAND ANNAT NICKEL, SOM I
STATEN KALIFORNIEN I USA HAR KLASSATS SOM
CANCERFRAMKALLANDE. MER INFORMATION FINNSPÅ WWW.P65WARNINGS.CA.GOV.
VAR FÖRSIKTIG: Koppla alltid urväxelströmsadaptern från väggen innan dukopplar ur den från syrgaskoncentratorn.
VAR FÖRSIKTIG: Placera inte koncentratorn så att det
är svårt att komma åt nätsladden.

PN MN232-C4 K | Bruksanvisning

SWE - 287

FreeStyle Comfort
VAR FÖRSIKTIG: Placera alltid syrgasslangarna och
nätsladdarna så att det inte finns risk för att någon
snubblar eller stryps.
VAR FÖRSIKTIG: Se till att fordonet startas och ärigång innan du kopplar in syrgaskoncentratorn om
syrgaskoncentratorn används med likströmsadapterni en bil, båt eller andra likströmskällor. Om man inte
följer dessa anvisningar kan det hända att strömkällan
inte förser syrgaskoncentratorn med ström.
VAR FÖRSIKTIG: Koppla ur och ta ut enheten ur bilen
när bilen i vilken syrgaskoncentratorn används stängs
av. Förvara inte syrgaskoncentratorn i en mycket varmeller kall bil, eller i andra liknande miljöer med hög eller
låg temperatur.
Obs! Portabel och mobil radiofrekvensutrustning för
kommunikation kan påverka medicinsk elektronisk
utrustning.
VAR FÖRSIKTIG: Lämna INTE
syrgaskoncentratorn eller nätaggregatet
inkopplat i bilen när tändningen är iAVSTÄNGT läge. Det kan tömma fordonets
batteri.
VAR FÖRSIKTIG: Om syrgaskoncentratorn
används eller förvaras utanför det normala
driftstemperaturområdet och omgivningstryckområdet
(altitud) kan prestandan påverkas, batteriets
drifttid kan minska och/eller batteriets laddningstid
kan öka. Se avsnittet med specifikationer iden här bruksanvisningen för information om
temperaturgränser vid förvaring och drift.

Obs! Se till att kanylen är helt instucken och ordentligt
fastsatt. Då kan syrgaskoncentratorn upptäcka
inandningar och ge syretillförsel på ett korrekt sätt. Under
inhalation kan du höra eller känna hur syre flödar till
näskanylens mynningar.
Obs! Litiumbatterier kan förlora sin kapacitet permanentom de utsätts för höga temperaturer när de är fulladdade
eller helt urladdade. Vid långtidsförvaring bör batterierna
ha 25–50 % laddning och förvaras vid temperaturer
mellan -20 °C och +20 °C (-4 °F till 68 °F).
Obs! Användning av vissa tillbehör för
syrgasadministrering som inte är avsedda att användas
med denna syrgaskoncentrator kan påverka dessprestanda negativt. Denna bruksanvisning beskriver vilka
tillbehör som rekommenderas.
Obs! Använd inte syrgaskoncentratorn utan att
luftintagsfiltret sitter på plats. Om ett andra filter
tillhandahålls, sätt i ersättningsfiltret innan du rengör
det smutsiga filtret. Rengör det smutsiga filtret i en varmtvålvattenlösning och torka det ordentligt innan det
används.
Obs! Tillverkaren rekommenderar inte att utrustningensteriliseras.
Obs! Det är aldrig någon risk att du tömmer ett rum påsyrgas när du använder syrgaskoncentratorn.
VARNING: SYRGASKONCENTRATORNFÅR INTE ANVÄNDAS ELLER ANSLUTAS
TILL ETT EXTERNT STRÖMUTTAG(VÄXELSTRÖM/LIKSTRÖM) UTAN ETTINSTALLERAT BATTERI. FÖRSÖK
INTE ATT SLÅ PÅ ENHETEN ELLERANSLUTA DEN TILL EXTERN STRÖM NÄR
BATTERIET ÄR BORTTAGET.

VAR FÖRSIKTIG: För syrgaskoncentratorer som ärutrustade med batterier: Förvara på en sval och torr
plats för att säkerställa livslängden på ditt batteri.
Förvaring av syrgaskoncentratorn under längre
perioder i höga temperaturer eller med ett fulladdat
eller helt urladdat batteri kan förkorta batteriets
totala livslängd. Försök inte att öppna batteriet;
det innehåller inga delar som kan repareras av
användaren. Håll batteriet borta från barn.

BATTERIET FÅR ENDAST TAS BORT NÄR ENHETEN
ÄR AVSTÄNGD OCH FRÅNKOPPLAD FRÅNNÄTAGGREGATET. FÖRSÖK INTE ATT TA BORTKRAFTENHETEN NÄR ENHETEN FORTFARANDEÄR PÅSLAGEN ELLER ANSLUTEN TILL
NÄTAGGREGATET.

VAR FÖRSIKTIG: Använd ENDAST de batterier som
tillverkaren tillhandahåller. Se till att batteriet kasseras
på ett korrekt sätt genom att kontakta återförsäljareneller de lokala myndigheterna angående bestämmelser
om kassering.

Obs! Om syrgaskoncentratorn under en längre tid har
förvarats utanför de normala driftsförhållanden som
visas i tabellen ovan, ska du låta den återfå normal
driftstemperatur innan du slår på den.

288 - SWEPN MN232-C4 K | Bruksanvisning

FreeStyle Comfort
Specifikationer
FreeStyle ComfortMått (med ett batteri)25,4 x 18,5 x 7,9 cm (10,0 x 7,3 x 3,1 tum)Mått (med dubbelt batteri)27,9 x 18,5 x 7,9 cm (11,0 x 7,3 x 3,1 tum)

Vikt (med ett batteripaket)

2,3 kg (5 lbs)

Vikt (med dubbelt batteripaket)

2,7 kg (6 lbs)

Nominell ljudnivå***39,93 dB (A) ± 0,42 dB (A) @ 2

Flödesinställningar

Pulsinställningar 1, 2, 3, 4, 5

Inställning för O2-utflöde

Inställning 1 – 210 ml/min
Inställning 2 – 420 ml/min
Inställning 3 – 630 ml/min
Inställning 4 – 840 ml/min
Inställning 5 – 1 050 ml/min(vid 20 andetag/min)

Syrgaskoncentration*

90 % (+5,5 %/-3 %)Syrgasövervakning

Ja

Max. utloppstryck<30 PSI

Standard för doseringskänslighet> -0,1 cm H2O (justerbar)

Växelström

100–240 VAC, 50–60 Hz, 120W Max

Likström

11–18 VDC (max 10 A)

Batteriladdningstid

Ett batteri: 3,5 timmar
Dubbelt batteri: 6,0 timmar

Batteripaketets drifttid

Drifttid för ett batteri: Inställning 2 – upp till 4 timmar
Drifttid för dubbelt batteri: Inställning 2 – upp till 8 timmar

Driftstemperatur**
Batterityp
Batterikapacitet

5 °C till 40 °C (41 °F till 104 °F)
Litiumjon
Enkelcell: 6700 mAh; Dubbel cell: 13400 mAh

Transport- och förvaringstemp. (enhet)-25 °C till +70 °C (-13 °F till +158 °F) upp till 3 048 m (10 000 fot)Transport- och förvaringstemp. (batteri)-25 °C till +70 °C (-13 °F till +158 °F) upp till 3 048 m (10 000 fot)

Drift vid höjd över havet**-381 till 3 048 m (-1 250 till 10 000 fot) (testats för 700–1 060 hPa)

Luftfuktighet vid drift

15–95 % relativ luftfuktighet (ej kondenserande)

Lagringsfuktighet

0 - 90% icke kondenserande

Begränsad garanti

3 år, enhet
1 år, batteri och nätaggregat* Baserat på ett atmosfäriskt tryck på 14,7 psi (101 kPa) vid 21 °C (70 °F)** Enheten fungerar i alla miljöer så länge miljön är med de angivna temperatur- och luftfuktighetsområdena. Användningutanför dessa driftspecifikationer kan begränsa koncentratorns förmåga att uppnå den specificerade syrgaskoncentrationen vid
flödeshastigheter med ett högre literantal. Uppvärmningsskärmen kan visas ända tills koncentratorn har nått målkoncentrationen
eller i upp till 15 minuter. Fortsätt att använda enheten under den här tiden.
*** Ljudnivå uppmätt per testmetod Nr. 14-1 10/2018 MDS-Hej.
Den förväntade livslängden är minst fem år. Den förväntade hållbarheten för utbyteskolonnerna är cirka ett år, om koncentratorn
förblir oanvänd under hela perioden.

PN MN232-C4 K | Bruksanvisning

SWE - 289

FreeStyle Comfort

Bruksanvisning

1. Placera FreeStyle Comfort på en lämplig plats för
användning.
a. Koncentratorn ska vara välventilerad så att
luftintagsfiltren och luftutloppet inte täcks.
b. Koncentratorn ska placeras så att det är lätt att
se och höra alla hörbara och synliga indikatorer
och larm.
c. Kontrollera att luftintagsfiltren sitter på platsinnan enheten används. Kontakta leverantören
av utrustningen om det saknas ett filter på
enheten.
2. Installera ett batteri i koncentratorn.
a. Rikta in batteriet med bottenhöljet frånFreeStyle Comforts mitt.

3. Anslut nätaggregatet för växelström eller likström
till strömkontakten som sitter på framsidan, längst
ner till höger på enheten.

4. Anslut en näskanyl till syrgasutloppets munstycke. Se
till att anslutningen sitter ordentligt före användning.
OBS! Användning av andra kanyler än de som
rekommenderats kan påverka prestandan.

b. Skjut batteriet inåt tills du hör ett klick som
bekräftar att batteriet låsts fast på plats.

290 - SWEPN MN232-C4 K | Bruksanvisning

Obs: Till utrustningsleverantören: Följande syretillbehör
rekommenderas för användning med FreeStyle Comfort:• Nasal kanyl med 7 fot (2,1 m) slang (maximalt LPM: 6
LPM): CAIRE artikelnummer CU002-1 5408-SEQ• Ytterligare information om rekommenderad tillbehör
finns tillgänglig online på www.caireinc.com.

FreeStyle Comfort
5. Slå på FreeStyle Comfort genom att trycka påströmknappen och välja den flödesinställning som
ordinerats med hjälp av knappen (+) för att öka eller
knappen (-) för att minska flödet.

Inställning för autoSAT-pulsvolym (ml)*Andningsfrekvens

1

2

3

4

5

15

10,5

21,0

31,5

42,0

52,5

20

10,5

21,0

31,5

42,0

52,5

25

10,5

21,0

31,5

42,0

42,0

30

10,5

21,0

31,5

35,0

35,0

35

10,5

21,0

30,0

30,0

30,0

40

10,5

21,0

26,3

26,3

26,3

*Toleransen är +/- 15 %Ett kort ljudlarm avges för att indikera att enheten
har slagits på.
Obs! När FreeStyle Comfort först slås på är
standardflödet inställt på 2. Om inställningen ändras
kommer enheten att startas med den föregående
flödesinställningen.
6. Placera näskanylen i ansiktet enligt de medföljandeanvisningarna från tillverkaren och andas normalt.
Du ska känna att en puff av syrgas tillförs genom
näskanylen varje gång du andas in.
7. Om du vill ändra flödesinställningen trycker du på
knappen (+) för att öka flödet eller på knappen (-) för
att minska flödet till önskad hastighet. Det är normalt
att det låter annorlunda när du byter flödesinställning.

Inställning för icke-autoSAT-pulsvolym (ml)*Andningsfrekvens

1

2

3

4

5

15

14,0

28,0

42,0

56,0

70,0

20

10,5

21,0

31,5

42,0

52,5

25

8,4

16,8

25,2

33,6

42,0

30

7,0

14,0

21,0

28,0

35,0

35

6,0

12,0

18,0

24,0

30,0

40

5,3

10,5

15,8

21,0

26,3

8. Stäng av FreeStyle Comfort genom att trycka påströmknappen.
Obs! Displayen försätts i viloläget och dämpas efter 20
sekunder. Tryck på valfri knapp för att aktivera skärmen
från viloläget.
9. Avancerade funktioner beskrivs i avsnittet
Anvisningar för menynavigering.

PRODUKTENS SPECIALFUNKTIONER
autoSAT®
CAIREs egenutvecklade autoSAT-funktion gör det
möjligt för FreeStyle Comfort att leverera en jämn
volym syre (bolus) per andetag, upp till en maximal
andningsfrekvens. När användarens andetagshastighetökar under aktiviteter som promenader, justerar
autoSAT-funktionen enhetens drift för att bibehålla eninställd bolusvolym.
UltraSense®
Den känsliga pulsdostriggern känner av
tryckförändringen under andningen och utlöser
syrgaspulsen vid inandning.
autoDOSE™
Om ett larm om utebliven andning inträffar utlöser den
här funktionen syrgaspulser motsvarande 20 andetag
per minut med enhetens aktuella flödesinställning tills
larmstatus har åtgärdats.
PN MN232-C4 K | Bruksanvisning

SWE - 291

FreeStyle Comfort

Nätaggregat

FreeStyle Comfort kan drivas på tre sätt:• laddningsbart batteri• Växelström• Likström
Information om hur dessa komponenter fungerar ges ide följande avsnitten.

TA UT BATTERIET

Tryck ner batterifrigöringsknappen bort från FreeStyle
Comfort. Håll knappen intryckt och dra ut batteriet ur
koncentratorn.
VARNING: SYRGASKONCENTRATORNMÅSTE VARA AVSTÄNGD OCH
FRÅNKOPPLAD FRÅN EXTERN STRÖM
INNAN BATTERIPAKETET TAS UT.

VARNING: ANVÄND ENDAST DENÄTAGGREGAT SOM BESKRIVS I DENHÄR ANVÄNDARHANDBOKEN. OM
NÄTAGGREGAT ELLER BATTERIPAKETFRÅN EN ANNAN TILLVERKARE ÄN CAIREANVÄNDS KAN ENHETEN SKADAS OCH
FUNGERA FELAKTIGT.

BATTERIPAKETFreeStyle Comfort kan drivas med det laddningsbara
batteripaketet som medföljer utrustningen.
Koncentratorn har inga driftsmässiga begränsningar
när den drivs med batteripaketet. Alla flödeshastigheterär tillgängliga vid batteridrift.

KONTROLLERA BATTERIPAKETETSLADDNING
INSTALLERA BATTERIET

Rikta in batteriet med bottenhöljet från FreeStyle
Comforts mitt. Skjut batteriet inåt tills du hör ett klick
som bekräftar att batteriet låsts fast på plats.

Se batterisymbolen och uppskattad återstående batteritid
på displayen för att kontrollera batteriladdningen när det
är installerat i FreeStyle Comfort.
80–100 %60–80 %

40–60 %

20–40 %

20 %

292 - SWEPN MN232-C4 K | Bruksanvisning

FreeStyle Comfort
Obs! Om batteriet håller på att laddas, visas en
kontaktindikator istället för återstående batteriprocent.
Om du vill kontrollera batteriladdningen när batteriet
inte är installerat i FreeStyle Comfort placerar
du batteriet plant och trycker på TEST-knappen.
Indikatorlamporna för batterimätaren (25–100 %) tänds
till höger om TEST-knappen för att visa batteripaketets
laddningsnivå.

DRIFTTID VID BATTERIDRIFT

När extern växelström eller likström kopplas från,växlar koncentratorn automatiskt över till batteridriftom ett laddat batteri är installerat. I tabellen nedan
visas normal drifttid för ett nytt batteripaket.
Batteripaketets drifttid förblir konstant oavsett din
andningsfrekvens.
Pulsinställning

Drifttid för ett batteri(8 celler)

Drifttid för dubbelt
batteri (16 celler)

1

upp till 8 timmar

upp till 16 timmar

2

upp till 4 timmar

upp till 8 timmar

3

upp till 3 timmar

upp till 6 timmar

4

upp till 2,25 timmar

upp till 4,5 timmar

5

upp till 2 timmar

upp till 4 timmar

NORMAL BATTERILADDNINGSTID

Det tar vanligtvis 3,5 timmar för ett batteri och
6,0 timmar för ett dubbelt batteri för att helt ladda
batteripaketet när det är helt urladdat.

FÖRSTA LADDNINGEN

Det nya batteriet som medföljer din FreeStyle Comfort
är bara delvis laddat när det levereras från fabriken.
Innan du använder enheten för första gången ska duladda batteripaketet helt med växelström eller likström.

ALLMÄN INFORMATION OM LADDNING

Obs! Batteriet kanske inte börjar laddningen direkt om
batteripaketet är helt urladdat.

BATTERIETS LIVSLÄNGD

FreeStyle Comfort-batterier är utformade för att hålla
för 300 laddnings-/urladdningscykler.

FreeStyle Comfort laddar batteripaketet när det
är installerat i koncentratorn och koncentratorn är
ansluten till en extern växelströms- eller likströmskälla(förutom på ett flygplan).
Batteriet är fulladdat när det visar 100 %.
Alternativt kan batteripaketet laddas med
bordsladdaren (tillval).

PN MN232-C4 K | Bruksanvisning

SWE - 293

FreeStyle Comfort

Nätaggregat för växelström och
likström
FreeStyle Comfort levereras som standard med
nätaggregat för växelström och likström.
I nätaggregatet för växelström ingår nedanstående.
a. Växelströmdel som drivs på 100–240 V,
50–60 Hz.
b. Nätsladd för att ansluta till strömdelen och
motsvarande nätströmsuttag.

Nätaggregatet för likström består av en sladd som
ansluts direkt till enheten och likströmssuttaget.

2. Anslut nätsladden till växelströmdelen.

VÄXELSTRÖM

Växelström kan användas hemma och överallt där
det finns vanliga nätströmsuttag. Freestyle Comfort
har full funktionalitet när den drivs med växelström.
FreeStyle Comfort laddar även batteripaketet när som
helst där växelström finns tillgängligt (utom i flygplan).
Batteripaketet laddas oavsett om FreeStyle Comfort ärpåslagen eller avstängd.
VARNING: ANSLUT INTESYRGASKONCENTRATORN TILL EXTERNSTRÖM NÄR ETT BATTERIPAKET INTE ÄR
INSTALLERAT.
Så här ansluter du Freestyle Comfort till växelström:
1. Anslut växelströmdelen till strömingången på
framsidan, längst ner på FreeStyle Comfort. Tryckinåt så att anslutningen sitter säkert.

294 - SWEPN MN232-C4 K | Bruksanvisning

3. Anslut den andra änden av nätsladden till ett
vägguttag eller ett annat lämpligt eluttag.

FreeStyle Comfort
4. Slå på FreeStyle Comfort genom att trycka påströmknappen och välja den flödesinställning som
ordinerats med hjälp av knappen (+) för att öka eller
knappen (-) för att minska flödet.

Obs! Om batteripaketet behöver laddas visas
batterisymbolen på displayen med ett enda orangefärgat
streck.

LIKSTRÖM

3. Anslut den andra änden av likströmssladden till ett
uttag i en bil eller ett annat likströmsuttag.

4. Slå på FreeStyle Comfort genom att trycka påströmknappen och välja den flödesinställning som
ordinerats med hjälp av knappen (+) för att öka eller
knappen (-) för att minska flödet.
Obs! Om batteripaketet behöver laddas visas
batterisymbolen på displayen med ett enda orangefärgat
streck.

Likström används vid anslutning till tillbehörsuttag
i bilar, båtar och andra motorfordon. FreeStyle
Comfort har full funktionalitet när den drivs med
likström. Freestyle Comfort laddar också vid
behov batteripaketet när likström är tillgängligt.
Batteripaketet laddas både när enheten är avstängd,påslagen och när den används.
VARNING: ANSLUT INTESYRGASKONCENTRATORN TILL EXTERNSTRÖM NÄR ETT BATTERIPAKET INTE ÄR
INSTALLERAT.
VARNING: ANVÄND INTE ENHETEN I ETT BILUTTAG
PÅ 12 V MED LIKSTRÖM MEDAN FORDONET ÄRAVSTÄNGT.
1. Starta motorfordonet.
2. Anslut nätaggregatet för likström till strömuttaget
på framsidan, längst ner på FreeStyle Comfort. Tryckinåt så att anslutningen sitter säkert.

PN MN232-C4 K | Bruksanvisning

SWE - 295

FreeStyle Comfort
Rengöring, skötsel och
rutinunderhåll
RUTINMÄSSIGT UNDERHÅLL

Förebyggande underhåll
Enheten är underhållsfri. Filter inuti enheten är
avsedda att hålla enhetens livslängd. Om enheten
inte fungerar korrekt ska enheten returneras tillleverantören för reparation. Service av interna
komponenter innanför höljet på FreeStyle Comfort ska
utföras av syrgasleverantören.
Användaren av enheten ansvarar för nedanstående
underhållsprocedurer.
• Varje vecka – rengöring av luftintagsfiltren
VARNING: FÖRSÖK INTE ATT ÖPPNAELLER TA BORT PLASTHÖLJET PÅSYRGASKONCENTRATORN. ENHETEN
FÅR ENDAST ÖPPNAS AV EN BEHÖRIGSERVICETEKNIKER.

LUFTINTAGSFILTER

Luft kommer in i FreeStyle Comfort genom
luftintagsfilter som sitter på framsidan, längst ner påsyrgaskoncentratorn. Filtren förhindrar att damm ochandra större partiklar i luften kommer in i enheten.
Luftintagsfiltren ska rengöras minst en gång i veckan.
Syrgasleverantören kan råda dig att rengöra dem
oftare, beroende på driftsförhållandena. Följ dessa
anvisningar för att rengöra luftintagsfiltren ordentligt.
1. Ta bort de två filtren från framsidan av FreeStyle
Comfort genom att trycka upp motsvarande flik och
dra filtren utåt.

VARNING: ANVÄND INTESYRGASKONCENTRATORN UTAN ATTLUFTINTAGSFILTRET SITTER PÅ PLATS.

SKÖTSEL AV BATTERIPAKETET

Låt inte batteripaketet komma i kontakt med vätska.
Försök inte att demontera eller öppna batteripaketet.
Låt inte batteriet komma i kontakt med öppen eld.
Du får inte tappa batteripaketet.
Håll batteripaketet utom räckhåll för barn.
Om batteripaketet verkar vara skadat ska du kontakta
syrgasleverantören innan du använder det.
Placera inte batteriet i eller i närheten av en
mikrovågsugn eller annan köksutrustning.

FÖRVARING AV BATTERIPAKETET

Syrgaskoncentratorns batteri behöver inte laddas ur
helt innan det laddas på nytt. Syrgaskoncentratorns
batteri bör laddas efter varje användning.
Lämna inte batteriet på en plats med mycket hög
värme, t.ex. i direkt solljus eller i en bil. Om du gör
det kan batteriets prestanda och livslängd försämras.
Det kan även leda till att batteriet alstrar värme,börjar ryka, tar eld eller exploderar. Batteripaketetska förvaras i en stabil miljö med låg luftfuktighet(lägre än 70 % relativ luftfuktighet) och utan frätande
gaser. För att förhindra rost ska förhållanden som kan
ge upphov till kondens undvikas, exempelvis snabba
växlingar i den omgivande temperaturen.
Vid förvaring i en månad eller längre tid
rekommenderas en laddningsnivå på mellan 50 % och
75 % (två till tre indikatorlampor lyser). När batteriet
laddats ska det tas ut ur syrgaskoncentratorn.

2. Tvätta båda filtren med tvål och varmt vatten.
3. Skölj filtren noggrant och avlägsna vattnet med en
mjuk luddfri trasa. Se till att filtren är helt torra innan
du sätter tillbaka dem.
4. Sätt tillbaka de torra, rena filtren.

296 - SWEPN MN232-C4 K | Bruksanvisning

FreeStyle Comfort
RENGÖRING

Stäng AV enheten och koppla bort den från växel- eller
likströmmen före all rengöring eller. Spreja INGET
direkt på ytterhöljet. Använd en fuktig (inte blöt)
trasa eller svamp. Spreja trasan eller svampen med
ett milt hushållsrengöring för att rengöra hölje ochnätaggregat. Gör som tillverkaren av rengöringsmedlet
föreskriver.
VARNING: STÄNG AV ENHETEN OCH
KOPPLA BORT NÄTSLADDEN FRÅNELUTTAGET INNAN DU RENGÖRENHETEN FÖR ATT UNDVIKA ELSTÖTAR
OCH BRÄNNSKADOR. ENDAST DINÅTERFÖRSÄLJARE ELLER EN BEHÖRIGSERVICETEKNIKER FÅR TA BORT HÖLJETELLER REPARERA ENHETEN.
VARNING: SÄKERSTÄLL ATTSYRGASKONCENTRATORN OCH BATTERIPAKETET
INTE BLIR BLÖTA OCH ATT VÄTSKOR INTE KOMMERIN I ENHETEN. DET KAN LEDA TILL FEL ELLER ATTENHETEN STÄNGS AV OCH KAN MEDFÖRA RISK FÖRELSTÖTAR ELLER BRÄNNSKADOR.

VARNING: ANVÄND EN FUKTAD (INTE BLÖT)DUK ELLER SVAMP FÖR ATT RENGÖRABATTERIPAKETET. SPREJA FÖRST ETT MILTHUSHÅLLSRENGÖRINGSMEDEL PÅ DUKEN ELLER
SVAMPEN OCH RENGÖR BATTERIPAKETETS HÖLJE.
VARNING: FÅR INTE RENGÖRAS MED ETTAUTOMATISKT TVÄTTSYSTEM.

BYTE AV KANYL

Följ alltid kanyltillverkarens anvisningar för korrekt
användning. Byt ut näskanylen eller syrgasslangenvar 3:e till 6:e månad (vid daglig användning,var 3:e månad) enligt kanyltillverkarens ellersyrgasleverantörens rekommendationer. Läkaren ellersyrgasleverantören ger dig instruktioner för rengöring
och byte.
Extra tillbehör för byte är tillgängliga via
syrgasleverantören.

VARNING: ANVÄND INTE VÄTSKA DIREKT PÅ
ENHETEN. OLÄMPLIGA KEMISKA MEDEL OMFATTAR,
MEN ÄR INTE BEGRÄNSADE TILL FÖLJANDE:
ALKOHOL OCH ALKOHOLBASERADE PRODUKTER,
PRODUKTER BASERADE PÅ KONCENTRERADKLORIN (ETYLENKLORID) OCH OLJEBASERADEPRODUKTER. DESSA PRODUKTER FÅR INTEANVÄNDAS FÖR ATT RENGÖRA KONCENTRATORNSPLASTHÖLJE EFTERSOM DE KAN SKADA ENHETEN.
VARNING: RENGÖR KÅPAN, KONTROLLPANELENOCH NÄTSLADDEN ENBART MED ETT MILTHUSHÅLLSRENGÖRINGSMEDEL OCH EN FUKTAD(INTE BLÖT) DUK ELLER SVAMP OCH TORKA SEDANALLA YTOR TORRA. SE TILL ATT VÄTSKA INTE
KOMMER IN I KONCENTRATORN. VAR SÄRSKILTNOGGRANN MED SYRGASUTLOPPET DÄR KANYLENANSLUTS OCH SE TILL ATT DET ÄR FRITT FRÅNDAMM, VATTEN OCH PARTIKLAR.

PN MN232-C4 K | Bruksanvisning

SWE - 297

FreeStyle Comfort

Larmtillstånd
Prioritet nr

Beskrivning
av larm

1

Systemfel
Högtemperatur påbatteriet

Display

Ljudsignal

Mer information/föreslagen åtgärd

1
pipsignalvar
20:e sek

Enheten avger ett larm när
batteritemperaturen börjar närma sig
säkerhetsavstängningsgränsen.
Om avstängningsgränsen nås, stängs
enheten av. Om extern strömförsörjningansluts visas larmskärmen och ett ljudlarm
fortsätter i 2 min.
Testa med ett batteri som har rumstemperatur.
Kontakta leverantören om problemet kvarstår.

2

Systemfel
Högtemperatur

1
pipsignalvar
20:e sek

Enheten avger ett larm när deninterna temperaturen börjar närma sig
avstängningsgränsen.
Om avstängningsgränsen nås stängs enheten
av. Larmskärmen visas och ett ljudlarm
fortsätter i 2 min.
Se till att det inte finns några blockeringar
i enhetens ventilationsöppningar. Kontakta
leverantören om problemet kvarstår.

3

4

Systemfel
Högt tryck

Inget andetag
detekterat

1
pipsignalvar
20:e sekOm avstängningsgränsen nås stängs enheten
av. Larmskärmen visas och ett ljudlarm
fortsätter i 2 min.

1
pipsignalvar
30:e sek

Inget andetag detekterat under 15 sekunder.

Kontakta leverantören om problemet kvarstår.

Aktiveras efter första andetaget eller när
uppvärmningsskärmen stängs av, beroendepå vad som inträffar först.
Kontrollera om kanylen har veck eller läckor.
Öka känslighetsinställningen. Kontakta
leverantören om problemet kvarstår.

5

6

298 - SWE

Lågbatterinivå
Avstängning

1
pipsignalvar
30:e sek

Larmet avges tills batteriet laddas ur och
enheten stängs av.

Uppvärmningingen

Enheten börjar automatisk dosering tills ett
andetag detekteras eller uppvärmningenavslutas, beroende på vad som inträffar först.

PN MN232-C4 K | Bruksanvisning

Ladda batteriet.

FreeStyle Comfort
Prioritet nr

Beskrivning
av larm

7

Display

Ljudsignal

Mer information

Låg koncentration1
pipsignalvar
30:e sek
i upp till
5 min

Kontakta leverantören om problemet kvarstår.

8

Systemfel
Flödeshastighet (OCDventilfel,
obstruktion av
gasväg)1
pipsignalvar
30:e sek

Kontakta leverantören om problemet kvarstår.

9Låg
batterinivå
Varning1
ljudsignal,
upprepasej

Ladda batteriet.

10

Andningsfrekvensöverskriden

1
pipsignalvar
60:e sek

Andningsfrekvensen överskrider 40 andetag/
min.

11

Systemfel
Fläktfelingen

Kontakta leverantören om problemet kvarstår.

12

SystemfelLågt tryck

ingen

Kontakta leverantören om problemet kvarstår.

13

Systemfel
Låg temp

ingenLåt enheten värmas upp till rumstemperatur.
Kontakta leverantören om problemet kvarstår.

Info

Start

1 informationspip

Skärmen visas vid påslagning och du bör
höra en pipsignal som bekräftar att signalen
fungerar.

Om dosen utlöses felaktigt ska
känslighetsinställningen minskas. Kontakta
leverantören om problemet kvarstår.

Startskärmen visar aktuell revision för fast
programvara längst ner i det högra hörnet.

PN MN232-C4 K | Bruksanvisning

SWE - 299

FreeStyle Comfort
Prioritet nr

Beskrivning
av larm

Display

Ljudsignal

Info

Skärm vid
normal drift
i autoSATläget (ingen
likström
ansluten)

ingen

Info

Skärm vid
normal drift
med autoSAT
inaktiverat(ingenlikström
ansluten)

ingen

Info

Ändradinställning

1 informationspip

Info

Inkoppling av
extern ström iavstängt läge

ingen

Info

Inkoppling av
extern ström idriftläget

ingen

Info

Frånkoppling
av extern
ström iavstängt läge

1 informationspip

InfoFrånkoppling
av extern
ström idriftläget(med batteri)

1 informationspip

300 - SWEPN MN232-C4 K | Bruksanvisning

Mer information

Visar CAIRE-skärmen i 2 sekunder, följt av
batteri- och kontaktstatus i 5 sekunder innanskärmen stängs av.

1 Visar CAIRE-skärmen i 2 sekunder och
stänger sedan av displayen.

FreeStyle Comfort
Prioritet nr

Beskrivning
av larm

Info

Display

Ljudsignal

Mer information

När upp- eller
nedknappen
trycks ner iviloläget visas
batteri- ochladdningsstatus på
displayen

ingen

Skärmen visas i 5 sekunder och stängs
sedan av.

Info

Flygplansläge
när enheten
drivs påbatteri

1 långljudsignal

Info

Flygplansläge
när enhetenär ansluten tillström

1 långljudsignal

PN MN232-C4 K | Bruksanvisning

SWE - 301

FreeStyle Comfort

Anvisningar för menynavigering
Menyn kan öppnas i viloläge eller driftläge. Om
ett larmtillstånd inträffar i driftläge avslutas menyn
automatiskt. Om inga knappar trycks in under
20 sekunder avslutas menyn automatiskt. I viloläge
kan du behöva trycka på menyknappen två gånger
för att aktivera enheten och sedan gå in i menyn.
Det finns fyra menyskärmar:
1. Statusskärm, visar revisionsnummer för fast
programvara, drifttid och batteriurladdningscykler

Flygplansläge

2. Flygplansläge (PÅ/AV)
3. Känslighet, justerbar (1–5)
4. autoSAT (PÅ/AV).

Känslighet

Exempel på statusskärm

ANVÄNDA INSTÄLLNINGSMENYN
Du når inställningsmenyn genom att trycka på meny-/
valknappen
i driftläge eller viloläge.
Tryck påför att bläddra genom menyerna.

Statussidan

302 - SWEPN MN232-C4 K | Bruksanvisning

autoSATFreeStyle Comfort
STATUS SIDA
Statussidan för Freestyle Comfort visar Firmware-revision, körningstider och antal battericykler
på det anslutna batteriet.

Flygplansläge på bekräftat

Upprepa ovanstående steg och växla till OFF (av) för
att stänga av flygplansläget.

Statussidan

Körtimmar visas i hela timmar.
Batteriladdningar är antalet cykler som det anslutna
batteriet har tömts och laddats upp.

FLYGPLANSLÄGE
FreeStyle Comfort levereras med flygplansläget
avstängt.

Om du trycker på
när skärmen visar Nextsymbolen (nästa) flyttar menysidan till nästa
skärmalternativ.
Flygplansläget är avsett för användning i driftläge.
Flygplansläget inaktiveras automatiskt om enheten
lämnas i viloläge med batteridrift.
Obs! Se avsnittet Aktivera flygplansläget för mer
information om flygplansläget.

KÄNSLIGHETSLÄGE
FreeStyle Comfort levereras från fabriken med
standardinställningen 2 för känslighetsläget och bör
inte behöva justeras under normala förhållanden.
Känslighetsinställningen kan justeras mellan värdena
1 och 5, där 1 innebär minst känslighet och 5 mest
känslighet.
Huvudskärm i flygplansläget

Tryck påmedan CHANGE-symbolen (ändra)
visas för att aktivera läget. Värdet växlar då till ordet
ON (på) i blått.

Skärmen i känslighetsläge vid inställning 2

Tryck påmedan CHANGE-symbolen (ändra)
visas för att justera känslighetsläget. Värdet växlar dåtill nästa siffra i blått.
Flygplansläge på har valts

Tryck påmedan Save-symbolen (spara) visas för
att bekräfta flygplansläge på. Värdet växlar då till ordet
ON (på) i svart.
Tryck påmedan Cancel-symbolen (avbryt) visas
för att avbryta valet av flygplansläge på.
Inställning 3 för känslighetsläget vald
PN MN232-C4 K | Bruksanvisning

SWE - 303FreeStyle Comfort
Tryck påmedan SAVE-symbolen (spara) visas för
att bekräfta inställning 3 för känslighetsläget. Värdet
växlar då till siffran 3 i svart.

AUTOSAT-LÄGEFreeStyle Comfort levereras med autoSAT-läget
aktiverat.

Tryck påmedan CANCEL-symbolen (avbryt) ärsynlig för att avbryta ändringen av känslighetsläge.

Huvudskärm i autoSAT-läget

Inställning 3 för känslighetsläget bekräftad

Tryck påmedan CHANGE-symbolen (ändra)
visas för att inaktivera läget. Värdet växlar då till ordet
OFF (av) i blått.

Inställning 5 (max) för känslighetsläget bekräftad
autoSAT-läge av har valts

Om du trycker på
när skärmen visar Nextsymbolen (nästa) flyttar menysidan till nästa
skärmalternativ.

Tryck påmedan SAVE-symbolen (spara) visas för
att bekräfta autoSAT-läge på. Värdet växlar då till ordet
OFF (av) i svart.
Tryck påmedan CANCEL-symbolen (avbryt)visas för att avbryta valet av autoSAT-läge på.

autoSAT-läge av bekräftat

Upprepa ovanstående steg och växla till ON (på) för att
aktivera autoSAT-läget.
Om du trycker på
när skärmen visar Nextsymbolen (nästa) flyttar menysidan till nästa
skärmalternativ.

304 - SWEPN MN232-C4 K | Bruksanvisning

FreeStyle Comfort
Felsökning
Problem

Trolig anledning

Lösning

FreeStyle Comfort visar ett larmoch avger en återkommandeljudsignal.

Se tabellen överlarmförhållanden.

Se tabellen över larmförhållanden.

Syrgaskoncentratorn startar
inte.

Batteriet är urladdat.

Förse enheten med ström genom växel- eller
likströmsuttag.

Inte ansluten till en externströmkälla.
Allmänt fel.

Se till att de externa kontakterna sitter säkert.
Utför en fulleffektscykel genom att ta ut batterietoch koppla från eventuell extern växelströms- eller
likströmskälla. Sätt sedan tillbaka batteriet ocheffektcykeln är klar. Anslut vid behov den externaströmkällan igen.
Kontakta leverantören av utrustningen och byt vid
behov till en annan syrgaskälla.

Alla övriga problem.

Utför en fulleffektscykel genom att ta ut batterietoch koppla från eventuell extern växelströms- eller
likströmskälla. Sätt sedan tillbaka batteriet ocheffektcykeln är klar. Anslut vid behov den externaströmkällan igen.
Byt till en annan tillgänglig syrgaskälla och kontakta
leverantören av utrustningen.

Obs! Vissa larm gör så att enheten stängs av.
Obs! Under en total strömförlust återställs larmsystemet (eventuella tidigare larm rensas) och ett larm visas endast om
tillståndet kvarstår när strömmen återställs.

PN MN232-C4 K | Bruksanvisning

SWE - 305

FreeStyle Comfort

Information inför resor

FreeStyle Comfort följer FAA:s riktlinjer för
användning ombord på trafikflygplan. Före varje resa
(med flyg, tåg, kryssningsfartyg osv.) ska du vidta
följande åtgärder.
• Kontrollera att koncentratorn är i ett välfungerandeskick.
• Kontakta researrangören minst 48 timmar före avresa
och informera dem om att du kommer att resa med
FreeStyle Comfort.
Du bör alltid ha med extra batterier när du reser. Vid
flygresor ska du se till att drifttiden med batteri räcker
för tiden på marken (incheckning, säkerhetskontrollosv.) och mellanlandningar. De flesta flygbolag kräver
att du har ett tillräckligt antal fulladdade batteripaket
för att driva enheten under minst 150 % av den
förväntade maximala restiden (flygtider, tid på marken
före och efter flygresan, anslutningar samt oförutsedda
förseningar).

1. Slå på den bärbara syrgaskoncentratorn med
ENDAST batteridrift, utan någon extern strömkälla
ansluten.

2. Aktivera flygplansläget genom någon av metoderna
nedan.
a. Följ anvisningarna för aktivering av flygplansläge
i avsnittet Anvisningar för menynavigering.

Obs! Vid flygresor bör extra batterier förvaras i en säker
transportväska.

FAA-BESTÄMMELSER

Reservbatterier måste skyddas var för sig för
att förhindra kortslutning. De kan placeras ioriginalförpackningen eller isoleras på annat sätt, t.ex.
genom att exponerade terminaler tejpas för eller varje
batteri placeras i en separat plast- eller skyddspåse, ochbör endast transporteras som handbagage.

b. Håll in både upp- och nedknappen i
5 sekunder tills du hör en lång pipsignal
och ser att flygplanssymbolen visas.

AKTIVERA FLYGPLANSLÄGET
Batteridrift
Den bärbara syrgaskoncentratorn kan användas på
ett flygplan ENDAST med batteridrift under normala
driftförhållanden, när en extern strömkälla inte är
ansluten

3. Koppla in den bärbara syrgaskoncentratorn när
flygplanssymbolen visas.

Extern ström
På ett flygplan, om ett eluttag är tillgängligt, kan
användas och dess läge är lämpligt för användning som
en POC-strömkälla, kan det fungera som en backup till
POC-batterier. I det fallet måste du följa nedanstående
rekommenderade förfarande för övergång från batteri
till flygplanets elsystem:
Obs! På grund av flygplanets begränsningar laddas inte
batterierna i flygplansläget.

306 - SWEPN MN232-C4 K | Bruksanvisning

FreeStyle Comfort
TillbehörSÄTTA I ENHETEN I TRANSPORTVÄSKAN

För korrekt funktion och säkerhet ska du endast
använda nedanstående tillbehör från CAIRE somlevereras via din syrgasleverantör. Om du använder
andra tillbehör än de som räknas upp nedan kan
koncentratorns prestanda och/eller säkerhet påverkas.

1. Skjut in enheten till botten av väskan och rikta infilter- och batteriöppningarna korrekt.

Standardtillbehör till FreeStyle Comfort
Transportväska

MI396-1

Ryggsäck

MI459-1

Axelrem

MI397-1

Nätaggregat för växelström (allasladdar medföljer)

PW036-1S

Likströmssladd

CD041-1

Externt partikelfilter, grovt (höger)FI226-1

Externt partikelfilter, grovt (vänster)FI227-1

2. Stäng blixtlåset runt botten av väskan.

Extra tillbehör som finns tillgängliga
Extra 8-cellsbatteri

BT034-1S

16-cellsbatteri

BT035-1S

Bordsladdare för batteripaket

BT036-1

Tillbehörsväska

MI372-2

Obs! Fler alternativ för landsspecifika nätsladdar kan vara
tillgängliga där detta anges ovan. Kontakta CAIRE eller dinsyrgasleverantör om du behöver beställa andra alternativ.

Anvisningar för transportväskan

FÄSTA AXELREMMEN
1. Axelremmen kan fästas antingen direkt på enheten
eller på transportväskan.
2. Snäpp fast remmen i ringen på enheten eller väskan.

Sätt i och förankra FreeStyle Comfort-enheten itransportväskan enligt anvisningarna nedan.

PN MN232-C4 K | Bruksanvisning

SWE - 307

FreeStyle Comfort

Bortskaffande och återanvändningÅterlämna alltid FreeStyle Comfort inklusive alla
komponenter till din hemsjukvårdsgivare för korrekt
avfallshantering. Du kan även kontakta ditt lokala
förvaltnings- eller kommunkontor för instruktioner
om korrekt avfallshantering av batteriet. Enheten kanåteranvändas mellan människor men måste återlämnas
till hemvårdsleverantören för upparbetning.

WEEE och RoHS
Syftet med denna symbol är att påminnaägaren av utrustningen att lämna tillbaka den till enåtervinningsanläggning när den ska kasseras, enligt
WEEE-direktivet om kassering av elektriskt avfall(Waste Electrical and Electronic Equipment).
Våra produkter uppfyller kraven i RoHS-direktivet(begränsning av användningen av vissa farliga ämnen
i elektriska och elektroniska produkter). De innehåller
inte mer än spårelement av bly och andra farliga
material.
VAR FÖRSIKTIG: Kontakta leverantöreneller de lokala myndigheterna för
anvisningar om korrekt avfallshantering av
batterier.

Material som har direktkontakt med användaren
Koncentratorns hölje.................................................................................................................................... PC/ABS
Koncentratorns kontrollpanel.................................................................................. Polyester EBG7 eller likvärdigt
Koncentratorhandtag............................................................................................................................ Polykarbonat
Syrgasutlopp, munstycke..................................................................................................................... Polykarbonat
Etikett på enheten.............................................................................................................................................Lexan
Sladdanslutningar...............................................................................................................Polykarbonat/vinylklorid
Nätsladd................................................................................................................................................. PVC, metall
Nätaggregat...................................................................................................................... Lexan 940 (polykarbonat)
Batteripaket.................................................................................................................................................. PC/ABS
Batteripaket, etiketter på nätaggregat................................................................................................... Polyesterfilm
Koncentratorns transportväska................................................................................... Cordura, neopren, nylonfoder
Rem................................................................................................... Polypropylen, acetyl, vinyl, cordura, neopren

EMC-tester
Vägledning och tillverkarens deklaration – elektromagnetiska emissioner
FreeStyle Comfort är avsedd för användning i den elektromagnetiska miljö som specificeras nedan. Kunden eller
användaren av FreeStyle Comfort ska säkerställa att den används i en sådan miljö.
Emissionstester
RF-emissioner
CISPR 11
RF-emissioner
CISPR 11
Harmoniska emissioner
IEC 61000-3-2
Spänningsvariationer/
flimmer IEC 61000-3-3

308 - SWEÖverensstämmelse

Elektromagnetisk miljö – vägledning

Grupp 1FreeStyle Comfort använder endast radiofrekvensenergi för interna
funktioner. Därför är RF-emissionerna väldigt låga och det ärinte sannolikt att de inverkar störande på elektronisk utrustning inärheten.

Klass B
Klass AÖverensstämmer

PN MN232-C4 K | Bruksanvisning

FreeStyle Comfort lämpar sig för användning i alla inrättningar,
inklusive hemmiljöer och sådana som är direktanslutna till det
allmänna elnätet med lågspänning som försörjer byggnader som
används som bostäder.

FreeStyle Comfort
Vägledning och tillverkarens deklaration – elektromagnetisk immunitet

FreeStyle Comfort är avsedd för användning i den elektromagnetiska miljö som specificeras nedan. Kunden eller
användaren av FreeStyle Comfort ska säkerställa att den används i en sådan miljö.
Immunitetstest
IEC 60601-testnivå
Överensstämmelsenivå Elektromagnetisk miljö – vägledning
Golven ska vara av trä, betong eller kakel.
Elektromagnetisk±8 kV kontakt±8 kV kontakt
Om golven är täckta med syntetmaterial
miljö – vägledningska den relativa fuktigheten vara minst±15 kV lufturladdning±15 kV lufturladdning
IEC 61000-4-230 %.
±2 kV för strömförsörj±2 kV för strömförsörjSnabba elektriska
ningsledningar
Nätströmmen ska vara samma som i enningsledningar
transienter/typisk professionell sjukvårdsinrättningtransientskurar
eller hemsjukvårdsmiljö.
±1
kV
för
in-/utgångsledIEC 61000-4-4
Ej tillämpligt
ningar
±1 kV ledning(ar) till
ledning(ar)Toppar
IEC 61000-4-5

Spänningsfall,korta avbrott och
spänningsvariationer
hosströmförsörjningens
inloppsledningar IEC
61000-4-11

Spänningsfrekvensens (50/60 Hz)
magnetfält, IEC
61000-4-8±2 kV normalläge för
växelströmledningar±1 kV ledning(ar) till
ledning(ar)±1 kV differentialläge för
växelströmledningar±2 kV normalläge för I/Okopplingar utomhus
0 % UT i 0,5 cykel (0°,0 % UT i 0,5 cykel (0°, 45°,45°, 90°, 135°, 180°,
90°, 135°, 180°, 225°,
225°, 270°, 315°)
270°, 315°)
0 % UT i 1 cykel (00)

0 % UT i 1 cykel (00)

70 % UT (30 % fall i UT) i
25/30 cykler (00)

70 % UT (30 % fall i UT) i
25/30 cykler (00)

0 % UT i
250/300 cykler (00)

0 % UT i
250/300 cykler (00)

3 A/m

3 A/m

Nätströmmen ska vara samma som i en
typisk professionell sjukvårdsinrättningeller hemsjukvårdsmiljö.

Nätströmmen ska vara samma som i en
typisk professionell sjukvårdsinrättningeller hemsjukvårdsmiljö. Om användaren
av FreeStyle Comfort behöver kontinuerlig
drift under strömavbrott rekommenderar vi
att FreeStyle Comfort drivs med avbrottsfri
kraftförsörjning eller med ett batteri.

Magnetfält som uppstår på grund
av spänningsfrekvens ska ligga
på nivåer som är typiska för enprofessionell sjukvårdsinrättning ochhemsjukvårdsmiljö.

OBS! UT är nätspänningen innan testnivån tillämpas.

OBS! I vissa fall kan Freestyle Comfort återställas efter en betydande elektrostatisk urladdning och kan kräva en
fulleffektscykel för att återuppta normal drift. Se felsökningsavsnittet för mer information.

PN MN232-C4 K | Bruksanvisning

SWE - 309

FreeStyle Comfort
Vägledning och tillverkarens deklaration – elektromagnetisk immunitet

FreeStyle Comfort är avsedd för användning i den elektromagnetiska miljö som specificeras nedan. Kunden eller
användaren av FreeStyle Comfort ska säkerställa att den används i en sådan miljö.
Immunitetstest
IEC 60601-testnivå
Överensstämmelsenivå Elektromagnetisk miljö – vägledning
Ledningsburen RF 10 Vrms
10 Vrms
Portabel och mobil RFkommunikationsutrustning ska inte
IEC 61000-4-6
150 kHz till 80 MHz
användas närmare någon del av
FreeStyle Comfort, inklusive kablarna,
än det rekommenderade minsta
separationsavstånd som beräknats med
10 V/mUtstrålad RF
10 V/mden ekvation som är tillämplig för sändarens
IEC 61000-4-3
80 MHz till 2700 MHz
frekvens.
Rekommenderat separationsavstånd

d = 1,2√ P
d = 1,2√ P = 80 MHz till 800 MHz
d = 1,2√ P = 800 MHz till 2 700 MHz
där P är sändarens maximala uteffekt i watt(W) enligt sändartillverkaren och d är det
rekommenderade separationsavståndet imeter (m).
Fältstyrkor från fasta RF-sändare, enligt vad
som fastställts genom en elektromagnetisk
anläggningsundersökninga, ska vara mindreän överensstämmelsenivån för varje
frekvensområdeb.
Störningar kan inträffa i närheten av
utrustning som är märkt med följandesymbol:OBS 1! Vid 80 MHz och 800 MHz gäller det högre frekvensområdet.
OBS 2! Dessa riktlinjer kanske inte gäller i alla situationer. Elektromagnetisk vågutbredning påverkas av absorption ochreflektion från byggnader, föremål och människor.
aFältstyrkor från fasta sändare, t.ex. basstationer för radiotelefoner (mobila/trådlösa) och fast mobilradio, amatörradio,
AM- och FM-radiosändningar och TV-sändningar, kan inte förutses teoretiskt med noggrannhet. För att fastställa
den elektromagnetiska miljön till följd av fasta RF-sändare bör en elektromagnetisk anläggningsundersökningövervägas. Om den uppmätta fältstyrkan på den plats där FreeStyle Comfort används överstiger den tillämpliga RFöverensstämmelsenivån enligt ovan, ska FreeStyle Comfort hållas under uppsikt för verifiering av normal drift. Om onormal
prestanda observeras kan ytterligare åtgärder bli nödvändiga, såsom att rikta om eller flytta FreeStyle Comfort.
bÖver frekvensområdet 150 kHz till 80 MHz ska fältstyrkorna vara mindre än 10 V/m.

310 - SWEPN MN232-C4 K | Bruksanvisning

FreeStyle Comfort
Rekommenderade separationsavstånd mellan bärbar och mobil RFkommunikationsutrustning och FreeStyle Comfort

FreeStyle Comfort är avsedd för användning i en elektromagnetisk miljö där utstrålade RF-störningar är under kontroll.
Kunden eller användaren av FreeStyle Comfort kan hjälpa till att förhindra elektromagnetiska störningar genom att
upprätthålla ett minimiavstånd mellan bärbar och mobil RF-kommunikationsutrustning (sändare) och FreeStyle Comfort
enligt rekommendationerna nedan, i enlighet med kommunikationsutrustningens maximala uteffekt.
Sändarens maximala
Separationsavstånd enligt sändarens frekvens
nominella uteffektmW

150 kHz till 80 MHz

d = 1,2√ P

80 MHz till 800 MHz

800 MHz till 2,5 GHz

d = 1,2√ P

d = 1,2√ P

0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
För sändare med en maximal uteffekt vilka inte listas ovan kan rekommenderat separationsavstånd d i meter (m) beräknas
med hjälp av den ekvationen som tillämpas för sändarens frekvens, där P är sändarens maximala uteffekt i watt (W) enligt
sändartillverkaren.
OBS 1! Vid 80 MHz och 800 MHz är det separationsavståndet för det högre frekvensområdet som gäller.
OBS 2! Dessa riktlinjer kanske inte gäller i alla situationer. Elektromagnetisk vågutbredning påverkas av absorption ochreflektion från byggnader, föremål och människor.

Klassificering

Typ av skydd mot elstöt:
Klass II
Skydd mot elstötar uppnås genom dubbel
isolering.

enligt RTCA/DO160 avsnitt 21, kategori M.
• CISPR 11/EN 55011 klass B grupp 1, ”Industriell,vetenskaplig och medicinsk (ISM) utrustning”
• FCC del 15, underavsnitt B – oavsiktligt
utstrålande enheter av klass B

Grad av skydd mot elektrisk stöt:
BF-typUtrustningen ger en viss grad av skydd
mot elstötar med avseende på		 1) tillåtna läckströmmar		2) skyddsjordsanslutningens tillförlitlighet(i förekommande fall).
		 Ej avsedd för applicering direkt på hjärtat.

IP22 – Skydd mot intrång av fingrar eller liknande
föremål som är större än 12,5 mm i diameter.
Vertikalt droppande vatten ska inte ha någon skadlig
effekt när höljet lutas i en vinkel på 15° från sin
normala position.

Oberoende tester gällande standarder för elektrisk
utrustning för medicinskt bruk:
Utrustningen har testats för att uppfylla kraven
enligt• IEC 60601-1 utgåva 3.1: Elektrisk utrustning
för medicinskt bruk – del 1: Allmänna fordringar
beträffande säkerhet.
• CAN/CSA C22.2 nr 60601-1-8 utgåva 2.1,
elektrisk utrustning för medicinskt bruk – del 1:
Allmänna fordringar beträffande säkerhet
Skydd mot potentiella elektromagnetiska störningar
eller andra störningar mellan utrustningen och andra
enheter.
• Utrustningen har testats för att uppfylla kraven i
EN 60601-1-2/IEC 60601-1-2, 4:e utgåvan.
• Utrustningen har testats för att uppfylla kravenTillåten metod för rengöring och desinfektion: Se
avsnittet ”Rengöring, skötsel och korrekt underhåll”i den här användarhandboken till FreeStyle.
Grad av säkerhet vid användning i närheten av
antändliga anestesigaser:
Utrustningen är inte lämplig för sådan användning.
Driftsätt: Kontinuerlig drift.

CAIRE och CAIRE Inc. är registrerade varumärken som tillhör CAIRE Inc. Besök vår
webbplats nedan för en fullständig lista över varumärken.
Varumärken: www.careinc.com/corporate/trademarks.
Copyright © 2021 CAIRE Inc. CAIRE Inc. förbehåller sig rätten att upphöra med
försäljning av sina produkter eller ändra priser, material, utrustning, kvalitet, beskrivningar,
specifikationer och/eller processer för sina produkter när som helst utan förvarning ochutan vidare förpliktelser eller följder. Vi förbehåller oss alla rättigheter som inte uttryckligen
anges häri om tillämpligt.

PN MN232-C4 K | Bruksanvisning

SWE - 311

FreeStyle Comfort

Säätimet ja järjestelmän tilan merkkivalot
ISO 7000; laitteessa käytettävät graafiset symbolit –hakemisto ja yhteenveto
Lue käyttöopas ennen käyttöä.
Sääntö 1641
Varastointi- tai käyttölämpötila-alue.
Sääntö 0632

Varastoinnin kosteusalue. Sääntö 2620

Neuvoston direktiivi 93/42/ETY lääkinnällisistä
laitteista
Valtuutettu edustaja Euroopan
yhteisössä
Jos tuotteen yksilöivässä
tunnistemerkinnässä (UDI) on CEmerkki, laite on direktiivin 93/42 /ETY lääkinnällisiä laitteita koskevien
vaatimusten mukainen. Symboli CE#### tarkoittaa ilmoitetun laitoksen
numeroa.
IEC 60417

Ilmanpainerajoitus. Sääntö 2621

Luokan II laite, kaksinkertaisesti
eristetty. Sääntö 5172

Valmistajan nimi ja osoite. Sääntö 3082

Virtakytkin. Sääntö 5009

Valmistuspäivämäärä. Sääntö 2497

ETL-sertifiointi
Turvallisuusvirasto CAN/CSA C22.2
nro 601.1 M90 lääketieteellisiä laitteita
vartenLuettelonumero. Sääntö 2493
Sarjanumero. Sääntö 2498

Sisäiset symbolit

Tämä puoli ylös. Sääntö 0623

Pidä kaukana syttyvistä materiaaleista,öljystä ja rasvasta.

Särkyvää, käsittele varoen.
Sääntö 0621

Ei saa purkaa osiin.

Säilytettävä kuivassa ja pidettävä
kuivana. Sääntö 0626

Lisää virtausta.

Pinoamisen lukumäärärajoitus.
Sääntö 2403

Vähennä virtausta.

ISO 7010: Graafiset symbolit—Turvavärit ja
turvamerkit—Rekisteröidyt turvamerkit

Valikkopainike.

Käyttöohjeet on luettava. Sääntö M002
Pidettävä kaukana avotulesta, tulesta ja
kipinöistä. Avoimet syttymislähteet ja
tupakointi kielletty. Sääntö P003

Täyttää RTCA DO160 -osan 21
luokan M vaatimukset. FAA SFAR
106 -vaatimus

Älä tupakoi yksikön lähellä tai kun
käytät sitä. Sääntö P002

Luokan II laite, kaksoiseristetty

Tyypin BF potilaskosketuksessa
oleva osa (sähköiskusuojauksen taso).
Sääntö 5333

FAA:n hyväksymä – POC

Varoitus. Sääntö W001

312 - FIN

RTCA/DO-160osa 21
Täyttää
luokan M
vaatimukset

PN MN232-C4 K | Käyttöopas

FreeStyle Comfort
21 CFR 801.15: Code of Federal Regulations Title 21(Yhdysvallat)
Yhdysvaltojen liittovaltion laki sallii
tämän laitteen myynnin vain lääkärilletai lääkärin määräyksestä.
Neuvoston direktiivi 2012/19/EU: sähkö- ja
elektroniikkalaiteromu (WEEE)
WEEE

IEC 60601-1: Sähkökäyttöiset lääkintälaitteet - Osa 1:
Yleiset vaatimukset turvallisuudelle ja olennaiselle
suorituskyvylle
Pisarasuojattu
Tämä tuote saattaa olla suojattu yhdellä tai useammalla
patentilla Yhdysvalloissa ja muissa maissa. Alla
olevalla verkkosivullamme on lueteltu kaikki
sovellettavat patentit. Pat.: www.caireinc.com/
corporate/patents/.

Happirikastimen toiminta

hengitystiesairautesi hoitoon.

Hengittämämme ilma sisältää noin 21 %happea, 78 % typpeä ja 1 % muita kaasuja.
FreeStyle® Comfort® -happirikastimeen imetään
huoneilmaa ilmanottoaukkojen kautta. Ilma kulkee
molekyylisiiviläksi kutsutun adsorboivan aineen läpi.
Vain happi läpäisee sen, joten typpi suodattuu pois.
Siksi käyttäjä saa hengitettäväkseen puhdasta happea.

Vaikka happihoito ei aiheuta riippuvuutta, se voi olla
vaarallista itsehoitona. Ota yhteys lääkäriin ennen
tämän happirikastimen käyttämistä. Happirikastimen
jälleenmyyjä esittelee, kuinka lääkärin määräämä
jatkuva syöttö säädetään.

Tehokas ja helposti mukana kuljetettava FreeStyle
Comfort -happirikastin käyttää kehittynyttä happea
säästävää tekniikkaa ja mahdollistaa kotihoidon.
Edistyksellinen virtaustoiminto tunnistaa, koska
käyttäjä hengittää sisään ja tuottaa happisykäyksen
kunkin sisäänhengitysjakson alkaessa. Tästä
toiminnosta on hyötyä harrastuksissa ja kodin
ulkopuolella liikuttaessa. Sen ansiosta happea ja akun
varausta kuluu vähemmän kuin jatkuvaa virtausta
käytettäessä. FreeStyle Comfort on kevyt, mikä
mahdollistaa sen kuljettamisen olkahihnassa. Laitetta
voi käyttää verkkovirralla (seinäpistoke), ajoneuvosta
saatavalla tasavirralla tai akun avulla.

Happirikastimen avulla syötetään täydentävää
happea käyttäjille, joiden keuhkot eivät pysty
siirtämään sitä riittävästi verenkiertoon. Kannettavissa
happirikastimissa ei ole happisäiliötä. Niitä ei tarvitse
täyttää, ja niiden akut voidaan ladata verkkovirralla tai
tasavirralla missä vain (lentokonetta lukuun ottamatta).
Siksi niitä voidaan käyttää helposti kotihoidossa tai
liikkeellä oltaessa. Happirikastimen käyttäminenedellyttää lääkärin antamaa laitetta. Sitä ei ole
tarkoitettu elämää ylläpitävään käyttöön.

Käyttötarkoitus
CAIRE FreeStyle Comfort -happ konsentraattori
on tarkoitettu lisähapen annosteluun. Laitetta ei ole
tarkoitettu elämän tukemiseen, eikä se tarjoa potilaan
tarkkailuvalmiuksia.

Lääkärin määräämä happihoito
Monet kärsivät erilaisista sydän-, keuhko- ja
hengitystiesairauksista. Monille heistä on runsaasti
hyötyä happihoidosta kotona, matkustettaessa tai
poissa kotoa.

Käyttäjät

Vaikka happihoitoa voidaan määrätä kaiken
ikäisille, tyypillinen käyttäjä on yli 65-vuotias, jolla
on jokin keuhkosairaus, kuten keuhkoahtauma.
Käyttäjillä on yleensä hyvät kognitiiviset kyvyt, ja
heidän täytyy pystyä ilmoittamaan, jos olo muuttuu
huonoksi. Jos käyttäjä ei pysty ilmoittamaan
olon muuttumisesta huonoksi tai lukemaan
ja ymmärtämään happirikastimen merkintöjä
ja käyttöohjetta, on suositeltavaa käyttää sitä
valvotusti. Jos happirikastimen käytön aikana
ilmenee epämukavuutta, käyttäjän tulee ottaa
yhteys terveydenhoidon ammattilaiseen. Lisäksi on
suositeltavaa, että käyttäjällä on varahappijärjestelmä(esimerkiksi happipullo) sähkökatkon ja
happirikastimen vian varalta. Happirikastimen
käyttäminen ei vaadi erityisiä taitoja.

Happi on kaasua, jota hengitysilmastamme on
21 %. Kehomme oikea toiminta riippuu tasaisesta
hapensaannista. Lääkärisi on määrännyt happihoitoa
PN MN232-C4 K | Käyttöopas

FIN - 313

FreeStyle Comfort

FreeStyle Comfort -happirikastimen purkaminen pakkauksestaan
Tarkista, että pakkaus sisältää
kaikki luetellut osat. Jos
jotakin puuttuu, ota heti yhteys
happirikastimen toimittajaan.

Vakiovarusteet• Kannettava happirikastin• Kantolaukku• Olkahihna• 8-kennoinen akkupakkaus• Vaihtovirtalähde• Virtajohto• Tasavirtalähde• Käyttöopas

Valinnaiset varusteet• 8-kennoinen lisäakkupakkaus• 16-kennoinen akkupakkaus• Pöytälaturi• Kannettava tarvikelaukku

Tutustuminen FreeStyle Comfort -happirikastimeen
Tutustu FreeStyle Comfortin tärkeimpiin ominaisuuksiin ja sen käyttöliittymään.

Laite edestä
Hihnakiinnitys

Käyttöliittymä

Happilähtö
Poistoilmaaukko

Ilmanotto/
suodattimet
Ilmanotto/
suodattimet

Virtaliitäntä

Akkupakkaus
Akun vapautuspainike
314 - FINPN MN232-C4 K | Käyttöopas

FreeStyle Comfort
Laite päältä
Näyttöruutu
Virtapainike

Lisää virtausta

Valikkopainike

Vähennä virtausta

Akun näyttöruutu
Virtausasetus

Akun varausnäyttö

autoSAT-merkki

Hälytysnäyttö

Jäljellä olevan
akun varaus
prosentteina

Bolussyötön
merkkivalo

Sähkövirran näyttöruutu

Virtausasetus
Akun varausnäyttö

Tila
hälytysnäytölle

Latauksenmerkkivalo

PN MN232-C4 K | Käyttöopas

FIN - 315

FreeStyle Comfort

Tärkeää!

Turvaohjeet on merkitty seuraavasti:
VAROITUS: TÄRKEITÄ TURVATIETOJAVAAROISTA, JOTKA VOIVAT AIHEUTTAAVAKAVIA VAMMOJA.
TÄRKEÄ HUOMAUTUS: tärkeää
tietoa FreeStyle Comfortin vaurioiden
estämiseksi.
Huomautus: nämä tiedot vaativat erityistä huomiota.

Käyttöindikaatiot
VAROITUS: ON TÄRKEÄÄ, ETTÄNOUDATAT LÄÄKÄRIN SINULLE
MÄÄRÄÄMÄÄ HAPPITASOA. ÄLÄ MUUTAVIRTAUSASETUSTA ILMAN LÄÄKÄRIN
MÄÄRÄYSTÄ.
VAROITUS: YHDYSVALTAIN LIITTOVALTION LAIN
MUKAISESTI TÄTÄ LAITETTA SAA OSTAA TAIVUOKRATA VAIN LÄÄKÄRIN TAI MUUN LISENSOIDUNTERVEYDENHUOLLON PALVELUNTARJOAJAN
MÄÄRÄYKSESTÄ.
VAROITUS: TÄTÄ YKSIKKÖÄ EI OLE TARKOITETTUELÄMÄÄ YLLÄPITÄVIIN SOVELLUKSIIN. VANHUS,LAPSI TAI MUU POTILAS, JOKA EI PYSTYILMAISEMAAN LAITTEEN KÄYTÖN AIHEUTTAMAAEPÄMUKAVAA OLOA, SAATTAA TARVITAYLIMÄÄRÄISTÄ TARKKAILUA. KUULO- JA/TAINÄKÖVAMMAISET KÄYTTÄJÄT VOIVAT TARVITAAPUA HÄLYTYSTEN TARKKAILUSSA.
VAROITUS: PULSSIANNOSTELUTILAN ASETUS ONMÄÄRITETTÄVÄ YKSILÖLLISESTI KUNKIN POTILAANTARPEIDEN MUKAISESTI LEVOSSA, LIIKUNNANAIKANA JA MATKALLA.
VAROITUS: POTILAAN TIETYNLAINEN HENGITYS EIEHKÄ LAUKAISE HAPPEA SÄÄSTÄVÄÄ TOIMINTOA.
VAROITUS: FREESTYLE COMFORT-HAPPIRIKASTINTA EI OLE TARKOITETTU
TRAKEOSTOOMAPOTILAAN KÄYTTÖÖN.
VAROITUS: RASKAANA OLEVIEN TAI IMETTÄVIEN
EI PIDÄ KÄYTTÄÄ TÄMÄN KÄYTTÖOHJEENSUOSITTELEMIA VARUSTEITA, SILLÄ NE SAATTAVATSISÄLTÄÄ FTALAATTEJA.

316 - FINPN MN232-C4 K | Käyttöopas

Käytön vasta-aiheet
VAROITUS: JOISSAKIN OLOSUHTEISSAVOI OLLA VAARALLISTA ANNOSTELLALISÄHAPPEA ILMAN LÄÄKÄRIN
MÄÄRÄYSTÄ. TÄTÄ LAITETTATULEE KÄYTTÄÄ VAIN LÄÄKÄRIN
MÄÄRÄYKSESTÄ.
VAROITUS: EI SAA KÄYTTÄÄ HELPOSTI SYTTYVIENANESTESIA-AINEIDEN LÄHELLÄ.
VAROITUS: KAIKKIEN SÄHKÖLAITTEIDEN TAVOIN
TÄMÄ HAPPIRIKASTIN EI OLE KÄYTETTÄVISSÄ,JOS SÄHKÖNSYÖTTÖ SIIHEN KATKEAA TAIPÄTEVÄ HUOLTOTEKNIKKO HUOLTAA SITÄ. TÄMÄ
HAPPIRIKASTIN EI SOVI KÄYTTÄJILLE, JOIDENTERVEYTEEN TILAPÄINEN KATKOS VAIKUTTAAHAITALLISESTI.

Turvallisuusohjeet
VAROITUS: TÄTÄ LAITETTA EI SAA
MUOKATA.
VAROITUS: LAITE ANNOSTELEE ERITTÄIN
HAPPIPITOISTA ILMAA, JOKA KIIHDYTTÄÄ
PALAMISTA. TUPAKOINTI JA AVOTULENKÄYTTÄMINEN ON KIELLETTYÄ SAMASSAHUONEESSA (1) TÄMÄN LAITTEEN KANSSA TAI(2) MUUN HAPPILISÄVARUSTEEN KANSSA. TÄMÄN
VAROITUKSEN LAIMINLYÖMINEN VOI AIHEUTTAAVAKAVAN TULIPALON, OMAISUUSVAHINKOJA,FYYSISIÄ VAMMOJA TAI KUOLEMAN.
VAROITUS: ÄLÄ KÄYTÄ LAITETTA AHTAASSATAI SULJETUSSA TILASSA (KUTEN PIENESSÄLAATIKOSSA TAI KÄSILAUKUSSA), JOSSAILMANVAIHTO ON HEIKKO. SE VOI AIHEUTTAA
HAPPIRIKASTIMEN YLIKUUMENEMISEN JA TEHONHEIKKENEMISEN.
VAROITUS: HAPPIRIKASTINTA EI SAA ALTISTAASAVULLE, ILMANSAASTEILLE EIKÄ HÖYRYILLE.
VAROITUS: JOS LAITTEESSA TAPAHTUUVAKAVA VAARATILANNE, KÄYTTÄJÄN TULEEILMOITTAA TAPAUKSESTA VÄLITTÖMÄSTIPALVELUNTARJOAJALLE JA / TAI VALMISTAJALLE.
VAKAVANA TAPAHTUMANA MÄÄRITELLÄÄNLOUKKAANTUMINEN, KUOLEMA TAI MAHDOLLISUUSAIHEUTTAA VAMMOJA / KUOLEMIA, JOS TAPAHTUMATOISTUU. KÄYTTÄJÄ VOI MYÖS ILMOITTAATAPAHTUMASTA SEN MAAN TOIMIVALTAISELLEVIRANOMAISELLE, JOSSA TAPAHTUMA TAPAHTUI.

FreeStyle Comfort
VAROITUS: JOS KÄYTETÄÄN
HAPPILISÄVARUSTEITA, JOITA EI OLE
SUOSITELTU TÄLLE HAPPIRIKASTIMELLE,LAITTEEN TEHO VOI HEIKENTYÄ.
SUOSITELLUT LISÄVARUSTEET ON
MAINITTU TÄSSÄ KÄYTTÖOPPAASSA.
VAROITUS: JOS HAPPIRIKASTIN ON PUDONNUT,
VAHINGOITTUNUT TAI ALTISTUNUT VEDELLE, OTA
YHTEYTTÄ KOTIHOITOPALVELUUSI, JOTTA LAITETARKISTETAAN JA TARVITTAESSA KORJATAAN. ÄLÄKÄYTÄ HAPPIRIKASTINTA, JOS SEN VIRTAJOHTO TAIPISTOKE ON VAHINGOITTUNUT.
VAROITUS: HAPPIRIKASTIMEN VAHINGOITTUMISENTAI HENKILÖVAMMAN VÄLTTÄMISEKSI KIINNITÄ SE
TIUKASTI ESIMERKIKSI VYÖLLÄ, KUN KULJETATSITÄ AJONEUVOSSA.
VAROITUS: YHDYSVALTAIN LIIKENNEMINISTERIÖN(DOT) JA YHDISTYNEIDEN KANSAKUNTIEN (YK)SÄÄDÖSTEN MUKAISESTI LAITTEEN AKKU ON
POISTETTAVA KAIKILLA KANSAINVÄLISILLÄLENNOILLA, KUN HAPPIRIKASTIN KIRJATAAN
MATKATAVARAKSI. KUN HAPPIRIKASTIN
TOIMITETAAN POSTITSE, AKKU ON POISTETTAVALAITTEESTA JA PAKATTAVA ASIANMUKAISESTI.
VAROITUS: ÄLÄ KÄYTÄ TÄTÄ LAITETTA, ENNEN
KUIN OLET LUKENUT TÄMÄN KÄYTTÖOPPAAN
JA YMMÄRTÄNYT, MITÄ SIINÄ SANOTAAN. JOSET YMMÄRRÄ VAROITUKSIA JA OHJEITA, OTA
YHTEYTTÄ LAITETOIMITTAJAAN ENNEN LAITTEENKÄYTTÄMISTÄ. MUUTOIN SEURAUKSENA VOI OLLAHENKILÖVAHINKO TAI LAITEVAURIO.
VAROITUS: JOS OLOSI ON EPÄMUKAVA TAIKOHTAAT LÄÄKETIETEELLISEN HÄTÄTILAN,
HAKEUDU VÄLITTÖMÄSTI LÄÄKÄRINHOITOON.
VAROITUS: JOS HAPPIRIKASTINTA KÄYTETÄÄNSUOSITELTUJEN KÄYTTÖLÄMPÖTILOJENULKOPUOLELLA, HAPPIRIKASTIMEN TOIMINTA
EI EHKÄ VASTAA TEKNISISSÄ TIEDOISSAILMOITETTUA. TARKISTA LÄMPÖTILAT TÄMÄN
KÄYTTÖOPPAAN TEKNISISTÄ TIEDOISTA.
VAROITUS: ÄLÄ KÄYTÄ ÖLJYÄ, RASVAA TAIMAAÖLJYPOHJAISIA TAI MUITA HELPOSTISYTTYVIÄ TUOTTEITA HAPPILISÄVARUSTEIDENTAI HAPPIRIKASTIMEN KANSSA. KÄYTÄ VAIN
VESIPOHJAISIA, HAPEN KANSSA YHTEENSOPIVIAKOSTEUSEMULSIOITA TAI VOITEITA. HAPPIKIIHDYTTÄÄ HELPOSTI SYTTYVIEN AINEIDEN
PALAMISTA.

VAROITUS: ÄLÄ TUKI TULO- TAI POISTOILMAAUKKOJA. ÄLÄ PUDOTA TAI ASETA MITÄÄN
ESINEITÄ LAITTEEN AUKKOIHIN. SE VOI AIHEUTTAA
HAPPIRIKASTIMEN YLIKUUMENEMISEN JA TEHONHEIKKENEMISEN.
VAROITUS: JOS AKKUA KÄYTETÄÄN VÄÄRIN,AKKU VOI KUUMETA TAI SYTTYÄ PALAMAAN,
MISTÄ VOI SEURATA VAKAVA VAMMA. AKKUA EISAA PUHKAISTA, LYÖDÄ, TALLOA EIKÄ PUDOTTAA,EIKÄ SITÄ SAA ALTISTAA VOIMAKKAILLE ISKUILLEEIKÄ TÄRÄHDYKSILLE. VAHINGOITTUNEEN AKUN
KÄYTTÄMINEN VOI AIHEUTTAA HENKILÖVAHINKOJA.
VAROITUS: ÄLÄ OIKOSULJE AKUN NAPOJAYHDISTÄMÄLLÄ NE TOISIINSA AVAINTEN TAIKOLIKOIDEN KALTAISILLA METALLIESINEILLÄ.
MUUTOIN VOI AIHEUTUA KIPINÖINTIÄ TAIVOIMAKASTA KUUMENEMISTA.
VAROITUS: KUN KANNETTAVAA HAPPIRIKASTINTAKÄYTETÄÄN ULKONA VAIHTOVIRTALÄHTEENKANSSA, KYTKE VIRTALÄHDE AINOASTAANVIKAVIRTAKATKAISIMELLA VARUSTETTUUNPISTORASIAAN.
VAROITUS: HAPPIRIKASTINTA EI SAA KÄYTTÄÄMUIDEN LAITTEIDEN VIERESSÄ TAI PINOTTUNAMUIDEN LAITTEIDEN KANSSA. JOS LAITETTAJOUDUTAAN KÄYTTÄMÄÄN MUIDEN LAITTEIDEN
VIERESSÄ TAI PINOTTUNA NIIDEN KANSSA,LAITETTA PITÄÄ VALVOA NORMAALIN TOIMINNANVARMISTAMISEKSI.
VAROITUS: MUIDEN KUIN MÄÄRITETTYJEN
KAAPELEIDEN TAI SOVITTIMIEN KÄYTTÄMINEN,LUKUUN OTTAMATTA LÄÄKINNÄLLISEN
SÄHKÖLAITTEEN VALMISTAJAN TOIMITTAMIA
KAAPELEITA JA SOVITTIMIA, JOTKA ON
TARKOITETTU SISÄISTEN OSIEN VARAOSIKSI,VOI LISÄTÄ HAPPIRIKASTIMEN PÄÄSTÖJÄ TAIVÄHENTÄÄ SEN SUOJAUSTA.
VAROITUS: KÄYTÄ VAIN SITÄ SÄHKÖJÄNNITETTÄ,JOKA ON MÄÄRITETTY LAITTEESEENKIINNITETYSSÄ TEKNISTEN TIETOJEN TARRASSA.
VAROITUS: ÄLÄ KÄYTÄ JATKOJOHTOJA TÄMÄN
LAITTEEN KANSSA ÄLÄKÄ KYTKE LIIKAASÄHKÖLAITTEITA SAMAAN PISTORASIAAN.
JATKOJOHTOJEN KÄYTTÄMINEN VOI HEIKENTÄÄ
LAITTEEN TEHOA HUOMATTAVASTI. JOSYHTEEN PISTORASIAAN KYTKETÄÄN LIIKAASÄHKÖLAITTEITA, SÄHKÖTAULU VOI KUORMITTUA,MIKÄ VOI LAUKAISTA KATKAISIMEN/SULAKKEEN TAI
TULIPALOON, JOS KATKAISIN/SULAKE EI TOIMI.

PN MN232-C4 K | Käyttöopas

FIN - 317

FreeStyle ComfortTÄRKEÄ HUOMAUTUS: Jos laiteantaa varoituksen tai jos havaitset,
että happirikastin ei toimi oikein,katso lisätietoja tämän käyttöoppaan
vianetsintäosiosta. Jos ongelma ei ratkea,
ota yhteyttä laitetoimittajaan.
TÄRKEÄ HUOMAUTUS: Noudata valmistajan ohjeita,
jotta takuu ei kumoudu.
VAROITUS: ÄLÄ KOHDISTA LAITTEESEENMITÄÄN HUOLTOTOIMIA LUKUUNOTTAMATTA TÄSSÄ KÄYTTÖOPPAASSA
MAINITTUJA RATKAISUJA. ÄLÄ POISTAPANEELEJA. VAIN LAITETOIMITTAJA TAIPÄTEVÄ HUOLTOTEKNIKKO SAA POISTAALAITTEEN PANEELIT TAI HUOLTAALAITETTA.
VAROITUS: KÄYTÄ VAIN VALMISTAJANSUOSITTELEMIA LISÄVARUSTEITA. MUIDEN
LISÄVARUSTEIDEN KÄYTTÄMISESTÄ VOI AIHEUTUAVAARAA, HAPPIRIKASTIN VOI VAHINGOITTUAVAKAVASTI TAI TAKUU VOI KUMOUTUA.
VAROITUS: VALMISTAJA SUOSITTELEE
VARAHAPPIJÄRJESTELMÄÄ SÄHKÖKATKON,LAITTEEN HÄLYTYSTILAN TAI MEKAANISEN VIAN
VARALTA. PYYDÄ LISÄTIETOJA TARVITTAVASTAVARAJÄRJESTELMÄSTÄ LÄÄKÄRILTÄSI TAILAITETOIMITTAJALTASI.
VAROITUS: ÄLÄ TUKI TULO- TAI POISTOILMAAUKKOJA. ÄLÄ PUDOTA TAI ASETA MITÄÄN
ESINEITÄ LAITTEEN AUKKOIHIN. SE VOI AIHEUTTAA
HAPPIRIKASTIMEN YLIKUUMENEMISEN JA TEHONHEIKKENEMISEN.
VAROITUS: TUPAKOINTI HAPPILAITTEENKÄYTTÄMISEN YHTEYDESSÄ ON YLEISIN SYY
TULIPALOIHIN JA NIIHIN LIITTYVIIN KUOLEMIIN.
NOUDATA NÄITÄ TURVAVAROITUKSIA:
VAROITUS: TUPAKOINTI, KYNTTILÖIDEN
POLTTAMINEN JA AVOTULEN KÄYTTÄMINEN ON
KIELLETTYÄ SAMASSA HUONEESSA TÄMÄN
LAITTEEN TAI HAPPILISÄVARUSTEIDEN KANSSA.
VAROITUS: TUPAKOINTI HAPPIKANYYLIN
KÄYTTÄMISEN AIKANA VOI AIHEUTTAAPALOVAMMOJA KASVOILLE JA MAHDOLLISESTIKUOLEMAN.
VAROITUS: JOS POLTAT, NÄMÄ 3 VAIHETTA VOIVATPELASTAA HENKESI: SAMMUTA HAPPIRIKASTIN,
IRROTA KANYYLI JA POISTU HUONEESTA, JOSSALAITE ON.

318 - FINPN MN232-C4 K | Käyttöopas

VAROITUS: "TUPAKOINTI KIELLETTY – HAPPILAITE
KÄYTÖSSÄ" -KYLTIN ON OLTAVA NÄKYVÄSTI
ESILLÄ KOTONA TAI MUUSSA TILASSA, MISSÄ
HAPPILAITETTA KÄYTETÄÄN. POTILAILLE
JA HEIDÄN HOITAJILLEEN ON KERROTTAVATUPAKOINNIN VAAROISTA LÄÄKINNÄLLISEN HAPEN
LÄHEISYYDESSÄ JA SITÄ KÄYTETTÄESSÄ.
VAROITUS: ÄLÄ KÄYTÄ HAPPIRIKASTINTA HELPOSTISYTTYVIEN KAASUJEN LÄHELLÄ. SE VOI AIHEUTTAANOPEAN PALON, MISTÄ VOI OLLA SEURAUKSENAOMAISUUSVAHINKOJA, FYYSISIÄ VAMMOJA TAIKUOLEMA.
VAROITUS: IRROTETUN KANYYLIN ASETTAMINENVAATTEIDEN, VUODEVAATTEIDEN, SOHVANTAI MUUN TEKSTIILIMATERIAALIN PÄÄLLE VOIAIHEUTTAA RÄJÄHTÄVÄN PALON, JOS KANYYLIALTISTUU SAVUKKEELLE, LÄMMÖNLÄHTEELLE TAIAVOTULELLE.
VAROITUS: ÄLÄ JÄTÄ NENÄKANYYLIA VAATTEIDEN,
VUODEVAATTEIDEN TAI ISTUINTYYNYJEN PÄÄLLETAI ALLE. JOS LAITTEEN VIRTA ON KYTKETTY
MUTTA SE EI OLE KÄYTÖSSÄ, HAPPI TEKEE
MATERIAALISTA HELPOSTI SYTTYVÄN. KUN
HAPPIRIKASTIN EI OLE KÄYTÖSSÄ, KATKAISE SIITÄVIRTA.
VAROITUS: HAPPIRIKASTIMEN PITÄÄ OLLAPOIS PÄÄLTÄ JA IRTI VERKKOVIRRASTA ENNENAKKUPAKKAUKSEN IRROTTAMISTA.
VAROITUS: PIENET OSAT VOIVATMUODOSTAA TUKEHTUMISVAARAN.
VAROITUS: ÄLÄ KOSKE POISTOILMA-AUKKOON YLI
10 SEKUNTIA PIDEMPÄÄ AIKAA, SILLÄ SE SAATTAAOLLA LÄMMIN.
VAROITUS: TUULI TAI VOIMAKAS VETO SAATTAAHAITATA HAPPIHOITOA.
VAROITUS: TÄMÄ TUOTE VOI ALTISTAA SINUTKEMIKAALEILLE, MUKAAN LUKIEN NIKKELI, JOKA
KALIFORNIAN OSAVALTION TIETOJEN MUKAANAIHEUTTAA SYÖPÄÄ. KATSO LISÄTIETOJA SIVULTA
WWW.P65WARNINGS.CA.GOV.
TÄRKEÄ HUOMAUTUS: irrota
aina virtajohto seinästä ennen
verkkovirtalaitteen irrottamista
happirikastimesta.
TÄRKEÄ HUOMAUTUS: älä sijoita happirikastinta niin,
että virtajohtoon on vaikea päästä käsiksi.

FreeStyle ComfortTÄRKEÄ HUOMAUTUS: aseta hapen syöttöputket ja
virtajohdot siten, että niihin ei voi kompastua eivätkä
ne aiheuta kuristumisen vaaraa.
TÄRKEÄ HUOMAUTUS: Kun käytät happirikastinta
auton, veneen tai muun ajoneuvon tasavirtalähteellä,
varmista, että ajoneuvo on käynnistetty ja käynnissä,
ennen kuin kytket happirikastimen. Jos näitä ohjeita ei
noudateta, seurauksena voi olla, että virtalähde ei anna
virtaa happirikastimeen.
TÄRKEÄ HUOMAUTUS: Kun auto, jossa käytät
happirikastinta, ei ole käynnissä, irrota ja poista laite
autosta. Älä säilytä happirikastinta hyvin kuumassa tai
kylmässä autossa tai vastaavassa ympäristössä, jonka
lämpötila on hyvin korkea tai matala.
Huomautus: kannettavat ja siirrettävät radiotaajuiset
viestintävälineet voivat häiritä sähkökäyttöisiä
lääkintälaitteita.
TÄRKEÄ HUOMAUTUS: ÄLÄ pidä
happirikastinta tai virtalähdettä kytkettynä
ajoneuvoon, jos auton virta on kytketty
pois. Tämä voi tyhjentää ajoneuvon akun.
TÄRKEÄ HUOMAUTUS: Jos happirikastinta käytetään
tai säilytetään sen normaalien käyttölämpötilarajojen ja
painealueen (korkeus) ulkopuolella, sen suorituskyky
voi heikentyä, akun käyttöaika voi lyhentyä ja/tai akun
lataamiseen kuluva aika voi pidentyä. Tarkista säilytysja käyttölämpötilarajat tämän käyttöoppaan teknisistä
tiedoista.
TÄRKEÄ HUOMAUTUS: Jos happirikastimessa on
akku: Säilytä viileässä ja kuivassa paikassa, jotta akun
käyttöikä ei lyhene. Jos happirikastinta säilytetään
pitkään korkeassa lämpötilassa tai täyteen ladatun tai
täysin tyhjentyneen akun kanssa, akun käyttöikä voi
lyhetä. Älä yritä avata akkua. Se ei sisällä huollettavia
osia. Pidä akut lasten ulottumattomissa.
TÄRKEÄ HUOMAUTUS: Käytä VAIN valmistajan
toimittamia akkuja. Hävitä akku laitetoimittajan
tai paikallisten viranomaisten kierrätysohjeiden
mukaisesti.

