Artemis Capture
User’s Manual
Version 3.0 - July 2015

Contents
1. Introduction .................................................................................................................................... 4
1.1 Key Features .............................................................................................................................. 4
2. Installing the Software .................................................................................................................... 5
2.1 Opening Artemis Capture ......................................................................................................... 5
3. The Main Screen ............................................................................................................................. 6
3.1 Tool Bar Buttons........................................................................................................................ 7
4. Exposure Window ........................................................................................................................... 8
4.1 Exposure Length........................................................................................................................ 8
4.2 Delay ......................................................................................................................................... 8
4.3 Lock ........................................................................................................................................... 8
4.4 Pre ............................................................................................................................................. 8
4.5 Binning ...................................................................................................................................... 8
4.6 Selecting a Subframe ................................................................................................................ 8
4.7 Autosave ................................................................................................................................... 9
5. Display Window ............................................................................................................................ 10
5.1 Histogram ................................................................................................................................ 10
5.2 Auto Stretch ............................................................................................................................ 10
5.3 Log ........................................................................................................................................... 10
5.4 Negative .................................................................................................................................. 10
5.5 Reticle...................................................................................................................................... 11
5.6 Correct Aspect Ratio ............................................................................................................... 11
5.7 Night Vision ............................................................................................................................. 11
6. Cooling .......................................................................................................................................... 12
6.1 Nonregulated Cooling ............................................................................................................. 12
6.2 Warming up the camera ......................................................................................................... 12
7. Taking an Exposure ....................................................................................................................... 13
7.1 Snapshot.................................................................................................................................. 13
7.2 Loop......................................................................................................................................... 13
7.3 Quick menu ............................................................................................................................. 13
7.4 Image Sequencer..................................................................................................................... 13
Page 2

7.5 Cancelling an exposure or a loop ............................................................................................ 14
8. Filter Wheel Control ...................................................................................................................... 14
9. Saving Images................................................................................................................................ 14
10. Focusing ...................................................................................................................................... 15
11. Autoguiding ................................................................................................................................. 16
11.1 Calibrating the Guide Axes .................................................................................................... 16
11.2 Dithering ............................................................................................................................... 17
12. Colour Cameras and the Bayer Matrix ........................................................................................ 18
13. Additional Features in Capture ................................................................................................... 19
13.1 Image Header ........................................................................................................................ 19
13.2 Marker Mode ........................................................................................................................ 19
13.3 Rotate 180° ........................................................................................................................... 19
13.4 Camera Information .............................................................................................................. 19
14. Further Information .................................................................................................................... 20

Page 3

1. Introduction
Firstly, congratulations on your purchase of an Atik Camera. Atik Cameras provide superior
performance, unparalleled ease of use and exceptional value for money. They are the result ofextensive research and development, each one having been designed and built with therequirements of the most demanding astroimager in mind, and will be your imaging companion for along time to come.
Artemis Capture is our very own standalone program which allows you to control your Atik camera,
capture and preview images and save them as standard FITS files. Solely designed for optimum
image capture, it is not an image processor, though it does provide the ability to manipulate thehistogram for mapping the 16-bit images to the display. Such adjustments are purely for display
purposes to help you get the most of your imaging sessions, and the data saved in the FITS file isexactly what is read from the camera (RAW).

1.1 Key Features
Special features of Artemis Capture include:• Pixel Binning• Sub-frame selection for rapid focusing• Automatic sequential capture, saved with sequential file names• Black and white level adjustment and log stretch for displaying images• Histogram function• Regulated cooling control• Motorised filter wheel control• Autoguider function with dithering• Focus Assist using data gathered from star images• Zoom feature• Display adjust for aspect ratio when using non-square pixels• Night-vision mode
• FITS file inspection

