PuraLev®
Pump Console for Life Science Applications

LCO-i100
For PuraLev® i30SU/MU & PuraLev® i100SU/MU& PuraLev® i600SU

Advanced Ultrapure Fluid Handling!

SYSTEM OVERVIEWPuraLev® i600SU
Pump System with
Single-Use Pump HeadINTRODUCTION
The PuraLev® Console integrates Levitronix® MagLev pumps with
LEVIFLOW® ultrasonic flow sensors and adds a sophisticated userinterface with enhanced connectivity features. The system allows
various operating modes including stand-alone pumping with flow
monitoring or flow control, pressure monitoring with up to 3
pressure sensors or pressure control and temperature monitoring.
The brain of the PuraLev® Console is an embedded PC with a 7”
multi-touch screen and it comes with software for flow or pressure

PuraLev® i30SU/MU
Pump System with
Single-Use and Multi-Use
Pump Head

control (or control with custom signal), system configuration andmonitoring of various sensors. The console can operate the pumpdrivers of the PuraLev® i30, PuraLev® i100 or PuraLev® i600 pumpsystem. Various integrated sensor signal converter circuits allow
connection of LEVIFLOW® single-use and clamp-on flow sensors,and single-use pressure and temperature sensors from suppliers
such as PendoTECH. The console can be placed horizontally on atable or mounted vertically on a wall. The display is rotatable bysoftware.
®

PuraLev i100SU/MU
Pump Systems with
Single-Use and Multi-Use
Pump HeadPuraLev® Fluid Handling
Console LCO-i100

PendoTECH Single Use
Pressure SensorsTM
PendoTECH Single Use
Temperature SensorsTM

SYSTEM BENEFITSPrecise flow monitoring or control.

Integrated signal converters for pressure & temperaturemonitoring.

Precise pressure limitation to avoid potential tube ruptures.



Plug and play stand-alone functionality.

Intuitive user interface with 7” color multi-touchscreen.

Data collection capabilities.



Easily configurable and customizable by software.

Custom recipe control for process automation.

Configurable user rights.

APPLICATIONS®

LEVIFLOW
Clamp-On
Flow Sensors
LFSC-iX Series
LFSC-iX.1 Series

LEVIFLOW® Clamp-On
Flow Sensors LFSC-D Series

LEVIFLOW® High Precision
Single-Use Flow Sensors

Figure 1: Component connectivity of PuraLev® fluid handling console

Fluid transfer and recirculation in upstream and downstream
bioprocessing.

Single-use and multi-use components available.



Ideal for RG2+ agents and toxic products due to inherent
maximum pressure limitation.



Depth filtration, microfiltration, ultrafiltration, diafiltration,sterile filtration.

Mixing, blending, dilution and filling.

Filter integrity testing.

SYSTEM CONFIGURATIONS
0.0[gallons / min]

0.5

1.0

1.5

2.0
16
Specific gravity = 1g/cm 3
Viscosity = 0.7cP
Liquid Temp.: 40°C

1.0

STAND-ALONE PUMP SYSTEM CONFIGURATION

1413000 rpm

120.8

With the standard stand-alone pump system configuration, as
illustrated in Figure 8 (for PuraLev® i30 and i100 pump system)
and Figure 9 (for PuraLev® i600 pump system), the speed of thepump can be set manually and the flow can be monitored with a

[bar]

[psi]

0.6

8

11000 rpm

60.4
10000 rpm49000 rpm
8000 rpm

0.2
7000 rpm

LEVIFLOW® flow sensor. The mode of operation can be easily
switched to a flow control setup, where the flow is set manually

1012000 rpm

26000 rpm
5000 rpm
0.0

0
0

1

2

3

4

5

6

7

8

[liters / min]Figure 3: Pressure/flow curves with PuraLev® i30SU (DCP-30.1/2 pump heads)on the 7” multi-touchscreen and is controlled by the integrated[gallons / min]

PC.

0.0

0.2

0.4

0.6

0.8

1.0

3.0

The USB interface on the rear side facilitates software updates
or data collection for debugging with a USB stick.

Specific gravity = 1g/cm3
Viscosity = 0.7cP
Liquid Temp.: 40°C
DC-Voltage = 24VDC
Pump Head: LPP-30.7

19000 rpm

2.5

18000 rpm

combination with the touch screen, simplifies the usage,

35
30

2.0
17000 rpm[bar]

The console comes with an intuitive software, which, in

40

2516000 rpm

1.5

[psi]

20
15000 rpm151.0
14000 rpm

configuration and monitoring of the system.