Huomautus: Varmista, että kanyyli on asetettu kokonaan
ja pitävästi. Tämä varmistaa, että happirikastin voi
tunnistaa hengityksen oikein hapen annostelua varten.
Sisäänhengityksen aikana kuulet tai tunnet, kuinka happi
virtaa nenäkanyylin haaroihin.
Huomautus: Jos litiumakut altistuvat erittäin kuumalle
lämpötilalle, kun ne on ladattu täyteen tai ne ovat täysin
tyhjiä, ne voivat menettää kapasiteettinsa pysyvästi.
Ennen akkujen varastoimista pitkään on suositeltavaa
ladata niitä 25–50 %. Suositeltava säilytyslämpötila on–20 –20 °C (–4–68 °F).
Huomautus: Jos käytetään happilisävarusteita, joita
ei ole suositeltu tälle happirikastimelle, sen teho voi
heikentyä. Suositellut lisävarusteet on mainittu tässä
käyttöoppaassa.
Huomautus: Älä käytä happirikastinta, jostuloilmasuodatin ei ole paikallaan. Jos mukana on toinen
suodatin, aseta varasuodatin, ennen kuin puhdistat
likaisen suodattimen. Puhdista likainen suodatin
lämpimällä saippuavesiliuoksella ja anna sen kuivua
kokonaan ennen käyttöä.
Huomautus: valmistaja ei suosittele tämän laitteen
sterilointia.
Huomautus: hapen loppumisen vaaraa ei ole huoneessa,
jossa happirikastinta käytetään.
VAROITUS: HAPPIRIKASTINTA EI SAAKÄYTTÄÄ TAI LIITTÄÄ ULKOISEEN TASATAI VAIHTOVIRTALÄHTEESEEN ILMANAKKUA. JOS LAITTEESTA ON IRROTETTUAKKU, ÄLÄ YRITÄ KÄYNNISTÄÄ
SITÄ TAI YHDISTÄÄ SITÄ ULKOISEENVIRTALÄHTEESEEN.
IRROTA AKKU VASTA KUN LAITTEESTA ON
KATKAISTU VIRTA JA SE ON IRROTETTU
VIRRANSYÖTÖSTÄ. ÄLÄ YRITÄ POISTAA AKKUA,KUN LAITTEESEEN ON KYTKETTY VIRTA TAI SE ON
YHDISTETTYNÄ VIRTALÄHTEESEEN.
Huomautus: jos happirikastinta on säilytetty pitkään sen
normaalien käyttölämpötilarajojen ulkopuolella (ne on
kuvattu edellä näkyvässä taulukossa), sen on annettava
palautua normaaliin käyttölämpötilaan ennen sen
käynnistämistä.

PN MN232-C4 K | Käyttöopas

FIN - 319

FreeStyle Comfort

Tekniset tiedot

FreeStyle Comfort
Mitat (yksi akku)

25,4 x 18,5 x 7,9 cm (10,0 x 7,3 x 3,1 in)

Mitat (kaksi akkua)

27,9 x 18,5 x 7,9 cm (11,0 x 7,3 x 3,1 in)

Paino (yksi akku)

2,3 kg (5 lbs)

Paino (kaksi akkua)

2,7 kg (6 lbs)

Nimellisäänitaso***

39,93 dB (A) ± 0,42 dB (A) @ 2

Virtausasetukset

Sykäysasetus 1, 2, 3, 4, 5

Hapen ulostuloasetus

Asetus 1–210 ml/min
Asetus 2–420 ml/min
Asetus 3–630 ml/min
Asetus 4–840 ml/min
Asetus 5–1 050 ml/min(hengitystaajuudella 20 bpm)

Happipitoisuus*

90 % (+5,5 %/-3 %)Happimittari

Kyllä

Maksimilähtöpaine<30 PSI

Annosherkkyyden oletusasetus> –0,1 cm H2O (säädettävissä)

verkkovirta

100–240 V AC, 50–60 Hz, 120W Max

tasavirta

11–18 V DC (10 A maks.)

Akun uudelleenlatausaika

Yksi akku: 3,5 tuntia
Kaksi akkua: 6,0 tuntia

Akun kesto

Yhden akun kesto: Asetus 2 - jopa 4 tuntia
Kahden akun kesto: Asetus 2 - jopa 8 tuntia

Käyttölämpötilat**
Akun tyyppi
Akun kapasiteetti
Kuljetus- ja varastointilämpötila (laite)
Kuljetus- ja varastointilämpötila (akku)
Käyttökorkeus**
Käyttöympäristön kosteus

5–40 °C (41–104 °F)
litium-ioni
Yksisoluinen: 6700 mAh; Kaksoissolu: 13400 mAh–25...70 °C (–13...+158 °F), maks. 3 048 m (10 000 ft)–25...70 °C (–13...+158 °F), maks. 3 048 m (10 000 ft)–381...+3 048 m (–1 250–10 000 ft) (testattu 700–1 060 hPa:nilmanpaineelle)
15–95 % suhteellinen kosteus (ei kondensoiva)

Varastointikosteus

0 - 90% ei tiivisty

Rajoitettu takuu

3 vuotta laitteelle
1 vuosi akuille ja virtalähteille* Perustuu ilmanpaineeseen 14,7 psi (101 kPa) lämpötilassa 21 °C (70 °F)** Laite toimii kaikissa ympäristöissä niin kauan kuin ympäristö on määritellyillä lämpötila- ja kosteusalueilla. Jos happirikastinta
käytetään muunlaisissa olosuhteissa, se ei ehkä toimi teknisten tietojensa mukaisesti. Lämmitysnäyttö saattaa jäädä näkyviin,
kunnes happirikastin on saavuttanut tavoitepitoisuuden tai 15 minuutiksi. Jatka laitteen käyttöä kyseisenä aikana.
*** Äänitaso mitattu testimenetelmää kohden Nr. 14-1 10/2018 MDS-Hei.
Odotettavissa oleva käyttöikä on vähintään viisi vuotta. Siiviläyksiköt kestävät noin vuoden, jos rikastin on tämän ajan
varastoituna.
320 - FINPN MN232-C4 K | Käyttöopas

FreeStyle Comfort
Käyttöohjeet
1. Sijoita FreeStyle Comfort käytön kannalta sopivaan
paikkaan.
a. Happirikastimen ympärillä tulee olla hyvä
ilmankierto, jotta ilmanottosuodattimet ja
ilmanpoistoaukot eivät tukkeutuisi.
b. Happirikastin tulee asettaa paikkaan, jossa
merkkivalot näkyvät ja hälytykset kuuluvat
esteettä.
c. Varmista, että ilmanottosuodattimet ovat
paikoillaan ennen laitteen käyttämistä. Jos
laitteesta puuttuu suodatin, ota yhteyttä laitteen
toimittajaan.

3. Työnnä virtajohto laitteen etupuolella oikeassaalareunassa olevaan liittimeen.

2. Asenna akkupakkaus happirikastimeen.
a. Kohdista akku FreeStyle Comfort-happirikastimen keskelle kotelon alaosaa.

4. Yhdistä nenäkanyyli hapen syöttöliitäntään.
Varmista ennen käyttämistä, että liitos on tiivis.

HUOMAUTUS: muiden kuin suositeltujen kanyyleiden
käyttö voi heikentää laitteen tehoa.

Huomautus: Laitevalmistajalle: Seuraavia hapen
annostelun lisävarusteita suositellaan käytettäväksi
FreeStyle Comfort kanssa:
b. Työnnä akkua, kunnes kuulet napsahduksen
osoituksena sen lukkiutumisesta paikalleen.

• nenänkanyyli, jossa on 7 jalkaa (2,1 m) letkua(suurin LPM: 6 LPM): CAIRE-osanumero 5408-SEQ• Lisätietoja suositelluista lisävarusteista on saatavillaosoitteessa www.caireinc.com.

PN MN232-C4 K | Käyttöopas

FIN - 321

FreeStyle Comfort
5. Käynnistä FreeStyle Comfort painamalla
virtapainiketta ja valitse haluamasi virtausasetus
lisäämällä virtausta (+) tai vähentämällä virtausta (-).

autoSAT-sykäystilavuuden (ml)*asetus
Hengitysnopeus

1

2

3

4

5

15

10,5

21,0

31,5

42,0

52,5

20

10,5

21,0

31,5

42,0

52,5

25

10,5

21,0

31,5

42,0

42,0

30

10,5

21,0

31,5

35,0

35,0

35

10,5

21,0

30,0

30,0

30,0

40

10,5

21,0

26,3

26,3

26,3

*Toleranssi on +/- 15 %Laite antaa lyhyen äänimerkin, joka osoittaa, ettälaite on päällä.

Ei-autoSAT-sykäystilavuuden (ml)*asetus

Huomautus: Kun FreeStyle Comfort käynnistetään
ensimmäistä kertaa, virtauksen oletusasetus on 2. Jos
asetusta muutetaan, laite muistaa edellisen asetuksen
uuden käynnistyksen yhteydessä.

Hengitysnopeus

1

2

3

4

5

15

14,0

28,0

42,0

56,0

70,0

6. Aseta nenäkanyyli kasvoillesi noudattaen sen
valmistajan ohjeita. Hengitä normaalisti. Kun
hengität sisään, tunnet, että nenäkanyylistä virtaa
happisykäys.

20

10,5

21,0

31,5

42,0

52,5

25

8,4

16,8

25,2

33,6

42,0

30

7,0

14,0

21,0

28,0

35,0

35

6,0

12,0

18,0

24,0

30,0

40

5,3

10,5

15,8

21,0

26,3

7. Voit muuttaa virtausasetusta painamalla lisää
virtausta (+) tai vähemmän virtausta (-) haluamasi
virtausmäärän saavuttamiseksi. Kun virtausasetus
vaihtuu, ääni muuttuu.

8. Käännä FreeStyle Comfort pois päältä painamalla
virtapainikkeesta.
Huomautus: Näyttöruutu siirtyy valmiustilaan ja himmenee 20 sekunnin kuluttua. Näyttö herää valmiustilasta,
kun painat mitä tahansa painiketta.
9. Tietoja erikoistoiminnoista on kohdassa Valikonselausohjeet.

TUOTTEEN ERITYISTOIMINNOT
autoSAT®
CAIREn patentoitu autoSAT-ominaisuus antaa
FreeStyle Comfort -laitteelle toimittaa tasaisen määrän
happea (bolus) henkeä kohden, maksimaaliseen
hengitysnopeuteen saakka. Kun käyttäjän
hengitysnopeus nousee sellaisten toimintojen aikana
kuin kävely, autoSAT-ominaisuus säätää laitteen
toimintaa pitämään asetettu bolusmäärä.
UltraSense®
Herkkä sykäysannoksen käynnistin havaitsee
hengenvedon aikaisen painemuutoksen ja antaa
happisykäyksen sisäänhengityksen aikana.
autoDOSE™
Ominaisuus antaa Hengitystä ei havaittu -hälytyksen
yhteydessä happisykäyksiä 20 hengenvetoa minuutissa-nopeudella laitteen senhetkisen virtausasetuksen
mukaisesti, kunnes hälytystila raukeaa.
322 - FINPN MN232-C4 K | Käyttöopas

FreeStyle Comfort
Virtalähteet
FreeStyle Comfort voi saada virtansa kolmesta eri
lähteestä. Niitä ovat:• ladattava akkupakkaus• vaihtovirta• tasavirta
Toiminnasta kerrotaan seuraavissa luvuissa.
VAROITUS: KÄYTÄ VAIN TÄSSÄ
KÄYTTÖOHJEESSA KUVATTUJAVIRTALÄHTEITÄ. MUIDEN KUIN CAIRENTOIMITTAMIEN VIRTALÄHTEIDEN
JA AKKUJEN KÄYTTÄMINEN VOIVAURIOITTAA LAITETTA JA AIHEUTTAATOIMINTAHÄIRIÖN.

AKUN IRROTTAMINEN

Työnnä akun vapautuspainiketta alaspäin ja poispäin
FreeStyle Comfort -happirikastimesta. Pidä painiketta
painettuna ja irrota akku happirikastimesta vetämällä.
VAROITUS: HAPPIRIKASTIMENPITÄÄ OLLA POIS PÄÄLTÄ JA
IRTI VERKKOVIRRASTA ENNENAKKUPAKKAUKSEN IRROTTAMISTA.

AKKUPAKKAUS

FreeStyle Comfort -happirikastinta voidaan käyttää
ladattavalla akulla, joka toimitetaan laitteen
mukana. Kun happirikastin saa virtaa akusta,
mitään toimintarajoituksia ei aiheudu. Kaikki
virtausvaihtoehdot ovat tällöin käytettävissä.

AKKUPAKKAUKSEN LATAUKSENTARKISTAMINENAKUN ASENTAMINEN

Kohdista akku FreeStyle Comfort -happirikastimen
keskelle kotelon alaosaa. Työnnä akkua, kunnes
kuulet napsahduksen osoituksena sen lukkiutumisesta
paikalleen.

Voit tarkistaa akun varaustason sen ollessa asennettuna
FreeStyle Comfort -happirikastimeen akkukuvakkeesta
ja jäljellä olevasta käyttöajasta laitteen näytöltä.
80–100 %60–80 %

40–60 %

20–40 %20 %PN MN232-C4 K | Käyttöopas

FIN - 323

FreeStyle Comfort
Huomautus: jos akku latautuu, näytöllä näkyy pistotulpan
kuvake prosenttien sijaan.
Kun akkua ei ole asennettu FreeStyle Comfort-happirikastimeen, voit tarkistaa sen varaustason
asettamalla sen tasaiselle alustalle ja painamalla TESTpainiketta. TEST-painikkeen oikealla puolella näkyvät
akkumittarin merkkivalot (25–100 %) ilmaisevat akun
lataustason.

AKKUKÄYTTÖAIKA

Kun rikastinta ei ole kytketty ulkoiseen tasa- tai
vaihtovirtaan, se siirtyy automaattisesti akkukäytölle,jos laitteeseen on asennettu ladattu akku. Uuden
akun kesto kuvataan alla näkyvässä taulukossa.
Hengitysnopeus ei vaikuta akun kestoon.
Sykäysasetus

Yhden (8-kennoisen)
akun kesto

Kaksoisakun(16-kennoisen) kesto

1

Jopa 8 tuntia

Jopa 16 tuntia

2

Jopa 4 tuntia

Jopa 8 tuntia

3

Jopa 3 tuntia

Jopa 6 tuntia

4

Jopa 2,25 tuntia

Jopa 4,5 tuntia

5

Jopa 2 tuntia

Jopa 4 tuntia

TYYPILLINEN AKUN LATAUSAIKA

Tyypillinen akun latausaika täysin tyhjälle akulle on
yhden akun kohdalla 3,5 tuntia, kaksoisakun kohdalla
6,0 tuntia.

LATAAMINEN ENSIMMÄISEN KERRANFreeStyle Comfort -happirikastimen mukana toimitettu
uusi akku on ladattu tehtaalla vain osittain. Ennen
akun käyttämistä ensimmäistä kertaa lataa se täyteen
verkko- tai tasavirran avulla.

YLEISTÄ TIETOA LATAAMISESTAFreeStyle Comfort lataa akkupakkauksen aina, kun
se on asennettuna rikastimeen ja kun rikastin on
kytkettynä ulkoiseen vaihto- tai tasavirtalähteeseen(lentokonetta lukuun ottamatta).
Akku on täyteen ladattu, kun näytöllä näkyy 100 %.
Akun voi ladata myös lisävarusteena hankittavalla
pöytälaturilla.

324 - FINPN MN232-C4 K | Käyttöopas

Huomautus: akku ei ehkä ota varausta vastaan, jos se on
täysin tyhjentynyt.

AKUN KÄYTTÖIKÄ

FreeStyle Comfort -akut on suunniteltu kestämään
vähintään 300 lataus- ja purkausjaksoa.

FreeStyle Comfort
Vaihto- ja tasavirtalähde
FreeStyle Comfort toimitetaan vakiona vaihto- jatasavirtalähteillä varustettuna.
Vaihtovirtalähde sisältää seuraavat osat:
a. Vaihtovirtayksikkö 100–240 V, 50–60 Hz.
b. Virtajohto, joka kytketään
vaihtovirtayksikköön ja vastaavaan liittimeen.

Tasavirtalähde muodostuu johdosta, joka kytketään
suoraan laitteeseen ja tasavirtaliitäntään.

VAIHTOVIRTA

2. Liitä vaihtovirtajohto vaihtovirtayksikköön.

Vaihtovirtaa käytetään kotona ja paikoissa, joissa sitäon saatavana. FreeStyle Comfort toimii vaihtovirralla.
FreeStyle Comfort sisältää myös vaihtovirta-akun,
joka voidaan ladata aina, kun vaihtovirtaa on saatavilla(paitsi lentokoneessa). Akku latautuu riippumatta siitä,
onko FreeStyle Comfort päällä vai ei.
VAROITUS: ÄLÄ KÄYNNISTÄ
HAPPIRIKASTINTA ILMAN SIIHENASENNETTUA AKKUA.
FreeStyle Comfort -rikastimen kytkeminen
vaihtovirtaan:
1. Kytke vaihtovirtayksikkö FreeStyle Comfort-laitteen etupuolella oikeassa alakulmassa olevaanliittimeen. Paina liitin kunnolla pohjaan, jotta liitäntä
on tiukka.

3. Työnnä vaihtovirtajohdon pistoke pistorasiaan.

PN MN232-C4 K | Käyttöopas

FIN - 325

FreeStyle Comfort
4. Käynnistä FreeStyle Comfort painamalla
virtapainiketta ja valitse haluamasi virtausasetus
lisäämällä virtausta (+) tai vähentämällä virtausta (-).

3. Työnnä tasavirtajohdon toinen pää
auton tasavirtapistokkeeseen tai muuhun
tasavirtapistokkeeseen.

Huomautus: jos akku on ladattava uudelleen, näytön
akkukuvakkeessa näky yksi oranssi palkki.

4. Käynnistä FreeStyle Comfort painamalla
virtapainiketta ja valitse haluamasi virtausasetus
lisäämällä virtausta (+) tai vähentämällä virtausta (-).

TASAVIRTA

Huomautus: jos akku on ladattava uudelleen, näytön
akkukuvakkeessa näky yksi oranssi palkki.

Tasavirtaa on saatavana autoissa, veneissä ja muissa
moottoriajoneuvoissa. FreeStyle Comfort toimii myös
tasavirralla. FreeStyle Comfort lataa akkupakkauksen
tarvittaessa aina, kun tasavirtaa on saatavilla.
Akku latautuu riippumatta siitä, onko happirikastin
käynnistetty vai ei.
VAROITUS: ÄLÄ KÄYNNISTÄ
HAPPIRIKASTINTA ILMAN SIIHENASENNETTUA AKKUA.
VAROITUS: ÄLÄ KÄYTÄ LAITETTA 12 VDC:NPISTOKKEESTA, JOS AJONEUVO EI OLE KÄYNNISSÄ.
1. Käynnistä moottoriajoneuvo.
2. Työnnä tasavirtajohto laitteen etupuolella oikeassaalareunassa olevaan liittimeen. Paina liitin kunnolla
pohjaan, jotta liitäntä on tiukka.

326 - FINPN MN232-C4 K | Käyttöopas

FreeStyle Comfort
Puhdistus, ylläpito ja huolto

AKUSTA HUOLEHTIMINEN

MÄÄRÄAIKAISKUNNOSSAPITOÄlä yritä purkaa tai avata akkua.

Ennaltaehkäisevä huolto
Laite ei tarvitse huoltoa. Laitteen sisällä olevien
suodattimien on tarkoitus kestää laitteen käyttöikä. Jos
laite ei toimi oikein, laite on palautettava toimittajalle
korjattavaksi. FreeStyle Comfort -happirikastimen
sisäosien huolto on annettava laitetoimittajan
tehtäväksi.
Laiteen käyttäjä on vastuussa seuraavista
huoltotoimenpiteistä.
• Ilmanottosuodattimet täytyy puhdistaa kerranviikossa.
VAROITUS: ÄLÄ YRITÄ AVATA TAIIRROTTAA HAPPIRIKASTIMENMUOVISUOJIA. LAITTEEN SAA AVATA VAIN
PÄTEVÄ HUOLTOTEKNIKKO.

Älä anna akun altistua nesteille.

Älä altista akkua avotulelle.
Älä pudota akkua.
Pidä akku poissa lasten ulottuvilta.
Jos akkupakkaus on vahingoittunut, ota yhteyttä
laitetoimittajaan ennen käyttöä.
Älä aseta akkua mikroaaltouunin tai lieden lähelle.

AKUN SÄILYTTÄMINEN

Happirikastimen akun varauksen ei tarvitse purkautua
kokonaan ennen kuin akku ladataan uudelleen. On
suositeltavaa ladata happirikastimen akku jokaisen
käyttökerran jälkeen.

ILMANOTTOSUODATTIMETÄlä altista akkua kuumuudelle, kuten
auringonpaisteelle. Älä jätä sitä autoon. Muutoin akun
suorituskyky voi heikentyä ja käyttöikä lyhentyä.
Lisäksi akku voi kuumentua, savuta, syttyä tuleen tai
räjähtää. Akkua tulee säilyttää vakaassa ympäristössä,jossa suhteellinen ilmankosteus alittaa 70 % ja jossa
ei esiinny korroosiota aiheuttavia kaasuja. Vältä
esimerkiksi äkillisiä lämpötilan vaihteluita, koska
ne saattavat aiheuttaa kosteuden tiivistymistä jaruostumista.

1. Irrota kaksi suodatinta FreeStyle Comfort-happirikastimen edestä työntämällä ylös kyseistä
kielekettä ja vetämällä ne ulos.

Jos akku on käyttämättä vähintään kuukauden,
on suositeltavaa ladata se 50–75-prosenttisesti(2–3 merkkivaloa palaa). Kun akku on ladattu,
irrota se happirikastimesta.

FreeStyle Comfort -rikastimeen otetaan ilmaa laitteen
edessä alareunassa sijaitsevien ilmanottosuodattimien
kautta. Suodattimet estävät ilmassa olevan pölyn jamuiden suurien partikkeleiden pääsyn laitteeseen.
Ilmanottosuodattimet täytyy puhdistaa vähintään
kerran viikossa. Laitetoimittaja saattaa neuvoa
puhdistamaan laitteen useammin käyttöolosuhteista
riippuen. Puhdista ilmanottosuodattimet toimimalla
seuraavasti.

2. Pese molemmat suodattimet lämpimällä
saippualiuoksella ja vedellä.
3. Huuhtele suodattimet huolellisesti ja kuivaa
ylimääräinen vesi pehmeällä, nukkaamattomalla
kankaalla. Varmista, että suodattimet ovat täysin
kuivia, ennen kuin laitat ne takaisin.
4. Asenna takaisin puhdas, kuiva ja suodatin.
VAROITUS: ÄLÄ KÄYTÄ
HAPPIRIKASTINTA, JOS
TULOILMASUODATIN EI OLE
PAIKALLAAN.
PN MN232-C4 K | Käyttöopas

FIN - 327

FreeStyle Comfort
PUHDISTAMINEN

Katkaise laitteesta virta ja irrota se vaihto- taitasavirtalähteestä ennen puhdistamista. ÄLÄ suihkuta
nesteitä suoraan ulkokoteloon. Käytä kosteaa(ei märkää) liinaa tai sientä. Suihkuta liinaan tai
sieneen mietoa kotitalouden puhdistusaine kotelon ja
virtalähteen puhdistamiseksi. Noudata puhdistusaineen
valmistajan ohjeita.
VAROITUS: KATKAISE LAITTEESTAVIRTA JA IRROTA PISTOKESÄHKÖPISTORASIASTA ENNEN KUIN
PUHDISTAT LAITTEEN. MUUTOINSEURAUKSENA ON SÄHKÖISKUNJA PALOVAMMAN VAARA. VAINLAITETOIMITTAJA TAI PÄTEVÄHUOLTOTEKNIKKO SAA POISTAALAITTEEN KANNET TAI HUOLTAALAITETTA.
VAROITUS: HUOLEHDI, ETTÄ HAPPIRIKASTIN
TAI AKKU EI PÄÄSE KASTUMAAN JA ETTÄLAITTEEN SISÄÄN EI PÄÄSE NESTEITÄ. MUUTOINSEURAUKSENA VOI OLLA TOIMINTAHÄIRIÖ TAI
LAITTEEN SAMMUMINEN, JA SÄHKÖISKUN TAIPALOVAMMOJEN RISKI VOI LISÄÄNTYÄ.
VAROITUS: ÄLÄ SUIHKUTA NESTETTÄ SUORAAN
LAITTEESEEN. ESIMERKIKSI SEURAAVATKEMIKAALIT OVAT HAITALLISIA: ALKOHOLI JA
ALKOHOLIPOHJAISET TUOTTEET, VÄKEVÖITYYNKLOORIIN POHJAUTUVAT TUOTTEET(ETYLEENIKLORIDI) JA ÖLJYPOHJAISET TUOTTEET.
HAPPIRIKASTIMEN MUOVIOSIA EI SAA PUHDISTAATÄLLAISILLA AINEILLA, KOSKA NE VAURIOITTAVATLAITETTA.
VAROITUS: PUHDISTA RUNKO, OHJAUSPANEELI
JA VIRTAJOHTO VAIN MIEDOLLA KOTIKÄYTTÖÖNTARKOITETULLA PUHDISTUSAINEELLA. LEVITÄ SITÄKOSTEALLA LIINALLA TAI SIENELLÄ. PYYHI PINNATKUIVIKSI. ÄLÄ PÄÄSTÄ HAPPIRIKASTIMEN SISÄÄN
MITÄÄN NESTETTÄ. KIINNITÄ ERITYISTÄ HUOMIOTA
KANYYLIN HAPPILIITTIMEEN JA VARMISTA, ETTÄ SE
ON PÖLYTÖN, KUIVA JA PUHDAS.

328 - FINPN MN232-C4 K | Käyttöopas

VAROITUS: PUHDISTA AKKU KOSTEALLA (EIMÄRÄLLÄ) LIINALLA TAI SIENELLÄ. SUIHKUTA
LIINAAN TAI PESUSIENEEN MIETOA KOTIKÄYTTÖÖNTARKOITETTUA PUHDISTUSAINETTA. PUHDISTAAKKUKOTELO.
VAROITUS: ÄLÄ PUHDISTA AUTOMAATTISISSAPESUJÄRJESTELMISSÄ.

KANYYLIN VAIHTAMINEN

Noudata aina kanyylin valmistajan käyttöohjeita. Vaihda
nenäkanyyli ja happiletkut 3–6 kuukauden välein (joslaite on päivittäisessä käytössä, 3 kuukauden välein)valmistajan suositusten mukaisesti ja kanyylin sekä
happilaitetoimittajan suositusten mukaisesti. Lääkärisi
tai happirikastimen toimittaja antaa puhdistus- ja
vaihto-ohjeet.
Varaosia on saatavana happirikastimen toimittajalta.

FreeStyle Comfort
Hälytystilat
Prioriteetti

Hälytyksen
kuvaus1

Järjestelmän
toimintahäiriö,
akun korkea
lämpötila

Näyttö

Äänimerkki

Lisätietoja/toimenpide-ehdotus

Yksi piippaus
20 sekunnin
väleinLaite antaa hälytyksen, jos akun lämpötila
lähestyy turvallisuussyistä tapahtuvan
sulkemisen kynnysarvoa.
Jos sulkemisen kynnysarvo saavutetaan, laite
sammuu. Jos ulkoinen virtalähde on liitettynä,
hälytysnäyttö tulee näkyviin ja äänimerkki
jatkuu kahden minuutin ajan.
Testaa laite niin, että akku on
huoneenlämpöinen. Jos ongelma jatkuu, otayhteyttä laitetoimittajaan.

2

Järjestelmän
toimintahäiriö,
korkea lämpötila

Yksi piippaus
20 sekunnin
väleinLaite antaa hälytyksen, jos sisäinen lämpötila
lähestyy sulkemisen kynnysarvoa.
Jos sulkemisen kynnysarvo saavutetaan,laite sammuu, hälytysnäyttö tulee näkyviin jaäänimerkki jatkuu kahden minuutin ajan.
Varmista, että laitteen ilma-aukoissa ei ole
tukoksia. Jos ongelma jatkuu, ota yhteyttä
laitetoimittajaan.

3

Järjestelmän
toimintahäiriö,
korkea paine

Yksi piippaus
20 sekunnin
väleinJos sulkemisen kynnysarvo saavutetaan,laite sammuu, hälytysnäyttö tulee näkyviin jaäänimerkki jatkuu kahden minuutin ajan.
Jos ongelma jatkuu, ota yhteyttä
laitetoimittajaan.

4

Hengitystä ei
havaittu

Yksi piippaus
30 sekunnin
väleinHengitystä ei havaittu 15 sekuntiin.
Aktivoituu ensimmäisen hengenvedon jälkeen
tai kun lämmitysnäyttö sammuu sen mukaan,
kumpi ehto täyttyy aikaisemmin.
Tarkista kanyyli vääntymien tai vuotojen
varalta. Lisää herkkyysasetusta. Jos ongelma
jatkuu, ota yhteyttä laitetoimittajaan.

5

6

Akun vähäisen
virran aiheuttama sammuttaminen

Yksi piippaus
30 sekunnin
väleinHälyttää, kunnes akun virta loppuu ja laite
sammuu.

Lämmitys

Ei mitään

Laite käynnistyy automaattisella annostelulla,
kunnes hengitys havaitaan tai lämmitys
loppuu, sen mukaan, kumpi ehto täyttyy
aikaisemmin.

Lataa akku.

PN MN232-C4 K | Käyttöopas

FIN - 329

FreeStyle Comfort
Prioriteetti

Hälytyksen
kuvausNäyttö

Äänimerkki

Lisätietoja

7

Alhainenpitoisuus

1 piippaus
Jos ongelma jatkuu, ota yhteyttä
30 sekunninlaitetoimittajaan.
välein enintään
5 minuuttia

8

Järjestelmän
toimintahäiriö,
virtausnopeus(OCDventtiilivika,
kaasureitin
tukkeutuminen)

Yksi piippaus
30 sekunnin
väleinJos ongelma jatkuu, ota yhteyttä
laitetoimittajaan.

9

Akku vähissä-varoitus

Yksi piippaus,
ei toistoa

Lataa akku.

10

Hengitystaajuus ylitetty

Yksi piippaus
60 sekunnin
väleinHengitystaajuus ylittää 40 bpm.

11

Järjestelmän
toimintahäiriö,
puhallinvika

Ei mitään

Jos ongelma jatkuu, ota yhteyttä
laitetoimittajaan.

12

Järjestelmän
toimintahäiriö,
alhainen paine

Ei mitään

Jos ongelma jatkuu, ota yhteyttä
laitetoimittajaan.

13

Järjestelmän
toimintahäiriö,
alhainenlämpötila

Ei mitään

Anna laitteen lämmetä huoneenlämpöiseksi.
Jos ongelma jatkuu, ota yhteyttä
laitetoimittajaan.

Info

Käynnistys

1 infopiippaus

Näkyy näytössä käynnistyksen yhteydessä,
kuulet yhden äänimerkin, joka vahvistaa, ettääänimerkki toimii.

Jos annos käynnistyy virheellisesti, pienennä
herkkyysasetusta. Jos ongelma jatkuu, otayhteyttä laitetoimittajaan.

Käynnistysnäyttö kertoo laiteversion oikeassa
alakulmassa.

330 - FINPN MN232-C4 K | Käyttöopas

FreeStyle Comfort
Prioriteetti

Hälytyksen
kuvausNäyttö

Äänimerkki

Info

Normaalikäyttönäyttö
autoSAT-tilassa(tasavirtaa ei
ole kytketty)

Ei mitään

Info

Normaalikäyttönäyttö
ja autoSAT
pois käytöstä(tasavirtaa ei
ole kytketty)

Ei mitään

Info

Asetustamuutettu

1 infopiippaus

Info

Kytke ulkoinen
virtalähde
laitteen ollessa
poissa päältä.

Ei mitään

Info

Kytke ulkoinen
virtalähde
laitteen ollessa
käynnissä.

Ei mitään

Info

Kytke ulkoinen
virtalähde irti
laitteen ollessa
poissa päältä.

1 infopiippaus

Info

Irrota ulkoinen
virtalähde
laitteen ollessa
käynnissä(akku
laitteessa).

1 infopiippaus

Lisätietoja

Näyttää CAIRE-näytön 2 sekunnin ajan ja sen
jälkeen akun ja kytkennän tilan 5 sekunnin
ajan, ennen kuin näyttö sammuu.

Näyttää CAIRE-näytön 2 sekunnin ajan, minkä
jälkeen näyttö sammuu.

PN MN232-C4 K | Käyttöopas

FIN - 331

FreeStyle Comfort
Prioriteetti

Hälytyksen
kuvausInfo

Näyttö

Äänimerkki

Lisätietoja

Valmiustilassa kun ylös- taialas-painiketta
painetaan,näytössä
näkyy akun ja
latauksen tila.

Ei mitään

Näyttö näkyy 5 sekuntia ja sammuu sitten.

Info

Lentokonetila
akkuvirralla

1 pitkäpiippausInfo

Lentokonetila,
kun kytkettynä

1 pitkäpiippaus

332 - FINPN MN232-C4 K | Käyttöopas

FreeStyle Comfort
Valikon selausohjeet
Voit avata valikon valmius- tai käyttötilassa. Jos
käyttötilassa ilmenee hälytystila, valikko sulkeutuu
automaattisesti. Jos mitään painiketta ei paineta
20 sekuntiin, valikko sulkeutuu automaattisesti.
Valmiustilassa valikkopainiketta voi olla tarpeen
painaa kaksi kertaa, jotta laite siirtyy käyttötilaan ja
voit avata valikon.
Valikkonäyttöjä on neljä:
1. tilanäyttö, jossa näkyvät laiteohjelmistonversionumero, käyttöaika ja akun purkausjaksot

Lentokonetila

2. lentokonetila (ON/OFF)
3. herkkyys: säädettävä (1–5)
4. autoSAT (ON/OFF)

Herkkyys

Esimerkki tilanäytöstä

ASETUSVALIKON KÄYTTÖ
Voit avata asetusvalikon painamalla valikko-/
valintapainiketta
valmiustilassa.

autoSAT, kun laite on käyttö- tai

Voit selata valikkoja painamalla painiketta.

Tilasivu

PN MN232-C4 K | Käyttöopas

FIN - 333

FreeStyle Comfort
TILASIVU
Freestyle Comfort -sivun tilasivulla näkyy liitetyn
akun laiteohjelmiston versio, aukioloajat ja parisyklien lukumäärä.

Jos haluat peruttaa lentokonetilan ottamisen käyttöön,
paina-painiketta Cancel (Peruuta) -kuvakkeen
ollessa näkyvissä.

Lentokonetila käytössä -asetus vahvistettu
Tilasivu

Käyttötunnit esitetään kokonaistunneina.
Akun purkautuminen on jaksojen lukumäärä, johon
liitetty akku on tyhjennetty ja ladattu.

LENTOKONETILA
Lentokonetila on pois käytöstä, kun FreeStyle Comfort
toimitetaan.

Kun haluat poistaa lentotilan käytöstä, toista edelläolevat vaiheet ja vaihda asetukseksi OFF.
Kun näytössä näkyy Next (Seuraava) -kuvake ja-painiketta valikkosivu siirtyy seuraavaan
painatnäyttövaihtoehtoon.
Lentotila on tarkoitettu käytettäväksi käyttötilassa.
Lentotila kytkeytyy automaattisesti pois käytöstä, joslaite jätetään valmiustilaan akkuvirralla.
Huomautus: lisätietoja lentotilasta on kohdassa
Lentotilan aktivointi.

HERKKYYSTILA

Lentokonetilan päänäyttö

Kytke tila päälle painamalla-painiketta Change(Muuta) -kuvakkeen ollessa näkyvissä. Arvoksi
muuttuu sininen ON (Käytössä) -sana.

Kun FreeStyle Comfort toimitetaan, sen oletusarvoisena
herkkyysasetuksena on 2. Tätä asetusta ei ole tarpeen
muuttaa normaaleissa olosuhteissa. Herkkyysasetukseksi
voidaan valita arvo välillä 1–5, jossa 1 on vähiten
herkkä ja 5 herkin.

Herkkyystilanäyttö, jossa asetuksena 2

Lentokonetila käytössä -asetus valittu

Vahvista lentokonetilan kytkeminen käyttöön
painamalla-painiketta Save (Tallenna) -kuvakkeen
ollessa näkyvissä. Arvoksi muuttuu musta ON(Käytössä) -sana.

334 - FINPN MN232-C4 K | Käyttöopas

Voit säätää herkkyystilaa painamalla-painiketta
Change (Muuta) -kuvakkeen ollessa näkyvissä.
Arvoksi muuttuu seuraava sininen numero.

FreeStyle Comfort
AUTOSAT-TILA
autoSAT-tila on käytössä, kun FreeStyle Comfort
toimitetaan.

Herkkyystilan asetus 3 valittu

Vahvista herkkyystilan arvoksi 3 painamalla-painiketta Save (Tallenna) -kuvakkeen ollessa
näkyvissä. Arvoksi muuttuu musta numero 3.
Voit peruuttaa herkkyystilan muutoksen painamalla-painiketta Cancel (Peruuta) -kuvakkeen ollessa
näkyvissä.

autoSAT-tilan päänäyttö

Voit kytkeä tilan pois käytöstä painamalla-painiketta Change (Muuta) -kuvakkeen ollessa
näkyvissä. Arvoksi muuttuu sininen OFF (Poiskäytöstä) -sana.

Herkkyystilan asetus 3 vahvistettu
autoSAT-tila pois käytöstä -asetus valittu

Vahvista autoSAT-tilan kytkeminen pois käytöstä
painamalla-painiketta Save (Tallenna) -kuvakkeen
ollessa näkyvissä. Arvoksi muuttuu musta OFF (Poiskäytöstä) -sana.
Voit peruuttaa autoSAT-tilan ottamisen käyttöön
painamalla-painiketta Cancel (Peruuta)-kuvakkeen ollessa näytössä.
Herkkyystilan asetus 5 (Max) vahvistettu

Kun näytössä näkyy Next (Seuraava) -kuvake ja
painat-painiketta valikkosivu siirtyy seuraavaan
näyttövaihtoehtoon.

autoSAT-tila pois käytöstä -asetus vahvistettu

Kun haluat kytkeä autoSAT-tilan käyttöön, toista edelläolevat vaiheet ja vaihda asetukseksi ON.
Kun näytössä näkyy Next (Seuraava) -kuvake ja
painat-painiketta valikkosivu siirtyy seuraavaan
näyttövaihtoehtoon.

PN MN232-C4 K | Käyttöopas

FIN - 335

FreeStyle Comfort

Vianmääritys
OngelmaTodennäköinen syy

Ratkaisu

FreeStyle Comfort näyttää
hälytyksen ja antaa katkonaistaäänimerkkiä.

Lue Hälytystilat-taulukko.

Lue Hälytystilat-taulukko.

Happirikastin ei käynnisty.

Akku on tyhjä.

Käynnistä laite AC- tai DC-liitännän kautta.

Ei kytketty sähkövirtaan.

Varmista ulkoisten liitäntöjen pitävyys.

Yleinen toimintahäiriö.

Suorita täydellinen kuittaus irrottamalla akku ja
mahdolliset ulkoiset AC- tai DC-virtalähteet. Asenna
akku sitten uudelleen. Kuittaus on valmis. Liitä laitetarvittaessa ulkoiseen virtalähteeseen.
Ota yhteys laitteen toimittajaan ja ota tarvittaessa
käyttöön vaihtoehtoinen happilähde.

Kaikki muut ongelmat

Suorita täydellinen kuittaus irrottamalla akku ja
mahdolliset ulkoiset AC- tai DC-virtalähteet. Asenna
akku sitten uudelleen. Kuittaus on valmis. Liitä laitetarvittaessa ulkoiseen virtalähteeseen.
Ota käyttöön vaihtoehtoinen happilähde ja ota
yhteys laitteen toimittajaan.

Huomautus: osa hälytyksistä aiheuttaa laitteen sammumisen.
Huomautus: sähkökatkoksen aikana hälytysjärjestelmä nollautuu (poistaa kaikki aiemmat hälytykset) ja näyttää vain
viimeisimmän hälytyksen, jos hälytys on edelleen voimassa, kun sähköä taas saa.

336 - FINPN MN232-C4 K | Käyttöopas

FreeStyle Comfort
Tietoja matkustamisesta
FreeStyle Comfort noudattaa Yhdysvaltainilmailuhallinto FAA:n ohjeita, jotka koskevat laitteen
käyttöä liikennelentokoneissa. Ennen matkustamista
esimerkiksi junassa, lentokoneessa tai laivassa:• Varmista, että happirikastin on hyvässä
käyttökunnossa.
• Ota yhteyttä liikennöitsijään vähintään 48 tuntia
ennen lähtöä ja ilmoita, että matkustat FreeStyle
Comfort mukanasi.
On suositeltavaa ottaa lisäakkuja mukaan matkalle. Jos
matkustat lentokoneella, varmista, että akussa riittää
virtaa esimerkiksi lähtöselvitystä ja turvatarkastusta
varten. Useimmat lentoyhtiöt edellyttävät, että sinulla
on mukana riittävä määrä ladattuja akkuja, jotta
akkuvirta riittää laitteen käyttämiseen vähintään
150 prosenttia odotetusta enimmäismatkustusajasta(lentoajat, maassaoloaika, aika ennen ja jälkeen
lennon, yhteyslennot ja odottamattomat viivästykset).
Huomautus: jos matkustat lentäen, on suositeltavaa, että
ylimääräiset akut ovat mukana turvallisessa kantolaukussa.

FAA:N MÄÄRÄYKSET

laite ole kytkettynä ulkoisen virtalähteeseen.

2. Ota lentokonetila käyttöön jollakin seuraavista
menetelmistä:
a. Ota lentotila käyttöön noudattamalla ohjeita,
jotka annetaan kohdassa Valikon selausohjeet.

b. Pidä ylös- ja alas-painikkeita painettuna
5 sekuntia, kunnes kuuluu pitkä piippaus ja
näyttöön tulee lentokonekuvake.

Vara-akut tulee suojata erikseen oikosulkujen
estämiseksi pitämällä ne alkuperäisessä
tuotepakkauksessa tai muutoin eristämällä navat,
esimerkiksi teippaamalla jokaisen akun navat
tai sijoittamalla ne erilliseen muovipussiin tai
suojapussiin, ja kuljettamalla ne käsimatkatavaroissa.

LENTOTILAN AKTIVOINTI
Akkuvirta
Kannettavaa happirikastinta (POC) voi käyttää
lentokoneessa normaalisti mutta VAIN akkuvirralla,
kun ulkoista virtalähdettä ei ole kytketty

3. Kun lentokonekuvake on näytössä, kytke laite
sähkövirtaan.

Ulkoinen virtalähde
Jos lentokoneessa on käytössä sähköpistorasia ja sen
sijainti soveltuu happirikastimen virtalähteelle, sitä
voidaan käyttää laitteen akkujen varajärjestelmänä.
Tällöin on ennen akkuvirrasta lentokoneen
sähkövirtaan siirtymistä toimittava seuraavasti:
1. Käynnistä happirikastin VAIN akkutilassa niin, ettei

Huomautus: lentokoneen rajoitusten vuoksi akkuja ei saa
ladata lentotilassa.

PN MN232-C4 K | Käyttöopas

FIN - 337

FreeStyle Comfort

Tarvikkeet

KANTOLAUKKUUN

Käytä yhdessä happirikastimen kanssa vain seuraavia
CAIREn toimittamia tarvikkeita. Muiden kuin
seuraavassa lueteltujen tarvikkeiden käyttäminen voi
heikentää happirikastimen tehoa ja/tai turvallisuutta.

1. Aseta laite laukun pohjalle. Kohdista suodattimet ja
akun aukot oikein.

FreeStyle Comfort -vakiovarusteet
Kantolaukku

MI396-1

Reppu

MI459-1Olkahihna

MI397-1

Vaihtovirtalähde (kaikki johdot mukana)

PW036-1S

Tasavirtajohto

CD041-1

Ulkoinen suurten hiukkasten suodatin(oikea)FI226-1

Ulkoinen suurten hiukkasten suodatin(vasen)FI227-1

2. Kiinnitä alaosan vetoketju.

Valinnaiset lisätarvikkeet
8-kennoinen lisäakkupakkaus

BT034-1S

16-kennoinen akkupakkaus

BT035-1S

Pöytälaturi

BT036-1

Kannettava tarvikelaukku

MI372-2

Huomautus: Muiden tarvikkeiden mukana voidaan toimittaa
tietyssä maassa käytettävä pistoke. Voit tilata muita
tarvikkeita ottamalla yhteyden CAIREen tai happirikastimen
toimittajaan.

Kantolaukun ohjeet
FreeStyle Comfort laitetaan kantolaukkuun ja laukku
suljetaan seuraavasti:

LAITTEEN ASETTAMINEN

338 - FINPN MN232-C4 K | Käyttöopas

OLKAHIHNAN KIINNITTÄMINEN
1. Olkahihna voidaan kiinnittää joko suoraan
laitteeseen tai laitteen kantolaukkuun.
2. Kiinnitä hinta laitteen tai laukun renkaaseen.

FreeStyle Comfort
Hävittäminen ja uudelleenkäyttö
Palauta aina FreeStyle Comfort kaikkine osineen
kotihoitoon hävitettäväksi. Voit myös ottaa yhteyttä
paikalliseen kaupungin tai kunnan viranomaiseen ja
kysyä ohjeita akun asianmukaisesta hävittämisestä.
Laitetta voidaan käyttää uudelleen ihmisten välillä,
mutta se on palautettava kotihoidon tarjoajalle
uudelleenkäsittelyä varten.

sähkö- ja elektroniikkalaiteromudirektiivin (WEEE)
mukaisesti.
Tuotteemme täyttävät vaarallisia aineita koskevan
RoHS-direktiivin vaatimukset. Tuotteissamme on vain
vähäisiä jäämiä lyijystä ja muista vaarallisista aineista.
TÄRKEÄ HUOMAUTUS: hävitä
akku laitetoimittajan tai paikallisten
viranomaisten kierrätysohjeiden
mukaisesti.

WEEE ja RoHS
Tämä symboli muistuttaa laitteen omistajaa
siitä, että elinkaarensa lopussa laite on kierrätettävä

Materiaalit, joiden kanssa käyttäjä joutuu suoraan kosketukseen
Happirikastimen kotelo................................................................................................................................ PC/ABS
Happirikastimen ohjauspaneeli...................................................................................EBG7-polyesteri tai vastaava
Happirikastimen kahva.................................................................................................................... Polykarbonaatti
Hapen syöttösuutin........................................................................................................................... Polykarbonaatti
Tyyppikilpi.......................................................................................................................................................Lexan
Johtoliitännät............................................................................................................. Polykarbonaatti/vinyylikloridi
Virtajohto..............................................................................................................................................PVC, metalli
Virtalähde..................................................................................................................... Lexan 940 (polykarbonaatti)
Akkupakkaus................................................................................................................................................ PC/ABS
Akun ja virtalähteen tarrat.................................................................................................................Polyesterikalvo
Happirikastimen kantolaukku................................................................................. Cordura, neopreeni, nylonvuori
Hihna.........................................................................................Polypropeeni, asetyyli, vinyyli, cordura, neopreeni

EMC-testaus
Ohjeistus ja valmistajan ilmoitus – sähkömagneettinen säteily
FreeStyle Comfort on tarkoitettu käytettäväksi alla määritetyssä sähkömagneettisessa ympäristössä. FreeStyle Comfort-asiakkaan tai -käyttäjän on varmistettava, että sitä käytetään sellaisessa ympäristössä.
Säteilytesti

Vastaavuus

Sähkömagneettinen ympäristö – ohjeistus

Ryhmä 1FreeStyle Comfort käyttää radiotaajuusenergiaa vain sisäisiin
toimintoihinsa. Tämän vuoksi radiotaajuussäteily on hyvin matalaa eikä
se todennäköisesti häiritse lähellä olevia elektronisia laitteita.

Luokka B
CISPR 11
Harmonisoidut päästöt
Luokka A
IEC 61000-3-2
Jännitteenvaihtelut/kohinapäästöt IEC 61000- Vastaa
3-3

FreeStyle Comfort soveltuu käytettäväksi kaikissa tiloissa,
kuten kotona ja tiloissa, jotka on liitetty suoraan julkiseen
pienjänniteverkkoon, joka syöttää sähköä kotitalouksille tarkoitettuihin
rakennuksiin.

Radiotaajuussäteily
CISPR 11
Radiotaajuussäteily

PN MN232-C4 K | Käyttöopas

FIN - 339

FreeStyle Comfort
Ohjeistus ja valmistajan ilmoitus – sähkömagneettinen häiriönsieto

FreeStyle Comfort on tarkoitettu käytettäväksi alla määritetyssä sähkömagneettisessa ympäristössä. FreeStyle Comfort-asiakkaan tai -käyttäjän on varmistettava, että sitä käytetään sellaisessa ympäristössä.
ImmuniteettiIEC 60601 -testaustaso Vastaavuustaso
Sähkömagneettinen ympäristö –testausohjeistus
Lattian tulee olla puuta, betonia tai
Sähkömagneettinen±8 kV kontakti±8 kV kontakti
keraamilaattaa. Jos lattia on peitetty
ympäristö – ohjeistus
synteettisellä materiaalilla, suhteellisen±15 kV ilman purkaus±15 kV ilman purkaus
IEC 61000-4-2
kosteuden on oltava vähintään 30 %.
±2 kV virtakaapelit
Transienttipurskeen ±2 kV virtakaapelit
Verkkovirran laadun tulee olla ammattisieto
ja kotihoidon hoitolaitosympäristöille
tarkoitettua.
IEC 61000-4-4±1 kV tulo-/lähtökaapelit Ei mitään±1 kV kaapelien välillä

Syöksyaalto
IEC 61000-4-5±2 kV yhteismuotoinen
jännitevaihtovirtajohdoissa±1 kV kaapelien välillä

Verkkovirran laadun tulee olla ammattija kotihoidon hoitolaitosympäristöille
tarkoitettua.

±1kV eromuotoinen
jännitevaihtovirtajohdoissa±2 kV yhteismuotoinen
jännite ulkoisilla
I/O-linjoilla
0 % UT 0,5 jakson ajan
0 % UT 0,5 jakson ajan
(0°, 45°, 90°, 135°, 180°, (0°, 45°, 90°, 135°, 180°,
225°, 270°, 315°)
225°, 270°, 315°)

Jännitteen vajoamat,lyhyet katkokset ja
jännitteenvaihteluttulosähkökaapeleissa, IEC 61000-4-11

0 % UT 1 jakson ajan (0 )070 % UT (30 %:n pudotus
UT:ssa) 25/30 jakson
ajan (00)
0 % UT 250/300 jakson
ajan (00)

Verkkovirran laadun tulee olla ammattija kotihoidon hoitolaitosympäristöille
0 % UT 1 jakson ajan (0 ) tarkoitettua. Jos FreeStyle Comfort-käyttäjä edellyttää jatkuvaa toimintaa
70 % UT (30 %:n pudotus virtakatkojenkin aikana, tällöin virta on
suositeltavaa syöttää FreeStyle Comfort
UT:ssa) 25/30 jakson-laitteeseen keskeytymättömästä
ajan (00)virtalähteestä (UPS) tai akusta.
00 % UT 250/300 jakson
ajan (00)

Verkkotaajuinen
Verkkotaajuisten magneettikenttien tason(50/60 Hz)
3 A/m
3 A/m
pitäisi vastata koti- tai sairaalarakennuksenmagneettikenttä IEC
tyypillistä tasoa.
61000-4-8
HUOMAA UT tarkoittaa AC (vaihtovirta) -verkkojännitettä ennen testaustason soveltamista.

HUOMAA Joissakin tapauksissa Freestyle Comfort voi nollautua täydellisen sähköstaattisen purkauksen seurauksena,jolloin normaali toiminta on palautettava täydellisellä kuittauksella. Katso lisätietoja kohdasta Vianmääritys.

340 - FINPN MN232-C4 K | Käyttöopas

FreeStyle Comfort
Ohjeistus ja valmistajan ilmoitus – sähkömagneettinen häiriönsieto

FreeStyle Comfort on tarkoitettu käytettäväksi alla määritetyssä sähkömagneettisessa ympäristössä. FreeStyle Comfort-asiakkaan tai -käyttäjän on varmistettava, että sitä käytetään sellaisessa ympäristössä.
Immuniteettites- IEC 60601 -testaustaso
Vastaavuustaso
Sähkömagneettinen ympäristö –tausohjeistus
10 Vrms
Jos kannettavia ja siirrettäviä
Johdettu
10 Vrms
radiotaajuus
radiotaajuuslaitteita käytetään FreeStyle
150 kHz – 80 MHz
Comfort -happirikastimen jonkin osan
IEC 61000-4-6
lähellä, mukaan lukien kaapelit, etäisyyden
on oltava vähintään kyseisen lähettimen
taajuuden perusteella laskettu suositeltu
10 V/m
10 V/metäisyys.
Säteilevä
80–2 700 MHz
radiotaajuus
IEC 61000-4-3

Suositeltu etäisyys

d = 1,2√ P
d = 1,2√ P = 80–800 MHz
d = 1,2√ P = 800–2 700 MHz
jossa P on lähettimen suurin lähtöteho
watteina (W) lähettimen valmistajan
ilmoituksen mukaisesti ja d on suositeltu
etäisyys metreinä (m).
Kiinteiden radiotaajuuslähettimien
kentänvoimakkuuksien (paikan
sähkömagneettisen mittauksen a mukaisesti)
on oltava alhaisempia kuin kunkintaajuusalueen säädöstenmukaisen tason. b
Häiriöitä voi esiintyä, jos lähellä on
seuraavalla symbolilla merkittyjä
laitteita:

HUOMAA 1 Korkea taajuusalue pätee tehoilla 80 MHz ja 800 MHz.
HUOMAA 2 Tämä ohjeistus ei ehkä päde kaikissa tilanteissa. Sähkömagneettiseen johtumiseen vaikuttavat pintojen,
esineiden ja ihmisten aiheuttama imeytyminen ja heijastuminen.
aKiinteiden lähettimien, kuten matka- ja langattomien puhelinten sekä satelliittiradioiden asemien, amatööriradioiden,
AM- ja FM-radiolähetysten ja televisiolähetysten kentänvoimakkuuksia ei voi teoreettisesti ennakoida tarkasti.
Kiinteän radiotaajuuslähettimen aiheuttaman sähkömagneettisen ympäristön arvioimiseksi on suoritettava paikan
sähkömagneettinen mittaus. Jos FreeStyle Comfort -happirikastimen käyttöympäristön kenttävoimakkuus ylittää yllä
esitetyn sovellettavan radiotaajuuden vaatimustenmukaisuustason, FreeStyle Comfort -happirikastinta tulee tarkkailla sennormaalin toiminnan varmistamiseksi. Jos toiminnassa havaitaan poikkeavuuksia, lisätoimenpiteet, kuten FreeStyle Comfort-laitteen uudelleensuuntaus tai -sijoitus, voivat olla tarpeen.
bTaajuusalueella 150 kHz – 80 MHz kentänvoimakkuuksien on oltava alle 10 V/m.

PN MN232-C4 K | Käyttöopas

FIN - 341

FreeStyle Comfort
Suositellut etäisyydet kannettavien ja siirrettävien radiotaajuusviestintälaitteiden sekä
FreeStyle Comfort -happirikastimen välillä

FreeStyle Comfort on tarkoitettu käytettäväksi sähkömagneettisessa ympäristössä, jossa radiotaajuuksien säteilyhäiriöitä
kontrolloidaan. FreeStyle Comfort -asiakas tai käyttäjä voi auttaa estämään sähkömagneettisia häiriöitä säilyttämällä
alla olevan suosituksen mukaisen vähimmäisvälimatkan kannettavan ja siirrettävän radiotaajuuslaitteen (lähettimet) ja
FreeStyle Comfort -happirikastimen välillä viestintälaitteen enimmäistehon mukaan.
Lähettimen suurin
Etäisyys lähettimen taajuuden mukaisesti
nimellislähtöteho
m
W

150 kHz – 80 MHz

80–800 MHz

800 MHz–2,5 GHz

d = 1,2√ P

d = 1,2√ P

d = 1,2√ P

0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Jos lähettimen suurinta lähtötehoa ei ole mainittu alla, suositeltu etäisyys d metreinä (m) voidaan arvioida yhtälöllä
lähettimen taajuuden perusteella, kun P on lähettimen suurin lähtöteho watteina (W) lähettimen valmistajan ilmoituksen
mukaisesti.
HUOMAA 1 Korkean taajuusalueen etäisyys pätee tehoilla 80 MHz ja 800 MHz.
HUOMAA 2 Tämä ohjeistus ei ehkä päde kaikissa tilanteissa. Sähkömagneettiseen johtumiseen vaikuttavat pintojen,
esineiden ja ihmisten aiheuttama imeytyminen ja heijastuminen.

Luokitus

Suojaus sähköiskuilta (tyyppi):
Luokka II	Laite on suojattu sähköiskuja vastaan
kaksoiseristämällä.

	• CISPR 11 / EN 55011 -standardin mukaan tämä
on luokkaan B ja ryhmään 1 kuuluva ISM-laite(Industrial, Scientific, and Medical).
	• FCC-osa 15, alaluku B – luokka B, tahattomia
päästöjä aiheuttavat laitteet

Suojausaste sähköiskua vastaan:
Tyyppi BF Laitteen suojaustaso sähköiskua vastaan
seuraavin osin:
		 1) sallittu vuotovirta		2) mahdollisen maadoituksen luotettavuus
		 Laitetta ei ole tarkoitettu käytettäväksi suoraan
sydämeen.

IP22 – Suojaus sormien tai muiden vastaavien
esineiden läpimitaltaan suurempien kuin 12,5 mm
sisäänpääsyltä. Pystysuoraan tippuva vesi ei ole
haitallista, vaikka kotelo on kallistuneena 15 astetta
normaalista asennostaan.

Riippumaton testaaminen lääkinnällisiä laitteitakoskevan standardin mukaisesti:
Täyttää testien mukaan• IEC 60601-1 versio 3.1: Sähkökäyttöiset
lääkintälaitteet - osa 1, Yleiset
turvallisuusvaatimukset.
• CAN/CSA C22.2 nro 60601-1-8 versio 2.1,
Sähkökäyttöiset lääkintälaitteet - osa 1: Yleiset
turvallisuusvaatimukset
Suojattu mahdollisia sähkömagneettisia ja muita
laitteen sekä muiden laitteiden välisiä häiriöitä vastaan.
	• Täyttää testien mukaan standardin EN 60601-1-2/
IEC 60601-1-2: Versio 4 vaatimukset.
	• Testattu RTCA/DO160 osan 21 luokan M
mukaisesti.

342 - FINPN MN232-C4 K | Käyttöopas

Sallittu puhdistus- ja infektioidenestomenetelmä:
Lisätietoja on FreeStyle-käyttöohjeen
puhdistamisesta, hoidosta ja kunnossapidosta
kertovassa osassa.
Turvallisuusaste herkästi syttyvien anestesiakaasujen
läheisyydessä:
Laite ei sovellu tällaiseen käyttöön.
Toimintatila: jatkuva käyttö

CAIRE ja CAIRE Inc. ovat CAIRE Inc: n rekisteröityjä tavaramerkkejä. Katso täydellinen
luettelo tavaramerkeistä alla olevalta verkkosivustoltamme.
Tavaramerkit: www.careinc.com/corporate/trademarks.
Tekijänoikeus © 2021 CAIRE Inc. pidättää oikeuden lakkauttaa tuotteitaan tai muuttaa
hintoja, materiaaleja, laitteita, laatua, nimityksiä, teknisiä tietoja ja/tai tuotteiden
valmistustapoja milloin tahansa ilman ennakkoilmoitusta ja ilman, että tästä aiheutuu sillelisävelvoitteita tai seuraamuksia. Kaikki oikeudet, joita tässä ei ilmaista suoraan, pidätetäänsoveltuvissa määrin.

FreeStyle ComfortΣτοιχεία ελέγχου χρήστη και ενδείξεις κατάστασης συστήματοςΕξουσιοδοτημένος αντιπρόσωπος στηνΕυρωπαϊκή ΚοινότηταΕάν η ετικέτα μοναδικού
αναγνωριστικού συσκευής (UDI)φέρει το σύμβολο CE ####, η συσκευή
συμμορφώνεται με τις απαιτήσεις τηςοδηγίας 93/42 / ΕΟΚ σχετικά με ταιατροτεχνολογικά προϊόντα. Το σύμβολοCE #### υποδεικνύει τον κοινοποιημένο
αριθμό σώματος.

ISO 7000· Γραφικά σύμβολα για χρήση στονεξοπλισμό — Ευρετήριο και σύνοψη
Διαβάστε το εγχειρίδιο χρήστη πριναπό τη λειτουργία. Reg. # 1641
Περιοχή θερμοκρασίας αποθήκευσης ήλειτουργίας. Reg. # 0632
Εύρος υγρασίας αποθήκευσης.
Reg. # 2620

IEC 60417
Εξοπλισμός κλάσης II, διπλά
μονωμένος Reg. # 5172

Περιορισμός ατμοσφαιρικής πίεσης.
Reg. # 2621

Διακόπτης λειτουργίας. Reg. # 5009

Όνομα και διεύθυνση τουκατασκευαστή. Reg. # 3082
Ημερομηνία κατασκευής. Reg. # 2497

Πιστοποίηση ETLΟργανισμός για την Ασφάλεια γιαCAN/CSA C22.2 Αρ. 601.1 M90 γιαιατρικό ηλεκτρικό εξοπλισμό

Αριθμός καταλόγου. Reg. # 2493
Σειριακός αριθμός. Reg. # 2498
Με αυτή την όψη προς τα πάνω.
Reg. # 0623

Εσωτερικά σύμβολαΜακριά από εύφλεκτα υλικά, λάδια
και γράσο.

Εύθραυστο, χειριστείτε με προσοχή.
Reg. # 0621

Μην αποσυναρμολογείτε.

Φυλάσσετε μακριά από τη βροχή,διατηρείτε στεγνό. Reg. # 0626

Αύξηση της ροής.
Όριο στοίβαξης σε αριθμό. Reg. # 2403
Μείωση της ροής.

ISO 7010: Γραφικά σύμβολα — Χρώματα ασφάλειας
και σήματα ασφάλειας — Καταχωρημένα σήματα
ασφάλειαςΑπαιτείται να διαβαστεί το εγχειρίδιο
οδηγιών. Reg. # M002
Μακριά από γυμνές φλόγες, φωτιά,σπίθες. Απαγορεύονται οι ανοιχτές
πηγές ανάφλεξης και το κάπνισμα.
Reg. # P003
Μην καπνίζετε κοντά στη μονάδα ήκατά τη διάρκεια της λειτουργίας τηςμονάδας. Reg. # P002
Εφαρμοζόμενο μέρος τύπουBF (βαθμός προστασίας απόηλεκτροπληξία). Reg. # 5333
Προειδοποίηση. Reg. # W001
Οδηγία του Συμβουλίου 93/42/ΕΟΚ για ταιατροτεχνολογικά προϊόντα

Κουμπί μενού.
RTCA/DO160 Section
21
Category M
CompliantΣυμμορφώνεται με την Κατηγορία Mτου προτύπου RTCA DO160
Ενότητα 21. Απαίτηση FAA SFAR 106
Εξοπλισμός Κατηγορίας ΙΙ, διπλής
μόνωσης
Εγκεκριμένος από την FAA – POC(φορητός συμπυκνωτής οξυγόνου)21 CFR 801.15: Κώδικας Ομοσπονδιακών
Κανονισμών (Code of Federal Regulations) των ΗΠΑ,Τίτλος 21Η ομοσπονδιακή νομοθεσία τωνΗΠΑ επιτρέπει την πώληση αυτής τηςσυσκευής μόνο από ιατρό ή κατόπιν
εντολής ιατρού.
PN MN232-C4 K | Εγχειρίδιο χρήστη

GRE - 343

FreeStyle ComfortΟδηγία του Συμβουλίου 2012/19/ΕΕ: οδηγία για τααπόβλητα ηλεκτρικού και ηλεκτρονικού εξοπλισμού
(ΑΗΗΕ)
ΑΗΗΕ

IEC 60601-1: Ιατρικός ηλεκτρικός εξοπλισμός
Μέρος 1 - Γενικές απαιτήσεις για βασική ασφάλεια καιουσιαστική απόδοση
Με προστασία από κάθετη πτώση
σταγόνωνΑυτό το προϊόν μπορεί να καλύπτεται από ένα ήπερισσότερα διπλώματα ευρεσιτεχνίας, των ΗΠΑ καιδιεθνή. Παρακαλούμε επισκεφθείτε τον δικτυακό τόποτης εταιρείας μας παρακάτω για τον κατάλογο τωνεφαρμοστέων διπλωμάτων ευρεσιτεχνίας. Διπλώματα
ευρεσιτεχνίας: www.caireinc.com/corporate/patents/.

Τι είναι ο συμπυκνωτής οξυγόνουΟ αέρας που αναπνέουμε περιέχει περίπου 21% οξυγόνο,78% άζωτο και 1% άλλα αέρια. Στο the FreeStyle®
Comfort®, ο αέρας του δωματίου αναρροφάται μέσαστο μηχάνημα μέσω των ανοιγμάτων εισαγωγής αέρα.
Στη συνέχεια, περνά μέσα από ένα απορροφητικό υλικόπου ονομάζεται «μοριακό κόσκινο». Αυτό το υλικό
ξεχωρίζει το οξυγόνο από το άζωτο και επιτρέπει μόνοστο οξυγόνο να περάσει. Το αποτέλεσμα είναι μια ροήοξυγόνου υψηλής συγκέντρωσης που παρέχεται στονχρήστη.
Το FreeStyle Comfort συνδυάζει προηγμένες τεχνολογίες
συμπύκνωσης οξυγόνου και διατήρησης οξυγόνου γιατη δημιουργία μιας φορητής συσκευής η οποία επιτρέπει
μέγιστη φορητότητα και κινητοποίηση. Η προηγμένη
χορήγηση παλμικής ροής ανιχνεύει γρήγορα πότεεισπνέει ο χρήστης και χορηγεί έναν παλμό οξυγόνου
στην αρχή της κάθε εισπνοής. Η χορήγηση παλμικής
ροής είναι ιδανική για δραστηριότητες και χρονικό
διάστημα μακριά από το σπίτι. Εξοικονομεί οξυγόνο καιπαρατείνει τη διάρκεια ζωής της μπαταρίας, σε αντίθεση
με τη συνεχή ροή. Το FreeStyle Comfort είναι αρκετά
ελαφρύ ώστε να μεταφέρεται μέσω ενός ιμάντα ώμου.
Μπορεί να λειτουργεί με εναλλασσόμενο ρεύμα (AC)(επιτοίχια παροχή), συνεχές ρεύμα (DC) (όχημα) ή ισχύαπό μπαταρία.

Προβλεπόμενη χρήση

Ο συμπυκνωτής οξυγόνου CAIRE FreeStyle Comfortπροορίζεται για τη χορήγηση συμπληρωματικού
οξυγόνου. Η συσκευή δεν προορίζεται για υποστήριξη
ζωής ούτε παρέχει δυνατότητες παρακολούθησης
ασθενούς.

Γιατί ο γιατρός σας συνταγογράφησε
οξυγόνοΠολλοί άνθρωποι πάσχουν από μια ποικιλία καρδιακών,πνευμονικών και άλλων αναπνευστικών παθήσεων.
Ένας σημαντικός αριθμός από αυτά τα άτομα μπορούν
να ωφεληθούν από συμπληρωματική οξυγονοθεραπείαστο σπίτι, όταν ταξιδεύουν ή ενώ συμμετέχουν σεκαθημερινές δραστηριότητες μακριά από το σπίτι.
Το οξυγόνο είναι ένα αέριο που αποτελεί το 21%του αέρα του περιβάλλοντος που αναπνέουμε και οοργανισμός μας εξαρτάται από μια σταθερή παροχή γιανα λειτουργεί κανονικά. Ο γιατρός σας συνταγογράφησε
344 - GREPN MN232-C4 K | Εγχειρίδιο χρήστη

μια συγκεκριμένη ροή ή ρύθμιση για την αντιμετώπισητης ιδιαίτερης αναπνευστικής σας κατάστασης.
Αν και το οξυγόνο είναι μια μη εθιστική θεραπεία,η αυθαίρετη οξυγονοθεραπεία μπορεί να είναι
επικίνδυνη. Πρέπει να ζητήσετε ιατρική συμβουλή πρινχρησιμοποιήσετε αυτόν τον συμπυκνωτή οξυγόνου.
Ο πάροχος εξοπλισμού που παρέχει τον εξοπλισμό
οξυγόνου σας θα σας δείξει πώς να ρυθμίσετε τονσυνταγογραφημένο ρυθμό ροής.

Προφίλ χειριστή
Οι συμπυκνωτές οξυγόνου προορίζονται για ναπαρέχουν συμπληρωματικό οξυγόνο σε χρήστες πουπάσχουν από δυσφορία λόγω παθήσεων που επηρεάζουν
την αποτελεσματικότητα των πνευμόνων τους ναμεταφέρουν οξυγόνο από τον αέρα στην κυκλοφορία
του αίματος. Οι φορητοί συμπυκνωτές οξυγόνου (POC)δεν αποθηκεύουν ή περιέχουν οξυγόνο. Δεν χρειάζονται
επαναπλήρωση και μπορούν να επαναφορτίζονται όπουυπάρχει διαθέσιμη ισχύς AC ή DC (εκτός εάν βρίσκονται
σε αεροσκάφος). Αυτό καθιστά τον χρήστη σχετικά
αυτάρκη από την άποψη της χρήσης στο σπίτι, τηςκινητοποίησης (τόσο εντός όσο και εκτός του σπιτιού),της κινητικότητας και του γενικού τρόπου ζωής. Η χρήση
συμπυκνωτή οξυγόνου απαιτεί συνταγή γιατρού και δενπροορίζεται για χρήση ζωτικής υποστήριξης.
Αν και η οξυγονοθεραπεία μπορεί να συνταγογραφηθεί
για χρήστες όλων των ηλικιών, ο τυπικός χρήστης
οξυγονοθεραπείας είναι ηλικίας άνω των 65 ετών καιπάσχει από μια ποικιλία αναπνευστικών παθήσεων,
συμπεριλαμβανομένης της χρόνιας αποφρακτικής
πνευμονοπάθειας (ΧΑΠ). Οι χρήστες τυπικά έχουν
καλές γνωστικές ικανότητες και πρέπει να μπορούν ναγνωστοποιούν τυχόν δυσφορία. Εάν ο χρήστης δεν είναι
σε θέση να γνωστοποιήσει τυχόν δυσφορία, ή δεν είναι
σε θέση να διαβάσει και να κατανοήσει την επισήμανσηκαι τις οδηγίες χρήσης του συμπυκνωτή, τότε η χρήση
συνιστάται μόνο υπό την επίβλεψη κάπου ατόμου πουμπορεί να το πράξει. Εάν γίνει αισθητή οποιαδήποτε
δυσφορία κατά τη διάρκεια της χρήσης του συμπυκνωτή,συνιστάται στους χρήστες να επικοινωνούν με τονπάροχο υγειονομικής περίθαλψής τους. Συνιστάται
επίσης στους χρήστες να έχουν διαθέσιμο εφεδρικό
οξυγόνο (δηλ. φιάλη οξυγόνου) σε περίπτωση διακοπής
ρεύματος ή βλάβης του συμπυκνωτή. Δεν απαιτούνται
άλλες μοναδικές δεξιότητες ή ικανότητες χρήσης για τηχρήση του συμπυκνωτή.

Αποσυσκευασία του FreeStyle ComfortΕπαληθεύστε ότι όλα ταπαρατιθέμενα εξαρτήματα
περιλαμβάνονται στησυσκευασία. Εάν λείπουν
οποιαδήποτε εξαρτήματα,επικοινωνήστε με τον πάροχο
οξυγόνου σας αμέσως.

Τυπικά παρελκόμενα• Φορητός συμπυκνωτής οξυγόνου
• Τσάντα μεταφοράς
• Ιμάντας ώμου• Πακέτο μπαταρίας 8 στοιχείων• Τροφοδοτικό AC• Καλώδιο τροφοδοσίας AC• Τροφοδοτικό DC• Εγχειρίδιο χρήστη

FreeStyle ComfortΠροαιρετικά παρελκόμενα
• Πρόσθετο πακέτο μπαταρίας
8 στοιχείων• Πακέτο μπαταρίας 16 στοιχείων• Επιτραπέζιος φορτιστής
• Τσάντα παρελκομένων «carry-all»Γνωριμία με το FreeStyle ComfortΕξοικειωθείτε με τα βασικά χαρακτηριστικά του FreeStyle Comfort και της διεπαφής χρήστη.

ΠρόσοψηΠροσάρτηση ιμάντα

Διεπαφή χρήστη

Έξοδος
οξυγόνουΈξοδος
αέραΕισαγωγή αέρα/φίλτραΕισαγωγή αέρα/φίλτραΓλωττίδα
υποδοχήςθύραςτροφοδοσίας
Πακέτο μπαταρίας
Κουμπί απελευθέρωσης
πακέτου μπαταρίας
PN MN232-C4 K | Εγχειρίδιο χρήστη

GRE - 345

FreeStyle ComfortΚάτοψη
Οθόνη ενδείξεων
Κουμπί
λειτουργίας

Αύξηση της ροήςΚουμπί μενού

Μείωση της ροήςΟθόνη ενδείξεων με μπαταρία
Ρύθμιση ροήςΈνδειξη μπαταρίας

Ένδειξη autoSATΕμφάνιση
συναγερμών

Ποσοστό
μπαταρίας πουαπομένειΈνδειξηχορήγησης
ενισχυμένης δόσης

Οθόνη ενδείξεων με τροφοδοσία AC/DCΡύθμιση ροήςΈνδειξη μπαταρίας

Πεδίο εμφάνισης
συναγερμών

346 - GREPN MN232-C4 K | Εγχειρίδιο χρήστη

Ένδειξη φόρτισης

FreeStyle ComfortΣημαντικό!
Οι οδηγίες ασφαλείας καθορίζονται ως εξής:ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΣΗΜΑΝΤΙΚΕΣΠΛΗΡΟΦΟΡΙΕΣ ΑΣΦΑΛΕΙΑΣ ΓΙΑΚΙΝΔΥΝΟΥΣ ΠΟΥ ΜΠΟΡΕΙ ΝΑΠΡΟΚΑΛΕΣΟΥΝ ΣΟΒΑΡΟ ΤΡΑΥΜΑΤΙΣΜΟ.
ΠΡΟΣΟΧΗ: Σημαντικές πληροφορίες
για την πρόληψη ζημιάς στο FreeStyle
Comfort.

Σημείωση: Πληροφορίες που χρήζουν ιδιαίτερης
προσοχής.

Ενδείξεις χρήσης
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΊΝΑΙ ΠΟΛΎΣΗΜΑΝΤΙΚΌ ΝΑ ΕΠΙΛΈΓΕΤΕ ΜΌΝΟΤΟ ΣΥΝΤΑΓΟΓΡΑΦΗΜΈΝΟ ΕΠΊΠΕΔΟ
ΟΞΥΓΌΝΟΥ. ΜΗΝ ΑΛΛΆΖΕΤΕ ΤΗΝΕΠΙΛΟΓΉ ΡΟΉΣ ΕΚΤΌΣ ΕΆΝ ΣΑΣ ΤΟΥΠΈΔΕΙΞΕ ΑΔΕΙΟΎΧΟΣ ΚΛΙΝΙΚΌΣ
ΓΙΑΤΡΌΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΟΜΟΣΠΟΝΔΙΑΚΉ ΝΟΜΟΘΕΣΊΑ(ΗΠΑ) ΠΕΡΙΟΡΊΖΕΙ ΤΗΝ ΠΏΛΗΣΗ Ή ΜΊΣΘΩΣΗ ΑΥΤΉΣ
ΤΗΣ ΣΥΣΚΕΥΉΣ ΑΠΌ Ή ΚΑΤ' ΕΝΤΟΛΉ ΓΙΑΤΡΟΎ ΉΆΛΛΟΥ ΑΔΕΙΟΎΧΟΥ ΠΑΡΌΧΟΥ ΥΓΕΙΟΝΟΜΙΚΉΣ
ΠΕΡΊΘΑΛΨΗΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΥΤΉ Η ΜΟΝΆΔΑ ΔΕΝΠΡΟΟΡΊΖΕΤΑΙ ΝΑ ΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ ΓΙΑ ΖΩΤΙΚΉ
ΥΠΟΣΤΉΡΙΞΗ. ΗΛΙΚΙΩΜΈΝΟΙ, ΠΑΙΔΙΑΤΡΙΚΟΊ ΉΆΛΛΟΙ ΧΡΗΣΤΕΣ ΑΝΊΚΑΝΟΙ ΝΑ ΓΝΩΣΤΟΠΟΙΉΣΟΥΝ
ΤΥΧΌΝ ΔΥΣΦΟΡΊΑ ΚΑΤΆ ΤΗ ΧΡΉΣΗ ΑΥΤΉΣ ΤΗΣΣΥΣΚΕΥΉΣ ΜΠΟΡΕΊ ΝΑ ΑΠΑΙΤΟΎΝ ΠΡΌΣΘΕΤΗΠΑΡΑΚΟΛΟΎΘΗΣΗ. ΧΡΗΣΤΕΣ ΜΕ ΜΕΙΩΜΈΝΗ ΑΚΟΉΉ/ΚΑΙ ΌΡΑΣΗ ΜΠΟΡΕΊ ΝΑ ΑΠΑΙΤΟΎΝ ΒΟΉΘΕΙΑ ΜΕΤΗΝ ΠΑΡΑΚΟΛΟΎΘΗΣΗ ΤΩΝ ΣΥΝΑΓΕΡΜΏΝ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΟΙ ΡΥΘΜΊΣΕΙΣ ΛΕΙΤΟΥΡΓΊΑΣ
ΠΑΛΜΙΚΉΣ ΔΌΣΗΣ ΠΡΈΠΕΙ ΝΑ ΚΑΘΟΡΊΖΟΝΤΑΙ ΓΙΑΚΆΘΕ ΧΡΗΣΤΗ ΕΞΑΤΟΜΙΚΕΥΜΈΝΑ ΓΙΑ ΤΙΣ ΑΝΆΓΚΕΣ
ΤΟΥ ΣΕ ΑΝΆΠΑΥΣΗ, ΚΑΤΆ ΤΗ ΔΙΆΡΚΕΙΑ ΆΣΚΗΣΗΣΚΑΙ ΌΤΑΝ ΤΑΞΙΔΕΎΕΙ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΟΡΙΣΜΕΝΕΣ ΑΝΑΠΝΕΥΣΤΙΚΕΣΠΡΟΣΠΑΘΕΙΕΣ ΤΟΥ ΑΣΘΕΝΟΥΣ ΜΠΟΡΕΙ ΝΑ ΜΗΝΕΝΕΡΓΟΠΟΙΗΣΟΥΝ ΤΗ ΛΕΙΤΟΥΡΓΙΑ ΔΙΑΤΗΡΗΣΗΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΟ FREESTYLE COMFORT ΔΕΝΠΡΟΟΡΙΖΕΤΑΙ ΓΙΑ ΧΡΗΣΗ ΜΕ ΤΡΑΧΕΙΟΤΟΜΗΜΕΝΟΑΣΘΕΝΗ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΓΚΥΕΣ Ή ΘΗΛΑΖΟΥΣΕΣΓΥΝΑΙΚΕΣ ΔΕΝ ΠΡΕΠΕΙ ΝΑ ΧΡΗΣΙΜΟΠΟΙΟΥΝΤΑ ΠΑΡΕΛΚΟΜΕΝΑ ΠΟΥ ΣΥΝΙΣΤΩΝΤΑΙ ΣΕ ΑΥΤΟΤΟ ΕΓΧΕΙΡΙΔΙΟ, ΚΑΘΩΣ ΜΠΟΡΕΙ ΝΑ ΠΕΡΙΕΧΟΥΝ
ΦΘΑΛΙΚΕΣ ΕΝΩΣΕΙΣ.

Αντενδείξεις χρήσης
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΣΕ ΟΡΙΣΜΕΝΕΣΠΕΡΙΠΤΩΣΕΙΣ, Η ΧΡΗΣΗ ΜΗΣΥΝΤΑΓΟΓΡΑΦΗΜΕΝΟΥ ΟΞΥΓΟΝΟΥΜΠΟΡΕΙ ΝΑ ΕΙΝΑΙ ΕΠΙΚΙΝΔΥΝΗ. ΑΥΤΗ ΗΣΥΣΚΕΥΗ ΠΡΕΠΕΙ ΝΑ ΧΡΗΣΙΜΟΠΟΙΕΙΤΑΙΜΟΝΟ ΟΤΑΝ ΕΧΕΙ ΣΥΝΤΑΓΟΓΡΑΦΗΘΕΙΑΠΟ ΓΙΑΤΡΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΔΕΝ ΠΡΟΟΡΙΖΕΤΑΙ ΓΙΑ ΧΡΗΣΗ
ΥΠΟ ΤΗΝ ΠΑΡΟΥΣΙΑ ΕΥΦΛΕΚΤΩΝ ΑΝΑΙΣΘΗΤΙΚΩΝ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΌΠΩΣ ΚΑΙ ΜΕ ΚΆΘΕ ΗΛΕΚΤΡΙΚΆ
ΤΡΟΦΟΔΟΤΟΎΜΕΝΗ ΣΥΣΚΕΥΉ, Ο ΧΡΉΣΤΗΣ ΜΠΟΡΕΊΝΑ ΑΝΤΙΜΕΤΩΠΊΣΕΙ ΠΕΡΙΌΔΟΥΣ ΕΚΤΌΣ ΛΕΙΤΟΥΡΓΊΑΣΩΣ ΑΠΟΤΈΛΕΣΜΑ ΔΙΑΚΟΠΉΣ ΗΛΕΚΤΡΙΚΟΎ
ΡΕΎΜΑΤΟΣ, Ή ΤΗΣ ΑΝΆΓΚΗΣ ΝΑ ΑΝΑΘΈΣΕΙΤΟ ΣΈΡΒΙΣ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΣΕΕΞΕΙΔΙΚΕΥΜΈΝΟ ΤΕΧΝΙΚΌ. Ο ΣΥΜΠΥΚΝΩΤΉΣ
ΟΞΥΓΌΝΟΥ ΔΕΝ ΕΊΝΑΙ ΚΑΤΆΛΛΗΛΟΣ ΓΙΑΟΠΟΙΟΝΔΉΠΟΤΕ ΧΡΗΣΤΗ Ο ΟΠΟΊΟΣ ΘΑ ΜΠΟΡΟΎΣΕΝΑ ΥΠΟΣΤΕΊ ΑΝΕΠΙΘΎΜΗΤΕΣ ΕΠΙΠΤΏΣΕΙΣ ΣΤΗΝΥΓΕΊΑ ΩΣ ΑΠΟΤΈΛΕΣΜΑ ΤΈΤΟΙΑΣ ΠΡΟΣΩΡΙΝΉΣ
ΔΙΑΚΟΠΉΣ.