Page 4

2. Installing the Software
Although we provide a CD-ROM including a full software installer, we recommend, where possible,that you download the core software installer from the downloads page of our website;www.atik-cameras.com/downloads
We frequently update our core software so this is simply to ensure you have the absolute latest
version that may not yet be available on the CD-ROM. We also regularly update our CD-ROM
installer, so if you aren’t in a position to download the core software installer, don’t worry, you’ll still
have a very recent version.
To install the software:• Download the core software installer OR insert the CD-ROM• Run the file “SetupArtemisUniversal.exe”• Follow the prompts, ensuring ‘Drivers’ is ticked and selecting any additional plug-ins you
may require• The software should now be installed
When connecting the camera to the computer, an “Installing Device Drivers” message should pop
up. Follow any prompts, clicking ‘continue anyway’ if asked.
Note: The camera’s fan will come on as soon as you power the camera on – this does not mean thecooling is on.
Once the driver is installed, the camera is ready to use.
2.1 Opening Artemis Capture
It is recommended you connect your camera to the computer before opening Capture. If opening
Artemis Capture without a camera connected, the following message will appear:

If you have intentionally not plugged in a camera, clicking ‘Ignore’ will open Artemis Capture, thougha number of features will be unavailable. This is useful if you would like to inspect previously
captured FITS files.
If you are expecting the camera to be recognised, ensure the camera is powered on (whereapplicable) and connected to the computer. Sometimes it can take Windows a few minutes to findand install the correct driver – try waiting a few minutes before pressing ‘Retry’. You can also check
whether the camera is showing in the devices window.

Page 5

3. The Main Screen

This is the main screen of Artemis Capture. This is where you can access menus and short cut
buttons and control which additional menus are displayed. The green rectangle represents the fullsize of the CCD sensor and this is where the most recent image will be displayed. The bar along thebottom displays the programs status, including whether the current image has been saved, thecurrent level of zoom, and whether the camera is idle, currently exposing or an image is being
downloaded. This is particularly useful to check on the status of longer exposures as the bar willcountdown the amount of time left.

Page 6

3.1 Tool Bar Buttons
A number of key features can be controlled through the Tool Bar Buttons on the main screen as
follows:
Save: Saves the current image
Autosave: Auto saves all captured images, according to the settings in the Exposure window
Display: Toggles the display window on/off
Exposure: Toggles the Exposure window on/off
Cooler control: Controls the camera cooler and temperature (on supported models)
Focus and Autoguider: Toggles focus-assist and autoguider window on/off
Sub-frame: Selects a specific area of the image instead of the full frame. Very useful for
focusing.
Lock settings: Locks all exposure control settings
Marker mode: Activates the image orientation assist
Snapshot: Initiates an exposure with duration as specified in the Exposure window
Loop: Toggles the loop exposure on/off. Useful for focusing and capturing sequences
Sequencer: Allows programming a sequence of images, which can use different
exposures/filters
Filter Wheel control: Controls the optional Atik EFW-U/EFW2 filter wheel

Page 7

4. Exposure Window
You can toggle the Exposure window by pressing the Exposure button
or by clicking VIEW > Exposure Settings from the main menu.
This window allows you to control the exposure settings, along withbinning and autosave functions.
4.1 Exposure Length
The exposure [ Exp.(s) ] is displayed in seconds, with the [ min. ] selector
adding 60, the [ sec. ] selector adding 1 and the subsequent selector
handling all decimal places. The shortest exposure available is 0.001s.
4.2 Delay
The Delay [ Dly .(s) ] function allows you specify a delay in seconds
between exposures when looping images. This is particularly useful for seeing changes in focus andfor guider stabilisation.
4.3 Lock
Checking this box locks the current exposure settings.
4.4 PreThe Pre checkbox allows the camera to function in a preview mode. This speeds up image download
time to roughly twice as fast, but the image quality will suffer. This is particularly useful when
focusing the camera and testing your equipment, but is best unchecked when taking final images.
4.5 Binning
BinX and BinY specify the level of binning for an image. It is usually best to keep these values thesame, in which case =BinX can be selected from the BinY options.
4.6 Selecting a Subframe
Selecting a subframe, or region of interest, can have a number of benefits including faster download
times and focusing on a single area. A subframe will be shown as a red rectangle within the green
rectangle on the main screen. Only the area within the red rectangle will be displayed on the next
exposure.