10

13000 rpm

0.558000 rpm
5000 rpm

0.0
0.0

0.5

1.0

10000 rpm 11000 rpm

1.5

2.0

2.5

3.0

12000 rpm

3.5

0
4.0

[liters / min]Figure 4: Pressure/flow curves with PuraLev® i30MU (LPP-30.7 pump head)
0.0

1.0

2.0

[gallons/min]

3.0

4.0

5.0

2.25
Specific gravity = 1 g/cm3
Viscosity = 0.7 cP
Liquid Temp.: 40°C
DC-Voltage = 24VDC
Pumphead: DCP-200.2

2.009000 rpm

1.75

30

25

1.50
[bar]

20

8000 rpm

1.25

[psi]

15

1.00
7000 rpm

0.75

10

0.50

6000 rpm
5000 rpm

0.25

54000 rpm

0.00
0.0

2.5

5.0

7.5

10.0

12.5

15.0

0
20.0

17.5

[liters/min]Figure 5: Pressure/flow curves with PuraLev® i100SU (DCP-200.2/3 pump heads)
0.0

1.0

2.0

[gallons/min]

3.0

4.0

5.0

2.50
35

10000 rpm

Figure 2: Start-up menu of touch screen

Specific gravity = 1 g/cm 3
Viscosity = 0.7 cP
Liquid Temp.: 40°C

2.25

30

2.00
9000 rpm
1.75

25

1.50
20

[bar]

8000 rpm

1.25

[psi]

15

1.00
0.75

10
7000 rpm

ADVANCED FLUID HANDLING CONFIGURATION

0.50
5
0.25
4000 rpm

Figure 10 illustrates the extended monitoring capabilities of the®

PuraLev console for advanced fluid handling applications such

6000 rpm

5000 rpm

0.00
0.0

2.5

5.0

7.5

10.0

12.5

15.0

0
20.0

17.5

[liters/min]Figure 6: Pressure/flow curves with PuraLev® i100MU (LPP-200.16/17 pump heads)

as depth filtration, microfiltration, ultrafiltration, diafiltration,[gallons / min]0.0

tangential flow filtration (TFF), sterile filtration or pressure control

3.5

for filling stations.

3.0

2.5

5.0

7.5

10.0

12.5

15.0

17.5

20.0
50

Specific gravity = 1 g/cm3
Viscosity = 0.7 cP
Liquid Temp.: 40 °C
DC-Voltage = 48 VDC
Pump Head: DCP-600.2

9000 rpm

40

2.5

The PuraLev® console includes two flow sensor connections andgauge amplifiers for connection of up to three single-use

2.0

30

8000 rpm

[bar]

[psi]

1.5

20
7000 rpmpressure sensors and an additional circuit for the connection of

1.0

a temperature sensor. Furthermore, there is a generic analog

0.5

10

6000 rpm
5000 rpm

input for connection of further external signals.

0.0

0
0

10

20

30

[liters / min]

40

50

60

70

Figure 7: Pressure/flow curves with PuraLev® i600SU (DCP-600.2 pump head)

80

SYSTEM CONFIGURATIONS

Fluid OutMulti-Use pump head
option for PuraLev-i30/100
2b

USB Interface for
Firmware Updatewith USB Stick

6

5

11a

3b

Flow Sensor

2b
2aSingleUse
Pump
Head

Fluid In

2d

Note: Direct driverconnection to console
possible.

3d

3a

PuraLev® Console

1

Flow Sensor
LFSC-D
LFS-SU

Integrated Pump
Driver OEM Model

Multi-Touch
Screen 7"(with Connector)

3d

Pump Head Socket
16a(Necessary for DCP
single-use pump heads)

16b
14Power toIntegrated
Driver16c

AC Supply
Input

AC/DC Supply

DC Supply Input

AC Mains Cable

On-Off Button®

Figure 8: Stand-alone pump system configuration of LCO-i100 console and PuraLev i30/100 pump systems with flow control or flow monitoring

Flow Sensor
LFSC-D
LFS-SU
5

Fluid Out
6
11a

1
USB Interface for
Firmware Updatewith USB Stick4b

4a

Fluid In

SingleUse
Pump
Head

PuraLev® Console

Flow Sensor

Integrated Pump Driver
OEM Model(with Connector)
Multi-Touch
Screen 7"

Note: Whenrunning with a Y
split box thedriver connector
can not be used
to attach anotheri30/i100 pump.