Οδηγίες ασφαλείας
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΔΕΝ ΕΠΙΤΡΈΠΕΤΑΙ
ΚΑΜΊΑ ΤΡΟΠΟΠΟΊΗΣΗ ΑΥΤΟΎ ΤΟΥΕΞΟΠΛΙΣΜΟΎ
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΥΤΉ Η ΣΥΣΚΕΥΉ ΠΑΡΈΧΕΙ
ΟΞΥΓΌΝΟ ΥΨΗΛΉΣ ΣΥΓΚΈΝΤΡΩΣΗΣ, ΤΟ ΟΠΟΊΟ
ΠΡΟΆΓΕΙ ΤΗΝ ΤΑΧΕΊΑ ΚΑΎΣΗ. ΜΗΝ ΕΠΙΤΡΈΠΕΤΕΤΟ ΚΆΠΝΙΣΜΑ Ή ΤΙΣ ΑΝΟΙΧΤΈΣ ΦΛΌΓΕΣ ΣΤΟΝΊΔΙΟ ΧΏΡΟ ΜΕ (1) ΑΥΤΉΝ ΤΗ ΣΥΣΚΕΥΉ, Ή (2)ΟΠΟΙΟΔΉΠΟΤΕ ΠΑΡΕΛΚΌΜΕΝΟ ΠΟΥ ΦΈΡΕΙ
ΟΞΥΓΌΝΟ. ΑΠΟΤΥΧΊΑ ΤΉΡΗΣΗΣ ΑΥΤΉΣ ΤΗΣΠΡΟΕΙΔΟΠΟΊΗΣΗΣ ΜΠΟΡΕΊ ΝΑ ΟΔΗΓΉΣΕΙ ΣΕΠΥΡΚΑΓΙΆ, ΚΑΤΑΣΤΡΟΦΉ ΙΔΙΟΚΤΗΣΊΑΣ Ή/ΚΑΙ ΝΑΠΡΟΚΑΛΈΣΕΙ ΦΥΣΙΚΌ ΤΡΑΥΜΑΤΙΣΜΌ Ή ΘΆΝΑΤΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΛΕΙΤΟΥΡΓΕΊΤΕ ΤΗ ΜΟΝΆΔΑΣΕ ΠΕΡΙΟΡΙΣΜΈΝΟ Ή ΣΤΕΝΌ ΧΏΡΟ (ΔΗΛ. ΜΙΚΡΉ
ΤΣΆΝΤΑ Ή ΤΣΆΝΤΑ ΧΕΙΡΌΣ) ΌΠΟΥ Ο ΑΕΡΙΣΜΌΣ
ΜΠΟΡΕΊ ΝΑ ΕΊΝΑΙ ΠΕΡΙΟΡΙΣΜΈΝΟΣ. ΚΆΤΙ ΤΈΤΟΙΟ
ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΕΙ ΥΠΕΡΘΈΡΜΑΝΣΗ ΤΟΥΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΚΑΙ ΝΑ ΕΠΗΡΕΆΣΕΙ
ΑΡΝΗΤΙΚΆ ΤΗΝ ΑΠΌΔΟΣΗ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΣΥΜΠΥΚΝΩΤΉΣ ΠΡΈΠΕΙ ΝΑΤΟΠΟΘΕΤΕΊΤΑΙ ΜΕ ΤΈΤΟΙΟΝ ΤΡΌΠΟ ΏΣΤΕ ΝΑΑΠΟΦΕΎΓΕΤΑΙ Ο ΚΑΠΝΌΣ, ΟΙ ΡΥΠΑΝΤΈΣ Ή ΟΙΑΝΑΘΥΜΙΆΣΕΙΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΣΕ ΠΕΡΊΠΤΩΣΗ ΠΟΥ ΣΥΜΒΕΊ
ΚΆΠΟΙΟ ΣΟΒΑΡΌ ΣΥΜΒΆΝ ΜΕ ΑΥΤΉΝ ΤΗ ΣΥΣΚΕΥΉ,Ο ΧΡΉΣΤΗΣ ΘΑ ΠΡΈΠΕΙ ΝΑ ΑΝΑΦΈΡΕΙ ΑΜΈΣΩΣΤΟ ΣΥΜΒΆΝ ΣΤΟΝ ΠΑΡΟΧΈΑ Ή / ΚΑΙ ΣΤΟΝΚΑΤΑΣΚΕΥΑΣΤΉ. ΈΝΑ ΣΟΒΑΡΌ ΣΥΜΒΆΝ ΟΡΊΖΕΤΑΙΩΣ ΤΡΑΥΜΑΤΙΣΜΌΣ, ΘΆΝΑΤΟΣ Ή ΠΙΘΑΝΌΤΗΤΑ
ΠΡΌΚΛΗΣΗΣ ΤΡΑΥΜΑΤΙΣΜΟΎ / ΘΑΝΆΤΟΥ ΣΕΠΕΡΊΠΤΩΣΗ ΕΠΑΝΆΛΗΨΗΣ ΤΟΥ ΣΥΜΒΆΝΤΟΣ.
Ο ΧΡΉΣΤΗΣ ΜΠΟΡΕΊ ΕΠΊΣΗΣ ΝΑ ΑΝΑΦΈΡΕΙ ΤΟΠΕΡΙΣΤΑΤΙΚΌ ΣΤΗΝ ΑΡΜΌΔΙΑ ΑΡΧΉ ΤΗΣ ΧΏΡΑΣ
ΌΠΟΥ ΣΥΝΈΒΗ ΤΟ ΠΕΡΙΣΤΑΤΙΚΌ.
PN MN232-C4 K | Εγχειρίδιο χρήστη

GRE - 347

FreeStyle ComfortΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΧΡΉΣΗ ΟΡΙΣΜΈΝΩΝ
ΠΑΡΕΛΚΟΜΈΝΩΝ ΧΟΡΉΓΗΣΗΣ
ΟΞΥΓΌΝΟΥ ΠΟΥ ΔΕΝ ΚΑΘΟΡΊΖΟΝΤΑΙ ΓΙΑΑΥΤΌΝ ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ
ΜΠΟΡΕΊ ΝΑ ΕΠΗΡΕΆΣΕΙ ΑΡΝΗΤΙΚΆ
ΤΗΝ ΑΠΌΔΟΣΗ. ΠΑΡΑΠΟΜΠΈΣ ΣΤΑΣΥΝΙΣΤΏΜΕΝΑ ΠΑΡΕΛΚΌΜΕΝΑ
ΠΑΡΈΧΟΝΤΑΙ ΣΕ ΑΥΤΌ ΤΟ ΕΓΧΕΙΡΊΔΙΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΆΝ Ο ΣΥΜΠΥΚΝΩΤΉΣ
ΟΞΥΓΌΝΟΥ ΈΧΕΙ ΠΈΣΕΙ, ΥΠΟΣΤΕΊ ΖΗΜΙΆ Ή ΕΚΤΕΘΕΊΣΕ ΝΕΡΌ, ΠΑΡΑΚΑΛΟΎΜΕ ΕΠΙΚΟΙΝΩΝΉΣΤΕΜΕ ΤΟΝ ΠΡΟΜΗΘΕΥΤΉ ΕΞΟΠΛΙΣΜΟΎ ΣΑΣ ΓΙΑΈΛΕΓΧΟ Ή ΠΙΘΑΝΉ ΕΠΙΣΚΕΥΉ ΤΗΣ ΣΥΣΚΕΥΉΣ.
ΜΗ ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥΕΆΝ ΈΧΕΙ ΥΠΟΣΤΕΊ ΖΗΜΙΆ ΤΟ ΚΑΛΏΔΙΟ Ή ΤΟ ΦΙΣΡΕΎΜΑΤΟΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΣΦΑΛΊΣΤΕ ΚΑΤΆΛΛΗΛΑ,ΠΡΟΣΔΈΣΤΕ ΜΕ ΙΜΆΝΤΑ Ή ΣΤΕΡΕΏΣΤΕ ΜΕ ΆΛΛΟΝ
ΤΡΌΠΟ ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΌΤΑΝΒΡΊΣΚΕΣΤΕ ΣΕ ΈΝΑ ΌΧΗΜΑ ΚΑΤΆ ΤΗ ΔΙΆΡΚΕΙΑ
ΜΕΤΑΦΟΡΆΣ ΓΙΑ ΝΑ ΑΠΟΤΡΈΨΕΤΕ ΖΗΜΙΆ ΉΤΡΑΥΜΑΤΙΣΜΌ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΟΙ ΚΑΝΟΝΙΣΜΟΊ ΤΟΥ ΥΠΟΥΡΓΕΊΟΥ
ΜΕΤΑΦΟΡΏΝ ΤΩΝ ΗΠΑ (DOT) ΚΑΙ ΤΩΝ ΗΝΩΜΈΝΩΝ
ΕΘΝΏΝ (UN) ΑΠΑΙΤΟΎΝ ΤΗΝ ΑΦΑΊΡΕΣΗ ΤΗΣΜΠΑΤΑΡΊΑΣ ΑΠΌ ΤΗ ΣΥΣΚΕΥΉ ΓΙΑ ΚΆΘΕ ΔΙΕΘΝΈΣ
ΑΕΡΟΠΟΡΙΚΌ ΤΑΞΊΔΙ ΌΤΑΝ Ο ΣΥΜΠΥΚΝΩΤΉΣ
ΟΞΥΓΌΝΟΥ ΕΛΈΓΧΕΤΑΙ ΩΣ ΑΠΟΣΚΕΥΉ. ΚΑΤΆ ΤΗΝΑΠΟΣΤΟΛΉ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ,Η ΜΠΑΤΑΡΊΑ ΠΡΈΠΕΙ ΕΠΊΣΗΣ ΝΑ ΑΦΑΙΡΕΊΤΑΙ ΑΠΌ ΤΗΣΥΣΚΕΥΉ ΚΑΙ ΝΑ ΣΥΣΚΕΥΆΖΕΤΑΙ ΚΑΤΆΛΛΗΛΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΛΕΙΤΟΥΡΓΕΙΤΕ ΑΥΤΟΝ ΤΟΝΕΞΟΠΛΙΣΜΟ ΧΩΡΙΣ ΝΑ ΕΧΕΤΕ ΠΡΩΤΑ ΔΙΑΒΑΣΕΙΚΑΙ ΚΑΤΑΝΟΗΣΕΙ ΤΟ ΠΑΡΟΝ ΕΓΧΕΙΡΙΔΙΟ. ΕΑΝ ΔΕΝΜΠΟΡΕΙΤΕ ΝΑ ΚΑΤΑΝΟΗΣΕΤΕ ΤΙΣ ΠΡΟΕΙΔΟΠΟΙΗΣΕΙΣΚΑΙ ΤΙΣ ΟΔΗΓΙΕΣ, ΕΠΙΚΟΙΝΩΝΗΣΤΕ ΜΕ ΤΟΝ ΠΑΡΟΧΟ
ΤΟΥ ΕΞΟΠΛΙΣΜΟΥ ΣΑΣ ΠΡΟΤΟΥ ΕΠΙΧΕΙΡΗΣΕΤΕ ΝΑΧΡΗΣΙΜΟΠΟΙΗΣΕΤΕ ΑΥΤΟΝ ΤΟΝ ΕΞΟΠΛΙΣΜΟ· ΣΕΑΝΤΙΘΕΤΗ ΠΕΡΙΠΤΩΣΗ, ΜΠΟΡΕΙ ΝΑ ΠΡΟΚΛΗΘΕΙΤΡΑΥΜΑΤΙΣΜΟΣ Ή ΥΛΙΚΗ ΖΗΜΙΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΆΝ ΑΙΣΘΆΝΕΣΤΕ ΔΥΣΦΟΡΊΑΉ ΠΑΡΟΥΣΙΆΖΕΤΕ ΕΠΕΊΓΟΥΣΑ ΙΑΤΡΙΚΉ ΚΑΤΆΣΤΑΣΗ,ΑΝΑΖΗΤΉΣΤΕ ΙΑΤΡΙΚΉ ΒΟΉΘΕΙΑ ΑΜΈΣΩΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΛΕΙΤΟΥΡΓΊΑ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ
ΟΞΥΓΌΝΟΥ ΕΚΤΌΣ ΤΩΝ ΠΡΟΔΙΑΓΡΑΦΏΝ
ΘΕΡΜΟΚΡΑΣΊΑΣ ΛΕΙΤΟΥΡΓΊΑΣ ΜΠΟΡΕΊ ΝΑΠΕΡΙΟΡΊΣΕΙ ΤΗ ΔΥΝΑΤΌΤΗΤΑ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ ΝΑΙΚΑΝΟΠΟΙΕΊ ΤΗΝ ΠΡΟΔΙΑΓΡΑΦΉ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ
ΟΞΥΓΌΝΟΥ. ΑΝΑΤΡΈΞΤΕ ΣΤΗΝ ΕΝΌΤΗΤΑ
ΠΡΟΔΙΑΓΡΑΦΏΝ ΤΟΥ ΠΑΡΌΝΤΟΣ ΕΓΧΕΙΡΙΔΊΟΥ ΓΙΑΤΑ ΌΡΙΑ ΘΕΡΜΟΚΡΑΣΊΑΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΛΆΔΙ,ΓΡΆΣΟ Ή ΠΡΟΪΌΝΤΑ ΜΕ ΒΆΣΗ ΠΕΤΡΈΛΑΙΟ Ή ΆΛΛΑΕΎΦΛΕΚΤΑ ΠΡΟΪΌΝΤΑ ΜΕ ΤΑ ΠΑΡΕΛΚΌΜΕΝΑ
ΠΟΥ ΦΈΡΟΥΝ ΟΞΥΓΌΝΟ Ή ΜΕ ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ
ΟΞΥΓΌΝΟΥ. ΠΡΈΠΕΙ ΝΑ ΧΡΗΣΙΜΟΠΟΙΟΎΝΤΑΙ ΜΌΝΟΛΟΣΙΌΝ Ή ΑΛΟΙΦΈΣ ΜΕ ΒΆΣΗ ΤΟ ΝΕΡΌ Ή ΣΥΜΒΑΤΈΣΜΕ ΤΟ ΟΞΥΓΌΝΟ. ΤΟ ΟΞΥΓΌΝΟ ΕΠΙΤΑΧΎΝΕΙ ΤΗΝΚΑΎΣΗ ΤΩΝ ΕΎΦΛΕΚΤΩΝ ΟΥΣΙΏΝ.
348 - GREPN MN232-C4 K | Εγχειρίδιο χρήστη

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΑΦΉΝΕΤΕ ΝΑ ΦΡΆΞΟΥΝΤΑ ΑΝΟΊΓΜΑΤΑ ΕΙΣΑΓΩΓΉΣ Ή ΕΞΌΔΟΥ ΑΈΡΑ.
ΜΗΝ ΑΦΉΝΕΤΕ ΝΑ ΠΈΣΟΥΝ ΚΑΙ ΜΗΝ ΕΙΣΆΓΕΤΕ
ΑΝΤΙΚΕΊΜΕΝΑ ΣΕ ΚΑΝΈΝΑ ΆΝΟΙΓΜΑ ΤΗΣΣΥΣΚΕΥΉΣ. ΚΆΤΙ ΤΈΤΟΙΟ ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΕΙ
ΥΠΕΡΘΈΡΜΑΝΣΗ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΚΑΙΝΑ ΕΠΗΡΕΆΣΕΙ ΑΡΝΗΤΙΚΆ ΤΗΝ ΑΠΌΔΟΣΗ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΛΑΝΘΑΣΜΈΝΗ ΧΡΉΣΗ ΤΗΣΜΠΑΤΑΡΊΑΣ ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΕΙ ΘΈΡΜΑΝΣΗΉ ΑΝΆΦΛΕΞΗ ΤΗΣ ΜΠΑΤΑΡΊΑΣ ΚΑΙ ΜΠΟΡΕΊΝΑ ΠΡΟΚΑΛΈΣΕΙ ΣΟΒΑΡΌ ΤΡΑΥΜΑΤΙΣΜΌ. ΔΕΝΕΠΙΤΡΈΠΕΤΑΙ ΝΑ ΤΡΥΠΉΣΕΤΕ, ΝΑ ΧΤΥΠΉΣΕΤΕ,ΝΑ ΠΑΤΉΣΕΤΕ Ή ΝΑ ΡΊΞΕΤΕ ΤΗΝ ΜΠΑΤΑΡΊΑ,Ή ΝΑ ΥΠΟΒΆΛΕΤΕ ΑΛΛΙΏΣ ΤΗΝ ΜΠΑΤΑΡΊΑ ΣΕΙΣΧΥΡΈΣ ΚΡΟΎΣΕΙΣ Ή ΚΡΑΔΑΣΜΟΎΣ. Η ΧΡΉΣΗ
ΜΙΑΣ ΚΑΤΕΣΤΡΑΜΜΈΝΗΣ ΜΠΑΤΑΡΊΑΣ ΜΠΟΡΕΊ ΝΑΠΡΟΚΑΛΈΣΕΙ ΠΡΟΣΩΠΙΚΌ ΤΡΑΥΜΑΤΙΣΜΌ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΒΡΑΧΥΚΥΚΛΏΝΕΤΕ ΤΙΣΜΕΤΑΛΛΙΚΈΣ ΕΠΑΦΈΣ ΤΗΣ ΜΠΑΤΑΡΊΑΣ ΜΕΜΕΤΑΛΛΙΚΆ ΑΝΤΙΚΕΊΜΕΝΑ, ΌΠΩΣ ΚΛΕΙΔΙΆ ΉΚΈΡΜΑΤΑ. ΚΆΤΙ ΤΈΤΟΙΟ ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΕΙ
ΣΠΙΝΘΉΡΕΣ Ή ΤΗΝ ΠΑΡΑΓΩΓΉ ΥΠΕΡΒΟΛΙΚΉΣ
ΘΕΡΜΌΤΗΤΑΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΚΑΤΆ ΤΗ ΧΡΉΣΗ ΤΟΥ ΦΟΡΗΤΟΎ
ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΣΕ ΕΞΩΤΕΡΙΚΌ ΧΏΡΟ ΜΕΤΟ ΤΡΟΦΟΔΟΤΙΚΌ AC, ΣΥΝΔΈΣΤΕ ΤΟ ΤΡΟΦΟΔΟΤΙΚΌ
ΑΠΟΚΛΕΙΣΤΙΚΆ ΣΕ ΜΙΑ ΠΡΊΖΑ ΜΕ ΔΙΑΚΌΠΤΗ ΒΛΆΒΗΣ
ΓΕΊΩΣΗΣ (GFI).
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΣΥΜΠΥΚΝΩΤΉΣ ΟΞΥΓΌΝΟΥ
ΔΕΝ ΠΡΈΠΕΙ ΝΑ ΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ ΠΑΡΑΚΕΊΜΕΝΑΣΕ Ή ΣΕ ΔΙΆΤΑΞΗ ΣΤΟΊΒΑΣ ΜΕ ΆΛΛΟ ΕΞΟΠΛΙΣΜΌ.
ΕΆΝ Η ΧΡΉΣΗ ΠΑΡΑΚΕΊΜΕΝΑ Ή ΣΕ ΔΙΆΤΑΞΗ
ΣΤΟΊΒΑΣ ΕΊΝΑΙ ΑΝΑΠΌΦΕΥΚΤΗ, Η ΣΥΣΚΕΥΉ ΠΡΈΠΕΙΝΑ ΠΑΡΑΚΟΛΟΥΘΕΊΤΑΙ ΓΙΑ ΝΑ ΕΠΑΛΗΘΕΎΕΤΑΙ ΗΚΑΝΟΝΙΚΉ ΛΕΙΤΟΥΡΓΊΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΧΡΉΣΗ ΚΑΛΩΔΊΩΝ ΚΑΙΠΡΟΣΑΡΜΟΓΈΩΝ ΕΚΤΌΣ ΑΠΌ ΤΑ ΠΡΟΒΛΕΠΌΜΕΝΑ,ΜΕ ΤΗΝ ΕΞΑΊΡΕΣΗ ΤΩΝ ΚΑΛΩΔΊΩΝ ΚΑΙΠΡΟΣΑΡΜΟΓΈΩΝ ΠΟΥ ΠΩΛΟΎΝΤΑΙ ΑΠΌ ΤΟΝΚΑΤΑΣΚΕΥΑΣΤΉ ΤΟΥ ΙΑΤΡΙΚΟΎ ΗΛΕΚΤΡΙΚΟΎ
ΕΞΟΠΛΙΣΜΟΎ ΩΣ ΑΝΤΑΛΛΑΚΤΙΚΆ ΓΙΑ ΕΣΩΤΕΡΙΚΆ
ΕΞΑΡΤΉΜΑΤΑ, ΜΠΟΡΕΊ ΝΑ ΟΔΗΓΉΣΕΙ ΣΕΑΥΞΗΜΈΝΕΣ ΕΚΠΟΜΠΈΣ Ή ΜΕΙΩΜΈΝΗ ΑΤΡΩΣΊΑ ΤΟΥΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΜΌΝΟ ΤΗΝΗΛΕΚΤΡΙΚΉ ΤΆΣΗ ΠΟΥ ΚΑΘΟΡΊΖΕΤΑΙ ΣΤΗΝ ΕΤΙΚΈΤΑ
ΠΡΟΔΙΑΓΡΑΦΏΝ ΠΟΥ ΦΈΡΕΙ Η ΣΥΣΚΕΥΉ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΚΑΛΏΔΙΑ
ΕΠΈΚΤΑΣΗΣ ΜΕ ΑΥΤΉΝ ΤΗ ΜΟΝΆΔΑ ΚΑΙ ΜΗΣΥΝΔΈΕΤΕ ΥΠΕΡΒΟΛΙΚΌ ΑΡΙΘΜΌ ΒΥΣΜΆΤΩΝ ΣΤΗΝΊΔΙΑ ΠΑΡΟΧΉ ΡΕΎΜΑΤΟΣ. Η ΧΡΉΣΗ ΚΑΛΩΔΊΩΝ
ΕΠΈΚΤΑΣΗΣ ΘΑ ΜΠΟΡΟΎΣΕ ΝΑ ΕΠΗΡΕΆΣΕΙ
ΑΡΝΗΤΙΚΆ ΤΗΝ ΑΠΌΔΟΣΗ ΤΗΣ ΣΥΣΚΕΥΉΣ.
ΥΠΕΡΒΟΛΙΚΌΣ ΑΡΙΘΜΌΣ ΒΥΣΜΆΤΩΝ ΣΕ ΜΊΑΠΑΡΟΧΉ ΜΠΟΡΕΊ ΝΑ ΟΔΗΓΉΣΕΙ ΣΕ ΥΠΕΡΦΌΡΤΩΣΗ
ΤΟΥ ΗΛΕΚΤΡΙΚΟΎ ΠΊΝΑΚΑ ΠΡΟΚΑΛΏΝΤΑΣ ΤΗΝΕΝΕΡΓΟΠΟΊΗΣΗ ΤΟΥ ΑΣΦΑΛΕΙΟΔΙΑΚΌΠΤΗ/ ΤΗΣΑΣΦΆΛΕΙΑΣ, Ή ΠΥΡΚΑΓΙΆ ΣΕ ΠΕΡΊΠΤΩΣΗ ΠΟΥΟ ΑΣΦΑΛΕΙΟΔΙΑΚΌΠΤΗΣ/ Η ΑΣΦΆΛΕΙΑ ΔΕΝΛΕΙΤΟΥΡΓΉΣΟΥΝ.

FreeStyle ComfortΠΡΟΣΟΧΗ: Σε περίπτωση που εκδηλωθεί
συναγερμός ή που παρατηρήσετε ότι οσυμπυκνωτής οξυγόνου δεν λειτουργεί
κανονικά, συμβουλευτείτε την ενότητα
«Αντιμετώπιση προβλημάτων» στο παρόν
εγχειρίδιο. Εάν δεν μπορείτε να επιλύσετε
το πρόβλημα, συμβουλευτείτε τον πάροχο
του εξοπλισμού σας.

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΣΉΜΑΤΑ «ΑΠΑΓΟΡΕΎΕΤΑΙ ΤΟΚΆΠΝΙΣΜΑ – ΧΡΉΣΗ ΟΞΥΓΌΝΟΥ» ΠΡΈΠΕΙ ΝΑΑΝΑΡΤΏΝΤΑΙ ΣΕ ΠΕΡΊΟΠΤΗ ΘΈΣΗ ΣΤΟ ΣΠΊΤΙ ΉΣΤΟΝ ΧΏΡΟ ΌΠΟΥ ΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ ΟΞΥΓΌΝΟ.
ΟΙ ΧΡΗΣΤΕΣ ΚΑΙ ΟΙ ΦΡΟΝΤΙΣΤΈΣ ΤΟΥΣ ΠΡΈΠΕΙ ΝΑΕΝΗΜΕΡΏΝΟΝΤΑΙ ΣΧΕΤΙΚΆ ΜΕ ΤΟΥΣ ΚΙΝΔΎΝΟΥΣ
ΤΟΥ ΚΑΠΝΊΣΜΑΤΟΣ ΥΠΌ ΤΗΝ ΠΑΡΟΥΣΊΑ, Ή ΚΑΤΆ ΤΗΔΙΆΡΚΕΙΑ ΤΗΣ ΧΡΉΣΗΣ, ΙΑΤΡΙΚΟΎ ΟΞΥΓΌΝΟΥ.

ΠΡΟΣΟΧΗ: Προκειμένου να αποφύγετε την ακύρωση
της εγγύησης, ακολουθείτε όλες τις οδηγίες τουκατασκευαστή.

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΤΟΝΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΥΠΌ ΤΗΝ ΠΑΡΟΥΣΊΑ
ΕΎΦΛΕΚΤΩΝ ΑΕΡΊΩΝ. ΑΥΤΌ ΜΠΟΡΕΊ ΝΑ ΟΔΗΓΉΣΕΙΣΕ ΤΑΧΕΊΑ ΚΑΎΣΗ ΠΡΟΚΑΛΏΝΤΑΣ ΚΑΤΑΣΤΡΟΦΉ
ΙΔΙΟΚΤΗΣΊΑΣ, ΣΩΜΑΤΙΚΟΎΣ ΤΡΑΥΜΑΤΙΣΜΟΎΣ ΉΘΆΝΑΤΟ.

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΕΠΙΧΕΙΡΉΣΕΤΕ
ΟΠΟΙΑΔΉΠΟΤΕ ΕΡΓΑΣΊΑ ΣΥΝΤΉΡΗΣΗΣ
ΕΚΤΌΣ ΑΠΌ ΤΙΣ ΠΙΘΑΝΈΣ ΛΎΣΕΙΣ ΠΟΥΠΑΡΑΤΊΘΕΝΤΑΙ ΣΤΟ ΠΑΡΌΝ ΕΓΧΕΙΡΊΔΙΟ.
ΜΗΝ ΑΦΑΙΡΕΊΤΕ ΤΑ ΚΑΛΎΜΜΑΤΑ· ΜΌΝΟΟ ΠΆΡΟΧΟΣ ΤΟΥ ΕΞΟΠΛΙΣΜΟΎ ΣΑΣΉ ΈΝΑΣ ΕΞΕΙΔΙΚΕΥΜΈΝΟΣ ΤΕΧΝΙΚΌΣ
ΣΈΡΒΙΣ ΕΠΙΤΡΈΠΕΤΑΙ ΝΑ ΑΦΑΙΡΈΣΕΙ ΤΑΚΑΛΎΜΜΑΤΑ Ή ΝΑ ΣΥΝΤΗΡΉΣΕΙ ΑΥΤΉΝΤΗ ΣΥΣΚΕΥΉ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΜΌΝΟΠΑΡΕΛΚΌΜΕΝΑ ΠΟΥ ΣΥΝΙΣΤΏΝΤΑΙ ΑΠΌ ΤΟΝΚΑΤΑΣΚΕΥΑΣΤΉ. Η ΧΡΉΣΗ ΟΠΟΙΩΝΔΉΠΟΤΕ ΆΛΛΩΝ
ΜΠΟΡΕΊ ΝΑ ΕΊΝΑΙ ΕΠΙΚΊΝΔΥΝΗ, ΝΑ ΠΡΟΚΑΛΈΣΕΙ
ΣΟΒΑΡΉ ΖΗΜΙΆ ΣΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΚΑΙΝΑ ΚΑΤΑΣΤΉΣΕΙ ΤΗΝ ΕΓΓΎΗΣΗ ΜΗ ΈΓΚΥΡΗ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΚΑΤΑΣΚΕΥΑΣΤΉΣ ΣΥΝΙΣΤΆ
ΜΙΑ ΕΝΑΛΛΑΚΤΙΚΉ ΠΗΓΉ ΣΥΜΠΛΗΡΩΜΑΤΙΚΟΎ
ΟΞΥΓΌΝΟΥ ΣΕ ΠΕΡΊΠΤΩΣΗ ΔΙΑΚΟΠΉΣ ΡΕΎΜΑΤΟΣ,ΚΑΤΆΣΤΑΣΗΣ ΣΥΝΑΓΕΡΜΟΎ Ή ΜΗΧΑΝΙΚΉΣ ΒΛΆΒΗΣ.
ΣΥΜΒΟΥΛΕΥΤΕΊΤΕ ΤΟΝ ΓΙΑΤΡΌ Ή ΤΟΝ ΠΆΡΟΧΟ
ΤΟΥ ΕΞΟΠΛΙΣΜΟΎ ΓΙΑ ΤΟΝ ΤΎΠΟ ΤΟΥ ΕΦΕΔΡΙΚΟΎ
ΣΥΣΤΉΜΑΤΟΣ ΠΟΥ ΑΠΑΙΤΕΊΤΑΙ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΑΦΉΝΕΤΕ ΝΑ ΦΡΆΞΟΥΝΤΑ ΑΝΟΊΓΜΑΤΑ ΕΙΣΑΓΩΓΉΣ Ή ΕΞΌΔΟΥ ΑΈΡΑ.
ΜΗΝ ΑΦΉΝΕΤΕ ΝΑ ΠΈΣΟΥΝ ΚΑΙ ΜΗΝ ΕΙΣΆΓΕΤΕ
ΑΝΤΙΚΕΊΜΕΝΑ ΣΕ ΚΑΝΈΝΑ ΆΝΟΙΓΜΑ ΤΗΣΣΥΣΚΕΥΉΣ. ΚΆΤΙ ΤΈΤΟΙΟ ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΕΙ
ΥΠΕΡΘΈΡΜΑΝΣΗ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΚΑΙΝΑ ΕΠΗΡΕΆΣΕΙ ΑΡΝΗΤΙΚΆ ΤΗΝ ΑΠΌΔΟΣΗ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΟ ΚΆΠΝΙΣΜΑ ΕΝΏΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ ΟΞΥΓΌΝΟ ΕΊΝΑΙ Η ΠΡΏΤΗ
ΑΙΤΊΑ ΤΡΑΥΜΑΤΙΣΜΏΝ ΑΠΌ ΠΥΡΚΑΓΙΆ ΚΑΙΣΧΕΤΙΖΌΜΕΝΩΝ ΘΑΝΆΤΩΝ. ΠΡΈΠΕΙ ΝΑ ΤΗΡΕΊΤΕ ΤΙΣΑΚΌΛΟΥΘΕΣ ΠΡΟΕΙΔΟΠΟΙΉΣΕΙΣ ΑΣΦΑΛΕΊΑΣ:ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΕΠΙΤΡΈΠΕΤΕ ΤΟ ΚΆΠΝΙΣΜΑ,ΤΑ ΚΕΡΙΆ Ή ΤΙΣ ΑΝΟΙΧΤΈΣ ΦΛΌΓΕΣ ΣΤΟΝ ΊΔΙΟ ΧΏΡΟΜΕ ΤΗ ΣΥΣΚΕΥΉ Ή ΤΑ ΠΑΡΕΛΚΌΜΕΝΑ ΠΟΥ ΦΈΡΟΥΝ
ΟΞΥΓΌΝΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΟ ΚΆΠΝΙΣΜΑ ΕΝΏ ΦΟΡΆΤΕ
ΚΆΝΟΥΛΑ ΟΞΥΓΌΝΟΥ ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΕΙ
ΕΓΚΑΎΜΑΤΑ ΤΟΥ ΠΡΟΣΏΠΟΥ ΚΑΙ ΠΙΘΑΝΏΣ
ΘΆΝΑΤΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΆΝ ΚΑΠΝΊΖΕΤΕ, ΤΑ ΑΚΌΛΟΥΘΑ
3 ΒΉΜΑΤΑ ΜΠΟΡΕΊ ΝΑ ΣΏΣΟΥΝ ΤΗ ΖΩΉ ΣΑΣ:ΑΠΕΝΕΡΓΟΠΟΙΉΣΤΕ ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ,ΑΦΑΙΡΈΣΤΕ ΤΗΝ ΚΆΝΟΥΛΑ ΚΑΙ ΒΓΕΊΤΕ ΑΠΌ ΤΟΝΧΏΡΟ ΌΠΟΥ ΒΡΊΣΚΕΤΑΙ Η ΣΥΣΚΕΥΉ.

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΑΦΑΊΡΕΣΗ ΤΗΣ ΚΆΝΟΥΛΑΣΚΑΙ Η ΑΠΌΘΕΣΉ ΤΗΣ ΕΠΆΝΩ ΣΕ ΕΝΔΎΜΑΤΑ,ΚΛΙΝΟΣΚΕΠΆΣΜΑΤΑ, ΚΑΝΑΠΈΔΕΣ, ΜΑΞΙΛΆΡΙΑ ΉΠΑΡΌΜΟΙΑ ΥΛΙΚΆ ΘΑ ΠΡΟΚΑΛΈΣΕΙ ΚΑΤΆΚΑΥΣΗ
ΑΕΡΊΟΥ ΝΈΦΟΥΣ ΌΤΑΝ ΕΚΤΕΘΕΊ ΣΕ ΤΣΙΓΆΡΟ, ΠΗΓΉΘΕΡΜΌΤΗΤΑΣ Ή ΦΛΌΓΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΑΦΉΝΕΤΕ ΤΗ ΡΙΝΙΚΉ
ΚΆΝΟΥΛΑ ΕΠΆΝΩ Ή ΚΆΤΩ ΑΠΌ ΕΝΔΎΜΑΤΑ,ΚΛΙΝΟΣΚΕΠΆΣΜΑΤΑ Ή ΜΑΞΙΛΆΡΙΑ ΚΑΡΈΚΛΑΣ.
ΕΆΝ Η ΜΟΝΆΔΑ ΕΝΕΡΓΟΠΟΙΗΘΕΊ ΑΛΛΆ ΔΕΝΧΡΗΣΙΜΟΠΟΙΗΘΕΊ, ΤΟ ΟΞΥΓΌΝΟ ΘΑ ΚΑΤΑΣΤΉΣΕΙΤΟ ΥΛΙΚΌ ΕΎΦΛΕΚΤΟ. ΔΙΑΣΦΑΛΊΣΤΕ ΌΤΙ Η ΙΣΧΎΣ
ΕΊΝΑΙ ΑΠΕΝΕΡΓΟΠΟΙΗΜΕΝΗ ΌΤΑΝ Ο ΣΥΜΠΥΚΝΩΤΉΣ
ΟΞΥΓΌΝΟΥ ΔΕΝ ΒΡΊΣΚΕΤΑΙ ΣΕ ΧΡΉΣΗ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΣΥΜΠΥΚΝΩΤΗΣ ΟΞΥΓΟΝΟΥΠΡΕΠΕΙ ΝΑ ΑΠΕΝΕΡΓΟΠΟΙΗΘΕΙ ΚΑΙ ΝΑΑΠΟΣΥΝΔΕΘΕΙ ΑΠΟ ΤΗΝ ΕΞΩΤΕΡΙΚΗ ΤΡΟΦΟΔΟΣΙΑΡΕΥΜΑΤΟΣ ΠΡΙΝ ΤΗΝ ΑΦΑΙΡΕΣΗ ΤΟΥ ΠΑΚΕΤΟΥΜΠΑΤΑΡΙΑΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΑ ΜΙΚΡΆ ΜΈΡΗΜΠΟΡΕΊ ΝΑ ΠΑΡΟΥΣΙΆΖΟΥΝ ΠΙΘΑΝΌ
ΚΊΝΔΥΝΟ ΠΝΙΓΜΟΎ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΠΟΦΎΓΕΤΕ ΤΗΝ ΕΠΑΦΉ ΜΕ ΤΟΆΝΟΙΓΜΑ ΑΠΑΓΌΜΕΝΟΥ ΑΈΡΑ ΓΙΑ ΠΕΡΙΣΣΌΤΕΡΟ
ΑΠΌ 10 ΔΕΥΤΕΡΌΛΕΠΤΑ ΚΑΘΏΣ ΜΠΟΡΕΊ ΝΑ ΕΊΝΑΙ
ΘΕΡΜΌ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΆΝΕΜΟΣ Ή ΤΑ ΙΣΧΥΡΆ
ΡΕΎΜΑΤΑ ΜΠΟΡΕΊ ΝΑ ΕΠΗΡΕΆΣΟΥΝ ΑΡΝΗΤΙΚΆ ΤΗΝΑΚΡΙΒΉ ΧΟΡΉΓΗΣΗ ΟΞΥΓΟΝΟΘΕΡΑΠΕΊΑΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΥΤΌ ΤΟ ΠΡΟΪΌΝ ΜΠΟΡΕΊΝΑ ΣΑΣ ΕΚΘΈΣΕΙ ΣΕ ΧΗΜΙΚΈΣ ΟΥΣΊΕΣ
ΣΥΜΠΕΡΙΛΑΜΒΑΝΟΜΈΝΟΥ ΤΟΥ ΝΙΚΕΛΊΟΥ, ΤΟ ΟΠΟΊΟ
ΕΊΝΑΙ ΓΝΩΣΤΌ ΣΤΗΝ ΠΟΛΙΤΕΊΑ ΤΗΣ ΚΑΛΙΦΌΡΝΙΑΌΤΙ ΠΡΟΚΑΛΕΊ ΚΑΡΚΊΝΟ. ΓΙΑ ΠΕΡΙΣΣΌΤΕΡΕΣ
ΠΛΗΡΟΦΟΡΊΕΣ, ΜΕΤΑΒΕΊΤΕ ΣΤΟ WWW.
P65WARNINGS.CA.GOV.
ΠΡΟΣΟΧΗ: Αποσυνδέετε πάντα τοτροφοδοτικό AC από τον τοίχο πριν τηναποσύνδεση του τροφοδοτικού AC απότον συμπυκνωτή οξυγόνου.
ΠΡΟΣΟΧΗ: Μην τοποθετείτε τον συμπυκνωτή
οξυγόνου σε θέση που καθιστά δύσκολη τηνπρόσβαση στο καλώδιο ρεύματος.
PN MN232-C4 K | Εγχειρίδιο χρήστη

GRE - 349

FreeStyle ComfortΠΡΟΣΟΧΗ: Τοποθετείτε πάντα τη σωλήνωση παροχής
οξυγόνου και τα καλώδια ρεύματος με τρόπο πουαποτρέπει τον κίνδυνο παραπατήματος ή τηνπιθανότητα τυχαίου στραγγαλισμού.
ΠΡΟΣΟΧΗ: Όταν χρησιμοποιείτε τον συμπυκνωτή
οξυγόνου σε αυτοκίνητο, σκάφος ή σε άλλες πηγές DCμε το τροφοδοτικό DC, βεβαιωθείτε ότι το όχημα έχειξεκινήσει και κινείται πριν συνδέσετε τον συμπυκνωτή
οξυγόνου. Η μη τήρηση αυτών των οδηγιών μπορεί ναέχει ως αποτέλεσμα την αποτυχία του τροφοδοτικού
να τροφοδοτήσει με ισχύ τον συμπυκνωτή οξυγόνου.
ΠΡΟΣΟΧΗ: Όταν το όχημα στο οποίο χρησιμοποιείτετη μονάδα συμπυκνωτή οξυγόνου είναι
απενεργοποιημένο, αποσυνδέστε και αφαιρέστε
τη συσκευή από το όχημα. Μην αποθηκεύετε τονσυμπυκνωτή οξυγόνου σε ένα πολύ θερμό ή ψυχρό
όχημα ή σε άλλο παρόμοιο περιβάλλον με υψηλή ήχαμηλή θερμοκρασία.

Σημείωση: Φορητός και κινητός εξοπλισμός
επικοινωνιών με ραδιοσυχνότητες (RF) μπορεί ναεπηρεάσει τον ιατρικό ηλεκτρικό εξοπλισμό.
ΠΡΟΣΟΧΗ: ΜΗΝ αφήνετε τον συμπυκνωτή
οξυγόνου ή το τροφοδοτικό συνδεδεμένοστο όχημα εάν η ανάφλεξη βρίσκεται στηθέση ΑΠΕΝΕΡΓΟΠΟΙΗΣΗΣ. Αυτό μπορεί ναεξαντλήσει την μπαταρία του οχήματος.
ΠΡΟΣΟΧΗ: Η λειτουργία ή αποθήκευση τουσυμπυκνωτή οξυγόνου εκτός του κανονικού εύρους
θερμοκρασίας λειτουργίας και πίεσης περιβάλλοντος
(υψόμετρο) μπορεί να επηρεάσει την απόδοση
και να μειώσει τη διάρκεια ζωής της μπαταρίας ή/και να αυξήσει τον χρόνο φόρτισης της μπαταρίας.
Ανατρέξτε στην ενότητα «Προδιαγραφές» στο παρόν
εγχειρίδιο για τα όρια θερμοκρασίας αποθήκευσης καιλειτουργίας.
ΠΡΟΣΟΧΗ: Για συμπυκνωτές οξυγόνου που είναι
εξοπλισμένοι με μπαταρίες: Αποθηκεύστε σε δροσερό
και ξηρό μέρος για να βοηθήσετε να διασφαλιστεί
η μακροβιότητα της μπαταρίας. Η αποθήκευση τουσυμπυκνωτή οξυγόνου για παρατεταμένες χρονικές
περιόδους σε υψηλές θερμοκρασίες ή με πλήρως
φορτισμένη ή πλήρως αποφορτισμένη μπαταρία
μπορεί να υποβαθμίσει τη συνολική διάρκεια ζωήςτης μπαταρίας. Μην επιχειρήσετε να ανοίξετε τηνμπαταρία, καθώς δεν υπάρχουν μέρη που επιδέχονται
συντήρηση στο εσωτερικό της μπαταρίας. Φυλάσσετε
τις μπαταρίες μακριά από παιδιά.
ΠΡΟΣΟΧΗ: ΧΡΗΣΙΜΟΠΟΙΕΙΤΕ ΜΟΝΟ τις μπαταρίες
που παρέχονται από τον κατασκευαστή. Για σωστή
απόρριψη, επικοινωνήστε με τον πάροχο τουεξοπλισμού σας ή με την αρμόδια υπηρεσία τοπικής
αυτοδιοίκησης για τις απαιτήσεις απόρριψης.

350 - GREPN MN232-C4 K | Εγχειρίδιο χρήστη

Σημείωση: Διασφαλίστε ότι η κάνουλα έχει εισαχθεί
και στερεωθεί πλήρως. Αυτό διασφαλίζει ότι οσυμπυκνωτής οξυγόνου μπορεί να ανιχνεύει σωστά
την εισπνοή για χορήγηση οξυγόνου. Κατά τηδιάρκεια της εισπνοής, θα πρέπει να ακούσετε ήνα αισθανθείτε τη ροή του οξυγόνου στα άκρα τηςρινικής κάνουλας.
Σημείωση: Οι μπαταρίες λιθίου μπορεί να χάσουν
μόνιμα τη χωρητικότητά τους όταν εκτεθούν σε υπερβολικά υψηλές θερμοκρασίες με τις μπαταρίες τελείως
φορτισμένες ή τελείως εξαντλημένες. Για μακροχρόνια
αποθήκευση, συνιστάται οι μπαταρίες να είναι φορτισμένες κατά 25 έως 50% και να παραμένουν εντός τουεύρους θερμοκρασίας -20°C – 20°C (-4°F – 68°F).
Σημείωση: Η χρήση ορισμένων παρελκομένων
χορήγησης οξυγόνου που δεν καθορίζονται γιαχρήση με αυτόν τον συμπυκνωτή οξυγόνου μπορεί ναεπηρεάσει αρνητικά την απόδοσή του. Παραπομπές
στα συνιστώμενα παρελκόμενα παρέχονται σε αυτότο εγχειρίδιο.
Σημείωση: Μη λειτουργείτε τον συμπυκνωτή οξυγόνου χωρίς να είναι τοποθετημένο το φίλτρο εισαγωγής αέρα. Εάν παρέχεται ένα δεύτερο φίλτρο, τοποθετήστε το ανταλλακτικό φίλτρο πριν να καθαρίσετε
το βρώμικο φίλτρο. Καθαρίστε το βρώμικο φίλτρο μεζεστό διάλυμα σαπουνιού και νερού και στη συνέχεια
στεγνώστε τελείως πριν από τη χρήση.
Σημείωση: Ο κατασκευαστής δεν συνιστά την αποστείρωση αυτού του εξοπλισμού.
Σημείωση: Δεν υπάρχει ποτέ κίνδυνος εξάντλησηςτου οξυγόνου σε ένα δωμάτιο όταν χρησιμοποιείτε τημονάδα συμπυκνωτή οξυγόνου.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΣΥΜΠΥΚΝΩΤΉΣ
ΟΞΥΓΌΝΟΥ ΔΕΝ ΠΡΈΠΕΙ ΝΑ ΛΕΙΤΟΥΡΓΕΊΉ ΝΑ ΣΥΝΔΈΕΤΑΙ ΣΕ ΕΞΩΤΕΡΙΚΉ
ΙΣΧΎ (AC/DC) ΧΩΡΊΣ ΤΟΠΟΘΕΤΗΜΈΝΗ
ΜΠΑΤΑΡΊΑ. ΜΗΝ ΕΠΙΧΕΙΡΉΣΕΤΕ ΝΑΕΝΕΡΓΟΠΟΙΉΣΕΤΕ ΤΗ ΜΟΝΆΔΑ Ή ΝΑ ΤΗΣΥΝΔΈΣΕΤΕ ΣΕ ΕΞΩΤΕΡΙΚΉ ΠΗΓΉ ΙΣΧΎΟΣ
ΌΤΑΝ ΤΟ ΠΑΚΈΤΟ ΜΠΑΤΑΡΊΑΣ ΈΧΕΙΑΦΑΙΡΕΘΕΊ.
ΑΦΑΙΡΈΣΤΕ ΤΟ ΠΑΚΈΤΟ ΜΠΑΤΑΡΊΑΣ ΜΌΝΟΌΤΑΝ Η ΜΟΝΆΔΑ ΕΊΝΑΙ ΑΠΕΝΕΡΓΟΠΟΙΗΜΕΝΗΚΑΙ ΑΠΟΣΥΝΔΕΔΕΜΈΝΗ ΑΠΌ ΤΗΝ ΤΡΟΦΟΔΟΣΊΑ
ΡΕΎΜΑΤΟΣ. ΜΗΝ ΕΠΙΧΕΙΡΉΣΕΤΕ ΝΑ ΑΦΑΙΡΈΣΕΤΕΤΟ ΠΑΚΈΤΟ ΜΠΑΤΑΡΊΑΣ ΕΝΏ Η ΜΟΝΆΔΑ ΕΊΝΑΙ
ΑΚΌΜΑ ΕΝΕΡΓΟΠΟΙΗΜΕΝΗ Ή ΣΥΝΔΕΔΕΜΈΝΗ ΣΤΗΝΤΡΟΦΟΔΟΣΊΑ ΡΕΎΜΑΤΟΣ.

Σημείωση: Εάν ο συμπυκνωτής οξυγόνου έχειαποθηκευτεί για παρατεταμένη χρονική περίοδο
εκτός των κανονικών περιβαλλοντικών συνθηκών
λειτουργίας του, θα πρέπει να αφεθεί να επιστρέψει
στην κανονική θερμοκρασία λειτουργίας προτού
ενεργοποιηθεί.

FreeStyle ComfortΠροδιαγραφές
FreeStyle ComfortΔιαστάσεις (με μονή μπαταρία)25,4 x 18,5 x 7,9 cm (10,0 x 7,3 x 3,1 in)Διαστάσεις (με διπλή μπαταρία)27,9 x 18,5 x 7,9 cm (11,0 x 7,3 x 3,1 in)Βάρος (με πακέτο μονής μπαταρίας)2,3 kg (5 lb)Βάρος (με πακέτο διπλής μπαταρίας)2,7 kg (6 lb)Ονομαστικό επίπεδο ήχου***

39,93 dB (A) ± 0,42 dB (A) @ 2Ρυθμίσεις ροήςΡύθμιση παλμού 1, 2, 3, 4, 5Ρύθμιση εξόδου O2Ρύθμιση 1 - 210 mL/minΡύθμιση 2 - 420 mL/minΡύθμιση 3 - 630 mL/minΡύθμιση 4 - 840 mL/minΡύθμιση 5 - 1050 mL/min(@ 20 bpm)Συμπύκνωση οξυγόνου*Σύστημα παρακολούθησης οξυγόνου
Μέγιστη πίεση εξόδου
Προεπιλεγμένη ευαισθησία δοσολόγησης

90% (+5,5% / -3%)Ναι<30 PSI> -0,1 cm H2O (ρυθμιζόμενη)Ισχύς AC

100–240 VAC, 50–60 Hz, 120W Max

Ισχύς DC

11–18 VDC (10 amp μέγ.)Χρόνος επαναφόρτισης μπαταρίας

Μονή μπαταρία: 3,5 ώρεςΔιπλή μπαταρία: 6,0 ώρεςΔιάρκεια πακέτου μπαταρίας

Διάρκεια μονής μπαταρίας: Ρύθμιση 2 - έως 4 ώρεςΔιάρκεια διπλής μπαταρίας: Ρύθμιση 2 - έως 8 ώρεςΘερμοκρασία λειτουργίας**Τύπος μπαταρίας
Χωρητικότητα μπαταρίας

5°C έως 40°C (41°F έως 104°F)Ιόντων λιθίου
Ενός κυττάρου: 6700 mAh; Διπλό κύτταρο: 13400 mAhΘερμοκρ. μεταφοράς και αποθήκευσης
(μονάδα)-25°C έως +70°C (-13°F έως +158°F) έως και τα 3048 m (10.000 ft)

Θερμοκρ. μεταφοράς και αποθήκευσης
(μπαταρία)-25°C έως +70°C (-13°F έως +158°F) έως και τα 3048 m (10.000 ft)

Υψόμετρο λειτουργίας**-381 έως 3048 m (-1250 έως 10.000 ft) (ελέγχθηκε στα 700 – 1060hPa)Υγρασία λειτουργίας

15 – 95% σχετική υγρασία (χωρίς συμπύκνωση υδρατμών)Υγρασία αποθήκευσης

0 - 90% χωρίς συμπύκνωσηΠεριορισμένη εγγύηση

3 έτη - μονάδα
1 έτος - μπαταρία και τροφοδοτικά

* Με βάση ατμοσφαιρική πίεση 101 kPa (14,7 psi) στους 21°C (70°F)** Η μονάδα θα λειτουργεί σε όλα τα περιβάλλοντα, εφόσον το περιβάλλον είναι με τα καθορισμένα εύρη θερμοκρασίας καιυγρασίας. Η λειτουργία εκτός αυτών των προδιαγραφών λειτουργίας μπορεί να περιορίσει τη δυνατότητα του συμπυκνωτή
να ικανοποιεί την προδιαγραφή του συμπυκνωτή οξυγόνου σε υψηλότερους ρυθμούς ροής λίτρων. Η οθόνη προθέρμανσης
μπορεί να συνεχίσει να εμφανίζεται μέχρι ο συμπυκνωτής να φθάσει τη συγκέντρωση-στόχο ή έως και 15 λεπτά. Συνεχίζετε ναχρησιμοποιείτε τη μονάδα κατά τη διάρκεια αυτού του χρονικού διαστήματος.
*** Η στάθμη θορύβου που μετράται ανά μέθοδο δοκιμής Αρ. 14-1 10/2018 MDS-Hi.
Η ελάχιστη αναμενόμενη ωφέλιμη διάρκεια ζωής είναι πέντε χρόνια. Η αναμενόμενη διάρκεια ζωής των στρωμάτων μοριακού
κόσκινου είναι περίπου ένα έτος εάν ο συμπυκνωτής παραμείνει αχρησιμοποίητος καθ' όλη τη χρονική περίοδο.
PN MN232-C4 K | Εγχειρίδιο χρήστη

GRE - 351

FreeStyle ComfortΟδηγίες λειτουργίας

1. Καθορίστε τη θέση και τοποθετήστε το FreeStyle
Comfort σε μια κατάλληλη θέση για χρήση.
α. Ο συμπυκνωτής πρέπει να αερίζεται καλά έτσιώστε τα φίλτρα εισαγωγής αέρα και η έξοδος
αέρα να μην εμποδίζονται.
β. Ο συμπυκνωτής πρέπει να τοποθετείται μετέτοιον τρόπο ώστε όλες οι ηχητικές και οπτικές
ενδείξεις και συναγερμοί να είναι εύκολα
ακουστοί και ορατοί.
γ. Βεβαιωθείτε ότι τα φίλτρα εισαγωγής αέραβρίσκονται στη θέση τους πριν τη λειτουργία
της μονάδας. Εάν ένα φίλτρο λείπει από τημονάδα, επικοινωνήστε με τον πάροχο τουεξοπλισμού σας.
2. Εγκαταστήστε ένα πακέτο μπαταρίας στον
συμπυκνωτή.
α. Ευθυγραμμίστε την μπαταρία με το κάτωπερίβλημα από το κέντρο του FreeStyle Comfort.

3. Εισάγετε το τροφοδοτικό AC ή DC μέσα στηνυποδοχή θύρας τροφοδοσίας στην πρόσοψη, στηνκάτω δεξιά πλευρά της μονάδας.

4. Συνδέστε μια ρινική κάνουλα στη θύρα εξόδου
οξυγόνου. Διασφαλίστε ότι η σύνδεση είναι ασφαλής
πριν τη λειτουργία.

ΣΗΜΕΙΩΣΗ: Η χρήση διαφορετικής(ών)κάνουλας(ών) από τη συνιστώμενη μπορεί ναεπηρεάσει την απόδοση.

Σημείωση: Για τον παροχέα εξοπλισμού: Τα παρακάτω
εξαρτήματα χορήγησης οξυγόνου συνιστώνται για χρήση
με το FreeStyle Comfort:β. Ολισθήστε την μπαταρία προς τα μέσα μέχρι ναακούσετε ένα κλικ για να επιβεβαιώσετε ότι έχεικλειδώσει στη θέση της.

352 - GREPN MN232-C4 K | Εγχειρίδιο χρήστη• Ρινική κάνουλα με σωλήνα (μέγιστο LPM: 6 LPM):Αριθμός 5408-SEQ• Πρόσθετες προτεινόμενες πληροφορίες για αξεσουάρ
είναι διαθέσιμες στο διαδίκτυο στη διεύθυνση www.
caireinc.com.

FreeStyle Comfort
5. Ενεργοποιήστε το FreeStyle Comfortπατώντας το κουμπί λειτουργίας και επιλέξτε τησυνταγογραφημένη ρύθμιση ροής χρησιμοποιώντας
τα κουμπιά αύξησης (+) ή μείωσης (-) ροής.

Ρύθμιση όγκου παλμού autoSAT*(mL)Αναπνευστικός
ρυθμός

1

2

3

4

5

15

10,5

21,0

31,5

42,0

52,5

20

10,5

21,0

31,5

42,0

52,5

25

10,5

21,0

31,5

42,0

42,0

30

10,5

21,0

31,5

35,0

35,0

35

10,5

21,0

30,0

30,0

30,0

40

10,5

21,0

26,3

26,3

26,3

*Η ανοχή είναι +/-15%Ρύθμιση όγκου παλμού μηautoSAT* (mL)Θα παραχθεί ένας σύντομος ηχητικός συναγερμός
υποδεικνύοντας ότι η συσκευή έχει ενεργοποιηθεί.

Σημείωση: Όταν το FreeStyle Comfort ενεργοποιείται
για πρώτη φορά, η προεπιλεγμένη ρύθμιση ροήςείναι 2. Εάν η ρύθμιση αλλάξει, η μονάδα θαενεργοποιηθεί στην προηγούμενη ρύθμιση ροής.
6. Τοποθετήστε τη ρινική κάνουλα στο πρόσωπό σαςσύμφωνα με τις οδηγίες του κατασκευαστή πουπαρέχονται με την κάνουλα και αναπνέετε κανονικά.
Θα πρέπει να αισθανθείτε την χορήγηση μιας ριπής
οξυγόνου διαμέσου της ρινικής κάνουλας κάθε φοράπου εισπνέετε.
7. Για να αλλάξετε τη ρύθμιση ροής, πατήστε τακουμπιά αύξησης (+) ή μείωσης (-) ροής μέχρι
τον επιθυμητό ρυθμό ροής. Είναι φυσιολογικό ναακούσετε μια διαφορά στον ήχο καθώς αλλάζετετις ρυθμίσεις ροής.

Αναπνευστικός
ρυθμός

1

2

3

4

5

15

14,0

28,0

42,0

56,0

70,0

20

10,5

21,0

31,5

42,0

52,5

25

8,4

16,8

25,2

33,6

42,0

30

7,0

14,0

21,0

28,0

35,0

35

6,0

12,0

18,0

24,0

30,0

40

5,3

10,5

15,8

21,0

26,3

8. Απενεργοποιήστε το FreeStyle Comfort πατώνταςτο κουμπί λειτουργίας.

Σημείωση: Η οθόνη ενδείξεων θα τεθεί σεκατάσταση αναμονής και θα σκοτεινιάσει μετά από
20 δευτερόλεπτα. Πατήστε οποιοδήποτε κουμπί γιανα ενεργοποιήσετε την οθόνη από την κατάσταση
αναμονής.
9. Για προηγμένα χαρακτηριστικά, βλ. Οδηγίες
πλοήγησης μενού.

ΕΙΔΙΚΆ ΧΑΡΑΚΤΗΡΙΣΤΙΚΆ ΠΡΟΪΌΝΤΟΣ
autoSAT®
Η ιδιόκτητη λειτουργία autoSAT της CAIRE επιτρέπειστο FreeStyle Comfort να παρέχει συνεχή όγκοοξυγόνου (bolus) ανά αναπνοή, μέχρι το μέγιστορυθμό αναπνοής. Καθώς αυξάνεται ο ρυθμός αναπνοής
του χρήστη κατά τη διάρκεια δραστηριοτήτων όπως τοβάδισμα, η λειτουργία autoSAT ρυθμίζει τη λειτουργίατης συσκευής για να διατηρήσει μια ρυθμισμένη
ένταση βλωμού.
UltraSense®
Αυτό το ευαίσθητο έναυσμα παλμικής δόσης ανιχνεύει
την αλλαγή πίεσης κατά τη διάρκεια της αναπνοής
σας και απελευθερώνει τον παλμό οξυγόνου ότανεισπνέετε.
autoDOSE™
Σε περίπτωση που προκύψει «συναγερμός
μη αναπνοής», αυτό το χαρακτηριστικό θα χορηγεί
παλμικό οξυγόνο σε ρυθμό 20 αναπνοών ανά λεπτό
στην τρέχουσα ρύθμιση ροής της μονάδας μέχρι να
PN MN232-C4 K | Εγχειρίδιο χρήστη GRE - 353
επιλυθεί η κατάσταση συναγερμού.

FreeStyle ComfortΤροφοδοτικά

Το FreeStyle Comfort μπορεί να τροφοδοτηθεί με τρεις
τρόπους:• Επαναφορτιζόμενο πακέτο μπαταρίας• Ισχύς AC• Ισχύς DCΠληροφορίες σχετικά με τη λειτουργία καθενός απόαυτά τα στοιχεία παρέχονται στις ακόλουθες ενότητες.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ
ΜΌΝΟ ΤΑ ΤΡΟΦΟΔΟΤΙΚΆ ΠΟΥΚΑΘΟΡΊΖΟΝΤΑΙ ΣΕ ΑΥΤΌ ΤΟ ΕΓΧΕΙΡΊΔΙΟ
ΧΡΉΣΤΗ. Η ΧΡΉΣΗ ΤΡΟΦΟΔΟΤΙΚΏΝ
ΚΑΙ ΠΑΚΈΤΩΝ ΜΠΑΤΑΡΊΑΣ ΆΛΛΩΝ
ΑΠΌ CAIRE ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΕΙ
ΖΗΜΙΆ ΣΤΗ ΜΟΝΆΔΑ ΚΑΙ ΑΚΑΤΆΛΛΗΛΗΛΕΙΤΟΥΡΓΊΑ.

ΑΦΑΙΡΕΣΗ ΤΗΣ ΜΠΑΤΑΡΙΑΣ

Πιέστε το κουμπί απελευθέρωσης της μπαταρίας
προς τα κάτω μακριά από το FreeStyle Comfort.
Ενώ ΚΡΑΤΑΤΕ αυτό το κουμπί, ολισθήστε τηνμπαταρία προς τα έξω, τραβώντας την μακριά από τονσυμπυκνωτή.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΣΥΜΠΥΚΝΩΤΉΣ
ΟΞΥΓΌΝΟΥ ΠΡΈΠΕΙ ΝΑΑΠΕΝΕΡΓΟΠΟΙΗΘΕΙ ΚΑΙ ΝΑΑΠΟΣΥΝΔΕΘΕΊ ΑΠΌ ΤΗΝ ΕΞΩΤΕΡΙΚΉ
ΤΡΟΦΟΔΟΣΊΑ ΡΕΎΜΑΤΟΣ ΠΡΙΝ ΤΗΝΑΦΑΊΡΕΣΗ ΤΟΥ ΠΑΚΈΤΟΥ ΜΠΑΤΑΡΊΑΣ.

ΠΑΚΕΤΟ ΜΠΑΤΑΡΙΑΣ

Το FreeStyle Comfort μπορεί να λειτουργήσει με
το επαναφορτιζόμενο πακέτο μπαταρίας, το οποίο
παρέχεται με τη συσκευή. Δεν υπάρχουν λειτουργικοί
περιορισμοί του συμπυκνωτή όταν λειτουργεί απότο πακέτο μπαταρίας. Όλοι οι ρυθμοί ροής είναι
διαθέσιμοι με ισχύ από την μπαταρία.

ΕΓΚΑΤΑΣΤΑΣΗ ΤΗΣ ΜΠΑΤΑΡΙΑΣ

Ευθυγραμμίστε την μπαταρία με το κάτω περίβλημααπό το κέντρο του FreeStyle Comfort. Ωθήστε τηνμπαταρία προς τα μέσα μέχρι να ακούσετε ένα κλικ γιανα επιβεβαιώσετε ότι έχει κλειδώσει στη θέση της.

ΕΛΕΓΧΟΣ ΤΗΣ ΦΟΡΤΙΣΗΣ ΤΟΥ ΠΑΚΕΤΟΥΜΠΑΤΑΡΙΑΣΓια να ελέγξετε τη φόρτιση της μπαταρίας ότανείναι εγκατεστημένη στο FreeStyle Comfort,συμβουλευτείτε το εικονίδιο μπαταρίας και τονεκτιμώμενο υπολειπόμενο χρόνο στην οθόνη.
80–100%60–80%

40–60%

20–40%

20%354 - GREPN MN232-C4 K | Εγχειρίδιο χρήστη

FreeStyle ComfortΣημείωση: Εάν η μπαταρία βρίσκεται σε διαδικασία
φόρτισης, θα εμφανίζεται μια ένδειξη βύσματος αντίγια το υπολειπόμενο ποσοστό φόρτισης.
Για να ελέγξετε τη φόρτιση της μπαταρίας όταν δεν είναι
εγκατεστημένη στο FreeStyle Comfort, ακουμπίστε τηνεπίπεδα και πατήστε το κουμπί «TEST». Οι ενδεικτικές
λυχνίες μετρητή μπαταρίας (25-100%) θα ανάψουν σταδεξιά του κουμπιού TEST υποδεικνύοντας το επίπεδο
της φόρτισης του πακέτου μπαταρίας.

ΔΙΑΡΚΕΙΑ ΛΕΙΤΟΥΡΓΙΑΣ ΜΕ ΜΠΑΤΑΡΙΑ

Όταν αποσυνδεθεί η εξωτερική ισχύς AC ή DC,
ο συμπυκνωτής θα αλλάξει αυτόματα σε χρήση
ισχύος από την μπαταρία εάν έχει εγκατασταθεί έναφορτισμένο πακέτο μπαταρίας. Ο πίνακας παρακάτω
δείχνει την τυπική διάρκεια για ένα καινούριο πακέτο
μπαταρίας. Η διάρκεια του πακέτου μπαταρίας θαπαραμείνει σταθερή ανεξάρτητα από τον αναπνευστικό
ρυθμό σας.
Ρύθμισηπαλμού

Διάρκεια μονής
μπαταρίας
(8 στοιχείων)Διάρκεια διπλής
μπαταρίας
(16 στοιχείων)1

έως 8 ώρεςέως 16 ώρες2

έως 4 ώρεςέως 8 ώρες3

έως 3 ώρεςέως 6 ώρες4

έως 2,25 ώρεςέως 4,5 ώρες5

έως 2 ώρεςέως 4 ώρεςΤΥΠΙΚΟΣ ΧΡΟΝΟΣ ΕΠΑΝΑΦΟΡΤΙΣΗΣΜΠΑΤΑΡΙΑΣΑΡΧΙΚΗ ΦΟΡΤΙΣΗΗ νέα μπαταρία που παρέχεται με το FreeStyle
Comfort είναι μόνο μερικώς φορτισμένη όταναποστέλλεται από το εργοστάσιο. Πριν τη χρήση γιαπρώτη φορά, πρέπει να φορτίσετε πλήρως το πακέτο
μπαταρίας χρησιμοποιώντας είτε ισχύ AC είτε DC.

ΓΕΝΙΚΕΣ ΠΛΗΡΟΦΟΡΙΕΣ ΦΟΡΤΙΣΗΣ

Το FreeStyle Comfort θα επαναφορτίζει το πακέτο
μπαταρίας κάθε φορά που είναι εγκατεστημένο στον
συμπυκνωτή και ο συμπυκνωτής είναι συνδεδεμένος
σε μια εξωτερική πηγή ισχύος AC ή DC (εκτός εάνβρίσκεται σε αεροσκάφος).

Ο τυπικός χρόνος για την πλήρη επαναφόρτιση τουπακέτου μπαταρίας από μια πλήρως αποφορτισμένη
μπαταρία είναι 3,5 ώρες για μια μονή μπαταρία και6,0 ώρες για μια διπλή μπαταρία.

Σημείωση: Η μπαταρία μπορεί να μην ξεκινήσει ναφορτίζεται αρχικά εάν το πακέτο μπαταρίας έχειεξαντληθεί πλήρως.

ΩΦΕΛΙΜΗ ΔΙΑΡΚΕΙΑ ΖΩΗΣ ΜΠΑΤΑΡΙΑΣ
Οι μπαταρίες του FreeStyle Comfort είναι
σχεδιασμένες για να διαρκούν για 300 κύκλους
φόρτισης/αποφόρτισης.

Η μπαταρία είναι πλήρως φορτισμένη όταν εμφανίζει
την ένδειξη 100%.
Εναλλακτικά, το πακέτο μπαταρίας μπορεί ναφορτιστεί χρησιμοποιώντας τον προαιρετικό
επιτραπέζιο φορτιστή.

PN MN232-C4 K | Εγχειρίδιο χρήστη

GRE - 355

FreeStyle ComfortΤροφοδοτικό AC και DC
Το FreeStyle Comfort διατίθεται τυπικά μετροφοδοτικά AC και DC.
Το τροφοδοτικό AC περιλαμβάνει τα ακόλουθα:α. Μονάδα τροφοδοτικού AC που λειτουργεί
μεταξύ 100-240 volts, 50-60Hz.
β. Καλώδιο τροφοδοσίας AC για τη σύνδεση
στο τροφοδοτικό και στην αντίστοιχη παροχή
AC.

Το τροφοδοτικό DC αποτελείται από ένα καλώδιο πουσυνδέεται απευθείας στη μονάδα και στην παροχή DC.

2. Συνδέστε το καλώδιο τροφοδοσίας AC στοτροφοδοτικό AC.

ΙΣΧΥΣ ACΗ ισχύς AC προορίζεται για χρήση στο σπίτι ήοπουδήποτε είναι διαθέσιμη ισχύς παροχής AC.
Το FreeStyle Comfort είναι πλήρως λειτουργικό
με ισχύ AC. Το FreeStyle Comfort επαναφορτίζει
επίσης το πακέτο μπαταρίας οποιαδήποτε στιγμή
είναι διαθέσιμη ισχύς AC (εκτός εάν βρίσκεται σεαεροσκάφος). Το πακέτο μπαταρίας θα επαναφορτιστεί
είτε το FreeStyle Comfort είναι ενεργοποιημένο είτεαπενεργοποιημένο.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΣΥΝΔΈΕΤΕ ΤΗΝΕΞΩΤΕΡΙΚΉ ΙΣΧΎ ΧΩΡΊΣ ΝΑ ΥΠΆΡΧΕΙ
ΕΓΚΑΤΕΣΤΗΜΈΝΟ ΈΝΑ ΠΑΚΈΤΟ
ΜΠΑΤΑΡΊΑΣ ΣΤΟΝ ΣΥΜΠΥΚΝΩΤΉ
ΟΞΥΓΌΝΟΥ.
Για τη σύνδεση του FreeStyle Comfort στην ισχύ AC:
1. Συνδέστε το τροφοδοτικό AC στην είσοδο
συνδέσμου ισχύος στην πρόσοψη, στην κάτω δεξιά
πλευρά του FreeStyle Comfort. Βεβαιωθείτε ότιωθήσατε προς τα μέσα έτσι ώστε η σύνδεση να είναι
ασφαλής.

356 - GREPN MN232-C4 K | Εγχειρίδιο χρήστη

3. Συνδέστε το αντίθετο άκρο του καλωδίου
τροφοδοσίας AC στην επιτοίχια πρίζα ή άλληκατάλληλη παροχή ρεύματος.

FreeStyle Comfort
4. Ενεργοποιήστε το FreeStyle Comfortπατώντας το κουμπί λειτουργίας και επιλέξτε τησυνταγογραφημένη ρύθμιση ροής χρησιμοποιώντας
τα κουμπιά αύξησης (+) ή μείωσης (-) ροής.

Σημείωση: Εάν το πακέτο μπαταρίας χρειάζεται
επαναφόρτιση, το εικονίδιο μπαταρίας στην οθόνη
θα έχει μία μόνο πορτοκαλί ράβδο.

ΙΣΧΥΣ DCΗ ισχύς DC προορίζεται για χρήση στις πρίζες
παρελκομένων αυτοκινήτων, σκαφών ή άλλων
μηχανοκίνητων οχημάτων. Το FreeStyle Comfortείναι πλήρως λειτουργικό για χρήση με ισχύ DC.
Το FreeStyle Comfort επαναφορτίζει επίσης τοπακέτο μπαταρίας εάν χρειάζεται, οποιαδήποτε στιγμή
είναι διαθέσιμη ισχύς DC. Το πακέτο μπαταρίας θαεπαναφορτιστεί είτε η μονάδα είναι απενεργοποιημένη,είτε είναι ενεργοποιημένη και σε χρήση.

3. Συνδέστε το αντίθετο άκρο του καλωδίου
τροφοδοσίας DC στην πρίζα του αυτοκινήτου ήάλλη παρόμοια παροχή αξεσουάρ DC.

4. Ενεργοποιήστε το FreeStyle Comfortπατώντας το κουμπί λειτουργίας και επιλέξτε τησυνταγογραφημένη ρύθμιση ροής χρησιμοποιώντας
τα κουμπιά αύξησης (+) ή μείωσης (-) ροής.

Σημείωση: Εάν το πακέτο μπαταρίας χρειάζεται
επαναφόρτιση, το εικονίδιο μπαταρίας στην οθόνη
θα έχει μία μόνο πορτοκαλί ράβδο.

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΣΥΝΔΈΕΤΕ ΤΗΝΕΞΩΤΕΡΙΚΉ ΙΣΧΎ ΧΩΡΊΣ ΝΑ ΥΠΆΡΧΕΙ
ΕΓΚΑΤΕΣΤΗΜΈΝΟ ΈΝΑ ΠΑΚΈΤΟ
ΜΠΑΤΑΡΊΑΣ ΣΤΟΝ ΣΥΜΠΥΚΝΩΤΉ
ΟΞΥΓΌΝΟΥ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΛΕΙΤΟΥΡΓΕΊΤΕ ΤΗ ΣΥΣΚΕΥΉ ΣΕΠΡΊΖΑ ΑΥΤΟΚΙΝΉΤΟΥ 12 VDC ΕΝΏ ΤΟ ΌΧΗΜΑ ΕΊΝΑΙ
ΑΠΕΝΕΡΓΟΠΟΙΗΜΈΝΟ.
1. Ενεργοποιήστε το μηχανοκίνητο όχημά σας.
2. Συνδέστε το τροφοδοτικό DC στην υποδοχή θύρας
τροφοδοσίας στην πρόσοψη, στην κάτω δεξιά
πλευρά του FreeStyle Comfort. Βεβαιωθείτε ότιωθήσατε προς τα μέσα έτσι ώστε η σύνδεση να είναι
ασφαλής.

PN MN232-C4 K | Εγχειρίδιο χρήστη

GRE - 357

FreeStyle Comfort
Καθαρισμός, φροντίδα καισυντήρηση ρουτίνας
ΣΥΝΤΗΡΗΣΗ ΡΟΥΤΙΝΑΣΠροληπτική συντήρηση
Η συσκευή δεν απαιτεί συντήρηση. Τα φίλτρα μέσαστη συσκευή προορίζονται να διαρκέσουν τη διάρκεια
ζωής της συσκευής. Εάν η συσκευή δεν λειτουργεί
σωστά, η μονάδα πρέπει να επιστραφεί στον πάροχο
για επισκευή. Η συντήρηση των εσωτερικών
εξαρτημάτων στο εσωτερικό του περιβλήματος τουFreeStyle Comfort πρέπει να πραγματοποιείται απότον πάροχο οξυγόνου.
Ο χειριστής της μονάδας είναι υπεύθυνος για τιςακόλουθες διαδικασίες συντήρησης.
• Εβδομαδιαία – Καθαρισμός των φίλτρωνεισαγωγής αέραΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΕΠΙΧΕΙΡΉΣΕΤΕΝΑ ΑΝΟΊΞΕΤΕ Ή ΝΑ ΑΦΑΙΡΈΣΕΤΕΤΑ ΠΛΑΣΤΙΚΆ ΚΑΛΎΜΜΑΤΑ ΤΟΥΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ. Η ΜΟΝΆΔΑ
ΠΡΈΠΕΙ ΝΑ ΑΝΟΊΓΕΤΑΙ ΜΌΝΟ ΑΠΌ ΈΝΑΝ
ΕΙΔΙΚΕΥΜΈΝΟ ΤΕΧΝΙΚΌ ΣΈΡΒΙΣ.

ΦΙΛΤΡΑ ΕΙΣΑΓΩΓΗΣ ΑΕΡΑΟ αέρας εισέρχεται στο FreeStyle Comfort μέσωτων φίλτρων εισαγωγής αέρα που βρίσκονται στηνπρόσοψη, στο κάτω μέρος του συμπυκνωτή οξυγόνου.
Αυτά τα φίλτρα εμποδίζουν τη σκόνη και άλλα μεγάλα
σωματίδια στον αέρα να εισέλθουν στη μονάδα.
Τουλάχιστον μία φορά την εβδομάδα, τα φίλτρα
εισαγωγής αέρα πρέπει να καθαρίζονται. Ο πάροχος
οξυγόνου σας μπορεί να σας συμβουλεύσει να τακαθαρίζετε συχνότερα, ανάλογα με τις συνθήκες
λειτουργίας. Ακολουθήστε αυτά τα βήματα για τονσωστό καθαρισμό των φίλτρων εισαγωγής αέρα.
1. Αφαιρέστε τα δύο φίλτρα από την πρόσοψη τουFreeStyle Comfort ωθώντας προς τα πάνω τηναντίστοιχη γλωττίδα και τραβώντας προς τα έξω.

4. Επανατοποθετήστε τα καθαρά, στεγνά φίλτρα.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΛΕΙΤΟΥΡΓΕΊΤΕ
ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΧΩΡΊΣΝΑ ΕΊΝΑΙ ΤΟΠΟΘΕΤΗΜΈΝΟ ΤΟ ΦΊΛΤΡΟ
ΕΙΣΑΓΩΓΉΣ ΑΈΡΑ.

ΦΡΟΝΤΙΔΑ ΤΟΥ ΠΑΚΕΤΟΥ ΜΠΑΤΑΡΙΑΣ

Μην αφήνετε το πακέτο μπαταρίας να εκτεθεί σε υγρά.
Μην επιχειρήσετε να αποσυναρμολογήσετε ή ναανοίξετε το πακέτο μπαταρίας.
Μην εκθέτετε την μπαταρία σε γυμνές φλόγες.
Μην αφήνετε το πακέτο μπαταρίας να πέσει.
Φυλάσσετε το πακέτο μπαταρίας μακριά από παιδιά.
Εάν το πακέτο μπαταρίας φαίνεται να έχει υποστεί
ζημιά, επικοινωνήστε με τον πάροχο οξυγόνου σαςπριν τη χρήση.
Μην τοποθετείτε την μπαταρία μέσα ή κοντά σεφούρνο μικροκυμάτων ή άλλες μαγειρικές συσκευές.

ΑΠΟΘΗΚΕΥΣΗ ΤΟΥ ΠΑΚΕΤΟΥ ΜΠΑΤΑΡΙΑΣ
Η μπαταρία του συμπυκνωτή οξυγόνου δενχρειάζεται να έχει αποφορτιστεί πλήρως πριν από τηνεπαναφόρτιση. Συνιστάται η φόρτιση της μπαταρίας
του συμπυκνωτή οξυγόνου μετά από κάθε χρήση.

Μην αφήνετε την μπαταρία σε μέρος εκτεθειμένο
σε υπερβολική θερμότητα, όπως σε άμεση ηλιακή
ακτινοβολία ή μέσα σε αυτοκίνητο. Αυτό θα μπορούσε
να προκαλέσει υποβάθμιση της απόδοσης τηςμπαταρίας και της διάρκειας ζωής της. Μπορεί επίσης
να προκαλέσει έκλυση θερμότητας ή καπνού απότην μπαταρία, ανάφλεξη ή έκρηξη της μπαταρίας.
Το πακέτο μπαταρίας πρέπει να αποθηκεύεται
σε σταθερό περιβάλλον που χαρακτηρίζεται απόχαμηλή υγρασία (ΣΥ υγρασία χαμηλότερη από70%) και απουσία διαβρωτικών αερίων. Για τηνπρόληψη της σκουριάς, αποφεύγετε συνθήκες πουμπορεί να δημιουργήσουν συμπύκνωση υδρατμών,όπως γρήγορες διακυμάνσεις της θερμοκρασίας τουπεριβάλλοντος.
Για αποθήκευση άνω του ενός μηνός, συνιστάται
αποφορτισμένη ή μερικώς αποφορτισμένη κατάστασηφόρτισης, μεταξύ 50% και 75% (δύο έως τρία LED).
Όταν είναι φορτισμένη, αφαιρέστε την μπαταρία απότον συμπυκνωτή οξυγόνου.

2. Πλύνετε και τα δύο φίλτρα σε ζεστό διάλυμα
σαπουνιού και νερού.
3. Ξεπλύνετε τα φίλτρα τελείως και αφαιρέστε τοπερίσσιο νερό με ένα μαλακό πανί που δεν αφήνει
χνούδι. Βεβαιωθείτε ότι τα φίλτρα είναι τελείως
στεγνά πριν τα επανατοποθετήσετε.
358 - GREPN MN232-C4 K | Εγχειρίδιο χρήστη

FreeStyle ComfortΚΑΘΑΡΙΣΜΟΣΑΠΕΝΕΡΓΟΠΟΙΗΣΤΕ τη μονάδα και αποσυνδέστε
την από την τροφοδοσία AC ή DC πριν απόοποιαδήποτε διαδικασία καθαρισμού. ΜΗΝ ψεκάζετε
απευθείας στο εξωτερικό περίβλημα. Χρησιμοποιήστεένα υγραμένο (όχι μουσκεμένο) πανί ή σφουγγάρι.
Ψεκάστε το πανί ή το σφουγγάρι με ένα διάλυμα ήπιου
οικιακό καθαριστικό για να καθαρίσετε το περίβλημακαι τα τροφοδοτικά. Προχωρήστε σύμφωνα με τιςοδηγίες του κατασκευαστή του καθαριστικού.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΠΕΝΕΡΓΟΠΟΙΗΣΤΕΤΗ ΜΟΝΑΔΑ ΚΑΙ ΑΠΟΣΥΝΔΕΣΤΕΤΟ ΚΑΛΩΔΙΟ ΡΕΥΜΑΤΟΣ ΑΠΟΤΗΝ ΗΛΕΚΤΡΙΚΗ ΠΑΡΟΧΗ ΠΡΙΝΤΟΝ ΚΑΘΑΡΙΣΜΟ ΤΗΣ ΜΟΝΑΔΑΣ
ΠΡΟΚΕΙΜΕΝΟΥ ΝΑ ΑΠΟΦΥΓΕΤΕΤΥΧΟΝ ΗΛΕΚΤΡΟΠΛΗΞΙΑ ΚΑΙ ΚΙΝΔΥΝΟΕΓΚΑΥΜΑΤΩΝ. ΜΟΝΟ Ο ΠΑΡΟΧΟΣ
ΤΟΥ ΕΞΟΠΛΙΣΜΟΥ ΣΑΣ Ή ΕΝΑΣΕΙΔΙΚΕΥΜΕΝΟΣ ΤΕΧΝΙΚΟΣ ΣΕΡΒΙΣ
ΕΠΙΤΡΕΠΕΤΑΙ ΝΑ ΑΦΑΙΡΕΣΕΙ ΤΑΚΑΛΥΜΜΑΤΑ Ή ΝΑ ΣΥΝΤΗΡΗΣΕΙ ΤΗΜΟΝΑΔΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΠΡΕΠΕΙ ΝΑ ΛΑΜΒΑΝΕΤΑΙΜΕΡΙΜΝΑ ΩΣΤΕ ΝΑ ΜΗ ΒΡΑΧΕΙ Ο ΣΥΜΠΥΚΝΩΤΗΣΟΞΥΓΟΝΟΥ ΚΑΙ ΤΟ ΠΑΚΕΤΟ ΜΠΑΤΑΡΙΑΣ Ή ΝΑΜΗ ΔΙΕΙΣΔΥΣΟΥΝ ΥΓΡΑ ΣΤΗ ΜΟΝΑΔΑ. ΚΑΤΙΤΕΤΟΙΟ ΜΠΟΡΕΙ ΝΑ ΠΡΟΚΑΛΕΣΕΙ ΔΥΣΛΕΙΤΟΥΡΓΙΑΉ ΑΠΕΝΕΡΓΟΠΟΙΗΣΗ, ΚΑΘΩΣ ΚΑΙ ΑΥΞΗΜΕΝΟΚΙΝΔΥΝΟ ΗΛΕΚΤΡΟΠΛΗΞΙΑΣ Ή ΕΓΚΑΥΜΑΤΩΝ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΧΡΗΣΙΜΟΠΟΙΕΙΤΕ ΥΓΡΟΑΠΕΥΘΕΙΑΣ ΕΠΑΝΩ ΣΤΗ ΜΟΝΑΔΑ. ΕΝΑΣΚΑΤΑΛΟΓΟΣ ΑΝΕΠΙΘΥΜΗΤΩΝ ΧΗΜΙΚΩΝ ΟΥΣΙΩΝΠΕΡΙΛΑΜΒΑΝΕΙ, ΑΛΛΑ ΟΧΙ ΠΕΡΙΟΡΙΣΤΙΚΑ, ΤΑΑΚΟΛΟΥΘΑ: ΑΛΚΟΟΛΗ ΚΑΙ ΠΡΟΪΟΝΤΑ ΜΕ ΒΑΣΗΑΛΚΟΟΛΗ, ΣΥΜΠΥΚΝΩΜΕΝΑ ΠΡΟΪΟΝΤΑ ΜΕΒΑΣΗ ΧΛΩΡΙΟ (ΧΛΩΡΙΟΥΧΟ ΑΙΘΥΛΕΝΙΟ) ΚΑΙΠΡΟΪΟΝΤΑ ΜΕ ΒΑΣΗ ΛΑΔΙ. ΑΥΤΑ ΔΕΝ ΠΡΕΠΕΙ ΝΑΧΡΗΣΙΜΟΠΟΙΟΥΝΤΑΙ ΓΙΑ ΤΟΝ ΚΑΘΑΡΙΣΜΟ ΤΟΥΠΛΑΣΤΙΚΟΥ ΠΕΡΙΒΛΗΜΑΤΟΣ ΤΟΥ ΣΥΜΠΥΚΝΩΤΗ,ΚΑΘΩΣ ΜΠΟΡΕΙ ΝΑ ΠΡΟΚΑΛΕΣΟΥΝ ΖΗΜΙΑ ΣΤΗΜΟΝΑΔΑ.

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΚΑΘΑΡΙΣΤΕ ΤΟ ΠΕΡΙΒΛΗΜΑ, ΤΟΝΠΙΝΑΚΑ ΕΛΕΓΧΟΥ ΚΑΙ ΤΟ ΚΑΛΩΔΙΟ ΡΕΥΜΑΤΟΣΜΟΝΟ ΜΕ ΕΝΑ ΗΠΙΟ ΟΙΚΙΑΚΟ ΚΑΘΑΡΙΣΤΙΚΟ ΠΟΥΑΠΛΩΝΕΤΕ ΜΕ ΥΓΡΑΜΕΝΟ (ΟΧΙ ΜΟΥΣΚΕΜΕΝΟ)ΠΑΝΙ Ή ΣΦΟΥΓΓΑΡΙ, ΚΑΙ ΣΤΗ ΣΥΝΕΧΕΙΑ
ΣΤΕΓΝΩΣΤΕ ΣΚΟΥΠΙΖΟΝΤΑΣ ΟΛΕΣ ΤΙΣ ΕΠΙΦΑΝΕΙΕΣ.
ΜΗΝ ΕΠΙΤΡΕΠΕΤΕ ΤΗ ΔΙΕΙΣΔΥΣΗ ΟΠΟΙΟΥΔΗΠΟΤΕΥΓΡΟΥ ΣΤΟ ΕΣΩΤΕΡΙΚΟ ΤΟΥ ΣΥΜΠΥΚΝΩΤΗ. ΔΩΣΤΕΙΔΙΑΙΤΕΡΗ ΠΡΟΣΟΧΗ ΣΤΗΝ ΕΞΟΔΟ ΟΞΥΓΟΝΟΥ ΓΙΑΤΗ ΣΥΝΔΕΣΗ ΤΗΣ ΚΑΝΟΥΛΑΣ ΠΡΟΚΕΙΜΕΝΟΥ ΝΑΔΙΑΣΦΑΛΙΣΕΤΕ ΟΤΙ ΠΑΡΑΜΕΝΕΙ ΑΠΑΛΛΑΓΜΕΝΗ ΑΠΟΣΚΟΝΗ, ΝΕΡΟ ΚΑΙ ΣΩΜΑΤΙΔΙΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΧΡΗΣΙΜΟΠΟΙΗΣΤΕ ΕΝΑΥΓΡΑΜΕΝΟ (ΌΧΙ ΜΟΥΣΚΕΜΕΝΟ) ΠΑΝΙ Ή ΣΦΟΥΓΓΑΡΙ
ΓΙΑ ΤΟΝ ΚΑΘΑΡΙΣΜΟ ΤΟΥ ΠΑΚΕΤΟΥ ΜΠΑΤΑΡΙΑΣ.
ΨΕΚΑΣΤΕ ΤΟ ΠΑΝΙ Ή ΤΟ ΣΦΟΥΓΓΑΡΙ ΜΕ ΕΝΑ ΗΠΙΟΟΙΚΙΑΚΟ ΚΑΘΑΡΙΣΤΙΚΟ ΓΙΑ ΝΑ ΚΑΘΑΡΙΣΕΤΕ ΤΟΠΕΡΙΒΛΗΜΑ ΤΟΥ ΠΑΚΕΤΟΥ ΜΠΑΤΑΡΙΑΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΚΑΘΑΡΊΖΕΤΕ ΜΕ ΑΥΤΌΜΑΤΟ
ΣΎΣΤΗΜΑ ΠΛΎΣΗΣ.

ΑΝΤΙΚΑΤΑΣΤΑΣΗ ΤΗΣ ΚΑΝΟΥΛΑΣ

Πάντα να ακολουθείτε τις οδηγίες του κατασκευαστήτης κάνουλας για σωστή χρήση. Αντικαθιστάτε τηρινική κάνουλα ή τη σωλήνωση οξυγόνου κάθε3 έως 6 μήνες (με καθημερινή χρήση, 3 μήνες) όπωςσυνιστάται από τον κατασκευαστή της κάνουλας ήτον πάροχο οξυγόνου σας. Ο γιατρός ή ο πάροχος
οξυγόνου σας θα σας παράσχουν οδηγίες καθαρισμού
και αντικατάστασης.
Επιπλέον αναλώσιμα για αντικατάσταση είναι
διαθέσιμα μέσω του παρόχου οξυγόνου σας.

PN MN232-C4 K | Εγχειρίδιο χρήστη

GRE - 359

FreeStyle Comfort

Καταστάσεις συναγερμούΑρ.
προτεραιότητας

Περιγραφή
συναγερμού

1Δυσλειτουργία
του συστήματος Υψηλή
θερμοκρασία
μπαταρίας

Εμφανιζόμενη ένδειξηΗχητικό σήμαΠερισσότερες λεπτομέρειες /Προτεινόμενη ενέργεια

1 μπιπ κάθε20 δευτερόλεπταΗ μονάδα θα παράγει συναγερμό όταν ηθερμοκρασία της μπαταρίας πλησιάζει τοόριο απενεργοποίησης ασφαλείας.
Εάν επιτευχθεί το όριο απενεργοποίησης,η μονάδα θα απενεργοποιηθεί. Εάν είναι
συνδεδεμένη εξωτερική τροφοδοσία, θαεμφανιστεί η οθόνη συναγερμού και οηχητικός συναγερμός θα συνεχιστεί για2 λεπτά.
Πραγματοποιήστε δοκιμή σεθερμοκρασία δωματίου. Εάν τοπρόβλημα επιμένει, επικοινωνήστε μετον πάροχο.

2Δυσλειτουργία
του συστήματος Υψηλή
θερμοκρασία

1 μπιπ κάθε20 δευτερόλεπταΗ μονάδα θα παράγει συναγερμό ότανη εσωτερική θερμοκρασία πλησιάζει τοόριο απενεργοποίησης.
Εάν επιτευχθεί το όριο απενεργοποίησης,η μονάδα θα απενεργοποιηθεί και θαεμφανιστεί η οθόνη συναγερμού και οηχητικός συναγερμός θα συνεχιστεί για2 λεπτά.
Διασφαλίστε ότι δεν υπάρχουν εμφράξεις
σε κανένα άνοιγμα αερισμού τηςμονάδας. Εάν το πρόβλημα επιμένει,επικοινωνήστε με τον πάροχο.

3Δυσλειτουργία
του συστήματος Υψηλή πίεση

1 μπιπ κάθε20 δευτερόλεπταΕάν επιτευχθεί το όριο απενεργοποίησης,η μονάδα θα απενεργοποιηθεί και θαεμφανιστεί η οθόνη συναγερμού και οηχητικός συναγερμός θα συνεχιστεί για2 λεπτά.
Εάν το πρόβλημα επιμένει,επικοινωνήστε με τον πάροχο.

4Δεν ανιχνεύτηκε
αναπνοή

1 μπιπ κάθε30 δευτερόλεπταΔεν ανιχνεύτηκε αναπνοή για15 δευτερόλεπτα.
Ενεργοποιείται μετά την πρώτη αναπνοή
ή αφού απενεργοποιηθεί η οθόνη
προθέρμανσης, όποιο από τα δύο συμβεί
πρώτο.
Ελέγξτε την κάνουλα για συστροφέςή διαρροές. Αυξήστε τη ρύθμιση
ευαισθησίας. Εάν το πρόβλημα επιμένει,επικοινωνήστε με τον πάροχο.

5Χαμηλή μπαταρία
- απενεργοποίηση

1 μπιπ κάθε30 δευτερόλεπταΟ συναγερμός παράγεται μέχρι τηνεξάντληση της μπαταρίας και την
απενεργοποίηση της μονάδας.
Φορτίστε την μπαταρία.

6

360 - GREΠροθέρμανση

PN MN232-C4 K | Εγχειρίδιο χρήστη

κανέναΗ μονάδα θα ξεκινήσει με αυτόματη
χορήγηση μέχρι να ανιχνευτεί αναπνοήή να λήξει η προθέρμανση, όποιο από ταδύο συμβεί πρώτο.

FreeStyle ComfortΑρ.
προτεραιότητας

Περιγραφή
συναγερμού

7

Εμφανιζόμενη ένδειξηΗχητικό σήμαΠερισσότερες λεπτομέρειες

Χαμηλή
συγκέντρωση

1 μπιπ κάθε30 δευτερόλεπτα,για έως 5 λεπτάΕάν το πρόβλημα επιμένει,επικοινωνήστε με τον πάροχο.

