Companion 5TM

User Manual (English)
Manuel d’utilisation (Français)Bedienungsanleitung (Deutsch)
Manual del usuario (Español)
Manuale dell’utente (Italiano)
Manual do utilizador (Português)Gebruikershandleiding (Nederlands)
Brugervejledning (Dansk)

Bruksanvisning (Norsk)
Bruksanvisning (Svenska)
Käyttöopas (Suomi)
Εγχειρίδιο χρήστη (Ελληνικά)Kullanıcı El Kitabı (Türkçe)Uživatelská příručka (Čeština)
Instrukcja obsługi (Polski)
Felhasználói kézikönyv (Magyar)Companion 5User Controls & System Status Indicators
Type BF applied part (degree ofprotection against electric shock).
Reg. # 5333

Internal Symbols

02

Oxygen Output

I/O On/Off Switch
No serviceable parts inside. Do notopen cover.

Council Directive 93/42/EEC; concerning
medical devices
Authorized representative in theEuropean Community
IEC 60417

ISO 7000; Graphical symbols for use on
equipment—Index and synopsis
Storage or operating temperature
range. Reg. # 0632

Class II Equipment, Double
Insulated Reg. # 5172
ETL Certification
Safety agency for CAN/CSA C22.2
No. 601.1 M90 for medical electricalequipment

Storage humidity range. Reg. # 2620
Name and address of manufacturer.
Reg. # 3082
Date of manufacturer. Reg. # 2497
Catalog Number. Reg. # 2493
Serial Number. Reg. # 2498

FCC Notice
21 CFR 801.15: Code of Federal Regulations
Title 21
Federal law restricts this device tosale by or on the order of a physician.
Council Directive 2012/19/EU: waste electricaland electronic equipment (WEEE)This way up. Reg. # 0623
Fragile, handle with care.
Reg. # 0621
Keep away from rain, keep dry.
Reg. # 0626
Read user’s manual before operation.
Reg. # 1641
Stacking limit by number. Reg #2403
ISO 7010: Graphical symbols—Safety colors
and safety signs—Registered safety signs
The instruction manual must be read.
Reg. # M002
Warning. Reg. # W001
Keep away from open flame, fire,
sparks. Open ignition source andsmoking prohibited. Reg. # P003
Do not smoke near unit or while
operating unit. Reg. # P002

2 - ENGPN 15062781-C4 J | User Manual

WEEE
IEC 60601-1: Medical electrical equipment Part
1 General requirements for basic safety andessential performance

IPX1

Drip Proof Equipment-IPX1: The
Companion 5 provides protection
against the harmful effects of
the ingress of liquids. (IPX1,
per IEC 60529)

FCC NOTICE:

This device may contain CYBLE-022001-00,
including the antenna 2450AT18B100 fromJohnson Technology, complies with par 15 of the
FCC rules. The device meets the requirements formodular transmitter approval as detailed in FCC
public Notice DA00-1407. Transmitter operation
is subject to the following two conditions (1) This
device may not cause harmful interference, and (2)This device must accept any interference received,
including interference that may cause undesired
operation.
This product may be covered by one or more
patents, US and international. Please visit our
website below for the listing of applicable patents.
Pat.: www.caireinc.com/corporate/patents/ .

Companion 5
Quick Start Guide
1 Unpack Your Companion 5

Companion 5

2 Getting to Know Your Companion 5
Review all Warnings, Cautions and additional
device information in the rest of this manual.
Become familiar with the key features of theCompanion 5 and the User Control Panel.

FRONT
Outer
Case

Power
SwitchLED
Display

OxygenOutlet
Barb

Hour
Meter

Flow
Meter
Humidifier
BottleSupportStand (Bottle
optional)

BACK
Carrying
Handle &Air Intake
Location

Cooling
Air Intake
Location

Air Intake
Location

PN 15062781-C4 J | User Manual

ExhaustVent

3 - ENG

Companion 5
The Companion 5 Components
Outer Case

Power Switch

Outlet Barb

LED Display

Hour Meter

Flow Meter

Humidifier
Bottle Support
Stand (Bottle
optional)

4 - ENG

Durable plastic case that encloses andprotects the inner components of the
CAIRE Companion 5.
ON/OFF power switch used to initiate
or stop power supply to the unit.
The ON position is indicated by the“l” symbol, and the OFF position isindicated by the “O” symbol.
Oxygen exits the Companion 5 here
after it has been filtered and concentrated. Either a single lumen nasal
cannula or standard oxygen tubing is
attached to deliver to the patient. The
maximum length of oxygen tubing
attached to the concentrator is 50 ft.
(15.2 m).
Contains a green, red and yellow* LED
light. The green light indicates normal
operation, and the yellow* and red
lights indicate alarms conditions. (Seethe Alarm Conditions section of the
instruction for more information.)
Digital display of the elapsed operation time of the concentrator. Displays
to the nearest tenth of an hour andcannot be reset. Displays diagnostic
alarm codes when the concentrator
experiences an alarm condition.
Used to adjust and display the flowrate of oxygen that is delivered. Flowrates range from 0.5 - 5 liters per
minute (LPM) and can be adjusted byturning the knob.
Location to attach a humidifier bottle.
The stand contains an elastic band
used to secure the bottle on the stand.
A humidifier bottle provides additional
moisture to the oxygen flow. The
bottle is filled with distilled water, andthe moisture is used to prevent dryingof the nasal tissues.
*Oxygen Concentrator Status
Indicator (OCSI) models only

PN 15062781-C4 J | User Manual

3 Attaching Humidifier Bottle andCannulaIf using a humidifier bottle, first fill the bottle with
distilled water to the manufacturer’s specified level.
Place bottle in designated support stand and securewith the elastic band. Connect the threaded endof the humidifier tube to the bottle, and the otherto the Companion 5 outlet barb. Connect oxygen
tubing and/or a nasal cannula to the outlet barb on
the humidifier bottle.

If not using a humidifier bottle, connect the oxygen
tubing and/or a nasal cannula directly to theCompanion 5 outlet barb.

Companion 5
4 Power On and Warm Up

6 Adjust Flow Setting to Prescribed
Level

Plug the AC power cord into an AC outlet.

Use the Flow Control Knob to select the flowprescribed by your physician.

Toggle the Power Switch to the On position.

Power
SwitchWhen the Companion 5 is powered on properly,
the green power indicator on the LED Display willlight up. All LED lights will illuminate upon startup. After the concentrator completes the warm-up
cycle, only the green light will remain on.
Note: After Initially powering on, the O2 light will
illuminate and “O2 UP” will display on the screen
until the concentration reaches specifications.

WARNING: IT IS VERY IMPORTANT TO
SELECT ONLY THE PRESCRIBED LEVEL OFOXYGEN. DO NOT CHANGE THE FLOWSELECTION UNLESS YOU HAVE BEEN
DIRECTED TO DO SO BY A LICENSED CLINICIAN. THE OXYGEN CONCENTRATOR
MAY BE USED DURING SLEEP UNDER
THE RECOMMENDATION OF A QUALIFIED
CLINICIAN.

Breathe normally through the nasal cannula.
To adjust flow rate:
Turn counter-clockwise to increase flow.
Turn clockwise to decrease flow.
The middle of the ball indicates flow rate. See
picture below.

Power
Indicator
LED

PN 15062781-C4 J | User Manual

5 - ENG

Companion 5
Important!

Safety Instructions are defined as follows:
WARNING: IMPORTANT SAFETY INFORMATION FOR HAZARDS THAT MIGHT
CAUSE SERIOUS INJURY.
CAUTION: Important information forpreventing damage to the Companion 5.
Note: Information needing special attention.