4c

Pump Head Socket(Necessary for DCP single-use
pump heads)

Power to
Driver19

Y-Split Box

DC Supply Input(with RS485)(Integrated DC/DC
Converter)

On-Off Button

1717b

AC Mains Cable
17a

AC/DC Supply(Sealed)AC Supply
Input

Figure 9: Stand-alone pump system configuration of LCO-i100 console and PuraLev® i600 pump system with flow control or flow monitoringSYSTEM CONFIGURATIONS

Fluid
In

7

8
10

Flow
Sensors
LFSC-iX
LFSC-iX.1

Single-Use
Pump Head

InlineTemp.
SensorPump Head Socket

Pressure
Sensor 1

Integrated Pump
Driver OEM Model9(with Connector)
11b

12

Pressure
Sensor 2

9

Pressure
Sensor 3

913

External
PLC

15
12
12

AC/DC
Supply11a

Flow
Sensors

5
6

LFSC-D
LFS-SU

AC
Mains
Cable

Fluid
Out
Multi-Touch
Screen 7"

AC Supply
Input

PuraLev ®
ConsoleOn-Off
ButtonFigure 10: Extended monitoring configuration of console LCO-i100 for advanced fluid handling in bioprocessing applications
Note 1: For PuraLev® i600 pump system the power supply and console connection concept shown in Figure 9 has to be used.

DIMENSIONS AND INTERFACES

1PuraLev® Console
LCO-i100.1

Note: VESA 100 mounting interface.

Note: The 4 rubber feet can be removed
for mounting the console on threads.
Multi-touch screen 7”

On-Off Button

Figure 11: Dimensions of PuraLev® console LCO-i100.1
Note 1: Dimensions in mm and [inch]. Un-tolerated dimensions are for reference only.

5

6

7

2

#

Interface
Connector

Description

1

DC Supply
Connector

Single phase DC connection.

2

Pump DriverPower connector for
IPD-30 and IPD-100 drivers.

--

3

Remote

RS485 for future purposes.

Not active.

4

PLC

Remote control.
Configurable designations.

2x digital outputs (outputs software configurable
to 24VDC), 2x digital inputs, 1 analog output (420 mA), 1x analog input (4-20mA)

5

Flow Sensor
LEVIFLOW®

Connection of flow sensors fromthe LFS-SU and LFSC-D series.

Circular 6 pin snap-in connector.
Purpose: Flow control or monitoring.

6a
6b
6c

Pressure Sensor 1
Pressure Sensor 2
Pressure Sensor 3 1

Connection for single-usepressure sensors 1, 2 and 3.

Needs adaptor cable (see Table 7)
Circular 6 pin snap-in connector.
Purpose: pressure monitoring.

7

Temperature Sensor

Connection for 1 single-use
temperature sensor.

Needs adaptor cable (see Table 7).
Circular 6 pin snap-in connector.
Purpose: Temperature monitoring.

8

Analog In
24VDC OutMulti purpose sensor input.

Current input 4-20 mA, 24 VDC supply.

9

RS485
24VDC Out

Sensor input for new clamp-on
flow sensors LFSC-iX/iX.1

RS485 bus with 24 VDC supply.

10

LAN Interface

TCP/IP Modbus

For connection to OPC devices or Levitronix®
Service Software.

11

USB Interface

Connection of USB stick.

Connection with protective cover.
Purpose: Data collection and software update.

1

11
10

Specification/Purpose/Note
24 VDC with external AC/DC supply.
Software configurable Autoresume function.
RS485 for connection of PuraLev-i600 with Y
split cable..

Note: All signals isolated.

8

3

9

4

Note: All connectors come delivered with a protective cap for sealing if unused.
When connected all connectors are at least IP66 rated.

On-off button

Figure 12: Interfaces of PuraLev® console LCO-i100.1

ORDER INFORMATION
Pos.