8Δυσλειτουργία
του συστήματος Ρυθμός ροής(βλάβη βαλβίδαςOCD, έμφραξη
διαδρομής αερίου)1 μπιπ κάθε30 δευτερόλεπταΕάν το πρόβλημα επιμένει,επικοινωνήστε με τον πάροχο.

9Προειδοποίηση
χαμηλής μπαταρίας

1 μπιπ, χωρίς
επανάληψηΦορτίστε την μπαταρία.

10Υπέρβαση τουαναπνευστικού
ρυθμού

1 μπιπ κάθε60 δευτερόλεπτα

Ο αναπνευστικός ρυθμός υπερβαίνει
τα 40 bpm (αναπνοές ανά λεπτό).

11Δυσλειτουργία
του συστήματος Βλάβη ανεμιστήρα

κανέναΕάν το πρόβλημα επιμένει,επικοινωνήστε με τον πάροχο.

12Δυσλειτουργία
του συστήματος Χαμηλή πίεση

κανέναΕάν το πρόβλημα επιμένει,επικοινωνήστε με τον πάροχο.

13Δυσλειτουργία
του συστήματος Χαμηλή
θερμοκρασία

κανέναΑφήστε τη μονάδα να προθερμανθεί
σε θερμοκρασία δωματίου. Εάν τοπρόβλημα επιμένει, επικοινωνήστε μετον πάροχο.

Πληροφοριακό

Εκκίνηση1 πληροφοριακό
μπιπ

Η οθόνη εμφανίζεται κατά τηνενεργοποίηση και θα πρέπει
να ακούσετε ένα μπιπ για ναεπιβεβαιωθεί ότι ο βομβητής λειτουργείΕάν η δόση ενεργοποιείται εσφαλμένα,μειώστε τη ρύθμιση ευαισθησίας. Εάντο πρόβλημα επιμένει, επικοινωνήστε
με τον πάροχο.

Η οθόνη εκκίνησης εμφανίζει
την τρέχουσα αναθεώρηση
υλικολογισμικού στην κάτωδεξιά γωνία.

PN MN232-C4 K | Εγχειρίδιο χρήστη

GRE - 361

FreeStyle ComfortΑρ.
προτεραιότητας

Περιγραφή
συναγερμού

Πληροφοριακό

Οθόνη κανονικής
λειτουργίας
σε τρόπο
autoSAT (χωρίς
συνδεδεμένη
τροφοδοσία DC)κανέναΠληροφοριακό

Οθόνη κανονικής
λειτουργίας
με το autoSAT
απενεργοποιημένο
(χωρίς
συνδεδεμένη
τροφοδοσία DC)ΚαμίαΠληροφοριακόΗ ρύθμιση άλλαξε

1 πληροφοριακόμπιπΠληροφοριακό

Σύνδεσηεξωτερικής
τροφοδοσίας στην
απενεργοποιημένη
κατάστασηκανέναΠληροφοριακό

Σύνδεσηεξωτερικής
τροφοδοσίας
στην κατάσταση
λειτουργίας

κανέναΠληροφοριακό

Αποσύνδεση
εξωτερικής
τροφοδοσίας στην
απενεργοποιημένη
κατάσταση1 πληροφοριακόμπιπΠληροφοριακό

Αποσύνδεση
εξωτερικής
τροφοδοσίας
στην κατάσταση
λειτουργίας
(με παρουσία
μπαταρίας)1 πληροφοριακό
μπιπ

362 - GREPN MN232-C4 K | Εγχειρίδιο χρήστη

Εμφανιζόμενη ένδειξηΗχητικό σήμαΠερισσότερες λεπτομέρειες

Εμφάνιση της οθόνης CAIRE για2 δευτερόλεπτα, ακολουθούμενηαπό την κατάσταση μπαταρίας καισύνδεσης για 5 δευτερόλεπτα πριν
απενεργοποιηθεί η οθόνη

Εμφάνιση της οθόνης CAIRE για2 δευτερόλεπτα και στη συνέχεια
απενεργοποίηση της οθόνης.

FreeStyle ComfortΑρ.
προτεραιότητας

Περιγραφή
συναγερμού

Πληροφοριακό

Εμφανιζόμενη ένδειξηΗχητικό σήμαΠερισσότερες λεπτομέρειες

Στην κατάσταση
αναμονής - ότανπατηθεί τοκουμπί Πάνω ήΚάτω, η οθόνη
εμφανίζει τηνκατάσταση
μπαταρίας καιφόρτισηςκανέναΗ οθόνη θα εμφανιστεί για 5 δευτερόλεπτακαι στη συνέχεια θα απενεργοποιηθεί

Πληροφοριακό

Λειτουργία
πτήσης μετροφοδοσία απόμπαταρία

1 μακρύ μπιπΠληροφοριακό

Λειτουργία
πτήσης ενώ είναι
συνδεδεμένο

1 μακρύ μπιπ

PN MN232-C4 K | Εγχειρίδιο χρήστη

GRE - 363

FreeStyle ComfortΟδηγίες πλοήγησης μενού
Μπορείτε να μπείτε στο μενού σε κατάσταση
αναμονής ή λειτουργίας. Σε κατάσταση λειτουργίας,εάν παρουσιαστεί κατάσταση συναγερμού, θα γίνει
αυτόματα έξοδος από το μενού. Εάν δεν πατηθεί
κανένα κουμπί για 20 δευτερόλεπτα, θα γίνει
αυτόματα έξοδος από το μενού. Σε κατάσταση
αναμονής, μπορεί να χρειαστεί να πατήσετε το κουμπί
μενού δύο φορές για να το αφυπνίσετε, στη συνέχεια
μπείτε στο μενού.
Υπάρχουν τέσσερις οθόνες μενού:Airplane Mode (Λειτουργία πτήσης)1. Οθόνη Status (Κατάσταση): Εμφανίζει τον αριθμό
έκδοσης υλικολογισμικού, τις ώρες λειτουργίας,τους κύκλους αποφόρτισης της μπαταρίας
2. Airplane mode (Λειτουργία πτήσης) (ON/OFF Ενεργό/Ανενεργό)3. Ευαισθησία: ρυθμιζόμενη (1-5)
4. autoSAT (ON/OFF - Ενεργό/Ανενεργό)Sensitivty (Ευαισθησία)Παράδειγμα οθόνης Status (Κατάσταση)ΛΕΙΤΟΥΡΓΊΑ ΜΕΝΟΎ ΡΥΘΜΊΣΕΩΝΗ είσοδος στο μενού ρυθμίσεων γίνεται πατώνταςτο κουμπί Μενού/Επιλογή
λειτουργίας ή αναμονής.

σε κατάστασηΓια κύλιση στα μενού, πατήστε τοΣελίδα Status (Κατάσταση)364 - GREPN MN232-C4 K | Εγχειρίδιο χρήστη

.

autoSATFreeStyle ComfortΕΛΙΔΑ ΚΑΤΑΣΤΑΣΗΣΗ σελίδα κατάστασης της ετικέτας Freestyle Comfort εμφανίζει την αναθεώρηση τουυλικολογισμικού, τις ώρες λειτουργίας και τοναριθμό κύκλων μπαταρίας στη συνημμένη
μπαταρία.

Για να ακυρώσετε την ενεργοποίηση της λειτουργίας
πτήσης, πατήστε
ενόσω το εικονίδιο CANCEL(Άκυρο) είναι ορατό.

Επιβεβαιωμένη ενεργή λειτουργία πτήσης

Σελίδα Status (Κατάσταση)Οι ώρες λειτουργίας εμφανίζονται σε ολόκληρεςώρες.
Η εκφόρτιση της μπαταρίας είναι ο αριθμός
κύκλων που έχει αποστραγγιστεί καιεπαναφορτιστεί.

AIRPLANE MODE (ΛΕΙΤΟΥΡΓΊΑ ΠΤΉΣΗΣ)Το FreeStyle Comfort αποστέλλεται με τη λειτουργία
πτήσης απενεργοποιημένη.

Για να απενεργοποιήσετε τη λειτουργία πτήσης,επαναλάβετε τα παραπάνω ενώ αλλάζετε σεOFF (Ανενεργό).
Το πάτημα τουόταν η οθόνη εμφανίζει τοσύμβολο «Next» (Επόμενο) θα μετακινήσει τη σελίδα
μενού στην επόμενη επιλογή οθόνης.
Η λειτουργία πτήσης προορίζεται για χρήση σεκατάσταση λειτουργίας. Η λειτουργία πτήσης θααπενεργοποιηθεί αυτόματα εάν η μονάδα αφεθεί
σε κατάσταση αναμονής με τροφοδοσία από τηνμπαταρία.

Σημείωση: Βλ. την ενότητα «Ενεργοποίηση
λειτουργίας πτήσης» για περισσότερες πληροφορίες
σχετικά με τη λειτουργία πτήσης.

ΛΕΙΤΟΥΡΓΊΑ SENSITIVITY (ΕΥΑΙΣΘΗΣΊΑ)Κύρια οθόνη Airplane Mode (Λειτουργία πτήσης)Για να ενεργοποιήσετε τη λειτουργία, πατήστε
ενόσω το εικονίδιο CHANGE (Αλλαγή) είναι ορατό.
Η τιμή θα αλλάξει στην μπλε ένδειξη «ON» (Ενεργό).

Το FreeStyle Comfort αποστέλλεται με μιαπροεπιλεγμένη ρύθμιση λειτουργίας Sensitivity(Ευαισθησία) 2 από το εργοστάσιο και δεν αναμένεται
να απαιτεί προσαρμογή υπό κανονικές συνθήκες.
Η ρύθμιση ευαισθησίας μπορεί να προσαρμοστεί
ανάμεσα στις τιμές 1-5, όπου 1 είναι η χαμηλότερη
ευαισθησία και 5 είναι η υψηλότερη ευαισθησία.

Οθόνη λειτουργίας Sensitivity (Ευαισθησία) στη ρύθμιση 2Επιλεγμένη ενεργή λειτουργία πτήσης

Για να επιβεβαιώσετε την ενεργοποίηση τηςλειτουργίας πτήσης, πατήστε
ενόσω το εικονίδιο
SAVE (Αποθήκευση) είναι ορατό. Η τιμή θα αλλάξειστη μαύρη ένδειξη «ON» (Ενεργό).

Για να ρυθμίσετε τη λειτουργία Sensitivity(Ευαισθησία), πατήστε
ενόσω το εικονίδιο
CHANGE (Αλλαγή) είναι ορατό. Η τιμή θα αλλάξει
στον επόμενο μπλε αριθμό.
PN MN232-C4 K | Εγχειρίδιο χρήστη

GRE - 365

FreeStyle ComfortΛΕΙΤΟΥΡΓΊΑ AUTOSATΤο FreeStyle Comfort αποστέλλεται με τη λειτουργία
autoSAT ενεργοποιημένη.

Λειτουργία Sensitivity (Ευαισθησία) με επιλεγμένη τη ρύθμιση 3Για να επιβεβαιώσετε τη ρύθμιση 3 στη λειτουργία
Sensitivity (Ευαισθησία), πατήστε
ενόσω τοεικονίδιο SAVE (Αποθήκευση) είναι ορατό. Η τιμή θααλλάξει στη μαύρη ένδειξη «3».
Για να ακυρώσετε την αλλαγή της λειτουργίας
Sensitivity (Ευαισθησία), πατήστε
ενόσω τοεικονίδιο CANCEL (Άκυρο) είναι ορατό.

Λειτουργία Sensitivity (Ευαισθησία) με
επιβεβαιωμένη τη ρύθμιση 3Κύρια οθόνη λειτουργίας autoSATΓια να απενεργοποιήσετε τη λειτουργία, πατήστε
ενόσω το εικονίδιο CHANGE (Αλλαγή) είναι
ορατό. Η τιμή θα αλλάξει στην μπλε ένδειξη «OFF»(Ανενεργό).

Επιλεγμένη ανενεργή λειτουργία autoSATΓια να επιβεβαιώσετε την απενεργοποίηση τηςλειτουργίας autoSAT, πατήστε
ενόσω το εικονίδιο
SAVE (Αποθήκευση) είναι ορατό. Η τιμή θα αλλάξειστη μαύρη ένδειξη «OFF» (Ανενεργό).
Για να ακυρώσετε την ενεργοποίηση της λειτουργίας
autoSAT, πατήστε
(Άκυρο) είναι ορατό.

ενόσω το εικονίδιο CANCEL

Λειτουργία Sensitivity (Ευαισθησία) με
επιβεβαιωμένη τη ρύθμιση 5 (Μέγιστη)Το πάτημα τουόταν η οθόνη εμφανίζει τοσύμβολο «Next» (Επόμενο) θα μετακινήσει τησελίδα μενού στην επόμενη επιλογή οθόνης.

Επιβεβαιωμένη ανενεργή λειτουργία autoSATΓια να ενεργοποιήσετε τη λειτουργία autoSAT,
επαναλάβετε τα παραπάνω ενώ αλλάζετε σεON (Ενεργό).
Το πάτημα τουόταν η οθόνη εμφανίζει τοσύμβολο «Next» (Επόμενο) θα μετακινήσει τησελίδα μενού στην επόμενη επιλογή οθόνης.
366 - GREPN MN232-C4 K | Εγχειρίδιο χρήστη

FreeStyle ComfortΑντιμετώπιση προβλημάτων
ΠρόβλημαΠιθανή αιτία

ΛύσηΤο FreeStyle Comfort εμφανίζει
συναγερμό και παράγει
διαλείπον ηχητικό σήμα.

Ανατρέξτε στον πίνακα
«Καταστάσεις συναγερμού».

Ανατρέξτε στον πίνακα «Καταστάσεις
συναγερμού».

Ο συμπυκνωτής οξυγόνου δενενεργοποιείται.

Η μπαταρία είναι
αποφορτισμένη.

Τροφοδοτήστε τη μονάδα μέσω παροχής AC ή DC.

Δεν έχει συνδεθεί σε εξωτερική
τροφοδοσία.
Γενική δυσλειτουργία.

Διασφαλίστε ότι οι εξωτερικές συνδέσεις είναι
ασφαλείς.
Πραγματοποιήστε πλήρη κύκλο απενεργοποίησηςεπανενεργοποίησης αφαιρώντας την μπαταρία
και αποσυνδέοντας οποιαδήποτε εξωτερική
πηγή τροφοδοσίας AC ή DC. Στη συνέχεια,
απενεγκαταστήστε την μπαταρία και ο κύκλος
απενεργοποίησης-επανενεργοποίησης έχειολοκληρωθεί. Επανασυνδέστε εξωτερική πηγήτροφοδοσίας εάν είναι επιθυμητό.
Επικοινωνήστε με τον πάροχο του εξοπλισμού σαςκαι αλλάξτε σε μια άλλη πηγή οξυγόνου όπως είναι
απαραίτητο.

Όλα τα άλλα προβλήματα.

Πραγματοποιήστε πλήρη κύκλο απενεργοποίησηςεπανενεργοποίησης αφαιρώντας την μπαταρία
και αποσυνδέοντας οποιαδήποτε εξωτερική
πηγή τροφοδοσίας AC ή DC. Στη συνέχεια,
απενεγκαταστήστε την μπαταρία και ο κύκλος
απενεργοποίησης-επανενεργοποίησης έχειολοκληρωθεί. Επανασυνδέστε εξωτερική πηγήτροφοδοσίας εάν είναι επιθυμητό.
Αλλάξτε σε μια άλλη πηγή οξυγόνου που υπάρχει
διαθέσιμη και επικοινωνήστε με τον πάροχο τουεξοπλισμού σας.

Σημείωση: Ορισμένοι συναγερμοί θα προκαλέσουν απενεργοποίηση της μονάδας.
Σημείωση: Κατά τη διάρκεια πλήρους διακοπής της τροφοδοσίας, γίνεται επαναφορά του συστήματος συναγερμών (απαλοιφή οποιωνδήποτε προηγούμενων συναγερμών) και θα εμφανιστεί συναγερμός μόνο εάν η κατάσταση εξακολουθεί να υπάρχει όταν αποκατασταθεί η τροφοδοσία.

PN MN232-C4 K | Εγχειρίδιο χρήστη

GRE - 367

FreeStyle ComfortΤαξιδιωτικές πληροφορίες
Το FreeStyle Comfort πληροί τις κατευθυντήριες
οδηγίες της FAA για χρήση επί εμπορικού
αεροσκάφους. Πριν από οποιοδήποτε ταξίδι
(με αεροπλάνο, τρένο, πλοίο κ.λπ.) πρέπει να:• Διασφαλίσετε ότι ο συμπυκνωτής είναι σε καλήκατάσταση λειτουργίας.
• Επικοινωνήστε με την εταιρεία με την οποία θαταξιδέψετε τουλάχιστον 48 ώρες πριν από τηναναχώρηση για να ενημερώσετε ότι θα ταξιδέψετε
με ένα FreeStyle Comfort.
Συνιστάται να έχετε πάντα μαζί σας επιπλέον
μπαταρίες στα ταξίδια. Για αεροπορικά ταξίδια,προβλέψτε αρκετή διάρκεια ζωής της μπαταρίας
για τον χρόνο στο έδαφος (check in πριν την πτήση,έλεγχο ασφάλειας, κ.λπ.) και ενδιάμεσες στάσεις.
Οι περισσότερες αεροπορικές εταιρείες απαιτούν ναέχετε επαρκή αριθμό πλήρως φορτισμένων πακέτων
μπαταρίας για την τροφοδοσία της συσκευής γιατουλάχιστον 150% της μέγιστης αναμενόμενηςδιάρκειας του ταξιδιού (χρόνοι πτήσης, χρόνος στοέδαφος πριν και μετά την πτήση, ανταποκρίσεις καιαπρόβλεπτες καθυστερήσεις).

Σημείωση: Εάν ταξιδεύετε αεροπορικώς, συνιστάται
να μεταφέρετε τις επιπλέον μπαταρίες σε μια ασφαλή
τσάντα μεταφοράς.

ΚΑΝΟΝΙΣΜΟΣ FAA

Οι εφεδρικές μπαταρίες πρέπει να προστατεύονταιατομικά ώστε να αποφεύγονται τα βραχυκυκλώματα,τοποθετώντας τις στην αρχική συσκευασία πώλησης
ή μονώνοντας τους ακροδέκτες με άλλον τρόπο,π.χ. καλύπτοντας με ταινία τους εκτεθειμένους
ακροδέκτες ή τοποθετώντας κάθε μπαταρία σε έναξεχωριστό πλαστικό σακουλάκι ή προστατευτική θήκη,και να μεταφέρονται μόνο στις χειραποσκευές.

αεροσκάφους:1. Ενεργοποιήστε τον POC χρησιμοποιώντας
τροφοδοσία από μπαταρία ΜΟΝΟ, όταν δεν είναι
συνδεδεμένη εξωτερική πηγή τροφοδοσίας.

2. Ενεργοποιήστε τη λειτουργία πτήσης
χρησιμοποιώντας μία από τις ακόλουθες μεθόδους:α. Ακολουθήστε τις οδηγίες για την ενεργοποίησητης λειτουργίας πτήσης στην ενότητα «Οδηγίες
πλοήγησης μενού».

β. Πατήστε και κρατήστε πατημένα και τα δύοκουμπιά Πάνω και Κάτω για 5 δευτερόλεπταμέχρι να ακούσετε ένα μακρύ ηχητικό σήμα και ναεμφανιστεί το σύμβολο αεροσκάφους.

ΕΝΕΡΓΟΠΟΙΗΣΗ ΛΕΙΤΟΥΡΓΙΑΣ ΠΤΗΣΗΣΤροφοδοσία από μπαταρία
Μπορεί να χρησιμοποιηθεί POC (φορητός
συμπυκνωτής οξυγόνου) σε αεροσκάφος σύμφωνα μετις κανονικές διαδικασίες λειτουργίας χρησιμοποιώντας
τροφοδοσία από μπαταρία ΜΟΝΟ, όταν δεν είναι
συνδεδεμένη εξωτερική πηγή τροφοδοσίας
Εξωτερική τροφοδοσίαΕπί του αεροσκάφους, όταν είναι διαθέσιμη πρίζα
ηλεκτρικού ρεύματος, και η θέση της είναι κατάλληληγια χρήση ως πηγή τροφοδοσίας του POC, μπορεί ναχρησιμεύσει ως εφεδρική στις μπαταρίες του POC.
Στην περίπτωση αυτή, πρέπει να τηρηθεί η ακόλουθησυνιστώμενη διαδικασία όσον αφορά τη μετάβαση απότην μπαταρία στην παροχή ηλεκτρικού ρεύματος του

368 - GREPN MN232-C4 K | Εγχειρίδιο χρήστη

3. Όταν εμφανιστεί το σύμβολο αεροσκάφους,συνδέστε τον POC.

Σημείωση: Λόγω περιορισμών των αεροσκαφών, οι
μπαταρίες δεν φορτίζονται σε λειτουργία πτήσης.

FreeStyle Comfort
Παρελκόμενα

ΤΟΠΟΘΕΤΗΣΗ ΤΗΣ ΜΟΝΑΔΑΣ ΣΤΗΝΤΣΑΝΤΑ ΜΕΤΑΦΟΡΑΣ

Για σωστή απόδοση και ασφάλεια, χρησιμοποιείτε
μόνο αυτά τα παρατιθέμενα παρελκόμενα πουπαρέχονται από την CAIRE μέσω του παρόχου
οξυγόνου σας. Η χρήση παρελκομένων που δενπαρατίθενται παρακάτω θα μπορούσε να επηρεάσει
αρνητικά την απόδοση ή/και ασφάλεια τουσυμπυκνωτή.

1. Τοποθετήστε τη μονάδα στο κάτω μέρος τηςτσάντας, ευθυγραμμίζοντας σωστά τα φίλτρα καιτα ανοίγματα μπαταρίας.

Τυπικά παρελκόμενα του FreeStyle ComfortΤσάντα μεταφοράς

MI396-1Σακίδιο πλάτης

MI459-1Ιμάντας ώμουMI397-1Τροφοδοτικό AC (συμπεριλαμβάνονταιόλα τα καλώδια)PW036-1SΚαλώδιο τροφοδοσίας DC

CD041-1Εξωτερικό φίλτρο χονδρών
σωματιδίων (δεξιά)FI226-1Εξωτερικό φίλτρο χονδρών
σωματιδίων (αριστερά)FI227-1

2. Κλείστε καλά το φερμουάρ γύρω από το κάτω μέροςτης θήκης.

Διαθέσιμα προαιρετικά εξαρτήματα
Πρόσθετο πακέτο μπαταρίας
8 στοιχείων

BT034-1SΠακέτο μπαταρίας 16 στοιχείων

BT035-1SΕπιτραπέζιος φορτιστής πακέτου
μπαταρίας

BT036-1Τσάντα παρελκομένων «carry-all»

MI372-2Σημείωση: Μπορεί να είναι διαθέσιμες πρόσθετες
επιλογές για καλώδια τροφοδοσίας ειδικά για τηχώρα στις περιπτώσεις που αναφέρονται παραπάνω.
Επικοινωνήστε με την CAIRE ή με τον πάροχο
οξυγόνου σας εάν χρειάζεται να παραγγείλετε
εναλλακτικές επιλογές.

ΤΟΠΟΘΕΤΗΣΗ ΤΟΥ ΙΜΑΝΤΑ ΩΜΟΥ
1. Ο ιμάντας ώμου μπορεί να προσαρτηθεί είτεαπευθείας στη μονάδα είτε στην τσάντα μεταφοράς.
2. Συνδέστε τον ιμάντα με το κλιπ στον δακτύλιο πουβρίσκεται στη μονάδα ή στην τσάντα.

Οδηγίες τσάντας μεταφοράς
Για να τοποθετήσετε και να ασφαλίσετε το FreeStyle
Comfort στην τσάντα μεταφοράς, ακολουθήστε τιςπαρακάτω οδηγίες.

PN MN232-C4 K | Εγχειρίδιο χρήστη

GRE - 369

FreeStyle ComfortΑπόρριψη και επαναχρησιμοποίηση
Πάντοτε να επιστρέφετε το FreeStyle Comfort,
συμπεριλαμβανομένων όλων των εξαρτημάτων, στονπάροχο κατ' οίκον εξοπλισμός για σωστή απόρριψη.
Μπορείτε επίσης να επικοινωνήσετε με τα τοπικά
γραφεία της πόλης σας για οδηγίες σχετικά με τησωστή απόρριψη της μπαταρίας. Η μονάδα μπορεί ναεπαναχρησιμοποιηθεί μεταξύ ανθρώπων, αλλά πρέπει
να επιστραφεί στον προμηθευτή εξοπλισμού.

ΑΗΗΕ και RoHSΑυτό το σύμβολο υπενθυμίζει στους κατόχους
του εξοπλισμού να τον επιστρέψουν σε εγκαταστάσεις
ανακύκλωσης στο τέλος της ωφέλιμης ζωής του,σύμφωνα με την Οδηγία για τα Απόβλητα Ηλεκτρικού
και Ηλεκτρονικού Εξοπλισμού (ΑΗΗΕ).
Τα προϊόντα μας συμμορφώνονται με την οδηγία γιατον περιορισμό των επικίνδυνων ουσιών (RoHS).
Δεν περιέχουν περισσότερο από ιχνοποσότητες
μολύβδου ή άλλων επικίνδυνων υλικών.
ΠΡΟΣΟΧΗ: Για σωστή απόρριψη,επικοινωνήστε με τον πάροχο τουεξοπλισμού σας ή με την αρμόδια
υπηρεσία τοπικής αυτοδιοίκησης γιαοδηγίες απόρριψης

Υλικά σε άμεση επαφή με τον χειριστή
Περίβλημα συμπυκνωτή.............................................................................................................................. PC/ABSΠίνακας ελέγχου συμπυκνωτή............................................................................. Πολυεστέρας EBG7 ή ισοδύναμο
Λαβή συμπυκνωτή........................................................................................................................... Πολυανθρακικό
Έξοδος οξυγόνου, ρύγχος................................................................................................................ Πολυανθρακικό
Ετικέτα μονάδας...............................................................................................................................................LexanΣύνδεσμοι καλωδίου......................................................................................... Πολυανθρακικό/χλωριούχο βινύλιο
Καλώδιο ρεύματος..............................................................................................................................PVC, μέταλλοΤροφοδοτικό................................................................................................................ Lexan 940 (πολυανθρακικό)Πακέτο μπαταρίας........................................................................................................................................ PC/ABSΕτικέτες πακέτου μπαταρίας, τροφοδοτικού......................................................................Πολυεστερική μεμβράνη
Θήκη μεταφοράς συμπυκνωτή............................................................... Cordura, νεοπρένιο, επένδυση από νάιλον
Ιμάντας............................................................................ Πολυπροπυλένιο, ακετύλιο, βινύλιο, Cordura, νεοπρένιο

Δοκιμές ηλεκτρομαγνητικές συμβατότητας (EMC)Οδηγία και δήλωση του κατασκευαστή – ηλεκτρομαγνητικές εκπομπές
Το FreeStyle Comfort προορίζεται για χρήση στο ηλεκτρομαγνητικό περιβάλλον που προσδιορίζεται παρακάτω. Οαγοραστής ή ο χρήστης του FreeStyle Comfort θα πρέπει να διασφαλίσει ότι το σύστημα χρησιμοποιείται σε ανάλογο
περιβάλλον.
Δοκιμή εκπομπών
Συμμόρφωση
Ηλεκτρομαγνητικό περιβάλλον – οδηγίες
Εκπομπές
ραδιοσυχνοτήτων (RF)CISPR 11Εκπομπές
ραδιοσυχνοτήτων (RF)CISPR 11Εκπομπές αρμονικών
IEC 61000‑3-2Διακυμάνσεις τάσης/εκπομπές αναλαμπής
IEC 61000-3-3
370 - GREΟμάδα 1Κατηγορία B
Κατηγορία ΑΣυμμορφώνεταιPN MN232-C4 K | Εγχειρίδιο χρήστη

Το FreeStyle Comfort χρησιμοποιεί ενέργεια ραδιοσυχνοτήτων(RF) μόνο για την εσωτερική λειτουργία του. Συνεπώς, οι εκπομπές
ραδιοσυχνοτήτων της συσκευής είναι πολύ χαμηλές και δεν είναι
πιθανό να προκαλέσουν παρεμβολή σε κοντινό ηλεκτρονικό εξοπλισμό.
Το FreeStyle Comfort είναι κατάλληλο για χρήση σε όλους τους χώρους,συμπεριλαμβανομένων των οικιακών εγκαταστάσεων και εκείνων πουείναι απευθείας συνδεδεμένοι στο δημόσιο δίκτυο παροχής χαμηλής
τάσης, που τροφοδοτεί κτίρια που χρησιμοποιούνται για οικιακούς
σκοπούς.

FreeStyle ComfortΟδηγίες και δήλωση κατασκευαστή – ηλεκτρομαγνητική ατρωσία

Το FreeStyle Comfort προορίζεται για χρήση στο ηλεκτρομαγνητικό περιβάλλον που προσδιορίζεται παρακάτω. Οαγοραστής ή ο χρήστης του FreeStyle Comfort θα πρέπει να διασφαλίσει ότι το σύστημα χρησιμοποιείται σε ανάλογο
περιβάλλον.
Δοκιμή ατρωσίας
Επίπεδο δοκιμής IECΕπίπεδοΗλεκτρομαγνητικό περιβάλλον –60601συμμόρφωσης
οδηγίεςΤα δάπεδα θα πρέπει να είναι
±8 kV επαφή
Ηλεκτρομαγνητικό
κατασκευασμένα από ξύλο, τσιμέντο ή±8 kV επαφή
περιβάλλον – οδηγίες
κεραμικά πλακίδια. Εάν τα δάπεδα είναι
±15 kV εκφόρτιση
±15 kV εκφόρτιση στον αέραεπιστρωμένα με συνθετικό υλικό, η σχετική
στον αέρα
IEC 61000-4-2
υγρασία πρέπει να είναι τουλάχιστον 30%.
Η ποιότητα των κυρίων γραμμών
±2 kV για γραμμές
±2 kV για γραμμές
τροφοδοσίας πρέπει να είναι ισοδύναμη
τροφοδοσίας ρεύματοςτροφοδοσίας
Ταχεία ηλεκτρική
με εκείνη μιας τυπικής επαγγελματικής
ρεύματος
πάροδος/ριπήεγκατάστασης υγειονομικής περίθαλψης
IEC 61000-4-4±1 kV για γραμμές εισόδου/και ενός τυπικού περιβάλλοντος
εξόδου
Δεν ισχύει
περίθαλψης κατ' οίκον.
±1 kV γραμμή(ές) προςγραμμή(ές)ΥπέρτασηIEC 61000-4-5±2 kV τρόπος λειτουργίας
±1 kV γραμμή(ές)κοινού σήματος στις γραμμές προς γραμμή(ές)
AC±1 kV διαφορικός τρόπος
λειτουργίας στις γραμμές AC±2 kV τρόπος λειτουργίας
κοινού σήματος στις γραμμές
I/O (εισόδου/εξόδου)Πτώσεις τάσης,σύντομες διακοπές
και διακυμάνσεις
τάσης στις
γραμμές παροχής
τροφοδοσίας
εισόδου IEC 610004-11

0% UT για 0,5 κύκλο
(0°, 45°, 90°, 135°,
0% UT για 0,5 κύκλο (0°, 45°,
180°, 225°, 270°,90°, 135°, 180°, 225°, 270°,
315°)
315°)
0% UT για 1 κύκλο (00)
70% UT (30% βύθιση στηνUT) για 25/30 κύκλους (00)
0% UT για 250/300 κύκλους
(00)

0% UT για 1 κύκλο
(00)
70% UT (30% βύθιση
στην UT) για 25/30
κύκλους (00)
0% UT για 250/300κύκλους (00)Η ποιότητα των κυρίων γραμμών
τροφοδοσίας πρέπει να είναι ισοδύναμη
με εκείνη μιας τυπικής επαγγελματικής
εγκατάστασης υγειονομικής περίθαλψης
και ενός τυπικού περιβάλλοντος
περίθαλψης κατ' οίκον.

Η ποιότητα των κυρίων γραμμών
τροφοδοσίας πρέπει να είναι ισοδύναμη
με εκείνη μιας τυπικής επαγγελματικής
εγκατάστασης υγειονομικής περίθαλψης
και ενός τυπικού περιβάλλοντος
περίθαλψης κατ' οίκον. Εάν ο χρήστης
του FreeStyle Comfort απαιτεί συνεχή
λειτουργία κατά τη διάρκεια διακοπών
ρεύματος, συνιστάται η τροφοδοσία
του FreeStyle Comfort να γίνεται απότροφοδοτικό αδιάλειπτης λειτουργίας ήμπαταρία.

Τα μαγνητικά πεδία συχνότητας ρεύματοςθα πρέπει να βρίσκονται σε επίπεδα
χαρακτηριστικά μιας τυπικής τοποθεσίας
3A/m
3A/m
σε μια επαγγελματική εγκατάστασηυγειονομικής περίθαλψης και ένα τυπικό
περιβάλλον περίθαλψης κατ' οίκον.
ΣΗΜΕΙΩΣΗ UT είναι η τάση κύριας παροχής τάσης δικτύου (AC) πριν την εφαρμογή του επιπέδου ελέγχου.

Μαγνητικό πεδίο
συχνότητας ισχύος
(50/60 Hz) IEC
61000-4-8ΣΗΜΕΙΩΣΗ Σε ορισμένες περιπτώσεις, το Freestyle Comfort μπορεί να κάνει επαναφορά μετά από σημαντική
ηλεκτροστατική εκφόρτιση και μπορεί να απαιτεί πλήρη κύκλο απενεργοποίησης-επανενεργοποίησης για να συνεχίσει
την κανονική λειτουργία. Βλ. την ενότητα «Αντιμετώπιση προβλημάτων» για περισσότερες λεπτομέρειες.

PN MN232-C4 K | Εγχειρίδιο χρήστη

GRE - 371

FreeStyle ComfortΟδηγίες και δήλωση κατασκευαστή – ηλεκτρομαγνητική ατρωσία

Το FreeStyle Comfort προορίζεται για χρήση στο ηλεκτρομαγνητικό περιβάλλον που προσδιορίζεται παρακάτω. Οαγοραστής ή ο χρήστης του FreeStyle Comfort θα πρέπει να διασφαλίσει ότι το σύστημα χρησιμοποιείται σε ανάλογο
περιβάλλον.
Δοκιμή ατρωσίας Επίπεδο δοκιμής IECΕπίπεδο συμμόρφωσης Ηλεκτρομαγνητικό περιβάλλον – οδηγίες
60601
Αγόμενες
10 Vrms
10 VrmsΟ φορητός και κινητός εξοπλισμός
ραδιοσυχνότητες 150 kHz έως 80 MHzεπικοινωνιών με ραδιοσυχνότητες δενπρέπει να χρησιμοποιείται πλησιέστερα
IEC 61000-4-6σε οποιοδήποτε εξάρτημα του FreeStyle
Comfort, συμπεριλαμβανομένων τωνκαλωδίων, από τη συνιστώμενη απόσταση10 V/m
10 V/mδιαχωρισμού που υπολογίζεται από τηνΑκτινοβολούμενες 80 MHz έως 2700 MHzεξίσωση που εφαρμόζεται στη συχνότητα
ραδιοσυχνότητες
του πομπού.
IEC 61000-4-3Συνιστώμενη απόσταση διαχωρισμού

d = 1,2√ P

d = 1,2√ P = 80 MHz έως 800 MHz
d = 1,2√ P = 800 MHz έως 2700 MHzόπου P είναι η μέγιστη τιμή ισχύος εξόδου
του πομπού σε watt (W) σύμφωνα με τονκατασκευαστή του πομπού και d είναι ησυνιστώμενη απόσταση διαχωρισμού σεμέτρα (m).
Η ισχύς πεδίου από σταθερούς πομπούς
ραδιοσυχνοτήτων, όπως καθορίζεται απόμία ηλεκτρομαγνητική μελέτη του χώρου,απρέπει να είναι χαμηλότερη από το επίπεδο
συμμόρφωσης σε κάθε εύρος συχνοτήτων.βΕνδέχεται να εμφανιστεί παρεμβολή στοπεριβάλλον εξοπλισμού που επισημαίνεταιμε το ακόλουθο σύμβολο:ΣΗΜΕΙΩΣΗ 1 Στα 80 MHz και τα 800 MHz, ισχύει το υψηλότερο εύρος συχνοτήτων.
ΣΗΜΕΙΩΣΗ 2 Αυτές οι οδηγίες μπορεί να μην ισχύουν σε όλες τις καταστάσεις. Η διάδοση των ηλεκτρομαγνητικών
κυμάτων επηρεάζεται από την απορρόφηση και την αντανάκλαση από δομές, αντικείμενα και ανθρώπους.

Η ισχύς πεδίου από σταθερούς πομπούς, όπως σταθμοί βάσης για ραδιοτηλέφωνα (κινητά/ασύρματα) και επίγεια κινητά
ραδιόφωνα, ερασιτεχνικές ραδιοφωνικές εκπομπές, ραδιοφωνικές αναμεταδόσεις AM και FM, καθώς και τηλεοπτικές
αναμεταδόσεις, δεν μπορεί να προβλεφθεί θεωρητικά με ακρίβεια. Για να αξιολογήσετε το ηλεκτρομαγνητικό περιβάλλον
λόγω των σταθερών πομπών ραδιοσυχνοτήτων, θα πρέπει να εξετάζεται το ενδεχόμενο διερεύνησης της θέσης. Εάνη μετρημένη ισχύς πεδίου στην τοποθεσία όπου χρησιμοποιείται το FreeStyle Comfort υπερβαίνει το παραπάνω
εφαρμοζόμενο επίπεδο συμμόρφωσης ραδιοσυχνοτήτων, το FreeStyle Comfort πρέπει να τεθεί υπό παρακολούθηση μεσκοπό την επαλήθευση της κανονικής λειτουργίας. Εάν παρατηρηθεί μη κανονική απόδοση, ενδέχεται να απαιτούνται
πρόσθετα μέτρα, όπως η αλλαγή προσανατολισμού ή θέσης του FreeStyle Comfort.

αβΣτο εύρος συχνοτήτων 150 kHz έως 80 MHz, η ισχύς πεδίου πρέπει να είναι μικρότερη από 10 V/m.

372 - GREPN MN232-C4 K | Εγχειρίδιο χρήστη

FreeStyle ComfortΣυνιστώμενες αποστάσεις διαχωρισμού μεταξύ φορητού και κινητού εξοπλισμού
επικοινωνιών με ραδιοσυχνότητες και του FreeStyle Comfort

Το FreeStyle Comfort προορίζεται για χρήση σε ηλεκτρομαγνητικό περιβάλλον στο οποίο οι διαταραχές από εκπομπές
ραδιοσυχνοτήτων είναι ελεγχόμενες. Ο αγοραστής ή ο χρήστης του FreeStyle Comfort μπορεί να βοηθήσει στην πρόληψη
των ηλεκτρομαγνητικών παρεμβολών, διατηρώντας μια ελάχιστη απόσταση μεταξύ του φορητού και κινητού εξοπλισμού
επικοινωνιών με ραδιοσυχνότητες (πομπών) και του FreeStyle Comfort, όπως συνιστάται παρακάτω, σύμφωνα με τημέγιστη ισχύ εξόδου του εξοπλισμού επικοινωνιών.
Απόσταση διαχωρισμού ανάλογα με τη συχνότητα του πομπού
Ονομαστική τιμήμέγιστης ισχύος
mεξόδου του πομπού
150 kHz έως 80 MHz
80 MHz έως 800 MHz
800 MHz έως 2,5 GHz

d = 1,2√ P
d = 1,2√ P
d = 1,2√ P
W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Για πομπούς με ονομαστική μέγιστη ισχύ εξόδου που δεν παρατίθεται παραπάνω, η συνιστώμενη απόστασηδιαχωρισμού d σε μέτρα (m) μπορεί να προσδιοριστεί με χρήση της εξίσωσης που εφαρμόζεται στη συχνότητα τουπομπού, όπου P είναι η μέγιστη τιμή ισχύος εξόδου του πομπού σε watt (W), σύμφωνα με τον κατασκευαστή του πομπού.
ΣΗΜΕΙΩΣΗ 1 Στα 80 MHz και 800 MHz, ισχύει η απόσταση διαχωρισμού για το υψηλότερο εύρος συχνοτήτων.
ΣΗΜΕΙΩΣΗ 2 Αυτές οι οδηγίες μπορεί να μην ισχύουν σε όλες τις καταστάσεις. Η διάδοση των ηλεκτρομαγνητικών
κυμάτων επηρεάζεται από την απορρόφηση και την αντανάκλαση από δομές, αντικείμενα και ανθρώπους.

Ταξινόμηση

Τύπος προστασίας από ηλεκτροπληξία:Κατηγορία II	Η προστασία από ηλεκτροπληξία
επιτυγχάνεται από τη διπλή μόνωση.
Βαθμός προστασίας από ηλεκτροπληξία:Τύπος BF Εξοπλισμός που παρέχει ένανσυγκεκριμένο βαθμό προστασίας από ηλεκτροπληξία
όσον αφορά
		 1) το επιτρεπόμενο ρεύμα διαρροής,		2) την αξιοπιστία της σύνδεσης προστατευτικής
γείωσης (αν υπάρχει).
		 Δεν προορίζεται για άμεση καρδιακή εφαρμογή.
Ανεξάρτητες δοκιμές για το πρότυπο για ιατρικές
ηλεκτρικές συσκευές:Δοκιμάστηκε για συμμόρφωση με• IEC 60601-1 έκδοση 3.1: Ιατρικές ηλεκτρικές
συσκευές – Μέρος 1 Γενικές απαιτήσεις για τηνασφάλεια.
• CAN/CSA C22.2 Αρ. 60601-1-8 έκδοση 2.1,Ιατρικές ηλεκτρικές συσκευές – Μέρος 1:Γενικές απαιτήσεις για την ασφάλεια
Προστασία έναντι πιθανών ηλεκτρομαγνητικών ήάλλων παρεμβολή μεταξύ του εξοπλισμού και άλλων
συσκευών.
	• Δοκιμάστηκε για συμμόρφωση με EN 60601-1-2/
IEC 60601-1-2: 4η έκδοση.

	• Δοκιμάστηκε για συμμόρφωση με το πρότυπο
RTCA/DO160 Ενότητα 21 Κατηγορία M.
	• CISPR 11 / EN 55011 Κατηγορία B Ομάδα 1,«Βιομηχανικές, επιστημονικές, και ιατρικές (ISM)συσκευές»	• FCC Μέρος 15, Τμήμα B – Κατηγορία BΑκούσιοι ακτινοβολητές
IP22 – Προστασία κατά της εισόδου δακτύλων ήπαρόμοιων αντικειμένων με διάμετρο μεγαλύτερη
από 12,5 mm. Η κάθετη πτώση σταγόνων νερού δενθα έχει επιβλαβείς επιπτώσεις όταν το περίβλημαβρίσκεται υπό γωνία κλίσης 15° από την κανονική
του θέση.
Επιτρεπόμενη μέθοδος καθαρισμού και ελέγχου τωνλοιμώξεων: Παρακαλώ ανατρέξτε στην ενότητα
«Καθαρισμός, φροντίδα και συντήρηση ρουτίνας»του παρόντος εγχειριδίου χρήστη του FreeStyle.
Βαθμός ασφάλειας της εφαρμογής υπό την παρουσία
εύφλεκτων αναισθητικών αερίων:Εξοπλισμός που δεν είναι κατάλληλος για τέτοια
εφαρμογή.
Τρόπος λειτουργίας: Συνεχής λειτουργία.
Η CAIRE και η CAIRE Inc. είναι σήματα κατατεθέντα της CAIRE Inc. Επισκεφθείτε τηνιστοσελίδα μας παρακάτω για μια πλήρη λίστα εμπορικών σημάτων.
Εμπορικά σήματα: www.careinc.com/corporate/trademarks.
Copyright © 2021 CAIRE Inc. Η CAIRE Inc. διατηρεί το δικαίωμα να διακόψει τα προϊόντα
της, ή να αλλάξει τις τιμές, τα υλικά, τον εξοπλισμό, την ποιότητα, τις περιγραφές,τις προδιαγραφές ή/και τις διαδικασίες για τα προϊόντα της ανά πάσα στιγμή χωρίς
προηγούμενη ειδοποίηση και χωρίς περαιτέρω υποχρέωση ή συνέπεια. Όλα ταδικαιώματα που δεν αναφέρονται ρητά στο παρόν διατηρούνται από την εταιρεία μας,ανάλογα με την περίπτωση.

PN MN232-C4 K | Εγχειρίδιο χρήστη

GRE - 373

FreeStyle Comfort

Kullanıcı Kontrolleri ve Sistem Durumu Göstergeleri
ISO 7000; Ekipmanda kullanılan grafik sembolleri - Dizinve özetTıbbi cihazlara ilişkin Konsey Direktifi 93/42/EEC
Avrupa Topluluğundaki yetkili temsilciÜrüne özgü cihaz tanımlayıcı (UDI)
etiketi üzerinde CE #### sembolü varsa,
cihaz tıbbi cihazlarla ilgili 93/42 / EEC
Direktifinin gerekliliklerine uygundur.
CE #### sembolü onaylanmış kuruluş
numarasını gösterir.

Çalıştırmadan önce kullanıcı el kitabını
okuyun. 1641 sayılı yönetmelik
Depolama veya çalışma sıcaklığı
aralığı.. 0632 sayılı yönetmelik
IEC 60417
Depolama nem aralığı.
2620 sayılı yönetmelik

Sınıf II Ekipman, İki Kat Yalıtımlı
5172 sayılı yönetmelik

Atmosfer basıncı sınırlaması.
2621 sayılı yönetmelik

Güç anahtarı. 5009 sayılı yönetmelikÜreticinin adı ve adresi.
3082 sayılı yönetmelik

ETL Sertifikası
Tıbbi elektrikli ekipmana yönelik
CAN/CSA C22.2 No. 601.1 M90 için
güvenlik ajansıÜretim tarihi. 2497 sayılı yönetmelik
Katalog Numarası.
2493 sayılı yönetmelik

Dahili Semboller

Seri Numarası. 2498 sayılı yönetmelik

Yanıcı malzemeler, yağ ve gresten uzak
tutun.

Üst kısmıdır. 0623 sayılı yönetmelikParçalarına ayırmayın.

Dikkat, kırılır. 0621 sayılı yönetmelikYağmurdan koruyun, kuru tutun.
0626 sayılı yönetmelikAkışı artırın.

Belirtilen sayının üzerinde istiflemeyin.
2403 sayılı yönetmelikAkışı azaltın.

ISO 7010: Grafik sembolleri - Güvenlik renkleri vegüvenlik işaretleri - Kayıtlı güvenlik işaretleri
Kullanıcı el kitabı okunmalıdır.
M002 sayılı yönetmelik
Açık alev, yangın ve kıvılcımdan uzak
tutun. Yakınında açık alev kaynağı
bulundurmak ve sigara içmek yasaktır.
P003 sayılı yönetmelikÜnitenin yakınındayken veya üniteyiçalıştırırken sigara içmeyin.
P002 sayılı yönetmelik
BF Tipi uygulamalı parça (elektrikçarpmasına karşı koruma derecesi).
5333 sayılı yönetmelik
Uyarı. W001 sayılı yönetmelik

374 - TURPN MN232-C4 K | Kullanıcı El Kitab

Menü düğmesi.
RTCA/DO160 Bölüm
21
M Kategorisi
ile Uyumlu

RTCA DO160 Bölüm 21 M Kategorisi
ile Uyumlu. FAA SFAR 106
gereksinimi
Sınıf II Ekipman, İki Kat Yalıtımlı

FAA Onaylı - POC
21 CFR 801.15: Federal Düzenlemeler Kanunu Başlık
21
Federal yasalar uyarınca bu cihazyalnızca bir hekim tarafından veya
hekimin talimatıyla satılabilir.

FreeStyle Comfort
Konsey Direktifi 2012/19/EU: atık elektrikli veelektronik ekipmanlar (WEEE)
WEEE

IEC 60601-1: Tıbbi elektrikli ekipman Bölüm 1 Temel
güvenlik ve temel performans için genel gereksinimler
Sızdırmaz
Bu ürün, ABD’de ve tüm dünyada geçerli bir veya
daha fazla patent kapsamında olabilir. Geçerli
patentlerin listesini görmek için lütfen aşağıdaki
web sitemizi ziyaret edin. Pat.: www.caireinc.com/
corporate/patents/.

Oksijen Yoğunlaştırıcı Nedir?
Soluduğumuz hava yaklaşık %21 oksijen, %78 azot
ve %1 oranında diğer gazlardan oluşur. FreeStyle®
Comfort®, odadaki havayı hava girişleri yoluyla içineçeker. Hava daha sonra moleküler elek adı verilen
tutucu bir maddeden geçirilir. Bu madde, oksijeni
azottan ayırır ve yalnızca oksijenin geçmesine
izin verir. Bunun sonucunda, kullanıcıya yüksek
yoğunlukta oksijen akışı sağlanır.
FreeStyle Comfort, gelişmiş oksijen yoğunlaştırıcı
ve oksijen koruma teknolojilerini birleştirerek
maksimum taşınabilirlik ve hareketliliğe olanak
tanıyan taşınabilir bir cihazdır. Gelişmiş puls akışı
sunumu, kullanıcının nefes alışını hızla algılar ve
alınan her nefesin başlangıcında bir puls oksijen sağlar.
Puls akışı sunumu, evin dışında yapılan faaliyetler vegeçirilen zaman için idealdir. Sürekli akışın aksine,oksijeni muhafaza eder ve pil ömrünü uzatır. FreeStyle
Comfort, omuz askısı ile taşınacak kadar hafiftir. AC(duvar prizi), DC (taşıt elektriği) veya pil gücüyleçalıştırılabilir.

Kullanım amacı
CAIRE FreeStyle Comfort Oksijen Yoğunlaştırıcı,
ilave oksijenin uygulanması için tasarlanmıştır. Cihazyaşam desteği için tasarlanmamıştır ve herhangi bir
hasta izleme yeteneği sağlamamaktadır.

Hekiminiz Neden Oksijen
Reçete Etti?
Birçok kişi çeşitli kalp, akciğer ve diğer solunum
hastalıklarından şikayetçidir. Bu tür hastalıklardanşikayetçi olan çok sayıda kişi evde, seyahat sırasında
veya evin dışında gündelik faaliyetlere katılırken ek
oksijen terapisinin faydasını görebilir.
Oksijen, soluduğumuz ortam havasının %21’inioluşturan bir gazdır ve vücudumuzun düzgün şekildeçalışması için sabit bir oksijen kaynağına ihtiyacı
vardır. Hekiminiz, solunum durumunuza faydasağlaması için bir akışı veya ayarı reçete etmiştir.

Oksijen bağımlılık yapmayan bir ilaç olsa daonaysız oksijen tedavisi tehlikeli olabilir. Bu
oksijen yoğunlaştırıcıyı kullanmadan önce hekimedanışmanız gerekir. Oksijen ekipmanınızı sağlayan
Ekipman Sağlayıcısı, reçete edilen akış hızının nasıl
ayarlanacağını gösterecektir.

Operatör ProfiliYoğunlaştırıcılar, akciğerlerin havadaki oksijeni
kan dolaşımına aktarma verimliliğini etkileyen
hastalıklardan dolayı rahatsızlık çeken kullanıcılara
ek oksijen sağlamak üzere tasarlanmıştır. Taşınabilir
oksijen yoğunlaştırıcılar (POC’ler) oksijen depolamaz
veya içermez. Yeniden doldurulmaları gerekmeyen
bu yoğunlaştırıcılar, bir AC veya DC güç kaynağının
mevcut olduğu her yerde şarj edilebilir (uçak dışında).
Bu da kullanıcıyı evde kullanım, (evin hem dışında
hem de içinde) yürüme, hareketlilik ve genel yaşam
tarzı konusunda görece bağımsız hale getirir. Oksijenyoğunlaştırıcı, bir hekim tarafından reçete edilmesi
durumunda kullanılır ve yaşam destek amaçlı kullanımiçin tasarlanmamıştır.
Oksijen terapisinin her yaştan kullanıcıya
reçete edilebilmesine karşın, oksijen terapisinin
uygulandığı kişiler genellikle 65 yaşın üstündedirve bu kişiler, Kronik Obstrüktif Akciğer Hastalığı
(KOAH) dahil olmak üzere çeşitli solunum
hastalıklarından şikayetçidir. Kullanıcıların bilişsel
yetileri genellikle iyi durumdadır ve rahatsız
olduklarını ifade edebilmeleri gerekir. Kullanıcı,
rahatsızlıklarını iletebilecek durumda değilse veya
yoğunlaştırıcı etiketiyle kullanım talimatlarını
okuyamayacak ve anlayamayacak durumdaysa,
cihazın yalnızca bu konularda yardımcı olabilecekbirinin gözetiminde kullanılması önerilir.
Yoğunlaştırıcının kullanımı sırasında herhangi bir
rahatsızlığın hissedilmesi durumunda kullanıcıların,sağlık hizmeti sağlayıcılarıyla iletişime geçmeleriönerilir. Kullanıcılara ayrıca, güç kesintisi veya
yoğunlaştırıcının arızalanması ihtimaline karşı yedek
oksijen kaynağı (yani silindir tüpte sağlanan oksijen)
bulundurmaları önerilir. Yoğunlaştırıcı kullanımı için
gerekli olan başka herhangi bir beceri veya yeterlilik
söz konusu değildir.
PN MN232-C4 K | Kullanıcı El Kitab

TUR - 375

FreeStyle Comfort

FreeStyle Comfort Ürününüzü Kutusundan Çıkarma
Listelenen tüm bileşenlerin
ambalajda mevcut olduğunu
doğrulayın. Eksik bir parça varsa
oksijen sağlayıcınızla hemen
iletişime geçin.

Standart Aksesuarlar• Taşınabilir Oksijen Yoğunlaştırıcı
• Taşıma Çantası• Omuz Askısı• 8 Hücreli Pil Takımı• AC Güç Kaynağı
• AC Güç Kablosu• DC Güç Kaynağı
• Kullanıcı El Kitabı

Opsiyonel Aksesuarlar• Ek 8 Hücreli Pil Takımı• 16 Hücreli Pil Takımı• Masaüstü Şarj Cihazı• Tüm Parçalar için AksesuarÇantasıFreeStyle Comfort Ürününüzü Tanıma
FreeStyle Comfort ve kullanıcı arayüzünün temel özelliklerini tanıyın.

Önden Görünüm
Askı Ek Parçası

Kullanıcı Arayüzü

OksijenÇıkışı
Hava Çıkışı

Hava Girişi/Filtreler
Hava Girişi/Filtreler
Güç Portu
Priz Tırnağı

Pil Takımı
Pil Takımını ÇıkarmaDüğmesi
376 - TURPN MN232-C4 K | Kullanıcı El Kitab

FreeStyle ComfortÜstten Görünüm
Görüntü EkranıAkışı Artırma

Güç Düğmesi

Menü Düğmesi

Akışı Azaltma

Pilde Görüntü EkranıAkış Ayarı

Pil Ekranı

autoSAT Göstergesi

Alarm Ekranı

Kalan Pil
Yüzdesi

Bolus Verilme
Göstergesi

AC/DC Güçte Görüntü EkranıAkış Ayarı
Pil Ekranı

Alarm Ekranı
AlanıŞarj Göstergesi

PN MN232-C4 K | Kullanıcı El Kitab

TUR - 377

FreeStyle ComfortÖnemli!

Güvenlik Talimatları aşağıdaki gibitanımlanmıştır:UYARI: CİDDİ YARALANMAYA NEDEN
OLABİLECEK TEHLİKELERE DAİRÖNEMLİ GÜVENLİK BİLGİLERİ.
DİKKAT: FreeStyle Comfort ünitesinin zarar
görmesini önlemek için gerekli bilgiler.

Not: Özellikle dikkat edilmesi gereken bilgiler.

Kullanım Endikasyonları
UYARI: YALNIZCA REÇETE EDILENOKSIJEN DÜZEYININ SEÇILMESI BÜYÜK
ÖNEM TAŞIR. LISANSLI BIR KLINISYENTARAFINDAN BU YÖNDE TALIMATVERILMEDIĞI SÜRECE AKIŞ SEÇIMINIDEĞIŞTIRMEYIN.
UYARI: FEDERAL (ABD) YASALAR, BU CIHAZINYALNIZCA BIR HEKIM VEYA LISANSLI BAŞKA BIRSAĞLIK HIZMETI SAĞLAYICISI TARAFINDAN YADA BU KIŞILERIN TALIMATIYLA SATILMASINI VEYAKIRALANMASINI GEREKLI KILAR.
UYARI: BU ÜNITE YAŞAM DESTEK ÜNITESI OLARAKKULLANILAMAZ. BU MAKINEYI KULLANIRKENRAHATSIZLIĞINI ILETEMEYECEK OLAN YAŞLILAR,
ÇOCUKLAR YA DA DIĞER KULLANICILARIN EK BIRYARDIMA IHTIYAÇLARI OLABILIR. İŞITME VE/VEYAGÖRME BOZUKLUKLARI OLAN KULLANICILAR,
ALARMLARI IZLEME KONUSUNDA YARDIMA IHTIYAÇDUYABILIR.
UYARI: PULS DOZU MODU AYARLARI, DINLENME,EGZERSIZ VE SEYAHAT SIRASINDAKI
IHTIYAÇLARINA GÖRE HER BIR KULLANICI IÇIN AYRIAYRI BELIRLENMELIDIR.

Kullanım Kontrendikasyonları
UYARI: BELİRLİ DURUMLARDA,REÇETELENMİŞ OLANIN DIŞINDA
OKSİJEN KULLANIMI TEHLİKELİ
OLABİLİR. BU CİHAZ YALNIZCA BİR HEKİMTARAFINDAN REÇETE EDİLDİĞİNDE
KULLANILMALIDIR.
UYARI: YANICI ANESTETİKLERİN BULUNDUĞU
YERLERDE KULLANILMAMALIDIR.
UYARI: ELEKTRIKLE ÇALIŞAN TÜM CIHAZLARDAOLDUĞU GIBI KULLANICI, ELEKTRIK KESINTISINEDENIYLE KULLANIM DIŞI KALMA SÜRELERIYLE
KARŞI KARŞIYA KALABILIR YA DA OKSIJEN
YOĞUNLAŞTIRICIYA YETKILI BIR TEKNISYENINSERVIS IŞLEMI UYGULAMASI GEREKEBILIR.
OKSIJEN YOĞUNLAŞTIRICI, BU TÜR GEÇICIKESINTILER SONUCUNDA SAĞLIK AÇISINDANOLUMSUZ SONUÇLARLA KARŞI KARŞIYAKALABILECEK KULLANICILAR IÇIN UYGUN
DEĞILDIR.

Güvenlik Kılavuzları
UYARI: BU EKIPMAN ÜZERINDEHERHANGI BIR DEĞIŞIKLIK YAPILMASIYASAKTIRUYARI: BU CIHAZ, HIZLI YANMAYI KOLAYLAŞTIRANYÜKSEK YOĞUNLUKLU OKSIJEN SAĞLAMAKTADIR.
(1) BU CIHAZIN VEYA (2) OKSIJEN TAŞIYANHERHANGI BIR AKSESUARIN BULUNDUĞU ODADASIGARA IÇILMESINE VEYA AÇIK ATEŞ OLMASINA
IZIN VERMEYIN. BU UYARIYA UYULMAMASI CIDDI
YANGINLARA, MADDI HASARA YOL AÇABILIRVE/VEYA FIZIKSEL YARALANMA YA DA ÖLÜMESEBEBIYET VEREBILIR.

UYARI: HASTANIN SOLUNUM ÇABALARI, KORUMAFONKSİYONUNU TETİKLEMEYEBİLİR.

UYARI: ÜNITEYI, HAVALANDIRMANIN SINIRLIOLABILECEĞI KISITLI VEYA SIKIŞIK BIR ALANDA(ÖR. KÜÇÜK BIR KILIF VEYA EL ÇANTASIIÇINDE) ÇALIŞTIRMAYIN. BU DURUM, OKSIJEN
YOĞUNLAŞTIRICININ AŞIRI ISINMASINA VE
PERFORMANSININ DÜŞMESINE NEDEN OLABILIR.

UYARI: FREESTYLE COMFORT, TRAKEOSTOMİUYGULANMIŞ HASTALARDA KULLANILMAK ÜZERETASARLANMAMIŞTIR.

UYARI: YOĞUNLAŞTIRICININ, DUMAN, KIRLETICIMADDE VEYA ISTEN UZAK BIR YERDEKONUMLANDIRILMASI GEREKIR.

UYARI: HAMİLE VEYA EMZİREN KADINLAR,FTALATLAR İÇERDİĞİ İÇİN BU EL KİTABINDA
ÖNERİLEN AKSESUARLARI KULLANMAMALIDIR.

UYARI: BU CIHAZLA CIDDI BIR OLAY OLMASIDURUMUNDA, KULLANICI OLAYI DERHALSAĞLAYICIYA VE / VEYA ÜRETICIYE BILDIRMELIDIR.
CIDDI BIR OLAY, OLAYIN TEKRARLANMASIDURUMUNDA YARALANMA, ÖLÜM VEYA YARALANMA/ ÖLÜME NEDEN OLMA POTANSIYELI OLARAKTANIMLANIR. KULLANICI AYRICA OLAYI, OLAYIN
MEYDANA GELDIĞI ÜLKEDEKI YETKILI MAKAMARAPOR EDEBILIR.

378 - TURPN MN232-C4 K | Kullanıcı El Kitab

FreeStyle Comfort
UYARI: BU OKSIJEN YOĞUNLAŞTIRICIDAKULLANIM IÇIN BELIRTILMEMIŞ OLAN
OKSIJEN UYGULAMA AKSESUARLARININ
KULLANILMASI, ÜNITE PERFORMANSININ
DÜŞMESINE YOL AÇABILIR. ÖNERILENAKSESUARLAR BU EL KITABINDA
BELIRTILMIŞTIR.
UYARI: OKSIJEN YOĞUNLAŞTIRICI DÜŞTÜYSE,HASAR GÖRDÜYSE YA DA SUYA MARUZ KALDIYSALÜTFEN ÜNITENIN INCELENMESI VEYA OLASIONARIMI IÇIN EKIPMAN SAĞLAYICINIZ ILE ILETIŞIME
GEÇIN. GÜÇ KABLOSU YA DA FIŞI HASARLIYSAOKSIJEN YOĞUNLAŞTIRICIYI KULLANMAYIN.
UYARI: HASARI VEYA YARALANMAYI ÖNLEMEKIÇIN TAŞIMA SIRASINDA TAŞIT IÇINDE OKSIJEN
YOĞUNLAŞTIRICIYI DÜZGÜN BIR ŞEKILDESABITLEYIN, KEMERLE BAĞLAYIN VEYA BAŞKA BIRŞEKILDE SABIT TUTUN.
UYARI: ABD ULAŞTIRMA BAKANLIĞI (DOT) VE
BIRLEŞMIŞ MILLETLER (BM) DÜZENLEMELERI,OKSIJEN YOĞUNLAŞTIRICININ BAGAJIÇINDE HAVAYOLUYLA TAŞINDIĞI TÜM
ULUSLARARASI SEYAHATLERDE PILIN CIHAZDANÇIKARILMASINI GEREKLI KILMAKTADIR. OKSIJEN
YOĞUNLAŞTIRICININ NAKLIYATI SIRASINDA DAPIL CIHAZDAN ÇIKARILMALI VE DÜZGÜN ŞEKILDEPAKETLENMIŞ OLMALIDIR.
UYARI: BU EKİPMANI, ÖNCELİKLE BU EL KİTABINIOKUYUP ANLADIĞINIZDAN EMİN OLDUKTANSONRA ÇALIŞTIRIN. UYARI VE TALİMATLARI
ANLAMAKTA SIKINTI YAŞIYORSANIZ BU EKİPMANIKULLANMADAN ÖNCE EKİPMAN SAĞLAYICINIZLAİLETİŞİME GEÇİN; AKSİ HALDE YARALANMA VEYAHASAR MEYDANA GELEBİLİR.
UYARI: HERHANGI BIR RAHATSIZLIĞINIZ YA DAACIL ILGILENILMESI GEREKEN BIR HASTALIĞINIZVARSA VAKIT KAYBETMEDEN TIBBI DESTEK IÇINBAŞVURUN.
UYARI: OKSIJEN YOĞUNLAŞTIRICININ, ÇALIŞTIRMASICAKLIĞI TEKNIK ÖZELLIKLERINDE BELIRTILENIN
DIŞINDA BIR SICAKLIKTA ÇALIŞTIRILMASI,YOĞUNLAŞTIRICININ OKSIJEN YOĞUNLUĞU TEKNIKÖZELLIKLERINE UYGUN ÇALIŞMA BECERISINISINIRLAYABILIR. SICAKLIK SINIRLARI IÇIN BU ELKITABININ TEKNIK ÖZELLIKLER BÖLÜMÜNE BAKIN.
UYARI: OKSIJEN TAŞINAN AKSESUARLAR VEYAOKSIJEN YOĞUNLAŞTIRICI ILE YAĞ, GRES YA DAPETROL BAZLI VEYA DIĞER YANICI ÜRÜNLERIKULLANMAYIN. YALNIZCA SU BAZLI, OKSIJENILE UYUMLU LOSYON YA DA MERHEMLERKULLANILMALIDIR. OKSIJEN, YANICI MADDELERINTUTUŞMASINI HIZLANDIRIR.

UYARI: HAVA GIRIŞI YA DA HAVA ÇIKIŞI EGZOZUNUNENGELLENMESINE IZIN VERMEYIN. CIHAZÜZERINDEKI AÇIKLIKLARDAN HIÇBIRININ IÇINE
NESNE DÜŞÜRMEYİN VEYA SOKMAYIN. BU DURUM,
OKSIJEN YOĞUNLAŞTIRICININ AŞIRI ISINMASINA VE
PERFORMANSININ DÜŞMESINE NEDEN OLABILIR.
UYARI: PILIN HATALI KULLANILMASI, PILIN
ISINMASINA VE TUTUŞMASINA YOL AÇARAK
CIDDI YARALANMALARA NEDEN OLABILIR. PILIKESINLIKLE DELMEYIN, DÜŞÜRMEYIN, PILE
VURMAYIN, PILIN ÜZERINE BASMAYIN VEYA PILIBAŞKA BIR ŞEKILDE SERT DARBELERE YA DAŞOKLARA MARUZ BIRAKMAYIN. HASARLI PILKULLANIMI, FIZIKSEL YARALANMAYA NEDEN
OLABILIR.
UYARI: PILIN METAL TEMAS UÇLARINI ANAHTAR
VEYA BOZUK PARA GIBI METALIK NESNELERLE KISA
DEVRE YAPTIRMAYIN. BU, KIVILCIM VEYA AŞIRI ISIOLUŞUMUNA NEDEN OLABILIR.
UYARI: TAŞINABILIR OKSIJEN YOĞUNLAŞTIRICIYI ACGÜÇ KAYNAĞI ILE DIŞ MEKANLARDA KULLANIRKEN,
GÜÇ KAYNAĞINI YALNIZCA TOPRAKLAMA KAÇAK
DEVRE KESICISI (GFI) BULUNAN BIR PRIZEBAĞLAYIN.
UYARI: OKSIJEN YOĞUNLAŞTIRICI, BAŞKA
EKIPMANLARLA BITIŞIK VEYA ÜST ÜSTEKONMUŞ OLARAK KULLANILMAMALIDIR.
BAŞKA EKIPMANLARLA BITIŞIK YA DA ÜST ÜSTEKONMUŞ OLARAK KULLANILMASI GEREKIYORSACIHAZIN NORMAL ŞEKILDE ÇALIŞIP ÇALIŞMADIĞIGÖZLEMLENMELIDIR.
UYARI: TIBBI ELEKTRIKLI EKIPMANIN ÜRETICISITARAFINDAN IÇ BILEŞENLERIN YEDEK PARÇALARI
OLARAK SATILAN KABLOLAR VE ADAPTÖRLER
HARIÇ OLMAK ÜZERE, BELIRTILENLERIN DIŞINDAKI
KABLO VE ADAPTÖRLERIN KULLANILMASI, OKSIJEN
YOĞUNLAŞTIRICININ BAĞIŞIKLIK SISTEMININ
ZAYIFLAYARAK DAHA FAZLA EMISYON SALMASINASEBEBIYET VEREBILIR.
UYARI: YALNIZCA CIHAZA YAPIŞTIRILMIŞSPESIFIKASYON ETIKETI ÜZERINDE BELIRTILENELEKTRIK VOLTAJINI KULLANIN.
UYARI: BU ÜNITEYI UZATMA KABLOSUYLAKULLANMAYIN VEYA AYNI ELEKTRIK PRIZINE ÇOK
SAYIDA FIŞ TAKMAYIN. UZATMA KABLOSUYLAKULLANILMASI CIHAZIN PERFORMANSINI OLUMSUZYÖNDE ETKILEYEBILIR. TEK BIR PRIZE ÇOK SAYIDA
FIŞ TAKILMASI, ELEKTRIK PANELINDE AŞIRI YÜKENEDEN OLARAK DEVRE KESICININ/SIGORTANIN
DEVREYE GIRMESINE YA DA DEVRE KESICI VEYASIGORTA ÇALIŞMAZSA YANGINA SEBEP OLABILIR.

PN MN232-C4 K | Kullanıcı El Kitab

TUR - 379

FreeStyle ComfortDİKKAT: Alarm durumunda veya
Oksijen Yoğunlaştırıcının doğru şekildeçalışmadığını gözlemlediğinizde, bu elkitabındaki Sorun Giderme bölümüne
bakın. Sorunu çözemezseniz EkipmanSağlayıcınıza danışın.
DİKKAT: Garantinin geçersiz hale gelmesini önlemekiçin üreticinin tüm talimatlarını izleyin.
UYARI: BU EL KITABINDA BELIRTILEN
OLASI ÇÖZÜMLER DIŞINDA HERHANGIBIR BAKIM IŞLEMINI DENEMEYIN.
KAPAKLARI ÇIKARMAYIN; SADECEEKIPMAN SAĞLAYICINIZ YA DA VASIFLIBIR SERVIS TEKNISYENI BU CIHAZINKAPAKLARINI ÇIKARMALI YA DA SERVISIŞLEMI GERÇEKLEŞTIRMELIDIR.
UYARI: SADECE ÜRETICI TARAFINDAN TAVSIYE
EDILEN AKSESUARLARI KULLANIN. BAŞKAHERHANGI BIR AKSESUARIN KULLANILMASIOKSIJEN YOĞUNLAŞTIRICINIZ IÇIN ZARARLIOLABILIR, CIDDI HASARA YOL AÇABILIR VEGARANTIYI GEÇERSIZ KILAR.
UYARI: ÜRETICI, ELEKTRIK KESINTISI, ALARM
DURUMU VEYA MEKANIK ARIZA IHTIMALINEKARŞI ALTERNATIF BIR TAMAMLAYICI OKSIJEN
KAYNAĞININ HAZIR BULUNDURULMASINI ÖNERIR.
GEREKLI OLAN YEDEK SISTEMIN TÜRÜNÜ
ÖĞRENMEK IÇIN HEKIMINIZE YA DA EKIPMAN
SAĞLAYICINIZA DANIŞIN.
UYARI: HAVA GIRIŞI YA DA HAVA ÇIKIŞI DELIKLERININ
ENGELLENMESINE IZIN VERMEYIN. CIHAZÜZERINDEKI AÇIKLIKLARDAN HIÇBIRININ IÇINE
NESNE DÜŞÜRMEYİN VEYA SOKMAYIN. BU DURUM,
OKSIJEN YOĞUNLAŞTIRICININ AŞIRI ISINMASINA VE
PERFORMANSININ DÜŞMESINE NEDEN OLABILIR.
UYARI: OKSIJEN KULLANIRKEN SIGARA IÇMEK,YANGINDAN KAYNAKLANAN YARALANMA VE BUNABAĞLI ÖLÜMLERIN EN YAYGIN NEDENIDIR. ŞUGÜVENLIK UYARILARINI IZLEMELISINIZ:UYARI: CIHAZIN YA DA OKSIJEN TAŞINANAKSESUARLARIN BULUNDUĞU ODADA SIGARAIÇILMESINE, MUM VEYA AÇIK ALEV BULUNMASINA
IZIN VERMEYIN.
UYARI: OKSIJEN KANÜLÜ TAKILIYKEN SIGARAIÇMEK, YÜZDE YANIKLARA NEDEN OLARAK ÖLÜMLE
SONUÇLANABILIR.
UYARI: SIGARA IÇIYORSANIZ BU 3 ADIM HAYATINIZIKURTARABILIR: OKSIJEN YOĞUNLAŞTIRICIYIKAPATIN, KANÜLÜ ÇIKARIN VE CIHAZIN BULUNDUĞUODADAN ÇIKIN.

380 - TURPN MN232-C4 K | Kullanıcı El Kitab

UYARI: EVDE YA DA OKSIJENIN KULLANILDIĞIHERHANGI BIR YERDE HER ZAMAN “SIGARAİÇILMEZ - OKSIJEN KAYNAĞI” BENZERI LEVHALARBULUNMALIDIR. KULLANICILAR VE BAKICILARI,
TIBBI OKSIJENIN BULUNDUĞU ORTAMDA VEYAKULLANILMASI SIRASINDA SIGARA IÇMENINTEHLIKELERI HAKKINDA BILGI SAHIBI OLMALIDIR.
UYARI: OKSIJEN YOĞUNLAŞTIRICINIZI YANICIGAZLARIN BULUNDUĞU ORTAMDA KULLANMAYIN.
BU DURUM HIZLA YANMAYA YOL AÇARAK MADDI
HASARA, VÜCUTTA YARALANMALARA VEYA ÖLÜMESEBEBIYET VEREBILIR.
UYARI: KANÜLÜN ÇIKARILMASI VE GIYSILERIN,YATAĞIN, KANEPENIN YA DA YASTIK BENZERIDIĞER EŞYALARIN ÜZERINE KONULMASI, SIGARA,ISI KAYNAĞI VEYA ALEVE MARUZ KALDIĞINDA
PARLAYARAK ALEV ALMASINA NEDEN OLACAKTIR.
UYARI: BURUN KANÜLÜNÜ GIYSILERIN, YATAKÖRTÜLERININ VEYA SANDALYE MINDERLERININÜZERINDE YA DA ALTINDA BIRAKMAYIN. ÜNITE AÇIKOLDUĞU HALDE KULLANIMDA DEĞILSE OKSIJEN BU
EŞYALARIN YANICI OLMASINA SEBEBIYET VERIR.
OKSIJEN YOĞUNLAŞTIRICININ KULLANILMADIĞI
SIRADA GÜCÜNÜN KAPALI KONUMDA OLDUĞUNDAN
EMIN OLUN.
UYARI: PİL TAKIMI ÇIKARILMADAN ÖNCE, OKSIJEN
YOĞUNLAŞTIRICI KAPALI KONUMDA OLMALIDIR VEHARİCİ BİR GÜÇ KAYNAĞINA BAĞLI OLMAMALIDIR.
UYARI: KÜÇÜK PARÇALAR, BOĞULMATEHLIKESINE NEDEN OLABILIR.
UYARI: SICAK OLABILECEĞI IÇIN EGZOZ HAVA
AĞZIYLA 10 SANIYEDEN UZUN SÜRE TEMASETMEKTEN KAÇININ.
UYARI: RÜZGAR VEYA GÜÇLÜ HAVA AKIMLARI,OKSIJEN TEDAVISININ DOĞRU ŞEKILDEUYGULANMASINI OLUMSUZ ETKILEYEBILIR.
UYARI: BU ÜRÜN, CALIFORNIA EYALETI TARAFINDANKANSERE NEDEN OLDUĞU BILINEN NIKEL GIBIKIMYASAL MADDELERE MARUZ KALMANIZAYOL AÇABILIR. DAHA FAZLA BILGI IÇIN WWW.
P65WARNINGS.CA.GOV WEB SITESINI ZIYARET EDIN.

FreeStyle ComfortDİKKAT: AC güç kaynağını OksijenYoğunlaştırıcıdan ayırmadan önce her
zaman AC güç kaynağını prizden çekin.
DİKKAT: Oksijen Yoğunlaştırıcıyı, güç kablosunaerişimi güçleştirecek şekilde konumlandırmayın.
DİKKAT: Oksijen kaynağı borularını ve güç kablolarını,
insanların takılmasını veya olası bir dolanmayı
engelleyecek biçimde yerleştirin.
DİKKAT: Oksijen Yoğunlaştırıcıyı otomobilde, teknede
veya DC güç kaynağına sahip diğer DC kaynaklarıüzerinde kullanırken, ürünü bağlamadan önce taşıtın
çalıştırıldığından ve çalışır durumda olduğundan eminolun. Bu talimatların uygulanmaması, güç kaynağının
Oksijen Yoğunlaştırıcıya güç sağlamamasıyla
sonuçlanabilir.
DİKKAT: Oksijen Yoğunlaştırıcı ünitesini kullandığınız
otomobil çalışmaz durumdayken, cihazın otomobilleolan bağlantısını kesin ve cihazı çıkarın. OksijenYoğunlaştırıcıyı çok sıcak veya soğuk bir otomobilde
veya yüksek ya da düşük sıcaklığa sahip diğer benzer
ortamlarda saklamayın.
Not: Taşınabilir ve mobil RF haberleşme ekipmanı
tıbbi elektrikli ekipmanı etkileyebilir.
DİKKAT: Marş KAPALI konumdayken,
Oksijen Yoğunlaştırıcıyı veya Güç
Kaynağını taşıta takılı şekilde BIRAKMAYIN.
Bu, taşıtın pilini tüketebilir.
DİKKAT: Oksijen Yoğunlaştırıcının normal çalışma
sıcaklığı aralığı ve ortam basıncı aralığı (irtifa)dışında çalıştırılması veya saklanması, performansı
etkileyebilir ve pil çalışma süresini azaltabilir ve/veya
pil şarj süresini artırabilir. Saklama ve çalışma sıcaklığı
sınırları için bu el kitabının Spesifikasyonlar bölümüne
başvurun.
DİKKAT: Pil donanımlı Oksijen Yoğunlaştırıcılariçin: Pilinizin uzun ömürlü olmasını sağlamak içinpilleri serin ve kuru bir konumda saklayın. OksijenYoğunlaştırıcınızın uzun bir süre boyunca yüksek
sıcaklıklarda veya tam şarj edilmiş ya da tamamenboşalmış bir pille saklanması, genel pil ömrünü
azaltabilir. Pilin içini açma girişiminde bulunmayın;
pil içinde servis uygulanabilen parça yoktur. Pilleriçocukların erişemeyeceği konumda tutun.
DİKKAT: YALNIZCA Üretici tarafından sağlanan pilleri
KULLANIN. Pillerin doğru şekilde elden çıkarılması
için imha gereksinimleriyle ilgili olarak EkipmanSağlayıcınız veya yerel idari birimle iletişime geçin.

Not: Kanülün tamamen takılı olduğundan ve sabitlendiğinden emin olun. Oksijen Yoğunlaştırıcı, bu
sayede oksijen uygulaması için nefes almayı doğru
şekilde algılar. Nefes alma sırasında burun kanülünün
sivri uçlarına oksijen akışı olduğunu duymanız ya dahissetmeniz gerekir.
Not: Lityum piller, tam şarj edilmiş veya tamamenboşaltılmış durumdayken aşırı yüksek sıcaklıklara
maruz bırakıldığında kalıcı kapasite kaybına uğrar.
Uzun süreli saklama için pillerin %25-50 oranında şarjedilmesi ve -20°C ila +20°C (-4°F ila 68°F) sıcaklık
aralığında tutulması önerilir.
Not: Bu oksijen yoğunlaştırıcıyla kullanıma uygunolduğu belirtilmemiş olan bazı oksijen uygulama
aksesuarlarının kullanılması, performansı düşürebilir.
Önerilen aksesuarlar bu el kitabında belirtilmiştir.
Not: Oksijen Yoğunlaştırıcıyı, hava girişi filtresi takılı
olmadan çalıştırmayın. İkinci bir filtre sağlanmışsa kirli
filtreyi temizlemeden önce yedek filtreyi takın. Kirli filtreyi ılık su ve sabun çözeltisinde temizleyin, ardından
kullanmadan önce tamamen kurutun.
Not: Üretici bu ekipmanın sterilize edilmesini önermemektedir.
Not: Oksijen yoğunlaştırıcı ünitenizi kullanırken
odadaki oksijenin tükenme tehlikesi asla söz konusudeğildir.
UYARI: OKSIJEN YOĞUNLAŞTIRICI,IÇERISINDE PIL OLMADAN
ÇALIŞTIRILMAMALI VE HARICIBIR (AC/DC) GÜÇ KAYNAĞINA
BAĞLANMAMALIDIR. PIL TAKIMIÇIKARILMIŞSA ÜNITEYI ÇALIŞTIRMAYIVEYA HARICI BIR GÜÇ KAYNAĞINA
BAĞLAMAYI DENEMEYIN.
PIL TAKIMINI ANCAK ÜNITE KAPALI KONUMAGETIRILDIKTEN VE ÜNITENIN GÜÇ KAYNAĞI
BAĞLANTISI KESILDIKTEN SONRA ÇIKARIN. ÜNITE
AÇIK KONUMDA VEYA GÜÇ KAYNAĞINA BAĞLI
DURUMDAYKEN PIL TAKIMINI ÇIKARMAYIN.

Not: Oksijen yoğunlaştırıcı yukarıdaki tabloda
belirtilen normal çevresel çalışma koşullarınındışında uzun bir süre boyunca saklanmışsa, üniteninçalıştırılmadan önce normal çalışma sıcaklığına geri
dönmesine izin verilmelidir.

PN MN232-C4 K | Kullanıcı El Kitab

TUR - 381

FreeStyle Comfort

Teknik Özellikler

FreeStyle Comfort
Boyutlar (Tek Pil ile)

25,4 x 18,5 x 7,9 cm (10,0 x 7,3 x 3,1 inç)

Boyutlar (Çift Pil ile)

27,9 x 18,5 x 7,9 cm (11,0 x 7,3 x 3,1 inç)Ağırlık (Tek Pil Takımı ile)

2,3 kg (5 lb)Ağırlık (Çift Pil Takımı ile)

2,7 kg (6 lb)

Anma Ses Düzeyi***39,93 dB (A) ± 0,42 dB (A) @ 2Akış Ayarları

Puls Ayarı 1, 2, 3, 4, 5

O2 Çıkış Ayarı

Ayar 1 - 210 mL/dk
Ayar 2 - 420 mL/dk
Ayar 3 - 630 mL/dk
Ayar 4 - 840 mL/dk
Ayar 5 - 1.050 mL/dk(20 bpm’de)Oksijen Yoğunluğu*%90 (+%5,5 / -%3)Oksijen Monitörü

Evet

Maksimum Çıkış Basıncı<30 PSI

Varsayılan Dozaj Hassasiyeti> -0,1 cm H2O (ayarlanabilir)

AC Güç

100-240 VAC, 50-60 Hz, 120W Max

DC Güç

11–18 VDC (10 maks amp)

Pil Şarj Süresi

Tek Pil: 3,5 saatÇift Pil: 6 saat

Pil Takımı Süreleri

Tek Pil Süresi: Ayar 2 - en fazla 4 saatÇift Pil Süresi: Ayar 2 - en fazla 8 saatÇalışma Sıcaklığı**Pil Türü
Pil Kapasitesi

5°C - 40°C (41°F - 104°F)
Lityum İyonTek hücre: 6700 mAh; Çift hücre: 13400 mAhTaşıma ve Depolama Sıcaklığı
(Ünite)3.048 m’ye kadar (10.000 fit) -25°C ila +70°C (-13°F ila +158°F)Taşıma ve Depolama Sıcaklığı (Pil)3.048 m’ye kadar (10.000 fit) -25°C ila +70°C (-13°F ila +158°F)Çalışma İrtifası**-381 ila 3.048 m (1.250 ila 10.000 fit) (700 - 1.060 hPa’da test edilmiştir)Çalışma Nemliliği

%15 - 95 bağıl nem (yoğuşmasız)

Depolama Nemi

0% - 90% Yoğuşmasız

Sınırlı GarantiÜnite için 3 yıl
Pil ve güç kaynağı için 1 yıl* 21°C (70°F) sıcaklıkta 14,7 psi (101 kPa) atmosfer basıncı baz alınmıştır
** Ortam, belirtilen sıcaklık ve nem aralıklarına sahip olduğu sürece, ünite tüm ortamlarda çalışacaktır. Bu kullanım koşullarıdışında çalıştırmak, yoğunlaştırıcının daha yüksek litreli akış hızlarında Oksijen Yoğunlaştırma koşullarını karşılama becerisinisınırlayabilir. Isınma ekranı, yoğunlaştırıcı hedef yoğunluğa ulaşana dek veya en fazla 15 dakika boyunca görünmeye devam
edebilir. Bu süre boyunca üniteyi kullanmaya devam edin.
*** Test yöntemi Nr. 14-1 10/2018 MDS-Merhaba. Öngörülen kullanım ömrü en az beş yıldır.
Elek yataklarının beklenen raf ömrü, yoğunlaştırıcı sürekli olarak kullanılmadığı takdirde yaklaşık bir yıldır.
382 - TURPN MN232-C4 K | Kullanıcı El Kitab

FreeStyle ComfortÇalıştırma Talimatları
1. FreeStyle Comfort cihazını, kullanım için uygun bir
konuma yerleştirin.
a. Hava girişi filtrelerinin ve hava çıkış kanalının
tıkanmaması için yoğunlaştırıcının yeterli bir
biçimde havalandırılması gereklidir.
b. Yoğunlaştırıcının, her türlü işitsel/görsel
gösterge ve alarmın kolayca görülebilecekve işitilebilecek biçimde konumlandırılması
gereklidir.
c. Üniteyi çalıştırmadan önce hava girişi
filtrelerinin yerinde olduğundan emin olun.
Ünitede bir filtre yoksa ekipman sağlayıcınızla
iletişime geçin.
2. Yoğunlaştırıcıya bir Pil Takımı takın.
a. Pili, FreeStyle Comfort’ın merkezindeki alt
muhafazayla hizalayın.

3. AC veya DC güç kaynağını, ünitenin ön sağ alt
tarafında bulunan güç portu prizine takın.

4. Burun kanülünü oksijen çıkış ucuna bağlayın.
Cihazı çalıştırmadan önce bağlantınıngerçekleştiğinden emin olun.

NOT: Önerilenler dışındaki kanüllerin kullanılması
performansı etkileyebilir.

Not: Ekipman Sağlayıcıya: FreeStyle Comfort ile kullanımiçin aşağıdaki oksijen uygulama aksesuarları önerilir:b. Pilin yerine oturup kilitlendiğini belirten bir tık
sesi duyana kadar pili içeri doğru kaydırın.

• 7 fit (2,1 m) tüplü Burun Kanülü (Maksimum LPM: 6
LPM): CAIRE Parça Numarası 5408-SEQ• Ek önerilen aksesuar bilgileri çevrimiçi olarak www.
caireinc.com adresinde mevcuttur.

PN MN232-C4 K | Kullanıcı El Kitab

TUR - 383

FreeStyle Comfort
5. Güç düğmesine basarak FreeStyle Comfort’ı açınve akışı artırma (+) veya azaltma (-) düğmelerini
kullanarak reçete edilmiş olan akış ayarınızı seçin.

autoSAT Puls Hacmi (mL)*Ayarı
Nefes
Hızı

1

2

3

4

5

15

10,5

21,0

31,5

42,0

52,5

20

10,5

21,0

31,5

42,0

52,5

25

10,5

21,0

31,5

42,0

42,0

30

10,5

21,0

31,5

35,0

35,0

35

10,5

21,0

30,0

30,0

30,0

40

10,5

21,0

26,3

26,3

26,3

*Tolerans +/- %15’tir

Cihaza güç verildiğini belirtmek için kısa bir sesli
alarm duyulacaktır.