Indications for Use
Intended Use

The Companion 5 is intended for the administrationof supplemental oxygen. The device is not intended
for life support, nor does it provide any patientmonitoring capabilities.
WARNING: IT IS VERY IMPORTANT TO
SELECT ONLY THE PRESCRIBED LEVEL OFOXYGEN. DO NOT CHANGE THE FLOWSELECTION UNLESS YOU HAVE BEEN
DIRECTED TO DO SO BY A LICENSED
CLINICIAN. THE PORTABLE OXYGENCONCENTRATOR MAY BE USED DURINGSLEEP UNDER THE RECOMMENDATION
OF A QUALIFIED CLINICIAN.

WARNING: TO ENSURE RECEIVING THE
THERAPEUTIC AMOUNT OF OXYGEN DELIVERYACCORDING TO YOUR MEDICAL CONDITION,
COMPANION 5 MUST BE USED ONLY AFTER ONE
OR MORE SETTINGS HAVE BEEN INDIVIDUALLY
DETERMINED OR PRESCRIBED FOR YOU AT YOURSPECIFIC ACTIVITY LEVELS. COMPANION 5 MUST
BE USED WITH THE SPECIFIC COMBINATION OR
PARTS AND ACCESSORIES THAT ARE IN LINE WITH
THE SPECIFICATION OF THE CONCENTRATORMANUFACTURER AND THAT WERE USED WHILE
YOUR SETTINGS WERE DETERMINED.
WARNING: THIS UNIT IS NOT TO BE USED FOR LIFE
SUPPORT. GERIATRIC, PEDIATRIC, OR ANY OTHERPATIENT UNABLE TO COMMUNICATE DISCOMFORTCAN REQUIRE ADDITIONAL MONITORING AND/OR A DISTRIBUTED ALARM SYSTEM TO CONVEY
THE INFORMATION ABOUT THE DISCOMFORT AND
OR THE MEDICAL URGENCY TO THE RESPONSIBLE
CARE GIVER TO AVOID HARM. PATIENTS WITH
HEARING AND/OR SIGHT IMPAIRMENT(S) MAY
NEED ASSISTANCE WITH MONITORING ALARMS.

WARNING: PREGNANT OR NURSING WOMENSHOULD NOT USE ACCESSORIES RECOMMENDED IN
THIS MANUAL, THEY MY CONTAIN PHTHALATES.
WARNING: FEDERAL (USA) LAW RESTRICTS THIS
DEVICE TO SALE OR RENTAL BY ORDER OF APHYSICIAN OR OTHER LICENSED HEALTH CAREPROVIDER.

Contraindications for Use
WARNING: IN CERTAIN CIRCUMSTANCES,
THE USE OF NON-PRESCRIBEDOXYGEN CAN BE HAZARDOUS. THIS
DEVICE SHOULD ONLY BE USED WHEN
PRESCRIBED BY A PHYSICIAN.

WARNING: NOT FOR USE IN THE PRESENCE OF
FLAMMABLE ANESTHETICS.
WARNING: AS WITH ANY ELECTRICALLY POWEREDDEVICE, THE USER MAY EXPERIENCE PERIODS OFNON-OPERATION AS A RESULT OF ELECTRICAL
POWER INTERRUPTION, OR THE NEED TO HAVE
THE COMPANION 5 SERVICED BY A QUALIFIEDTECHNICIAN. THE COMPANION 5 IS NOTAPPROPRIATE FOR ANY PATIENT WHO WOULD
EXPERIENCE ADVERSE HEALTH CONSEQUENCES AS
THE RESULT OF SUCH TEMPORARY INTERRUPTION.

Safety Guidelines
WARNING: NO MODIFICATION OF THISEQUIPMENT IS PERMITTEDWARNING: THE MANUFACTURER RECOMMENDS AN
ALTERNATE SOURCE OF SUPPLEMENTAL OXYGENIN THE EVENT OF A POWER OUTAGE, ALARM
CONDITION, OR MECHANICAL FAILURE. CONSULTYOUR PHYSICIAN OR EQUIPMENT PROVIDER FOR
THE TYPE OF RESERVE SYSTEM REQUIRED.
WARNING: THIS DEVICE SUPPLIES HIGHCONCENTRATION OXYGEN THAT PROMOTESRAPID BURNING. DO NOT ALLOW SMOKING OR
OPEN FLAMES WITHIN TWO (2) METERS OF (1)THIS DEVICE, OR (2) ANY OXYGEN-CARRYINGACCESSORY. FAILURE TO OBSERVE THIS WARNINGCAN RESULT IN SEVERE FIRE, PROPERTY DAMAGE,
AND/OR CAUSE PHYSICAL INJURY OR DEATH.
WARNING: DO NOT OPERATE UNIT IN A
RESTRICTED OR CONFINED SPACE (I.E., A SMALL
CASE OR HANDBAG) WHERE VENTILATION CAN
BE LIMITED. THIS CAN CAUSE THE OXYGENCONCENTRATOR TO OVERHEAT AND IMPAIRPERFORMANCE.
WARNING: THE CONCENTRATOR SHOULD
BE LOCATED AS TO AVOID SMOKE,
POLLUTANTS OR FUMES.

6 - ENGPN 15062781-C4 J | User ManualCompanion 5
WARNING: THE USE OF SOME OXYGENADMINISTRATING ACCESSORIES NOT SPECIFIEDFOR THIS OXYGEN CONCENTRATOR MAY IMPAIR
ITS PERFORMANCE. RECOMMENDED ACCESSORIESARE REFERENCED WITHIN THIS MANUAL.
WARNING: IF THE OXYGEN CONCENTRATOR HAS
BEEN DROPPED, DAMAGED OR EXPOSED TO WATER
PLEASE CONTACT YOUR HOME CARE PROVIDERFOR INSPECTION OR POSSIBLE REPAIR OF THE
DEVICE. DO NOT USE THE OXYGEN CONCENTRATORIF IT HAS A DAMAGED POWER CORD OR PLUG.
WARNING: DO NOT ALLOW EITHER THE AIR INTAKE
OR THE AIR OUTLET VENTS TO BECOME BLOCKED.
DO NOT DROP OR INSERT ANY OBJECTS INTOANY OPENINGS ON THE DEVICE. THIS CAN CAUSETHE OXYGEN CONCENTRATOR TO OVERHEAT AND
IMPAIR PERFORMANCE.

WARNING: THE MANUFACTURER
RECOMMENDS AN ALTERNATE
SOURCE OF SUPPLEMENTAL OXYGENIN THE EVENT OF A POWER OUTAGE,
ALARM CONDITION, OR MECHANICALFAILURE. CONSULT YOUR PHYSICIAN OR
EQUIPMENT PROVIDER FOR THE TYPE OF
RESERVE SYSTEM REQUIRED.
WARNING: THIS DEVICE SUPPLIES HIGHCONCENTRATION OXYGEN THAT PROMOTESRAPID BURNING. DO NOT ALLOW SMOKING OR
OPEN FLAMES WITHIN THE SAME ROOM OF (1)THIS DEVICE, OR (2) ANY OXYGEN-CARRYINGACCESSORY. FAILURE TO OBSERVE THIS WARNINGCAN RESULT IN SEVERE FIRE, PROPERTY DAMAGE,
AND/OR CAUSE PHYSICAL INJURY OR DEATH.