Component

PuraLev® Console

1

Article Name

Article #LCO-i100.1-01

100-30501

LCO-i100.3-01

100-30539(FDA CFR 21 Part 11 ready)Characteristics

Value / Feature

Supply Voltage / Power

24 VDC with 50 W or 160 W supply (see Table 7)

Housing Rating / Weight

IP65 2 / 1.6 kg

Interfaces

Multi-touch screen 7”, 3 pressure sensors, 2 flow sensors, 1 temp. sensor
PLC (2 digital in/outputs, 1 analog in/output), LAN, USB

Pump Motors

PuraLev® i30/ i100 integrated driver. PuraLev® i600SU with relevant Y cable.
(Only one driver can be connected at the same time).

Certifications

CE, IECEE CB scheme, ETL (NRTL) listed.

Table 1: Basic specifications of standard LCO-i100 console.
Note 1: AC/DC power supply and AC mains cable for various countries to be ordered according to Table 7.

Component 2

Pos.

Note 2: Tested and designed for IP66.

Article Name

Article #Characteristics

Value / Feature

DCP-30.2 (Barb 3/8”) 1 3
DCP-30.1 (Triclamp 3/8”) 1
DCP-30.5 (Barb 1/4”)

100-90968
100-90959
100-91691

Wet Materials
Sterilization Methods

PP (FDA, USP Class VI, BSE/TSE/animal-free)Gamma radiation up to 40 kGy.

Wet Materials / Fittings
Sterilization Methods

PEEK, PTFE, EPDM (FDA, USP Class VI, BSE/TSE/animal-free) / Triclamp 3/8”)
CIP (clean in place), Autoclaving

Housing
Cable / Connectors
Note SU Drivers

Epoxy (anti-corrosive) coated aluminum, IP65 (tested for IP67)
1 x 0.3m cable with PVC jacket / circular connector
IPD-30.8-03-02 for pump heads DCP-30.1/2, IPD-30.8-03-07 for DCP-30.5

2aSingle-Use Pump Head

2b

Multi-Use Pump Head

LPP-30.7 (no drain port)

100-91557

2c

Driver

IPD-30.8-03-02 (SU)IPD-30.8-03-07 (SU)IPD-30.8-03-06 (MU)

100-10104
100-10246
100-10212

2d

Pump Head Socket

PHS-i30.1

100-90947

Mounting Type / Material

Bayonet mount with locking pin made of anodized aluminum

Single-Use Pump Head

DCP-200.2 (Triclamp) 1
DCP-200.3 (Barb) 1 3

100-90734
100-90792

Wet Materials
Max. Flow / Max. Diff.-Pressure
Sterilization Methods

PP (FDA, USP Class VI, BSE/TSE/animal-free) /17.4 liters/min (4.6 gallons/min) / 2 bar (29 psi)
Gamma radiation up to 40 kGy.

LPP-200.17 (with drain port)
LPP-200.16 (no drain port)

100-90864
100-90863

Wet Materials / Fittings
Max. Flow / Max. Diff.-Pressure
Sterilization Methods

PVDF, PFA and EPDM (FDA, USP Class VI, BSE/TSE/animal-free) / Triclamp ½”
17.4 liters/min (4.6 gallons/min) / 2.4 bar (35 psi)
CIP (clean in place), Autoclaving

100-10105
100-10137

Housing
Cable / Connectors

Epoxy (anti-corrosive) coated aluminum, IP65 (tested for IP67)
1x 0.3m cable with PVC jacket / circular connector

100-91053

Mounting Type / Material

Bayonet mount with locking pin made of anodized aluminum

Wet Materials
Max. Flow / Max. Diff.-Pressure
Sterilization Methods

PP (FDA, USP Class VI, BSE/TSE/animal-free) /75 liters/min (20 gallons/min) / 3.1 bar (45 psi)
Gamma radiation up to 40 kGy.

(Needs pump head socket.)(MBP-i30.1 included)

3a(Needs pump head socket.)

3bMulti-Use Pump Head

3c

Driver

IPD-100.3-03-02 (SU)IPD-100.3-03-03 (MU)

3d

Pump Head Socket

PHS-i100.1
DCP-600.2 (Triclamp)1100-90784

Triclamp or Barb ½”

Triclamp 1”

4aSingle-Use Pump Head

4b

Driver

IPD-600.3-30-02

100-10197

Housing
Cable / Connectors

Epoxy (anti-corrosive) coated aluminum, IP65 (tested for IP67)
1x 3m cable with PVC jacket / circular connector

4cPump Head Socket

PHS-600.1

100-90696

Mounting Type / Material

Bayonet mount with locking pin made of anodized aluminum(Needs pump head socket.)