Not: FreeStyle Comfort açıldığında, varsayılanakış ayarı 2’dir. Ayar değiştirilirse ünite önceki akışayarında açılacaktır.
6. Burun kanülünü, kanülle birlikte verilen üretici
talimatlarına uygun olarak yüzünüze yerleştirin ve
normal şekilde nefes alıp vermeye başlayın. Her
soluk aldığınızda, burun kanülünden verilen biroksijen akışı hissetmeniz gerekir.
7. Akış ayarını değiştirmek için akışı artırma (+)
veya azaltma (-) düğmeleriyle istenen akış hızını
ayarlayın. Akış ayarlarını değiştirdiğinizde farklı bir
ses duymanız normaldir.

autoSAT olmayan Puls Hacmi (mL)*Ayarı
Nefes
Hızı

1

2

3

4

5

15

14,0

28,0

42,0

56,0

70,0

20

10,5

21,0

31,5

42,0

52,5

25

8,4

16,8

25,2

33,6

42,0

30

7,0

14,0

21,0

28,0

35,0

35

6,0

12,0

18,0

24,0

30,0

40

5,3

10,5

15,8

21,0

26,3

8. Güç düğmesine basarak FreeStyle Comfort’ı
kapatın.

Not: Görüntü ekranı, bekleme moduna girecek ve 20
saniye sonra kararacaktır. Ekranı bekleme modundançıkarmak için herhangi bir düğmeye basın.
9. Gelişmiş Özellikler için Menü Gezinme
Talimatlarına bakın

ÖZEL ÜRÜN ÖZELLIKLERI
autoSAT®
CAIRE’nin tescilli autoSAT özelliği, FreeStyle
Comfort’un maksimum nefes hızına kadar nefesbaşına sabit bir oksijen hacmi (bolus) vermesini sağlar.
Kullanıcının yürüme gibi aktiviteler sırasında nefes
hızı arttıkça, autoSAT özelliği ayarlanmış bir bolus
hacmini korumak için cihazın çalışmasını ayarlar.
UltraSense®
Bu hassas puls dozu tetikleyicisi, solunumunuz
sırasındaki basınç değişimini algılar ve siz nefes
alırken oksijen pulsu verir.
autoDOSE™
“Solunum yok alarmı” verildiğinde bu özellik, alarm
durumu giderilene kadar ünitenin mevcut akış ayarıyla
dakikada 20 nefeslik oksijen sağlar.
384 - TURPN MN232-C4 K | Kullanıcı El Kitab

FreeStyle Comfort
Güç Kaynakları
FreeStyle Comfort’a üç şekilde güç verilebilir:• Şarj Edilebilir Pil Takımı• AC Güç• DC Güç
Bu bileşenlerin her birinin çalıştırılmasına ilişkin
bilgiler, aşağıdaki bölümlerde verilmiştir.
UYARI: YALNIZCA BU KULLANICIEL KITABINDA BELIRTILEN GÜÇ
KAYNAKLARINI KULLANIN. CAIRETARAFINDAN ÜRETILMEMIŞ GÜÇ
KAYNAKLARI VE PIL TAKIMLARININ
KULLANILMASI, ÜNITEYE ZARAR
VEREBILIR VE ÜNITENIN DOĞRU BIR
BIÇIMDE ÇALIŞMAMASINA YOL AÇABILIR.

PİLİN ÇIKARILMASI

Pil çıkarma düğmesini FreeStyle Comfort cihazından
aşağı doğru itin. Bu düğmeyi TUTARKEN, piliyoğunlaştırıcıdan kaydırarak çekip çıkarın.
UYARI: PIL TAKIMI ÇIKARILMADAN ÖNCE,OKSIJEN YOĞUNLAŞTIRICI KAPALIKONUMDA OLMALIDIR VE HARICI BIR
GÜÇ KAYNAĞINA BAĞLI OLMAMALIDIR.

PİL TAKIMIFreeStyle Comfort, cihazla birlikte verilen şarjedilebilir pil takımıyla çalıştırılabilir. Pil takımıylaçalıştırılırken, yoğunlaştırıcının çalıştırılmasına ilişkin
herhangi bir sınırlama yoktur. Pille çalıştırılırken tümakış hızları kullanılabilir.

PİL TAKIMININ DOLULUK DÜZEYİNİN
KONTROL EDİLMESİ
PİLİN TAKILMASI

Pili, FreeStyle Comfort’ın merkezindeki alt
muhafazayla hizalayın. Pilin yerine oturup
kilitlendiğini belirten bir tık sesi duyana kadar pili içeridoğru itin.

Pilin FreeStyle Comfort cihazına takılıyken doluluk
düzeyini kontrol etmek için pil simgesine ve ekranda
gösterilen kalan tahmini süreye bakın.
%80-100%60-80%40-60%20-40%20

PN MN232-C4 K | Kullanıcı El Kitab

TUR - 385

FreeStyle Comfort
Not: Pil şarj ediliyorsa kalan şarj yüzdesi yerine fişgöstergesi gösterilecektir.
Pilin FreeStyle Comfort cihazına takılı olmadığı
durumda doluluk düzeyini kontrol etmek için pili yereserin ve TEST düğmesine basın. TEST düğmesininsağındaki pil durumu gösterge ışıkları (%25-100)
yanarak pil takımının doluluk düzeyini gösterecektir.

PİL GÜCÜYLE ÇALIŞTIRMA SÜRESİ

Harici AC veya DC güç bağlantısı kesildiğinde, şarjedilmiş bir pil takılmışsa yoğunlaştırıcı otomatik olarak
pil gücüne geçer. Yeni bir pil takımının tipik kullanımsüreleri aşağıdaki tabloda gösterilmektedir. Solunum
hızınıza bakılmaksızın, pil takımından çalışma süresi
sabittir.
Puls
Ayarı

Tek (8 Hücreli) Pil
SüresiÇift (16 Hücreli) Pil
Süresi1

en fazla 8 saat

en fazla 16 saat2

en fazla 4 saat

en fazla 8 saat3

en fazla 3 saat

en fazla 6 saat

4

en fazla 2,25 saat

en fazla 4,5 saat

5

en fazla 2 saat

en fazla 4 saatTİPİK PİL ŞARJ ETME SÜRESİ
İLK KEZ ŞARJ ETMEFreeStyle Comfort cihazınızla birlikte verilen yeni
pil, fabrikadan çıktığında kısmen şarj edilmiştir. İlk
kullanımdan önce, AC veya DC güç kullanarak pil
takımını tamamen şarj etmeniz gerekir.

GENEL ŞARJ ETME BİLGİLERİ

Pillerin yoğunlaştırıcıya takılı, yoğunlaştırıcının daharici bir AC veya DC güç kaynağına bağlı olduğu her
zaman, FreeStyle Comfort pil takımını dolduracaktır(uçak dışında).
Pil, %100 gösterildiğinde tamamen şarj olmuştur.
Bunun yerine pil takımı, opsiyonel masaüstü şarj cihazı
kullanılarak da şarj edilebilir.

386 - TURPN MN232-C4 K | Kullanıcı El Kitab

Pil takımınızı tam şarjı bir pilden tamamen şarj etmekiçin gereken tipik süre, tek pil için 3,5 saat ve çift pil
için 6 saattir.

Not: Pil, pil paketi tamamen boşalmışsa başlangıçtaşarj etmeye başlamayabilir.

PİL HİZMET ÖMRÜFreeStyle Comfort PİLLERİ, 300 şarj/boşalma
döngüsü kadar dayanacak şekilde tasarlanmıştır.

FreeStyle Comfort
AC ve DC Güç Kaynağı
FreeStyle Comfort, standart olarak AC ve DC güç
kaynaklarıyla birlikte sunulur.
AC güç kaynağı şunları içerir:a. 100-240 volt, 50-60 Hz arasında çalışan AC
Güç adaptörü.
b. Güç adaptörü ve ilgili AC priziyle bağlantı
kurmak için kullanılan AC Güç kablosu.

DC güç kaynağı, doğrudan üniteye ve DC prizine
takılan bir kablodan oluşmaktadır.

2. AC güç kablosunu AC güç adaptörüne bağlayın.

AC GÜÇ

AC güç, evde veya standart AC prizlerinin mevcutolduğu her yerde kullanılabilir. FreeStyle Comfort, AC
güçle eksiksiz çalışır. FreeStyle Comfort, AC gücün
verildiği her zaman pil takımını da şarj edecektir (uçakdışında). FreeStyle Comfort’ın çalışır durumda olup
olmamasına bakılmaksızın, pil takımı şarj edilir.
UYARI: OKSIJEN YOĞUNLAŞTIRICIYA PILTAKIMININ TAKILMADIĞI DURUMLARDAÜNITEYI HARICI BIR GÜÇ KAYNAĞINA
BAĞLAMAYIN.
FreeStyle Comfort ünitesini AC Güç kaynağına
bağlamak için:
1. AC güç adaptörünü, FreeStyle Comfort’ın sağ altköşesinden ön taraftaki güç konnektörü girişine
bağlayın. Bağlantının gerçekleşmesi için içeri doğru
itin.

3. AC güç kablosunun diğer ucunu prize veya uygun
olan başka bir elektrik çıkışına bağlayın.

PN MN232-C4 K | Kullanıcı El Kitab

TUR - 387

FreeStyle Comfort
4. Güç düğmesine basarak FreeStyle Comfort’ı açınve akışı artırma (+) veya azaltma (-) düğmelerini
kullanarak reçete edilmiş olan akış ayarınızı seçin.

3. DC güç kablosunun diğer ucunu otomobile veya
başka bir DC aksesuar çıkışına bağlayın.

Not: Pil takımının şarj edilmesi gerekirse ekrandaki pilsimgesinde tek bir turuncu çubuk olacaktır.

DC GÜÇ

DC güç, otomobil, tekne veya diğer motorlu taşıtlarınaksesuar çıkışlarından kullanım içindir. FreeStyle
Comfort, DC güçle eksiksiz şekilde kullanılabilir.
FreeStyle Comfort, DC gücün verildiği her zaman pil
takımını da şarj eder. Pil takımı, ünite açık olduğunda
da kapalı olduğunda da şarj olur.
UYARI: OKSIJEN YOĞUNLAŞTIRICIYA PILTAKIMININ TAKILMADIĞI DURUMLARDAÜNITEYI HARICI BIR GÜÇ KAYNAĞINA
BAĞLAMAYIN.
UYARI: CIHAZI, TAŞIT ÇALIŞIR DURUMDA DEĞILKEN
12 VDC ARABA PRIZLERINDE ÇALIŞTIRMAYIN.
1. Motorlu taşıtınızı çalıştırın.
2. DC güç kaynağını, FreeStyle Comfort’ın sağ alt
tarafından ön kısımdaki güç portu prizine bağlayın.
Bağlantının gerçekleşmesi için içeri doğru itin.

388 - TURPN MN232-C4 K | Kullanıcı El Kitab

4. Güç düğmesine basarak FreeStyle Comfort’ı açınve akışı artırma (+) veya azaltma (-) düğmelerini
kullanarak reçete edilmiş olan akış ayarınızı seçin.

Not: Pil takımının şarj edilmesi gerekirse ekrandaki pilsimgesinde tek bir turuncu çubuk olacaktır.

FreeStyle Comfort
Temizlik, Özen ve Düzenli Bakım
RUTİN BAKIM

PİL TAKIMININ BAKIMI

Pil takımının sıvıya maruz kalmasına izin vermeyin.

Önleyici bakım
Cihaz bakım gerektirmez. Cihazın içindeki filtreler,
cihazın kullanım ömrünü uzatmak üzere tasarlanmıştır.
Cihaz düzgün çalışmıyorsa, ünite onarım içintedarikçiye iade edilmelidir. FreeStyle Comfort
kabininin içindeki dahili bileşenlerin bakımı, oksijensağlayıcısı tarafından yapılmalıdır.

Pil takımını sökmeye veya açmaya çalışmayın.

Ünitenin operatörü, şu bakım prosedürlerinden
sorumludur.
• Haftalık - Hava Girişi Filtrelerinin Temizlenmesi

Pil takımınızın zarar gördüğü anlaşılırsa üniteyi
kullanmadan önce oksijen sağlayıcınızla görüşün.

UYARI: OKSIJEN YOĞUNLAŞTIRICININ
PLASTIK KAPAĞINI AÇMAYA VEYAÇIKARMAYA ÇALIŞMAYIN. ÜNITEYALNIZCA GEREKLI YETKINLIĞE SAHIPBIR SERVIS TEKNISYENI TARAFINDANAÇILMALIDIR.

Pili açık aleve maruz bırakmayın.
Pil takımını düşürmeyin.
Pil takımını çocuklardan uzak tutun.

Pili mikrodalga fırın veya başka bir pişirme cihazının
yakınına koymayın.

PİL TAKIMININ SAKLANMASIOksijen yoğunlaştırıcı pilinin şarj edilmeden öncetamamen boşalması gerekmez. Oksijen yoğunlaştırıcı
pilinin her kullanımdan sonra şarj edilmesi önerilir.

HAVA GİRİŞİ FİLTRELERİ

Pili, doğrudan güneş ışığı veya bir otomobilin içi gibiaşırı sıcaklığa maruz kalacağı bir yerde bırakmayın.
Bunu yapmak, pilin performansının düşmesine veömrünün kısalmasına yol açabilir. Ayrıca pilin ısı veya
duman yaymasına, tutuşmasına veya patlamasına
yol açabilir. Pil takımı, düşük nemin (%70’ten düşük
bağıl nem) olduğu ve paslandırıcı gazların olmadığı
kararlı bir ortamda saklanmalıdır. Paslanmayı önlemekiçin ortam sıcaklığında hızlı dalgalanmalar gibiyoğunlaşmaya yol açabilecek durumlardan kaçının.

1. İlgili tırnağa bastırıp dışarı çekerek iki filtreyi
FreeStyle Comfort’ın ön kısmından çıkarın.

Bir ay veya daha uzun süreli saklama için boşaltılmış
veya %50 ile %75 arası (iki veya üç LED) kısmen
şarj edilmiş durumda saklama önerilir. Şarj edildikten
sonra, pili oksijen yoğunlaştırıcıdan çıkarın.

Hava, oksijen yoğunlaştırıcının alt tarafından, ön
kısımdaki hava girişi filtreleriyle FreeStyle Comfort’agirer. Bu filtreler, tozun ve havadaki diğer büyük
partiküllerin üniteye girişini engeller.
Hava girişi filtrelerinin haftada en az bir kez
temizlenmesi gerekir. Oksijen sağlayıcınız,çalışma koşullarına bağlı olarak filtreleri daha sık
temizlemenizi önerebilir. Hava girişi filtrelerini doğru
şekilde temizlemek için bu adımları uygulayın.

2. İki filtreyi de ılık sabunlu suda yıkayın.
3. Filtreleri durulayın ve kalan suyu tüy bırakmayanyumuşak bir havluyla silin. Filtreleri yerleştirmedenönce filtrelerin tamamen kuru olduğundan emin olun.
4. Temiz ve kuru filtreleri tekrar takın.
UYARI: OKSIJEN YOĞUNLAŞTIRICIYI,HAVA GIRIŞI FILTRESI TAKILI OLMADAN
ÇALIŞTIRMAYIN.

PN MN232-C4 K | Kullanıcı El Kitab

TUR - 389

FreeStyle ComfortTEMİZLEME

Her türlü temizleme önce üniteyi KAPALI konuma
getirin ve AC veya DC gücü bağlantısını kesin. Dışmuhafazaya doğrudan ürün PÜSKÜRTMEYİN. Nemli(ıslak olmayan) bir bez veya sünger kullanın. Kabini
ve güç kaynaklarını temizlemek için beze veya süngere
hafif bir ev temizleyici çözeltisi püskürtün. Temizlik
malzemesi üreticisinin talimatlarına uygun olarak
devam edin.
UYARI: ÜNİTEYİ TEMİZLEMEDEN ÖNCE,KAZARA ELEKTRİK ÇARPMASI VE YANMATEHLİKESİNİ ÖNLEMEK İÇİN ÜNİTEYİ
KAPATIN VE GÜÇ KABLOSUNU ELEKTRİK
PRİZİNDEN ÇIKARIN. YALNIZCA EKİPMAN
SAĞLAYICINIZ VEYA YETKİLİ BİR SERVİS
TEKNİSYENİ ÜNİTENİN KAPAKLARINIÇIKARMALI YA DA ÜNİTE ÜZERİNDE
SERVİS İŞLEMİ GERÇEKLEŞTİRMELİDİR.
UYARI: OKSİJEN YOĞUNLAŞTIRICININ VE PİL
TAKIMININ ISLANMAMASINA VE ÜNİTEYE SIVIGİRMEMESİNE ÖZEN GÖSTERİLMELİDİR. BU DURUM
ARIZAYA YA DA KAPANMAYA YOL AÇABİLİR VE
ELEKTRİK ÇARPMASI VEYA YANMA RİSKİNDE
ARTIŞA NEDEN OLABİLİR.
UYARI: DOĞRUDAN ÜNİTE ÜZERİNDE SIVIKULLANMAYIN. İSTENMEYEN KİMYASAL MADDELER,
BİR SINIRLAMA OLMAKSIZIN ŞUNLARI İÇERİR:ALKOL VE ALKOL BAZLI ÜRÜNLER, KONSANTREKLOR BAZLI ÜRÜNLER (ETİLEN KLORÜR) VE YAĞBAZLI ÜRÜNLER. ÜNİTEYE ZARAR VEREBİLECEK BUÜRÜNLER YOĞUNLAŞTIRICI ÜZERİNDEKİ PLASTİKGÖVDEDE KULLANILMAMALIDIR.

390 - TURPN MN232-C4 K | Kullanıcı El Kitab

UYARI: KABİN, KONTROL PANELİ VE GÜÇ
KABLOSUNU YALNIZCA NEMLİ (ISLAK OLMAYAN)BİR BEZ VEYA SÜNGER KULLANARAK EV TİPİHAFİF BİR TEMİZLİK MADDESİYLE TEMİZLEYİN
VE ARDINDAN TÜM YÜZEYLERİ KURULAYIN.
YOĞUNLAŞTIRICIYA SIVI GİRMESİNE İZİN VERMEYİN.
KANÜL BAĞLANTISININ OKSİJEN ÇIKIŞININ TOZ, SU
VE PARTİKÜLLERDEN ARINMIŞ OLDUĞUNDAN EMİNOLMAK İÇİN BU ÇIKIŞA ÖZELLİKLE ÖZEN GÖSTERİN.
UYARI: PİL TAKIMINI TEMİZLEMEK İÇİN NEMLİ (ISLAKOLMAYAN) BİR BEZ VEYA SÜNGER KULLANIN. ÖNCEBEZE VEYA SÜNGERE EV TİPİ HAFİF BİR TEMİZLİKMADDESİ PÜSKÜRTÜN VE PİL TAKIMININ DIŞ
MUHAFAZASINI TEMİZLEYİN.
UYARI: OTOMATIK TEMIZLIK SISTEMIYLE ETMEYINTEMIZLEMEYIN.

KANÜLÜN DEĞİŞTİRİLMESİ

Doğru kullanım için daima kanül üreticisinin
talimatlarını izleyin. Burun kanülünü veya oksijen
hortumunu, kanül üreticisinin veya oksijensağlayıcınızın önerilerine uygun olarak üreticininönerisine göre 3 ila 6 ayda bir (her gün kullanılıyorsa
3 ayda bir) değiştirin. Temizleme ve değiştirme
talimatlarını, hekiminiz veya oksijen sağlayıcınızverecektir.
İlave değiştirme kaynakları, oksijen sağlayıcınız
aracılığıyla sağlanır.

FreeStyle Comfort
Alarm DurumlarıÖncelikNo.

Alarm
Açıklaması

1

Sistem Arızası
Yüksek Pil
Sıcaklığı

Ekran

Sesli SinyalDiğer Ayrıntılar/Önerilen EylemHer
20 saniyede
1 bip sesiÜnite, pil sıcaklığı emniyet kapatma eşiğine
yaklaşırken alarm verecektir.
Kapatma eşiğine ulaşılırsa ünite
kapatılacaktır. Harici güç bağlanırsa alarm
ekranı görünecek ve sesli alarm 2 dakika
boyunca devam edecektir.
Pili oda sıcaklığında test edin. Sorun devam
ederse hizmet sağlayıcınızla iletişime geçin.

2

Sistem Arızası
Yüksek
SıcaklıkHer
20 saniyede
1 bip sesiÜnite, dahili sıcaklık kapatma eşiğine
yaklaşırken alarm verecektir.
Kapatma eşiğine ulaşılırsa ünite kapatılacak,
alarm ekranı görünecek ve sesli alarm 2
dakika boyunca devam edecektir.
Ünitenin deliklerinin herhangi bir şekilde
tıkalı olmadığından emin olun. Sorun devam
ederse hizmet sağlayıcınızla iletişime geçin.

3

Sistem Arızası
Yüksek BasınçHer
20 saniyede
1 bip sesi

Kapatma eşiğine ulaşılırsa ünite kapatılacak,
alarm ekranı görünecek ve sesli alarm 2
dakika boyunca devam edecektir.
Sorun devam ederse hizmet sağlayıcınızla
iletişime geçin.

4

Nefes
Algılanmadı

Her
30 saniyede
1 bip sesi

15 saniye boyunca nefes algılanmadı.
İlk nefesten veya ısınma ekranı kapatıldıktan
(hangisi önce gerçekleşirse) sonra
etkinleştirilir.
Kanülde herhangi bir bükülme veya sızıntı
olup olmadığını kontrol edin. Hassasiyet
ayarını artırır. Sorun devam ederse hizmetsağlayıcınızla iletişime geçin.

5Düşük Pil
Nedeniyle
KapatmaHer
30 saniyede
1 bip sesi

Pil gücü tükenene ve ünite kapanana dek
alarm verilir.

6

IsınmayokÜnite, nefes algılanana veya ısınma sonaerene (hangisi önce gerçekleşirse) dek
otomatik dozlamayla başlayacaktır.

7

Düşük
Yoğunluk

En fazla
5 dakikaboyunca her
30 saniyede
1 bip sesi

Sorun devam ederse hizmet sağlayıcınızla
iletişime geçin.

Pili şarj edin.

PN MN232-C4 K | Kullanıcı El Kitab

TUR - 391

FreeStyle ComfortÖncelikNo.

Alarm
Açıklaması

8

Ekran

Sesli SinyalDiğer Ayrıntılar

Sistem Arızası
Akış Hızı(OCD Değeri
Hatası, Gaz
YolununTıkanması)Her
30 saniyede
1 bip sesi

Sorun devam ederse hizmet sağlayıcınızla
iletişime geçin.

9Düşük Pil
Uyarısı

1 bip ses,
tekrarsız

Pili şarj edin.

10

Nefes Hızı
Aşıldı

Her
60 saniyede
1 bip sesi

Nefes Hızı 40 bpm’yi aşıyor.

11

Sistem Arızası
Fan Hatası

yok

Sorun devam ederse hizmet sağlayıcınızla
iletişime geçin.

12

Sistem Arızası
Düşük Basınçyok

Sorun devam ederse hizmet sağlayıcınızla
iletişime geçin.

13

Sistem Arızası
Düşük Sıcaklık

yokÜnitenin oda sıcaklığına kadar ısınmasınısağlayın. Sorun devam ederse hizmetsağlayıcınızla iletişime geçin.

BilgiBaşlangıç

1bilgilendirici
bip sesi

Ekran açılışta görünür ve elektrik zilininçalıştığını onaylayan bir bip sesi duymanız
gerekirDoz yanlış bir şekilde tetikleniyorsa
hassasiyet ayarını azaltın. Sorun devam
ederse hizmet sağlayıcınızla iletişime geçin.

Başlangıç ekranı, sağ alt köşede geçerli
ürün yazılımı revizyonunu gösterir.
Bilgi

autoSAT
ModundaNormalÇalışmaEkranı (DCGücü bağlı
değil)

392 - TURPN MN232-C4 K | Kullanıcı El Kitab

yokFreeStyle ComfortÖncelikNo.

Alarm
Açıklaması

Ekran

Sesli Sinyal

Bilgi

autoSAT
devre dışıyken
NormalÇalışmaEkranı (DCGücü bağlı
değil)

Yok

Bilgi

Ayar
Değiştirildi

1bilgilendirici
bip sesi

Bilgi

Kapalı
durumunda
harici Güçbağlayın

yok

Bilgi

Çalışmadurumunda
harici güçbağlayın

yok

Bilgi

Kapalı
durumunda
harici gücü
kesin

1bilgilendirici
bip sesi

BilgiÇalışmadurumundaki
harici gücü
kesin (pilvarken)1bilgilendirici
bip sesiDiğer Ayrıntılar

2 saniye boyunca CAIRE ekranı ve ardından
ekran kapatılmadan önce 5 saniye boyunca
Pil ve Fiş durumu görünür

2 saniye boyunca CAIRE ekranı görünür ve
ardından ekran kapatılır.

PN MN232-C4 K | Kullanıcı El Kitab

TUR - 393

FreeStyle ComfortÖncelikNo.

Alarm
Açıklaması

Bilgi

Ekran

Sesli
SinyalDiğer Ayrıntılar

BeklemeModunda- yukarıveya aşağı
düğmesinebasıldığında,ekran pili veşarj durumunu
gösteriryok

Ekran, 5 saniye boyunca görünüp kapatılır

Bilgi

Pil gücünde
Uçak Modu

1 uzun bip
sesi

Bilgi

Prize
takılıyken
Uçak Modu

1 uzun bip
sesi

394 - TURPN MN232-C4 K | Kullanıcı El Kitab

FreeStyle Comfort
Menü Gezinme Talimatları
Bekleme veya Çalışma modunda Menüye
girebilirsiniz. Çalışma Modundayken bir alarm
durumu oluştuğunda menüden otomatik olarak çıkılır.
20 saniye boyunca herhangi bir düğmeye basılmazsa
Menüden otomatik olarak çıkılır. Bekleme modunda,üniteyi uyandırmak için Menü düğmesine iki kez
basmanız ve ardından Menüye girmeniz gerekir.
Dört adet Menü ekranı vardır:
1. Durum ekranı: Ürün yazılımı revizyon numarasını,çalışma süresini, pil boşalma döngülerini gösterir

Uçak Modu

2. Uçak modu (AÇIK/KAPALI)
3. Hassasiyet: ayarlanabilir (1-5)
4. autoSAT (AÇIK/KAPALI)

HassasiyetÖrnek Durum Ekranı

AYARLAR MENÜSÜNÜ ÇALIŞTIRMA
Ayarlar menüsüne, Çalışma veya Bekleme
modlarındayken Menü/Seç düğmesinebasılarak
girilir.
Menülerde gezinmek için

autoSATdüğmesine basın.

Durum Sayfası

PN MN232-C4 K | Kullanıcı El Kitab

TUR - 395

FreeStyle Comfort
DURUM SAYFASIEkrandaki değer yerine siyah renkli “AÇIK” görünür.

Serbest Stil Konforunun Durum Sayfası, takılı pilüzerindeki Bellenim Revizyonunu, Çalışma Saatlerini ve pil döngüsü sayısını gösterir.

Uçak Modunu iptal etmek için İPTAL simgesi
görünürken
düğmesine basın.

Uçak Modu Açık Onaylı
Durum SayfasıÇalışma saatleri tam saat olarak gösterilir.
Pil Deşarjları, takılı pilin boşaltıldığı ve yenidenşarj edildiği döngü sayısıdırUÇAK MODUFreeStyle Comfort, Uçak Modu kapalı bir şekilde gelir.

Uçak Modunu kapatmak için KAPALI seçeneğine
geçerken yukarıdaki adımları tekrarlayın.
Ekranda “İleri” simgesi görünürken
düğmesinebasıldığında, menü sayfası sonraki ekran seçeneğine
geçer.
Uçak Modu, Çalışma Modunda kullanım içindir. Ünite
pille çalıştırılırken Bekleme modunda olması halinde,
Uçak Modu otomatik olarak devre dışı kalacaktır.

Not: Uçak Modu hakkında daha fazla bilgi almak için
Uçak Modunu Etkinleştirme bölümüne bakın.

HASSASIYET MODU

Uçak Modu Ana Ekranı

Bu modu açmak için DEĞİŞTİR simgesi görünürken
düğmesine basın. Ekrandaki değer yerine mavirenkli “AÇIK” görünür.

FreeStyle Comfort, fabrikadan Hassasiyet Modu 2’eayarlı bir şekilde gönderilir ve normal koşullarda
herhangi bir ayar gerektirmez. Hassasiyet Ayarı,
1-5 arasında düzenlenebilir. Burada 1 en düşük
hassasiyet, 5 ise en yüksek hassasiyettir.

2. Ayarda Hassasiyet Modu Ekranı
Uçak Modu Açık Seçili

Uçak Modunu açmayı onaylamak için KAYDET
simgesi görünürken
düğmesine basın.

396 - TURPN MN232-C4 K | Kullanıcı El Kitab

Hassasiyet Modunu ayarlamak için DEĞİŞTİR simgesi
görünürken
düğmesine basın. Ekrandaki değer
yerine bir sonraki mavi sayı görünür.

FreeStyle Comfort
AUTOSAT MODUFreeStyle Comfort, autoSAT Modu açık bir şekilde
gelir.

Hassasiyet Modu Ayarı 3 Seçili

Hassasiyet Modunda 3. Ayarı onaylamak için
KAYDET simgesi görünürken
düğmesine basın.
Ekrandaki değer yerine siyah renkli “3” görünür.
Hassasiyet Modu değişimini iptal etmek için İPTAL
simgesi görünürken
simgesine basın.

autoSAT Modu Ana Ekranı

Bu modu kapatmak içi, DEĞİŞTİR simgesi
görünürken
düğmesine basın. Ekrandaki değer
yerine mavi renkli “KAPALI” görünür.

Hassasiyet Modu Ayarı 3 Onaylı
autoSAT Modu Kapalı Seçili

autoSAT Modunun kapalı olduğunu onaylamak için
KAYDET simgesi görünürken
düğmesine basın.
Ekrandaki değer yerine siyah renkli “KAPALI”
görünür.
autoSAT Modunu iptal etmek için İPTAL simgesi
görünürken
düğmesine basın.
Hassasiyet Modu Ayarı 5 (Maks) Onaylı

Ekranda “İleri” simgesi görünürken
düğmesinebasıldığında, menü sayfası sonraki ekran seçeneğine
geçer.

autoSAT Modu Kapalı Onaylı

autoSAT Modunu açmak için AÇIK seçeneğine
geçerken yukarıdaki adımları tekrarlayın.
Ekranda “İleri” simgesi görünürken
düğmesinebasıldığında, menü sayfası sonraki ekran seçeneğine
geçer.

PN MN232-C4 K | Kullanıcı El Kitab

TUR - 397

FreeStyle Comfort

Sorun Giderme
Sorun

Olası NedenÇözümFreeStyle Comfort alarm verir
ve kesintili bip sesi çalar.

Alarm Durumları tablosuna
bakın.

Alarm Durumları tablosuna bakın.

Oksijen yoğunlaştırıcı
açılmıyor.

Pil boşalmıştır.

Üniteye AC veya DC çıkışıyla güç verin.

Harici güce bağlanmadı.

Harici bağlantıların güvenli olduğundan emin olun.

Genel arıza.

Pili çıkararak ve AC veya DC harici güç kaynağını
keserek gücü tamamen kapatıp açın. Ardından
pili yeniden takın ve gücü kapatıp açmayı
tamamlayın. İsterseniz harici güç kaynağını yenidenbağlayabilirsiniz.
Ekipman Sağlayıcınızla iletişime geçin ve gerektiği
şekilde başka bir oksijen kaynağına geçiş yapın.

Tüm diğer sorunlar.

Pili çıkararak ve AC veya DC harici güç kaynağını
keserek gücü tamamen kapatıp açın. Ardından
pili yeniden takın ve gücü kapatıp açmayı
tamamlayın. İsterseniz harici güç kaynağını yenidenbağlayabilirsiniz.
Mevcut olan başka bir oksijen kaynağına geçiş
yapın ve Ekipman Sağlayıcınızla iletişime geçin.

Not: Bazı alarmlar, ünitenin kapatılmasına yol açacaktır.
Not: Tam bir güç kaybı sırasında alarm sistemi sıfırlanır (önceki alarmlar silinir) ve yalnızca güç tekrar verildikten
sonra durum hala devam ediyorsa bir alarm verilir.

398 - TURPN MN232-C4 K | Kullanıcı El Kitab

FreeStyle Comfort
Seyahat Bilgileri

1. Hiçbir harici güç kaynağı bağlı olmadan
YALNIZCA Pil Gücünü kullanarak POC’yi açın.

FreeStyle Comfort, ticari uçaklarda kullanım için FAA
yönergelerini karşılamaktadır. Her türlü seyahatten(hava, deniz, demiryolu vs.) önce:• Yoğunlaştırıcının iyi ve çalışır durumda olduğundanemin olun.
• Hareket etmeden en az 48 saat önce seyahatşirketinizle iletişime geçerek, FreeStyle Comfortünitesiyle seyahat edeceğinizi bildirin.
Seyahatlerde ekstra pil taşımanız önerilir. Havayolu
ile seyahat durumunda, yerde geçirilen zamanı (uçuşöncesi giriş, güvenlik vs.) ve aktarma için bekleme
süresini de göz önünde bulundurarak yeterli pil ömrüsağlayın. Çoğu havayolu şirketi, öngörülen maksimum
yolculuk süresinin en az %150’si (uçuş süreleri, uçuşöncesi ve sonrası yerde geçirilen süre, aktarmalarve beklenmedik gecikmeler) kadar bir süre boyunca
cihaza güç verecek kadar yeterli sayıda tam dolu pil
takımı bulundurmanızı gerekli kılar.

Not: Uçuş sırasında ek pillerin güvenli bir taşıma
çantasında taşınması önerilir.

FAA DÜZENLEMELERİ

Yedek pillerin, orijinal satış ambalajına veya bu
mümkün olmadığında terminallerin yalıtılmasıyla,yani açıkta duran terminallerin üstü bantlanarak veya
pillerin her biri ayrı bir plastik poşete ya da koruyucu
bir keseye koyularak ve yalnızca el bagajında taşınarak
kısa devrelere karşı ayrı ayrı korunması gerekir.

2. Aşağıdaki yöntemlerden birini kullanarak Uçak
Modunu etkinleştirin:a. Uçak Modunu etkinleştirmek için Menü Gezinme
Talimatları bölümündeki talimatlara uyun.

b. Uzun bir bip sesi duyana ve Uçak sembolü
görene kadar Yukarı ve Aşağı düğmelerini 5 saniye
boyunca birlikte basılı tutun.

UÇAK MODUNU ETKİNLEŞTİRME
Pil Gücü
POC, hiçbir harici güç kaynağı bağlanmadan
YALNIZCA Pil Gücü kullanılarak normal çalışma
prosedürlerine göre uçakta kullanılabilir

3. Uçak sembolü göründüğünde POC’yi prize takın.

Harici Güç
Uçakta elektrik prizi varsa, çalışır durumdaysa
ve konumu POC güç kaynağı olarak kullanılması
için uygunsa priz, POC pillerinin yedeği olarak
kullanılabilir. Bu durumda, pilden uçaktaki elektrik
gücüne geçişle ilgili olarak aşağıda önerilen
prosedürün uygulanması gerekir:

Not: Uçaklarla ilgili sınırlamalar nedeniyle, piller Uçak
Modunda şarj olmaz.

PN MN232-C4 K | Kullanıcı El Kitab

TUR - 399

FreeStyle Comfort

Aksesuarlar

Uygun performans ve güvenlik için yalnızca listede
belirtilen ve oksijen sağlayıcınız aracılığıyla CAIRE
tarafından tedarik edilen aksesuarları kullanın. Aşağıda
listelenenlerin dışında aksesuarların kullanılması,yoğunlaştırıcının performansını ve/veya güvenliğini
olumsuz etkileyebilir.

ÜNİTENİN TAŞIMA ÇANTASINAYERLEŞTİRİLMESİ
1. Filtreleri ve pil boşluklarını hizalayarak üniteyiçantanın dibine yerleştirin.

FreeStyle Comfort Standart AksesuarlarıTaşıma Çantası

MI396-1

Sırt Çantası

MI459-1

Omuz Askısı

MI397-1

AC Güç Kaynağı (Tüm Kablolar
Dahildir)

PW036-1SDC Güç Kablosu

CD041-1

Harici Büyük Partikül Filtresi (Sağ)

FI226-1

Harici Büyük Partikül Filtresi (Sol)FI227-1

2. Çantanın dibindeki fermuarı kapatın.

Mevcut Opsiyonel Aksesuarlar
Ek 8 Hücreli Pil Takımı

BT034-1S

16 Hücreli Pil Takımı

BT035-1S

Pil Takımı Masaüstü Şarj Cihazı

BT036-1

Tüm Parçalar için Aksesuar Çantası

MI372-2

Not: Yukarıda belirtildiği durumlarda, ülkelere özel güç
kabloları için ilave seçenekler mevcut olabilir. Sipariş
için alternatif seçenekler gerektiğinde CAIRE veya
oksijen sağlayıcınızla görüşün.

Taşıma Çantası Talimatları
FreeStyle Comfort ünitenizi taşıma çantasınayerleştirmek ve sabitlemek için aşağıdaki talimatlarıyerine getirin.

400 - TURPN MN232-C4 K | Kullanıcı El Kitab

OMUZ ASKISININ TAKILMASI
1. Omuz askısı, doğrudan üniteye veya taşıma
çantasına takılabilir.
2. Askıyı, ünite veya çantadaki halkaya geçirin.

FreeStyle Comfort
Elden Çıkarma ve Yeniden Kullanım
FreeStyle Comfort ve tüm bileşenlerini, atılmak üzere
mutlaka ekipman hizmeti sağlayıcınıza iade edin. Pilindoğru bir biçimde elden çıkarılması için şehrinizdeki
yerel idari birimlerle de iletişime geçebilirsiniz. Ünite
insanlar arasında yeniden kullanılabilir, ancak yenidenişleme için ekipman sağlayıcısına iade edilmelidir.

WEEE ve RoHS

Bu sembol, ekipmanın sahiplerine ekipmanı
kullanım ömrü sonunda Atık Elektrikli ve
Elektronik Ekipmanlar (WEEE) Direktifine
uygun olarak bir geri dönüşüm tesisine iadeetmelerini hatırlatmaya yöneliktir.

Ürünlerimiz, Tehlikeli Maddelerin Kısıtlanması(RoHS) direktifine uyar. Eser miktardan fazla kurşun
veya başka tehlikeli malzeme içermezler.
DİKKAT: Doğru bir imha işlemi için,imha talimatlarıyla ilgili olarak ekipmansağlayıcınız veya yerel idari birimle
iletişime geçin

Operatörle Doğrudan Temas Halindeki MalzemelerYoğunlaştırıcı Çantası.................................................................................................................................. PC/ABSYoğunlaştırıcı Kontrol Paneli.................................................................................... Polyester EBG7 veya eşdeğeri
Yoğunlaştırıcı Sapı................................................................................................................................ Polikarbonat
Oksijen Çıkışı, Ağız.............................................................................................................................. PolikarbonatÜnite Etiketi................................................................................................................................................... Leksan
Kablo Bağlantıları..............................................................................................................Polikarbonat/Vinil klorür
Güç Kablosu............................................................................................................................................PVC, Metal
Güç Kaynağı......................................................................................................................Lexan 940(Polikarbonat)
Pil Takımı..................................................................................................................................................... PC/ABS
Pil Takımı, Güç Kaynağı Etiketleri..................................................................................................... Polyester filmYoğunlaştırıcı Taşıma Çantası................................................................................Cordura, Neopren, Naylon Astar
Askı....................................................................................................Polipropilen, Asetil, Vinil, Cordura, Neopren

EMC Testi
Yönergeler ve Üretici Beyanı­- Elektromanyetik Emisyonlar
FreeStyle Comfort ünitesinin, aşağıda belirtilen elektromanyetik ortamda kullanımı amaçlanmıştır. FreeStyle
Comfort müşterisi veya kullanıcısı, ünitenin bu tür bir ortamda kullanılmasını sağlamalıdır.
Emisyon Testi
RF emisyonları
CISPR 11
RF emisyonları
CISPR 11
Harmonik emisyonlar
IEC 61000-3-2
Voltaj dalgalanmaları/
titreme emisyonları IEC
61000-3-3

Uyumluluk

Elektromanyetik Ortam - Yönerge

Grup 1FreeStyle Comfort, RF enerjisini yalnızca dahili fonksiyonu için kullanır.
Bu nedenle, RF emisyonları oldukça düşüktür ve bu emisyonların,
yakındaki elektronik ekipmanla herhangi bir parazite neden olması
beklenmez.

Sınıf B
Sınıf A
Uyumludur

FreeStyle Comfort, konutlar ve konutsal amaçlarla kullanılan binalara
tedarik sağlayan kamusal düşük voltajlı güç kaynağı şebekesinebağlı olan tesisler de dahil olmak üzere, tüm tesislerde kullanım içinuygundur.

PN MN232-C4 K | Kullanıcı El Kitab

TUR - 401

FreeStyle Comfort
Yönergeler ve Üretici Beyanı-Elektromanyetik Bağışıklık

FreeStyle Comfort ünitesinin, aşağıda belirtilen elektromanyetik ortamda kullanımı amaçlanmıştır. FreeStyle Comfortmüşterisi veya kullanıcısı, ünitenin bu tür bir ortamda kullanılmasını sağlamalıdır.
Bağışıklık testi
IEC 60601 test düzeyi
Uyumluluk düzeyi
Elektromanyetik ortam - yönerge
Zemin ahşap, beton veya seramik karo
Elektromanyetik±8 kV kontakt±8 kV kontakt
olmalıdır. Zemin sentetik materyalle
ortam - yönerge
kaplanmışsa bağıl nem en az %30±15 kV hava tahliyesi±15 kV hava tahliyesi
IEC 61000-4-2
olmalıdır.
güç besleme hatları için
güç besleme hatları için
Elektriksel hızlı geçiş/ ±2 kVŞebeke gücü, tipik bir profesyonel sağlık
±2 kV
patlamatesisinin ve evde bakım ortamlarının
kalitesinde olmalıdır.
IEC 61000-4-4giriş/çıkış hatları için
Yok
±1 kV
hattan hatta ±1 kV

Dalgalanma
IEC 61000-4-5

AC hatlarında ±2 kV
ortak modhattan hatta ±1 kV

AC hatlarında ±1 kV
diferansiyelŞebeke gücü, tipik bir profesyonel sağlık
tesisinin ve evde bakım ortamlarının
kalitesinde olmalıdır.

Dış mekan G/Ç hatlarında±2 kV ortak mod0,5 döngü (0°, 45°, 90°, 0,5 döngü (0°, 45°, 90°,135°, 180°, 225°, 270°,
135°, 180°, 225°, 270°,315°) için %0 UT315°) için %0 UT

Güç beslemegiriş hatlarında
gerilim düşmeleri,kısa kesintiler vegerilim değişimleri
IEC 61000-4-11

Şebeke gücü, tipik bir profesyonel sağlık
tesisinin ve evde bakım ortamlarının
kalitesinde olmalıdır. FreeStyle Comfort
1 döngü (00) için %0 UT
1 döngü (00) için %0 UTkullanıcısı için şebeke elektriği kesintileri
sırasında çalışmanın sürdürülmesi
25/30 döngü (00) için %70 25/30 döngü (00) için %70 gerekiyorsa FreeStyle Comfort ünitesine
T
T
UT (U ’de %30 düşüş)UT (U ’de %30 düşüş)kesintisiz güç kaynağından veya pilden güç
verilmesi önerilir.
250/300 döngü (00) için
%0 UT

250/300 döngü (00) için
%0 UT

Güç frekansı manyetik alanları, profesyonelsağlık tesisi ve evde bakım ortamlarındabulunan tipik bir konuma özgü düzeylerde
olmalıdır.
NOT: UT, test düzeyinin uygulanmasından önce A.C. şebeke gerilimini ifade eder.

Güç frekansı(50/60 Hz) manyetik
alanı IEC 61000-4-8

3 A/m

3 A/m

NOT Bazı durumlarda, Freestyle Comfort önemli bir Elektrostatik Boşalmadan sonra sıfırlanabilir ve normal çalışmaya
dönmesi için gücün tamamen kapatılıp yeniden açılması gerekebilir. Daha fazla ayrıntı için Sorun Giderme bölümüne bakın.

402 - TURPN MN232-C4 K | Kullanıcı El Kitab

FreeStyle Comfort
Yönergeler ve Üretici Beyanı-Elektromanyetik Bağışıklık

FreeStyle Comfort ünitesinin, aşağıda belirtilen elektromanyetik ortamda kullanımı amaçlanmıştır. FreeStyle Comfortmüşterisi veya kullanıcısı, ünitenin bu tür bir ortamda kullanılmasını sağlamalıdır.
Bağışıklık testi
IEC 60601 test düzeyi
Uyumluluk düzeyi
Elektromanyetik ortam - yönergeİletilen RF
10 Vrms
10 VrmsTaşınabilir ve mobil RF iletişim ekipmanı,
kablolar da dahil olmak üzere FreeStyle
IEC 61000-4-6
150 kHz - 80 MHz
Comfort ünitesinin herhangi bir parçasına,vericinin frekansı için geçerli denklem
kullanılarak hesaplanmış önerilen ayrım
mesafesinden yakın olmayacak şekildeIşınan RF
10 V/m
10 V/mkullanılmalıdır.
IEC 61000-4-3
80 MHz - 2.700 MHzÖnerilen= , ayrım mesafesid  1.2 P= ,d  1.2 P = 80 MHz - 800 MHz= ,d  1.2 P = 800 MHz - 2700 MHz
Burada P, verici üreticisine göre vat (W)cinsinden vericinin maksimum çıkış gücü
derecesidir ve d, metre (m) cinsindenönerilen ayrım mesafesidir.
Sabit RF vericilerinin, elektromanyetik saha
etüdüyle belirlenen alan kuvvetleri,a her bir
frekans aralığında uyumluluk düzeyindendüşük olmalıdır. b
Şu sembolle işaretlenmiş ekipmanın
yakınında parazit oluşabilir:

NOT 1: 80 MHz ve 800 MHz’de, daha yüksek frekans aralığı geçerlidir.
NOT 2: Bu yönergeler her durumda geçerli olmayabilir. Elektromanyetik yayılım; yapılar, nesneler ve insanlar tarafından
gerçekleştirilen absorpsiyondan ve yansımadan etkilenir.
aRadyo frekansı ile çalışan telefonların (cep telefonları/kablosuz telefonlar) baz istasyonları ve hareketli kara telsizleri,amatör telsizler, AM ve FM radyo yayını ve TV yayını gibi sabit vericilerden kaynaklanan alan kuvvetleri, teorik olarakdoğru bir biçimde tahmin edilemez. Sabit RF vericilerinden kaynaklanan elektromanyetik ortamı değerlendirmek için bir
elektromanyetik saha araştırmasının yapılması göz önünde bulundurulmalıdır. FreeStyle Comfort ünitesinin kullanıldığı
konumdaki alan kuvveti, yukarıdaki geçerli RF uyumluluk düzeyini aşıyorsa FreeStyle Comfort normal çalıştığının
doğrulanması için gözlemlenmelidir. Anormal performans gözlemlenirse, FreeStyle Comfort ünitesinin yönünün veya
konumunun değiştirilmesi gibi ek önlemler gerekli olabilir.
b

150 kHz ile 80 MHz frekans aralığında alan kuvvetleri 10 V/m’den düşük olmalıdır.

PN MN232-C4 K | Kullanıcı El Kitab

TUR - 403

FreeStyle ComfortTaşınabilir ve hareketli RF haberleşme ekipmanları ile FreeStyle Comfort arasında
bırakılması önerilen mesafeler

FreeStyle Comfort ünitesinin, ışınan RF bozulmalarının kontrol altında olduğu bir elektromanyetik alanda kullanımı
amaçlanmıştır. FreeStyle Comfort müşterisi veya kullanıcısı, elektromanyetik girişimi önlemeye yardımcı olmak içintaşınabilir ve hareketli RF haberleşme ekipmanları (vericiler) ile FreeStyle Comfort arasında, haberleşme ekipmanının
maksimum çıkış gücüne uygun olarak aşağıda belirtildiği gibi bir minimum mesafe bırakabilir.
Vericinin nominal
Vericinin frekansına göre ayrım mesafesi
maksimum çıkış gücümW

150 kHz - 80 MHzd= 1,.2 P

80 MHz - 800 MHz

d = 1,.2 P

800 MHz - 2,5 GHz

d = 1,.2 P

0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Yukarıda belirtilmemiş bir maksimum çıkış gücüne sahip olduğu belirtilen vericiler için vericinin üreticisinin sağladığı bilgilere
göre P’nin vericinin maksimum çıkış gücünü vat (W) cinsinden temsil ettiği denklem vericisinin frekansına uygulanarak,önerilen ayrım mesafesi olan d, metre (m) cinsinden hesaplanabilir.
NOT 1 80 MHz ve 800 MHz’de, daha yüksek frekans aralığına yönelik ayrım mesafesi geçerlidir.
NOT 2: Bu yönergeler her durumda geçerli olmayabilir. Elektromanyetik yayılım; yapılar, nesneler ve insanlar tarafından
gerçekleştirilen absorpsiyondan ve yansımadan etkilenir.

Sınıflandırma

Elektrik çarpmasına karşı koruma türü:
Sınıf II	Elektrik çarpmasına karşı koruma,çift yalıtımla elde edilir.
Elektrik çarpmasına karşı koruma derecesi:
BF Tipi Ekipman, şunlara ilişkin olarak, elektrik		 çarpmasına karşı belirli bir derecede koruma sağlar:		 1) izin verilen kaçak akım;
		2) koruyucu topraklama bağlantısının (varsa)
güvenilirliği.
		 Doğrudan kardiyak uygulama için tasarlanmamıştır.

	• RTCA/DO160 Bölüm 21 Kategori M uygunluğu
test edilmiştir.
	• CISPR 11 / EN 55011 Sınıf B Grup 1, “Endüstriyel,
Bilimsel ve Tıbbi (ISM) Ekipman”
	• FCC Kısım 15, Alt Kısım B - Sınıf B KasıtsızRadyasyon Yayan Cihazlar
IP22 - Parmakların veya çapı 12,5 mm’den büyük
olan nesnelerin girişine karşı koruma. Dikey olarak
damlayan su, muhafaza normal konumundan 15°’lik
bir açıyla eğildiğinde zararlı bir etkiye neden olmaz.

Tıbbi Elektrikli Ekipman Standardı için bağımsız test:
Şu standartlara uygunluğu test edilmiştir:• IEC 60601-1 sürüm 3.1: Tıbbi Elektrikli
ekipman - Bölüm 1 Genel Güvenlik
Gereksinimleri.
• CAN/CSA C22.2 No. 60601-1-8 sürüm 2.1, Tıbbi
Elektrikli Ekipman - Kısım 1: Genel Güvenlik
Gereksinimleri ile uyumlu bulunmuştur

İzin verilen temizlik ve enfeksiyon kontrolü yöntemi:
Lütfen bu FreeStyle Kullanıcı El Kitabının “Temizlik,
Bakım ve Uygun Servis İşlemleri” bölümünü
inceleyin.

Ekipman ile diğer cihazlar arasında potansiyel
elektromanyetik veya diğer parazite karşı koruma.
	• EN 60601-1-2/IEC 60601-1-2: 4. Sürüm ile uyumlu
olarak test edilmiştir.

Çalışma modu: Sürekli çalışma.

Yanıcı anestezik gazların mevcut olduğu ortamda
uygulama güvenliği derecesi:
Ekipman bu tür bir uygulama için uygun değildir.

CAIRE ve CAIRE Inc., CAIRE Inc.’in tescilli ticari markalarıdır. Ticari
markaların tam listesi için lütfen web sitemizi ziyaret edin. Ticari markalar:www.careinc.com/corporate/trademarks.

404 - TURPN MN232-C4 K | Kullanıcı El Kitab

Telif hakkı © 2021 CAIRE Inc. CAIRE Inc. herhangi bir zamanda, önceden
haber vermeksizin veya ek bir yükümlülük ya da sonuç olmaksızınürünlerinin üretimini durdurma veya ürünlerinin fiyatları, materyalleri,ekipmanı, kaliteyi, tanımları, teknik özellikleri ve/veya süreçlerini değiştirme
hakkını saklı tutar. Bu belgede açıkça ifade edilmeyen tüm haklar, geçerli
olduğu üzere tarafımızca saklıdır.

FreeStyle ComfortUživatelské ovládací prvky a ukazatele stavu systému
ISO 7000: Grafické značky pro použití na zařízeních –rejstřík a přehledPřed použitím si přečtěte uživatelskou
příručku. Reg. č. 1641
Rozsah skladovacích nebo provozních
teplot. Reg. č. 0632

Rozsah vlhkosti při skladování.
Reg. č. 2620

Směrnice Rady 93/42/EHS o zdravotnických
pomůckáchZplnomocněný zástupce v Evropském
společenství
Pokud je na štítku jedinečného
identifikátoru prostředku (UDI)
produktu uveden symbol CE ####,
zařízení splňuje požadavky směrnice93/42 / EHS týkající se zdravotnických
prostředků. Symbol CE #### označuječíslo oznámeného subjektu.
IEC 60417
Zařízení třídy II s dvojitou izolací.
Reg. č. 5172

Omezení atmosférického tlaku.
Reg. č. 2621

Vypínač. Reg. č. 5009

Název a adresa výrobce. Reg. č. 3082
Datum výroby. Reg. č. 2497

Certifikace ETL
Bezpečnostní agentura pro normu
CAN/CSA C22.2, č. 601.1 M90 pro
zdravotnické elektrické přístroje

Katalogové číslo. Reg. č. 2493
Sériové číslo. Reg. č. 2498

Interní symboly

Touto stranou nahoru. Reg. č. 0623

Chraňte před hořlavými materiály, olejia mazivy.

Křehké, manipulujte opatrně.
Reg. č. 0621

Nerozebírejte.

Chraňte před deštěm, uchovávejte
v suchu. Reg. č. 0626

Zvyšte průtok.

Omezení počtu kusů kladených na sebe.
Reg. č. 2403

Snižte průtok.

ISO 7010: Grafické značky – bezpečnostní barvy
a bezpečnostní značky – registrované bezpečnostníznačkyJe nezbytné přečíst příručku k obsluze.
Reg. č. M002Chraňte před otevřeným plamenem,ohněm, jiskrami. Zákaz kouření
a manipulace s plamenem. Reg. č. P003

Tlačítko Menu.
RTCA/DO160, část 21
Category M
CompliantV souladu s předpisem RTCA/DO-160,
část 21, kategorie M. Požadavky FAA
SFAR 106.

V blízkosti jednotky ani při manipulacis ní nekuřte. Reg. č. P002Zařízení třídy II, dvojitá izolacePříložná část typu BF (stupeň ochrany
proti úrazu elektrickým proudem).
Reg. č. 5333

Schváleno úřadem FAA – PKK(přenosný kyslíkový koncentrátor)Varování. Reg. č. W001

PN MN232-C4 K | Uživatelská příručka

CZE – 405

FreeStyle Comfort
21 CFR 801.15: Sbírka federálních předpisů, hlava 21
Tato zařízení smí být podle federálních
zákonů prodávána pouze lékaři nebo na
jejich příkaz.
Směrnice Rady 2012/19/EU: o odpadních elektrických
a elektronických zařízeních (OEEZ)OEEZ

Co je kyslíkový koncentrátor
Vzduch, který dýcháme, obsahuje přibližně
21 % kyslíku, 78 % dusíku a 1 % dalších plynů.
Zařízení FreeStyle® Comfort® funguje tak,že je prostorový vzduch nasáván do přístroje
prostřednictvím otvorů pro přívod vzduchu. Následně
prochází přes adsorpční materiál zvaný molekulové
síto. Tento materiál oddělí kyslík od dusíku a umožní
pouze průchod kyslíku. Výsledkem je proud vysoce
koncentrovaného kyslíku dodávaný uživateli.
Přístroj FreeStyle Comfort kombinuje vyspělý
kyslíkový koncentrátor s technologií hospodaření
s kyslíkem za účelem vytvoření přenosného zařízení,které umožní maximální přenosnost při chůzi.
Pokrokový systém dodávání kyslíku prostřednictvím
pulzního proudění rychle rozezná, kdy se uživatel
nadechuje, a na počátku každého nádechu poskytne
proud kyslíku. Dodávka pulzním prouděním je
ideální pro různé aktivity a dobu pobytu mimo
domov. Na rozdíl od nepřetržitého proudění lépehospodaří s kyslíkem a prodlužuje životnostbaterie. Přístroj FreeStyle Comfort je tak lehký, že
je možné jej přenášet pomocí popruhu na rameno.
Může být napájen střídavým proudem (ze zásuvky)
i stejnosměrným proudem (ve vozidle) nebo bateriemi.

Zamýšlené použití
Koncentrátor kyslíku CAIRE FreeStyle Comfort jeurčen k podávání doplňkového kyslíku. Zařízení neníurčeno pro podporu života a neposkytuje žádné funkce
pro sledování pacienta.

Proč vám lékař předepsal kyslík
Mnoho lidí trpí různými srdečními, plicními a jinými
dýchacími onemocněními. Významné množství těchto
lidí může profitovat z dodatečné léčby kyslíkem doma,na cestách nebo při každodenních aktivitách mimo
domov.
Vzduch, který dýcháme, je z 21 % tvořen právě tímto
plynem, kyslíkem. Aby lidské tělo správně fungovalo,potřebuje jeho nepřetržitý přísun. Lékař vám předepsal
406 – CZEPN MN232-C4 K | Uživatelská příručka

IEC 60601-1: Zdravotnické elektrické přístroje, část 1:
Všeobecné požadavky na základní bezpečnost
a nezbytnou funkčnost
Chráněno proti kapající voděNa tento výrobek se může v USA i mezinárodně
vztahovat jeden nebo více patentů. Seznam platných
patentů najdete na níže uvedené webové stránce.
Pat.: www.caireinc.com/corporate/patents/.

míru proudění nebo nastavení přístroje podle
konkrétního onemocnění dýchacího ústrojí.
Přestože kyslík není návyková látka, neoprávněná
léčba kyslíkem může být nebezpečná. Před použitím
tohoto kyslíkového koncentrátoru se musíte poradit
s lékařem. Dodavatel kyslíkového přístroje vám
názorně ukáže způsob nastavení předepsanéhoprůtočného množství.

Profil uživatele
Koncentrátory jsou určeny k dodávce doplňkového
kyslíku uživatelům, kteří trpí obtížemi v důsledku
onemocnění ovlivňujícího účinnost jedné z plic připřenosu kyslíku ze vzduchu do krevního oběhu.
Přenosné kyslíkové koncentrátory (PKK) neskladují
ani neobsahují kyslík. Není potřeba je doplňovat a lze
je dobít na každém místě s dostupným střídavým nebo
stejnosměrným napájením (kromě letadel). Díky tomuje uživatel relativně soběstačný z hlediska domácího
použití, přecházení (uvnitř domu i venku), mobility
a celkového životního stylu. Použití kyslíkového
koncentrátoru vyžaduje předpis lékaře a není určeno
k použití pro podporu života.
Přestože léčbu kyslíkem lze předepsat uživatelům
všech věkových kategorií, typický pacient pro léčbu
kyslíkem je starší 65 let a trpí některou z nemocídýchacích cest, včetně chronické obstrukční plicní
nemoci (CHOPN). Uživatelé mají obvykle dobré
kognitivní schopnosti a musí být schopni informovato bolestech. Pokud uživatel není schopen informovato bolestech nebo nedokáže přečíst a chápat značení
a pokyny k použití koncentrátoru, doporučuje sepoužití koncentrátoru pouze pod dohledem osoby,která je toho schopná. Pokud uživatelé při použití
koncentrátoru cítí nějaké bolesti, doporučuje se,
aby kontaktovali svého poskytovatele zdravotnípéče. Zároveň se uživatelům doporučuje, aby mělik dispozici záložní kyslík (tj. kyslíkovou láhev) propřípad výpadku proudu nebo poruchy koncentrátoru.
K použití koncentrátoru nejsou nutné žádné další
jedinečné dovednosti ani uživatelské schopnosti.

Rozbalení vašeho přístroje FreeStyle ComfortOvěřte, že všechny uvedené
součásti jsou součástí balení.
Pokud některé položky chybí,okamžitě kontaktujte dodavatele
kyslíku.

Standardní příslušenství• Přenosný kyslíkový koncentrátor• Přenosná taška• Ramenní popruh• 8článková baterie• Napájecí zdroj pro střídavý proud• Napájecí kabel pro střídavý proud• Napájecí zdroj pro stejnosměrný
proud• Uživatelská příručka

FreeStyle Comfort
Volitelné příslušenství• Záložní 8článková baterie• 16článková baterie• Stolní nabíječka• Přenosná taška pro veškeré
příslušenstvíSeznámení s přístrojem FreeStyle Comfort
Seznamte se s hlavními prvky přístroje FreeStyle Comfort a jeho uživatelským rozhraním.

Čelní pohled
Poutko na popruhUživatelské
rozhraníVýstupkyslíkuVýstupvzduchuPřívod vzduchu /filtryPřívod vzduchu /filtryKryt samice
napájecího
konektoru
BaterieTlačítko pro
vysunutí baterie
PN MN232-C4 K | Uživatelská příručka

CZE – 407

FreeStyle Comfort
Horní pohled
Obrazovka displejeZvýšení průtoku

Tlačítko vypínače

Tlačítko MenuSnížení průtoku

Obrazovka displeje při použití baterie
Nastavení průtoku

Zobrazení baterie

Ukazatel autoSAT
Zobrazení výstrah

Ukazatel podáníboluZbývajícíprocento
nabití baterieObrazovka displeje při použití napájecího zdroje pro střídavý i stejnosměrný proud

Nastavení průtoku
Zobrazení baterie

Prostor pro zobrazenívýstrah408 – CZEPN MN232-C4 K | Uživatelská příručka

Ukazatel dobíjení

FreeStyle ComfortDůležité!
Bezpečnostní pokyny jsou určeny
následujícím způsobem:
VAROVÁNÍ: DŮLEŽITÉ BEZPEČNOSTNÍ
INFORMACE O NEBEZPEČÍCH, KTERÁMOHOU ZPŮSOBIT VÁŽNÁ ZRANĚNÍ.
UPOZORNĚNÍ: Důležité informace, jakzabránit poškození přístroje FreeStyle
Comfort.

Poznámka: Informace, které vyžadují zvláštní
pozornost.

Indikace pro použití
VAROVÁNÍ: JE VELMI DŮLEŽITÉ ZVOLITPOUZE PŘEDEPSANOU ÚROVEŇ
KYSLÍKU. NEMĚŇTE VÝBĚR PRŮTOKU,POKUD VÁM TO NEBYLO NAŘÍZENO
OŠETŘUJÍCÍM LÉKAŘEM.
VAROVÁNÍ: V USA SMÍ BÝT PODLE FEDERÁLNÍCH
ZÁKONŮ TATO ZAŘÍZENÍ PRODÁVÁNA
A PRONAJÍMÁNA POUZE LÉKAŘI ČI JINÝMI
LICENCOVANÝMI POSKYTOVATELI ZDRAVOTNÍ PÉČENEBO NA JEJICH PŘÍKAZ.
VAROVÁNÍ: TATO JEDNOTKA NENÍ URČENA
K POUŽITÍ PRO PODPORU ŽIVOTA.
U GERIATRICKÝCH, PEDIATRICKÝCH NEBO
JAKÝCHKOLI JINÝCH UŽIVATELŮ, KTEŘÍNEDOKÁŽOU INFORMOVAT O BOLESTECH PŘIPOUŽÍVÁNÍ TOHOTO PŘÍSTROJE, MŮŽE BÝT NUTNÝ
DALŠÍ DOHLED. UŽIVATELÉ S POŠKOZENÝM
SLUCHEM A/NEBO ZRAKEM MOHOU POTŘEBOVATPOMOC PŘI SLEDOVÁNÍ ALARMŮ.
VAROVÁNÍ: NASTAVENÍ REŽIMU PULZNÍCH DÁVEK
MUSÍ BÝT STANOVENO PRO KAŽDÉHO UŽIVATELEINDIVIDUÁLNĚ PODLE JEHO VLASTNÍCH POTŘEB
BĚHEM KLIDU, CVIČENÍ A CESTOVÁNÍ.
VAROVÁNÍ: NĚKTERÉ POKUSY PACIENTA O DÝCHÁNÍ
NEMUSÍ SPUSTIT HOSPODÁRNOU FUNKCI.
VAROVÁNÍ: PŘÍSTROJ FREESTYLE COMFORT NENÍURČEN K POUŽITÍ U PACIENTŮ, KTEŘÍ PRODĚLALI
TRACHEOTOMII.
VAROVÁNÍ: TĚHOTNÉ NEBO KOJÍCÍ ŽENY BYNEMĚLY POUŽÍVAT PŘÍSLUŠENSTVÍ DOPORUČENÉV TÉTO PŘÍRUČCE, JELIKOŽ MŮŽE OBSAHOVATFTALÁTY.

Kontraindikace pro použití
VAROVÁNÍ: ZA URČITÝCH OKOLNOSTÍ
MŮŽE BÝT POUŽITÍ NEPŘEDEPSANÉHOKYSLÍKU NEBEZPEČNÉ. TOTO ZAŘÍZENÍ
BY MĚLO BÝT POUŽÍVÁNO POUZE TEHDY,POKUD BYLO PŘEDEPSÁNO LÉKAŘEM.
VAROVÁNÍ: NENÍ URČENO K POUŽITÍ V PŘÍTOMNOSTIHOŘLAVÝCH ANESTETIK.
VAROVÁNÍ: STEJNĚ JAKO U JINÝCH ELEKTRICKYNAPÁJENÝCH ZAŘÍZENÍ MŮŽE DOJÍT K SITUACI,KDY PŘÍSTROJ NEBUDE MOCI FUNGOVAT KVŮLI
PŘERUŠENÍ NAPÁJENÍ, PŘÍPADNĚ BUDE NUTNÉPROVÉST SERVIS KYSLÍKOVÉHO KONCENTRÁTORUKVALIFIKOVANÝM TECHNIKEM. KYSLÍKOVÝ
KONCENTRÁTOR NENÍ VHODNÝ PRO ŽÁDNÉHOUŽIVATELE, JENŽ BY BYL KVŮLI TOMUTO
DOČASNÉHO PŘERUŠENÍ VYSTAVEN NEPŘÍZNIVÝM
DŮSLEDKŮM NA ZDRAVÍ.

Bezpečnostní pokyny
VAROVÁNÍ: NEJSOU POVOLENY ŽÁDNÉ
ÚPRAVY TOHOTO ZAŘÍZENÍ.
VAROVÁNÍ: TENTO PŘÍSTROJ DODÁVÁ VYSOCE
KONCENTROVANÝ KYSLÍK, KTERÝ ZNAČNĚ
PODPORUJE HOŘENÍ. VYVARUJTE SE KOUŘENÍ
NEBO OTEVŘENÝCH PLAMENŮ VE STEJNÉMÍSTNOSTI, KDE SE NACHÁZÍ (1) TENTO PŘÍSTROJ
NEBO (2) JAKÉKOLI PŘÍSLUŠENSTVÍ PŘENÁŠEJÍCÍ
KYSLÍK. NEDODRŽENÍ TOHOTO VAROVÁNÍ MŮŽEVÉST K VÁŽNÉMU POŽÁRU A ŠKODÁM NA MAJETKU
A/NEBO ZPŮSOBIT ZRANĚNÍ ČI SMRT.
VAROVÁNÍ: NEPOUŽÍVEJTE JEDNOTKUV OMEZENÉM NEBO STÍSNĚNÉM PROSTORU (TJ.
V MALÉM POUZDRU NEBO RUČNÍ TAŠCE), V NĚMŽMŮŽE BÝT OMEZENO VĚTRÁNÍ. MOHLO BY DOJÍT
K PŘEHŘÁTÍ KYSLÍKOVÉHO KONCENTRÁTORUA SNÍŽENÍ JEHO VÝKONU.
VAROVÁNÍ: KONCENTRÁTOR BY MĚL BÝTUMÍSTĚN TAK, ABY BYL CHRÁNĚN PŘED KOUŘEM,ZNEČIŠTĚNÍM ČI VÝPARY.
VAROVÁNÍ: V PŘÍPADĚ VÁŽNÉHO INCIDENTU STÍMTO ZAŘÍZENÍM BY MĚL UŽIVATEL OKAMŽITĚ
NAHLÁSIT INCIDENT POSKYTOVATELI A / NEBO
VÝROBCI. VÁŽNÝ INCIDENT JE DEFINOVÁN JAKO
ZRANĚNÍ, SMRT NEBO POTENCIÁL ZPŮSOBITZRANĚNÍ / SMRT V PŘÍPADĚ OPAKOVANÉHOVÝSKYTU INCIDENTU. UŽIVATEL MŮŽE TAKÉNAHLÁSIT INCIDENT PŘÍSLUŠNÉMU ORGÁNU V
ZEMI, KDE K INCIDENTU DOŠLO.

PN MN232-C4 K | Uživatelská příručka

CZE – 409

FreeStyle Comfort
VAROVÁNÍ: POUŽITÍ URČITÝCH
PŘÍSLUŠENSTVÍ PRO VEDENÍ KYSLÍKU,KTERÁ NEJSOU URČENA PRO TENTO
KYSLÍKOVÝ KONCENTRÁTOR, MŮŽEVÉST KE SNÍŽENÍ JEHO VÝKONU.
DOPORUČENÁ PŘÍSLUŠENSTVÍ JSOUUVEDENA V TÉTO PŘÍRUČCE.

VAROVÁNÍ: ZABRAŇTE TOMU, ABY DOŠLOK ZABLOKOVÁNÍ PŘÍVODU VZDUCHU A VÝSTUPUVYFUKOVANÉHO VZDUCHU. NEUPUSŤTE ANINEVKLÁDEJTE ŽÁDNÉ PŘEDMĚTY DO OTVORŮ
PŘÍSTROJE. MOHLO BY DOJÍT K PŘEHŘÁTÍ
KYSLÍKOVÉHO KONCENTRÁTORU A SNÍŽENÍ JEHOVÝKONU.

VAROVÁNÍ: POKUD KYSLÍKOVÝ KONCENTRÁTORSPADL, BYL POŠKOZEN NEBO SE DOSTAL DO
KONTAKTU S VODOU, OBRAŤTE SE NA SVÉHOPOSKYTOVATELE DOMÁCÍ ZDRAVOTNÍ PÉČE,ABY PROVEDL KONTROLU NEBO PŘÍPADNOUOPRAVU ZAŘÍZENÍ. NEPOUŽÍVEJTE KYSLÍKOVÝ
KONCENTRÁTOR, POKUD MÁ POŠKOZENÝ
NAPÁJECÍ KABEL NEBO ZÁSTRČKU.

VAROVÁNÍ: NESPRÁVNÉ POUŽITÍ BATERIE MŮŽEZPŮSOBIT, ŽE SE BATERIE ZAHŘEJE A VZNÍTÍ, COŽMŮŽE VÉST K VÁŽNÝM ZRANĚNÍM. DEJTE POZOR,ABYSTE BATERII NEPROPÍCHLI, NEUDEŘILI S NÍ,
NESTOUPLI NA NI ČI JI NEUPUSTILI A ABYSTE JINEVYSTAVOVALI JINÝM SILNÝM ÚDERŮM NEBO
NÁRAZŮM. POUŽITÍ POŠKOZENÉ BATERIE MŮŽEZPŮSOBIT ZRANĚNÍ OSOB.

VAROVÁNÍ: PŘI PŘEPRAVĚ VE VOZIDLE KYSLÍKOVÝ
KONCENTRÁTOR PEČLIVĚ PŘIPOUTEJTE NEBO
JINÝM ZPŮSOBEM ZAJISTĚTE, ABYSTE ZABRÁNILI
POŠKOZENÍ NEBO ZRANĚNÍ.

VAROVÁNÍ: NEZKRATUJTE KOVOVÉ KONTAKTY
BATERIE KOVOVÝMI PŘEDMĚTY, JAKO JSOU KLÍČE
NEBO MINCE. TO MŮŽE ZPŮSOBIT JISKRY NEBO
VÉST K TVORBĚ NADMĚRNÉHO TEPLA.

VAROVÁNÍ: PŘEDPISY MINISTERSTVA DOPRAVY USA(DOT) A ORGANIZACE SPOJENÝCH NÁRODŮ (OSN)VYŽADUJÍ PŘI VŠECH MEZINÁRODNÍCH LETECHVYJMUTÍ BATERIE ZE ZAŘÍZENÍ V PŘÍPADĚ, ŽE JEKYSLÍKOVÝ KONCENTRÁTOR ODBAVEN JAKOZAVAZADLO. TAKÉ PŘI PŘEPRAVĚ KYSLÍKOVÉHOKONCENTRÁTORU MUSÍ BÝT BATERIE ZE ZAŘÍZENÍ
VYJMUTA A ŘÁDNĚ ZABALENA.

VAROVÁNÍ: PŘI POUŽITÍ PŘENOSNÉHOKYSLÍKOVÉHO KONCENTRÁTORU VE VENKOVNÍMPROSTŘEDÍ S NAPÁJENÍM STŘÍDAVÝM PROUDEMZAPOJTE NAPÁJENÍ VŽDY POUZE DO ZÁSUVKY
S OCHRANOU PROTI ZEMNÍMU SPOJENÍ (GFI).

VAROVÁNÍ: TENTO PŘÍSTROJ NEPOUŽÍVEJTE,DOKUD SI NEPŘEČTETE A NEPOCHOPÍTE TUTOPŘÍRUČKU. POKUD NEROZUMÍTE VAROVÁNÍMA POKYNŮM, KONTAKTUJTE DODAVATELE ZAŘÍZENÍ,DŘÍVE NEŽ PŘÍSTROJ POUŽIJETE, JINAK MŮŽEDOJÍT KE ZRANĚNÍ NEBO ŠKODÁM.
VAROVÁNÍ: POKUD CÍTÍTE BOLESTI NEBO SE
NACHÁZÍTE V NALÉHAVÉM ZDRAVOTNÍM STAVU,OKAMŽITĚ VYHLEDEJTE LÉKAŘSKOU POMOC.
VAROVÁNÍ: PROVOZ KYSLÍKOVÉHOKONCENTRÁTORU MIMO URČENÝ ROZSAHPROVOZNÍCH TEPLOT MŮŽE OMEZIT SCHOPNOSTKONCENTRÁTORU SPLNIT SPECIFIKACE
KYSLÍKOVÉHO KONCENTRÁTORU. MEZNÍ HODNOTYTEPLOTY NAJDETE V ČÁSTI TÉTO PŘÍRUČKY
VĚNOVANÉ SPECIFIKACÍM.
VAROVÁNÍ: NA PŘÍSLUŠENSTVÍ PŘENÁŠEJÍCÍ
KYSLÍK A NA KYSLÍKOVÝ KONCENTRÁTOR
NEPOUŽÍVEJTE OLEJE, MAZIVA, MATERIÁLY NABÁZI ROPY ANI JINÉ HOŘLAVÉ VÝROBKY. POUŽÍVAT
BY SE MĚLY POUZE ROZTOKY A BALZÁMY NA BÁZIVODY, KTERÉ JSOU KOMPATIBILNÍ S KYSLÍKEM.
KYSLÍK URYCHLUJE SPALOVÁNÍ HOŘLAVÝCH
LÁTEK.

410 – CZEPN MN232-C4 K | Uživatelská příručka

VAROVÁNÍ: KYSLÍKOVÝ KONCENTRÁTOR BY SENEMĚL POUŽÍVAT V BLÍZKOSTI JINÝCH ZAŘÍZENÍ
ANI BÝT ULOŽEN NA NICH ČI POD NIMI. POKUD SE
TAKOVÉMU POUŽITÍ NELZE VYHNOUT, ZAŘÍZENÍ
BY MĚLO BÝT SLEDOVÁNO, ABY SE OVĚŘIL JEHOBĚŽNÝ PROVOZ.
VAROVÁNÍ: POUŽITÍ JINÝCH NEŽ K TOMUURČENÝCH KABELŮ A ADAPTÉRŮ, S VÝJIMKOUKABELŮ A ADAPTÉRŮ PRODÁVANÝCH VÝROBCEMZDRAVOTNICKÉHO ELEKTRICKÉHO PŘÍSTROJE
JAKO NÁHRADNÍ DÍLY ZA VNITŘNÍ SOUČÁSTI,MŮŽE VÉST KE SNÍŽENÍ ODOLNOSTI KYSLÍKOVÉHOKONCENTRÁTORU.
VAROVÁNÍ: POUŽÍVEJTE POUZE ELEKTRICKÉNAPĚTÍ UVEDENÉ NA SPECIFIKAČNÍM ŠTÍTKU
PŘIPEVNĚNÉM NA PŘÍSTROJI.
VAROVÁNÍ: V KOMBINACI S TÍMTO VÝROBKEM
NEPOUŽÍVEJTE PRODLUŽOVACÍ KABELY ANINEZAPOJUJTE PŘÍLIŠ MNOHO ZÁSTRČEK
DO STEJNÉ ELEKTRICKÉ ZÁSUVKY. POUŽITÍPRODLUŽOVACÍCH KABELŮ NEPŘÍZNIVĚ OVLIVŇUJE
VÝKON ZAŘÍZENÍ. PŘÍLIŠ MNOHO ZÁSTRČEK
V JEDNÉ ZÁSUVCE MŮŽE VÉST K PŘETÍŽENÍ
ELEKTRICKÉHO PANELU, COŽ ZPŮSOBÍ AKTIVACIJISTIČE/POJISTKY NEBO DOKONCE VZNÍCENÍ,POKUD SE JISTIČ/POJISTKA NEUVEDE DO
PROVOZU.

FreeStyle Comfort
UPOZORNĚNÍ: V případě výstrahy, nebokdyž si všimnete, že kyslíkový koncentrátor
nefunguje správně, přečtěte si část tétopříručky věnovanou řešení potíží. Pokud sevám nepodaří problém vyřešit, požádejte
o radu dodavatele přístroje.
UPOZORNĚNÍ: Abyste zabránili zneplatnění záruky,dodržujte všechny pokyny výrobce.
VAROVÁNÍ: NEPOKOUŠEJTE SEO ŽÁDNOU ÚDRŽBU S VÝJIMKOU
MOŽNÝCH ŘEŠENÍ UVEDENÝCH V TÉTOPŘÍRUČCE. NESNÍMEJTE KRYTY. SNÍMATKRYTY A PROVÁDĚT SERVIS TOHOTO
PŘÍSTROJE SMÍ POUZE DODAVATELZAŘÍZENÍ NEBO KVALIFIKOVANÝSERVISNÍ TECHNIK.
VAROVÁNÍ: POUŽÍVEJTE POUZE PŘÍSLUŠENSTVÍDOPORUČENÉ VÝROBCEM. POUŽITÍ JAKÉHOKOLIJINÉHO PŘÍSLUŠENSTVÍ MŮŽE BÝT NEBEZPEČNÉA MŮŽE ZPŮSOBIT VÁŽNÉ POŠKOZENÍ VAŠEHOKYSLÍKOVÉHO KONCENTRÁTORU A POVEDEK ZNEPLATNĚNÍ ZÁRUKY.
VAROVÁNÍ: VÝROBCE DOPORUČUJE ALTERNATIVNÍZDROJ DOPLŇKOVÉHO KYSLÍKU PRO PŘÍPADY
VÝPADKU PROUDU NEBO STAVU ALARMU ČI
MECHANICKÉ ZÁVADY PŘÍSTROJE. O TYPUPOŽADOVANÉHO ZÁLOŽNÍHO SYSTÉMU SE
PORAĎTE SE SVÝM LÉKAŘEM NEBO DODAVATELEM
PŘÍSTROJE.
VAROVÁNÍ: ZABRAŇTE TOMU, ABY DOŠLOK ZABLOKOVÁNÍ PŘÍVODU VZDUCHU A VÝSTUPUVYFUKOVANÉHO VZDUCHU. NEUPUSŤTE ANINEVKLÁDEJTE ŽÁDNÉ PŘEDMĚTY DO OTVORŮ
PŘÍSTROJE. MOHLO BY DOJÍT K PŘEHŘÁTÍ
KYSLÍKOVÉHO KONCENTRÁTORU A SNÍŽENÍ JEHOVÝKONU.
VAROVÁNÍ: KOUŘENÍ PŘI POUŽÍVÁNÍ KYSLÍKU JE
HLAVNÍ PŘÍČINOU ZRANĚNÍ ZPŮSOBENÝCH OHNĚMA S TÍM SOUVISEJÍCÍCH ÚMRTÍ. MUSÍTE DODRŽOVAT
NÁSLEDUJÍCÍ BEZPEČNOSTNÍ VAROVÁNÍ:VAROVÁNÍ: VYVARUJTE SE KOUŘENÍ A POUŽITÍ
SVÍČEK ČI OTEVŘENÉHO OHNĚ VE STEJNÉMÍSTNOSTI, KDE SE NACHÁZÍ PŘÍSTROJ NEBO
PŘÍSLUŠENSTVÍ PŘENÁŠEJÍCÍ KYSLÍK.

VAROVÁNÍ: V DOMĚ NEBO NA JINÉM MÍSTĚ,KDE SE POUŽÍVÁ KYSLÍK, MUSÍ BÝT NÁPADNĚVYSTAVENY CEDULE „ZÁKAZ KOUŘENÍ – POUŽÍVÁNÍ
KYSLÍKU“. UŽIVATELÉ A JEJICH PEČOVATELÉ MUSÍBÝT INFORMOVÁNI O NEBEZPEČÍCH KOUŘENÍ
V PŘÍTOMNOSTI LÉKAŘSKÉHO KYSLÍKU NEBO PŘIJEHO POUŽITÍ.
VAROVÁNÍ: NEPOUŽÍVEJTE KYSLÍKOVÝ
KONCENTRÁTOR V PŘÍTOMNOSTI HOŘLAVÝCH
PLYNŮ. TO MŮŽE VÉST K RYCHLÉMU HOŘENÍ,KTERÉ ZPŮSOBÍ ŠKODY NA MAJETKU, TĚLESNÁ
ZRANĚNÍ NEBO SMRT.
VAROVÁNÍ: ODSTRANĚNÍ KANYLY A JEJÍ ULOŽENÍ
NA OBLEČENÍ, POVLEČENÍ, POHOVKU NEBO JINÝVÝPLŇOVÝ MATERIÁL ZPŮSOBÍ PO KONTAKTUS CIGARETOU, ZDROJEM TEPLA NEBO PLAMENEMOKAMŽITÝ OHEŇ.
VAROVÁNÍ: NOSNÍ KANYLU NENECHÁVEJTE NAOBLEČENÍ, POKRÝVCE POSTELE ČI POLSTROVÁNÍ
KŘESLA ANI POD NIMI. POKUD JE JEDNOTKA
ZAPNUTÁ, ALE NEPOUŽÍVÁ SE, KYSLÍK UČINÍ
Z TĚCHTO MATERIÁLŮ HOŘLAVÉ LÁTKY. ZAJISTĚTE,ABY BYL KYSLÍKOVÝ KONCENTRÁTOR VYPNUTÝ,KDYŽ SE NEPOUŽÍVÁ.
VAROVÁNÍ: PŘED VYJMUTÍM BATERIE MUSÍBÝT KYSLÍKOVÝ KONCENTRÁTOR VYPNUTÝA ODPOJENÝ OD EXTERNÍHO NAPÁJENÍ.
VAROVÁNÍ: MALÉ DÍLY MOHOUPŘEDSTAVOVAT POTENCIÁLNÍ RIZIKOUDUŠENÍ.
VAROVÁNÍ: VYVARUJTE SE KONTAKTU S OTVORYVYFUKOVANÉHO VZDUCHU NA DOBU DELŠÍ NEŽ
10 SEKUND, PROTOŽE MOHOU BÝT TEPLÉ.
VAROVÁNÍ: VÍTR ČI SILNÝ PRŮVAN MŮŽENEPŘÍZNIVĚ OVLIVNIT SPRÁVNÝ PRŮBĚH LÉČBY
KYSLÍKEM.
VAROVÁNÍ: TENTO VÝROBEK VÁS MŮŽE VYSTAVITCHEMICKÝM LÁTKÁM, VČETNĚ NIKLU, KTERÝ JE VESTÁTĚ KALIFORNIE POVAŽOVÁN ZA KARCINOGEN.
PODROBNĚJŠÍ INFORMACE NAJDETE NA WWW.
P65WARNINGS.CA.GOV.

VAROVÁNÍ: KOUŘENÍ PŘI NOŠENÍ KYSLÍKOVÉ
KANYLY MŮŽE ZPŮSOBIT POPÁLENINY OBLIČEJEA MOŽNÁ I SMRT.
VAROVÁNÍ: POKUD KOUŘÍTE, MOHOU VÁMNÁSLEDUJÍCÍ 3 KROKY ZACHRÁNIT ŽIVOT: VYPNĚTE
KYSLÍKOVÝ KONCENTRÁTOR, SEJMĚTE KANYLU
A OPUSŤTE MÍSTNOST, V NÍŽ SE TENTO PŘÍSTROJNACHÁZÍ.

PN MN232-C4 K | Uživatelská příručka

CZE – 411

FreeStyle Comfort
UPOZORNĚNÍ: Před odpojením napájecího
zdroje pro střídavý proud od kyslíkového
koncentrátoru vždy nejdříve odpojte
napájecí zdroj ze sítě.
UPOZORNĚNÍ: Neumisťujte kyslíkový koncentrátortak, aby byl obtížný přístup k napájecímu kabelu.
UPOZORNĚNÍ: Přívodní hadice kyslíku a napájecí
kabely vždy umístěte takovým způsobem, aby sezabránilo riziku zakopnutí nebo možnému náhodnému
uškrcení.
UPOZORNĚNÍ: Při používání kyslíkového
koncentrátoru v automobilu, na lodi nebo u jiných
zdrojů stejnosměrného proudu v kombinaci
s napájecím zdrojem pro stejnosměrný proud předpřipojením kyslíkového koncentrátoru zajistěte, aby
bylo vozidlo nastartované a v chodu. Nedodržení
těchto pokynů může vést k tomu, že napájecí zdroj
nebude do kyslíkového koncentrátoru přivádět
napájení.
UPOZORNĚNÍ: Když je automobil, ve kterém používáte
kyslíkový koncentrátor, vypnutý, přístroj odpojte
a vyjměte jej z automobilu. Neskladujte kyslíkový
koncentrátor ve velmi horkém nebo chladném
automobilu ani v žádném jiném podobném prostředí
s vysokými nebo nízkými teplotami.

Poznámka: Přenosná a mobilní radiofrekvenční
komunikační zařízení mohou mít vliv na zdravotnické
elektrické přístroje.
UPOZORNĚNÍ: NENECHÁVEJTE kyslíkový
koncentrátor ani napájecí zdroj zapojený
do vozidla, když je zapalování ve VYPNUTÉ
poloze. Pokud tak učiníte, může dojít
k vybití baterie vozidla.
UPOZORNĚNÍ: Používání nebo skladování kyslíkového
koncentrátoru mimo rozsah jeho běžných provozních
teplot a rozsah okolního tlaku (nadmořské výšky)může ovlivnit výkon a snížit provozní dobu baterienebo zvýšit dobu nabíjení baterie. Mezní hodnoty
skladovací a provozní teploty najdete v části tétopříručky věnované specifikacím.
UPOZORNĚNÍ: U kyslíkových koncentrátorů
vybavených bateriemi: Skladujte je na chladném
a suchém místě, abyste pomohli zajistit dlouhouživotnost baterie. Dlouhodobé skladování kyslíkového
koncentrátoru při vysokých teplotách nebo s plněnabitou či zcela vybitou baterií může snížit celkovouživotnost baterie. Nepokoušejte se otevřít baterii –uvnitř baterie nejsou žádné díly určené k servisu.
Baterie uchovávejte mimo dosah dětí.
UPOZORNĚNÍ: POUŽÍVEJTE POUZE baterie dodanévýrobcem. Za účelem řádné likvidace baterie požádejte
o pokyny dodavatele zařízení nebo místní státní úřad.