WARNING: DO NOT OVERFILL THE OPTIONAL
HUMIDIFIER. FILL THE OPTIONAL HUMIDIFIERWITH WATER ONLY TO THE LEVEL SHOWN BY THE
MANUFACTURER OF THE HUMIDIFIER.

WARNING: THE USE OF SOME OXYGENADMINISTRATING ACCESSORIES NOT SPECIFIEDFOR THIS OXYGEN CONCENTRATOR MAY IMPAIR
ITS PERFORMANCE. RECOMMENDED ACCESSORIESARE REFERENCED WITHIN THIS MANUAL.

WARNING: DO NOT OPERATE THIS EQUIPMENTWITHOUT FIRST READING AND UNDERSTANDING
THIS MANUAL. IF YOU ARE UNABLE TO
UNDERSTAND THE WARNINGS AND INSTRUCTIONS,CONTACT YOUR EQUIPMENT PROVIDER BEFOREATTEMPTING TO USE THIS EQUIPMENT; OTHERWISE,
INJURY OR DAMAGE MAY RESULT.

WARNING: IF THE OXYGEN CONCENTRATOR HAS
BEEN DROPPED, DAMAGED OR EXPOSED TO WATER
PLEASE CONTACT YOUR HOME CARE PROVIDERFOR INSPECTION OR POSSIBLE REPAIR OF THE
DEVICE. DO NOT USE THE OXYGEN CONCENTRATORIF IT HAS A DAMAGED POWER CORD OR PLUG.

WARNING: IF YOU FEEL DISCOMFORT OR ARE
EXPERIENCING A MEDICAL EMERGENCY, SEEKMEDICAL ASSISTANCE IMMEDIATELY.

WARNING: USE ONLY SPARE PARTSRECOMMENDED BY THE MANUFACTURER TO
ENSURE PROPER FUNCTION AND TO AVOID THE
RISK OF FIRE AND BURNS.

WARNING: OPERATING THE OXYGENCONCENTRATOR OUTSIDE OF THE OPERATIONALTEMPERATURE SPECIFICATIONS CAN LIMIT THE
CONCENTRATOR’S ABILITY TO MEET OXYGENCONCENTRATION SPECIFICATION. REFER TO THE
SPECIFICATION SECTION OF THIS MANUAL FORTEMPERATURE LIMITS.
WARNING: USE NO OIL, GREASE, OR PETROLEUMBASED OR OTHER FLAMMABLE PRODUCTS WITH
THE OXYGEN-CARRYING ACCESSORIES OR THE
OXYGEN CONCENTRATOR. ONLY WATER BASED,OXYGEN COMPATIBLE LOTIONS OR SALVES SHOULD
BE USED. OXYGEN ACCELERATES THE COMBUSTION
OF FLAMMABLE SUBSTANCES.
WARNING: THE OXYGEN CONCENTRATOR SHOULD
NOT BE USED ADJACENT TO OR STACKED WITH
OTHER EQUIPMENT. IF ADJACENT OR STACKED
USE IS UNAVOIDABLE, THE DEVICE SHOULD BE
OBSERVED TO VERIFY NORMAL OPERATION.

WARNING: DO NOT LUBRICATE FITTINGS,CONNECTIONS, TUBING OR OTHER ACCESSORIES OF
THE OXYGEN CONCENTRATOR TO AVOID THE RISKOF FIRE AND BURNS.
WARNING: SMOKING WHILE USING OXYGEN IS
THE NUMBER ONE CAUSE OF FIRE INJURIES AND
RELATED DEATHS. YOU MUST FOLLOW THESE
SAFETY WARNINGS:WARNING: DO NOT ALLOW SMOKING, CANDLES,OR OPEN FLAMES IN THE SAME ROOM WITH THE
DEVICE OR THE OXYGEN-CARRYING ACCESSORIES.
WARNING: SMOKING WHILE WEARING AN OXYGENCANNULA MAY CAUSE FACIAL BURNS AND
POSSIBLY DEATH.
WARNING: IF YOU SMOKE, THESE 3 STEPSMAY SAVE YOUR LIFE: TURN OFF THE OXYGENCONCENTRATOR, TAKE OFF THE CANNULA, ANDLEAVE THE ROOM WHERE THIS DEVICE IS LOCATED.
WARNING: “NO SMOKING – OXYGEN IN USE”
SIGNS MUST BE PROMINENTLY DISPLAYED IN THE
HOME, OR WHERE OXYGEN IS IN USE. PATIENTSAND THEIR CAREGIVERS MUST BE INFORMEDABOUT THE DANGERS OF SMOKING IN THE
PRESENCE OF, OR WHILE USING, MEDICAL OXYGEN.

PN 15062781-C4 J | User Manual

7 - ENG

Companion 5
WARNING: DO NOT USE YOUR OXYGENCONCENTRATOR IN THE PRESENCE OF FLAMMABLE
GASES. THIS CAN RESULT IN RAPID BURNINGCAUSING PROPERTY DAMAGE, BODILY INJURIESOR DEATH.
WARNING: REMOVING THE CANNULA AND
PUTTING IT ON CLOTHING, BEDDING, SOFAS, OR
OTHER CUSHION MATERIAL WILL CAUSE A FLASHFIRE WHEN EXPOSED TO A CIGARETTE, HEATSOURCE, OR FLAME.
WARNING: DO NOT LEAVE A NASAL CANNULA ON
OR UNDER CLOTHING, BED COVERINGS OR CHAIRCUSHIONS. IF THE UNIT IS TURNED ON BUT NOT
IN USE, THE OXYGEN WILL MAKE THE MATERIALFLAMMABLE. SET THE I/0 POWER SWITCH TO
THE 0 (OFF) POSITION WHEN THE OXYGENCONCENTRATOR IS NOT IN USE.

WARNING: DO NOT ATTEMPT ANY
MAINTENANCE OTHER THAN THE
POSSIBLE SOLUTIONS LISTED WITHIN
THIS MANUAL. DO NOT REMOVE
COVERS, ONLY YOUR EQUIPMENTPROVIDER OR A QUALIFIED SERVICETECHNICIAN SHOULD REMOVE THE
COVERS OR SERVICE THIS DEVICE.
WARNING: ONLY ACCESSORIES RECOMMENDED BY
THE MANUFACTURER. USE OF ANY OTHER MAY BE
HAZARDOUS, CAUSE SERIOUS DAMAGE TO YOUROXYGEN CONCENTRATOR AND WILL VOID THE
WARRANTY.
CAUTION: Always disconnect AC power
supply from the wall before disconnection the AC power supply from theoxygen concentrator.
CAUTION: Always place oxygen supply tubing
and power cords in a manner that prevents a trip
hazard.
CAUTION: To prevent a void warranty,
follow all manufacturers’ instructions.
Note: Portable and mobile RF communications equipment can effect medical electrical equipment.

WARNING: ALWAYS PLACE THE OXYGEN SUPPLYTUBING AND POWER CORDS IN A MANNER
THAT PREVENTS TRIP HAZARD OR POSSIBLE
ACCIDENTAL STRANGULATION.

8 - ENGPN 15062781-C4 J | User Manual

WARNING: THIS PRODUCT CAN EXPOSE YOU
TO CHEMICALS INCLUDING NICKEL, WHICH
IS KNOWN TO THE STATE OF CALIFORNIA TO
CAUSE CANCER. FOR MORE INFORMATION,
GO TO WWW.P65WARNINGS.CA.GOV.
WARNING: IN THE EVENT THERE IS A SERIOUS
INCIDENT OCCURRING WITH THIS DEVICE,THE USER SHOULD IMMEDIATELY REPORT THE
INCIDENT TO THE PROVIDER AND/OR THE MANUFACTURER. A SERIOUS INCIDENT IS DEFINEDAS AN INJURY, DEATH, OR POTENTIAL TO CAUSE
INJURY/DEATH SHOULD THERE BE A REOCCURRENCE OF THE INCIDENT. THE USER CAN ALSOREPORT THE INCIDENT TO THE COMPETENTAUTHORITY IN THE COUNTRY WHERE THE INCIDENT OCCURRED.