Table 2: Specification of standard pump motors and single-use pump heads compatible with LCO-i100 console.
Note 1: Gamma irradiation option available.

Note 2: See Levitronix® technical brochures of PuraLev® i30SU/i30MU/100SU/i100MU/600SU for detailed specifications and other configurations.

Note 3: Sterile fitting options available.

1
2c

2c

2b

2a

2d

3c

3c

4b

4a
3a

3d

3b
4c

Figure 13: LCO-i100 console with compatible drivers and pump heads

ORDER INFORMATION
Pos.
5a
5b
5c
5d
5e
5f
5g
5h
5i
5j

Component

Article Name

Article #Fitting

Wet Material

Note

LEVIFLOW® Single-Use
Flow Sensors

LFS-03SU-Z (0.8 lpm)
LFS-03SU-Z-SC1 (0.8 lpm) 1
LFS-06SU-Z (8 lpm)
LFS-06SU-Z-SC1 (8 lpm) 1
LFS-10SU-Z (20 lpm)
LFS-10SU-Z-SC1 (20 lpm) 1
LFS-15SU-Z (50 lpm)
LFS-15SU-Z-SC1 (50 lpm) 1
LFS-20SU.1-Z (80 lpm)
LFS-20SU-Z-SC1 (80 lpm) 1

100-30375
100-30418
100-30377
100-30394
100-30397
100-30408
100-30412
100-30431
100-30483
100-30464

Triclamp 3/8”
Triclamp 3/8”
Triclamp 3/8”
Triclamp 3/8”
Triclamp 1/2”
Triclamp 1/2”
Triclamp 1”
Triclamp 1”
Triclamp 1”
Triclamp 1”

Polypropylene (FDA, USP Class VI,
BSE/TSE/Animal free)Gamma stable for up to 40 kGy.

See Levitronix® technical brochure of
LFS-SU single-use sensor series formore detailed specifications and forother configurations and sensor sizes.

Table 3: Specification of LEVIFLOW® single-use high-precision (1% accuracy of reading) flow sensors compatible with LCO-600 console.
Note 1: Extended calibration for wider 1% accuracy range.

Pos.

Note 2: All flow sensors available with gamma irradiation (see LEVIFLOW® product literature for more details).

Component

Article Name

Article #

6a
6b
6c
6d
6e

LEVIFLOW® Clamp-On
Flow Sensors LFSC-D

LFSC-05D-001 (1 lpm)
LFSC-08D-001 (4 lpm)
LFSC-12D-007 (20 lpm)
LFSC-22D-005 (80 lpm)
LFSC-30D-001 (160 lpm)

100-30407
100-30396
100-30390
100-30391
100-30414

7a
7b
7c
7d
7e
7f

LEVIFLOW® Clamp-On
Flow Sensors LFSC-iX

LFSC-i06X-002 (1 lpm)
LFSC-i10X-001 (4 lpm)
LFSC-i16X-001 (20 lpm)
LFSC-i19X-001 (50 lpm)
LFSC-i25X-001 (80 lpm)
LFSC-i35X-001 (160 lpm)

100-30485
100-30477
100-30482
100-30479
100-30480
100-30481

Calibrated Liquid

Water @ 20°C/37°C

TubingTube: ID x OD

Silicone
CFlex® 1

1/8” x 1/4”
1/4” x 3/8”
3/8” x 9/16”
3/4" x 1”
1” x 13/8”

See Levitronix® technical brochures of
LFSC-D and LFSC-iX of clamp-on
sensor series for more detailed
specifications and other sensor sizes.

1/8” x 1/4”
1/4” x 3/8”
3/8” x 5/8”
1/2" x 3/4"
3/4" x 1”
1” x 13/8”

Silicone
CFlex®
TPE

Water @ 20°C/37°C

Note

Table 4: Specification of LEVIFLOW® clamp-on flow sensors for flexible tubings compatible with LCO-i100 console.
Note 1: C-Flex® is a registered trademark of Saint Gobain Performance Plastics, 2015. All rights reserved.