412 – CZEPN MN232-C4 K | Uživatelská příručka

Poznámka: Přesvědčte se, že je kanyla zcela zasunutá a zajištěná. To zajistí, aby kyslíkový koncentrátor
mohl správně rozeznat nadechnutí a dodávat kyslík.
Během nádechu byste měli slyšet nebo cítit, jak kyslík
proudí do hrotů nosní kanyly.
Poznámka: Lithiové baterie mohou permanentně
ztrácet kapacitu, když jsou v plně nabitém nebo zcela
vybitém stavu vystaveny extrémně vysokým teplotám.
Za účelem dlouhodobého skladování se doporučuje,aby baterie byly nabité na 25–50 % a uchovávaly sev teplotním rozsahu –20 až +20 °C (–4 až 68 °F).
Poznámka: Použití určitých příslušenství pro vedení
kyslíku, která nejsou specifikována pro použití s tímto
kyslíkovým koncentrátorem, může vést ke snížení
jeho výkonu. Doporučená příslušenství jsou uvedena
v této příručce.
Poznámka: Nepoužívejte kyslíkový koncentrátor
bez nainstalovaného filtru na přívodu vzduchu. Je-li
k dispozici druhý filtr, vložte do přístroje náhradnífiltr před vyčištěním zaneseného filtru. Zanesený filtr
vyčistěte v teplé mýdlové vodě a před použitím jejřádně vysušte.
Poznámka: Výrobce nedoporučuje sterilizaci tohotozařízení.
Poznámka: Při použití kyslíkového koncentrátoru
nikdy nehrozí unikání kyslíku do místnosti.
VAROVÁNÍ: KYSLÍKOVÝ KONCENTRÁTOR
BY SE NEMĚL POUŽÍVAT ANI PŘIPOJOVATK EXTERNÍMU ZDROJI STŘÍDAVÉHONEBO STEJNOSMĚRNÉHO PROUDU,POKUD NENÍ NAINSTALOVANÁ BATERIE.
JESTLIŽE JE BATERIE VYJMUTÁ,NEPOKOUŠEJTE SE JEDNOTKU ZAPNOUTANI PŘIPOJIT K EXTERNÍMU ZDROJI
NAPÁJENÍ.
BATERII VYJMĚTE POUZE VE CHVÍLI, KDY JE
JEDNOTKA VYPNUTÁ A ODPOJENÁ OD PŘÍVODU
NAPÁJENÍ. NEPOKOUŠEJTE SE BATERII VYJMOUT,
KDYŽ JE JEDNOTKA STÁLE ZAPNUTÁ NEBO
PŘIPOJENÁ K PŘÍVODU NAPÁJENÍ.

Poznámka: Pokud byl kyslíkový koncentrátor delšídobu uskladněn mimo běžné provozní podmínkyznázorněné ve výše uvedené tabulce, nechte jej předzapnutím zahřát/vychladnout na běžnou provozní
teplotu.

FreeStyle Comfort
Specifikace
FreeStyle ComfortRozměry (s jednou baterií)25,4 × 18,5 × 7,9 cm (10,0 × 7,3 × 3,1 in.)Rozměry (s dvojitou baterií)27,9 × 18,5 × 7,9 cm (11,0 × 7,3 × 3,1 in.)Hmotnost (s jednou baterií)

2,3 kg (5 lb)

Hmotnost (s dvojitou baterií)2,7 kg (6 lb)

Jmenovitá hladina hluku***

39,93 dB (A) ± 0,42 dB (A) @ 2

Nastavení průtoku

Pulzní nastavení 1, 2, 3, 4, 5

Nastavení výstupu O2

Nastavení 1 – 210 ml/min
Nastavení 2 – 420 ml/min
Nastavení 3 – 630 ml/min
Nastavení 4 – 840 ml/min
Nastavení 5 – 1050 ml/min(při 20 bpm)

Koncentrace kyslíku*
Monitor kyslíku

90 % (+5,5 % / –3 %)AnoMaximální výstupní tlak

< 206,8 kPa (30 psi)Výchozí citlivost dávkování> –0,1 cm H2O (nastavitelné)

Napájení střídavým proudem

100–240 V st, 50–60 Hz, 120W Max

Napájení stejnosměrným proudem

11–18 V ss (max. 10 A)

Doba nabíjení baterie

Jedna baterie: 3,5 hodiny
Dvojitá baterie: 6,0 hodinVýdrž bateriíVýdrž jedné baterie: Nastavení 2 – až 4 hodinyVýdrž dvojité baterie: Nastavení 2 – až 8 hodin

Provozní teplota**
Typ baterie
Kapacita baterie

5 až 40 °C (41 až 104 °F)
Lithium Ion
Jeden článek: 6700 mAh; Dvojitá buňka: 13400 mAhPřepravní a skladovací teplota (jednotka)–25 až +70 °C (–13 až +158 °F) do 3048 m (10 000 stop)Přepravní a skladovací teplota (baterie)–25 až +70 °C (–13 až +158 °F) do 3048 m (10 000 stop)

Provozní nadmořská výška**Provozní vlhkost
Vlhkost při skladování
Omezená záruka

−381 až 3048 m (−1250 až 10 000 stop) (testováno na 700–1060 hPa)
15–95 % relativní vlhkosti (bez kondenzace)
0 - 90% bez kondenzace
3 roky na jednotku
1 rok na baterii a napájecí zdroje* Na základě atmosférického tlaku 101 kPa (14,7 psi) při 21 °C (70 °F)** Jednotka bude fungovat ve všech prostředích, dokud bude prostředí ve specifikovaném rozmezí teplot a vlhkosti. Provoz mimo
tyto provozní specifikace může omezit schopnost kyslíkového koncentrátoru splnit uvedené parametry u vyšších průtočných
množství. Obrazovka rozehřívání se může nadále zobrazovat, dokud koncentrátor nedosáhne cílové koncentrace nebo po dobu až
15 minut. Během této doby jednotku nadále používejte.
*** Hladina akustického tlaku měřená podle zkušební metody č. 14-1 10/2018 MDS-Hi.
Předpokládaná životnost je nejméně pět let. Očekávaná skladovací životnost lůžek s molekulovým sítem je přibližně jeden rok,
pokud není koncentrátor během celého období používán.

PN MN232-C4 K | Uživatelská příručka

CZE – 413

FreeStyle Comfort

Provozní pokyny

1. Umístěte přístroj FreeStyle Comfort na vhodné místo.
a. Koncentrátor by měl být dobře větrán – filtry napřívodu vzduchu a výstup vzduchu by neměly
být nijak blokovány.
b. Koncentrátor by měl být umístěn takovým
způsobem, aby všechny zvukové a vizuální
ukazatele a výstrahy byly snadno slyšet a vidět.
c. Před použitím přístroje se přesvědčte, že jsou
nainstalovány filtry na přívodu vzduchu. Pokudpřístroj nemá filtr, kontaktujte dodavatele zařízení.

3. Zasuňte konektor napájecího zdroje pro střídavý
nebo stejnosměrný proud do samice napájecího
konektoru v pravém dolním rohu přední části
přístroje.

2. Nainstalujte baterii do koncentrátoru.
a. Zarovnejte baterii se spodní středovou částí
krytu přístroje FreeStyle Comfort.
4. Připojte nosní kanylu k výstupu kyslíku. Předpoužitím se přesvědčte, že zapojení je zajištěné.

POZNÁMKA: Použití jakýchkoli kanyl, které nebyly
doporučeny, může ovlivnit výkon.

b. Zasuňte baterii dovnitř, dokud neuslyšítezacvaknutí, které potvrdí zajištění baterie na
správném místě.

Poznámka: Poskytovateli zařízení: K použití s FreeStyle
Comfort se doporučuje následující příslušenství pro
podávání kyslíku:• Nasální kanyla s hadicí o délce 2,1 m (maximální LPM:
6 LPM): CAIRE Číslo dílu 5408-SEQ• Další doporučené informace o příslušenství jsou kdispozici online na www.caireinc.com.

414 – CZEPN MN232-C4 K | Uživatelská příručka

FreeStyle Comfort
5. Stisknutím tlačítka vypínače zapněte přístroj
FreeStyle Comfort a vyberte předepsané nastaveníprůtoku pomocí tlačítek ke zvýšení (+) nebo snížení
(–) průtoku.

Nastavení pulzního objemu (ml)*– autoSAT
Frekvencedýchání

1

2

3

4

5

15

10,5

21,0

31,5

42,0

52,5

20

10,5

21,0

31,5

42,0

52,5

25

10,5

21,0

31,5

42,0

42,0

30

10,5

21,0

31,5

35,0

35,0

35

10,5

21,0

30,0

30,0

30,0

40

10,5

21,0

26,3

26,3

26,3

*Tolerance je ±15 %Uslyšíte krátkou zvukovou výstrahu, která
signalizuje, že došlo k zapnutí přístroje.

Nastavení pulzního objemu (ml)* –nikoli autoSAT

Poznámka: Při prvním zapnutí přístroje FreeStyle
Comfort je průtok nastaven na hodnotu 2. Pokud
dojde ke změně nastavení, přístroj se příště zapne
s posledním nastavením průtoku.

Frekvencedýchání

1

2

3

4

5

15

14,0

28,0

42,0

56,0

70,0

20

10,5

21,0

31,5

42,0

52,5

6. Nasaďte si nosní kanylu na obličej dle pokynů
výrobce dodaných s kanylou a dýchejte běžným
způsobem. Při každém nadechnutí byste měli cítit
dávku kyslíku vycházející z nosní kanyly.

25

8,4

16,8

25,2

33,6

42,0

30

7,0

14,0

21,0

28,0

35,0

35

6,0

12,0

18,0

24,0

30,0

40

5,3

10,5

15,8

21,0

26,3

7. Nastavení průtoku na požadované průtočnémnožství změníte pomocí tlačítek pro zvýšení
(+) nebo snížení (–) průtoku. Při změně nastaveníprůtoku je běžné, že uslyšíte rozdílné zvuky.

8. Vypněte přístroj FreeStyle Comfort stisknutím
tlačítka vypínače.

Poznámka: Obrazovka displeje přejde do pohotovostního režimu a po 20 sekundách zhasne. Obrazovku
probudíte z pohotovostního režimu stisknutím
kteréhokoli tlačítka.
9. Pokud jde o pokročilé funkce, viz Pokyny pro
navigaci v menu

SPECIÁLNÍ FUNKCE PRODUKTU
autoSAT®
Patentovaná funkce CAIRE autoSAT umožňuje
FreeStyle Comfort dodávat konstantní objem kyslíku(bolus) na dech až do maximální rychlosti dechu.
Protože se dech uživatele zvyšuje během činností, jako
je chůze, funkce autoSAT upraví činnost zařízení tak,aby udržovala nastavený bolusový objem.
UltraSense®
Tato citlivá spoušť pulzní dávky detekuje změnu tlakuběhem vašeho dýchání a uvolňuje proud kyslíku, kdyžse nadechnete.
autoDOSE™
Pokud dojde ke spuštění poplachu „nebyl zjištěn
nádech“, tato funkce spustí proud kyslíku při
20 nádeších za minutu se stávajícím nastavenímprůtoku, dokud nebude stav poplachu vyřešen.
PN MN232-C4 K | Uživatelská příručka

CZE – 415

FreeStyle Comfort

Napájecí zdrojePřístroj FreeStyle Comfort lze napájet třemi způsoby:• Nabíjecí baterie• Napájení střídavým proudem• Napájení stejnosměrným proudem
Informace o provozu všech těchto součástí jsou
uvedeny v následujících odstavcích.
VAROVÁNÍ: POUŽÍVEJTE POUZENAPÁJECÍ ZDROJE UVEDENÉ V TÉTOUŽIVATELSKÉ PŘÍRUČCE. POUŽITÍ
NAPÁJECÍCH ZDROJŮ NEBO BATERIÍ,KTERÉ NEPOCHÁZÍ OD SPOLEČNOSTICAIRE, MŮŽE ZPŮSOBIT POŠKOZENÍ
PŘÍSTROJE A JEHO NESPRÁVNÝ
PROVOZ.

VYJMUTÍ BATERIE

Stiskněte tlačítko pro vysunutí baterie dolů směrem
od přístroje FreeStyle Comfort. DRŽTE tlačítko
a vytáhněte baterii směrem od koncentrátoru.
VAROVÁNÍ: PŘED VYJMUTÍM BATERIEMUSÍ BÝT KYSLÍKOVÝ KONCENTRÁTORVYPNUTÝ A ODPOJENÝ OD EXTERNÍHO
NAPÁJENÍ.

BATERIE

Přístroj FreeStyle Comfort lze používat s dobíjecí
baterií, která je dodávána s přístrojem. Při použití
kyslíkového koncentrátoru s baterií neexistují žádná
provozní omezení. Při napájení baterií jsou dostupnávšechna průtočná množství.

KONTROLA STAVU NABITÍ BATERIEINSTALACE BATERIE

Zarovnejte baterii se spodní středovou částí krytu
přístroje FreeStyle Comfort. Zatlačte baterii dovnitř,dokud neuslyšíte zacvaknutí, které potvrdí zajištění
baterie na správném místě.

Když je baterie nainstalována v přístroji FreeStyle
Comfort, zjistíte její stav nabití na displeji pomocí
ikony baterie a odhadované zbývající doby.
80–100 %60–80 %

40–60 %

20–40 %

20 %

416 – CZEPN MN232-C4 K | Uživatelská příručka

FreeStyle Comfort
Poznámka: Pokud se baterie nabíjí, objeví se na
displeji místo zbývajícího procenta nabití ukazatel
zástrčky.
Chcete-li zjistit stav nabití u baterie, která nenínainstalována v přístroji FreeStyle Comfort, položte
ji a stiskněte tlačítko TEST. Rozsvítí se signalizační
kontrolky baterie (25–100 %) umístěné vpravo odtlačítka TEST, které informují o úrovni nabití baterie.

PROVOZNÍ VÝDRŽ BATERIE

Je-li externí napájecí zdroj pro střídavý nebo
stejnosměrný proud odpojen a je nainstalovaná
nabitá baterie, koncentrátor se automaticky přepne
na napájení z baterie. Níže uvedená tabulka ukazuje
typickou výdrž nové baterie. Výdrž baterie zůstaneneměnná bez ohledu na vaši frekvenci dýchání.
Pulzní
nastaveníVýdrž jedné(8článkové) baterieVýdrž dvojité(16článkové)baterie1

až 8 hodinaž 16 hodin2

až 4 hodiny

až 8 hodiny

3až 3 hodinyaž 6 hodin4až 2,25 hodiny

až 4,5 hodiny

5až 2 hodiny

až 4 hodiny

TYPICKÁ DOBA NABÍJENÍ BATERIEPRVNÍ DOBÍJENÍ

Nová baterie dodávaná s přístrojem FreeStyle Comfort
je při expedici z výrobního závodu nabita pouzečástečně. Před prvním použitím byste měli baterii
dobít pomocí napájecího zdroje pro střídavý nebo
stejnosměrný proud.

OBECNÉ INFORMACE O DOBÍJENÍ

Typická doba plného dobití zcela vybité baterie je
3,5 hodiny u jedné baterie a 6 hodin u dvojité baterie.

Poznámka: Pokud byla baterie zcela vybitá, nemusí
se začít dobíjet ihned.

ŽIVOTNOST BATERIE

Baterie přístroje FreeStyle Comfort jsou navrženy tak,aby vydržely 300 cyklů nabití/vybití.

Přístroj FreeStyle Comfort bude dobíjet bateriipokaždé, když je nainstalována do koncentrátoru a tenje připojen k externímu napájecímu zdroji pro střídavý
nebo stejnosměrný proud (kromě letadel).
Baterie je plně nabitá, když ukazuje stav 100 %.
Baterii lze také dobíjet pomocí volitelné stolnínabíječky.

PN MN232-C4 K | Uživatelská příručka

CZE – 417

FreeStyle Comfort

Napájecí zdroj pro střídavý
a stejnosměrný proudPřístroj FreeStyle Comfort je standardně dodáváns napájecími zdroji pro střídavý i stejnosměrný proud.
Napájecí zdroj pro střídavý proud obsahuje následujícípoložky:a. Napájecí adaptér pro střídavý proud, který
pracuje s napětím 100–240 V při frekvenci
50–60 Hz.
b. Napájecí kabel pro střídavý proud určený
k připojení k napájecímu adaptéru a příslušné
zásuvce se střídavým proudem.

Napájecí zdroj pro stejnosměrný proud se skládá
z kabelu, který se zapojí přímo do jednotky a do
zásuvky se stejnosměrným proudem.
2. Připojte napájecí kabel pro střídavý proudk napájecímu adaptéru pro střídavý proud.

NAPÁJENÍ STŘÍDAVÝM PROUDEM

Napájení střídavým proudem je určeno pro použití
doma nebo na místech, kde je k dispozici standardnízásuvka se střídavým proudem. Přístroj FreeStyle
Comfort je v kombinaci s napájením střídavým
proudem plně funkční. Přístroj FreeStyle Comfort taképři každém připojení k napájení střídavým proudem
dobíjí baterii (kromě letadel). Baterie se dobíjí bez
ohledu na to, zda je přístroj FreeStyle Comfort zapnutý
nebo vypnutý.
VAROVÁNÍ: NEPŘIPOJUJTE EXTERNÍ
NAPÁJENÍ, POKUD NENÍ BATERIENAINSTALOVANÁ V KYSLÍKOVÉM
KONCENTRÁTORU.
Přístroj FreeStyle Comfort připojíte k napájení
střídavým proudem následujícím způsobem:
1. Zapojte napájecí adaptér pro střídavý proud donapájecí zásuvky v pravém dolním rohu přední části
přístroje FreeStyle Comfort. Dejte pozor, abyste
konektor zatlačili až nadoraz a připojení tak bylo
zajištěné.

418 – CZEPN MN232-C4 K | Uživatelská příručka

3. Opačný konec napájecího kabelu pro střídavý
proud zapojte do zásuvky ve zdi nebo jiné vhodné
elektrické zásuvky.

FreeStyle Comfort
4. Stisknutím tlačítka vypínače zapněte přístroj
FreeStyle Comfort a vyberte předepsané nastaveníprůtoku pomocí tlačítek ke zvýšení (+) nebo snížení
(–) průtoku.

Poznámka: Pokud je potřeba dobít baterii, ikona baterie na displeji bude mít jeden oranžový sloupeček.

NAPÁJENÍ STEJNOSMĚRNÝM PROUDEM

Napájení stejnosměrným proudem je určeno propoužití se zásuvkami v automobilech, na lodích nebov jiných motorových vozidlech. Přístroj FreeStyle
Comfort je v kombinaci s napájením stejnosměrným
proudem plně funkční. V případě potřeby přístroj
FreeStyle Comfort také při každém připojení
k napájení stejnosměrným proudem dobíjí baterii(kromě letadel). Baterie se dobíjí bez ohledu na to,zda je přístroj vypnutý nebo zapnutý a používá se.

3. Opačný konec napájecího kabelu pro stejnosměrný
proud zapojte do zásuvky automobilu se
stejnosměrným proudem.

4. Stisknutím tlačítka vypínače zapněte přístroj
FreeStyle Comfort a vyberte předepsané nastaveníprůtoku pomocí tlačítek ke zvýšení (+) nebo snížení
(–) průtoku.

Poznámka: Pokud je potřeba dobít baterii, ikona baterie na displeji bude mít jeden oranžový sloupeček.

VAROVÁNÍ: NEPŘIPOJUJTE EXTERNÍ
NAPÁJENÍ, POKUD NENÍ BATERIENAINSTALOVANÁ V KYSLÍKOVÉM
KONCENTRÁTORU.
VAROVÁNÍ: NEPOUŽÍVEJTE PŘÍSTROJ ZAPOJENÝ
DO ZÁSUVKY V AUTOMOBILU S NAPĚTÍM 12 V SS,KDYŽ JE VOZIDLO VYPNUTÉ.
1. Nastartujte motor vozidla.
2. Zapojte konektor napájecího zdroje pro
stejnosměrný proud do samice napájecího konektoru
v pravém dolním rohu přední části přístroje
FreeStyle Comfort. Dejte pozor, abyste konektor
zatlačili až nadoraz a připojení tak bylo zajištěné.

PN MN232-C4 K | Uživatelská příručka

CZE – 419

FreeStyle ComfortČištění, péče a pravidelná údržba
PRAVIDELNÁ ÚDRŽBAPreventivní údržba
Zařízení nevyžaduje údržbu. Filtry uvnitř zařízení jsou
určeny k tomu, aby vydržely životnost zařízení. Pokudzařízení nepracuje správně, měla by být jednotka
vrácena poskytovateli k opravě. Servis vnitřních
součástí umístěných pod krytem přístroje FreeStyle
Comfort smí provádět pouze dodavatel kyslíku.
Provozovatel přístroje je zodpovědný za následující
postupy údržby.
• Jednou týdně – vyčistěte filtry na přívodu vzduchu
VAROVÁNÍ: NEPOKOUŠEJTE SE OTEVŘÍTANI SEJMOUT PLASTOVÉ KRYTY
KYSLÍKOVÉHO KONCENTRÁTORU.
PŘÍSTROJ SMÍ OTVÍRAT POUZEKVALIFIKOVANÝ SERVISNÍ TECHNIK.

FILTRY NA PŘÍVODU VZDUCHU

Vzduch vstupuje do přístroje FreeStyle Comfort přesfiltry na přívodu vzduchu umístěné v přední spodníčásti kyslíkového koncentrátoru. Tyto filtry zabraňují
tomu, aby do přístroje pronikal prach nebo jiné velkéčástice, které se nachází ve vzduchu.
Filtry na přívodu vzduchu by se měly čistit alespoň
jednou týdně. Dodavatel kyslíku vám může v závislosti
na provozních podmínkách doporučit častější čištění
filtrů. Za účelem řádného čištění filtrů na přívodu
vzduchu dodržujte následující kroky.
1. Vyjměte oba filtry z přední části přístroje FreeStyle
Comfort zatažením za příslušný jazýček a jejichvytažením směrem ven.

VAROVÁNÍ: NEPOUŽÍVEJTEKYSLÍKOVÝ KONCENTRÁTOR BEZ
NAINSTALOVANÉHO FILTRU NA PŘÍVODUVZDUCHU.

ÚDRŽBA BATERIEChraňte baterii před kontaktem s kapalinami.
Nepokoušejte se baterii rozebírat ani otevírat.
Nevystavujte baterii otevřeným plamenům.
Nepouštějte baterii na zem.
Uchovávejte baterii mimo dosah dětí.
Pokud baterie vykazuje známky poškození, předpoužitím kontaktujte dodavatele kyslíkového
koncentrátoru.
Nepokládejte baterii do mikrovlnné trouby či jiných
kuchyňských zařízení ani do jejich blízkosti.

SKLADOVÁNÍ BATERIE

Baterie kyslíkového koncentrátoru nemusí být předdobíjením zcela vybitá. Doporučuje se dobíjet baterii
kyslíkového koncentrátoru po každém použití.
Nenechávejte baterii na místech vystavených
nadměrnému teplu, například na přímém slunečnímsvětle nebo v automobilu. Pokud je na takovém místě
necháte, může to způsobit snížení výkonu bateriea zkrácení její životnosti. Může to vést také k tomu,
že baterie bude produkovat teplo či kouř, dojde
k jejímu vznícení nebo vybuchne. Baterie by měla být
skladována ve stabilním prostředí s nízkou vlhkostí(nižší než 70 % relativní vlhkosti), ve kterém senevyskytují korozivní plyny. Abyste zabránili vzniku rzi,předejděte situacím, při kterých může dojít ke srážení
vody, například rychlému střídání okolní teploty.
Za účelem dlouhodobého skladování (jeden měsíc
a déle) se doporučuje stav částečného nabití/vybitímezi 50 % až 75 % (dvě až tři kontrolky LED).
Jakmile je baterie nabitá, vyjměte ji z kyslíkového
koncentrátoru.

2. Umyjte oba filtry v teplé mýdlové vodě.
3. Řádně filtry opláchněte a zbývající vodu otřete
měkkým hadříkem bez žmolků. Přesvědčte se, žefiltry jsou před opětovnou instalací zcela suché.
4. Nainstalujte zpět čisté a suché filtry.

420 – CZEPN MN232-C4 K | Uživatelská příručka

FreeStyle ComfortČIŠTĚNÍ

Před jakýmkoli čištěním nebo přístroj VYPNĚTE
a odpojte od napájení střídavým či stejnosměrným
proudem. Na vnější kryt přímo NESTŘÍKEJTE.
Používejte vlhký (nikoli mokrý) hadřík nebo houbičku.
Za účelem vyčištění krytu a napájecích zdrojů
postříkejte hadřík nebo houbičku jemným čistič pro
domácnost roztokem. Postupujte dle pokynů výrobcečisticího prostředku.
VAROVÁNÍ: PŘED ČIŠTĚNÍM PŘÍSTROJVYPNĚTE A ODPOJTE NAPÁJECÍKABEL ZE SÍŤOVÉ ZÁSUVKY, ABYSTEZABRÁNILI NEBEZPEČÍ NÁHODNÉHOÚRAZU ELEKTRICKÝM PROUDEM
A VZNIKU POPÁLENIN. SNÍMAT KRYTYA PROVÁDĚT SERVIS TOHOTO PŘÍSTROJE
SMÍ POUZE DODAVATEL ZAŘÍZENÍ NEBOKVALIFIKOVANÝ SERVISNÍ TECHNIK.
VAROVÁNÍ: JE NUTNÉ DBÁT NA TO, ABY SEKYSLÍKOVÝ KONCENTRÁTOR A BATERIE
NENAMOČILI A ABY SE DO NICH NEDOSTALA ŽÁDNÁKAPALINA. MOHLO BY TO VÉST K PORUŠE NEBO
VYNUCENÉMU VYPNUTÍ A ZVÝŠIT NEBEZPEČÍ
ÚRAZU ELEKTRICKÝM PROUDEM NEBO VZNIKUPOPÁLENIN.
VAROVÁNÍ: NEPOUŽÍVEJTE KAPALINY PŘÍMO
NA PŘÍSTROJ. SEZNAM NEŽÁDOUCÍCHCHEMICKÝCH LÁTEK OBSAHUJE MIMOJINÉ NÁSLEDUJÍCÍ POLOŽKY: ALKOHOLA VÝROBKY NA BÁZI ALKOHOLU, VÝROBKY
NA BÁZI KONCENTROVANÉHO CHLORU(ETHYLENDICHLORID) A VÝROBKY NA BÁZI OLEJE.
TYTO LÁTKY SE NESMÍ POUŽÍVAT K ČIŠTĚNÍ
PLASTOVÉHO KRYTU KONCENTRÁTORU, PROTOŽEMOHOU PŘÍSTROJ POŠKODIT.

VAROVÁNÍ: KRYT, OVLÁDACÍ PANEL A NAPÁJECÍKABEL OČISTĚTE POUZE POMOCÍ JEMNÉHODOMÁCÍHO ČISTICÍHO PROSTŘEDKU NANESENÉHOVLHKÝM (NIKOLI MOKRÝM) HADŘÍKEM NEBO
HOUBIČKOU A NÁSLEDNĚ VŠECHNY POVRCHYVYTŘETE DOSUCHA. ZABRAŇTE VNIKNUTÍJAKÉKOLI KAPALINY DO VNITŘNÍ ČÁSTI
KONCENTRÁTORU. VĚNUJTE ZVLÁŠTNÍ POZORNOSTVÝSTUPU KYSLÍKU URČENÉMU K PŘIPOJENÍ
KANYLY, UJISTĚTE SE, ŽE NA NĚM NENÍ ZACHYCENPRACH, VODA ANI JINÉ ČÁSTICE.
VAROVÁNÍ: K ČIŠTĚNÍ BATERIE POUŽIJTE VLHKÝ(NIKOLI MOKRÝ) HADŘÍK NEBO HOUBIČKU.
NEJDŘÍVE NASTŘÍKEJTE NA HADŘÍK NEBO
HOUBIČKU JEMNÝ DOMÁCÍ ČISTICÍ PROSTŘEDEKA OČISTĚTE KRYT BATERIE.
VAROVÁNÍ: K ČIŠTĚNÍ NEPOUŽÍVEJTE
AUTOMATIZOVANÉ MYCÍ SYSTÉMY.

VÝMĚNA KANYLYVždy postupujte podle pokynů ke správnému použití
od výrobce kanyl. Nosní kanylu nebo hadici vedoucí
kyslík vyměňte po každých 3 až 6 měsících (přidenním použití po 3 měsících) podle doporučenívýrobce nebo dodavatele kyslíku. Pokyny k čištění
a výměně vám poskytne lékař nebo dodavatel kyslíku.
Další produkty pro případ výměny jsou dostupné
u dodavatele kyslíku.

PN MN232-C4 K | Uživatelská příručka

CZE – 421

FreeStyle Comfort

Stavy alarmů
Priorita

Popisvýstrahy

1

Zobrazení na displeji

Zvukový
signálPodrobnosti / navrhované opatření

Systémováporucha,vysoká teplota
baterie

1 pípnutíkaždých
20 sekundPřístroj vydá výstrahu, když se teplota baterieblíží mezní hodnotě bezpečnostního vypnutí.
Když je dosaženo mezní hodnoty vypnutí,přístroj se vypne. Je-li připojeno externínapájení, objeví se obrazovka výstrahy
a po dobu 2 minut se bude ozývat zvukovávýstraha.
Testujte s baterií při pokojové teplotě. Pokud
problém přetrvává, kontaktujte dodavatele.

2

Systémováporucha,vysoká teplota

1 pípnutíkaždých
20 sekundPřístroj vydá výstrahu, když se vnitřní teplotablíží mezní hodnotě vypnutí.
Když je dosaženo mezní hodnoty vypnutí,přístroj se vypne, objeví se obrazovkavýstrahy a po dobu 2 minut se bude ozývat
zvuková výstraha.
Ujistěte se, že žádný otvor jednotky nenízablokovaný. Pokud problém přetrvává,
kontaktujte dodavatele.

3

Systémováporucha,vysoký tlak

1 pípnutíkaždých
20 sekundKdyž je dosaženo mezní hodnoty vypnutí,přístroj se vypne, objeví se obrazovkavýstrahy a po dobu 2 minut se bude ozývat
zvuková výstraha.
Pokud problém přetrvává, kontaktujte
dodavatele.

4

Nebyl zjištěn
nádech1 pípnutíkaždých
30 sekund

Po dobu 15 sekund nebyl zjištěn nádech.
Aktivuje se po prvním nádechu nebo pozhasnutí obrazovky rozehřívání (podle toho,co nastane dříve).
Zkontrolujte, zda kanyla není zalomená
nebo zda na ní nejsou netěsnosti. Zvyšte
nastavení citlivosti. Pokud problém přetrvává,
kontaktujte dodavatele.

5

Vypnutí kvůli
vybité baterii

1 pípnutíkaždých
30 sekundVýstraha je aktivní, dokud nedojde k úplnému
vybití baterie a vypnutí přístroje.
Nabijte baterii.

6

Rozehřívání

žádnýPřístroj zahájí provoz automatickým
dávkováním, dokud nezjistí nádech nebo
neskončí rozehřívání (podle toho, co nastanedříve).

7

Nízká
koncentrace

1 pípnutíkaždých
30 sekund,trvá až
5 minut

Pokud problém přetrvává, kontaktujte
dodavatele.

422 – CZEPN MN232-C4 K | Uživatelská příručka

FreeStyle Comfort
Priorita #Popis výstrahy

8

Zobrazení na displeji

Zvukový
signálPodrobnosti

Systémováporucha,průtočné
množství(porucha ventilu
OCD, překážka
v rozvodu plynu)

1 pípnutíkaždých
30 sekund

Pokud problém přetrvává, kontaktujte
dodavatele.

9

Varování vybitébaterie

1 pípnutí,bez
opakováníNabijte baterii.

10Překročená
frekvencedýchání1 pípnutíkaždých
60 sekund

Frekvence dýchání překročila hodnotu
40 nádechů za minutu.

11

Systémováporucha, závada
ventilátoru

žádný

Pokud problém přetrvává, kontaktujte
dodavatele.

12

Systémováporucha, nízký
tlak

žádný

Pokud problém přetrvává, kontaktujte
dodavatele.

13

Systémováporucha, nízká
teplotažádný

Nechte jednotku zahřát na pokojovou
teplotu. Pokud problém přetrvává,
kontaktujte dodavatele.

Informace

Spuštění

1 informační
pípnutíObrazovka se objeví při zapnutí. Mělibyste slyšet jedno pípnutí, abyste sepřesvědčili, že bzučák funguje.

Jestliže je dávka nesprávně spouštěna,snižte nastavení citlivosti. Pokud problémpřetrvává, kontaktujte dodavatele.

Obrazovka spuštění ukazuje v pravém
spodním rohu aktuální verzi firmwaru.
Informace

Obrazovkaběžnéhochodu v režimu
autoSAT (nenípřipojen žádný
napájecí zdroj
stejnosměrného
proudu)žádný

PN MN232-C4 K | Uživatelská příručka

CZE – 423

FreeStyle Comfort
Priorita

Popis výstrahy

Informace

Obrazovkaběžného chodu
s deaktivovaným
autoSAT (nenípřipojen žádný
napájecí zdroj
stejnosměrného
proudu)žádný

InformaceZměna nastavení

1 informační
pípnutíInformacePřipojeníexterního
napájení vevypnutém stavu

žádný

InformacePřipojeníexterního
napájení za
provozužádný

Informace

Odpojeníexterního
napájení vevypnutém stavu

1 informační
pípnutíInformace

Odpojeníexterního
napájeníza provozu(s nainstalovanou
baterií)1 informační
pípnutí424 – CZE

Zobrazení na displeji

PN MN232-C4 K | Uživatelská příručka

Zvukový
signálPodrobnosti

Na 2 sekundy se objeví obrazovka
s logem společnosti CAIRE, potom se na
5 sekund zobrazí stav baterie a připojení
k napájení a následně obrazovka
zhasne.

Na 2 sekundy se objeví obrazovka
s logem společnosti CAIRE a potom se
displej vypne.

FreeStyle Comfort
Priorita

Popis výstrahy

Informace

Zobrazení na displeji

Zvukový
signálPodrobnosti

V pohotovostnímrežimu při
stisknutí tlačítka
„nahoru“ nebo„dolů“ se na
displeji zobrazístav bateriea nabíjení

žádnýObrazovka se objeví na 5 sekund a potom
zhasne.

InformaceRežim Letadlo
při napájení
z baterie

1 dlouhépípnutíInformaceRežim Letadlo
při zapojení dosítě

1 dlouhépípnutíPN MN232-C4 K | Uživatelská příručka

CZE – 425

FreeStyle Comfort

Pokyny pro navigaci v menu
Do menu můžete vstoupit v pohotovostním režimu
nebo v provozním režimu. V provozním režimu,jestliže nastane stav alarmu, bude menu automaticky
opuštěno. Jestliže po dobu 20 sekund nedojde ke
stisknutí žádného tlačítka, menu bude automaticky
opuštěno. V pohotovostním režimu budete možná
muset stisknout tlačítko Menu dvakrát, abyste
jednotku probudili, pak vstoupíte do menu.
Existují čtyři obrazovky menu:
1. Stavová obrazovka: zobrazuje číslo rev. firmwaru,
dobu provozu, cykly vybití baterieRežim Letadlo

2. Režim Letadlo (ZAP./VYP.)
3. Citlivost: nastavitelná (1–5)
4. autoSAT (ZAP./VYP.)

CitlivostPříklad stavové obrazovkyOVLÁDÁNÍ MENU NASTAVENÍ
Do menu Nastavení vstoupíte stisknutím tlačítka
Menu/Zvolit
v provozním nebo pohotovostnímrežimu.
Pro procházení nabídkami posunujte

Stavová stránka

426 – CZEPN MN232-C4 K | Uživatelská příručka.

autoSATFreeStyle Comfort
STAV STRÁNKY
Stavová stránka Freestyle Comfort zobrazuje revizi
firmwaru, provozní hodiny a počet cyklů baterie napřipojené baterii.

Chcete-li zrušit zapnutí režimu Letadlo, stiskněte
když je vidět ikona CANCEL (ZRUŠIT).

,Zapnutí režimu Letadlo potvrzeno

Stavová stránka

Provozní hodiny jsou zobrazeny v celých hodinách.
Vybíjení baterií je počet cyklů, kdy byla připojená
baterie vybita a dobita.

REŽIM LETADLOFreeStyle Comfort se expeduje s vypnutým režimem
Letadlo.

Chcete-li režim Letadlo vypnout, opakujte výšeuvedené, zatímco přepnete na OFF (VYP.).
Stisknutí, když obrazovka ukazuje symbol„Next“ (Další), posune stránku menu k další volitelnémožnosti obrazovky.
Režim Letadlo je určen k použití v provozním režimu.
Režim Letadlo bude automaticky deaktivován, jestliže
bude jednotka ponechána v pohotovostním režimu při
napájení z baterie.

Poznámka: Podrobnější informace o režimu Letadlo
najdete v části Aktivace režimu Letadlo.

REŽIM CITLIVOST

Hlavní obrazovka režimu Letadlo

FreeStyle Comfort se expeduje s výchozím nastavenímz výrobního závodu pro režim Citlivost v hodnotě 2
a za normálních podmínek by neměl vyžadovat úpravu.
Nastavení citlivosti lze seřizovat mezi hodnotami 1–5,
kde 1 je nejnižší citlivost a 5 je nejvyšší citlivost.

Chcete-li tento režim zapnout, stiskněte
, když jevidět ikona CHANGE (ZMĚNIT). Hodnota se přepne
na modré slovo „ON“ (ZAP.).

Obrazovka režimu Citlivost při nastavení 2

Je zvoleno Režim Letadlo ON (zapnutý)Chcete-li upravit režim Citlivost, stiskněte
,když je vidět ikona CHANGE (ZMĚNIT). Hodnota
se přepne na další modrou číslici.

Pro potvrzení zapnutí režimu Letadlo stiskněte
,když je vidět ikona SAVE (ULOŽIT). Hodnota sepřepne na černé slovo „ON“ (ZAP.).

PN MN232-C4 K | Uživatelská příručka

CZE – 427

FreeStyle ComfortREŽIM AUTOSATFreeStyle Comfort se expeduje se zapnutým režimem
autoSAT.

Je zvoleno nastavení 3 režimu Citlivost

Pro potvrzení nastavení 3 v režimu Citlivost stiskněte
, když je vidět ikona SAVE (ULOŽIT). Hodnota sepřepne na černou hodnotu „3“.
Chcete-li zrušit změnu režimu Citlivost, stiskněte
když je vidět ikona CANCEL (ZRUŠIT).

,Hlavní obrazovka režimu autoSAT

Chcete-li tento režim vypnout, stiskněte
, když jevidět ikona CHANGE (ZMĚNIT). Hodnota se přepne
na modré slovo „OFF“ (VYP.).

Nastavení 3 režimu Citlivost potvrzeno
Je zvoleno Režim autoSAT Off (vypnutý)Pro potvrzení vypnutí režimu autoSAT stiskněte
, když je vidět ikona SAVE (ULOŽIT). Hodnota sepřepne na černé slovo „OFF“ (VYP.).
Chcete-li zrušit zapnutí režimu autoSAT, stiskněte
, když je vidět ikona CANCEL (ZRUŠIT).
Nastavení 5 (max.) režimu Citlivost potvrzeno

Stisknutí, když obrazovka ukazuje symbol„Next“ (Další), posune stránku menu k další volitelnémožnosti obrazovky.

Vypnutí režimu autoSAT potvrzeno

Chcete-li režim autoSAT zapnout, opakujte výšeuvedené, zatímco přepnete na ON (ZAP.).
Stisknutí, když obrazovka ukazuje symbol„Next“ (Další), posune stránku menu k další volitelnémožnosti obrazovky.

428 – CZEPN MN232-C4 K | Uživatelská příručka

FreeStyle Comfort
Odstraňování potíží
Problém

Pravděpodobná příčina

Řešení

Přístroj FreeStyle Comfort
zobrazuje výstrahu a vydávánepřetržité pípání.

Viz tabulka „Druhy alarmů“.

Viz tabulka „Druhy alarmů“.

Kyslíkový koncentrátor se
nezapne.

Baterie je vybitá.

Zapojte přístroj do zásuvky se střídavým nebo
stejnosměrným proudem.

Přístroj není připojen
k externímu napájení.
Obecná porucha.

Přesvědčte se, že jsou externí přípojky zajištěné.
Proveďte kompletní cyklus odpojení napájení –vyjměte baterii a odpojte všechny napájecí zdroje
pro střídavý a stejnosměrný proud. Následně
dokončete cyklus opětovnou instalací baterie.
V případě potřeby znovu připojte externí zdroj
napájení.
Kontaktujte dodavatele přístroje a podle potřeby
přejděte na jiný zdroj kyslíku.

Všechny ostatní problémy.

Proveďte kompletní cyklus odpojení napájení –vyjměte baterii a odpojte všechny napájecí zdroje
pro střídavý a stejnosměrný proud. Následně
dokončete cyklus opětovnou instalací baterie.
V případě potřeby znovu připojte externí zdroj
napájení.
Podle dostupnosti přejděte na jiný zdroj kyslíku
a kontaktujte dodavatele přístroje.

Poznámka: Některé výstrahy způsobí vypnutí přístroje.
Poznámka: Během úplného výpadku napájení dojde k resetu výstražného systému (všechny předchozí výstrahy
se vymažou) a výstraha se zobrazí pouze tehdy, když po obnovení napájení daný stav stále existuje.

PN MN232-C4 K | Uživatelská příručka

CZE – 429

FreeStyle Comfort

Informace o cestováníPřístroj FreeStyle Comfort splňuje předpisy
Federální letecké správy (FAA) pro použití na palubě
komerčních letů. Před každou cestou (letadlem,
vlakem, lodí atd.) byste měli:• zajistit, aby byl koncentrátor v dobrém provozním
stavu,• kontaktovat dopravní společnost alespoň 48 hodinpřed odletem/odjezdem a informovat ji o faktu, žebudete cestovat s přístrojem FreeStyle Comfort.
Na cestách se doporučuje mít vždy s sebou náhradníbaterie. Při cestování letadlem počítejte s dostatečným
nabitím baterie kvůli času strávenému na letišti(odbavení před odletem, bezpečnostní prohlídka
atd.) i případným mezipřistáním. Většina leteckých
společností vyžaduje, abyste měli s sebou dostatečnémnožství plně nabitých baterií, které poskytnou
napájení přístroje na minimálně 150 % maximálníočekávané délky trvání celé cesty (délka letů, letištnízáležitosti před odletem a po příletu, následné spoje
a neočekávaná zpoždění).

Poznámka: V případě cestování letadlem doporučujeme mít s sebou náhradní baterie v zabezpečenépřenosné tašce.

PŘEDPISY FEDERÁLNÍ LETECKÉ SPRÁVY
(FAA)

1. Příruční kyslíkový koncentrátor zapněte POUZEs baterií, když není připojen žádný externí zdroj
napájení.

2. Aktivujte režim Letadlo s použitím jednoho
z následujících způsobů:a. Postupujte podle pokynů pro aktivování režimu
Letadlo v části Pokyny pro navigaci v menu.

b. Stiskněte a držte tlačítka Nahoru i Dolů po dobu
5 sekund, dokud neuslyšíte dlouhé pípnutí a dokud
neuvidíte, že se objevil symbol letadla.

Náhradní baterie musí být jednotlivě chráněny proti
zkratům uložením do původního obalu, v němž byly
zakoupeny, nebo jinou izolací jejich pólů, např.
obalením odhalených pólů páskou nebo vložením
každé baterie do samostatné plastové tašky čiochranného vaku. Baterie se smí převážet pouzev příručním zavazadle.

AKTIVACE REŽIMU LETADLO

3. Když je zobrazen symbol letadla, zapojte přenosný
kyslíkový koncentrátor do sítě.

Napájení bateriíPříruční kyslíkový koncentrátor lze v letadle používat
běžným provozním způsobem POUZE v případě
napájení z baterie, kdy není připojen žádný externízdroj napájení.
Externí napájení
Pokud je v letadle dostupná elektrická zásuvka
umístěná tak, že ji lze použít jako zdroj napájení
příručního kyslíkového koncentrátoru, může posloužit
jako záložní řešení pro baterie. V tomto případě jenutné dodržovat následující doporučený postup propřechod z napájení baterií na napájení z elektrické sítěletadla:430 – CZEPN MN232-C4 K | Uživatelská příručka

Poznámka: Kvůli omezením na palubách letadel
nedochází v režimu Letadlo k nabíjení baterií.

FreeStyle ComfortPříslušenstvíZa účelem dosažení správného výkonu a bezpečnostipoužívejte pouze níže uvedené příslušenství
společnosti CAIRE dodávané prostřednictvím vašeho
dodavatele kyslíku. Použití příslušenství, které neníuvedeno níže, by mohlo nepříznivě ovlivnit výkon
nebo bezpečný provoz koncentrátoru.
Standardní příslušenství přístroje FreeStyle ComfortPřenosná taška

MI396-1

Batoh

MI459-1

Ramenní popruh

MI397-1

Napájecí zdroj pro střídavý proud(včetně všech kabelů)PW036-1S

Napájecí kabel pro stejnosměrný
proud

CD041-1

Externí filtr hrubých částic (pravý)FI226-1

Externí filtr hrubých částic (levý)FI227-1VLOŽENÍ PŘÍSTROJE DO PŘENOSNÉ
TAŠKY
1. Vložte přístroj do spodní části tašky a řádně
zarovnejte filtry a otvory baterie.

Dostupné volitelné příslušenstvíZáložní 8článková baterie

BT034-1S

16článková baterie

BT035-1S

Stolní nabíječka bateriíBT036-1Přenosná taška pro veškeré
příslušenstvíMI372-2
2. Zajistěte pomocí zipu kolem spodní části vaku.

Poznámka: Další možnosti mohou být dostupné pro
napájecí kabely v konkrétních zemích (je-li uvedenovýše). Potřebujete-li objednat alternativní příslušenství, kontaktujte společnost CAIRE nebo svého
dodavatele kyslíku.

Pokyny k přenosné tašce
Za účelem vložení přístroje FreeStyle Comfort dopřenosné tašky a jeho zajištění dodržujte následující
pokyny.

INSTALACE RAMENNÍHO POPRUHU
1. Ramenní popruh lze připevnit buď přímo k přístroji,nebo k přenosné tašce.
2. Připněte popruh ke kroužku na přístroji nebo natašce.

PN MN232-C4 K | Uživatelská příručka

CZE – 431

FreeStyle Comfort

Likvidace a opětovné použití
Za účelem řádné likvidace vždy vraťte přístroj
FreeStyle Comfort včetně všech součástí poskytovateli
vybavení. Můžete také kontaktovat místní městský
úřad a požádat o pokyny k řádné likvidaci baterie.
Jednotka může být znovu použita mezi lidmi, ale musíbýt vrácena poskytovateli vybavení k přepracování.

Směrnice OEEZ a RoHS
Tento symbol má majitelům zařízení
připomenout, že jej mají podle směrnice o odpadních
elektrických a elektronických zařízeních (OEEZ) na

konci životnosti odevzdat sběrnému recyklačnímu
zařízení.
Naše výrobky odpovídají směrnici o omezenípoužívání některých nebezpečných látek v elektrických
a elektronických zařízeních (RoHS). Neobsahují vícenež stopové množství olova ani jiných nebezpečných
materiálů.
UPOZORNĚNÍ: Za účelem řádné likvidacepožádejte o pokyny dodavatele zařízení
nebo místní státní úřad.

Materiály, se kterými se uživatel dostane do přímého kontaktu
Kryt koncentrátoru....................................................................................................................................... PC/ABS
Ovládací panel koncentrátoru........................................................................Polyester EBG7 nebo jeho ekvivalent
Rukojeť koncentrátoru......................................................................................................................... PolykarbonátVýstup kyslíku, tryska.......................................................................................................................... PolykarbonátŠtítek jednotky.................................................................................................................................................Lexan
Konektory kabelu.............................................................................................................Polykarbonát/vinylchlorid
Napájecí kabel............................................................................................................................................ PVC, kov
Napájecí zdroj.................................................................................................................. Lexan 940 (polykarbonát)
Baterie.......................................................................................................................................................... PC/ABSŠtítky baterie a napájecích zdrojů.................................................................................................Polyesterová fóliePřenosná taška koncentrátoru........................................................................ Cordura, neopren, nylonová podšívka
Popruh...............................................................................................Polypropylen, acetyl, vinyl, Cordura, neopren

Testování elektromagnetické kompatibility (EMC)
Pokyny a prohlášení výrobce – elektromagnetické emisePřístroj FreeStyle Comfort je určen k použití v níže specifikovaném elektromagnetickém prostředí. Zákazník nebo uživatelpřístroje FreeStyle Comfort by měl zajistit, aby byl přístroj používán v takovém prostředí.
Zkouška emisí
RF emiseCISPR 11
RF emiseCISPR 11
Harmonické emise,
IEC 61000-3-2
Kolísání napětí /generování kmitavého
signálu,
IEC 61000-3-3
432 – CZE

Kompatibilita

Elektromagnetické prostředí – pokyny

Skupina 1Přístroj FreeStyle Comfort používá radiofrekvenční energii pouze
pro své vnitřní funkce. Jeho vysokofrekvenční emise jsou tedy velmi
nízké a je nepravděpodobné, že by mohly způsobovat rušení blízkých
elektronických zařízení.

Třída BTřída A
Kompatibilní

PN MN232-C4 K | Uživatelská příručkaPřístroj FreeStyle Comfort je vhodný k použití ve všech zařízeníchvčetně domácností a zařízení přímo napojených na veřejnou síť
nízkého napětí, která napájí budovy užívané za účelem bydlení.

FreeStyle Comfort
Pokyny a prohlášení výrobce – elektromagnetická odolnostPřístroj FreeStyle Comfort je určen k použití v níže specifikovaném elektromagnetickém prostředí. Zákazník nebo uživatelpřístroje FreeStyle Comfort by měl zajistit, aby byl přístroj používán v takovém prostředí.
Zkouška odolnosti IEC 60601, zkušební
Úroveň kompatibility
Elektromagnetické prostředí – pokynyúroveň
Podlahy by měly být dřevěné, betonové
Elektromagnetické±8 kV kontakt±8 kV kontaktnebo z keramické dlažby. Jsou-li podlahy
prostředí – pokyny
pokryty syntetickým materiálem, relativní±15 kV vzduch±15 kV vzduch
IEC 61000-4-2
vlhkost by měla dosahovat alespoň 30 %.
±2 kV pro napájecí
±2
kV
pro
napájecíRychlé elektrické
Kvalita zdroje napájení by měla být
vedení
vedení
přechodné jevy /na úrovni typického profesionálního
skupiny impulzů
zdravotnického prostředí a typického±1 kV pro vstupní/
prostředí pro domácí zdravotní péči.
IEC 61000-4-4
Nehodí sevýstupní vedení±1 kV mezi vedeními

Rázový impulz
IEC 61000-4-5±2 kV v běžném režimu
na střídavém vedení±1 kV v diferenčnímrežimu na střídavém
vedení

±1 kV mezi vedeními

Kvalita zdroje napájení by měla být
na úrovni typického profesionálního
zdravotnického prostředí a typického
prostředí pro domácí zdravotní péči.

±2 kV v běžném režimu
na vnějším vst./výst.
vedení
0 % UT po dobu 0,5 cyklu 0 % UT po dobu 0,5 cyklu(0°, 45°, 90°, 135°, 180°, (0°, 45°, 90°, 135°, 180°,
225°, 270°, 315°)
225°, 270°, 315°)
Krátkodobé poklesynapětí, krátkápřerušení a pomalézměny napětí napřívodním vedenínapájecích zdrojů,IEC 61000-4-11

Síťová frekvence(50/60 Hz),magnetické pole,
IEC 61000-4-8

Kvalita zdroje napájení by měla být
na úrovni typického profesionálního
zdravotnického prostředí a typického
prostředí pro domácí zdravotní péči.
Pokud uživatel přístroje FreeStyle Comfortvyžaduje nepřetržitý provoz během
70 % UT (30% pokles
70 % UT (30% pokles
v UT) po dobu 25/30 cyklů v UT) po dobu 25/30 cyklů přerušovaného napájení, doporučuje senapájení přístroje FreeStyle Comfort ze
(00)
(00)
zdroje nepřetržitého napájení nebo baterie.

0 % UT po dobu 1 cyklu(00)

0 % UT po dobu 1 cyklu(00)

0 % UT po dobu
250/300 cyklů (00)

0 % UT po dobu
250/300 cyklů (00)

3 A/m

3 A/m

POZNÁMKA: UT je střídavé napětí sítě před použitím zkušební úrovně.

Magnetická pole síťové frekvence byměla být na úrovních charakteristických
pro typické místo v profesionálním
zdravotnickém prostředí nebo prostředí pro
domácí zdravotní péči.

POZNÁMKA: V určitých případech může dojít k restartování přístroje FreeStyle Comfort v reakci na výrazný
elektrostatický výboj. Za účelem obnovení běžného provozu může být nutné provést kompletní cyklus napájení.
Podrobnější informace najdete v části „Řešení potíží“.

PN MN232-C4 K | Uživatelská příručka

CZE – 433

FreeStyle Comfort
Pokyny a prohlášení výrobce – elektromagnetická odolnostPřístroj FreeStyle Comfort je určen k použití v níže specifikovaném elektromagnetickém prostředí. Zákazník nebo uživatelpřístroje FreeStyle Comfort by měl zajistit, aby byl přístroj používán v takovém prostředí.
Zkouška
IEC 60601, zkušební
Úroveň kompatibility
Elektromagnetické prostředí – pokyny
odolnostiúroveň
RF vedením
10 Vrms
10 VrmsPřenosná a mobilní radiofrekvenční
komunikační zařízení by se neměla používat
IEC 61000-4-6
150 kHz až 80 MHz
v menší vzdálenosti od jakékoli části
přístroje FreeStyle Comfort včetně kabelů,než je doporučená oddělovací vzdálenost
vypočítaná z rovnice platné pro frekvenci
RF zářením
10 V/m
10 V/mvysílače.
IEC 61000-4-3
80 MHz až 2700 MHz
d = 1,2√ P
Doporučená oddělovací vzdálenostd = 1,2√ P = 80 MHz až 800 MHz
d = 1,2√ P = 800 MHz až 2700 MHz
kde P je maximální jmenovitý výstupnívýkon vysílače ve wattech (W) podleúdajů výrobce vysílače a d je doporučenáoddělovací vzdálenost v metrech (m).
Intenzita pole pevných radiofrekvenčních
vysílačů zjištěná elektromagnetickým
průzkumem lokalitya by měla být nižší než
povolená úroveň v každém z frekvenčních
pásem.b
K rušení může docházet v blízkosti zařízení
označených následujícím symbolem:POZNÁMKA 1: U frekvencí 80 MHz a 800 MHz platí vyšší frekvenční rozsah.
POZNÁMKA 2: Tyto pokyny nemusí platit ve všech situacích. Na šíření elektromagnetických vln má vliv míra jejichpohlcování budovami, předměty a lidmi i míra jejich odrazu od nich.
aIntenzitu pole pevných vysílačů, například základnových stanic radiových (mobilních/bezdrátových) telefonů a pozemníchmobilních radiostanic, amatérských vysílaček, rozhlasového vysílání na AM a FM frekvencích a televizního vysílání,nelze předem teoreticky přesně stanovit. K posouzení elektromagnetického prostředí z hlediska přítomnosti pevných
radiofrekvenčních vysílačů je třeba zvážit provedení elektromagnetického průzkumu lokality. Přesahuje-li naměřená
intenzita pole v lokalitě, kde je zařízení FreeStyle Comfort používáno, příslušnou výše uvedenou radiofrekvenční úroveň,je třeba ověřit běžný provoz přístroje FreeStyle Comfort. Vykazuje-li přístroj FreeStyle Comfort během provozu odchylky,může být nutné provedení určitých opatření, např. změna orientace přístroje nebo jeho přemístění.
b

Ve frekvenčním rozsahu 150 kHz až 80 MHz by měla být intenzita pole nižší než 10 V/m.

434 – CZEPN MN232-C4 K | Uživatelská příručka

FreeStyle Comfort
Doporučené oddělovací vzdálenosti mezi přenosnými a mobilními radiofrekvenčními
komunikačními zařízeními a přístrojem FreeStyle ComfortPřístroj FreeStyle Comfort je určen k použití v elektromagnetickém prostředí, v němž je vyzařované radiofrekvenční rušení
regulováno. Zákazník nebo uživatel přístroje FreeStyle Comfort může předcházet elektromagnetickému rušení tím, že budedodržovat níže uvedenou minimální vzdálenost mezi přenosnými a mobilními radiofrekvenčními komunikačními zařízeními(vysílači) a přístrojem FreeStyle Comfort podle maximálního výstupního výkonu komunikačního zařízení.
Jmenovitý maximální
Vzdálenost podle frekvence vysílače
výstupní výkon
mvysílače
150 kHz až 80 MHz
80 MHz až 800 MHz
800 MHz až 2,5 GHz

d = 1,2√ P

d = 1,2√ P

d = 1,2√ P

W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
U vysílačů se jmenovitým maximálním výstupním výkonem, jenž není uveden výše, lze doporučenou vzdálenost dv metrech (m) odhadnout pomocí rovnice platné pro frekvenci vysílače, kde P je maximální jmenovitý výstupní výkon
vysílače ve wattech (W) podle údajů výrobce vysílače.
POZNÁMKA 1: U frekvencí 80 MHz a 800 MHz platí vzdálenost pro vyšší frekvenční rozsah.
POZNÁMKA 2: Tyto pokyny nemusí platit ve všech situacích. Na šíření elektromagnetických vln má vliv míra jejichpohlcování budovami, předměty a lidmi i míra jejich odrazu od nich.

Klasifikace

Druh ochrany před úrazem elektrickým proudem:Třída II	Ochrany před úrazem elektrickým
proudem je dosaženo použitím dvojité
izolace.
Stupeň ochrany před úrazem elektrickým proudem:
Typ BF Zařízení poskytující zvláštní stupeň
ochrany před úrazem elektrickým proudem týkající se		 1) přípustného svodového proudu,		2) spolehlivosti ochranného uzemnění (je-li
k dispozici).
		 Není určeno k přímému použití na srdci.
Nezávislé testování podle normy pro zdravotnické
elektrické přístroje:
Testováno za účelem souladu s předpisy• IEC 60601-1, vydání 3.1: Zdravotnické
elektrické přístroje – část 1: Všeobecnépožadavky na bezpečnost.
• CAN/CSA C22.2 No. 60601-1-8, vydání 2.1,
Zdravotnické elektrické přístroje – část 1:
Všeobecné požadavky na bezpečnost.
Ochrana před potenciálním elektromagnetickým nebojiným rušením mezi přístrojem a jinými zařízeními.
	• Testováno za účelem souladu s normou
EN 60601-1-2 / IEC 60601-1-2: 4. vydání.

	• Testováno za účelem souladu s normou RTCA/
DO-160, část 21, kategorie M.
	• CISPR 11 / EN 55011 třída B, skupina 1,„Průmyslové, vědecké a zdravotnické (ISM)zařízení“.
	• FCC, část 15, oddíl B – třída B Neúmyslné zdrojezáření
IP 22 – Ochrana před vniknutím prstů nebo podobných
těles s průměrem větším než 12,5 mm. Svisle
kapající voda by neměla mít škodlivý vliv, pokudje přístroj nakloněn pod úhlem 15° od své běžné
polohy.
Povolené metody čištění a kontroly infekce: Viz část„Čištění, péče a pravidelná údržba“ této uživatelsképříručky k přístroji FreeStyle.
Stupeň bezpečnosti použití za přítomnosti hořlavých
anestetických plynů:Přístroj není určen k tomuto druhu použití.
Provozní režim: Nepřetržitý provoz.
CAIRE a CAIRE Inc. jsou registrované ochranné známky společnosti CAIRE
Inc. Úplný seznam ochranných známek naleznete na níže uvedené webové
stránce. Ochranné známky: www.careinc.com/corporate/trademarks.
Copyright © 2021 CAIRE Inc. Společnost CAIRE Inc. si vyhrazuje právo
kdykoli bez předchozího upozornění a bez dalších závazků či důsledků
přestat vyrábět své výrobky nebo změnit ceny, materiály, zařízení, kvalitu,
popisy, specifikace svých výrobků nebo procesy související s těmito
výrobky. Veškerá práva, která zde nejsou výslovně uvedena, si vyhrazujeme
jako uplatnitelná.
PN MN232-C4 K | Uživatelská příručka

CZE – 435

FreeStyle Comfort

Elementy sterowania i wskaźniki stanu systemu
ISO 7000: Symbole graficzne stosowane
w urządzeniach — indeks i streszczenie
Przed rozpoczęciem użytkowania
przeczytać instrukcję obsługi.
Reg. # 1641

Dyrektywa Rady 93/42/EWG dotycząca wyrobów
medycznych
Autoryzowany przedstawiciel
w Europie.
Jeżeli na etykiecie niepowtarzalnego
identyfikatora urządzenia (UDI)
znajduje się symbol CE ####,
urządzenie to spełnia wymagania
dyrektywy 93/42 / EWG dotyczącej
wyrobów medycznych. Symbol
CE #### wskazuje numer jednostki
notyfikowanej.

Zakres temperatur przechowywania lub
pracy. Reg. # 0632

Zakres wilgotności podczas
przechowywania. Reg. # 2620

IEC 60417

Limit ciśnienia atmosferycznego.
Reg. # 2621

Urządzenie klasy II, z podwójną
izolacją. Reg. # 5172

Nazwa i adres producenta. Reg. # 3082

Przełącznik zasilania. Reg. # 5009

Data produkcji. Reg. # 2497

Certyfikat ETL
Agencja bezpieczeństwa
CAN/CSA C22.2 No. 601.1 M90 dla
elektrycznych urządzeń medycznych.

Numer katalogowy. Reg. # 2493
Numer seryjny. Reg. # 2498
Góra. Reg. # 0623

Symbole wewnętrzne
Przechowywać z dala od materiałówłatwopalnych, olejów i smarów.

Ostrożnie. Reg. # 0621
Chronić przed deszczem.
Przechowywać w suchym miejscu.
Reg. # 0626

Nie rozbierać.

Maksymalna liczba opakowań, któremożna na sobie ustawić. Reg. # 2403

Zwiększanie przepływu.

ISO 7010: Symbole graficzne — kolory i znaki
ostrzegawcze — zarejestrowane znaki ostrzegawcze

Zmniejszanie przepływu.

Przeczytać instrukcję obsługi.
Reg. # M002
Przechowywać z dala od nieosłoniętego
płomienia, ognia i źródeł iskrzenia.
Chronić przed źródłem zapłonu.
Zakaz palenia. Reg. # P003
Nie palić w pobliżu urządzenia ani
podczas jego obsługi. Reg. # P002Część klasy BF (stopień ochrony przedporażeniem elektrycznym). Reg. # 5333
Ostrzeżenie. Reg. # W001

436 – POLPN MN232-C4 K | Instrukcja obsługi

Przycisk menu.
Zgodnez RTCA/
DO-160
sekcja 21
kategoria M

Zgodne z RTCA DO160 sekcja 21,
kategoria M. Wymaganie FAA
SFAR 106.
Urządzenie klasy II, podwójnie
izolowane.
Zatwierdzone przez FAA — przenośny
koncentrator tlenu.

FreeStyle Comfort
21 CFR 801.15: Kodeks przepisów federalnych, tytuł
21
Prawo federalne dopuszcza sprzedaż
tego urządzenia wyłącznie przez lekarza
lub na jego zlecenie.
Dyrektywa Rady 2012/19/UE w sprawie zużytegosprzętu elektrycznego i elektronicznego (WEEE)
WEEE

Czym jest koncentrator tlenu?
Powietrze, którym oddychamy, zawiera: tlen
(około 21%), azot (78%) i inne gazy (1%).
W urządzeniu FreeStyle® Comfort® za pomocą
specjalnych wlotów pobierane jest powietrze
z otoczenia. Następnie jest ono filtrowane przez
materiał absorbujący zwany sitem cząsteczkowym.
Zatrzymuje on azot, przepuszczając jedynie tlen.
W efekcie użytkownikowi podawane jest powietrzez wysokim stężeniem tlenu.
FreeStyle Comfort to połączenie koncentratora
i oszczędzacza tlenu, którego efektem jest przenośne
urządzenie zapewniające maksymalną mobilność
i aktywność. Zaawansowany system podawania
impulsowego szybko wykrywa, kiedy użytkownik
wdycha powietrze, i podaje impuls tlenu na początku
każdego wdechu. Podawanie impulsowe doskonale
sprawdza się, gdy użytkownik jest aktywny lub
znajduje się poza domem. W odróżnieniu od trybu
ciągłego przepływu, pozwala oszczędzać tlen
i wydłuża czas pracy na akumulatorze. Urządzenie
FreeStyle Comfort jest na tyle lekkie, że można
je nosić na pasku na ramieniu. Może być zasilaneprądem AC (gniazdo ścienne), DC (samochód) lub
z akumulatora.

Przeznaczenie
Koncentrator tlenu FreeStyle Comfort CAIRE jest
przeznaczony do podawania dodatkowego tlenu.
Urządzenie nie jest przeznaczone do podtrzymywaniażycia ani nie zapewnia możliwości monitorowania
pacjenta.

Dlaczego lekarz przepisał podawanie
tlenu?
Wiele osób cierpi na różne choroby serca, płuc i dróg
oddechowych. Niejednokrotnie tlenoterapia może im
pomóc w codziennym życiu w domu i w podróży.
Tlen to gaz stanowiący nawet 21% powietrza, którym
oddychamy. Jego ciągłe dostarczanie jest warunkiem
prawidłowego funkcjonowania naszego organizmu.

IEC 60601-1: Medyczne urządzenia elektryczne— część 1: Wymagania ogólne dotyczące
bezpieczeństwa podstawowego oraz funkcjonowania
zasadniczegoSprzęt kroploszczelny
Ten produkt może być chroniony jednym lub wieloma
patentami w Stanach Zjednoczonych i innych krajach.
Odpowiednie patenty wyszczególnione są w naszej
witrynie pod adresem podanym poniżej. Pat.: www.
caireinc.com/corporate/patents/.

Lekarz przepisuje ustawienie przepływu tlenu
dostosowane do choroby układu oddechowego, naktórą cierpi użytkownik.
Pomimo że tlen jest lekiem nieuzależniającym,samowolne stosowanie go może być niebezpieczne.
Przed użyciem koncentratora tlenu należy zasięgnąć
porady lekarza. Dostawca sprzętu do podawania tlenu
zaprezentuje, jak ustawić przepisany przepływ gazu.

Profil operatora
Koncentratory służą do suplementacji tlenuu użytkowników doświadczających dyskomfortu
z powodu schorzeń wpływających na zdolność płuc
do transportowania tlenu z powietrza do układu
krwionośnego. W przenośnych koncentratorach tlenu
nie przechowuje się tlenu. Nie trzeba ich uzupełniać,a ładować je można wszędzie tam, gdzie dostępna jest
sieć AC lub DC (z wyjątkiem samolotów). Dzięki temuużytkownicy mogą w dużym stopniu samodzielnie
obsługiwać urządzenie oraz prowadzić bardziej aktywny
i mobilny tryb życia w domu i poza nim. Z koncentratora
tlenu należy korzystać wyłącznie na zalecenie lekarza.
Nie jest to system podtrzymywania życia.
Mimo że tlenoterapię można przepisać pacjentom
w dowolnym wieku, użytkownikami koncentratorów
są zazwyczaj osoby w wieku powyżej 65 lat
cierpiące na różne choroby układu oddechowego,na przykład przewlekłą obturacyjną chorobę płuc
(POChP). Użytkownik powinien mieć dobre
zdolności poznawcze i musi być w stanie informować
o potencjalnie odczuwanym dyskomforcie. Jeśli
użytkownik nie jest zdolny do informowaniao dyskomforcie lub nie jest w stanie przeczytać ani
zrozumieć etykiet i instrukcji koncentratora, zaleca się,aby użytkowanie odbywało się pod nadzorem osoby
posiadającej takie zdolności. Jeśli podczas korzystaniaz koncentratora odczuwany jest dyskomfort, należy
skontaktować się z pracownikiem służby zdrowia.
Zaleca się, aby na wypadek braku zasilania lub awarii
koncentratora użytkownik wyposażył się w zapasoweźródło tlenu (np. tlen w butli). Do korzystaniaz koncentratora nie są potrzebne żadne inne specjalne
umiejętności.
PN MN232-C4 K | Instrukcja obsługi

POL – 437

FreeStyle Comfort

Rozpakowywanie koncentratora FreeStyle ComfortNależy upewnić się, że
w opakowaniu znajdują się
wszystkie wyszczególnione
komponenty. Jeśli stwierdzone
zostaną braki, należy natychmiast
skontaktować się z dostawcą
rozwiązania do podawania tlenu.

Akcesoria standardowe• Przenośny koncentrator tlenu• Torba• Pasek na ramię
• Akumulator 8-ogniwowy• Zasilacz prądu przemiennego (AC)• Przewód zasilania AC• Zasilacz prądu stałego (DC)• Instrukcja obsługi

Akcesoria opcjonalne• Dodatkowy akumulator
8-ogniwowy• Akumulator 16-ogniwowy• Ładowarka biurkowa• Torba na wszystkie akcesoria

Podstawowe informacje na temat koncentratora FreeStyle ComfortPoniżej opisano ważne części oraz interfejs użytkownika koncentratora FreeStyle Comfort.

Widok z przodu
Ucho mocowania paska

Wylotpowietrza

InterfejsużytkownikaZłączewylotowe
tlenu

Wlot/filtry
powietrza
Wlot/filtry
powietrza

Zatyczka
gniazdazasilania
Akumulator
Przycisk zwalniania
akumulatora
438 – POLPN MN232-C4 K | Instrukcja obsługi

FreeStyle Comfort
Widok z góryWyświetlacz
Zwiększanie
przepływu

Przycisk zasilania

Przycisk menu

Zmniejszanie
przepływu

Wyświetlacz na akumulatorze
Ustawienie przepływuWskaźnik stanu
akumulatoraWskaźnik autoSATWyświetlacz
alarmuProcentowy
poziomnaładowania
akumulatoraWskaźnikdostarczania bolusaWyświetlacz na zasilaczu AC/DC

Ustawienie przepływuWskaźnik stanu
akumulatora

Miejscewyświetlania alarmuWskaźnikładowania

PN MN232-C4 K | Instrukcja obsługi

POL – 439

FreeStyle ComfortWażne!

Definicje instrukcji bezpieczeństwa:
OSTRZEŻENIE: WAŻNA INFORMACJA
DOTYCZĄCA ZAGROŻEŃ, KTÓRE MOGĄSPOWODOWAĆ POWAŻNE OBRAŻENIA.
PRZESTROGA: Ważna informacja
pozwalająca uniknąć uszkodzeniaurządzenia FreeStyle Comfort.

Uwaga: Informacja wymagająca szczególnej uwagi.

Wskazania do zastosowania
OSTRZEŻENIE: ZAWSZE USTAWIAĆ
PRZEPISANY PRZEPŁYW TLENU.
USTAWIENIE PRZEPŁYWU MOŻNAZMIENIAĆ TYLKO NA POLECENIE
LEKARZA.
OSTRZEŻENIE: PRAWO FEDERALNE (STANYZJEDNOCZONE) DOPUSZCZA SPRZEDAŻ TEGOURZĄDZENIA WYŁĄCZNIE PRZEZ LEKARZA LUB
PRACOWNIKA SŁUŻBY ZDROWIA ALBO NA JEGOZLECENIE.
OSTRZEŻENIE: URZĄDZENIE TO NIE JEST
SYSTEMEM PODTRZYMYWANIA ŻYCIA. OSOBYW PODESZŁYM WIEKU, DZIECI ORAZ INNIUŻYTKOWNICY, KTÓRZY NIE SĄ W STANIEINFORMOWAĆ O DYSKOMFORCIE, MOGĄ WYMAGAĆ
DODATKOWEGO MONITOROWANIA PODCZASKORZYSTANIA Z TEGO KONCENTRATORA.
UŻYTKOWNICY Z WADAMI SŁUCHU LUB WZROKUMOGĄ WYMAGAĆ POMOCY W MONITOROWANIU
ALARMÓW.
OSTRZEŻENIE: USTAWIENIA TRYBU IMPULSOWEGONALEŻY SKONFIGUROWAĆ DLA KAŻDEGO
UŻYTKOWNIKA ODDZIELNIE, BIORĄC POD UWAGĘJEGO POTRZEBY PODCZAS ODPOCZYNKU,ĆWICZEŃ I W PODRÓŻY.
OSTRZEŻENIE: CZASAMI CZYNNOŚĆ ODDECHOWA
MOŻE NIE WYWOŁAĆ AKTYWACJI FUNKCJI
OSZCZĘDZANIA.
OSTRZEŻENIE: PRODUKTU NIE NALEŻY STOSOWAĆU PACJENTÓW PO TRACHEOTOMII.
OSTRZEŻENIE: KOBIETY W CIĄŻY ANI KARMIĄCE NIE
POWINNY STOSOWAĆ AKCESORIÓW ZALECANYCHW TEJ INSTRUKCJI, PONIEWAŻ MOGĄ ONEZAWIERAĆ FTALANY.

440 – POLPN MN232-C4 K | Instrukcja obsługi

Przeciwwskazania do zastosowania
OSTRZEŻENIE: W NIEKTÓRYCHOKOLICZNOŚCIACH STOSOWANIE TLENU
BEZ ZALECENIA LEKARZA MOŻE BYĆ
NIEBEZPIECZNE. UŻYWAĆ URZĄDZENIA,
TYLKO JEŚLI ZOSTAŁO PRZEPISANEPRZEZ LEKARZA.
OSTRZEŻENIE: NIE STOSOWAĆ W OBECNOŚCI
ŁATWOPALNYCH ŚRODKÓW ZNIECZULAJĄCYCH.
OSTRZEŻENIE: KONCENTRATOR JEST
URZĄDZENIEM ELEKTRYCZNYM, W ZWIĄZKUZ CZYM NIE DZIAŁA PODCZAS PRZERWW ZASILANIU. MOŻE TEŻ WYMAGAĆ SERWISOWANIAPRZEZ WYKWALIFIKOWANEGO SPECJALISTĘ.
NIE STOSOWAĆ KONCENTRATORA TLENU
U UŻYTKOWNIKÓW, W PRZYPADKU KTÓRYCHTEGO RODZAJU PRZERWY W DZIAŁANIU MOGŁYBY
DOPROWADZIĆ DO NIEPOŻĄDANYCH SKUTKÓWZDROWOTNYCH.

Wytyczne dotyczące bezpieczeństwa
OSTRZEŻENIE: MODYFIKOWANIE TEGOURZĄDZENIA JEST ZABRONIONE.
OSTRZEŻENIE: TO URZĄDZENIE DOSTARCZAMIESZANKĘ O WYSOKIM STĘŻENIU TLENU,
KTÓRA POWODUJE SZYBKIE SPALANIE.
W POMIESZCZENIACH, W KTÓRYCH JEST (1) TOURZĄDZENIE LUB ZNAJDUJĄ SIĘ (2) AKCESORIAZAWIERAJĄCE TLEN, NIE WOLNO PALIĆ ANIUŻYWAĆ OTWARTEGO OGNIA. ZIGNOROWANIETEGO OSTRZEŻENIA MOŻE DOPROWADZIĆ DOPOŻARU, USZKODZENIA MIENIA, OBRAŻEŃ CIAŁA,A NAWET ŚMIERCI.
OSTRZEŻENIE: NIE UŻYTKOWAĆ URZĄDZENIAW OGRANICZONEJ LUB ZAMKNIĘTEJ PRZESTRZENI(NP. MAŁEJ WALIZCE LUB TOREBCE), GDZIEWENTYLACJA JEST UTRUDNIONA. MOŻETO SPOWODOWAĆ PRZEGRZEWANIE SIĘ
KONCENTRATORA TLENU I MIEĆ NEGATYWNYWPŁYW NA JEGO PRACĘ.
OSTRZEŻENIE: USTAWIAJĄC KONCENTRATOR,UNIKAĆ DYMU, ZANIECZYSZCZEŃ POWIETRZAI OPARÓW.
OSTRZEŻENIE: W PRZYPADKU WYSTĄPIENIAPOWAŻNEGO INCYDENTU Z TYM URZĄDZENIEM,
UŻYTKOWNIK POWINIEN NATYCHMIAST ZGŁOSIĆ
INCYDENT DOSTAWCY I / LUB PRODUCENTOWI.
POWAŻNY INCYDENT JEST DEFINIOWANY JAKO
URAZ, ŚMIERĆ LUB MOŻLIWOŚĆ SPOWODOWANIA
OBRAŻEŃ / ŚMIERCI W PRZYPADKU PONOWNEGO
WYSTĄPIENIA ZDARZENIA. UŻYTKOWNIK MOŻERÓWNIEŻ ZGŁOSIĆ INCYDENT WŁAŚCIWEMU
ORGANOWI W KRAJU, W KTÓRYM ZDARZENIEMIAŁO MIEJSCE.

FreeStyle Comfort
OSTRZEŻENIE: KORZYSTANIEZ NIEKTÓRYCH AKCESORIÓW DO
PODAWANIA TLENU, KTÓRYCH
NIE WYSZCZEGÓLNIONO WŚRÓD
KOMPONENTÓW DO STOSOWANIA Z TYM
URZĄDZENIEM, MOŻE MIEĆ NEGATYWNYWPŁYW NA JEGO DZIAŁANIE.
ZALECANE AKCESORIA WYMIENIONE SĄW NINIEJSZEJ INSTRUKCJI.
OSTRZEŻENIE: JEŚLI DOSZŁO DO UPUSZCZENIA,
USZKODZENIA LUB ZAMOCZENIA KONCENTRATORATLENU, SKONTAKTOWAĆ SIĘ Z DOSTAWCASPRZĘTU ZDROWOTNEJ W CELU SPRAWDZENIAI W RAZIE POTRZEBY NAPRAWY. NIE UŻYWAĆ
KONCENTRATORA TLENU, JEŚLI MA USZKODZONEPRZEWÓD LUB WTYCZKĘ ZASILANIA.
OSTRZEŻENIE: ABY UNIKNĄĆ USZKODZENIA LUB
OBRAŻEŃ PODCZAS TRANSPORTU W POJEŹDZIE,ODPOWIEDNIO ZABEZPIECZYĆ, PRZYPIĄĆ PASEMLUB ZAPEWNIĆ INNĄ OCHRONĘ KONCENTRATORATLENU.
OSTRZEŻENIE: PRZEPISY DEPARTAMENTU
TRANSPORTU STANÓW ZJEDNOCZONYCH ORAZ
ORGANIZACJI NARODÓW ZJEDNOCZONYCH(ONZ) WYMAGAJĄ ODŁĄCZENIA AKUMULATORAOD KONCENTRATORA TLENU W PRZYPADKUODPRAWIANIA GO JAKO BAGAŻ PRZED WSZELKIMILOTAMI MIĘDZYNARODOWYMI. PRZYGOTOWUJĄC
KONCENTRATOR DO WYSYŁKI, RÓWNIEŻ NALEŻY
ODŁĄCZYĆ I ODPOWIEDNIO ZAPAKOWAĆAKUMULATOR.

OSTRZEŻENIE: UWAŻAĆ, ABY NIE ZABLOKOWAĆ
WLOTÓW ANI WYLOTU POWIETRZA. NIE
WRZUCAĆ ANI NIE WKŁADAĆ PRZEDMIOTÓW
DO OTWORÓW W OBUDOWIE URZĄDZENIA.
MOŻE TO SPOWODOWAĆ PRZEGRZEWANIESIĘ KONCENTRATORA TLENU I MIEĆ NEGATYWNYWPŁYW NA JEGO PRACĘ.
OSTRZEŻENIE: NIEPRAWIDŁOWE UŻYWANIEAKUMULATORA MOŻE DOPROWADZIĆ DO JEGOROZGRZANIA LUB ZAPŁONU, A W EFEKCIE DOPOWAŻNYCH OBRAŻEŃ. AKUMULATORA NIE WOLNOPRZEBIJAĆ, UDERZAĆ, DEPTAĆ, UPUSZCZAĆ ANI
W INNY SPOSÓB NARAŻAĆ NA SILNE UDARY LUB
WSTRZĄSY. KORZYSTANIE Z USZKODZONEGOAKUMULATORA MOŻE DOPROWADZIĆ DO OBRAŻEŃ
CIAŁA.
OSTRZEŻENIE: NIE ZWIERAĆ METALOWYCHSTYKÓW AKUMULATORA METALOWYMIPRZEDMIOTAMI, TAKIMI JAK KLUCZE CZY MONETY.
MOŻE TO WYGENEROWAĆ ISKRY LUB DUŻĄ ILOŚĆ
CIEPŁA.
OSTRZEŻENIE: JEŚLI PRZENOŚNY KONCENTRATORTLENU ZNAJDUJE SIĘ NA ZEWNĄTRZI JEST ZASILANY PRĄDEM PRZEMIENNYM,PODŁĄCZAĆ ZASILACZ WYŁĄCZNIE DO GNIAZDAZ WYŁĄCZNIKIEM RÓŻNICOWOPRĄDOWYM.
OSTRZEŻENIE: NIE USTAWIAĆ KONCENTRATORATLENU W POBLIŻU INNEGO SPRZĘTU ANI NA NIM.
JEŚLI NIE MA INNEJ MOŻLIWOŚCI, MONITOROWAĆ
PRAWIDŁOWE DZIAŁANIE URZĄDZENIA.

OSTRZEŻENIE: PRZED ROZPOCZĘCIEM OBSŁUGI
URZĄDZENIA BEZWZGLĘDNIE PRZECZYTAĆI ZROZUMIEĆ INSTRUKCJĘ OBSŁUGI. JEŚLI
UŻYTKOWNIK NIE JEST W STANIE ZROZUMIEĆ
OSTRZEŻEŃ I INSTRUKCJI, PRZED ROZPOCZĘCIEM
UŻYTKOWANIA SPRZĘTU NALEŻY SKONTAKTOWAĆ
SIĘ Z DOSTAWCĄ. W PRZECIWNYM RAZIE MOŻEDOJŚĆ DO OBRAŻEŃ CIAŁA LUB USZKODZENIAURZĄDZENIA.

OSTRZEŻENIE: KORZYSTANIE Z PRZEWODÓWI ADAPTERÓW INNYCH NIŻ WSKAZANE,Z WYJĄTKIEM PRZEWODÓW I ADAPTERÓWSPRZEDAWANYCH PRZEZ PRODUCENTAMEDYCZNEGO URZĄDZENIA ELEKTRYCZNEGO,
JAKO CZĘŚCI ZAMIENNYCH DLA KOMPONENTÓW
WEWNĘTRZNYCH MOŻE SPOWODOWAĆ
ZWIĘKSZENIE EMISJI LUB OBNIŻENIE ODPORNOŚCI
KONCENTRATORA TLENU NA ZAKŁÓCENIA.

OSTRZEŻENIE: W RAZIE ODCZUWANIA
DYSKOMFORTU LUB WYSTĄPIENIA ZAGROŻENIAZDROWIA NATYCHMIAST SKORZYSTAĆ Z POMOCYLEKARSKIEJ.

OSTRZEŻENIE: PODŁĄCZAĆ URZĄDZENIE TYLKO DO
ZASILANIA O NAPIĘCIU WSKAZANYM NA ETYKIECIEZ DANYMI TECHNICZNYMI ZNAJDUJĄCEJ SIĘ NA
KONCENTRATORZE.

OSTRZEŻENIE: JEŚLI KONCENTRATORTLENU BĘDZIE UŻYWANY W WARUNKACHWYKRACZAJĄCYCH POZA ZAKRES TEMPERATURROBOCZYCH, GENEROWANE STĘŻENIE TLENU
MOŻE BYĆ NIŻSZE NIŻ W DANYCH TECHNICZNYCH.
INFORMACJE NA TEMAT LIMITÓW TEMPERATUR
DOSTĘPNE SĄ W TEJ INSTRUKCJI, W CZĘŚCIZAWIERAJĄCEJ DANE TECHNICZNE.

OSTRZEŻENIE: NIE PODŁĄCZAĆ URZĄDZENIA ZAPOMOCĄ PRZEDŁUŻACZY. NIE PODŁĄCZAĆ ZBYTWIELU URZĄDZEŃ DO TEGO SAMEGO GNIAZDAZASILANIA. ZASTOSOWANIE PRZEDŁUŻACZY
MOŻE MIEĆ NEGATYWNY WPŁYW NA DZIAŁANIE
KONCENTRATORA. PODŁĄCZENIE ZBYT WIELUURZĄDZEŃ DO JEDNEGO GNIAZDA MOŻESPOWODOWAĆ PRZECIĄŻENIE SKRZYNKI
ROZDZIELCZEJ I W EFEKCIE AKTYWACJĘ
WYŁĄCZNIKA/BEZPIECZNIKA LUB POŻAR, JEŚLI
WYŁĄCZNIK/BEZPIECZNIK NIE ZADZIAŁA.

OSTRZEŻENIE: DO KONSERWACJI AKCESORIÓWZAWIERAJĄCYCH TLEN ANI DO KONCENTRATORATLENU NIE UŻYWAĆ OLEJU, SMARU, PRODUKTÓW
OPARTYCH NA ROPIE NAFTOWEJ ANI INNYCH
ŚRODKÓW ŁATWOPALNYCH. WYKORZYSTYWAĆ
TYLKO BALSAMY I MAŚCI NA BAZIE WODY, KTÓRE
MOŻNA STOSOWAĆ W OBECNOŚCI TLENU.
TLEN PRZYŚPIESZA SPALANIE ŁATWOPALNYCHSUBSTANCJI.