Companion 5
Specifications
For proper use of the device, the following chart
provides important information concerning therecommended operating environments, or operating
conditions.

Specifications
Flow Rates
O2
Concentration
Dimensions
WeightSound
Pressure Level
Power
Consumption
OperatingTemperature
Operating
Humidity
Storage
Temperature
Storage Humidity
Electrical

0.5–5.0 LPM
90% (+5.5%/-3%)
21.5x12.5x13.5” (54.6x31.8x34.2 cm)
36 lbs (16.3 kg)
50 dB(A) at 2.0 LPM
285 W @ 2 LPM, 350 W Maximum
41º F to 104º F (5° to 40° C)
15% - 90% at an 82.4° F (28° C)
dew point-13º F to 158º F (-25º C to 70º C)
0% - 90% Non-Condensing

Use no extension cords. Use no electrical outlets controlled by a switch.
Operating-1253–9879 feet
Altitude(-382–3011 meters)
Operating Time up to 24-hours a day

The expected service life of the equipment is aminimum of five years.
WARNING: USE OF DEVICE OUTSIDE OF
SPECIFIED OPERATING CONDITIONS IS
EXPECTED TO ADVERSELY AFFECT THE
FLOWRATE AND PERCENTAGE OF OXYGENAND CONSEQUENTLY THE QUALITY OF
THE THERAPY.

rear and underneath the cabinet, and also allow forexhaust air to freely leave the exhaust vent at thebottom left of the device.
Air intakes are located on the upper back andbottom of the concentrator. Locate the unit so that
there is at least 12 inches (30 cm) of space between
the concentrator and any walls, furniture, curtains,or other obstructions.
Place the device such that the alarms can be heard.
Position the oxygen supply tubing in such a waythat it does not kink or occlude.
WARNING: DO NOT USE YOUR OXYGENCONCENTRATOR IN THE PRESENCE OF
FLAMMABLE GASES. THIS CAN RESULTIN RAPID BURNING CAUSING PROPERTYDAMAGE, BODILY INJURIES OR DEATH.
USE NO OIL, GREASE, OR PETROLEUMBASED OR OTHER FLAMMABLE
PRODUCTS WITH THE OXYGENCARRYING ACCESSORIES OR THE
OXYGEN CONCENTRATOR. ONLY WATERBASED, OXYGEN COMPATIBLE LOTIONSOR SALVES SHOULD BE USED. OXYGENACCELERATES THE COMBUSTION OF
FLAMMABLE SUBSTANCES.

WARNING: THIS DEVICE SUPPLIES HIGHCONCENTRATION OXYGEN THAT PROMOTESRAPID BURNING. DO NOT ALLOW SMOKING OR
OPEN FLAMES WITHIN THE SAME ROOM OF (1)THIS DEVICE, OR (2) ANY OXYGEN-CARRYINGACCESSORY. FAILURE TO OBSERVE THIS WARNINGCAN RESULT IN SEVERE FIRE, PROPERTY DAMAGE,
AND/OR CAUSE PHYSICAL INJURY OR DEATH.

Note: If the Oxygen Concentrator has been stored foran extended period of time outside its normal operating temperature range, the unit should be allowed toreturn to normal operating temperature before being
turned on. (Refer to the Specifications section in thismanual.)Proper Placement of the Companion 5

Select a location for the device that avoids theintake of smoke, fumes and pollutants. Correct
placement of the device should allow intake of
air through the three air intake locations at the top
PN 15062781-C4 J | User Manual

9 - ENG

Companion 5
Operating Instructions
Before OperatingThis user manual serves as your reference to help
you operate and maintain the device. If you haveany questions or concerns please call your homecare provider.
Important! DO NOT attempt to operate theCompanion 5 without first reading the Safety
Guidelines section of this manual. Please follow
all of the operating instructions. Please observe all
Warnings on the device and in the Users Manual. In
order to reduce the risk of fire, personal injury andserious damage to the Companion 5, please observe
all of the safety precautions.
The CAIRE Companion 5 does not create its
own oxygen. It produces highly concentrated
oxygen from surrounding room air and delivers
it to you. As it naturally exists, room air contains
approximately only 21% oxygen by volume.
Room air is drawn into the concentrator by acompressor. The air then travels through a seriesof filters and a molecular sieve bed that chemicallyadsorbs nitrogen molecules. As a result, theoxygen-enriched air that exits the concentrator isdelivered at 87–95% oxygen by volume.
WARNING: CLEAN THE CABINET,CONTROL PANEL, AND POWER CORD
ONLY WITH A MILD HOUSEHOLDCLEANER APPLIED WITH A DAMP (NOT
WET) CLOTH OR SPONGE, AND THEN
WIPE ALL SURFACES DRY. DO NOTALLOW ANY LIQUID TO GET INSIDE
THE DEVICE. PAY SPECIAL ATTENTION
TO THE OXYGEN OUTLET FOR THE
CANNULA CONNECTION TO MAKE SURE
IT REMAINS FREE OF DUST, WATER, AND
PARTICLES.

CAUTION: Do not allow either the
air intake or the air outlet vents tobecome blocked. This can cause the
Oxygen Concentrator to overheat andimpair performance. Do not operate the
Oxygen Concentrator without the air
intake filter in place. If a second filter
is provided, insert the “replacement”
filter before you clean the dirty filter.
Clean the dirty filter in a warm soap
and water solution then dry thoroughly
prior to use.

10 - ENG

PN 15062781-C4 J | User Manual

Step 1: Positioning Your Companion 5 for Use
Place the Companion 5 in a well-ventilated, well
lit area. Be sure the air inlet and exhaust vents are
not obstructed. Position the Companion 5 so that
all audible and visual indicators or alarms can beeasily seen and heard.
Plug the device into an AC Power outlet.
Step 2: If the Humidifier Bottle WILL NOT be Used

Air Intake
Location

Cooling
Air Intake
Location

ExhaustVent
Air Intake
Locationa. Connect a nasal cannula or oxygen tubing
directly to the oxygen outlet barb.

b. Proceed to step 4.
Step 3: If the Humidifier Bottle WILL be Useda. Unscrew the lid from the humidifier bottle.
b. Fill the bottle using distilled water. Make surethe water level is between the manufacturer’s
specified maximum and minimum levels shown on
the bottle.
c. Secure the lid of the humidifier bottle. Ensurethat there are no leaks.
d. Connect a nasal cannula or standard oxygen
tubing to the outlet barb on the humidifier bottle.
Note: Use of a humidifier not specified for use withthis concentrator might affect its performance.