Page 8

The easiest way to specify a subframe is by using the Subframe definition button. After clicking
the button, a subframe can be drawn using the mouse. This can also be accessed by selecting
CAMERA > Subframe on the main menu, or by holding the SHIFT key and drawing a box with themouse.
The StartX and StartY boxes in the exposure window will then respectively display the horizontal andvertical pixel values for the start of the subframe from the top left-hand corner of the full frame(green rectangle). The Width and Height values in the exposure window display the size of thesubframe itself. These values can also be altered directly to create a subframe.
A subframe can be removed either by deselecting the Subframe definition button, pressing the ‘Full
Frame’ button in the exposure window, or by manually re-entering the full pixel size of the sensor in
the width and height values.
4.7 Autosave
Due to the quantity of images often required in astroimaging, it may be preferable to use the‘autosave’ feature. The destination folder when autosaving can be set in the exposure window,
along with the sequence format and sequence number.

Page 9

5. Display Window
The display window can be toggled using the display window
buttonor by going to VIEW > Display Options. This allows
you to adjust the way in which the 16-bit image is mapped to thePC’s display.
Note: Changing these settings only alters how the image isdisplayed within Capture, and not the image itself. The image
saved will be from the raw camera data, regardless of how thedisplay settings may alter the image on screen.
5.1 Histogram
The histogram provides a simple way to visually check the pixel
intensity distribution of each image. The red marker shows where the black level is set while thegreen marker displays where the white level is set.
5.2 Auto Stretch
The auto stretch function allows the program to automatically adjust the black and white levels eachtime an image is captured. The levels are selected in such a way as to only allow a few pixels to falloutside the displayed range.
If ‘Auto stretch’ is unchecked, you can manually set the black and white levels. These will remain thesame until new levels are selected, or auto stretch is re-selected.
Hint: This is particularly useful for checking things such as star saturation. If you set the black level toaround 40,000, this will allow you to see how many stars in your image are coming close tosaturation point. You may then adjust your exposure length accordingly.
5.3 Log
The log value controls how grey shades are distributed in the image. Setting this at zero provides alinear map, while a positive value will increase contrast at the lower end and a negative value will
increase the contrast at the higher end.
5.4 Negative
The Negative button will invert the colours of an image. This can be particularly useful when looking
for faint objects as it is often easier to determine such details against a light background than it isagainst a dark background.

Page 10

5.5 Reticle
This simply adds a reticle to the image, allowing you to determine the centre of the frame. This canhelp with alignment and image composition.
5.6 Correct Aspect Ratio
This adjusts the displayed image according to the physical dimensions of the pixels on the CCD
sensor for cameras models that do not have square pixels. If the box is checked then the height ofthe displayed image is scaled so that a square area on the screen corresponds with a square area inthe sky.
5.7 Night VisionThis option changes the colour scheme to red in order to help preserve visual dark adaptation whenusing the software at night.
Selecting Night Vision mode may also change the colour settings in other programs – these will be
restored either when Night Vision is unchecked, or when Capture is closed.
Hint: It’s possible to save the Night Vision colour scheme for your general use. To do this, selectNight Vision in Capture, right-click on the Windows desktop, and select ‘Properties’ or ‘Personalize’,
depending on your version of Windows. You should then be able to save this as a theme. It’s
advisable to save your current theme before doing this, so you’re able to return to it when you nolonger need night vision.

Page 11

6. Cooling
The cooler menu can be toggled by pressing the Adjust Cooler
Settings button

or by selecting VIEW > Cooler Settings.

