High Vacuum
Pump

Turbo Molecular Pumping System [YTP series Ver.B]
YTP Series version B is a high vacuum pumping system with a ceramic ball bearing turbo molecular pump.
Vacuum pumping from atmospheric pressure to high vacuum is possible. ULVAC prepares two types, SA typewhich allows the automatic pumping by START / STOP switch and M type which is a manual operation. In
addition, the back pump can be selected from rotary pump as a standard specification, or dry scroll pump as
the other selection.

YTP-50MB-DRY

YTP-300SABFeatures

Applications

Preparing various models according to the application

Evacuation system from atmospheric pressure to highThe back pump can be selected from rotary pump as astandard specification, or dry scroll pump as the otherselection.

vacuumAdopting a superior power failure countermeasure

Evacuation system of various experiment and R&D
equipments
Vacuum substitution of high purity gas

A solenoid valve is equipped as standard on the fore
valve to prevent reverse flow and failure at power
failure.

Roughing evacuation system of UHV system

Possible to check the condition of rough vacuum pumpPossible to check the pressure during rough vacuum
pumping due to the Pirani vacuum gauge is equipped
as a standard parts.
Vacuum pumping to the Ultra High Vacuum area ispossible.
The installation of optional baking heater allows thepumping to the Ultra High Vacuum area by degassing
function. (※ As product specification value)

Evacuation system of analytical equipment

High Vacuum Pump

function

Turbo Molecular Pumping System[YTP series

Ver.B]

Specifications
YTP-50MB
YTP-50SAB

Model

YTP-50MB-DRY
YTP-50SAB-DRY
ICF114

Inlet port

ICF152

VG65

VG100

ISO63-K

ISO100-K
10-6PaUltimate pressure*1 *2Before baking

10-6PaPumping speed*3

After baking-810 Pa

10-8PaN270 L/s

280 L/s

H249 L/s

Outlet port
Baking pump

220 L/s

G3/4

NW16G1

NW1650Hz

50 L/min90 L/min

200 L/min

250 L/min

60Hz60 L/min

100 L/min

240 L/min

300 L/min

Pirani vacuum gauge（2.7×103 ～ 10-1Pa）

TMP back pressure and rough pressure monitering
Fore valve

Solenoid valve for rotary pump automatic vent system(Solenoid valve)

Input power source

Single phase AC100V（50/60Hz）

Input cable length

5m with 3P outlet

Power capacity

1.2 kVA0.9 kVA

Power consumption during stable operation

0.6 kVA

0.4 kVA

Weight55 kg

57 kg

*1
*3

YTP-300MB-DRY
YTP-300SAB-DRY

YTP-300MB
YTP-300SAB

Baking is only available with ICF flange type.
Without a protective net.

*2

1.7 kVA

1.6 kVA
1.0 kVA

86 kg

85 kg

A digit number is expressed.(Display only the digits of power.)

Dimensions
YTP-50 Ver.B(The figue below shows DRY specifications)YTP-300 Ver.B(The figue below shows RP specifications)YTP-50MB
YTP-50SAB

YTP-50MB-DRY
YTP-50SAB-DRY

A580

580

A'592

592

B

400

C1Unit：mm

YTP-300MB
YTP-300SAB

YTP-300MB-DRY
YTP-300SAB-DRY

A650650

A'665665

450

B450500

430430B'492542

C2412412

C1440

440

D311311

C2428428

E

6565

D311311

F7070

E6565

G160

160

F100100

H470

463

G100100

H576

576

www.ulvac.co.jp/en

ULVAC, Inc. Components Division
Overseas Sales in Japan TEL +81-467-89-2261
USA : ULVAC Technologies, Inc.
TEL +1-978-686-7550
GERMANY : ULVAC GmbHTEL +49-89-960909-0
CHINA : ULVAC (SHANGHAI) Trading Co.,Ltd. TEL +86-21-6127-6618
TAIWAN : ULVAC TAIWAN, Inc.
TEL +886-3-579-5688
KOREA : ULVAC KOREA, Ltd.
TEL +82-31-683-2922
SINGAPORE : ULVAC SINGAPORE PTE LTD TEL +65-6542-2700

PHILIPPINES : ULVAC SINGAPORE PTE LTD Philippines Branch TEL +63-46-437-7700
VIETNAM : ULVAC SINGAPORE PTE LTD Vietnam Representative Office TEL +84-28-3821-2367
INDONESIA : ULVAC SINGAPORE PTE LTD Indonesia Representative Office TEL +62-21-8990-5070
THAILAND : ULVAC (THAILAND) LTD
TEL +66-2-7388883
MALAYSIA : ULVAC MALAYSIA SDN. BHD.
TEL +60-3-5121-4700
INDIA : ULVAC, Inc., India Branch
TEL +91-40-27007006

This catalog is subject to change without notice.
This catalog is published in order to sell this product in overseas. Please see Japanese catalog when purchasing in Japan.

1551E-2/2019042000SP