Companion 5
g. Place the bottle in its designated space on the
Humidifier Bottle Support Stand.
h. Secure the bottle on the stand using the elastic
band.

e. Connect the threaded end of the humidifier tube
to the lid of the humidifier bottle.

i. Proceed to step 4.
Step 4: Power On and Warm Up
a. Plug the electrical cord of the CAIRE
Companion 5 into an AC outlet.
WARNING: INSPECT THE ELECTRICAL
CORD FOR DAMAGE BEFORE USE.
IF THE CORD IS DAMAGED, DO
NOT PLUG IT INTO AN ELECTRICAL
OUTLET OR ATTEMPT TO OPERATE THE
CONCENTRATOR.

f. Connect the other end of the humidifier tube to
the outlet barb of the CAIRE Companion 5.

b. Press the power switch in the “ON” (I) position.
When the Companion 5 is powered on properly,
the green indicator on the LED Display will light
up. All LED lights will illuminate upon start-up.
After the concentrator completes the warm-up
cycle, only the green light will remain on.
Note: After initially powering ON the device, pleaseallow up to ten (10) minutes for the device to reach its
performance specifications.

PN 15062781-C4 J | User Manual

11 - ENG

Companion 5
WARNING: IT IS VERY IMPORTANT TO
SELECT ONLY THE PRESCRIBED LEVEL OFOXYGEN. DO NOT CHANGE THE FLOWSELECTION UNLESS YOU HAVE BEEN
DIRECTED TO DO SO BY A LICENSED CLINICIAN. THE OXYGEN CONCENTRATOR
MAY BE USED DURING SLEEP UNDER
THE RECOMMENDATION OF A QUALIFIED
CLINICIAN.

Step 6: Verify Flow and Breathe Normally
Step 5: Adjust Flow Control Ratea. Turn the flow control knob to the oxygen flowrate (LPM) prescribed by your physician.

a. Verify that oxygen is indeed flowing through thenasal cannula and that there are no kinks, bends,
or blockages in the tubing.
Note: Ensure the cannula is fully inserted and secure.
You should hear or feel oxygen flow to the prongs ofthe nasal cannula.

b. Properly position your nasal cannula and breathe
normally through the nasal cannula.
Step 7: Turn Off
a. Press the power switch in the “OFF” (O) position
when the CAIRE Companion 5 is no longer inuse.

To adjust flow rate:
Turn counter-clockwise to increase flow.
Turn clockwise to decrease flow.
The middle of the ball indicates flow rate. The
image below indicates a flow rate of 4.0 LPM.

CAUTION: Use only accessories recommended by the manufacturer. Use of any
other may be hazardous, cause serious
damage to your oxygen concentrator
and will void the warranty.
CAUTION: Do not use extension cords with this unit
or connect too many plugs into the same electricaloutlet. The use of extension cords could adversely
affect the performance of the device. Too manyplugs into one outlet can result in an overload tothe electrical panel causing the breaker/fuse to activate or fire if the breaker or fuse fails to operate.
CAUTION: Use of cables and adapters other thanthose specified, with the exception of cables andadapters sold by the manufacturer of the medical
electrical equipment as replacement parts forinternal components, may result in increased
emissions of decreased immunity of the Oxygen
Concentrator.
WARNING: DO NOT ALLOW EITHER THE
AIR INTAKE OR THE AIR OUTLET VENTSTO BECOME BLOCKED. THIS CAN CAUSETHE OXYGEN CONCENTRATOR TO OVERHEAT AND IMPAIR PERFORMANCE.

12 - ENG

PN 15062781-C4 J | User ManualCompanion 5
Note: To Equipment Provider: The following oxygen
administration accessories are recommended for use
with the Companion 5:• Nasal Cannula: CAIRE Part Number CU002-1• Humidifier Adaptor Tubing: CAIRE Part number
20843882• Humidifier Bottle: CAIRE Part Number HU003-1• Firebreak: CAIRE Part Number 20629671
A firebreak is required for use with any cannula.
• CAIRE offers a firebreak intended to be used in conjunction with the oxygen concentrator. The firebreak is
a thermal fuse to stop the flow of gas in the event thatthe downstream cannula or oxygen tubing is ignited
and burns to the firebreak. It is placed in-line with thenasal cannula or oxygen tubing between the patient
and the oxygen outlet of the Companion 5. For properuse of the firebreak, always refer to the manufacturer’s instructions (included with each firebreak kit).
• For any additional recommended accessories, please
see the Accessories Catalog (PN MLLOX0010) available on www.caireinc.com.

PN 15062781-C4 J | User Manual

13 - ENG

Companion 5
Alarm Conditions
Your CAIRE Companion 5 uses a combination of
two or three (for models with OCSI) LED lights
and an audible alarm to alert you when there is amalfunction with your concentrator.
1) Green LED (Green)- Indicates normal operation.

Red2) Red LED ( ) - Indicates system malfunction.
See Alarm Conditions Table below for moredetails.

Yellow

3) Yellow LED () - Indicates low oxygen concentration. See Alarm Conditions Table below
for more details.
The Alarm Conditions Table on the following pageshows the different alarm conditions that can bedisplayed by the concentrator. It outlines possible
causes and the actions you should take if thealarm is experienced. In the case of any alarm, it
is recommended that you contact your healthcare
provider as soon as possible.
WARNING: DO NOT IGNORE ALARMSWARNINGS: THE MANUFACTURER RECOMMENDS AN ALTERNATE SOURCE OF
SUPPLEMENTAL OXYGEN IN THE EVENTOF A POWER OUTAGE, ALARM CONDITION, OR MECHANICAL FAILURE. CONSULT YOUR PHYSICIAN OR EQUIPMENTPROVIDER FOR THE TYPE OF RESERVESYSTEM REQUIRED.

14 - ENG

PN 15062781-C4 J | User Manual

Above: Companion 5

Companion 5
Alarm Conditions Table
AudibleAlarm

ColoredLED

Off(Green)Cause
The concentrator is
working properly.

The concentrator has
lost power but thepower switch is still in
the “ON” position.

Intermittent

Your Action
None

Verify that the concentrator’s electrical cord is plugged
into an outlet and that the outlet has power. Try adifferent outlet. If the problem continues, connect toa back-up oxygen supply and contact your healthcare
provider.

(Red)

Intermittent

System Malfunction

Turn the power switch in the “OFF” position
immediately. Disconnect the AC Power from the walloutlet. Wait 5 minutes. Connect the AC Power back into
the wall outlet and turn the concentrator back on again.
If the alarm continues, service is required. Connect toa backup oxygen supply and contact your healthcare
provider immediately.

Product Flow Rate Too
High or Too Low

1. Ensure that the cannula is not kinked or blocked. If
used with a humidifier bottle, ensure that it is filledproperly and not creating a blockage.
2. Ensure that the Companion 5 has proper ventilation. It
needs to be at least 12 inches (30 cm) from any surface
to ensure the vents are not blocked.
3. If the problem persists, switch to an alternate sourceof oxygen and contact healthcare provider for assistance.

(Red)

Intermittent(Red)

Intermittent(Yellow)

The Companion 5 hasdetected low oxygen
levels.

The Companion 5 hasdetected low oxygen
levels.

Intermittent

1. Ensure the air intake filter and exhaust locations are
not clogged or restricted. 2. Ensure the Companion 5 isin a well-ventilated area. Make sure there are at least
12 inches between the back and sides of the Companion
and any obstructions (furniture, curtain, etc.).
3. If the condition persists, switch to an alternate
source of oxygen and contact your healthcare provider
immediately.

(Yellow)

PN 15062781-C4 J | User Manual

15 - ENG

Companion 5
Troubleshooting
It is recommended that you contact your healthcare
provider if your concentrator displays any
alarms or if you feel that your concentrator is
not working properly. You should not attempt to
repair the concentrator yourself. Repair should
only be performed by a trained service technician.

However, there are some troubleshooting steps thatyou can take if you experience problems with your
CAIRE Companion 5. They are outlined in thefollowing table. You can use this chart by following
the checks for your problem in numerical order.