The cooling will be off by default and Status will display ‘Off’. The
temperature will show the current temperature of the camera
and will vary depending on oundings.
To turn the cooler on, select the desired setpoint using the arrows and press the ‘Cooler On’ button.
It’s best to then wait for up to 5 minutes to allow the temperature to fully stabilize.
A note about Setpoints
The level of cooling your camera can achieve is influenced by your surroundings. The figures wesupply are cooling deltas, and specify how far below the ambient temperature the camera can cool.
So, for example, a camera with a cooling delta of -30°C will be able to cool to 30°C below theambient temperature, meaning that if your imaging location is 10°C on a given night, your camera
will be able to achieve -20°C. If your imaging location is 20°C, your camera will be able to achieve 10°C, and so on.
It’s often more important to image at an accurate, regulated, repeatable temperature than it is tosimply operate at maximum cooling, and for this reason we recommend setting the cooling at aconsistent and controllable level for your environment.
This means you can maintain a stable temperature throughout an entire nights imaging where thetemperature may fluctuate slightly. This is also the case when imaging across a number of nights
where the temperature may change more noticeably.

6.1 Nonregulated CoolingOn cameras where the cooling is unregulated, the temperature will be displayed as ‘unknown’. The
cooler can only be turned on or off and will always cool to the maximum possible level.
6.2 Warming up the camera
After switching the cooler on, the button will change to ‘Warm up’. When pressed, this will initiate
the power down sequence, which will raise the temperature slowly. This is necessary to avoid
thermal shocks on the CCD and should always be performed before turning the power off. Closing
Capture will also initiate the power down sequence, though it is necessary to wait a few minutes
after closing Capture for this to be performed before powering down either the computer or thecamera.

Page 12

7. Taking an Exposure
There are a number of different ways to initiate an exposure in Capture, and which you use will
depend on your needs.
7.1 Snapshot
The snapshot button
will initiate a single exposure based on the current settings in the
exposure window. This can also be done by selecting CAMERA > SNAPSHOT through the main menu.
7.2 Loop
The loop button
will start taking repeated images based on the settings in the exposure
window. Settings can be changed during looped exposures, but will not take effect until the next
exposure starts. A loop can also be initiated by selecting CAMERA > LOOP on the main menu.
This is particularly useful when focusing the camera or setting up the autoguiding function.

7.3 Quick menu
The quick menu provides a number of shortcuts to pre-specified exposures. These canbe initiated both through the Quick menu and through the corresponding F1-F8 shortcuts.
These can be modified by selecting QUICK > Save… so you can create personalised
shortcuts for up to 8 of your most used exposure settings, depending on your needs. As
part of this, you can specify the level of binning and whether this initiates a single
exposure or a loop.

7.4 Image Sequencer
Capture also features an image sequencer
which allows you to take multiple images in a
pre-specified sequence. This can be accessedby pressing the Sequencer button
or by
selecting CAMERA > Sequencer from the mainmenu.
The sequencer allows you to specify settings
for up to 10 different exposures, including
length, binning and filter wheel position. You
can set how many times the sequence is to berepeated and in what order
Page 13(1,1,1…2,2,2…3,3,3 or 1,2,3…1,2,3…1,2,3). You can also set a delay between exposures and delay
before the start of the sequence, both of which may be useful for guider stabilization.
When you are happy with your sequence, press Run to initiate it.
Through this window you can also save and load sequences.
Images taken during the sequence are automatically saved according to the settings in the Exposure
window, along with any file suffix specified in the Sequencer window.
7.5 Cancelling an exposure or a loop
An exposure can be cancelled by unselecting the Snapshot button whilst it’s in progress. You can also
stop a loop by unselecting the loop button whilst imaging is in progress.
If you want to cancel the exposure early, but still keep the result, hold the shift key while deselecting
the exposure button. The image will then be downloaded and displayed in the main window.