Pos.
8a
8b
8c
8d
8e
8f
8g

Component

Article Name

Article #LEVIFLOW® Clamp-On
Flow Sensors LFSC-iX.1

LFSC-i05X.1-001 (0.2/1 lpm)
LFSC-i09X.1-001 (4 lpm)
LFSC-i13X.1-001 (20 lpm)
LFSC-i16X.1-001 (50 lpm)
LFSC-i20X.1-001 (80 lpm)
LFSC-i23X.1-001 (160 lpm)
LFSC-i36X.1-001 (400 lpm)

100-30490
100-30491
100-30492
100-30498
100-30499
100-30509
100-30487

Calibrated Liquid

SU Flow Pipe

Fitting, ID

Note

Water @ 20/37°C

LFP-05SU / 05SU.1
LFP-09SU
LFP-13SU
LFP-16SU
LFP-20SU
LFP-23SU / 23SU.1
LFP-36SU

Barb, 1/8”
Barb, 1/4"
Barb, 3/8”
Barb, 1/2"
Triclamp / 3/4"
Triclamp / 7/8+1”
Triclamp / 13/8”

See relevant Levitronix® technicalbrochures for more details on flow
pipes, specifications, other sensor sizes
and configurations.

Table 5: Specification of LEVIFLOW® clamp-on flow converters LFSC-ixxX.1 (for single-use flow pipes of the LFP-SU series) compatible with LCO-i100 console.

5c

6b

6a
5i

5d

5g
5e

5f

7b

7c

7d

6d

6c

5a
5b

7a

5j

5h
H

6e

7f

7e

8a

8e

8d

8c

8b

8f

8a

8g

8e

8b

8f

Figure 14: LEVIFLOW® flow sensors compatible with LCO-i100 console.

8d

8c

8g

ORDER INFORMATION

Pos.

Component

Article Name

Article #Fittings

Specification

Wet Material

Note

9a
9b
9c
9d

PendoTECH Single-Use
Pressure Sensors

PREPS-N-038
PREPS-N-5-5
PREPS-N-1-1
PREPS-N-15-15

190-10375
190-10373
190-10374
190-10478

3/8” Hose Barb
3/4" Sanitary Clamp
1" Sanitary Clamp
1.5” Sanitary Clamp 1

Pressure range:-0.79 to 5.2 bar

Polysulfone

See PendoTECH literature for moredetailed specifications.

10a
10b

Single-Use
Temperature Sensor

TEMPS-N-038
TEMPS-N-050

190-10377
190-10376

3/8” Hose Barb
1/2” Hose Barb

Temp. range:
0 – 70°C

Polysulfone
Stainless steel

See PendoTECH literature for moredetailed specifications.

Table 6: Specification of single-use pressure and temperature sensors compatible with console LCO-i100.
Note 1: ID is 13/8”.

9d
10b

9a

9c

10a

9b

Figure 15: Pressure sensors compatible with LCO-i100 console.

ORDER INFORMATION
Pos.

Component

11a

LEVIFLOW®
Interconnect Cable

Part #

Characteristics

Special Feature / Description

LFI-C.1-30 (3m)

190-10308

Cable Jacket Material
Main Purpose

PVC
Connection between LFSC-D and LFS-SU flow sensors and console.

11b

IP Cable Signal
4 Wires

ICS-3.1-30 (3m)

190-10447

Cable Jacket Material
Main Purpose

PUR
Connection between LFSC-iX/iX.1 flow sensors and console.

12

Pressure Sensor
Adaptor Cable

LPE-1.1-30 (3m)

190-10354

General Specifications
Main Purpose

Circular console connector, watertight on console side.
Connection of single-use pressure sensors to console.

13

Temperature Sensor
Adaptor Cable

LTE-1.1-30 (3m)

190-10353

General Specifications
Main Purpose

Circular console connector, watertight on console side.
Connect single-use temp. sensors to console. For barb sensors and dip probe.

14

IP Adaptor Cable Hybrid

ICH-1.2-30 (3m)

190-10329

General Specifications
Main Purpose

Circular console and driver side connector, watertight.
Extend cable of drivers.

15

IP Cable Signal
12 Wires

ICS-2.1-50 (5 m)

190-10347

Cable Material / Wires
Connection In / Out
Main Purpose

PVC jacket / 12x 0.14 mm2 and shielding
Connector with screw type plug for open wire connection / Circular Hirose type
General connection to PLC of console.

16a

Desktop AC/DC Power
SupplyVEC50US24 IC915

100-40019

Voltage Output / Input
Basic Dimensions
Certifications, Standards
Note

24VDC, 50W/ 90 – 264 VAC, 47-63 Hz
116 x 52 x 31 mm
IEC60950-1, EN60950-1, UL/cUL60950-1
Connector for direct connection to supply of console with cable length 1.2 m.