User Troubleshooting Table
ProblemThe
concentrator
is turned on,
but it is notrunning.

You areexperiencing
low or nooxygen flow.

Possible Cause

Your Action

1. The electrical cord is not plugged in
to an outlet.

1. Plug in the electrical cord.

2. The electrical outlet is not providing
power.

2. Check your household fuse and circuit. Try a different
outlet.

3. Internal failure.

3. Connect to a back-up oxygen supply. Contact yourhealthcare provider immediately.

1. The flow control setting has beenchanged.

1. Verify that the flow control knob is set on yourappropriate Liter Per Minute (LPM) setting prescribed byyour doctor.

2. Tubing has been disconnected.

2. Verify that the tubing is tight at all connections (outlet
barb, humidifier bottle, water trap, etc). Re-secure any
connections if necessary.

3. The oxygen tubing or cannula iskinked.

3. Check for kinks or blockages in your tubing. Repair ifneeded.

4. There is a leak in your cannula or
tubing.

4. Inspect all tubing and the cannula for leaks. You cando this by either listening or feeling for escaping air.
Replace leaky tubing or cannula.

5. Air flow into the concentrator isimpeded.

5. Verify that nothing is blocking the inlets of theconcentrator. Be sure it is at least 12 inches (30 cm)
away from furniture, walls, or other obstructions.

6. Internal failure.

6. Connect to a back-up oxygen supply. Contact yourhealthcare provider immediately.

Note: If your Companion 5 is still alarming after using the Alarm Conditions and Troubleshooting tables, pleasecontact Technical Service at 1-800-482-2473 with the alarm code displayed on the LCD Display.

16 - ENG

PN 15062781-C4 J | User ManualCompanion 5
Cleaning, Care & Routine MaintenanceRoutine MaintenanceRoutine maintenance consists of changing thehandle gross particle filter once per week (ifinstalled).

Note: The Manufacturer does not recommend thesterilization of this equipment.
Note: Do not operate the Oxygen Concentrator
without the air intake filter in place.

User Care and Cleaning of the Device

Cannula Replacement
Replace your supply tubing and cannula on aregular basis as recommended by your home careprovider. Your physician or home care provider
will provide you with cleaning, disinfection andreplacement information.
Note: Always follow the cannula manufacturer’s
instructions for proper use. Replace the disposable
cannula as recommended by the cannula manufacturer
or your Equipment Provider. Additional supplies areavailable from your Equipment Provider.

Air Intake Filter (Cartridge)
The Air Intake Filter should only be accessed orchanged by your healthcare provider.
Gross Particle Filter:
The gross particle filter is located on the very front
bottom of the Companion 5.

GrossParticle
Filter

You should inspect and clean it once weekly. To
clean the filter, use the following process:

Humidifier Bottle (if Applicable)
You should check your humidifier bottle daily
to ensure that its water level is between thespecified minimum and maximum levels. Refill
the bottle as necessary using distilled water.
Clean the humidifier bottle in accordance with the
manufacturers’ instructions.
Cleaning Procedure:Turn OFF the Companion 5 and disconnect fromAC power before any cleaning or disinfecting
activity. DO NOT spray the outer case directly.
Use a damp (not wet) cloth or sponge. Spray thecloth or sponge with a mild detergent solution toclean the cabinet and power cord. To disinfect theCompanion 5, use Lysol® Brand II disinfectant (orequivalent). Dilute as directed by the manufacturerof the cleaning product, but do not spray liquid
directly on the Companion 5.
WARNING: ELECTRICAL SHOCK HAZARD.
TURN OFF THE UNIT AND DISCONNECT
THE POWER CORD FROM THE
ELECTRICAL OUTLET BEFORE YOU CLEAN
THE UNIT TO PREVENT ACCIDENTAL
ELECTRICAL SHOCK AND BURN HAZARD.
ONLY YOUR EQUIPMENT PROVIDER
OR A QUALIFIED SERVICE TECHNICIANSHOULD REMOVE THE COVERS ORSERVICE THE UNIT.
WARNING: ELECTRICAL SHOCK HAZARD. TURN OFF
THE UNIT AND DISCONNECT THE POWER CORD
FROM THE ELECTRICAL OUTLET BEFORE YOU CLEAN
THE UNIT TO PREVENT ACCIDENTAL ELECTRICALSHOCK AND BURN HAZARD. ONLY YOUREQUIPMENT PROVIDER OR A QUALIFIED SERVICETECHNICIAN SHOULD REMOVE THE COVERS ORSERVICE THE UNIT.

1) Remove the gross particle filter.
2) Wash the filter in warm tap water using a mild
soap detergent solution.
3) Rinse the filter thoroughly with clear tap waterand squeeze out the excess water.
4) Allow the filter to air dry.
5) Reinsert the filter in the cabinet.
PN 15062781-C4 J | User Manual

17 - ENG

Companion 5
WARNING: ELECTRICAL SHOCK HAZARD.
DISCONNECT THE POWER CORD FROM THE
ELECTRICAL OUTLET BEFORE YOU CLEAN THE UNIT
TO PREVENT ACCIDENTAL ELECTRICAL SHOCK AND
BURN HAZARD. ONLY YOUR EQUIPMENT PROVIDER
OR A QUALIFIED SERVICE TECHNICIAN SHOULDREMOVE THE COVERS OR SERVICE THE UNIT.
WARNING: CARE SHOULD BE TAKEN TO PREVENTTHE OXYGEN CONCENTRATOR FROM GETTING WET
OR ALLOWING FLUIDS TO ENTER THE UNIT. THISCAN CAUSE A MALFUNCTION OR SHUT DOWN,
AND CAUSE AN INCREASED RISK FOR ELECTRICALSHOCK OR BURNS.
WARNING: IT CAN BE UNSAFE TO INTERCONNECT
THE COMPANION 5 WITH OTHER EQUIPMENT.

WARNING: CLEAN THE CABINET, CONTROL PANEL,
AND POWER CORD ONLY WITH A MILD HOUSEHOLDCLEANER APPLIED WITH A DAMP (NOT WET) CLOTH
OR SPONGE, AND THEN WIPE ALL SURFACES DRY.
DO NOT ALLOW ANY LIQUID TO GET INSIDE THE
DEVICE. PAY SPECIAL ATTENTION TO THE OXYGENOUTLET FOR THE CANNULA CONNECTION TO MAKE
SURE IT REMAINS FREE OF DUST, WATER, AND
PARTICLES.
WARNING: DO NOT USE LIQUID DIRECTLY ON THE
UNIT. A LIST OF UNDESIRABLE CHEMICAL AGENTS
INCLUDES BUT IS NOT LIMITED TO THE FOLLOWING:
ALCOHOL AND ALCOHOL-BASED PRODUCTS,
CONCENTRATED CHLORINE-BASED PRODUCTS(ETHYLENE CHLORIDE), AND OIL- BASED PRODUCTS(PINE-SOL®, LESTOIL®). THESE ARE NOT TO BE USEDTO CLEAN THE PLASTIC HOUSING ON OXYGENCONCENTRATOR, AS THEY CAN DAMAGE THE
UNIT’S PLASTIC.

DisposalAlways return Companion 5, including all components, to your homecare provider for proper
disposal. You can also contact your local city or
town offices for instructions on proper disposal of
the battery.

WEEE and RoHS
This symbol is to remind the equipment
owners to return it to a recycling facility at theend of its life, per Waste Electrical and Electronic
Equipment (WEEE) Directive.