8. Filter Wheel Control
Any Atik Filter Wheel can be controlled in Artemis Capture. To control the Filter Wheel manually,
select the filter wheel buttonor select CAMERA > FILTER WHEEL from the main menu. This
allows you to select which filter number you wish you to use and changing the filter number will
move the wheel to the selected position.
As mentioned, the filter wheel can also be controlled automatically using the image sequencer.

9. Saving Images
Single images can be saved by pressing the ‘Save as’ button
or by selecting FILE > Save Image
As… in the main menu. This will allow you to select a destination and name and save your image.
As previously mentioned, there is also an autosave function available. This can be selected in the
exposure window by checking the box, by pressing the ‘Autosave’ button
FILE>Autosave in the main menu.

or by selecting

Note: Unless using the sequencer, Artemis Capture will NOT save images unless either the autosave
function is selected, or you manually save the desired images. Please make sure you are saving yourimages.

Page 14

10. Focusing
As accurate focusing is such an important part of astroimaging, Capture contains a Focus Assist
feature to help you achieve the best focus possible.
Roughly focus your telescope and take an exposure sufficient to see stars. You can then activate the
Focus Assist window by double-clicking on a star. You can also use the Tracking buttonor select
VIEW> Tracking to open the window, which will then ask you to double-click a star to select it.
For best results, choose a star
which is bright but notoverexposed (you can check thisusing the method described in
section 5.2 on page 10) and which
has a relatively clear area around
it. Capture will select the brightest star in the vicinity of your selection and draw a small box around
it. It’s then a good idea to select a subframe around your chosen star to reduce the image download
time. You will also want to zoom in on the star so it’s highly magnified as this makes it easier tovisually verify the image quality.
Once a star is selected, the window will display the width of the star (FWHM) and its maximum
brightness (Bri). These are in bold to help you see them when you’re some distance from themonitor.
You are trying to get the brightness
to a maximum and the width to aminimum. This will give the
sharpest and brightest star images.
Press the Loop button to begin taking repeated exposures and adjust your telescope’s focus until it’sat an optimum level both visually and with the maximum and minimum values.
Capture will track the star from one exposure to the next by searching for the brightest pixel in thearea around the star’s previous location. This is useful as the star may drift by a number of pixels
depending on the magnification and stability of the telescope.
You can use the Tracking Search Radius to set how many pixels Capture will search around the star’s
previous location to find it. This figure should be slightly larger than the radius of the star, but
smaller than the distance to the next star. You can also use the measurement radius to measure thestar’s properties once located.

Page 15

11. Autoguiding
The autoguiding function in Capture allows you to use a second camera to guide on a reference starand send tracking information to your mount, allowing you to take much longer exposures. If alreadyusing Capture with your main camera, you must open a second instance of Capture. The camera will
be displayed in the Artemis Capture header, allowing you to identify which instance is running whichcamera.
Once you have focused your guide camera on your selected guide star using the process described in
Section 10 on page 15, check the ‘Setup Guiding’ box to access guider options.

Use the ASCOM chooser to select the mount that guiding commands will be sent to. Your mountmust be setup using ASCOM before you use the autoguide function.
To setup guiding, uncheck the ‘Autosave Images’ box (unless you have a particular reason to saveyour guide images) and begin a looped exposure.
11.1 Calibrating the Guide Axes
During calibration, Capture tests the mount in four directions to measure the resulting motion of theguide star. You can use the Calibration time [Calib. time (ms)] to set the duration of pulse that will be
issued during calibration. This can vary depending on factors such as focal length and mount speed,but 2/3 seconds is usually sufficient.
Page 16

The Min guide (ms) and Max guide (ms) options allow you to set the minimum and maximum pulse
duration that will be used for guide corrections.
Although you can also select which axes will be used for correction during guiding with the ‘Guideaxes’ box, all axes are enabled during calibration.
The Feedback RA and Feedback Dec options set how many tenths of the apparent motion of theguide star Capture will try to correct for. If you find your star is deviating a lot, it may be worth
lowering these values.
The stars deviation will be shown in the graphs. You can monitor these to ensure correct guiding.
11.2 Dithering
When guiding in Capture, it is also possible to dither. Dithering introduces very small shift indirection between exposures which can aid in the removal of hot pixels. You can select the numberof pixels to dither uses the Dither (pixels) selection box. 2-5 pixels is usually sufficient.

