NewLife® Intensity® 10High Flow Stationary Concentrator

TIME-PROVEN DESIGNVERSATILE OXYGEN DELIVERY
EASE OF SERVICE

O2 is what we do.

Trusted therapeutic oxygen solution designed
for medical equipment fleets world wide
Built-to-last Oxygen Delivery Innovation
Since being introduced in 2009, the New Life Intensity 10 has been a trusted high flow stationary concentrator
addressing patients’ oxygen delivery needs in over 100 countries including specialty applications in home andmedical care facilities.
Time-proven design

Versatile oxygen delivery

Ease of service• Industry-leading reliability—
greater than 99% reliability rateout of the box and less than 2%*
warranty repair rate
• Comprehensive warranty—3-year
warranty on the device• Durability—engineered with anoversized compressor and sieve
beds to minimize service• Oxygen delivery—up to 95.5%
concentration at up to 10 LPM,
meeting a wide range of patientprescriptions• High outlet pressure—the firstoxygen concentrator offering
20 psi to support specialty
applications and up to 200 feet ofoxygen tubing• Energy efficient—reduced power
consumption, minimizing theburden on the patient’s energy bill

• Minimal maintenance—provenquality design and performance tominimize required service• Modular design—fewer
components reduce time andeffort to service the device• Enhanced diagnostics—onboard
alarm alerts enable quick
troubleshooting*data on fileO2 is what we do.

NewLife Intensity 10 Specifications
FLOW RATE
2.0 – 10.0 LPM

POWER
600 W

DIMENSIONS
27.5 x 16.5 x 14.5 inches(69.9 x 41.9 x 36.8 cm)

MAXIMUM TUBING
200 ft (61 m)

WEIGHT
58 lbs (26.3 kg)
SOUND LEVEL
58 dB(A)WARRANTY
3 years
MAINTENANCE SCHEDULE
Felt Filter – 1 Year Replacement
Intake Filter – Clean Weekly

OUTLET PRESSURE
20 psig (138 kPA)

Accessories
NASAL CANNULA
Adult
7 ft. (2.1 m)
CU002-1
25 ft. (7.6 m) CU002-4

HUMIDIFIER BOTTLE
Item Number HU003-1
Flow Rate 6-15 LPM HU014-1

MASK
Adult, Concentration(Over the Ear Style)

7 ft. (2.1 m)
CU006-1

Pediatric
7 ft. (2.1 m)
CU002-7
Pediatric, Concentration
7 ft. (2.1 m)
CU007-1

OXYGEN TUBING
7 ft. (2.1 m)
CU004-2
25 ft. (7.6 m) CU004-3
50 ft. (15.2 m) CU004-1

OXYGEN OUTLET ADAPTOR
Item Numbers
Black F0025-1
Green B-776945-00

SureFlowTM

Cost effective oxygen delivery for multiple
patients from a single source
Multi-patient Flow Control
Combine the NewLife Intensity 10
with the multi-patient SureFlow
oxygen flow station to costeffectively serve multiple patientswith a single oxygen source.
SureFlow is a uniquely engineered
flow station that effectively
administers medical-grade oxygenfrom an oxygen concentrator or
another single oxygen source.
SureFlow is configured with five
low-flow flow meters, and can beused within a wide range of settings
to accommodate a broad varietyof patient needs, enabling oxygen
delivery for critical care up to10 LPM.
Positioning• Set on a table or desktop• Wall-mounted

Diverse applications• Pediatric-to-adult use in hospitals,
clinics, emergency shelters• Underdeveloped areas with
reduced to no infrastructure fortherapeutic oxygen delivery• Low flow pediatric oxygen needs
in a critical care nursery in low to
middle income countries

Ideal usage• Serve up to five individuals per
SureFlow unit
• Connect two or more SureFlow
units together to serve moreindividuals

SureFlow Specifications
FLOWMETERS
Low-flow flowmeters can be usedfrom 1/10 LPM to 1 LPM with.1/10 LPM increments with backlines for proper viewing angle
FLOWMETER ACCURACY±10% or 200mL whichever is greater
MAXIMUM INLET PRESSURE
20 psig (138 kPa)

DIMENSIONS
5.8 x 9.7 x 7.2 inches(14.7 x 24.6 x 18.3 cm)
WEIGHT
3.28 lb (1.49 kg)
WARRANTY
3 years on parts
INTERNATIONAL APPROVALS
CE 0459 (FM069-2 only)
ISO 13485:2016-certified

Find out more about NewLife Intensity 10and accessory products at

SEE PRODUCT WARRANTY STATEMENT FOR COMPLETE INFORMATION.
Please consult the applicable product instructions for use for product indications,contraindications, warnings, precautions, and detailed safety information.

CAIREinc.com© 2022 CAIRE Inc. All Rights Reserved. ML-CONC0084 H