18 - ENG

PN 15062781-C4 J | User Manual

Our products will comply with the restriction ofHazardous Substances (RoHS) directive. They will
not contain more than trace amounts of lead orother hazardous material content.
CAUTION: For proper disposal, contact
your equipment provider or local government agency for disposal instructions

Companion 5
Commandes d’utilisation et indicateurs d’état du système
Symboles internes

02

Pièce appliquée de type BF (degré
de protection contre les déchargesélectriques). Nº enreg. 5333

Sortie d’oxygène

I/O Interrupteur marche/arrêt
Ne contient aucune pièce nécessitantun entretien. Ne pas ouvrir le capot.

Directive du Conseil européen 93/42/CEE relative
aux dispositifs médicaux
Représentant agréé au sein de l’Union
européenne
CEI 60417

ISO 7000 : symboles graphiques à utiliser surl’équipement – index et résumé
Température de stockage ou defonctionnement. Nº enreg. 0632

Équipement de classe II à double
isolation Reg. # 5172
Certification ETL
Agence de sécurité pour CAN/CSA
C22.2 no 601.1 M90 pour les appareilsélectriques médicaux

Plage d’humidité de stockage..
Nº enreg. 2620
Nom et adresse du fabricant.
Nº enreg. 3082
Date de fabrication. Nº enreg. 2497

Avertissement de la FCC
21 CFR 801.15 : code de la réglementation
fédérale, chapitre 21

Référence catalogue. Nº enreg. 2493
Numéro de série. Nº enreg. 2498
Haut. Nº enreg. 0623
Fragile, à manipuler avec soin.
Nº enreg. 0621
Maintenez l’unité à l’abri de l’humidité.
Nº enreg. 0626
Lisez le manuel d’utilisation avant lamise en marche. Nº enreg. 1641
Limite d’empilement par nombre.
Nº enreg. 2403.
ISO 7010 : symboles graphiques – couleurs desécurité et étiquettes de sécurité – étiquettes desécurité enregistrées
Le manuel d’instruction doit être lu.
Nº enreg. M002
Avertissement. Nº enreg. W001
Tenez l’unité à l’écart des flammesnues, du feu, des étincelles. Sourced’incendie ouverte interdite etinterdiction de fumer. Nº enreg. P003
Ne fumez pas à proximité de l’appareilou pendant son fonctionnement.
Nº enreg. P002

La législation fédérale restreint la ventede cet appareil par un médecin ou sur
ordonnance de celui-ci.
Directive du Conseil européen 2012/19/UE : déchetsd’équipements électriques et électroniques (DEEE)
DEEE
CEI 60601-1 : Appareil électro-médical –Partie 1 – Règles générales de sécurité de baseet performances essentielles

IPX1

Équipement protégé contre les gouttesd’eau - IPX1 : le Companion 5n’assure aucune protection contre leseffets dangereux de la pénétration deliquides. (IPX1, conformément à ladirective CEI 60529)

AVERTISSEMENT DE LA FCC :Ce dispositif peut contenir un module
CYBLE-022001-00, y compris l’antenne
2450AT18B100 de Johnson Technology, conforme avec
l’article 15 de la réglementation FCC. Le dispositif est
conforme aux exigences relatives à l’approbation del’émetteur modulaire tel que décrit dans l’avertissement
public de la FCC DA00-1407. Le fonctionnement del’émetteur est soumis aux deux conditions suivantes :(1) ce dispositif ne peut provoquer d’interférences
nuisibles, et (2) ce dispositif doit accepter touteinterférence reçue, notamment les interférences
susceptibles d’entraîner un fonctionnement non souhaité.
Ce produit peut être protégé par un ou plusieurs brevets
américains ou internationaux. Consultez notre site
Web ci-dessous pour voir la liste des brevets. Brevet :www.caireinc.com/corporate/patents/
PN 15062781-C4 J | Manuel d’utilisation

19 - FRE

Companion 5
Guide de démarrage rapide
1 Déballage du système Companion 5

Companion 5

2 Familiarisation avec votre systèmeCompanion 5
Veillez à lire la totalité des avertissements, mises engarde et informations additionnelles sur l’appareildans le reste de ce manuel. Familiarisez-vous avec lesprincipales caractéristiques du système Companion 5 etle panneau de commande.
AVANTBoîtier
extérieur

Interrupteur

Écran
LED

Port desortieRaccordcranté
de sortie

CompteurhoraireDébitmètre

Supportdu flacond’humidificateur(Flacon en
option)ARRIÈREPoignée detransport etemplacementd’admission
d’air

Emplacementd’admission
d’air derefroidissement

Emplacementd’admission d’air

20 - FREPN 15062781-C4 J Manuel d’utilisation

Orificed’évacuation

Companion 5
Composants du système Companion 5Boîtier extérieur Boîtier en plastique durable qui
contient et protège les composantsinternes du système CAIRE
Companion 5.
Interrupteur
Interrupteur de MARCHE/ARRÊT
permettant de démarrer ou d’arrêter
l’alimentation de l’appareil. La
position MARCHE est indiquée par lesymbole « l », et la position ARRÊT
par le symbole « O ».
Raccord cranté L’oxygène ressort du systèmede sortie
Companion 5 ici après avoir été
filtré et concentré. Une canulenasale à lumière simple ou une
tubulure d’oxygène standard est
fixée pour l’administration aupatient. La longueur maximale destubulures d’oxygène raccordées
au concentrateur est de 15,2 m(50 pieds).
Écran LED
Contient un voyant vert, rouge etjaune*. Le voyant vert indique unfonctionnement normal, tandis que lesvoyants jaune* et rouge indiquent desconditions d’alarme. (Reportez-vousà la section Conditions d’alarme desinstructions pour plus d’informations.)
Compteur
Affichage numérique du temps
horairede fonctionnement écoulé duconcentrateur. Affiche le dixième leplus proche d’une heure et ne peutpas être réinitialisé. Affiche les codesd’alarme de diagnostic lorsque leconcentrateur subit une conditiond’alarme.
Débitmètre
Permet de régler et d’afficher le débit
d’oxygène administré. Les débits
sont compris entre 0,5 et 5 litres par
minute (LPM) et peuvent être réglés
en tournant le bouton.
Emplacement de raccordement d’un
Supportflacon d’humidificateur. Le support
du flacond’humidificateur comporte une bande élastiquepermettant de fixer le flacon au(Flacon en
support. Un flacon d’humidificateur
option)assure l’humidification supplémentaire
du flux d’oxygène. Le flacon est remplid’eau distillée et l’humidité évite ledessèchement des tissus nasaux.
*Modèles dotés d’un indicateurd’état du concentrateur d’oxygène
uniquement

3 Fixation du flacon d’humidificateur etde la canule
En cas d’utilisation d’un flacon d’humidificateur,commencez par remplir le flacon d’eau distillée
au niveau indiqué par le fabricant. Placez le flacondans le support désigné et fixez-le avec la bandeélastique. Raccordez l’extrémité filetée du tube del’humidificateur au flacon, et l’autre extrémité auraccord cranté du système Companion 5. Connectez
une tubulure d’oxygène et/ou une canule nasale au
raccord cranté de sortie du flacon d’humidificateur.

Si vous n’utilisez pas de flacon d’humidificateur,connectez la tubulure d’oxygène et/ou une canulenasale directement au raccord cranté de sortie dusystème Companion 5.

PN 15062781-C4 J | Manuel d’utilisation

21 - FRE

Companion 5
4 Mise en marche et préchauffage

6 Réglage du niveau de débit prescrit

Branchez le cordon d’alimentation CA sur une prisesecteur.