Page 17

12. Colour Cameras and the Bayer Matrix
Nearly all of our cameras are available in either mono or colour versions. These are the same interms of all functionality except the way they capture colour data. Whilst mono cameras can capturecolour data through the use of filters, colour cameras do it by placing red, green and blue filters overindividual pixels in a pattern called a Bayer Matrix. This can then be converted to a colour image
through a process called Debayering.
Capture can display a colour image directly from the exposure. You can control this through the
COLOUR option in the main menu.
If Automatic is selected, Capture will automatically debayer the image according to the camera
settings and display a colour version.
You can manually select how the image is displayed
through the COLOUR > ADVANCED menu.
• RAW will display the raw data from thecamera.
• Colour will display a synthesized colour image
from the RAW data.
• Luminance will display a synthesised
monochrome version of the image.
• Bayer Offsets allows you to manually alter theway the image is debayered.
The Auto setting will automatically select the correct offsets.
As mentioned, these settings are purely for display purposes. Images will be saved from the RAW
camera data and will need debayering outside Capture for further processing. This can be done in avariety of astrophotography processing software, including our own image processer, Dawn.
Note: If you use binning with a colour camera, you will only capture a monochrome image. This isbecause binning combines different colours of the bayer matrix, rendering it unable to collectcorrect colour data.

Page 18

13. Additional Features in Capture
13.1 Image Header
Through the FILE>Image Header… option you can specify anumber of details to be embedded in the image header,such as the object, scope details and observer name. You
can also record the exposure length to 0.1 second
precision. This can help reduce the need for you to takeadditional notes for future reference, particularly if you’replanning to send images out for publication, re-process
them at a later date, or enter them in one of ourcompetitions.
13.2 Marker Mode
Marker mode is a handy tool to help align your images between sessions, particularly if you have to
set up from scratch every time. It allows you to select ‘marker’ stars which you can use to align yourcamera on the next session. This is particularly useful for minimising rotation and helps cut down on
the area that may be lost in post-processing.
Simply select the Marker mode button
and then either double-click, or right-click and selectinsert marker, to play a marker on the reference stars you would like to use. A crosshair will then be
visible on top of the star. It’s best to use bright stars as they will be easier to find later on, and will
require shorter exposures when aligning.
If you are unhappy with a marker, right-click above it and select delete marker.
When you are happy with all your markers, right-click and select save markers. We find 2 -3 is usually
enough.
To reload your markers, right-click on an image with marker mode selected and choose load
markers. You can now adjust your camera rotation and use your mount controls to align your image
to your previous session.
Once no longer needed, right-click and select delete all markers to remove them.
13.3 Rotate 180°
Selecting CAMERA> ROTATE 180° will rotate the image by 180°.
13.4 Camera Information
Selecting CAMERA>Info will display some information about the camera, namely the model,
manufacturer, serial number and protocol. This is particularly helpful if you no longer have a recordof the serial number elsewhere and need to contact us for support.

Page 19

14. Further Information
If you need any further information on Artemis Capture or anything else to do with your Atik
Camera, you can find answers to a number of frequently asked questions at http://www.atikcameras.com/faqs/
You might also like to join our forum, where Atik users can go to discuss all things astrophotography,
as well as offer advice and guidance to one another. http://forums.atik-cameras.com/index.php
If you have a support issue that isn’t covered here, you can always get in touch with our support
team by emailing support@atik-cameras.com
We wish you clear skies and happy imaging!

Page 20