PN MN232-C4 K | Instrukcja obsługi

POL – 441

FreeStyle Comfort
PRZESTROGA: W przypadku alarmu
lub zaobserwowania nieprawidłowego
działania koncentratora tlenu zapoznać
się z częścią tej instrukcji poświęconą
rozwiązywaniu problemów. Jeśli nie
uda się rozwiązać problemu, należy
skontaktować się z dostawcą sprzętu.
PRZESTROGA: Przestrzegać instrukcji producenta.
W przeciwnym razie gwarancja będzie nieważna.
OSTRZEŻENIE: NIE PODEJMOWAĆ
PRÓB SERWISOWANIA URZĄDZENIA,Z WYJĄTKIEM POTENCJALNYCHROZWIĄZAŃ PROBLEMÓWWYMIENIONYCH W TEJ INSTRUKCJI. NIE
ZDEJMOWAĆ POKRYW. SERWISOWAĆ
URZĄDZENIE I ZDEJMOWAĆ POKRYWYMOŻE WYŁĄCZNIE DOSTAWCA SPRZĘTU
LUB WYKWALIFIKOWANY TECHNIK.
OSTRZEŻENIE: UŻYWAĆ TYLKO AKCESORIÓWZALECANYCH PRZEZ PRODUCENTA. STOSOWANIEINNYCH MOŻE BYĆ NIEBEZPIECZNE, SPOWODOWAĆ
POWAŻNE USZKODZENIE KONCENTRATORA TLENUI DOPROWADZIĆ DO UNIEWAŻNIENIA GWARANCJI.
OSTRZEŻENIE: PRODUCENT ZALECA, ABY W RAZIEPRZERWY W ZASILANIU, SYTUACJI ALARMOWEJ
LUB AWARII MECHANICZNEJ DOSTĘPNE BYŁOALTERNATYWNE ŹRÓDŁO TLENU. ABY WYBRAĆRODZAJ SYSTEMU REZERWOWEGO, NALEŻYSKONSULTOWAĆ SIĘ Z DOSTAWCĄ SPRZĘTU.
OSTRZEŻENIE: UWAŻAĆ, ABY NIE ZABLOKOWAĆ
WLOTÓW ANI WYLOTU POWIETRZA. NIE
WRZUCAĆ ANI NIE WKŁADAĆ PRZEDMIOTÓW
DO OTWORÓW W OBUDOWIE URZĄDZENIA.
MOŻE TO SPOWODOWAĆ PRZEGRZEWANIE SIĘ
KONCENTRATORA TLENU I MIEĆ NEGATYWNYWPŁYW NA JEGO PRACĘ.
OSTRZEŻENIE: PALENIE PODCZAS UŻYWANIA
TLENU JEST GŁÓWNĄ PRZYCZYNĄ OBRAŻEŃ
I ŚMIERCI SPOWODOWANYCH PRZEZ OGIEŃ.
NALEŻY BEZWZGLĘDNIE PRZESTRZEGAĆ
PONIŻSZYCH OSTRZEŻEŃ BEZPIECZEŃSTWA:
OSTRZEŻENIE: W POMIESZCZENIACH, W KTÓRYCH
JEST TO URZĄDZENIE LUB ZNAJDUJĄ SIĘ
AKCESORIA ZAWIERAJĄCE TLEN, NIE WOLNOPALIĆ, UŻYWAĆ ŚWIECZEK ANI OTWARTEGO OGNIA.
OSTRZEŻENIE: PALENIE PODCZAS STOSOWANIA
WĄSÓW TLENOWYCH MOŻE DOPROWADZIĆ DO
OPARZEŃ TWARZY, A NAWET ŚMIERCI.
OSTRZEŻENIE: PALĄCY UŻYTKOWNICY POWINNIPOSTĘPOWAĆ ZGODNIE Z NASTĘPUJĄCĄPROCEDURĄ, KTÓRA MOŻE IM URATOWAĆ ŻYCIE:WYŁĄCZYĆ KONCENTRATOR TLENU, WYJĄĆ WĄSYI OPUŚCIĆ POMIESZCZENIE, W KTÓRYM ZNAJDUJESIĘ URZĄDZENIE.

442 – POLPN MN232-C4 K | Instrukcja obsługi

OSTRZEŻENIE: W DOMU LUB W MIEJSCUPODAWANIA TLENU W WIDOCZNYCH MIEJSCACH
UMIEŚCIĆ OSTRZEŻENIA „ZAKAZ PALENIA— PODAWANIE TLENU”. POINFORMOWAĆ
UŻYTKOWNIKÓW I ICH OPIEKUNÓWO ZAGROŻENIACH ZWIĄZANYCH Z PALENIEMW OBECNOŚCI UŻYTKOWNIKA LUB PODCZASPODAWANIA TLENU MEDYCZNEGO.
OSTRZEŻENIE: NIE UŻYWAĆ KONCENTRATORATLENU W OBECNOŚCI ŁATWOPALNYCH GAZÓW.
MOŻE TO SPOWODOWAĆ NAGŁY ZAPŁON,KTÓREGO EFEKTEM BĘDZIE USZKODZENIE MIENIA,
OBRAŻENIE CIAŁA LUB ŚMIERĆ.
OSTRZEŻENIE: JEŚLI WĄSY ZOSTANĄ WYJĘTEZ NOSA I UMIESZCZONE NA UBRANIU, POŚCIELI,SOFIE LUB INNYM MIĘKKIM MATERIALE, A W
POBLIŻU BĘDZIE ZNAJDOWAĆ SIĘ PAPIEROS,
ŹRÓDŁO CIEPŁA LUB PŁOMIEŃ, MOŻE DOJŚĆ DO
WYBUCHU POŻARU.
OSTRZEŻENIE: NIE ZOSTAWIAĆ WĄSÓW
TLENOWYCH NA UBRANIU, POŚCIELI CZYPODUSZKACH ANI POD NIMI. GDY WŁĄCZONEURZĄDZENIE NIE JEST UŻYWANE, OBECNOŚĆ
TLENU ZWIĘKSZA RYZYKO ZAPALENIA SIĘ
MATERIAŁU. WYŁĄCZAĆ ZASILANIE ZAWSZE, GDY
KONCENTRATOR TLENU NIE JEST UŻYWANY.
OSTRZEŻENIE: PRZED ODŁĄCZENIEM
AKUMULATORA OD KONCENTRATORA TLENU
URZĄDZENIE NALEŻY BEZWZGLĘDNIE WYŁĄCZYĆ
I ODŁĄCZYĆ OD ZEWNĘTRZNEGO ŹRÓDŁAZASILANIA.
OSTRZEŻENIE: MAŁE CZĘŚCI
MOGĄ STWARZAĆ ZAGROŻENIE
ZADŁAWIENIEM.
OSTRZEŻENIE: NIE DOTYKAĆ WYLOTU POWIETRZAPRZEZ DŁUŻEJ NIŻ 10 SEKUND, PONIEWAŻ MOŻEMIEĆ WYSOKĄ TEMPERATURĘ.
OSTRZEŻENIE: WIATR ORAZ SILNE PRZECIĄGI
MOGĄ MIEĆ NEGATYWNY WPŁYW NA PRECYZJĘ
DZIAŁANIA URZĄDZENIA PODCZAS TLENOTERAPII.
OSTRZEŻENIE: STOSOWANIE TEGO PRODUKTUMOŻE NARAŻAĆ NA KONTAKT Z NIKLEM, KTÓRYW STANIE KALIFORNIA UZNAWANY JEST ZA
SUBSTANCJĘ RAKOTWÓRCZĄ. DODATKOWE
INFORMACJE DOSTĘPNE SĄ NA STRONIE WWW.
P65WARNINGS.CA.GOV.

FreeStyle Comfort
PRZESTROGA: Zawsze odłączać zasilacz
AC najpierw od gniazda sieciowego,
a dopiero później od koncentratora tlenu.
PRZESTROGA: Ustawić koncentrator tlenu w taki
sposób, aby był swobodny dostęp do przewodu
zasilania.
PRZESTROGA: Zadbać o takie rozmieszczenie
przewodów doprowadzania tlenu i zasilania, aby nie
doszło do przypadkowego potknięcia ani uduszenia.
PRZESTROGA: Jeśli koncentrator tlenu używany
jest w samochodzie, na łodzi lub podłączony dodowolnego źródła zasilania prądem stałym, uruchomić
pojazd przed podłączeniem urządzenia. W przeciwnym
razie zasilacz może nie doprowadzać prądu dokoncentratora tlenu.
PRZESTROGA: Po wyłączeniu silnika samochodu,
w którym używany jest koncentrator tlenu, odłączyć
i wyjąć z niego urządzenie. Nie pozostawiać
koncentratora tlenu w bardzo rozgrzanym lubwyziębionym samochodzie ani w innych miejscacho wysokiej lub niskiej temperaturze.
Uwaga: Przenośny sprzęt komunikacyjny
wykorzystujący częstotliwości radiowe możeoddziaływać na elektryczny sprzęt medyczny.
PRZESTROGA: NIE pozostawiać
koncentratora tlenu ani zasilaczapodłączonego, gdy silnik pojazdu jestWYŁĄCZONY. Może to spowodować
wyczerpanie akumulatora pojazdu.
PRZESTROGA: Korzystanie z koncentratora tlenu lub
przechowywanie go w warunkach wykraczających
poza normalny zakres temperatur i ciśnień roboczych(wysokość n.p.m.) może mieć wpływ na jego działanie,
skrócić czas pracy na akumulatorze, a także wydłużyć
czas ładowania akumulatora. Informacje na temat
limitów temperatur roboczych i przechowywaniadostępne są w tej instrukcji, w części zawierającej
dane techniczne.
PRZESTROGA: Koncentratory tlenu wyposażone
w akumulatory należy przechowywać w chłodnym
i suchym miejscu. Pozwoli to wydłużyć
czas eksploatacji akumulatora. Długotrwałe
przechowywanie koncentratora tlenu w wysokiej
temperaturze lub z całkowicie naładowanym bądźrozładowanym akumulatorem może doprowadzić do
skrócenia okresu eksploatacji ogniw. Nie podejmować
prób otwarcia akumulatora. Nie ma w nim części
serwisowanych. Trzymać akumulatory z dala od dzieci.
PRZESTROGA: UŻYWAĆ WYŁĄCZNIE akumulatorów
dostarczonych przez producenta. Aby dowiedzieć
się, jak prawidłowo zutylizować akumulator, należy
skontaktować się z dostawcą sprzętu lub odpowiednim
organem władz lokalnych.

Uwaga: Zadbać, aby wąsy były prawidłowo zamontowane. Dzięki temu koncentrator tlenu będzie mógł
prawidłowo wykrywać wdech i podawać tlen. Podczas
wdychania powinien być słyszalny przepływ tlenu dokońcówki nosowej.
Uwaga: Akumulatory litowe mogą trwale utracić pojemność, jeśli przy pełnym naładowaniu lub rozładowaniubędą narażone na ekstremalnie wysokie temperatury.
Jeśli akumulator ma być długo przechowywany, należy
naładować go w 25–50% i zapewnić warunki temperatur w przedziale od -4°F do 68°F (od -20°C do 20°C).
Uwaga: Korzystanie z niektórych akcesoriów do podawania tlenu, których nie wyszczególniono wśród komponentów do stosowania z tym urządzeniem, możemieć negatywny wpływ na jego działanie. Zalecane
akcesoria wymienione są w niniejszej instrukcji.
Uwaga: Nie używać koncentratora tlenu bezzałożonego filtra wlotu powietrza. Jeśli dostarczono
filtr zastępczy, założyć go przed przystąpieniem do
czyszczenia filtra głównego. Wymyć zanieczyszczony
filtr w ciepłym roztworze mydła w wodzie, a następnie
przed zamontowaniem wysuszyć.
Uwaga: Producent nie zaleca sterylizowania tegosprzętu.
Uwaga: Podczas używania koncentratora tlenu nie
istnieje ryzyko wyczerpania zapasu tlenu w pomieszczeniu.
OSTRZEŻENIE: NIE NALEŻY PODŁĄCZAĆ
KONCENTRATORA TLENU DO
ZEWNĘTRZNEGO ŹRÓDŁA ZASILANIA(AC/DC) BEZ ZAMONTOWANEGOAKUMULATORA. NIE WŁĄCZAĆ
ANI NIE PODŁĄCZAĆ URZĄDZENIA,
GDY AKUMULATOR JEST ODŁĄCZONY.
USUWAĆ AKUMULATOR TYLKO WTEDY, GDYURZĄDZENIE JEST WYŁĄCZONE I ODŁĄCZONEOD ŹRÓDŁA ZASILANIA. NIE ODŁĄCZAĆAKUMULATORA, GDY KONCENTRATOR JEST
WŁĄCZONY LUB PODŁĄCZONY DO ŹRÓDŁAZASILANIA.

Uwaga: Jeśli koncentrator tlenu był przez dłuższy
okres przechowywany w warunkach wykraczających
poza normalne przedziały robocze zaprezentowane
w tabeli powyżej, przed włączeniem go należy poczekać, aż uzyska standardową temperaturę pracy.

PN MN232-C4 K | Instrukcja obsługi

POL – 443

FreeStyle Comfort

Dane techniczne

Wymiary (z akumulatorem
standardowym)

FreeStyle Comfort
10,0 x 7,3 x 3,1 cala (25,4 x 18,5 x 7,9 cm)

Wymiary (z akumulatorempowiększonym)

11,0 x 7,3 x 3,1 cala (27,9 x 18,5 x 7,9 cm)

Masa (z akumulatorem
standardowym)

5 funtów (2,3 kg)

Masa (z akumulatorempowiększonym)

6 funtów (2,7 kg)

Nominalny poziom hałasu***

39,93 dB (A) ± 0,42 dB (A) @ 2

Ustawienia przepływu

Ustawienie impulsowe 1, 2, 3, 4, 5

Ustawienie podawania O2(przy 20 oddechach/min)Stężenie tlenu*
Czujnik tlenuCiśnienie maksymalne na wylocieDomyślna czułość dozowania

Ustawienie 1 — 210 ml/min
Ustawienie 2 — 420 ml/min
Ustawienie 3 — 630 ml/min
Ustawienie 4 — 840 ml/min
Ustawienie 5 — 1050 ml/min
90% (+5,5/-3%)
Tak<30 psi>-0,1 cm H2O (regulowane)

Zasilanie prądem przemiennym (AC) 100–240 V AC, 50–60 Hz, 120W Max
Zasilanie prądem stałym (DC)Typ baterii
Pojemność baterii

11–18 V DC (maks. 10 A)
Litowo-jonowej
Pojedyncze ogniwo: 6700 mAh; Podwójna komórka: 13400 mAh

Czas ładowania akumulatora

Akumulator standardowy: 3,5 godz.
Akumulator powiększony: 6 godz.

Czasy pracy na akumulatorze

Czas pracy na akumulatorze standardowym: Ustawienie 2 — do 4 godzin
Czas pracy na akumulatorze powiększonym: Ustawienie 2 — do 8 godzin

Dopuszczalna temperatura podczas Od 41°F do 104°F (od 5°C do 40°C)
pracy**
Temp. podczas transportu
i przechowywania (urządzenie)

Od -13°F do +158°F (od -25°C do +70°C) na wysokości maks. 10 000 stóp
(3048 m)Temp. podczas transportu
i przechowywania (akumulator)

Od -13°F do +158°F (od -25°C do +70°C) na wysokości maks. 10 000 stóp
(3048 m)Dopuszczalna wysokość pracy**-1250–10 000 stóp (-381–3048 m) (testowano przy 700–1060 hPa)

Dopuszczalna wilgotność podczas
pracy

Wilgotność względna 15–95% (bez kondensacji)

Wilgotność przechowywania
Ograniczona gwarancja

0 - 90% bez kondensacji
3 lata na urządzenie
1 rok na akumulator i zasilacze* Przy ciśnieniu atmosferycznym wynoszącym 14,7 psi (101 kPa) w temperaturze 70°F (21°C).
** Urządzenie będzie działać we wszystkich środowiskach, o ile otoczenie będzie miało określone zakresy temperatur iwilgotności. Jeśli urządzenie będzie używane w warunkach wykraczających poza te wartości robocze, stężenie tlenu generowane
przy wyższych ustawieniach przepływu może być niższe niż w danych technicznych. Ekran przygotowania może być wyświetlany
do czasu uzyskania przez koncentrator wymaganego stężenia (maksymalnie 15 minut). W tym czasie można normalnie użytkowaćurządzenie.
*** Poziom hałasu mierzony według metody badawczej Nr. 14-1 10/2018 MDS-Hi.
Przewidywany okres eksploatacji urządzenia to co najmniej pięć lat. Przewidywany okres przydatności wkładów sitowych wynosi
około roku, jeśli koncentrator nie jest używany przez cały ten czas.
444 – POLPN MN232-C4 K | Instrukcja obsługi

FreeStyle Comfort
Instrukcja obsługi
1. Ustaw urządzenie FreeStyle Comfort w miejscu
odpowiednim do użytkowania.
a. Należy zapewnić odpowiednią wentylację
i zadbać, aby filtry wlotów i wylot powietrza nie
były zablokowane.
b. Koncentrator należy ustawić tak, aby wszystkiewskaźniki oraz alarmy dźwiękowe i graficzne
były dobrze widoczne i słyszalne.
c. Przed uruchomieniem urządzenia upewnij się,że filtry wlotów powietrza są zamontowane.
Jeśli brakuje filtra, skontaktuj się z dostawcą
sprzętu.
2. Podłącz akumulator do koncentratora.
a. Ustaw akumulator centralnie względem dolnejczęści obudowy urządzenia FreeStyle Comfort.

3. Podłącz zasilacz AC lub DC do gniazda
znajdującego się w prawej dolnej części z przoduurządzenia.

4. Podłącz wąsy tlenowe do króćca wylotowego tlenu.
Przed rozpoczęciem użytkowania upewnij się, żepołączenie jest prawidłowe.

UWAGA: Stosowanie wąsów innych niż zalecanemoże powodować pogorszenie wydajności.

b. Wsuń akumulator, aż usłyszysz kliknięcie
potwierdzające, że zatrzasnął się na miejscu.

Uwaga: do dostawcy sprzętu: Do FreeStyle Comfort
zalecane są następujące akcesoria do podawania tlenu:• Kaniula nosowa z rurką o długości 7 stóp (2,1 m) (maks.
LPM: 6 LPM): Numer katalogowy CAIRE 5408-SEQ• Dodatkowe informacje na temat zalecanych akcesoriów
są dostępne online na stronie www.caireinc.com.

PN MN232-C4 K | Instrukcja obsługi

POL – 445

FreeStyle Comfort
5. Włącz koncentrator FreeStyle Comfort, naciskając
przycisk zasilania, i wybierz przepisane ustawienie
przepływu za pomocą przycisków regulacji (+) i (-).

Ustawienie objętości impulsu
autoSAT (ml)*Częstotliwość
oddechów

1

2

3

4

5

15

10,5

21,0

31,5

42,0

52,5

20

10,5

21,0

31,5

42,0

52,5

25

10,5

21,0

31,5

42,0

42,0

30

10,5

21,0

31,5

35,0

35,0

35

10,5

21,0

30,0

30,0

30,0

40

10,5

21,0

26,3

26,3

26,3

* Tolerancja wynosi +/-15%

Zostanie wyemitowany krótki sygnał dźwiękowy
wskazujący, że urządzenie zostało uruchomione.

Uwaga: Przy pierwszym włączeniu koncentratora
FreeStyle Comfort domyślne ustawienie przepływu to
2. Jeśli zostanie zmienione, urządzenie zapamięta je
przy następnym uruchomieniu.
6. Załóż wąsy tlenowe na nos zgodnie z dostarczoną
wraz z nimi instrukcją producenta i oddychaj
normalnie. Przy każdym wdechu powinien być
wyczuwalny strumień tlenu dostarczanego przez wąsy.
7. Aby zmienić ustawienie przepływu, naciskaj
przyciski regulacji (+) i (-), aż uzyskasz wymaganenatężenie. Różnica w dźwięku podczas zmieniania
ustawień jest normalna.

Ustawienie objętości impulsu bez
autoSAT (ml)*Częstotliwość
oddechów

1

2

3

4

5

15

14,0

28,0

42,0

56,0

70,0

20

10,5

21,0

31,5

42,0

52,5

25

8,4

16,8

25,2

33,6

42,0

30

7,0

14,0

21,0

28,0

35,0

35

6,0

12,0

18,0

24,0

30,0

40

5,3

10,5

15,8

21,0

26,3

8. Wyłącz koncentrator FreeStyle Comfort, naciskając
przycisk zasilania.

Uwaga: Wyświetlacz przełączy się w tryb gotowości
i zostanie przygaszony po 20 sekundach. Naciśnij
dowolny przycisk, aby wybudzić wyświetlacz z trybu
gotowości.
9. Funkcje zaawansowane: patrz Instrukcje poruszania
się po menu.

SPECJALNE FUNKCJE PRODUKTU
autoSAT®
Zastrzeżona przez CAIRE funkcja autoSAT umożliwia
FreeStyle Comfort dostarczanie stałej objętości
tlenu (bolusa) na oddech, do maksymalnej częstości
oddechów. Gdy częstość oddechów użytkownika
wzrasta podczas czynności takich jak chodzenie,
funkcja autoSAT dostosowuje działanie urządzenia,
aby utrzymać ustawioną objętość bolusa.
UltraSense®
Ten czuły regulator dawkowania wykrywa zmianę
ciśnienia podczas oddychania i uwalnia impuls tlenu
przy każdym wdechu.

446 – POLPN MN232-C4 K | Instrukcja obsługi

autoDOSE™
Jeśli wystąpi alarm „brak oddechu”, funkcja tabędzie impulsowo podawała tlen z częstotliwością
20 oddechów na minutę, przy bieżącym ustawieniu
przepływu, aż do zlikwidowania alarmu.

FreeStyle Comfort
Zasilanie
Koncentrator FreeStyle Comfort może być zasilany na
trzy sposoby:• Akumulator• Zasilanie prądem przemiennym (AC)• Zasilanie prądem stałym (DC)Informacje na temat działania każdego z tych
komponentów dostępne są poniżej.
OSTRZEŻENIE: KORZYSTAĆ WYŁĄCZNIEZ ZASILACZY WSKAZANYCH W TEJINSTRUKCJI OBSŁUGI. STOSOWANIEZASILACZY I AKUMULATORÓW FIRM
INNYCH NIŻ CAIRE MOŻE SPOWODOWAĆUSZKODZENIE I NIEPRAWIDŁOWE
DZIAŁANIE URZĄDZENIA.

ODŁĄCZANIE AKUMULATORA

Naciśnij przycisk zwalniania akumulatora w dół,w kierunku przeciwnym do urządzenia FreeStyle
Comfort. TRZYMAJĄC ten przycisk, wysuń
akumulator z koncentratora.
OSTRZEŻENIE: PRZED ODŁĄCZENIEM
AKUMULATORA KONCENTRATOR TLENU
NALEŻY BEZWZGLĘDNIE WYŁĄCZYĆ
I ODŁĄCZYĆ OD ZEWNĘTRZNEGOŹRÓDŁA ZASILANIA.

AKUMULATOR

Urządzenie FreeStyle Comfort może być zasilane
przez dostarczany wraz z nim akumulator.
Funkcjonalność koncentratora nie jest w żaden
sposób ograniczona podczas pracy na akumulatorze.
Przy zasilaniu z akumulatora dostępne są wszystkie
ustawienia natężenia przepływu.

SPRAWDZANIE STANU NAŁADOWANIAAKUMULATORA
PODŁĄCZANIE AKUMULATORA

Ustaw akumulator centralnie względem dolnej części
obudowy urządzenia FreeStyle Comfort. Wsuńakumulator, aż usłyszysz kliknięcie potwierdzające, żezatrzasnął się na miejscu.

Stan naładowania akumulatora, gdy jest on podłączony
do urządzenia FreeStyle Comfort, można sprawdzić,
spoglądając na ikonę baterii i szacowny pozostały czas
pracy prezentowany na wyświetlaczu.
80–100%60–80%

40–60%

20–40%

20%

PN MN232-C4 K | Instrukcja obsługi

POL – 447

FreeStyle Comfort
Uwaga: Jeśli trwa ładowanie akumulatora, zamiast
procentowego poziomu naładowania wyświetlanybędzie wskaźnik ładowania.
Aby sprawdzić poziom naładowania akumulatora,
gdy nie jest podłączony do koncentratora FreeStyle
Comfort, połóż go na płaskiej powierzchni i naciśnij
przycisk TEST. Po jego prawej stronie zaświecą się
kontrolki akumulatora (25–100%), wskazując poziom
naładowania ogniw.

PIERWSZE ŁADOWANIE

Nowy akumulator dostarczany wraz z urządzeniem
FreeStyle Comfort jest tylko częściowo naładowany
w chwili opuszczania fabryki. Przed pierwszym
użyciem należy go całkowicie naładować za pomocą
zasilacza AC lub DC.

INFORMACJE OGÓLNE NA TEMATŁADOWANIA

Akumulator jest ładowany zawsze, gdy jest
zainstalowany w koncentratorze FreeStyle Comfortpodłączonym do zewnętrznego źródła zasilania AC lub
DC (z wyjątkiem samolotu).
Akumulator jest w pełni naładowany, gdy wskaźnik
pokazuje 100%.
Można go też ładować za pomocą opcjonalnej

448 – POLPN MN232-C4 K | Instrukcja obsługiładowarki biurkowej.

CZAR PRACY NA AKUMULATORZE

Gdy zasilacz AC lub DC zostanie odłączony,koncentrator automatycznie przełączy się na zasilaniez akumulatora, jeśli jest on zainstalowany. W tabeliponiżej zaprezentowano typowe czasy pracy dla
nowego akumulatora. Częstotliwość oddechu nie mawpływu na czas pracy na akumulatorze.
Ustawienie impulsowe

Akumulator
standardowy (8ogniw)

Akumulatorpowiększony (16ogniw)

1

do 8 godz.

do 16 godz.

2

do 4 godz.

do 8 godz.

3

do 3 godz.

do 6 godz.

4

do 2,25 godz.

do 4,5 godz.

5

do 2 godz.

do 4 godz.

TYPOWY CZAS ŁADOWANIAAKUMULATORA

Zazwyczaj całkowite naładowanie w pełni
rozładowanego akumulatora zajmuje 3,5 godziny
dla modelu standardowego i 6 godzin w przypadku
powiększonego.

Uwaga: Jeśli akumulator został całkowicie rozładowany, jego ładowanie może nie rozpocząć się od razu.

OKRES EKSPLOATACJI AKUMULATORA

Akumulatory FreeStyle Comfort wytrzymują 300 cykliładowania/rozładowania.

FreeStyle Comfort
Zasilacz AC i DCUrządzenie FreeStyle Comfort dostarczane jest
standardowo z zasilaczami AC i DC.
Na zasilacz AC składają się:a. Zasilacz właściwy w obudowie, pracujący
w zakresie napięć 100–240 V i częstotliwości
50–60 Hz.
b. Przewód zasilania AC, który podłącza się dozasilacza i gniazda prądu przemiennego.

Zasilacz DC składa się z przewodu, który podłącza się
bezpośrednio do urządzenia i gniazda prądu stałego.

ZASILANIE PRĄDEM PRZEMIENNYM(AC) Z zasilania AC można korzystać w domu,

2. Podłącz przewód zasilania AC do zasilacza prądu
przemiennego.

gdy dostępne jest standardowe gniazdo siecioweprądu przemiennego. Koncentrator FreeStyle
Comfort zasilany prądem przemiennym jest w pełni
funkcjonalny. Urządzenie FreeStyle Comfort ładuje
akumulator zawsze, gdy jest podłączone do źródła
zasilania AC (z wyjątkiem samolotu). Ładowanie
odbywa się bez względu na to, czy koncentrator
FreeStyle Comfort jest włączony czy wyłączony.
OSTRZEŻENIE: NIE PODŁĄCZAĆ DOZEWNĘTRZNEGO ŹRÓDŁA ZASILANIA,GDY AKUMULATOR NIE JESTZAINSTALOWANY W KONCENTRATORZETLENU.

3. Podłącz drugi koniec przewodu zasilania AC dogniazda ściennego lub innego.

Aby podłączyć urządzenie FreeStyle Comfort doźródła zasilania AC:
1. Podłącz zasilacz AC do złącza zasilania w prawej
dolnej części, z przodu koncentratora FreeStyle
Comfort. Upewnij się, że wtyczka jest prawidłowowciśnięta i podłączona.

PN MN232-C4 K | Instrukcja obsługi

POL – 449

FreeStyle Comfort
4. Włącz koncentrator FreeStyle Comfort, naciskając
przycisk zasilania, i wybierz przepisane ustawienie
przepływu za pomocą przycisków regulacji (+) i (-).

Uwaga: Jeśli konieczne jest naładowanie akumulatora,
na ikonie baterii na wyświetlaczu widoczny będzie
jeden pomarańczowy pasek.

ZASILANIE PRĄDEM STAŁYM (DC)

Zasilacz DC służy do podłączania do gniazd
w samochodach, łodziach i innych pojazdach.
Koncentrator FreeStyle Comfort zasilany prądem
stałym jest w pełni funkcjonalny. W razie potrzebyurządzenie FreeStyle Comfort ładuje akumulator, gdy
jest podłączone do źródła zasilania DC. Ładowanie
odbywa się bez względu na to, czy urządzenie jestwłączone czy wyłączone.
OSTRZEŻENIE: NIE PODŁĄCZAĆ DOZEWNĘTRZNEGO ŹRÓDŁA ZASILANIA,GDY AKUMULATOR NIE JESTZAINSTALOWANY W KONCENTRATORZETLENU.
OSTRZEŻENIE: NIE UŻYWAĆ URZĄDZENIAPODŁĄCZONEGO DO GNIAZDA SAMOCHODOWEGO
12 V DC PRZY WYŁĄCZONYM SILNIKU.
1. Uruchom pojazd.
2. Podłącz zasilacz DC do gniazda zasilania w prawej
dolnej części, z przodu koncentratora FreeStyle
Comfort. Upewnij się, że wtyczka jest prawidłowowciśnięta i podłączona.

450 – POLPN MN232-C4 K | Instrukcja obsługi

3. Podłącz drugi koniec przewodu zasilania DC dogniazda zasilania w samochodzie.

4. Włącz koncentrator FreeStyle Comfort, naciskając
przycisk zasilania, i wybierz przepisane ustawienie
przepływu za pomocą przycisków regulacji (+) i (-).

Uwaga: Jeśli konieczne jest naładowanie akumulatora, na ikonie baterii na wyświetlaczu widoczny będzie
jeden pomarańczowy pasek.

FreeStyle Comfort
Czyszczenie, pielęgnacja i rutynowe
czynności konserwacyjne
RUTYNOWE CZYNNOŚCIKONSERWACYJNE

Konserwacja zapobiegawczaUrządzenie nie wymaga konserwacji. Filtry wewnątrz
urządzenia mają trwać przez cały okres użytkowaniaurządzenia. Jeśli urządzenie nie działa prawidłowo,urządzenie należy zwrócić dostawcy w celu naprawy.
Serwisowanie podzespołów znajdujących się wewnątrz
obudowy koncentratora FreeStyle Comfort możebyć wykonywane tylko przez dostawcę sprzętu do
podawania tlenu.
Osoba obsługująca urządzenie odpowiedzialna jest zanastępujące czynności konserwacyjne.
• Raz na tydzień — wyczyścić filtry wlotów powietrza.
OSTRZEŻENIE: NIE OTWIERAĆ ANI NIE
ZDEJMOWAĆ PLASTIKOWYCH POKRYW
KONCENTRATORA TLENU. MOŻE TOROBIĆ JEDYNIE WYKWALIFIKOWANYSERWISANT.

FILTRY WLOTÓW POWIETRZA

Powietrze doprowadzane jest do koncentratora
FreeStyle Comfort przez filtry wlotów powietrzaznajdujące się z przodu urządzenia na dole. Filtry
te zapobiegają dostawaniu się do urządzenia kurzu
i innych dużych cząstek znajdujących się w powietrzu.
Należy je czyścić co najmniej raz w tygodniu.
Dostawca sprzętu do podawania tlenu może zalecaćczęstsze czyszczenie w zależności od warunków
eksploatacji. Aby odpowiednio wyczyścić filtry na
wlotach powietrza, należy wykonać następujące
czynności:1. Zdejmij dwa filtry z przodu urządzenia FreeStyle
Comfort, naciskając odpowiednie uszko w góręi pociągając na zewnątrz.

OSTRZEŻENIE: NIE UŻYWAĆ
KONCENTRATORA TLENU BEZ
ZAŁOŻONEGO FILTRA WLOTU
POWIETRZA.

KONSERWACJA AKUMULATORA

Nie narażaj akumulatora na działanie płynów.
Nie rozbieraj go ani nie otwieraj.
Nie narażaj na działanie otwartego ognia.
Nie upuszczaj.
Trzymaj z dala od dzieci.
Jeśli na akumulatorze widoczne są uszkodzenia,
przed użyciem skontaktuj się z dostawcą sprzętu do
podawania tlenu.
Nie umieszczaj akumulatora wewnątrz lub w pobliżu
kuchenki mikrofalowej ani innych urządzeń
kuchennych.

PRZECHOWYWANIE AKUMULATORA

Akumulatora koncentratora tlenu nie trzeba całkowicie
rozładowywać przed ładowaniem. Zaleca się ładować
go po każdym użyciu urządzenia.
Nie zostawiaj akumulatora w miejscach, w których
narażony byłby na działanie wysokiej temperatury, na
przykład silnie nasłonecznionych lub w samochodzie.
W przeciwnym razie może dojść do pogorszenia jegowydajności i skrócenia czasu eksploatacji. Możeteż dojść do nagrzania akumulatora, wydostawania
się z niego dymu, zapłonu, a nawet wybuchu.
Akumulator należy przechowywać w miejscu
o ustabilizowanych warunkach i niskiej wilgotności
(poniżej 70% wilgotności względnej), wolnym od
gazów korozyjnych. Aby zapobiec rdzewieniu, unikaj
warunków, w których dochodzi do kondensacji, np.
nagłych zmian temperatury otoczenia.
Jeśli akumulator będzie przechowywany przez miesiąc
lub dłużej, zaleca się rozładować go lub naładowaćczęściowo do poziomu od 50% do 75% (od dwóch dotrzech diod LED). Po naładowaniu odłącz akumulator
od koncentratora tlenu.

2. Wymyj oba filtry w ciepłym roztworze mydła
w wodzie.
3. Dokładnie wypłucz filtry i wytrzyj miękką,niepozostawiającą włókien ściereczką. Przed
ponownym założeniem upewnij się, że filtry są
całkowicie suche.
4. Zamontuj czyste i suche filtry.
PN MN232-C4 K | Instrukcja obsługi

POL – 451

FreeStyle Comfort
CZYSZCZENIE

Przed czyszczeniem WYŁĄCZ urządzenie i odłącz
je od zasilania. NIE spryskuj obudowy zewnętrznej.
Stosuj wilgotną (nie mokrą) szmatkę lub gąbkę.
Spryskaj szmatkę lub gąbkę łagodnym roztworem
do środek czyszczący do gospodarstwa domowego,
aby wyczyścić obudowę i zasilacze. Postępuj zgodnie
z instrukcjami producenta środka czyszczącego.
OSTRZEŻENIE: ABY UNIKNĄĆ
PRZYPADKOWEGO PORAŻENIA
PRĄDEM LUB POPARZENIA, PRZEDPRZYSTĄPIENIEM DO CZYSZCZENIAURZĄDZENIA NALEŻY JE WYŁĄCZYĆ
I WYJĄĆ PRZEWÓD ZASILANIAZ GNIAZDA. SERWISOWAĆ
KONCENTRATOR I ZDEJMOWAĆPOKRYWY MOŻE WYŁĄCZNIE DOSTAWCASPRZĘTU LUB WYKWALIFIKOWANYTECHNIK.
OSTRZEŻENIE: UWAŻAĆ, ABY NIE ZAMOCZYĆ
KONCENTRATORA TLENU I ABY DO WEWNĄTRZ
NIE DOSTAŁ SIĘ ŻADEN PŁYN. MOGŁOBY TO
SPOWODOWAĆ USZKODZENIE LUB WYŁĄCZENIEURZĄDZENIA I ZWIĘKSZYĆ RYZYKO PORAŻENIA
PRĄDEM I POPARZEŃ.
OSTRZEŻENIE: NIE STOSOWAĆ PŁYNÓWBEZPOŚREDNIO NA URZĄDZENIU. NIEKTÓRE
ŚRODKI CHEMICZNE, KTÓRYCH NIE NALEŻYSTOSOWAĆ: ALKOHOL I PRODUKTY NA BAZIEALKOHOLU, KONCENTRATY NA BAZIE CHLORU(CHLOREK ETYLENU) ORAZ PRODUKTY NA BAZIEOLEJU. ŚRODKÓW TYCH NIE NALEŻY UŻYWAĆ
DO CZYSZCZENIA PLASTIKOWEJ OBUDOWYURZĄDZENIA, PONIEWAŻ MOGĄ JE USZKODZIĆ.

452 – POLPN MN232-C4 K | Instrukcja obsługi

OSTRZEŻENIE: DO CZYSZCZENIA OBUDOWY,
PANELU STEROWANIA I PRZEWODU ZASILANIA
UŻYWAĆ WYŁĄCZNIE ŁAGODNEGO DOMOWEGOŚRODKA CZYSZCZĄCEGO I WILGOTNEJ SZMATKI(NIE MOKREJ) LUB GĄBKI. PO CZYSZCZENIUWYTRZEĆ DO SUCHA. UWAŻAĆ, ABY ŻADEN PŁYNNIE DOSTAŁ SIĘ DO WNĘTRZA URZĄDZENIA.
SZCZEGÓLNĄ UWAGĘ ZWRÓCIĆ NA ZŁĄCZE
WYLOTOWE TLENU SŁUŻĄCE DO PODŁĄCZANIAWĄSÓW — UPEWNIĆ SIĘ, ŻE NIE MA NA NIM KURZU,WODY ANI INNYCH ZANIECZYSZCZEŃ.
OSTRZEŻENIE: DO CZYSZCZENIA AKUMULATORA
UŻYWAĆ WILGOTNEJ (NIE MOKREJ) SZMATKI
LUB GĄBKI. SPRYSKAĆ SZMATKĘ LUB GĄBKĘ
ŁAGODNYM DOMOWYM ŚRODKIEM CZYSZCZĄCYMI WYTRZEĆ OBUDOWĘ AKUMULATORA.
OSTRZEŻENIE: NIE UŻYWAĆ MYJEKAUTOMATYCZNYCH CZYSZCZENIA.

WYMIANA WĄSÓW

Bezwzględnie przestrzegać instrukcji prawidłowegoużytkowania dostarczonej przez producenta wąsów.
Wymieniaj wąsy tlenowe lub przewody tlenowe co
3–6 miesięcy (jeśli urządzenie używane jest codziennie,
co 3 miesiące) zgodnie z zaleceniami producenta
komponentu lub dostawcy sprzętu do podawania tlenu.
Instrukcje czyszczenia i wymiany powinien dostarczyć
lekarz lub dostawca sprzętu do podawania tlenu.
Części zamienne do urządzenia dostępne są u dostawcysprzętu do podawania tlenu.

FreeStyle Comfort
Stany alarmowe
Nrpriorytetu

Opis alarmu

1

Awariasystemu —wysokatemperatura
akumulatoraWyświetlacz

Sygnał
dźwiękowy

Dodatkowe informacje / sugerowane
działanie

1 sygnałco 20 sUrządzenie wyemituje alarm, gdy
temperatura akumulatora zbliży się dolimitu powodującego wyłączenie awaryjne.
Jeśli limit zostanie osiągnięty, urządzenie
się wyłączy. Jeśli będzie podłączonezewnętrzne źródło zasilania, przez 2 minutybędzie wyświetlany alarm na ekranie oraz
emitowany sygnał dźwiękowy.
Wykonaj test, używając akumulatora
w temperaturze pokojowej. Jeśli problem
nie ustąpi, skontaktuj się z dostawcą.

2

Awariasystemu —wysokatemperatura

1 sygnałco 20 sUrządzenie wyemituje alarm, gdy
temperatura wewnętrzna zbliży się do limitu
powodującego wyłączenie.
Jeśli limit zostanie osiągnięty, urządzenie
się wyłączy. Przez 2 minuty będzie
wyświetlany alarm na ekranie oraz
emitowany sygnał dźwiękowy.
Upewnij się, że nie ma żadnych
niedrożności na wlotach i wylotach
powietrza urządzenia. Jeśli problem nieustąpi, skontaktuj się z dostawcą.

3

Awariasystemu —wysokieciśnienie

1 sygnałco 20 sJeśli limit zostanie osiągnięty, urządzenie
się wyłączy. Przez 2 minuty będzie
wyświetlany alarm na ekranie oraz
emitowany sygnał dźwiękowy.
Jeśli problem nie ustąpi, skontaktuj sięz dostawcą.

4

Nie wykryto
oddechu

1 sygnałco 30 sNie wykryto oddechu przez 15 sekund.
Aktywacja po pierwszym oddechu lubwyłączeniu ekranu przygotowania,
w zależności od tego, co nastąpi szybciej.
Sprawdź wąsy pod kątem zagięć
i przecieków. Zwiększ ustawienie czułości.
Jeśli problem nie ustąpi, skontaktuj sięz dostawcą.

5

6

Wyłączenie
z powodu
niskiegopoziomunaładowania
akumulatora

1 sygnałco 30 sPrzygotowanie

Brak

Alarm emitowany do momentu wyczerpania
akumulatora i wyłączenia urządzenia.
Naładuj akumulator.

Urządzenie uruchomi się i będzie działało
w trybie automatycznego dozowania do
czasu wykrycia wdechu lub zakończenia
przygotowania, w zależności od tego, conastąpi szybciej.

PN MN232-C4 K | Instrukcja obsługi

POL – 453

FreeStyle Comfort
Nrpriorytetu

Opis alarmu

7Wyświetlacz

Sygnał
dźwiękowy

Dodatkowe informacje

Niskie stężenie

1 sygnał co
30 s przez
maks. 5 minJeśli problem nie ustąpi, skontaktuj sięz dostawcą.

8

Awariasystemu —natężenie
przepływu
(awaria zaworu
oszczędzacza
tlenu, blokada
linii gazu)

1 sygnał co
30 sJeśli problem nie ustąpi, skontaktuj sięz dostawcą.

9

Ostrzeżenie
o niskim
poziomieakumulatora

1 sygnał,bez
powtarzania

Naładuj akumulator.

10

Zbyt duża
częstotliwość
oddechu

1 sygnał co
60 sCzęstotliwość oddechu przekracza 40
oddechów/min.

11

Awariasystemu —uszkodzenie
wentylatora

BrakJeśli problem nie ustąpi, skontaktuj sięz dostawcą.

12

Awariasystemu —niskie ciśnienie

BrakJeśli problem nie ustąpi, skontaktuj sięz dostawcą.

13

Awariasystemu —niska
temperatura

Brak

Pozwól, aby urządzenie ogrzało się dotemperatury pokojowej. Jeśli problem nieustąpi, skontaktuj się z dostawcą.

Info

Uruchomienie

1 sygnałinformacyjny

Informacja na wyświetlaczu po
uruchomieniu oraz pojedynczy sygnałdźwiękowy potwierdzający sprawność
brzęczyka.

Jeśli dawka jest nieprawidłowo
wyzwalana, należy zmniejszyć
ustawienie czułości. Jeśli problem nieustąpi, skontaktuj się z dostawcą.

Na ekranie w prawym dolnym roguwyświetlana jest bieżąca wersja
oprogramowania układowego.
454 – POLPN MN232-C4 K | Instrukcja obsługi

FreeStyle Comfort
Nrpriorytetu

Opis alarmuWyświetlacz

Sygnał
dźwiękowy

Info

Normalny ekranroboczy w trybie
autoSAT(zasilanie DC
niepodłączone)

Brak

Info

Normalny ekranroboczy przywyłączonej
funkcji autoSAT(zasilanie DC
niepodłączone)

Brak

Info

Zmianaustawienia

1 sygnałinformacyjny

InfoPodłączyćzewnętrzneźródłozasilania przywyłączonymurządzeniu

Brak

Info

Podłączyćzewnętrzneźródło zasilania
przy włączonymurządzeniu

Brak

Info

Odłączyćzewnętrzneźródłozasilania przywyłączonymurządzeniu

1 sygnałinformacyjny

InfoOdłączyćzewnętrzneźródło zasilania
przy włączonymurządzeniu (zzainstalowanym
akumulatorem)

1 sygnałinformacyjny

Dodatkowe informacjeWyświetla ekran CAIRE przez 2 sekundy,następnie stan baterii i zasilania przez 5s, po czym wyświetlacz się wyłącza.

Wyświetla ekran CAIRE przez 2 sekundy,po czym wyświetlacz się wyłącza.

PN MN232-C4 K | Instrukcja obsługi

POL – 455

FreeStyle Comfort
Nrpriorytetu

Opis alarmu

InfoWyświetlacz

Sygnał
dźwiękowy

Dodatkowe informacje

W trybie
gotowości —gdy naciśnięty
zostanie
przycisk w góręlub w dół,na ekraniewyświetlony
zostanie stan
akumulatorai ładowania

BrakWyświetlacz włączy się na 5 s,a następnie wyłączy.

Info

Tryb
samolotowy
przy zasianiu
z akumulatora

1 długisygnał

Info

Trybsamolotowy
przy zasilaniu
sieciowym

1 długisygnał

456 – POLPN MN232-C4 K | Instrukcja obsługi

FreeStyle Comfort
Instrukcje poruszania się po menu
Do menu można przejść w trybie gotowości lub
w trybie działania. Jeśli w trybie działania wystąpi
alarm, menu zostanie automatycznie zamknięte. Jeśli
przez 20 sekund nie będą naciskane żadne przyciski,menu zostanie automatycznie zamknięte. W trybie
gotowości może być konieczne dwukrotne naciśnięcie
przycisku Menu w celu wybudzenia urządzenia, abymożliwe było przejście do menu.
Dostępne są cztery ekrany menu:
1. Ekran stanu: pokazuje wersję oprogramowania
układowego, czas pracy, liczbę cykli ładowania
akumulatora

Tryb samolotowy

2. Tryb samolotowy (ON/OFF) (WŁ./WYŁ.)3. Czułość: regulowana (1–5)
4. autoSAT (ON/OFF) (WŁ./WYŁ.)Czułość

Przykład ekranu stanuDZIAŁANIE MENU USTAWIEŃ
Aby przejść do menu ustawień, należy nacisnąć
przycisk Menu/Selectw trybie działania lub
gotowości.
Do przewijania menu służą przyciski

autoSAT.

Strona stanu

PN MN232-C4 K | Instrukcja obsługi

POL – 457

FreeStyle Comfort
STRONA STATUSU
Strona stanu Freestyle Comfort pokazuje wersję
oprogramowania układowego, godziny pracy iliczbę cykli akumulatora na podłączonym akumulatorze.

Wybrane włączenie trybu samolotowego

Aby potwierdzić włączenie trybu samolotowego,naciśnij, gdy widoczna jest ikona SAVE(ZAPISZ). Wartość zmieni się na czarne słowo „ON”(WŁ.).

Strona stanu

Aby anulować włączenie trybu samolotowego, naciśnij
, gdy widoczna jest ikona CANCEL (ANULUJ).

Godziny pracy są wyświetlane w pełnych godzinach.
Rozładowania baterii to liczba cykli, po których
podłączona bateria została wyczerpana i naładowana.

TRYB SAMOLOTOWY
Koncentrator FreeStyle Comfort jest dostarczanyz wyłączonym trybem samolotowym.

Potwierdzone włączenie trybu samolotowego

Aby wyłączyć tryb samolotowy, powtórz powyższe
kroki, przestawiając stan na pozycję OFF (WYŁ.).
Naciśnięcie przycisku, gdy na ekranie jest
widoczny symbol „Dalej”, powoduje przejście donastępnego ekranu opcji.

Ekran trybu samolotowego

Aby ten tryb włączyć, naciśnij
, gdy widoczna
jest ikona CHANGE (ZMIEŃ). Wartość zmieni się
na niebieskie słowo „ON” (WŁ.).

Tryb samolotowy jest przeznaczony do użycia w trybie
pracy. Tryb samolotowy zostanie automatyczniewyłączony, gdy urządzenie zostanie pozostawione
w trybie gotowości na zasilaniu akumulatorowym.

Uwaga: Więcej informacji na temat trybu samolotowego zawiera rozdział Włączanie trybu samolotowego.

TRYB USTAWIANIA CZUŁOŚCI
Koncentrator FreeStyle Comfort jest dostarczanyz fabrycznie ustawionym trybem czułości 2 i w
normalnych warunkach nie powinien wymagać
regulacji. Ustawienie czułości można zmieniać
w zakresie 1–5, gdzie 1 to najmniejsza, a 5 największa
wartość czułości.

458 – POLPN MN232-C4 K | Instrukcja obsługi

FreeStyle Comfort
widoczny symbol „Dalej”, powoduje przejście donastępnego ekranu opcji.

TRYB AUTOSAT
Koncentrator FreeStyle Comfort jest dostarczanyz włączonym trybem autoSAT.
Ekran trybu ustawiania czułości z wartością 2

Aby zmienić czułość, naciśnij
, gdy widoczna jest
ikona CHANGE (ZMIEŃ). Wartość zmieni się na
kolejną i zostanie wyświetlona na niebiesko.

Ekran główny trybu autoSAT

Aby ten tryb wyłączyć, naciśnij
, gdy widoczna
jest ikona CHANGE (ZMIEŃ). Wartość zmieni się na
niebieskie słowo „OFF” (WYŁ.).
Wybrane ustawienie czułości 3

Aby potwierdzić wybrane ustawianie czułości
3, naciśnij
, gdy widoczna jest ikona SAVE(ZAPISZ). Wartość zmieni się na „3” i zostaniewyświetlona na czarno.
Aby anulować ustawianie czułości, naciśnij
widoczna jest ikona CANCEL (ANULUJ).

, gdy
Wybrane wyłączenie trybu autoSAT

Aby potwierdzić wyłączenie trybu autoSAT, naciśnij
, gdy widoczna jest ikona SAVE (ZAPISZ).
Wartość zmieni się na czarne słowo „OFF” (WYŁ.).
Aby anulować włączenie trybu autoSAT, naciśnij
gdy widoczna jest ikona CANCEL (ANULUJ).

,

Potwierdzone ustawienie czułości 3

Potwierdzone wyłączenie trybu autoSAT

Aby włączyć tryb autoSAT, powtórz powyższe kroki,
przestawiając stan na pozycję ON (WŁ.).
Potwierdzone ustawienie czułości 5 (maksymalne)Naciśnięcie przycisku, gdy na ekranie jestNaciśnięcie przycisku, gdy na ekranie jest
widoczny symbol „Dalej”, powoduje przejście donastępnego ekranu opcji.

PN MN232-C4 K | Instrukcja obsługi

POL – 459

FreeStyle Comfort

Rozwiązywanie problemów
ProblemMożliwa przyczyna

RozwiązanieUrządzenie FreeStyle Comfort
pokazuje alarm i emituje
przerywany sygnał dźwiękowy.

Sprawdzić w tabeli Stany
alarmowe.

Sprawdzić w tabeli Stany alarmowe.

Koncentrator tlenu nie włącza
się.

Akumulator jest rozładowany.

Podłączyć urządzenie do gniazda AC lub DC.

Niepodłączony do zewnętrznegoźródła zasilania.

Upewnić się, że połączenia zewnętrzne są
bezpieczne.

Awaria ogólna.

Wykonać pełny cykl zasilania, wyjmując akumulator
i odłączając wszelkie zewnętrzne źródła zasilania
AC lub DC. Następnie ponownie zamontować
akumulator — cykl zasilania zostanie zakończony.
W razie potrzeby ponownie podłączyć zewnętrzneźródło zasilania.
Skontaktować się z dostawcą sprzętu i w razie
potrzeby skorzystać z innego źródła tlenu.

Wszystkie inne problemy.

Wykonać pełny cykl zasilania, wyjmując akumulator
i odłączając wszelkie zewnętrzne źródła zasilania
AC lub DC. Następnie ponownie zamontować
akumulator — cykl zasilania zostanie zakończony.
W razie potrzeby ponownie podłączyć zewnętrzneźródło zasilania.
Skorzystać z innego źródła tlenu, jeśli jestdostępne, i skontaktować się z dostawcą sprzętu.

Uwaga: Niektóre alarmy powodują wyłączenie urządzenia.
Uwaga: W przypadku całkowitej utraty zasilania system alarmowy jest resetowany (czyszczone są wszelkie wcześniejsze alarmy). Po przywróceniu zasilania wyświetlane są tylko te alarmy, których warunki wciąż są spełnione.

460 – POLPN MN232-C4 K | Instrukcja obsługi

FreeStyle Comfort
Informacje dla podróżujących
Koncentrator FreeStyle Comfort spełnia zalecenia
Federalnej Administracji Lotnictwa w zakresieużytkowania urządzeń na pokładzie samolotów
cywilnych. Przed każdą podróżą (samolotem,pociągiem, statkiem itd.) należy:• Upewnić się, że koncentrator jest w dobrym stanie
i działa.
• Skontaktować się z przewoźnikiem co najmniej
48 godzin przed wyruszeniem i poinformować goo zamiarze podróżowania z urządzeniem FreeStyle
Comfort.
Podczas podróży zaleca się zawsze mieć przysobie zapasowe akumulatory. W przypadku
podróży lotniczych należy zadbać, aby energii
z akumulatorów wystarczyło też na czas spędzony
na lotnisku (odprawa, kontrole bezpieczeństwa itp.)
oraz przerwy w podróży. Większość linii lotniczych
wymaga, aby pasażer miał ze sobą taką liczbę w pełni
naładowanych akumulatorów, aby zapewnić zasilanieurządzenia przez co najmniej 150% przewidywanego
maksymalnego czasu podróży (czasy przelotów,
pobyt na lotnisku przed lotem i po, połączenia
i nieoczekiwane opóźnienia).

Uwaga: W przypadku podróży lotniczej zaleca się
przygotować dodatkowe akumulatory w bezpiecznej
torbie.

PRZEPISY FEDERALNEJ ADMINISTRACJILOTNICTWA

Akumulatory zapasowe należy oddzielniezabezpieczyć przed zwarciem, umieszczając jew oryginalnym opakowaniu lub inaczej odizolowując
styki, np. oklejając je taśmą lub umieszczając każdy
akumulator w oddzielnej torebce foliowej lub
woreczku ochronnym. Akumulatory można wnosić
tylko w bagażu podręcznym.

urządzenia z zasilania akumulatorowego nazewnętrzne, należy postępować zgodnie z następującą
procedurą:1. Włącz przenośny koncentrator tlenu
z zainstalowanym akumulatorem i bez zasilania zeźródła zewnętrznego.

2. Włącz tryb samolotowy, używając jednejz następujących metod:
a. Wykonaj czynności mające na celu włączenie
trybu samolotowego, opisane w sekcji Instrukcje
poruszania się po menu.

b. Naciśnij i przytrzymaj przyciski „w górę” i „wdół” przez 5 sekund, aż wyemitowany zostanie
długi sygnał dźwiękowy, a na ekranie wyświetlony
zostanie symbol samolotu.

WŁĄCZANIE TRYBU SAMOLOTOWEGO
Zasilanie akumulatorowe
Przenośnego koncentratora tlenu można używać
w samolocie w normalny sposób, gdy jest zasilany
TYLKO z akumulatora, a zewnętrzne źródło zasilania
nie jest podłączone.

3. Po pojawieniu się symbolu podłącz przenośny
koncentrator tlenu do gniazda zasilania.

Zasilanie zewnętrzneJeśli w samolocie dostępne jest działające gniazdo
elektryczne, z którego można zasilać przenośny
koncentrator tlenu, może ono stanowić alternatywne
źródło zasilania. W takim wypadku, aby przełączyć

Uwaga: Ze względu na ograniczenia obowiązujące
w samolocie akumulatory nie ładują się w trybie
samolotowym.

PN MN232-C4 K | Instrukcja obsługi

POL – 461

FreeStyle Comfort

Akcesoria

Aby zapewnić odpowiednie działanie i bezpieczeństwo,należy używać wyłącznie wyszczególnionych
akcesoriów dostarczanych przez firmę CAIRE zapośrednictwem dostawcy sprzętu do podawania
tlenu. Korzystanie z akcesoriów niewymienionych
na poniższej liście może mieć negatywny wpływ na
działanie lub bezpieczeństwo koncentratora.
Standardowe akcesoria FreeStyle Comfort
Torba

MI396-1

PlecakMI459-1WKŁADANIE URZĄDZENIA DO TORBY

Pasek na ramię

MI397-1

Zasilacz AC (wraz ze wszystkimi
przewodami)

PW036-1S

1. Umieść urządzenie na dole torby, ustawiając je tak,aby otwory na filtry i akumulator znalazły się wewłaściwym miejscu.

Zasilacz DC

CD041-1Zewnętrzny filtr dużych cząstek
(prawy)FI226-1Zewnętrzny filtr dużych cząstek
(lewy)

FI227-1Dostępne akcesoria opcjonalne
Dodatkowy akumulator 8-ogniwowy

BT034-1S

Akumulator 16-ogniwowy

BT035-1SŁadowarka biurkowa

BT036-1

Torba na wszystkie akcesoria

MI372-2

2. Zamknij zamek na dole torby.

Uwaga: Jeśli tak napisano, mogą być dostępne
specjalne przewody zasilania dla poszczególnychkrajów. Jeśli potrzebny jest alternatywny przewód,należy skontaktować się z firmą CAIRE lub dostawcą
sprzętu do podawania tlenu.

Instrukcje dotyczące torby
Aby umieścić i zabezpieczyć urządzenie FreeStyle
Comfort w torbie, należy postępować zgodnie
z poniższymi instrukcjami.

PRZYCZEPIANIE PASKA NA RAMIĘ
1. Pasek na ramię można przyczepić do urządzenia
lub torby.
2. Przypnij pasek do ucha urządzenia lub torby.

462 – POLPN MN232-C4 K | Instrukcja obsługi

FreeStyle Comfort
Utylizacja i ponowne użycie
W celu prawidłowej utylizacji urządzenie FreeStyle
Comfort wraz z wszystkimi komponentami należy
oddać do dostawca sprzętu. Można też skontaktowaćsię z urzędem gminy lub miasta, aby dowiedzieć się,jak prawidłowo zutylizować akumulator. Urządzeniemoże być ponownie wykorzystywane między
ludźmi, ale musi zostać zwrócone do dostawcy opieki
domowej w celu ponownego przetworzenia.

Dyrektywy WEEE i RoHSsprzętu elektrycznego i elektronicznego (WEEE) pozakończeniu eksploatacji należy go oddać do zakładu
utylizacji.
Nasze produkty są zgodne z Dyrektywą w sprawie
ograniczenia stosowania niektórych niebezpiecznych
substancji w sprzęcie elektrycznym i elektronicznym(RoHS). Zawierają jedynie śladowe ilości ołowiu
i innych niebezpiecznych materiałów.

Ten symbol przypomina właścicielomsprzętu, że zgodnie z Dyrektywą w sprawie zużytego

PRZESTROGA: Aby dowiedzieć się, jak
prawidłowo zutylizować urządzenie, należy
skontaktować się z dostawcą sprzętu lub
odpowiednim organem władz lokalnych.

Materiały, z którymi osoba obsługująca urządzenie ma bezpośredni lubpośredni kontakt
Obudowa koncentratora............................................................................................................................... PC/ABS
Panel sterowania koncentratora............................................................................Poliester EBG7 lub odpowiednik
Uchwyt koncentratora..............................................................................................................................Poliwęglan
Złącze wylotowe tlenu, dysza..................................................................................................................Poliwęglan
Etykieta urządzenia..........................................................................................................................................LexanZłącza na przewodach.............................................................................................. Poliwęglan/polichlorek winylu
Przewód zasilania.....................................................................................................................................PCV, metal
Zasilacz............................................................................................................................... Lexan 940 (poliwęglan)
Akumulator.................................................................................................................................................. PC/ABS
Etykiety na akumulatorze, zasilaczach..........................................................................................Folia poliestrowa
Futerał na koncentrator............................................................................. Cordura, neopren, podszewka nylonowa
Pasek.................................................................................................Polipropylen, Acetyl, winyl, cordura, neopren

Testy kompatybilności elektromagnetycznej
Wskazówki i deklaracja producenta — emisja promieniowania elektromagnetycznegoUrządzenie FreeStyle Comfort jest przeznaczone do użytku w środowisku elektromagnetycznym opisanym poniżej. Klient
lub użytkownik urządzenia FreeStyle Comfort powinien upewnić się, że jest ono używane w takim środowisku.
Test emisji

Zgodność

Emisje o częstotliwości fal
radiowych
Grupa 1CISPR 11
Emisje o częstotliwości fal
radiowych
Klasa BCISPR 11
Emisje harmoniczne
Klasa AIEC 61000-3-2
Fluktuacja napięcia
/ emisje migotania
ZgodneIEC 61000-3-3Środowisko elektromagnetyczne — wskazówkiUrządzenie FreeStyle Comfort wykorzystuje energię o częstotliwości
radiowej wyłącznie do wewnętrznego działania. W związku z tym
emisje o częstotliwości radiowej są niewielkie i nie powinny
powodować żadnych zakłóceń w działaniu znajdującego się w pobliżu
sprzętu elektronicznego.

Urządzenie FreeStyle Comfort jest dostosowane do działania
w każdym środowisku, w tym w warunkach domowych oraz miejscachbezpośrednio podłączanych do publicznej sieci zasilania o niskimnapięciu, która zasila budynki mieszkalne.

PN MN232-C4 K | Instrukcja obsługi

POL – 463

FreeStyle Comfort
Wskazówki i deklaracja producenta — odporność elektromagnetycznaUrządzenie FreeStyle Comfort jest przeznaczone do użytku w środowisku elektromagnetycznym opisanym poniżej. Klient
lub użytkownik urządzenia FreeStyle Comfort powinien upewnić się, że jest ono używane w takim środowisku.
Test odporności
Poziom testu IEC 60601 Poziom zgodności
Środowisko elektromagnetyczne —wskazówki
Podłogi powinny być drewniane, betonoweŚrodowisko±8 kV przy dotyku±8 kV przy dotyku
lub wyłożone płytkami ceramicznymi.
elektromagnetyczne
Jeśli podłogi pokryte są materiałem±15 kV przy wylocie±15 kV przy wylocie— wskazówki
syntetycznym, wilgotność względna
powietrza
powietrza
IEC 61000-4-2
powinna wynosić przynajmniej 30%.
±2 kV dla linii zasilających ±2 kV dla linii
Nagły skok / impulsJakość zasilania sieciowego powinna
zasilających
elektryczny
być typowa dla profesjonalnej placówki±1 kV dla linii wejścia/szpitalnej lub środowiska opieki domowej.
IEC 61000-4-4wyjścia
Nie dotyczy±1 kV tryb różnicowy

Skok napięcia
IEC 61000-4-5±2 kV tryb wspólny na
liniach AC±1 kV tryb różnicowy±1 kV tryb różnicowy na
liniach AC±2 kV tryb wspólnyna liniach We/Wy nazewnątrz
0% UT przez 0,5 cyklu
0% UT przez 0,5 cyklu(0°, 45°, 90°, 135°, 180°, (0°, 45°, 90°, 135°, 180°,
225°, 270°, 315°)
225°, 270°, 315°)

Spadki, krótkie
przerwy orazzmiany napięcia na
liniach wejściowych
zasilania IEC 610004-11

0% UT przez 1 cykl (00)

0% UT przez 1 cykl (00)

70% UT (30% spadek
w UT) przez 25/30 cykli(00)

70% UT (30% spadek
w UT) przez 25/30 cykli(00)

0% UT przez 250/300
cykli (00)

0% UT przez 250/300
cykli (00)Pole magnetyczne
o częstotliwości
3 A/m
3 A/m
źródła zasilania(50/60 Hz)
IEC 61000-4-8
UWAGA: UT to napięcie sieci AC przed zastosowaniem poziomu testowego.

Jakość zasilania sieciowego powinna
być typowa dla profesjonalnej placówki
szpitalnej lub środowiska opieki domowej.

Jakość zasilania sieciowego powinna
być typowa dla profesjonalnej placówki
szpitalnej lub środowiska opieki domowej.
Jeśli użytkownik urządzenia FreeStyle
Comfort wymaga ciągłego działania
w trakcie przerw w zasilaniu, zaleca się
wyposażenie koncentratora FreeStyle
Comfort w zasilacz awaryjny (UPS) lub
akumulator.

Pole magnetyczne o częstotliwości
źródła zasilania powinno być na
poziomie charakterystycznym dla typowej
profesjonalnej placówki szpitalnej lubśrodowiska opieki domowej.

UWAGA: W niektórych przypadkach urządzenie Freestyle Comfort może zresetować się po dużym wyładowaniu
elektrostatycznym. W celu przywrócenia normalnego działania może być konieczne wykonanie pełnego cyklu zasilania.
Szczegółowe informacje zawarte są w części poświęconej rozwiązywaniu problemów.

464 – POLPN MN232-C4 K | Instrukcja obsługi

FreeStyle Comfort
Wskazówki i deklaracja producenta — odporność elektromagnetycznaUrządzenie FreeStyle Comfort jest przeznaczone do użytku w środowisku elektromagnetycznym opisanym poniżej. Klient
lub użytkownik urządzenia FreeStyle Comfort powinien upewnić się, że jest ono używane w takim środowisku.
Test odporności Poziom testu IEC 60601 Poziom zgodności
Środowisko elektromagnetyczne —wskazówki
Przewodzona
10 Vrms
10 Vrms
Przenośny sprzęt radiokomunikacyjny nieczęstotliwość
powinien być używany bliżej dowolnej części
od 150 kHz do 80 MHz
radiowaurządzenia FreeStyle Comfort, włącznie
z przewodami, niż zalecana odległość
IEC 61000-4-6
wyliczona ze wzoru odpowiedniego dlaczęstotliwości nadajnika.
10 V/m
10 V/mPromieniowanieo częstotliwości
radiowej
IEC 61000-4-3

od 80 MHz do 2700 MHz

d = 1,2√ P

Zalecana odległość pomiędzy urządzeniami

d = 1,2√ P = od 80 MHz do 800 MHz
d = 1,2√ P = od 800 MHz do 2700 MHz
gdzie P jest maksymalną wartością mocywyjściowej nadajnika w watach (W) zgodnie
z danymi producenta nadajnika, a d jest
zalecaną odległością między urządzeniami
w metrach (m).
Natężenia pola pochodzącego
z umieszczonych na stałe nadajników
pracujących na częstotliwościachradiowych, określone na podstawie pomiaru
poziomu zakłóceń elektromagnetycznych
w miejscu montażua, powinny być mniejszeniż poziom zgodności w każdym zakresieczęstotliwościb.
Zakłócenia mogą występować w pobliżu
urządzeń oznaczonych następującymi
symbolami:

UWAGA 1: Przy 80 MHz i 800 MHz stosowany jest zakres większych częstotliwości.
UWAGA 2: Te zalecenia mogą nie mieć zastosowania we wszystkich sytuacjach. Na propagację elektromagnetyczną
wpływa pochłanianie i odbijanie od struktur, przedmiotów i ludzi.
aNatężeń pola generowanych przez nadajniki stałe, takie jak radiostacje bazowe (komórkowe/bezprzewodowe)do telefonów, przenośne aparaty radiowe, amatorskie stacje radiowe, urządzenia do nadawania na falachśrednich i ultrakrótkich oraz telewizyjnych, nie można teoretycznie dokładnie przewidzieć. Aby ocenić środowisko
elektromagnetyczne powstałe na skutek działania nadajników pracujących na falach radiowych, należy rozważyć
przeprowadzenie pomiarów w miejscu instalacji. Jeśli pomiary natężeń pola w miejscu, w którym używany jest koncentrator
serii FreeStyle Comfort, przekraczają odpowiedni poziom zgodności fal radiowych podany powyżej, urządzenie FreeStyle
Comfort należy monitorować pod kątem prawidłowego działania. Jeśli zostanie zaobserwowane nieprawidłowe działanie,może być konieczne podjęcie dodatkowych środków takich jak zmiana orientacji lub ustawienia urządzenia FreeStyle
Comfort.
b

Przy zakresie częstotliwości od 150 kHz do 80 MHz, natężenia pola powinny wynosić mniej niż 10 V/m.

PN MN232-C4 K | Instrukcja obsługi

POL – 465

FreeStyle Comfort
Zalecane odległości między przenośnym sprzętem komunikacyjnym pracującym naczęstotliwościach radiowych a urządzeniem FreeStyle ComfortUrządzenie FreeStyle Comfort powinno być używane w środowisku elektromagnetycznym, w którym zakłócenia
promieniowania o częstotliwościach radiowych są pod kontrolą. Klient lub użytkownik urządzenia FreeStyle Comfort możepomóc zapobiec zakłóceniom elektromagnetycznym poprzez utrzymywanie minimalnej odległości pomiędzy przenośnym
sprzętem komunikacyjnym działającym na częstotliwościach radiowych (nadajnikami) a modułem FreeStyle Comfort w sposób
opisany poniżej, zgodnie z maksymalną mocą wyjściową sprzętu komunikacyjnego.
Maksymalna wartość
Odległość w zależności od częstotliwości nadajnika
mocy wyjściowejmnadajnika
od 150 kHz do 80 MHz
od 80 MHz do 800 MHz
od 800 MHz do 2,5 GHz

d = 1,2√ P

d = 1,2√ P

d = 1,2√ P

W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
W przypadku nadajników, których maksymalna moc wyjściowa nie jest wymieniona powyżej, zalecaną odległość pomiędzy
urządzeniami d w metrach (m) można oszacować, stosując odpowiednie równanie dla częstotliwości nadajnika, gdzie
P jest maksymalną wartością mocy wyjściowej nadajnika w watach (W), zgodnie z danymi producenta nadajnika.
UWAGA 1: Przy 80 MHz i 800 MHz stosowana jest odległość dla zakresu większych częstotliwości.
UWAGA 2: Te zalecenia mogą nie mieć zastosowania we wszystkich sytuacjach. Na propagację elektromagnetyczną
wpływa pochłanianie i odbijanie od struktur, przedmiotów i ludzi.

Klasyfikacja

Typ zabezpieczenia przed porażeniem elektrycznym:
Klasa II	Zabezpieczenie przed porażeniem
elektrycznym uzyskiwane jest poprzez
podwójną izolację.
Stopień ochrony przed porażeniem elektrycznym:
Typ BF Urządzenie zapewniające określony
stopień zabezpieczenia przed porażeniem
elektrycznym w zakresie:		 1) dopuszczalnego upływu prądu,		2) niezawodności uziemienia ochronnego (jeśli
zastosowane).
		 Urządzenie nie jest przeznaczone do
bezpośredniego zastosowania kardiologicznego.
Niezależne testy dla celów sprawdzenia zgodności
z normami dla elektrycznych urządzeń medycznych:
Testy wykazały zgodność z normami• IEC 60601-1 edycja 3.1: Medyczne urządzenia
elektryczne — Cześć 1. Wymagania ogólne
dotyczące bezpieczeństwa.
• CAN/CSA C22.2 No. 60601-1-8 wyd. 2.1,
Medyczne urządzenia elektryczne — Część 1.
Wymagania ogólne dotyczące bezpieczeństwa.
Ochrona przed potencjalnymi zakłóceniami
elektromagnetycznymi i innymi pomiędzy
urządzeniami a innym sprzętem.
	• Testy wykazały zgodność z normą EN 60601-1-2/
IEC 60601-1-2. Wyd. 4.
	• Testy wykazały zgodność z normą RTCA/DO160
sekcja 21 kategoria M.
	• CISPR 11 / EN 55011 klasa B grupa 1,„Urządzenia przemysłowe, naukowe i medyczne”.
466 – POLPN MN232-C4 K | Instrukcja obsługi

	• FCC część 15 punkt B — klasa B, Źródło
promieniowania niezamierzonego.
IP22 — Ochrona przed wkładaniem palców i innychciał o średnicy większej niż 12,5 mm. Krople wodyspadające pionowo nie mają negatywnego wpływu,gdy obudowa jest odchylona od normalnej pozycji
pod kątem 15°.
Dozwolone metody czyszczenia i zapobiegania
przenoszeniu chorób zakaźnych: Informacjedostępne są w części „Czyszczenie, pielęgnacja
i rutynowe czynności konserwacyjne” niniejszej
instrukcji obsługi urządzenia FreeStyle.
Poziom bezpieczeństwa w przypadku użytkowania
w obecności łatwopalnych gazów znieczulających:Urządzenie nie jest przeznaczone do zastosowania
w takich warunkach.
Tryb działania: Praca ciągła.

CAIRE i CAIRE Inc. są zastrzeżonymi znakami towarowymi firmy CAIRE
Inc. Aby uzyskać pełną listę znaków towarowych, odwiedź naszą stronę
internetową poniżej.
Znaki towarowe: www.careinc.com/corporate/trademarks.
Copyright © 2021 CAIRE Inc. Firma CAIRE Inc. zastrzega sobie prawo do
wstrzymania sprzedaży produktów, zmiany cen, materiałów, sprzętu, jakości,opisów, danych technicznych lub procesów związanych z produktami
w dowolnym momencie, bez wcześniejszego powiadomienia oraz bez
dodatkowych zobowiązań i konsekwencji. Firma zastrzega sobie wszystkie
prawa, które nie zostały wyraźnie określone w niniejszym dokumencie (jeśli
ma to zastosowanie).

FreeStyle Comfort
Felhasználói vezérlők és a rendszer állapotjelzői
ISO 7000-es szabvány szerinti, a készülék
használatára vonatkozó grafikus szimbólumok –Tárgymutató és összegzésÜzemeltetés előtt olvassa el afelhasználói kézikönyvet.
Reg. sz.: 1641
Tárolási vagy működési hőmérsékleti
tartomány. Reg. sz.: 0632

PTárolási páratartalom. Reg. sz.: 2620

Az orvostechnikai eszközökről szóló 93/42/EGK
tanácsi irányelv
Meghatalmazott képviselő az Európai
Közösségben.
Ha a termék egyedi eszköz
azonosítójának (UDI) címkéjén
CE #### szimbólum szerepel,
akkor az eszköz megfelel a
93/42 / EGK irányelvnek az
orvostechnikai eszközökre vonatkozó
követelményeinek. A CE ####
szimbólum jelzi a bejelentett szervezet
számát.
IEC 60417
II. osztályú berendezés, kettős
szigetelésű. Reg. sz.: 5172

Légköri nyomás szerinti korlátozás.
Reg. sz.: 2621

Főkapcsoló. Reg. sz.: 5009

A gyártó neve és címe. Reg. sz.: 3082
Gyártás dátuma. Reg. sz.: 2497

ETL igazolás
CAN/CSA biztonsági hivatala szerinti
C22.2 601.1 M90 számú, a gyógyászati
villamos készülékekre vonatkozóan.

Katalógusszám. Reg. sz.: 2493
Sorozatszám. Reg. sz.: 2498
Álló helyzetet jelölő nyilak.
Reg. sz.: 0623

Belső szimbólumok
Tartsa távol a gyúlékony anyagoktól,
olajtól és zsírtól!

Vigyázat, törékeny! Reg. sz.: 0621
Óvja az esőtől, tartsa szárazon.
Reg. sz.: 0626

Ne szerelje szét!

Egymásra helyezési korlát darabszám
alapján. Reg. sz.: 2403

Áramlási sebesség növelése.

ISO 7010-es szabvány: Grafikus szimbólumok –Biztonsági színek és biztonsági jelzések – Regisztrált
biztonsági jelzések
Olvassa el a használati útmutatót.
Reg. sz.: M002
Tartsa távol nyílt lángtól, tűztől,szikrától. A dohányzás és a nyílt lánghasználata tilos! Reg. sz.: P003
Az egység közelében vagy annaküzemeltetésekor tilos a dohányzás!
Reg. sz.: P002
BF típusú alkalmazott alkatrész(elektromos áramütés elleni védelem
fokozata). Reg. sz.: 5333

Áramlási sebesség csökkentése.
Menü gomb.
RTCA/DO160 Section
21
Category M
Compliant

RTCA DO160 21. fejezet
M kategóriának megfelelő.
FAA SFAR 106 követelmény.
II. osztályú berendezés, kettős
szigetelésű.
FAA által engedélyezett – POC.

Figyelmeztetés. Reg. sz.: W001
PN MN232-C4 K | Felhasználói kézikönyv

HUN – 467

FreeStyle Comfort
21 CFR 801.15: Szövetségi rendeletek kódexe –21. cím
A szövetségi törvény jelen készülékértékesítését orvos által vagy orvosi
rendelvényre korlátozza.
A 2012/19/EU tanácsi irányelv az elektromos éselektronikus berendezések hulladékairól
WEEE

Mit jelent az, hogy
oxigénkoncentrátor?
A levegő, amit belélegzünk, körülbelül 21% oxigént,
78% nitrogént és 1% egyéb gázokat tartalmaz. A
FreeStyle® Comfort® esetében a készülék a szoba
levegőjét szívja be a légbeszívó nyílásokon keresztül.
Ezután ezt egy adszorbens anyagon, úgynevezettmolekuláris szűrőn keresztül vezeti át. Ez az anyag
elkülöníti az oxigént a nitrogéntől, és csak az oxigént
hagyja átáramolni. Ennek eredményeként nagy
oxigénkoncentrációjú gázáram jut el a felhasználóhoz.
A FreeStyle Comfort egyesíti a modern
oxigénkoncentrátort és az oxigénmegőrző
technológiákat, hogy egy olyan hordozható készüléket
hozzon létre, amely maximális hordozhatóság
biztosít, hogy a beteg szabadon tudjon mozogni. A
fejlett impulzusáramlású oxigénbeviteli készülék
gyorsan érzékeli a felhasználó belégzését, és minden
belégzés kezdetekor egy oxigénimpulzust adagol. Az
impulzusáramlású gázbevitel ideális a mozgáshoz
és az otthontól távol végzett tevékenységekhez. A
folyamatos gázáramláshoz képest hatékonyabban
gazdálkodik az oxigénnel, és meghosszabbítja az
akkumulátor üzemidejét is. A FreeStyle Comfort
elég könnyű ahhoz, hogy vállszíjon magával vihesse.
Működtethető váltóáramról (fali csatlakozóról),egyenáramról (gépjárműben) vagy akkumulátorról is.

Rendeltetésszerű használat
A CAIRE FreeStyle Comfort oxigénkoncentrátor
kiegészítő oxigén beadására szolgál. A készüléket neméletmentésre szánják, és nem nyújt semmilyen betegfigyelési képességet.

Miért írt fel Önnek az orvos oxigént?
Sokan élnek különböző szív-, tüdő- és más légzőszervi
betegséggel. Ezeknek a betegeknek a nagy részénél
hasznos lehet a kiegészítő oxigénterápia otthonukban,
utazás közben vagy az otthonon kívüli napi
tevékenységek során.
Az oxigén egy gáz, amely az általunk belélegzett
levegő 21%-át teszi ki, és amely folyamatosanszükséges ahhoz, hogy a szervezetünk megfelelően
tudjon működni. Orvosa kifejezetten az Ön légzőszervi
problémájának megfelelő áramlási sebességet vagy

468 - HUNPN MN232-C4 K | Felhasználói kézikönyv

IEC 60601-1-es szabvány: Gyógyászati villamos
készülékek. 1. rész: Az alapvető biztonságra
és a lényeges működésre vonatkozó általános
követelmények
Cseppálló
Ez a termék egy vagy több USA-beli és nemzetközi
szabadalom védelme alatt állhat. A vonatkozó
szabadalmak listájának megtekintéséhez, kérjük,
látogasson el weboldalunkra az alábbi címen.
Szabadalmak: www.caireinc.com/corporate/patents/.
beállítást írt fel Önnek.
Bár az oxigén használata nem okoz függőséget, afelügyelet nélkül végzett oxigénterápia veszélyes
lehet. Az oxigénkoncentrátor használata előtt
forduljon orvoshoz és kérjen tanácsot. A berendezés
forgalmazója, aki leszállítja az oxigénberendezést,
megmutatja Önnek, hogy hogyan kell a felírt áramlási
sebességet beállítani.

Kezelői profil
A koncentrátor arra szolgál, hogy kiegészítő oxigént
juttasson azoknak a betegeknek, akiknél valamilyen
betegség károsan befolyásolja tüdejük arra irányuló
képességét, hogy a levegő oxigénjét a vérbe
juttassa, és ezért panaszaik vannak. A hordozható
oxigénkoncentrátorok (POC készülékek) nem tárolnak
és nem tartalmaznak oxigént. Nem kell utántölteni
ezeket, és az akkumulátoruk bármikor feltölthető az
elérhető váltó- vagy egyenáramú áramforrásról (kivévea repülőgépeken). Ezáltal a felhasználó viszonylagosönellátásra képes akár otthoni, akár mozgás közbeni
használat esetén (mind a házon belül, mind azon
kívül), mobilissé válik, és általánosságban elmondható,
hogy a megszokott életét élheti. Az oxigénkoncentrátor
használatához orvosi rendelvényre van szükség, és
nem alkalmas életfenntartó használatra.
Bár oxigénterápiát bármilyen életkorú betegnek
fel lehet írni, a felhasználók általában 65 évnél
idősebb betegek, akiknek különböző légzőrendszeri
betegségük van, mint például a krónikus obstruktívtüdőbetegség (COPD). A felhasználók általában jó
kognitív képességgel rendelkeznek, és képesnekkell lenniük jelezni, ha rosszul érzik magukat. Haa beteg nem képes közölni, ha rosszul van, vagy
ha nem tudja elolvasni és megérteni a koncentrátor
címkéjét és kezelési útmutatóját, akkor a használatcsak olyan személy felügyelete mellett javasolt, aki
képes ezekre. Ha a koncentrátor használata közben
bármilyen kellemetlenséget érez, akkor javasoljuk,
hogy beszéljen egészségügy szakemberrel. Szintén
javasoljuk, hogy rendelkezzen tartalék oxigénforrással
(azaz oxigénpalackkal) arra az esetre, ha áramszünet
van, vagy a koncentrátor meghibásodik. A koncentrátor
használatához nincs szükség semmilyen egyéb
különleges készségre vagy képességre.

A FreeStyle Comfort kicsomagolása
Ellenőrizze, hogy minden
felsorolt alkotóelem
megtalálható-e a csomagolásban.
Ha bármelyik tétel hiányzik,azonnal forduljon aforgalmazóhoz.

Szokásos tartozékok• Hordozható oxigénkoncentrátor• Hordtáska• Vállszíj• 8 cellás akkumulátorcsomag• Váltóáramú tápegység• Váltóáramú tápkábel• Egyenáramú tápegység• Felhasználói kézikönyv

FreeStyle Comfort
Opcionális tartozékok• Kiegészítő 8 cellás
akkumulátorcsomag• 16 cellás akkumulátorcsomag• Asztali töltő
• Tartozék hordtáska

A FreeStyle Comfort bemutatása
Ismerje meg a FreeStyle Comfort legfontosabb funkcióit és a felhasználói felületét.
Hordszíj
csatlakoztatója

Elülső nézet

Felhasználói
felületOxigénkimenet
Levegőkimenet

Légbeszívó/szűrők
Légbeszívó/szűrőkTápcsatlakozó
aljzattakarófedele

Akkumulátorcsomag

Akkumulátorcsomagkioldógombja
PN MN232-C4 K | Felhasználói kézikönyv

HUN – 469

FreeStyle ComfortFelső nézet
Kijelző
Áramlási sebesség
növelése

Főkapcsoló

Menü gombÁramlási sebesség
csökkentése

Kijelző akkumulátoros működésnél
Áramlás erőssége

Akkumulátor
töltöttsége

autoSAT jelző
Riasztási üzenet

Akkumulátor
töltöttségének
százalékos
kijelzése

Bólusbejuttatásjelző

Kijelző külső (váltó- vagy egyenáramú) áramforrásnál

Áramlás erőssége
Akkumulátor
töltöttsége

Riasztási üzenet
helye

470 - HUNPN MN232-C4 K | Felhasználói kézikönyv

Töltésjelző

FreeStyle Comfort
Fontos!

A használat ellenjavallatai

A következő biztonsági figyelmeztetések
vannak érvényben:
FIGYELMEZTETÉS: FONTOS BIZTONSÁGI
INFORMÁCIÓK OLYAN VESZÉLYEKKELKAPCSOLATBAN, AMELYEK SÚLYOSEGÉSZSÉGKÁROSODÁST OKOZHATNAK.
VIGYÁZAT: Fontos tudnivalók, hogy
elkerülhesse a FreeStyle Comfort
megrongálódását.

Megjegyzés: Különös figyelmet igénylő információk.

Alkalmazási terület
FIGYELMEZTETÉS: NAGYON FONTOS,
HOGY CSAK A FELÍRT OXIGÉNSZINTETÁLLÍTSA BE. CSAK AKKOR MÓDOSÍTSA A
KIVÁLASZTOTT ÁRAMLÁSI SEBESSÉGET,HA EGY ILLETÉKES ORVOSTÓL ERREKAP UTASÍTÁST.
FIGYELMEZTETÉS: AZ AMERIKAI EGYESÜLT
ÁLLAMOK SZÖVETSÉGI TÖRVÉNYEI A JELEN
BERENDEZÉS FORGALMAZÁSÁT ÉS BÉRLETÉTCSAK ORVOS VAGY MÁS EGÉSZSÉGÜGYI
SZAKEMBER RENDELVÉNYÉRE ENGEDÉLYEZIK.
FIGYELMEZTETÉS: A KÉSZÜLÉKET NEM
SZABAD ÉLETFENNTARTÁSRA HASZNÁLNI. HA
A KÉSZÜLÉKET IDŐSKORÚ, GYERMEKKORÚ
VAGY BÁRMILYEN MÁS OLYAN FELHASZNÁLÓHASZNÁLJA, AKI NEM KÉPES JELEZNI, HA ROSSZULVAN A KÉSZÜLÉK HASZNÁLATA KÖZBEN, AKKORTOVÁBBI MEGFIGYELÉSRE LEHET SZÜKSÉG. AHALLÁS- ÉS/VAGY LÁTÁSSÉRÜLT FELHASZNÁLÓK
SEGÍTSÉGRE SZORULHATNAK A RIASZTÁSOK
FIGYELÉSÉBEN.
FIGYELMEZTETÉS: MINDEN FELHASZNÁLÓESETÉBEN EGYÉNILEG KELL MEGHATÁROZNIAZ IMPULZUS-MÓD BEÁLLÍTÁSAIT A NYUGALMI,FIZIKAI AKTIVITÁS ÉS UTAZÁS KÖZBENISZÜKSÉGLETÜKNEK MEGFELELŐEN.
FIGYELMEZTETÉS: A BETEG EGYES LÉGZÉSIERŐFESZÍTÉSEI ESETLEG NEM INDÍTJÁK EL AZOXIGÉNMEGTAKARÍTÁSI FUNKCIÓT.
FIGYELMEZTETÉS: A FREESTYLE COMFORT NEM
HASZNÁLHATÓ TRACHEOTOMIZÁLT BETEGEKNÉL.
FIGYELMEZTETÉS: TERHES VAGY SZOPTATÓ NŐK
NEM HASZNÁLHATJÁK A JELEN KÉZIKÖNYVBENAJÁNLOTT TARTOZÉKOKAT, MERT AZOKFTALÁTOKAT TARTALMAZHATNAK.

FIGYELMEZTETÉS: A NEM ORVOSÁLTAL ELRENDELT OXIGÉNHASZNÁLAT
BIZONYOS KÖRÜLMÉNYEK KÖZÖTT
VESZÉLYES IS LEHET. A KÉSZÜLÉKETCSAK ORVOS ÁLTAL ELRENDELTESETBEN SZABAD HASZNÁLNI.
FIGYELMEZTETÉS: NE HASZNÁLJA A KÉSZÜLÉKETGYÚLÉKONY ALTATÓSZEREK JELENLÉTÉBEN!
FIGYELMEZTETÉS: MINT MINDEN ELEKTROMOS
KÉSZÜLÉK ESETÉBEN, A FELHASZNÁLÓ ITT ISTAPASZTALHAT OLYAN IDŐSZAKOKAT, AMIKORA KÉSZÜLÉK NEM MŰKÖDIK. ENNEK LEHETOKA ÁRAMSZÜNET, ILLETVE SZERVIZMUNKÁRAIS SZÜKSÉG LEHET, AMIT SZAKKÉPZETTTECHNIKUSNAK KELL ELVÉGEZNIE. AZ
OXIGÉNKONCENTRÁTOR HASZNÁLATA NEM
MEGFELELŐ OLYAN FELHASZNÁLÓ SZÁMÁRA,AKINÉL AZ ILYEN ÁTMENETI ÜZEMKIESÉSEKSÚLYOS EGÉSZSÉGÜGYI KÖVETKEZMÉNYEKKEL
JÁRNAK.