16b

Desktop AC/DC Power
Supply

GST160A24-R7B IC915

100-40020

Voltage Output / Input
Basic Dimensions
Safety Approvals
Note

24VDC, 160W
/ 85 – 264 VAC, 47-63 Hz
175 x 72 x 35 mm
UL60950-1, CSA C22.2, TUV EN60950-1
Connector for direct connection to supply of console with cable length 1.2 m.

AMC-1.1 (2m) / AMC-1.2 (2.5m)
AMC-1.3 / AMC-1.4 (2.5m)
AMC-1.5 (2.5m)

190-10331 / 32
190-10333 / 34
190-10335

CountryCountryCountry

US, Canada / Germ., Denm., Norw., Finl., Belg., Netherland, Sweden, Austria
Japan / Switzerland
United Kingdom

DDP-600-US48-SC

100-40026

Voltage Output / Input
Basic Dimensions
Certifications, Standards

48 VDC (600 W) / 100 – 277 VAC
250 x 125 x 60 mm / 2 x 200 mm
CB, CSA/UL, IP65

(for power supply 17a)

AMC-2.1 / AMC-2.2
AMC-2.3 / AMC-2.4
AMC-2.5

190-10336 /37
190-10338 /39
190-10338 /40

CountryCountryCountry
Cable Specifications

US, Canada / Germ., Denm., Norw., Finl., Belg., Netherl., Sweden, Austria
Japan / Switzerland
United Kingdom
Length = 3m, black color, watertight connector on power supply side

Specifications
IP Y-Split Cable

ICY-4.1-02

190-10476
Main Purpose

PVC cable jacket with PC split box, PUR sealing, circular type connectors.
Integrated DC/DC converter for 48 VDC to 24 VDC / 20 W.
Splits cable to connect power supply (3a) and console to driver IPD-600.

Specifications
Main Purpose

PVC jacket, connectors: circular type to circular type.
Fieldbus connection of driver from Y-split cable to console.

16c

17a

17b

18

19

Article NameAC Mains Cables(for Desktop power supply 16a/b)

AC/DC Power Supply(for Stand-Alone configuration)

AC Mains Cables

IP Cable Signal 4 Wires

ICS-1.5-30 (3 m)

190-10468

Table 7: Specification of standard cables and AC/DC desktop supplies for console.

13

12

11a

15

14

11b

16a

16b
16c

17b

18

17a

Figure 16: Accessories (Cables and AC/DC supplies).

19

NOTES PAGELEVITRONIX® THE COMPANY

Levitronix is the world-wide leader in magnetically levitated bearingless motor technology. Levitronix was the first company to introduce
bearingless motor technology to the Semiconductor, Medical and Life Science markets. The company is ISO 9001 certified. Production andquality control facilities are located in Switzerland. In addition, Levitronix is committed to bring other highly innovative products like the
LEVIFLOW® flowmeter series or the PuraLev® consoles to the market.

Headquarter andEuropean Contact

US Contact

Japan Contact

Taiwan Contact

Levitronix GmbH
Bändliweg 30
CH-8048 Zurich
Switzerland

Levitronix Technologies Inc.
10 Speen Street, Suite 102
Framingham, Massachusetts 01701
USA

Levitronix Japan K.K.
Wing Eight 5floor, 4-16-4
Asakusabashi, Taito-ku
Tokyo, 111-0053 Japan

Levitronix Taiwan
5F, No. 251, Dong Sec. 1, Guangming
6th Rd., Chu Pei City,
Hsin-Chu 302, Taiwan, R.O.C.

Phone:
E-Mail:Phone:
E-Mail:Phone:
E-Mail:Phone:
E-Mail:+41 44 974 4000
salesEurope@levitronix.com+1 508 861 3800
salesUS@levitronix.com+81 3 5823 4193
salesJapan@levitronix.com+886 3 657 6209
salesAsia@levitronix.com

This document and its content are the property of Levitronix® and shall not be reproduced, distributed, disclosed
or used for manufacturing or sale of Levitronix ® products without the expressed written consent of Levitronix®.
First Release: 20-Jul-2016

Last Update: 02-Feb-2024

PL-4061-00, Rev08, DCO# 24-013