Biztonsági irányelvek
FIGYELMEZTETÉS: A KÉSZÜLÉKEN TILOS
MÓDOSÍTÁST VÉGEZNI!
FIGYELMEZTETÉS: EZ A KÉSZÜLÉK NAGY
KONCENTRÁCIÓJÚ OXIGÉNT AD LE, AMI ELŐSEGÍTIA GYORS ÉGÉST. NE ENGEDJE MEG A DOHÁNYZÁST
VAGY NYÍLT LÁNG HASZNÁLATÁT ABBAN AHELYISÉGBEN, AHOL (1) EZ A KÉSZÜLÉK VAGY (2)BÁRMILYEN OXIGÉNT SZÁLLÍTÓ TARTOZÉK VAN.
ENNEK A FIGYELMEZTETÉSNEK A FIGYELMEN KÍVÜL
HAGYÁSA SÚLYOS TÜZET, ANYAGI KÁROKAT ÉS/VAGY FIZIKAI SÉRÜLÉST VAGY HALÁLT OKOZHAT.
FIGYELMEZTETÉS. NE MŰKÖDTESSE AZ
EGYSÉGET KISMÉRETŰ VAGY ZÁRT TEREKBEN(PL. KIS TOKBAN VAGY KÉZITÁSKÁBAN), AHOL
KORLÁTOZOTT A LÉGCSERE LEHETŐSÉGE.
EZ UGYANIS AZ OXIGÉNKONCENTRÁTOR
TÚLHEVÜLÉSÉT ÉS A TELJESÍTMÉNYÉNEKROMLÁSÁT OKOZHATJA.
FIGYELMEZTETÉS: A KONCENTRÁTORT OLYANHELYRE KELL TENNI, AHOL EL LEHET KERÜLNI A
FÜSTÖT, A SZENNYEZŐ ANYAGOKAT ÉS A PÁRÁT.
FIGYELMEZTETÉS: HA KOMOLY ESEMÉNY TÖRTÉNIKEZZEL A KÉSZÜLÉKKEL, A FELHASZNÁLÓNAKAZONNAL ÉRTESÍTENIE KELL AZ ESEMÉNYT ASZOLGÁLTATÓVAL ÉS / VAGY A GYÁRTÓVAL.
A SÚLYOS ESEMÉNYT SÉRÜLÉS, HALÁL VAGYSÉRÜLÉS / HALÁL OKAKÉNT KELL MEGHATÁROZNI,HA AZ ESEMÉNY MEGISMÉTLŐDIK. A FELHASZNÁLÓJELENTHETI AZ ESEMÉNYT ANNAK AZ ORSZÁGNAK
AZ ILLETÉKES HATÓSÁGÁHOZ IS, AHOL AZ
ESEMÉNY BEKÖVETKEZETT.

PN MN232-C4 K | Felhasználói kézikönyv

HUN – 471

FreeStyle Comfort
FIGYELMEZTETÉS: A NEM EHHEZ AZOXIGÉNKONCENTRÁTORHOZ VALÓEGYES OXIGÉNBEVITELI TARTOZÉKOKHASZNÁLATA BEFOLYÁSOLHATJAA TELJESÍTMÉNYT. A JAVASOLTTARTOZÉKOK LISTÁJA MEGTALÁLHATÓEBBEN A KÉZIKÖNYVBEN.

FIGYELMEZTETÉS: NE HAGYJA, HOGYAKÁR A LÉGBESZÍVÓ, AKÁR A LÉGKIMENETINYÍLÁSOK TAKARVA LEGYENEK. NE EJTSENVAGY HELYEZZEN BE SEMMILYEN TÁRGYAT AKÉSZÜLÉK SEMELYIK NYÍLÁSÁBA. EZ UGYANISAZ OXIGÉNKONCENTRÁTOR TÚLHEVÜLÉSÉT ÉS ATELJESÍTMÉNYÉNEK ROMLÁSÁT OKOZHATJA.

FIGYELMEZTETÉS: HA AZ OXIGÉNKONCENTRÁTORT
LEEJTETTE, AZ MEGSÉRÜLT VAGY VÍZ ÉRTE,AKKOR KERESSE MEG AZ OTTHONI EGÉSZSÉGÜGYI
ELLÁTÓJÁT, HOGY AZ ELLENŐRIZHESSE, ÉS
ESETLEG MEGJAVÍTSA A KÉSZÜLÉKET. NE
HASZNÁLJA AZ OXIGÉNKONCENTRÁTORT, HA A
TÁPKÁBEL VAGY A CSATLAKOZÓJA MEGSÉRÜLT.

FIGYELMEZTETÉS: AZ AKKUMULÁTOR HELYTELEN
HASZNÁLATA MIATT FELHEVÜLHET VAGYMEGGYULLADHAT AZ AKKUMULÁTOR, ÉS SÚLYOS
SÉRÜLÉST OKOZHAT. NE LYUKASSZA KI, NE
ÜSSÖN RÁ, NE TAPOSSON RÁ, NE EJTSE LE AZAKKUMULÁTORT, ÉS MÁS ERŐBEHATÁSNAK VAGYRÁZKÓDÁSNAK SE TEGYE AZT KI. A KÁROSODOTTAKKUMULÁTOR HASZNÁLATA SZEMÉLYI SÉRÜLÉST
OKOZHAT.

FIGYELMEZTETÉS: AZ ÖVVEL VAGY EGYÉBMÓDON MEGFELELŐEN RÖGZÍTSE AZOXIGÉNKONCENTRÁTORT, HA GÉPJÁRMŰBENHASZNÁLJA, HOGY ELKERÜLJE A SÉRÜLÉST VAGYA KÉSZÜLÉK KÁROSODÁSÁT.
FIGYELMEZTETÉS: AZ AMERIKAI EGYESÜLT
ÁLLAMOK KÖZLEKEDÉSI MINISZTÉRIUMÁNAK
(DOT) ÉS AZ EGYESÜLT NEMZETEK (UN) ELŐÍRÁSAIMEGKÖVETELIK AZ AKKUMULÁTOR KIVÉTELÉT A
KÉSZÜLÉKBŐL REPÜLŐN TÖRTÉNŐ UTAZÁSKOR,HA AZ OXIGÉNKONCENTRÁTORT CSOMAGKÉNTADJÁK FEL. AZ OXIGÉNKONCENTRÁTOR
SZÁLLÍTÁSAKOR IS KI KELL VENNI AZ
AKKUMULÁTORT A KÉSZÜLÉKBŐL, ÉS
MEGFELELŐEN BE KELL CSOMAGOLNI.
FIGYELMEZTETÉS: NE HASZNÁLJA A BERENDEZÉSTADDIG, AMÍG EL NEM OLVASTA ÉS MEG NEM
ÉRTETTE EZT A KÉZIKÖNYVET. HA NEM ÉRTHETŐK
AZ ÖN SZÁMÁRA A FIGYELMEZTETÉSEK ÉSAZ UTASÍTÁSOK, FORDULJON A BERENDEZÉSFORGALMAZÓJÁHOZ, MIELŐTT MEGKÍSÉRELNÉHASZNÁLNI. ELLENKEZŐ ESETBENEGÉSZSÉGKÁROSODÁS VAGY A BERENDEZÉSKÁROSODÁSA KÖVETKEZHET BE.
FIGYELMEZTETÉS: HA ROSSZUL ÉRZI MAGÁT VAGYEGÉSZSÉGÜGYI VÉSZHELYZETBE KERÜL, AZONNALHÍVJON ORVOSI SEGÍTSÉGET.
FIGYELMEZTETÉS: HA A MŰSZAKI JELLEMZŐKNÉLMEGADOTT ÜZEMELTETÉSI HŐMÉRSÉKLETENKÍVÜL MŰKÖDTETI AZ OXIGÉNKONCENTRÁTORT,AKKOR A KONCENTRÁTOR CSAK KORLÁTOZOTTANTUD MEGFELELNI AZ OXIGÉNKONCENTRÁCIÓSJELLEMZŐKNEK. A HŐMÉRSÉKLETIHATÁRÉRTÉKEKET MEGTALÁLJA A KÉZIKÖNYVMŰSZAKI JELLEMZŐKRŐL SZÓLÓ RÉSZÉBEN.
FIGYELMEZTETÉS: NE HASZNÁLJON OLAJAT, ZSÍRT,PETRÓLEUMALAPÚ VAGY MÁS ÉGHETŐ TERMÉKETAZ OXIGÉNT SZÁLLÍTÓ TARTOZÉKOKHOZ VAGY AZOXIGÉNKONCENTRÁTORHOZ. CSAK OXIGÉNNELÖSSZEFÉRHETŐ, VÍZALAPÚ KRÉMEKET ÉSKENŐCSÖKET SZABAD HASZNÁLNI. AZ OXIGÉNFELGYORSÍTJA AZ ÉGHETŐ ANYAGOK ÉGÉSÉT.

472 - HUNPN MN232-C4 K | Felhasználói kézikönyv

FIGYELMEZTETÉS: FÉMTÁRGGYAL, PL. KULCCSALVAGY PÉNZÉRMÉVEL NE ZÁRJA RÖVIDRE AZ
AKKUMULÁTOR ÉRINTKEZŐIT. EZ SZIKRÁZÁST
OKOZHAT VAGY HŐ TERMELŐDÉSÉHEZ VEZETHET.
FIGYELMEZTETÉS: HA A HORDOZHATÓOXIGÉNKONCENTRÁTORT VÁLTÓÁRAMRÓLHASZNÁLJA A HÁZON KÍVÜL, AKKORA TÁPEGYSÉGET CSAK RÖVIDZÁRLATIMEGSZAKÍTÓVAL ELLÁTOTT CSATLAKOZÓBACSATLAKOZTASSA.
FIGYELMEZTETÉS: AZ OXIGÉNKONCENTRÁTOR
NEM HASZNÁLHATÓ MÁS BERENDEZÉS MELLETT,ALATT VAGY FÖLÖTT ELHELYEZVE. HA NEM
LEHET ELKERÜLNI, HOGY A KÉSZÜLÉK MÁS
BERENDEZÉS MELLETT, ALATT VAGY FELETTLEGYEN ELHELYEZVE, ELLENŐRIZNI KELL, HOGY
MEGFELELŐEN MŰKÖDIK-E.
FIGYELMEZTETÉS: A MEGADOTTAKTÓL ELTÉRŐ
KÁBELEK ÉS ADAPTEREK ALKALMAZÁSA – A
GYÓGYÁSZATI VILLAMOS KÉSZÜLÉK GYÁRTÓJA
ÁLTAL A BELSŐ ALKATRÉSZEK CSERÉJÉREÁRUSÍTOTT KÁBELEK ÉS ADAPTEREK KIVÉTELÉVEL– AZ OXIGÉNKONCENTRÁTOR MEGNÖVEKEDETTKIBOCSÁTÁSÁT ÉS CSÖKKENT ZAVARTŰRÉSÉTEREDMÉNYEZHETI.
FIGYELMEZTETÉS: CSAK A KÉSZÜLÉKREHELYEZETT, MŰSZAKI ADATOKAT TARTALMAZÓ
CÍMKÉN MEGHATÁROZOTT ELEKTROMOS
FESZÜLTSÉGET SZABAD HASZNÁLNI.
FIGYELMEZTETÉS: NE HASZNÁLJON AZ EGYSÉGHEZ
HOSSZABBÍTÓT, ÉS NE CSATLAKOZTASSON TÚL
SOK DUGÓT UGYANAHHOZ AZ ELEKTROMOSALJZATHOZ. A HOSSZABBÍTÓ HASZNÁLATA
KÁROSAN BEFOLYÁSOLHATJA A KÉSZÜLÉK
TELJESÍTMÉNYÉT. HA TÚL SOK FOGYASZTÓ
VAN CSATLAKOZTATVA EGY ALJZATHOZ, AZ
AZ ELEKTROMOS PANEL TÚLTERHELÉSÉTEREDMÉNYEZHETI, AMITŐL A MEGSZAKÍTÓ/
BIZTOSÍTÉK AKTIVÁLÓDHAT, ILLETVE TŰZ
KELETKEZHET, HA A MEGSZAKÍTÓ VAGY A
BIZTOSÍTÉK MEGHIBÁSODIK.

FreeStyle ComfortVIGYÁZAT: Ha riasztási esemény fordul
elő, vagy ha Ön úgy találja, hogy az
oxigénkoncentrátor nem működik
megfelelően, lapozza fel a jelen kézikönyv
Hibaelhárítás című fejezetét. Ha nem
tudja megoldani a problémát, forduljon aberendezés forgalmazójához.
VIGYÁZAT: A garancia érvényességének megőrzéseérdekében kövesse a gyártó minden utasítását.
FIGYELMEZTETÉS: NE KÍSÉRELJEN MEG
ELVÉGEZNI SEMMILYEN KARBANTARTÁSI
MUNKÁLATOT A JELEN KÉZIKÖNYVBENFELSOROLTAKON KÍVÜL. NE TÁVOLÍTSA
EL A BURKOLATOKAT; CSAK ABERENDEZÉS FORGALMAZÓJA VAGYEGY SZAKKÉPZETT SZERVIZTECHNIKUS
TÁVOLÍTHATJA EL A BURKOLATOKAT,
VAGY SZERVIZELHETI A KÉSZÜLÉKET.
FIGYELMEZTETÉS: CSAK A GYÁRTÓ ÁLTAL
AJÁNLOTT TARTOZÉKOKAT HASZNÁLJA. AZ EGYÉBTARTOZÉKOK HASZNÁLATA VESZÉLYES LEHET,KÁROSODHAT AZ OXIGÉNKONCENTRÁTOR, ÉS AGARANCIA ELVESZTÉSÉVEL JÁR.
FIGYELMEZTETÉS: A GYÁRTÓ AZT JAVASOLJA,HOGY TARTSON MAGÁNÁL KIEGÉSZÍTŐOXIGÉNFORRÁST ARRA AZ ESETRE, HAÁRAMSZÜNET LENNE, ILLETVE RIASZTÁSI
KÖRÜLMÉNY VAGY MŰSZAKI HIBA LÉPNE FEL.
ÉRDEKLŐDJÖN ORVOSÁNÁL VAGY A BERENDEZÉSFORGALMAZÓJÁNÁL, HOGY MILYEN TÍPUSÚTARTALÉKRENDSZERRE VAN SZÜKSÉGE.
FIGYELMEZTETÉS: NE HAGYJA, HOGYAKÁR A LÉGBESZÍVÓ, AKÁR A LÉGKIMENETINYÍLÁSOK TAKARVA LEGYENEK. NE EJTSENVAGY HELYEZZEN BE SEMMILYEN TÁRGYAT AKÉSZÜLÉK SEMELYIK NYÍLÁSÁBA. EZ UGYANISAZ OXIGÉNKONCENTRÁTOR TÚLHEVÜLÉSÉT ÉS ATELJESÍTMÉNYÉNEK ROMLÁSÁT OKOZHATJA.
FIGYELMEZTETÉS: AZ OXIGÉN HASZNÁLATAKÖZBENI DOHÁNYZÁS A TŰZZEL KAPCSOLATOSSÉRÜLÉSEKNEK ÉS HALÁLESETEKNEK AZ ELSŐSZÁMÚ KIVÁLTÓ OKA. AZ ALÁBBI BIZTONSÁGI
FIGYELMEZTETÉSEKET SZIGORÚAN BE KELLTARTANI:FIGYELMEZTETÉS: NE ENGEDJE MEG ADOHÁNYZÁST, GYERTYA VAGY NYÍLT LÁNGHASZNÁLATÁT ABBAN A HELYISÉGBEN, AHOLA KÉSZÜLÉKET VAGY OXIGÉNT SZÁLLÍTÓTARTOZÉKAIT HASZNÁLJA.

FIGYELMEZTETÉS: „TILOS A DOHÁNYZÁS –OXIGÉNHASZNÁLAT” FELIRATÚ TÁBLÁKATKELL JÓL LÁTHATÓ MÓDON ELHELYEZNIAZ OTTHONÁBAN, VAGY AHOL AZ OXIGÉNTHASZNÁLJA. A FELHASZNÁLÓKAT ÉSGONDOZÓIKAT TÁJÉKOZTATNI KELL, HOGYMILYEN VESZÉLYEKKEL JÁR AZ ORVOSI OXIGÉNJELENLÉTÉBEN VAGY ANNAK HASZNÁLATAKÖZBEN TÖRTÉNŐ DOHÁNYZÁS.
FIGYELMEZTETÉS: NE HASZNÁLJA AZOXIGÉNKONCENTRÁTORT GYÚLÉKONY GÁZOKJELENLÉTÉBEN. ELLENKEZŐ ESETBEN GYORSÉGÉS ALAKULHAT KI, AMI ANYAGI KÁROKAT,SZEMÉLYI SÉRÜLÉST VAGY HALÁLT IS OKOZHAT.
FIGYELMEZTETÉS: A KANÜL LEVÉTELE ÉSRUHÁRA, ÁGYNEMŰRE, KANAPÉRA VAGY MÁSPÁRNAANYAGRA HELYEZÉSE FELLOBBANÓ TÜZET
OKOZ, HA CIGARETTÁVAL, HŐFORRÁSSAL VAGYLÁNGGAL KERÜL ÉRINTKEZÉSBE.
FIGYELMEZTETÉS: NE HAGYJA AZ ORRKANÜLT ARUHÁZATON, ÁGYTAKARÓN VAGY SZÉKHUZATON,
ILLETVE AZOK ALATT. HA AZ EGYSÉGETBEKAPCSOLJA, DE NEM HASZNÁLJA, AKKOR AZOXIGÉN EZEKET AZ ANYAGOKAT GYÚLÉKONNYÁTESZI. ELLENŐRIZZE, HOGY KIKAPCSOLTA-E AZOXIGÉNKONCENTRÁTORT, HA NEM HASZNÁLJA.
FIGYELMEZTETÉS: AZ OXIGÉNKONCENTRÁTORT
KI KELL KAPCSOLNI ÉS KI KELL HÚZNI
A FALI CSATLAKOZÓBÓL, MIELŐTT AZ
AKKUMULÁTORCSOMAGOT KIVESZI.
FIGYELMEZTETÉS: A KIS ALKATRÉSZEKFULLADÁSVESZÉLYT OKOZHATNAK.
FIGYELMEZTETÉS: NE ÉRINTSE MEG A LÉGKIMENETINYÍLÁST 10 MÁSODPERCNÉL HOSSZABB IDEIG,
MERT MELEG LEHET.
FIGYELMEZTETÉS: A SZÉL VAGY AZ ERŐS HUZATKÁROSAN BEFOLYÁSOLHATJA A PONTOSOXIGÉNSZÁLLÍTÁST AZ OXIGÉNTERÁPIA SORÁN.
FIGYELMEZTETÉS: EBBŐL A TERMÉKBŐLKÉMIAI ANYAGOK – TÖBBEK KÖZÖTT
NIKKEL – OLDÓDHATNAK KI, AMI KALIFORNIAÁLLAMBAN ISMERTEN RÁKKELTŐ. TOVÁBBI
INFORMÁCIÓÉRT LÁTOGASSON EL A
WWW.P65WARNINGS.CA.GOV OLDALRA.

FIGYELMEZTETÉS: A DOHÁNYZÁS AZ OXIGÉNKANÜL
VISELÉSE KÖZBEN AZ ARC MEGÉGÉSÉTOKOZHATJA ÉS HALÁLHOZ VEZETHET.
FIGYELMEZTETÉS: HA DOHÁNYZIK, AZ ALÁBBI 3
LÉPÉSÉLETMENTŐ LEHET: KAPCSOLJA KI AZOXIGÉNKONCENTRÁTORT, VEGYE LE A KANÜLT, ÉS
HAGYJAEL A HELYISÉGET, AHOL A KÉSZÜLÉK TALÁLHATÓ.
PN MN232-C4 K | Felhasználói kézikönyv

HUN – 473

FreeStyle ComfortVIGYÁZAT: Először mindig a fali
csatlakozóról válassza le a tápegységet,mielőtt az oxigénkoncentrátorról választanále.
VIGYÁZAT: Ne tegye olyan helyre az
oxigénkoncentrátort, ahonnan nehéz elérni a tápkábelt.
VIGYÁZAT: Az oxigénszállító csöveket és a tápkábelt
mindig úgy helyezze el, hogy ne lehessen bennük
megbotlani, és ne tudjanak véletlenül rátekeredni a
beteg nyakára.
VIGYÁZAT: Ha autóban, hajóban, vagy más
egyenáramú áramforrásról használja az
oxigénkoncentrátort, akkor az oxigénkoncentrátor
bekapcsolása előtt indítsa be a járművet. Ha nem tartja
be ezt az utasítást, akkor lehet, hogy az áramforrás
nem biztosít energiát az oxigénkoncentrátornak.
VIGYÁZAT: Ha leállítja a gépjárművet, amelyben az
oxigénkoncentrátort használja, akkor húzza ki azáramból a készüléket és vegye ki a járműből. Ne tartsa
az oxigénkoncentrátort nagyon meleg vagy nagyon
hideg járműben, vagy más magas vagy alacsonyhőmérsékletű környezetben.
Megjegyzés: A hordozható és mobil RF távközlési
berendezések befolyásolhatják a gyógyászati
villamos készülékek működését.
VIGYÁZAT: NE hagyja az
oxigénkoncentrátort vagy a tápegységet ajárműre csatlakoztatva, ha a gyújtás KI van
kapcsolva. Ebben az esetben lemerülhet ajármű akkumulátora.
VIGYÁZAT: Az oxigénkoncentrátor működtetése
vagy tárolása a normál hőmérsékleti és környezeti
nyomás (magasság) tartományon kívül hatással lehet
a teljesítményre, és csökkentheti az akkumulátorüzemidejét, és/vagy megnőhet az akkumulátor
feltöltési ideje. A tárolási és üzemeltetési hőmérsékleti
határértékeket megtalálja a kézikönyv Műszaki
jellemzők című részében.
VIGYÁZAT: Az akkumulátorral felszerelt
oxigénkoncentrátorok esetében: Hűvös és száraz
helyen tárolja, hogy biztosítsa az akkumulátorélettartamát. Ha sokáig meleg helyen tárolja az
oxigénkoncentrátort, vagy teljesen feltöltött,illetve teljesen lemerült akkumulátorral tárolja, az
csökkentheti az akkumulátor élettartamát. Ne próbálja
meg felnyitni az akkumulátort; nincsenek bennejavítható alkatrészek. Az akkumulátort tartsa távol agyermekektől.
VIGYÁZAT: CSAK a gyártó által biztosított
akkumulátort HASZNÁLJA. Az akkumulátor helyesártalmatlanításához vegye fel a kapcsolatot aberendezés forgalmazójával, vagy a veszélyes
hulladékok ügyében illetékes helyi állami hivatallal.

474 - HUNPN MN232-C4 K | Felhasználói kézikönyv

Megjegyzés: Ellenőrizze, hogy a kanült teljesenbehelyezte és megfelelően rögzül-e. Ez biztosítja,hogy az oxigénkoncentrátor megfelelően tudjaérzékelni a belégzést az oxigénellátáshoz. Belégzés
alatt hallania és éreznie kell az oxigén áramlását azorrkanül ágaiban.
Megjegyzés: Ha az akkumulátor teljesen fel van töltve
vagy teljesen lemerült, és szélsőségesen meleghőmérséklet éri, akkor a lítium-akkumulátorok kapacitása maradandóan csökkenhet. Hosszú idejű tárolás
esetén az akkumulátorok 25–50%-os feltöltöttsége
és a –20 °C – 20 °C (–4 °F – 68 °F) hőmérsékleti
tartomány az ajánlott.
Megjegyzés: Bizonyos oxigénadagoló tartozékok
használata, amelyek nincsenek meghatározva azezzel az oxigénkoncentrátorral való használatra, ronthatják a készülék teljesítményét. A javasolt tartozékok
listája megtalálható ebben a kézikönyvben.
Megjegyzés: Ne működtesse az oxigénkoncentrátort
behelyezett légbeszívó szűrő nélkül! Ha van másikszűrője is, akkor a szennyezett szűrő megtisztítása
előtt helyezze be a pótszűrőt. Meleg, szappanos vízzel tisztítsa meg a szennyezett szűrőt, majd használatelőtt teljesen szárítsa meg.
Megjegyzés: A gyártó nem javasolja a berendezéssterilizálását.
Megjegyzés: Attól soha nem kell tartani, hogy az
oxigénkoncentrátor használata közben a helyiségben
elfogy az oxigén.
FIGYELMEZTETÉS: AZOXIGÉNKONCENTRÁTOR NEM SZABAD
MŰKÖDTETNI, VAGY KÜLSŐ (VÁLTÓVAGY EGYENÁRAMÚ) ÁRAMFORRÁSRACSATLAKOZTATNI, HA AZ AKKUMULÁTOR
NINCS BEHELYEZVE. NE PRÓBÁLJA
MEG BEKAPCSOLNI VAGY KÜLSŐ
ÁRAMFORRÁSRA CSATLAKOZTATNIAZ EGYSÉGET, HA KIVETTE BELŐLE AZAKKUMULÁTORCSOMAGOT.
CSAK AKKOR VEGYE KI AZAKKUMULÁTORCSOMAGOT, HA KIKAPCSOLTAAZ EGYSÉGET, ÉS KIHÚZTA AZT AZ ELEKTROMOSALJZATBÓL. NE PRÓBÁLJA MEG ÚGY KIVENNI AZ
AKKUMULÁTORCSOMAGOT, HOGY AZ EGYSÉGMÉG BE VAN KAPCSOLVA ÉS NINCS KIHÚZVA AZ
ELEKTROMOS ALJZATBÓL.

Megjegyzés: Ha az oxigénkoncentrátort hosszú időnát a fenti táblázatban látható normális üzemeltetési
körülményeken kívül eső feltételek között tárolták,
akkor bekapcsolás előtt meg kell várni, hogy elérje anormális üzemeltetési hőmérsékletet.

FreeStyle ComfortMűszaki jellemzők
FreeStyle Comfort
Méretek (egy akkumulátorral)

25,4 × 18,5 × 7,9 cm (10,0 × 7,3 × 3,1 in)

Méretek (kettős akkumulátorral)

27,9 × 18,5 × 7,9 cm (11,0 × 7,3 × 3,1 in)Tömeg (egyakkumulátorcsomaggal)

2,3 kg (5 lbs)

Tömeg (két akkumulátorcsomag
esetén)2,7 kg (6 lbs)

Nominális zajszint***

39,93 dB (A) ± 0,42 dB (A) @ 2Áramlás beállításai

Impulzusbeállítások 1, 2, 3, 4, 5

(20/perc légzésszámnál)

Kimeneti O2 beállításai

1. beállítás – 210 ml/perc
2. beállítás – 420 ml/perc
3. beállítás – 630 ml/perc
4. beállítás – 840 ml/perc
5. beállítás – 1050 ml/perc

Oxigénkoncentráció*90% (+5,5% / –3%)

Oxigénmonitor

Igen

Maximális kimeneti nyomás< 30 PSI

Alapértelmezett adagolásérzékenység> -0,1 cm H2O (beállítható)

Váltóáram

100–240 VAC, 50–60 Hz, 120W Max

Egyenáram

11–18 VDC (max. 10 A)

Elem típusa

lítium-ion

Az akkumulátor kapacitása

Egy cellás: 6700 mAh; Kettős cella: 13400 mAh

Akkumulátorfeltöltési idő

Egy akkumulátor esetén: 3,5 óra
Két akkumulátor esetén: 6,0 óra

Akkumulátorcsomag üzemidejeÜzemeltetési hőmérséklet**Egy akkumulátor üzemideje: 2. beállítás – legfeljebb 4 óraKettős akkumulátor üzemideje: 2. beállítás – legfeljebb 8 óra
5 °C – 40 °C (41 °F – 104 °F)

Szállítási és tárolási hőmérséklet
(egység)–25 °C – +70 °C (–13 °F – +158 °F) legfeljebb 3048 m (10 000 ft)
magasságig

Szállítási és tárolási hőmérséklet
(akkumulátor)–25 °C – +70 °C (–13 °F – +158 °F) legfeljebb 3048 m (10 000 ft)
magasságigÜzemeltetési magasság**–381–3048 m (–1250–10 000 láb) (700–1060 hPa-ra tesztelve)Üzemeltetési páratartalom

15–95% relatív páratartalom (nem lecsapódó)

Tárolási páratartalom

0 - 90% Nem kondenzáló

Korlátozott garancia

3 év az egységre
1 év az akkumulátorra és a tápegységekre* 101 kPa (14,7 psi) légköri nyomás alapján, 21 °C-on (70 °F)** Az egység minden környezetben működni fog, amíg a környezet a megadott hőmérsékleti és páratartalom-tartományban van.
A megadott üzemeltetési körülményeken kívül történő üzemeltetés esetén nagyobb áramlási sebesség mellett a koncentrátor csak
korlátozottan tud megfelelni az oxigénkoncentrációs jellemzőknek. A bemelegedési képernyő látható, amíg a koncentrátor el nem
éri a célkoncentrációt, vagy legfeljebb 15 percig. Használja tovább az egységet ezalatt az idő alatt.
*** Vizsgálati módszer szerint mért zajszint Nr. MDS-Szia.
A várt üzemi élettartam legalább öt év. A szűrőágy feltételezett felhasználhatósági időtartama kb. egy év, ha a koncentrátort nem
használja ezalatt.
PN MN232-C4 K | Felhasználói kézikönyv

HUN – 475

FreeStyle Comfort

Kezelési utasítások

1. Keressen egy megfelelő helyet a FreeStyle Comfort
használatához.
a. A koncentrátornak megfelelő légáramlásra
van szüksége, így a légbeszívó szűrőket és alégkimeneti nyílást nem szabad eltorlaszolni.
b. A koncentrátort úgy kell elhelyezni, hogy afény- és hangjelzéseket, valamint a riasztásokat
könnyen észlelhesse.
c. Mielőtt bekapcsolná az egységet, ellenőrizze,
hogy a légbeszívó szűrők a helyükön
vannak-e. Ha az egységből hiányzik egy szűrő,akkor vegye fel a kapcsolatot a berendezés
forgalmazójával.

3. Csatlakoztassa a váltó- vagy az egyenáramú
tápegységet az elülső oldal jobb alsó részén lévőelektromos aljzathoz.

2. Helyezzen be egy akkumulátorcsomagot akoncentrátorba.
a. Igazítsa az akkumulátort a FreeStyle Comfort
burkolata alsó részének középvonalához.

4. Csatlakoztassa az orrkanült az oxigénkimenethez.
Használat előtt ellenőrizze a csatlakoztatást.

MEGJEGYZÉS: Az ajánlottakon kívüli kanül(ök)
használata hatással lehet a teljesítményre.

Megjegyzés: A berendezés szolgáltatójának: A következő
oxigénbeviteli kiegészítők használata ajánlott az FreeStyle Comfort:b. Kattanásig csúsztassa befelé az akkumulátort,
hogy az biztosan rögzüljön a helyén.

476 - HUNPN MN232-C4 K | Felhasználói kézikönyv• Orrkanül 7,1 láb (2 m) csővel (maximális LPM: 6 LPM):
CAIRE cikkszám: 5408-SEQ• Az ajánlott tartozékokról további információk találhatók awww.caireinc.com webhelyen.

FreeStyle Comfort
5. A főkapcsolóval kapcsolja be a FreeStyle Comfort
készüléket, és válassza ki az előírt áramlási
sebességet az áramlási sebességet növelő (+) vagy
csökkentő (-) gombot használva.

autoSAT impulzustérfogat(ml)*beállítása
Légzésszám

1

2

3

4

15

10,5

20

10,5

5

21,0

31,5

42,0

52,5

21,0

31,5

42,0

52,5

25

10,5

21,0

31,5

42,0

42,0

30

10,5

21,0

31,5

35,0

35,0

35

10,5

21,0

30,0

30,0

30,0

40

10,5

21,0

26,3

26,3

26,3

*Tűréshatár: +/– 15%

Egy rövid hangjelzés jelzi, hogy a készüléket
bekapcsolta.

Megjegyzés: Amikor először kapcsolja be a FreeStyle
Comfort készüléket, az áramlási sebesség alapértelmezett beállítása 2. Ha a beállítást megváltoztatja,
akkor az egység bekapcsolásakor a beállított áramlási
sebesség lesz érvényes.
6. A kanül gyártójának utasításai szerint helyezze
az arcára az orrkanült, és lélegezzen normálisan.
Minden belégzéskor érezni fogja, ahogy egy adag
oxigén jut az orrába az orrkanülön át.
7. Ha meg akarja változtatni az áramlási sebességet,akkor nyomja meg az áramlási sebességet növelő (+)
vagy csökkentő (-) gombot, amíg el nem éri a kívánt
beállítást. Természetes, hogy az áramlási sebesség
megváltoztatásával a hang is változik.

nem-autoSAT impulzustérfogat (ml)*beállítása
Légzésszám

1

2

3

4

5

15

14,0

28,0

42,0

56,0

70,0

20

10,5

21,0

31,5

42,0

52,5

25

8,4

16,8

25,2

33,6

42,0

30

7,0

14,0

21,0

28,0

35,0

35

6,0

12,0

18,0

24,0

30,0

40

5,3

10,5

15,8

21,0

26,3

8. A főkapcsoló megnyomásával tudja kikapcsolni a
FreeStyle Comfort készüléket.

Megjegyzés: A kijelző készenléti állapotba vált, és
20 másodperc múlva elsötétül. Készenléti állapotbanbármelyik gombot megnyomva visszahozza a képernyőt a készenléti állapotból.
9. A speciális funkciókat lásd a Menü navigáció
utasításoknál

A TERMÉK SPECIÁLIS FUNKCIÓI
autoSAT®
A CAIRE szabadalmaztatott autoSAT funkciója
lehetővé teszi a FreeStyle Comfort számára, hogy
lélegzetenként egységes mennyiségű oxigént (bolus)
szállítson a maximális légzési sebességig. Mivela felhasználó légzési sebessége növekszik olyan
tevékenységek során, mint a séta, az autoSAT funkció
beállítja az eszköz működését a beállított bolus
hangerő fenntartása érdekében.
UltraSense®
Ez a szenzitív, impulzusadagolást elindító érzékelő
figyeli a légzés során bekövetkező nyomásváltozást, ésa belégzéskor adagolja az oxigénimpulzust.
autoDOSE™
Ha „nincs légzés riasztás” következne be, ez a funkció
adagolja az oxigént 20/perces légzésszámmal az
egység aktuális áramlási sebesség beállítása mellett,amíg a riasztási állapot meg nem szűnik.
PN MN232-C4 K | Felhasználói kézikönyv

HUN – 477

FreeStyle ComfortÁramellátás

A FreeStyle Comfort három módon kaphat áramot:• Tölthető akkumulátorcsomag• Váltóáram
• Egyenáram
Az egyes módokhoz kapcsolódó működtetésről akövetkező részekben kap tájékoztatást.
FIGYELMEZTETÉS: CSAK AZ EBBENA FELHASZNÁLÓI KÉZIKÖNYVBENLEÍRT ÁRAMELLÁTÁSI MÓDOKATHASZNÁLJA. A NEM CAIREGYÁRTMÁNYÚ TÁPEGYSÉGEKÉS AKKUMULÁTORCSOMAGOKHASZNÁLATA KÁROSÍTHATJA AZ
EGYSÉGET, ÉS HELYTELEN MŰKÖDÉST
EREDMÉNYEZHET.

AZ AKKUMULÁTOR ELTÁVOLÍTÁSA

Nyomja a FreeStyle Comfort készülékhez képest lefelé
az akkumulátor kioldógombját. Miközben benyomva
TARTJA ezt a gombot, csúsztassa ki az akkumulátort
úgy, hogy kihúzza azt a koncentrátorból.
FIGYELMEZTETÉS: AZOXIGÉNKONCENTRÁTORT KI KELL
KAPCSOLNI ÉS KI KELL HÚZNI AFALI CSATLAKOZÓBÓL, MIELŐTT AZ
AKKUMULÁTORCSOMAGOT KIVESZI.

AKKUMULÁTORCSOMAG

A FreeStyle Comfort működtethető a készülékhez
mellékelt tölthető akkumulátorcsomagról. A
koncentrátor működése nincs korlátozva, ha az
akkumulátorcsomagról működik. Minden áramlási
sebesség a rendelkezésre áll akkumulátoros áramellátás
esetén.

AZ AKKUMULÁTORCSOMAG
TÖLTÖTTSÉGÉNEK ELLENŐRZÉSEAZ AKKUMULÁTOR BEHELYEZÉSE

Igazítsa az akkumulátort a FreeStyle Comfort
burkolata alsó részének középvonalához. Kattanásig
nyomja befelé az akkumulátort, hogy az biztosanrögzüljön a helyén.

A FreeStyle Comfort készülékbe behelyezett
akkumulátor töltöttségét az akkumulátor ikon és abecsült hátralévő működési idő mutatja a kijelzőn.
80–100%60-80%

40-60%

20-40%

20%478 - HUNPN MN232-C4 K | Felhasználói kézikönyv

FreeStyle Comfort
Megjegyzés: Az akkumulátor töltése közben egy
dugasz jelzés jelenik meg a töltöttségi állapot
százalékos kijelzése helyett.
Ha az akkumulátor töltöttségét úgy kívánja ellenőrizni,
hogy az akkumulátor nincs benne a FreeStyle Comfort
készülékben, akkor helyezze azt vízszintes felületre, és
nyomja meg a TEST (Teszt) gombot. Az akkumulátor
töltöttségét jelző fények (25–100%) kigyulladnak
a TEST (Teszt) gomb jobb oldalán, mutatva az
akkumulátorcsomag töltöttségi szintjét.

ELSŐ TÖLTÉS

A FreeStyle Comfort készülékhez mellékelt új
akkumulátor csak részlegesen van feltöltve, amikora gyárból kiszállítják. Az első használat előtt váltóvagy egyenáramú áramforrásból töltse fel teljesen az
akkumulátorcsomagot.

ÁLTALÁNOS TÖLTÉSI TUDNIVALÓK

A FreeStyle Comfort készülék mindig tölti az
akkumulátorcsomagot, ha az be van helyezve
a koncentrátorba, és a koncentrátor külső váltó- vagy
egyenáramú áramforrásra van csatlakoztatva (kivévea repülőn).
Ha az akkumulátor teljesen fel van töltve, akkor
a kijelzőn a 100% jelzés látható.
Alternatívaként az akkumulátorcsomag az opcionális
asztali töltővel is tölthető.

AZ AKKUMULÁTOR MŰKÖDÉSI IDEJE

Ha lecsatlakoztatta a külső váltó- vagy egyenáramú
áramforrásról, akkor a koncentrátor automatikusan
az akkumulátor használatára vált, ha be van helyezve
egy feltöltött akkumulátor. A lenti táblázatban egy új
akkumulátorcsomag jellemző működési ideje látható.
Az akkumulátorcsomag működési ideje állandó,
függetlenül a légzésszámtól.
Impulzusbeállítás

Egyes(8 cellás)
akkumulátorműködési
idejeKettős(16 cellás)
akkumulátorműködési
ideje

1

legfeljebb
8 óra

legfeljebb
16 óra2

legfeljebb
4 óra

legfeljebb
8 óra

3

legfeljebb
3 óra

legfeljebb
6 óra

4

legfeljebb
2,25 óra

legfeljebb
4,5 óra5

legfeljebb
2 óra

legfeljebb
4 óra

AZ AKKUMULÁTOR JELLEMZŐ FELTÖLTÉSI
IDEJE
Az akkumulátorcsomag jellemző teljes feltöltési ideje
teljesen lemerült akkumulátor esetén 3,5 óra egyes
akkumulátornál, és 6,0 óra kettős akkumulátornál.

Megjegyzés: Az akkumulátorcsomag teljes lemerülése
esetén bizonyos esetekben az akkumulátorcsomagelőször nem kezd töltődni.

AZ AKKUMULÁTOR ÜZEMI ÉLETTARTAMA
A FreeStyle Comfort akkumulátorokat 300 feltöltési/
lemerülési ciklusra tervezték.

PN MN232-C4 K | Felhasználói kézikönyv

HUN – 479

FreeStyle Comfort

Váltóáramú és egyenáramú
tápegység
A FreeStyle Comfort készüléket mindig váltó- és
egyenáramú tápegységgel együtt szállítják.
A váltóáramú tápegység az alábbiakat tartalmazza:
a. Váltóáramú adapter, amely 100–240 volt
feszültségen és 50–60 Hz frekvencián működik.
b. Váltóáramú tápkábel, amely összeköti az
adaptert és a megfelelő fali aljzatot.

Az egyenáramú tápegység egy kábelt tartalmaz, amely
közvetlenül köti össze az egységet és az egyenáramú
aljzatot.

2. Csatlakoztassa a váltóáramú tápkábelt a váltóáramú
tápegységhez.

VÁLTÓÁRAM

Váltóáramot használnak az otthonokban, ésbárhol, ahol a szabványos váltóáramú áramellátás
rendelkezésre áll. A FreeStyle Comfort
teljes mértékben működőképes váltóáramú
áramellátás esetén. A FreeStyle Comfort tölti
is az akkumulátorcsomagot, ha váltóáramra
van csatlakoztatva (kivéve a repülőn). Az
akkumulátorcsomag töltése független attól, hogy a
FreeStyle Comfort be van-e kapcsolva vagy sem.
FIGYELMEZTETÉS: NE CSATLAKOZTASSA
KÜLSŐ ÁRAMFORRÁSHOZ, HA
AZ AKKUMULÁTORCSOMAG
NINCS BEHELYEZVE AZOXIGÉNKONCENTRÁTORBA.
A FreeStyle Comfort csatlakoztatása váltóáramú
áramforráshoz:
1. Csatlakoztassa a váltóáramú tápegységet a
FreeStyle Comfort elülső oldalának jobb alsó részén
található áramcsatlakozóhoz. Jól nyomja bele, hogy
biztosan csatlakoztassa.

480 - HUNPN MN232-C4 K | Felhasználói kézikönyv

3. A váltóáramú tápkábel másik végét csatlakoztassa afali vagy egyéb, megfelelő elektromos aljzathoz.

FreeStyle Comfort
4. A főkapcsolóval kapcsolja be a FreeStyle Comfort
készüléket, és válassza ki az előírt áramlási
sebességet az áramlási sebességet növelő (+) vagy
csökkentő (-) gombot használva.

3. Az egyenáramú tápkábel másik végét csatlakoztassa
az autó vagy más jármű egyenáramú aljzatához.

Megjegyzés: Ha tölteni kell az akkumulátorcsomagot,
akkor a kijelzőn az akkumulátor ikonon egyetlen
narancssárga sáv látható.

4. A főkapcsolóval kapcsolja be a FreeStyle Comfort
készüléket, és válassza ki az előírt áramlási
sebességet az áramlási sebességet növelő (+) vagy
csökkentő (-) gombot használva.

EGYENÁRAM

Megjegyzés: Ha tölteni kell az akkumulátorcsomagot,
akkor a kijelzőn az akkumulátor ikonon egyetlen
narancssárga sáv látható.

A készülék egyenárammal az autók, a hajók és egyéb
gépjárművek elektromos csatlakozóiról használható.
A FreeStyle Comfort teljes mértékben működőképes
egyenáramú áramellátás esetén. A FreeStyle Comfort
szükség esetén tölti is az akkumulátorcsomagot, haegyenáramot kap. Az akkumulátorcsomag töltése
független attól, hogy az egység be van-e kapcsolva
vagy sem.
FIGYELMEZTETÉS: NE CSATLAKOZTASSA
KÜLSŐ ÁRAMFORRÁSHOZ, HA
AZ AKKUMULÁTORCSOMAG
NINCS BEHELYEZVE AZOXIGÉNKONCENTRÁTORBA.
FIGYELMEZTETÉS: NE MŰKÖDTESSE A KÉSZÜLÉKETAZ AUTÓ 12 V-OS EGYENÁRAMÁRÓL ÚGY, HOGY A
JÁRMŰ MOTORJA NEM JÁR.
1. Indítsa be a gépjárművet.
2. Csatlakoztassa az egyenáramú tápegységet a
FreeStyle Comfort elülső oldalának jobb alsó részén
található áramcsatlakozóhoz. Jól nyomja bele, hogy
biztosan csatlakoztassa.

PN MN232-C4 K | Felhasználói kézikönyv

HUN – 481

FreeStyle Comfort
Tisztítás, ápolás és rendszeres
karbantartás
RENDSZERES KARBANTARTÁS

Megelőző karbantartás
A készülék nem igényel karbantartást. Az eszköz
belsejében lévő szűrőknek célja az eszköz élettartama.
Ha az eszköz nem működik megfelelően, az egységetvissza kell adni a szállítóhoz javítás céljából. A
FreeStyle Comfort házán belüli belső alkatrészek
szervizelését csak a készülék forgalmazója végezheti.
Az egység használója az alábbi karbantartásiműveletekért felelős.
• Hetente – a légbeszívó szűrők tisztítása.
FIGYELMEZTETÉS: NE PRÓBÁLJA MEG
FELNYITNI VAGY ELTÁVOLÍTANI AZOXIGÉNKONCENTRÁTOR MŰANYAGBURKOLATAIT. AZ EGYSÉGET CSAKSZAKKÉPZETT SZERVIZTECHNIKUS
NYITHATJA FEL.

LÉGBESZÍVÓ SZŰRŐK

A levegő a FreeStyle Comfort légbeszívó szűrőin
keresztül jut be az oxigénkoncentrátor elülső részén,alul. Ezek a szűrők akadályozzák meg, hogy a por
vagy más nagyobb részecskék az egységbe jussanak.
Hetente legalább egyszer meg kell tisztítani alégbeszívó szűrőket. A forgalmazó azt tanácsolhatja,
hogy az üzemeltetési körülményektől függően akár
gyakrabban is végezze el a tisztítást. A légbeszívószűrők helyes megtisztítása érdekében kövesse az
alábbi lépéseket.
1. A FreeStyle Comfort elülső oldalából távolítsa el a
két szűrőt úgy, hogy megnyomja a megfelelő füleket
és kihúzza a szűrőket a készülékből.

2. Mindkét szűrőt meleg, szappanos vízben mossa át.
3. Alaposan öblítse le a szűrőket, majd puha,szálmentes törlőkendővel törölje le a felesleges vizet.
Ellenőrizze, hogy a szűrők teljesen megszáradtak-e,mielőtt visszahelyezi azokat.
4. Helyezze vissza a tiszta, száraz szűrőket.
FIGYELMEZTETÉS: NE MŰKÖDTESSE AZOXIGÉNKONCENTRÁTORT BEHELYEZETTLÉGBESZÍVÓ SZŰRŐ NÉLKÜL!
482 - HUNPN MN232-C4 K | Felhasználói kézikönyv

AZ AKKUMULÁTORCSOMAG GONDOZÁSA
Ne hagyja, hogy nedvesség érje az
akkumulátorcsomagot!

Ne próbálja meg szétszerelni vagy kinyitni az
akkumulátorcsomagot!
Ne tegye ki az akkumulátort nyílt lángnak!
Ne ejtse le az akkumulátorcsomagot!
Tartsa gyermekektől távol az akkumulátorcsomagot!
Ha az akkumulátorcsomag sérültnek tűnik, akkor
használat előtt vegye fel a kapcsolatot a forgalmazóval.
Ne tegye az akkumulátort mikrohullámú sütőbe vagy
egyéb konyhai készülékbe, illetve ezek közelébe!

AZ AKKUMULÁTORCSOMAG TÁROLÁSA

Töltés előtt nem kell teljesen lemeríteni az
oxigénkoncentrátor akkumulátorát. Minden használat
után ajánlott feltölteni az oxigénkoncentrátor
akkumulátorát.
Ne tegye olyan helyre az akkumulátort, ahol nagyhőhatásnak van kitéve, pl. közvetlen napfényre vagy
egy autóba. A hő rontja az akkumulátor teljesítményét
és az élettartamát is. Hő hatására az akkumulátor is hőt
termelhet, füstölhet, kigyulladhat vagy felrobbanhat.
Az akkumulátorcsomagot állandó környezeti
körülmények között, alacsony páratartalomban(kevesebb, mint 70% relatív páratartalom), korrozív
gázoktól mentes helyen kell tárolni. A rozsdásodásmegelőzése érdekében kerülje a páralecsapódásra
hajlamosító körülményeket, például a környezetihőmérséklet gyors ingadozását.
Egy hónapos vagy annál hosszabb tároláshoz alemerült vagy 50% és 75% közötti (két vagy három
LED) részlegesen töltött állapot ajánlott. Ha feltöltötte,vegye ki az akkumulátort az oxigénkoncentrátorból.

FreeStyle Comfort
TISZTÍTÁS

Kapcsolja KI az egységet, és válassza le a váltóáramú
vagy egyenáramú feszültségről, mielőtt bármilyen
tisztításba vagy belekezdene. SEMMIT NE
permetezzen közvetlenül a külső burkolatra.
Használjon nedves (nem átitatott) kendőt vagy
szivacsot. A törlőkendőre vagy szivacsra permetezzen
enyhe hatású háztartási tisztítószer oldatot, hogy
megtisztítsa a burkolatot és a tápegységet. Folytassa atisztítást a tisztítószer gyártójának utasításai alapján.
FIGYELMEZTETÉS: AZ EGYSÉG
TISZTÍTÁSA ELŐTT KAPCSOLJAKI AZT, ÉS A TÁPKÁBELT HÚZZAKI AZ ELEKTROMOS ALJZATBÓL,HOGY ELKERÜLJE AZ ÁRAMÜTÉST
VAGY AZ ÉGÉSI SÉRÜLÉST. CSAK ABERENDEZÉS FORGALMAZÓJA VAGYEGY SZAKKÉPZETT SZERVIZTECHNIKUS
TÁVOLÍTHATJA EL A BURKOLATOKAT,
VAGY SZERVIZELHETI AZ EGYSÉGET.
FIGYELMEZTETÉS: ÓVATOSAN KELL ELJÁRNI,HOGY AZ OXIGÉNKONCENTRÁTORT ÉS AZAKKUMULÁTORCSOMAGOT NE ÉRJE NEDVESSÉG,
VALAMINT NE KERÜLJÖN FOLYADÉK AZ EGYSÉGBE.
EZ HIBÁS MŰKÖDÉST VAGY ÜZEMSZÜNETET
EREDMÉNYEZHET, ÉS MEGNŐ AZ ÁRAMÜTÉS VAGYAZ ÉGÉSI SÉRÜLÉS KOCKÁZATA.
FIGYELMEZTETÉS: NE HASZNÁLJON FOLYADÉKOTKÖZVETLENÜL AZ EGYSÉGEN! A MELLŐZENDŐVEGYI ANYAGOK LISTÁJA TÖBBEK KÖZÖTT, DE NEMKIZÁRÓLAGOSAN AZ ALÁBBIAKAT TARTALMAZZA:ALKOHOL ÉS ALKOHOL ALAPÚ TERMÉKEK,KONCENTRÁLT KLÓR ALAPÚ TERMÉKEK(ETILÉN-KLORID), ÉS OLAJ ALAPÚ TERMÉKEK.
EZEK NEM HASZNÁLHATÓAK A KONCENTRÁTOR
MŰANYAGBURKOLATÁNAK TISZTÍTÁSÁRA, MIVELKÁROSÍTHATJÁK AZ EGYSÉGET.

FIGYELMEZTETÉS: A KÉSZÜLÉK HÁZÁNAK,KEZELŐPULTJÁNAK ÉS TÁPKÁBELÉNEKTISZTÍTÁSÁRA CSAK ENYHE HÁZTARTÁSI
TISZTÍTÓSZERREL BENEDVESÍTETT (NEMÁTITATOTT) KENDŐT VAGY SZIVACSOTHASZNÁLJON, MAJD TÖRÖLJÖN SZÁRAZRA MINDENFELÜLETET. ÜGYELJEN ARRA, NEHOGY FOLYADÉKKERÜLJÖN A KONCENTRÁTORBA. FORDÍTSON
KÜLÖNÖS FIGYELMET A KANÜLHÖZ VEZETŐOXIGÉNKIMENETI CSATLAKOZÓRA, HOGY AZMENTES MARADJON MINDEN PORTÓL, VÍZTŐL ÉS
SZEMCSÉTŐL.
FIGYELMEZTETÉS: EGY NEDVES (NEMÁTITATOTT) TÖRLŐKENDŐT VAGY SZIVACSOTHASZNÁLJON AZ AKKUMULÁTORCSOMAG
MEGTISZTÍTÁSÁHOZ. ELŐSZÖR PERMETEZZENHÁZTARTÁSI TISZTÍTÓSZERT A TÖRLŐKENDŐRE
VAGY A SZIVACSRA, ÉS EZZEL TISZTÍTSA MEG AZ
AKKUMULÁTORCSOMAG BURKOLATÁT.
FIGYELMEZTETÉS: NE TISZTÍTSA AUTOMATAMOSÓBERENDEZÉSBEN!

A KANÜL CSERÉJE

Mindig kövesse a kanül gyártójának a helyes
használatra vonatkozó utasításait. 3–6 havonta
cserélje a kanült és az oxigénvezetéket (napi használat
esetén 3 havonta) a gyártó ajánlásai szerint, a kanül
gyártójának vagy a készülék forgalmazójának az
ajánlásai szerint. Az orvos vagy az oxigénkoncentrátor
forgalmazója ellátja Önt a tisztításra és a cserérevonatkozó utasításokkal.
A cseréhez szükséges plusz kellékek beszerezhetők az
oxigénkoncentrátor forgalmazójától.

PN MN232-C4 K | Felhasználói kézikönyv

HUN – 483

FreeStyle Comfort

Riasztási körülmények
Fontossági
sorrend

Riasztásleírása1

Kijelző

Hangjelzés

További részletek/Javasolt művelet

Rendszermeghibásodás,
akkumulátor
magas hőmérséklet

1 sípolás
20 másodpercenként

Az egység riaszt, ha az akkumulátorhőmérséklete megközelíti abiztonsági lekapcsolás határértékét.
Ha eléri a lekapcsolási határértéket,az egység kikapcsol. Ha külső
áramforrásra van csatlakoztatva,
a riasztási képernyő jelenik meg,és a riasztási hangjelzés 2 percig
hallható.
Szobahőmérsékletű akkumulátorraltesztelje. Ha a probléma továbbra isfennáll, forduljon a forgalmazóhoz.

2

Rendszermeghibásodás,magashőmérséklet

1 sípolás
20 másodpercenként

Az egység riaszt, ha a belső
hőmérséklet megközelíti alekapcsolás határértékét.
Ha eléri a lekapcsolási határértéket,az egység kikapcsol, megjelenik ariasztási képernyő, és a riasztási
hangjelzés 2 percig hallható.
Ellenőrizze, hogy semmi nem zárja
el az egység légkimeneti nyílásait.
Ha a probléma továbbra is fennáll,
forduljon a forgalmazóhoz.

3

Rendszermeghibásodás,magas nyomás

1 sípolás
20 másodpercenként

Ha eléri a lekapcsolási határértéket,az egység kikapcsol, megjelenik ariasztási képernyő, és a riasztási
hangjelzés 2 percig hallható.
Ha a probléma továbbra is fennáll,
forduljon a forgalmazóhoz.

4

Nem észlelhető
légzés1 sípolás
30 másodpercenként

15 másodperce nem észlel légzést.
Az első légzés, vagy a bemelegedési
képernyő lekapcsolása aktiválja,amelyik előbb következik be.
Ellenőrizze, hogy a kanül nem
tört-e meg, vagy nincs-e rajta
repedés. Állítsa nagyobb fokozatra
az érzékenységet. Ha a probléma
továbbra is fennáll, forduljon aforgalmazóhoz.

5

Akkumulátor
lemerül,lekapcsolás

1 sípolás
30 másodpercenként

Riasztás, amíg az akkumulátor lenem merül és az egység ki nem
kapcsol.
Töltse az akkumulátort.

6

Bemelegedés

484 - HUNPN MN232-C4 K | Felhasználói kézikönyv

nincs

Az egység megkezdi az
automatikus adagolást, amíg
légzést nem észlel, vagy abemelegedés véget nem ér, amelyikelőbb következik be.

FreeStyle Comfort
Fontossági
sorrend

Riasztás leírása

7

Kijelző

Hangjelzés

További részletek

Alacsony
koncentráció

1 sípolás
30 másodpercenként,
legfeljebb
5 percig

Ha a probléma továbbra
is fennáll, forduljon aforgalmazóhoz.

8

Rendszermeghibásodás,áramlási sebesség (OCD szelep
meghibásodása,
légvezeték
eltömődése)

1 sípolás
30 másodpercenként

Ha a probléma továbbra
is fennáll, forduljon aforgalmazóhoz.

9

Akkumulátor
lemerül
figyelmeztetés

1 sípolás,
nincs ismétlés

Töltse az akkumulátort.

10

Légzésszámtúllépés

1 sípolás
60 másodpercenként

A légzésszám meghaladja a 40/
perc értéket.
Ha az adagolást téves növelése
történt, akkor csökkentse azérzékenységet. Ha a probléma
továbbra is fennáll, forduljon aforgalmazóhoz.

11

Rendszermeghibásodás,
ventilátorhiba

nincs

Ha a probléma továbbra
is fennáll, forduljon aforgalmazóhoz.

12

Rendszermeghibásodás,
alacsony nyomás

nincs

Ha a probléma továbbra
is fennáll, forduljon aforgalmazóhoz.

13

Rendszermeghibásodás,
alacsonyhőmérséklet

nincs

Várja meg, amíg az egység
szobahőmérsékletremelegszik. Ha a probléma
továbbra is fennáll, forduljon aforgalmazóhoz.

Tájékoztató

Indítás

1 tájékoztató
sípolás

A képernyő megjelenik
bekapcsoláskor, és megszólal
egy sípolás, amely jelzi, hogy ahangszóró működik.
Az indulási képernyőn megjelenika firmware aktuális verziója ajobb alsó sarokban.
PN MN232-C4 K | Felhasználói kézikönyv

HUN – 485

FreeStyle Comfort
Fontossági
sorrend

Riasztás leírása

Tájékoztató

Normál működési
képernyő autoSATüzemmódban
(nincsegyenáramú
tápellátás
csatlakoztatva)

nincs

Tájékoztató

Normál működési
képernyő
autoSATüzemmód
letiltva (nincs
egyenáramú
tápellátás
csatlakoztatva)

Nincs

Tájékoztató

Beállítás
megváltoztatva

1 tájékoztató
sípolás

Tájékoztató

Kikapcsoltállapotban külső
áramforrásra
csatlakozott

nincs

TájékoztatóMűködés közben
csatlakoztatjakülsőáramforrásra

nincs

Tájékoztató

Kikapcsoltállapotban
leválasztja
a külső
áramforrásról

1 tájékoztató
sípolás

TájékoztatóMűködés közben
leválasztja
a külső
áramforrásról
(az akkumulátor
behelyezve)

1 tájékoztató
sípolás

486 - HUN

Kijelző

PN MN232-C4 K | Felhasználói kézikönyv

Hangjelzés

További részletek

2 másodpercre megjelenika CAIRE képernyő, amit
5 másodpercig az akkumulátor
és a csatlakozó állapota követ,mielőtt kikapcsol a képernyő.

Megjelenik a CAIRE képernyő
2 másodpercre, majd a kijelző
kikapcsol.

FreeStyle Comfort
Fontossági
sorrend

Riasztásleírása

Tájékoztató

Kijelző

Hangjelzés

További részletek

Készenléti
módban – haa fel vagy
le gombot
megnyomja,
a képernyőn
megjelenik az
akkumulátor
töltöttsége
és a töltés
jelzésenincs

A képernyő 5 másodpercre
megjelenik, majd lekapcsol.

Tájékoztató

Repülőgép
üzemmód
akkumulátorrólműködve

1 hosszúsípolás

Tájékoztató

Repülőgép
üzemmód
hálózatra
csatlakoztatva

1 hosszúsípolás

PN MN232-C4 K | Felhasználói kézikönyv

HUN – 487

FreeStyle Comfort

Menünavigációs utasítások
Készenléti vagy Futtatási üzemmódban tud a menübe
belépni. Ha Futtatási üzemmódban riasztási állapot
következik be, akkor automatikusan kilép a menüből.
Ha 20 másodpercig semmilyen gombot sem nyom
meg, akkor automatikusan kilép a menüből. Készenlétiüzemmódban lehet, hogy kétszer is meg kell nyomnia
a Menü gombot, hogy felébressze az egységet, és betudjon lépni a menübe.
Négy menüképernyő van:
1. Állapot képernyő: megjelenik a firmware-verzió
száma, a futtatási idő és az akkumulátor lemerülési
ciklusainak a száma

Repülőgép üzemmód

2. Repülőgép üzemmód (KI/BE)
3. Érzékenység: beállítható (1-5)
4. autoSAT (KI/BE)

ÉrzékenységÁllapot képernyő példaBEÁLLÍTÁSOK MENÜ MŰKÖDÉSE
A Beállítások menübe a Menu/Select (Menü/
Kiválasztás) gombmegnyomásával tud lépni
Futtatási és Készenléti módban.
A menük végiggörgetéséhez nyomja meg agombot.

Állapot lap

488 - HUNPN MN232-C4 K | Felhasználói kézikönyv

autoSATFreeStyle ComfortÁLLAPOT LAP
A Freestyle Comfort állapotoldalán látható afirmware verziója, az üzemórák száma és az
akkumulátor ciklusainak száma a csatlakoztatottakkumulátoron.

A Repülőgép üzemmód megerősítéséhez nyomja meg
agombot, ha a képernyőn a SAVE (MENTÉS)
ikon látható. A kijelzőn látható szó fekete „ON” (BE)szóra vált.
A Repülőgép üzemmód törléséhez nyomja meg agombot, ha a képernyőn a CANCEL (TÖRLÉS) ikon
látható.

Állapot lap

A futási órákat teljes órákban mutatjuk be.
Az akkumulátor lemerülése: a csatlakoztatott akkumulátor lemerülési és újratöltési ciklusainak száma.

REPÜLŐGÉP ÜZEMMÓD
A FreeStyle Comfort kikapcsolt Repülőgép
üzemmóddal kerül kiszállításra.

Repülőgép üzemmód BE megerősítve

A Repülőgép mód kikapcsolásához ismételje meg a
fentieket, míg az OFF (KI) szóra vált.
A
gombot megnyomva, miközben a képernyőn a„Next” (Következő) szimbólum látható, a menü oldalt
a következő képernyő opcióra váltja.
A Repülőgép üzemmód Futtatás üzemmódban
való használatra szolgál. A Repülőgép üzemmód
automatikusan letiltásra kerül, ha az egység
akkumulátorról működik, és Készenléti módban
hagyta.

Megjegyzés: Lásd a Repülőgép üzemmód aktiválása
részt, ha többet kíván megtudni a Repülőgép
üzemmódról.
Repülőgép üzemmód főképernyője

Az üzemmód bekapcsolásához nyomja meg agombot, ha a képernyőn a CHANGE (MÓDOSÍTÁS)ikon látható. A kijelzőn látható szó kék „ON” (BE)szóra vált.

ÉRZÉKENYSÉG MÓD
A FreeStyle Comfort készüléket a 2-as érzékenység
gyári beállítással szállítják, és normál körülmények
között ezt nem szükséges módosítani. Az
érzékenységet 1 és 5 között lehet beállítani, ahol az
1-es érték a legkisebb, az 5-ös pedig a legnagyobbérzékenységet jelöli.

Repülőgép üzemmód BE kiválasztva
Érzékenység mód képernyője 2-es beállítással

Az érzékenység beállításához nyomja meg agombot, ha a képernyőn a CHANGE (MÓDOSÍTÁS)ikon látható. A kijelzőn látható érték a következő kék
számra vált.
PN MN232-C4 K | Felhasználói kézikönyv

HUN – 489

FreeStyle Comfort

3-as érzékenységi mód kiválasztva

A 3. beállítás megerősítéséhez az érzékenység módban
nyomja meg agombot, ha a képernyőn a SAVE
(MENTÉS) ikon látható. A kijelzőn látható érték fekete„3”-ra vált.

autoSAT üzemmód főképernyője

Az üzemmód kikapcsolásához nyomja meg agombot, ha a képernyőn a CHANGE (MÓDOSÍTÁS)ikon látható. A kijelzőn látható szó kék „OFF” (KI)szóra vált.

Az érzékenység törléséhez nyomja meg agombot,ha a képernyőn a CANCEL (TÖRLÉS) ikon látható.

autoSAT üzemmód Be kiválasztva

3-as érzékenységi mód megerősítve

Az autoSAT üzemmód kikapcsolásának
megerősítéséhez nyomja meg agombot, ha aképernyőn a SAVE (MENTÉS) ikon látható. A kijelzőn
látható szó fekete „OFF” (KI) szóra vált.
Az autoSAT üzemmód bekapcsolásának törléséhez
nyomja meg agombot, ha a képernyőn a
CANCEL (TÖRLÉS) ikon látható.

5-ös (maximális) érzékenységi mód megerősítveA
gombot megnyomva, miközben a képernyőn a„Next” (Következő) szimbólum látható, a menü oldalt
a következő képernyő opcióra váltja.

autoSAT mód Ki megerősítve

AUTOSAT ÜZEMMÓD

Az autoSAT mód bekapcsolásához ismételje meg a
fentieket, míg az ON (BE) szóra vált.

A FreeStyle Comfort bekapcsolt autoSAT üzemmóddal
kerül kiszállításra.

A
gombot megnyomva, miközben a képernyőn a„Next” (Következő) szimbólum látható, a menü oldalt
a következő képernyő opcióra váltja.

490 - HUNPN MN232-C4 K | Felhasználói kézikönyv

FreeStyle Comfort
Hibaelhárítás
Probléma

Lehetséges ok

Megoldás

A FreeStyle Comfort
készüléken megjelenik egy
riasztás és szaggatott sípolás
hallatszik.

Tekintse meg a riasztási
körülmények táblázatát.

Tekintse meg a riasztási körülmények táblázatát.

Az oxigénkoncentrátor nem
kapcsolható be.

Az akkumulátor lemerült.

A váltóáramú vagy az egyenáramú aljzaton
keresztül lássa el az egységet árammal.

Nincs csatlakoztatva a külső
áramforrásra.
Általános meghibásodás.

Ellenőrizze, hogy a külső csatlakozók megfelelően
csatlakoznak-e.
Végezzen teljes ki- és bekapcsolási ciklust úgy,hogy kiveszi az akkumulátort, és leválasztja aváltó- vagy egyenáramú külső áramforrásról.
Ezután helyezze vissza az akkumulátort, és aki- és bekapcsolási ciklus kész. Szükség esetén
csatlakoztassa vissza a külső áramforráshoz.
Forduljon a berendezés forgalmazójához, ésszükség esetén váltson másik oxigénforrásra.

Minden egyéb probléma.

Végezzen teljes ki- és bekapcsolási ciklust úgy,hogy kiveszi az akkumulátort, és leválasztja aváltó- vagy egyenáramú külső áramforrásról.
Ezután helyezze vissza az akkumulátort, és aki- és bekapcsolási ciklus kész. Szükség esetén
csatlakoztassa vissza a külső áramforráshoz.
Váltson másik oxigénforrásra, ha van, és vegye fel
a kapcsolatot a berendezés forgalmazójával.

Megjegyzés: Egyes riasztások miatt az egység leáll.
Megjegyzés: Teljes áramkimaradás esetén a riasztási rendszer alaphelyzetbe áll (minden korábbi riasztás törlődik), és
csak az a riasztás jelenik meg, amelynek a feltételei még az áramellátás helyreállása után is fennállnak.

PN MN232-C4 K | Felhasználói kézikönyv

HUN – 491

FreeStyle Comfort

Utazási tudnivalók

A FreeStyle Comfort megfelel a kereskedelmi repülők
fedélzetén történő használattal kapcsolatos FAA
irányelveknek. Minden utazás előtt (repülőn, vonaton,
hajón, stb.) az alábbiakat kell elvégeznie:• Ellenőrizni, hogy a koncentrátor megfelelően
működik-e.
• Az indulás előtt legalább 48 órával vegye fel akapcsolatot az utasszállítóval, és tájékoztassa arról,
hogy FreeStyle Comfort készülékkel fog utazni.
Ajánlott midig tartalék akkumulátort vinni az útra.
Légi utazásnál a földi tennivalókra (repülés előtti
becsekkolás, biztonsági ellenőrzés, stb.) és azátszállásokra is legyen elég az akkumulátor töltöttsége.
Sok légitársaság előírja, hogy az utazás becsült
maximális időtartamának (repülési idő, repülés előtti
és utáni földi tennivalók, csatlakozások, és váratlan
késések) 150%-ára legyen elegendő számú, teljesen
feltöltött akkumulátorcsomag Önnél.

be a POC-t úgy, hogy az nincs külső áramforrásra
csatlakoztatva.

2. Engedélyezze a Repülőgép üzemmódot az egyik
alábbi módszert használva:a. A Repülőgép üzemmód engedélyezéséhez
kövesse a Menünavigációs utasítások részt.

Megjegyzés: Repülés során a tartalék akkumulátorokat
biztonságos hordtáskában ajánlott tartani.

FAA-SZABÁLYOZÁS

A tartalék akkumulátoroknál külön-külön kellmegakadályozni a rövidzárlat kialakulását úgy,hogy az eredeti gyári csomagolásba rakja vissza
azokat, vagy más módon leszigeteli az érintkezőket,pl. szigetelőszalaggal leragasztja a szabadon lévőérintkezőket, vagy külön nejlonzacskóba vagyvédőtasakba helyezi mindegyik akkumulátort, és csaka kézi poggyászban viszi magával.

b. Tartsa lenyomva mind a Fel, mind a Le gombot
5 másodpercig, amíg egy hosszú sípolást nem hall,és meg nem jelenik a Repülőgép szimbólum.

REPÜLŐGÉP-ÜZEMMÓD AKTIVÁLÁSA
Akkumulátoros áramellátás
A POC a normál működési eljárások szerint
használható repülőgépen CSAK akkumulátorról úgy,hogy nincs külső áramforrásra csatlakoztatva.

3. Ha a repülőgép szimbólum megjelent, akkor
csatlakoztassa a POC-t az aljzathoz.

Külső áramforrásról történő áramellátás
Ha a repülőgépen van elektromos csatlakozóaljzat,
ami működőképes, és megfelelő helyen is van a
POC áramforrásaként, akkor az használható a POC
akkumulátorok biztonsági tartalékaként. Ebben az
esetben az alábbi eljárás végrehajtása a javasolt, hogy
az akkumulátorról a repülőgép áramforrására váltson:
1. CSAK az akkumulátor áramát használva kapcsolja

492 - HUNPN MN232-C4 K | Felhasználói kézikönyv

Megjegyzés: A repülőgépre vonatkozó korlátozás miatt az
akkumulátorok nem töltődnek Repülőgép-üzemmódban.

FreeStyle Comfort
Tartozékok
A megfelelő teljesítmény és a biztonság érdekében
kizárólag az oxigénkoncentrátor forgalmazóján
keresztül a CAIRE által szállított, itt felsorolt
tartozékokat használja. Az alább felsoroltaktól eltérő
tartozékok használata károsan befolyásolhatja akoncentrátor teljesítményét és/vagy a biztonságot.

AZ EGYSÉG BEHELYEZÉSE A
HORDTÁSKÁBA
1. Az egységet a táska aljára helyezze úgy, hogy aszűrők és az akkumulátor nyílásai a megfelelő
helyzetben legyenek.

A FreeStyle Comfort szokásos tartozékai
Hordtáska

MI396-1

Hátitáska

MI459-1

VállszíjMI397-1

Váltóáramú tápegység (mindenkábellel együtt)

PW036-1S

Egyenáramú tápkábel

CD041-1

Durva részecskék külső szűrője (jobb
oldali)

FI226-1

Durva részecskék külső szűrője (baloldali)

FI227-1

2. A táska alján húzza össze a cipzárt.

Opcionálisan rendelhető tartozékok
Kiegészítő 8 cellás
akkumulátorcsomag

BT034-1S

16 cellás akkumulátorcsomag

BT035-1S

Asztali akkumulátorcsomag-töltő

BT036-1

Tartozék hordtáska

MI372-2

Megjegyzés: További opciók lehetnek elérhetők az
országspecifikus tápkábelekhez, ahol fent jelölve van.
Forduljon a CAIRE vállalathoz vagy az oxigénkoncentrátor
forgalmazójához, ha másik változatot kell rendelnie.

A hordtáskára vonatkozó utasítások

A VÁLLSZÍJ FELHELYEZÉSE
1. A vállszíjat közvetlenül az egységre vagy ahordtáskára is felhelyezheti.
2. Csatolja a szíjat az egységen vagy a táskán lévőgyűrűre.

A FreeStyle Comfort készülékének a hordtáskába
való behelyezéséhez és rögzítéséhez kövesse az alábbi
utasításokat.

PN MN232-C4 K | Felhasználói kézikönyv

HUN – 493

FreeStyle ComfortÁrtalmatlanítás és újrafelhasználás
Minden esetben juttassa el a FreeStyle Comfort
készüléket az összes alkatrészével együtt az felszerelés
egészségügyi ellátójához a megfelelő ártalmatlanítás
céljából. Az akkumulátor megfelelő ártalmatlanítására
vonatkozó utasításokkal kapcsolatban a helyiönkormányzattal is kapcsolatba léphet. Az egységetújra felhasználhatják emberek között, de vissza kelladni az felszerelés szolgáltatónak újrafeldolgozás
céljából.

szóló irányelv (WEEE) szerint egy újrahasznosító
létesítménybe kell juttatnia.
A termékeink megfelelnek a veszélyes anyagokkorlátozására (RoHS) vonatkozó irányelvnek.
Legfeljebb nyomokban tartalmazhatnak ólmot vagy
egyéb veszélyes anyagot.

WEEE és RoHS irányelvek
Ez a szimbólum a berendezés tulajdonosátarra figyelmezteti, hogy a készüléket
annak élettartama végén az elektromos éselektronikus készülékek hulladékainak kezeléséről

VIGYÁZAT: A helyes ártalmatlanításhoz
vegye fel a kapcsolatot a berendezés
forgalmazójával, vagy a helyi állami
hivatallal az ártalmatlanításra vonatkozó
utasításokért.

A működtetővel közvetlenül érintkező anyagok
Koncentrátor burkolata................................................................................................................................ PC/ABS
Koncentrátor vezérlőpanelje.......................................................... EBG7 poliészter vagy ezzel egyenértékű anyag
Koncentrátor fogantyúja....................................................................................................................... Polikarbonát
Oxigénkimenet, csőcsatlakozás............................................................................................................ Polikarbonát
Egység címkéje................................................................................................................................................Lexan
Kábelcsatlakozók............................................................................................................... Polikarbonát/vinil-klorid
Tápkábel.....................................................................................................................................................PVC, fém
Tápegység.........................................................................................................................Lexan 940 (polikarbonát)
Akkumulátorcsomag.................................................................................................................................... PC/ABS
Akkumulátorcsomag és tápegység címkéi......................................................................................... Poliészter-film
Koncentrátor hordtáska..............................................................................................Cordura, neoprén, nejlonbélés
Szíj....................................................................................................... Polipropilén, acetil, vinil, Cordura, neoprén

EMC-tesztÚtmutató és gyártói nyilatkozat – Elektromágneses kibocsátás
A FreeStyle Comfort készülék az alábbiakban megadott elektromágneses környezetben való használatra készült. A
FreeStyle Comfort készülék vásárlójának vagy felhasználójának kell biztosítania, hogy azt ilyen környezetben használják.
Kibocsátási teszt
RF-kibocsátás
CISPR 11
RF-kibocsátás
CISPR 11
Harmonikus kibocsátások
IEC 61000-3-2
Feszültségingadozások/feszültségrezgések
IEC 61000-3-3

494 - HUN

Megfelelőség

Elektromágneses környezet – útmutatás

1. csoport

A FreeStyle Comfort csak a belső működéséhez használ RF energiát.
Ezért a rádiófrekvenciás kibocsátása nagyon alacsony, és nem
valószínű, hogy interferenciát okoz a közelében lévő elektronikus
berendezésekben.

B. osztály
A. osztály
Megfelel

PN MN232-C4 K | Felhasználói kézikönyv

A FreeStyle Comfort minden létesítményben használható, beleértve
a lakóépületeket, valamint a lakossági célokat szolgáló épületeket
ellátó alacsony feszültségű, nyilvános hálózatra közvetlenül csatlakozóépületeket is.

FreeStyle ComfortÚtmutató és gyártói nyilatkozat – Elektromágneses zavartűrés

A FreeStyle Comfort készülék az alábbiakban megadott elektromágneses környezetben való használatra készült. A
FreeStyle Comfort készülék vásárlójának vagy felhasználójának kell biztosítania, hogy azt ilyen környezetben használják.
Zavartűrési
IEC 60601-es mérőszint Megfelelőségi szint
Elektromágneses környezet – útmutatás
vizsgálat
Elektromágneses
A padlóburkolat fa, beton vagy kerámia±8 kV érintkezéses±8 kV érintkezéses
környezet –járólap legyen. Ha a padló műanyaggal vanútmutatásburkolva, a relatív páratartalom legalább±15 kV levegőn át±15 kV levegőn át
30% legyen.
IEC 61000-4-2
Elektromos gyors
tranziens/burst
IEC 61000-4-4

Túlfeszültség
IEC 61000-4-5±2 kV tápvezetékek
esetében±1 kV bemeneti/kimeneti
vezetékek esetében±1 kV vezeték(ek) ésvezeték(ek) között
±2kV normál üzemmód
váltóáramú vezetékeknél±2 kV tápvezetékek
esetében
Nem alkalmazható±1 kV vezeték(ek) ésvezeték(ek) között

±1kV differenciálóüzemmód váltóáramú
vezetékeknél±2kV normál üzemmód
kültéri I/O vezetékeknél
0% UT 0,5 ciklusig
0% UT 0,5 ciklusig(0°, 45°, 90°, 135°, 180°, (0°, 45°, 90°, 135°, 180°,
225°, 270°, 315°)
225°, 270°, 315°)

Feszültségesések,rövidáramkimaradások és
feszültségváltozások
a bemeneti
tápvezetékekben
IEC 61000-4-11

0% UT 1 ciklusig (0 )0% UT 1 ciklusig (0 )70% UT (30% esés az UTértékben) 25/30 ciklusig(00)

70% UT (30% esés az UTértékben) 25/30 ciklusig(00)

0% UT 250 / 300 ciklusig(00)

0% UT 250 / 300 ciklusig
(00)

0

0

A hálózati áram minőségének kórházi vagy
otthoni ápolási környezetre jellemzőnek
kell lennie.

A hálózati áram minőségének kórházi vagy
otthoni ápolási környezetre jellemzőnek
kell lennie.

A hálózati áram minőségének kórházi vagy
otthoni ápolási környezetre jellemzőnek
kell lennie. Ha a FreeStyle Comfort
felhasználója folyamatos működést
kíván hálózati áramkimaradás esetén is,ajánlott a FreeStyle Comfort szünetmentes
tápegységről vagy akkumulátorról történő
működtetése.

Ipari frekvenciájú
A hálózati frekvenciás mágneses térnek
(50/60 Hz) mágneses 3 A/m
3 A/ma tipikus kórházi vagy otthoni ápolásimező IEC 61000-4-8
környezetre jellemzőnek kell lennie.
MEGJEGYZÉS: Az UT a váltakozó áramú hálózat feszültsége a tesztszint alkalmazása előtt.

MEGJEGYZÉS: Egyes esetekben a FreeStyle Comfort alapállapotba állhat vissza egy jelentősebb elektrosztatikus
kisülést követően, és esetleg teljes ki-/bekapcsolási ciklust kell végezni, hogy visszaálljon a normális működés. A további
részleteket lásd a Hibaelhárítás című fejezetben.

PN MN232-C4 K | Felhasználói kézikönyv

HUN – 495

FreeStyle ComfortÚtmutató és gyártói nyilatkozat – Elektromágneses zavartűrés

A FreeStyle Comfort készülék az alábbiakban megadott elektromágneses környezetben való használatra készült. A
FreeStyle Comfort készülék vásárlójának vagy felhasználójának kell biztosítania, hogy azt ilyen környezetben használják.
Zavartűrési
IEC 60601-es mérőszint Megfelelőségi szint
Elektromágneses környezet – útmutatás
vizsgálat
Vezetett RF
10 Vrms
10 Vrms
A hordozható és mobil rádiófrekvenciás
hírközlő berendezéseket tilos közelebb
IEC 61000-4-6
150 kHz-től 80 MHz-ig
vinni a FreeStyle Comfort készülékhez,
illetve annak bármely részéhez (a kábeleket
is beleértve), mint a jeladó működési
frekvenciájára vonatkozó egyenlettel
Sugárzott RF
10 V/m
10 V/mkiszámítható ajánlott elkülönítési távolság.
IEC 61000-4-3
80 MHz-től 2700 MHz-ig
Ajánlott elkülönítési távolság

d = 1,2√ P
d = 1,2√ P = 80 MHz – 800 MHz
d = 1,2√ P = 800 MHz – 2700 MHz
ahol P az adó gyártó által megadottlegnagyobb kimeneti teljesítménye
wattban (W), és d az ajánlott elkülönítési
távolság méterben (m).
A telepített rádiófrekvenciás adókból
származó térerőnek, ahogy az a helyi
elektromágneses mérésekből kiderül,akisebbnek kell lennie, mint az egyes
frekvenciatartományokban elfogadható
szint. b
A következő szimbólummal ellátott
berendezések közelében interferencia
léphet fel:

1. MEGJEGYZÉS: 80 MHz-en és 800 MHz-en a nagyobb frekvenciatartomány érvényes.
2. MEGJEGYZÉS: Ezek az irányelvek nem minden esetben érvényesek. Az elektromágneses hullámok terjedését
befolyásolja az építmények, tárgyak és emberek elnyelő és visszaverő hatása.
a
A helyhez kötött adók, például a rádiótelefonok (mobil/vezeték nélküli) és mobil földi rádiók bázisállomásai, az amatőr
rádiók, az AM és FM rádióadások és a televízióadások elektromágneses térerősségét nem lehet elméleti úton pontosan
meghatározni. A telepített rádiófrekvenciás adókészülékekből származó elektromágneses környezet felmérése érdekébenérdemes lehet helyi elektromágneses felmérést végezni. Ha a mért térerősség meghaladja a vonatkozó rádiófrekvenciás
megfelelési szintet azon a helyen, ahol a FreeStyle Comfort készüléket használják, akkor a FreeStyle Comfort készüléket
figyelni kell, és meg kell győződni a zavartalan működéséről. Ha rendellenes működést tapasztal, akkor további
intézkedésekre, például a FreeStyle Comfort készülék elfordítására vagy áthelyezésére lehet szükség.
b

A 150 kHz – 80 MHz-es frekvenciatartományban az elektromágneses térerősségnek 10 V/m-nél kisebbnek kell lennie.

496 - HUNPN MN232-C4 K | Felhasználói kézikönyv

FreeStyle Comfort
Ajánlott elkülönítési távolságok a hordozható és mobil rádiófrekvenciás távközlési
eszközök, illetve a FreeStyle Comfort készülék között

A FreeStyle Comfort készüléket olyan elektromágneses környezetben való használatra tervezték, amelyben a sugárzott
rádiófrekvenciás zavarok szabályozva vannak. A FreeStyle Comfort készülék vásárlója vagy felhasználója segíthet az
elektromágneses interferencia megelőzésében azzal, hogy betartja a hordozható és mobil rádiófrekvenciás távközlési
eszközök (jeladók) és a FreeStyle Comfort készülék közötti ajánlott elkülönítési távolságot, a kommunikációs berendezés
maximális kimeneti teljesítményének megfelelően.
A jeladó névleges
Elkülönítési távolság a jeladó frekvenciájától függően
maximális kimenetimteljesítménye
150 kHz-től 80 MHz-ig
80 MHz-től 800 MHz-ig
800 MHz-től 2,5 GHz-ig
d = 1,2√ P
d = 1,2√ P
d = 1,2√ P
W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Olyan jeladók esetében, amelyek névleges maximális kimeneti teljesítménye nem szerepel a fenti táblázatban, a d jelűajánlott elkülönítési távolság (méterben, m) a jeladó frekvenciájára alkalmazható egyenlettel becsülhető meg, ahol P ajeladó névleges maximális kimeneti teljesítménye wattban (W) a jeladó gyártója által megadott adatok szerint.
1. MEGJEGYZÉS: 80 MHz-nél és 800 MHz-nél a magasabb frekvenciára vonatkozó elkülönítési távolság érvényes.
2. MEGJEGYZÉS: Ezek az irányelvek nem minden esetben érvényesek. Az elektromágneses hullámok terjedését
befolyásolja az építmények, tárgyak és emberek elnyelő és visszaverő hatása.

BesorolásÁramütés elleni védelem típusa:
II. osztály	Az áramütés elleni védelmet dupla
szigetelés biztosítja.
Áramütés elleni védelem mértéke:
BF típus Az áramütés ellen bizonyos fokú
védelmet nyújtó berendezés a következőkkel
szemben		 1) megengedhető szivárgóáram;		2) védőföldelés csatlakozásának megbízhatósága
(ha van).
		 Nem szolgál közvetlen kardiológiai alkalmazásra.
A Gyógyászati villamos készülékekre vonatkozó
szabványnak való megfelelés független vizsgálata:
A vizsgálat az• IEC 60601-1 szabvány 3.1-es kiadásávalösszhangban készült: Gyógyászati villamos
készülékek – 1. rész: Általános biztonsági
követelmények.
• CAN/CSA C22.2 60601-1-8 2.1 számú kiadás,
Gyógyászati villamos készülékek – 1. rész:Általános biztonsági követelmények.
A berendezés és az egyéb eszközök közötti potenciális
elektromágneses vagy egyéb interferencia elleni
védelem.
	• A vizsgálatot az EN 60601-1-2/IEC 60601-1-2:
4. kiadású szabvánnyal összhangban végezték.
	• A vizsgálatot az RTCA/DO160 szabvány 21. rész,
M kategóriával összhangban végezték.
	• CISPR 11/EN 55011 B. osztály 1. csoport, „Ipari,tudományos és orvosi (ISM) berendezések”	• FCC 15. rész, B. alrész – B. osztály Nem
szándékos sugárzók

IP22 – Ujjak és 12,5 mm-nél nagyobb átmérőjű
hasonló tárgyakkal szembeni védelem. Afüggőlegesen csepegő víz nem károsítja, ha aburkolat legfeljebb 15°-ban dől meg a normális
helyzetéhez képest.
Megengedett tisztítási és fertőzésvédelmi módszer:
Lásd a jelen Freestyle kézikönyv „Tisztítás, ápolás
és helyes karbantartás” című részét.
Az alkalmazás biztonságának foka gyúlékony
altatógázok jelenlétében:A berendezés nem alkalmas ilyen használatra.
Működtetési mód: Folyamatos üzemelés.

CAIRE i CAIRE Inc. są zastrzeżonymi znakami towarowymi firmy
CAIRE Inc. Aby uzyskać pełną listę znaków towarowych, odwiedź
naszą stronę internetową poniżej.Znaki towarowe: www.careinc.
com/corporate/trademarks.
Copyright © 2021 CAIRE Inc. A CAIRE Inc. fenntartja a jogot
termékei megszüntetésére vagy termékei árának, anyagainak,berendezéseinek, minőségének, leírásainak, specifikációinak és/
vagy folyamatainak bármikor, előzetes értesítés és kötelezettség
és következmény vállalása nélkül való módosítására. Minden itt
kifejezetten nem meghatározott jogot adott esetben fenntartunk.
PN MN232-C4 K | Felhasználói kézikönyv

HUN – 497

FreeStyle Comfort

Notes

498 - ENGPN MN232-C4 K | User ManualFreeStyle Comfort
Notes

PN MN232-C4 K | User Manual

ENG - 499

www.c a ire inc .c om

CAIRE Inc.
2200 Airport Industrial Dr., Ste. 500
Ball Ground, GA 30107 U.S.A.
Medical Product Service GmbH
Borngasse 20
35619 Braunfels, Germany
Tel: +49 (0) 6442-962073
E-mail: info@mps-gmbh.eu
CAIRE and CAIRE Inc. are registered trademarks of CAIRE Inc. Please
visit our website below for a full listing of trademarks. Trademarks:
www.caireinc.com/corporate/trademarks/.
© Copyright 2021 CAIRE Inc. All Rights Reserved. CAIRE Inc.
reserves the right to discontinue its products, or change the prices,materials, equipment, quality, descriptions, specifications and/or
processes to its products at any time without prior notice and with nofurther obligation or consequence. All rights not expressly stated herein
are reserved by us, as applicable.

|MN232-C4-KQ~
4 Nov 2021

PN MN232-C4 K