Utilisez le bouton de contrôle du débit poursélectionner le débit prescrit par le médecin.

Placez l’interrupteur d’alimentation en position Marche.

Interrupteur

Lorsque le système Companion 5 est correctement
allumé, le voyant d’alimentation vert présent surl’écran LED s’allume. Tous les voyants s’allument
au démarrage. À la fin du cycle de préchauffage duconcentrateur, seul le voyant vert reste allumé.
Remarque: après la mise sous tension initiale, levoyant O2 s’allume et «O2 UP» s’affiche à l’écran
jusqu’à ce que la concentration atteigne lesspécifications.

AVERTISSEMENT : IL EST TRÈSIMPORTANT DE SÉLECTIONNER LE
DÉBIT D’OXYGÈNE QUI VOUS A ÉTÉPRESCRIT. NE MODIFIEZ PAS LE RÉGLAGE
DE DÉBIT SANS AVIS MÉDICAL. LE
CONCENTRATEUR D’OXYGÈNE PEUT
ÊTRE UTILISÉ PENDANT LE SOMMEIL DU
PATIENT SOUS LA RECOMMANDATION
D’UN CLINICIEN QUALIFIÉ.
Respirez normalement à l’aide de la canule nasale.
Pour régler le débit :Tournez le bouton dans le sens contraire des aiguilles
d’une montre pour augmenter le débit.
Tournez le bouton dans le sens des aiguilles d’une
montre pour réduire le débit.
Le milieu de la bille indique le débit. Voir l’image
ci-dessous.

Voyantd’alimentation

Correct

22 - FREPN 15062781-C4 J Manuel d’utilisation

Companion 5
Important !

Les instructions de sécurité se définissent comme
indiqué ci-dessous :AVERTISSEMENT : INFORMATIONSIMPORTANTES DE SÉCURITÉ RELATIVESAUX DANGERS POUVANT CAUSER DESBLESSURES GRAVES.
MISE EN GARDE : informations
importantes pour éviter tout
endommagement du systèmeCompanion 5.
Remarque : informations exigeant une attention
particulière.

Indications d’utilisation
Utilisation conforme

Le système Companion 5 est conçu pourl’administration d’oxygène d’appoint. Ce dispositif
n’est pas destiné aux applications de réanimation ou
de maintien des fonctions vitales, et n’offre aucune
capacité de surveillance du patient.
AVERTISSEMENT : IL EST TRÈSIMPORTANT DE SÉLECTIONNER LE
DÉBIT D’OXYGÈNE QUI VOUS A ÉTÉPRESCRIT. NE MODIFIEZ PAS LE RÉGLAGE
DE DÉBIT SANS AVIS MÉDICAL. LE
CONCENTRATEUR D’OXYGÈNE PEUT
ÊTRE UTILISÉ PENDANT LE SOMMEIL DU
PATIENT SOUS LA RECOMMANDATION
D’UN CLINICIEN QUALIFIÉ.

AVERTISSEMENT : POUR GARANTIRL’ADMINISTRATION DE LA DOSE THÉRAPEUTIQUE
D’OXYGÈNE EN FONCTION DE VOTRE ÉTATDE SANTÉ, LE SYSTÈME COMPANION 5 DOIT
UNIQUEMENT ÊTRE UTILISÉ APRÈS AVOIR EFFECTUÉUN OU PLUSIEURS RÉGLAGES DÉTERMINÉSINDIVIDUELLEMENT, OU SELON LES PRESCRIPTIONSCORRESPONDANT À VOS NIVEAUX D’ACTIVITÉSPÉCIFIQUES. LE SYSTÈME COMPANION 5
DOIT ÊTRE UTILISÉ AVEC LA COMBINAISONSPÉCIFIQUE OU AVEC LES PIÈCES ET ACCESSOIRESRESPECTANT LES SPÉCIFICATIONS DU FABRICANT
DU CONCENTRATEUR ET UTILISÉS LORS DE LA
DÉTERMINATION DE VOS RÉGLAGES.
AVERTISSEMENT : CET APPAREIL NE DOIT PAS
ÊTRE UTILISÉ À DES FINS DE RÉANIMATION. LESPATIENTS GÉRIATRIQUES, PÉDIATRIQUES OU TOUTAUTRE TYPE DE PATIENT N’ÉTANT PAS EN MESURE
DE COMMUNIQUER UN MALAISE PEUVENT AVOIRBESOIN D’UNE SURVEILLANCE SUPPLÉMENTAIRE.
IL PEUT ÉGALEMENT ÊTRE NÉCESSAIRE DE
DISTRIBUER UN SYSTÈME D’ALARME POUR
TRANSMETTRE LES INFORMATIONS RELATIVES
AU MALAISE ET/OU À L’URGENCE MÉDICALE AU
SOIGNANT POUR ÉVITER TOUTE BLESSURE. LESPATIENTS SOUFFRANT DE DÉFICIENCE AUDITIVEET/OU VISUELLE PEUVENT AVOIR BESOIN D’UNE
ASSISTANCE POUR SURVEILLER LES ALARMES.
AVERTISSEMENT : LES FEMMES ENCEINTES OUQUI ALLAITENT NE DOIVENT PAS UTILISER LESACCESSOIRES RECOMMANDÉS DANS CE MANUEL,CAR ILS PEUVENT CONTENIR DES PHTALATES.
AVERTISSEMENT : CONFORMÉMENT À LA LOIFÉDÉRALE DES ÉTATS-UNIS, LA VENTE ET LA
LOCATION DE CET APPAREIL DOIVENT SE FAIRE SURORDONNANCE MÉDICALE OU PAR UN PRESTATAIRE
DE SOINS MUNI D’UNE LICENCE.

Contre-indications d’utilisation
AVERTISSEMENT : DANS CERTAINS CAS,L’UTILISATION D’OXYGÈNE SANS AVISMÉDICAL PEUT S’AVÉRER DANGEREUSE.
CET APPAREIL DOIT ÊTRE UTILISÉUNIQUEMENT SUR PRESCRIPTIONMÉDICALE.
AVERTISSEMENT : NE PAS UTILISER CET APPAREIL
EN PRÉSENCE D’ANESTHÉSIQUES INFLAMMABLES.
AVERTISSEMENT : COMME AVEC TOUTAPPAREIL ÉLECTRIQUE, L’UTILISATEUR PEUT
ÊTRE CONFRONTÉ À DES PÉRIODES DE NONFONCTIONNEMENT EN RAISON D’UNE PANNE
D’ALIMENTATION ÉLECTRIQUE OU À LA
NÉCESSITÉ DE FAIRE ENTRETENIR LE SYSTÈME
COMPANION 5 PAR UN TECHNICIEN QUALIFIÉ.
LE SYSTÈME COMPANION 5 N’EST PAS ADAPTÉAUX PATIENTS POUR LESQUELS UNE TELLE
INTERRUPTION TEMPORAIRE POURRAIT AVOIR DESCONSÉQUENCES NÉFASTES SUR LA SANTÉ.
PN 15062781-C4 J | Manuel d’utilisation

23 - FRE

Companion 5
Consignes de sécurité
AVERTISSEMENT : AUCUNEMODIFICATION DE CET ÉQUIPEMENTN’EST AUTORISÉEAVERTISSEMENT : LE FABRICANT RECOMMANDE DEDISPOSER D’UNE AUTRE SOURCE D’ALIMENTATIONEN OXYGÈNE EN CAS DE COUPURE DE COURANT,DE DÉCLENCHEMENT D’ALARME DE L’APPAREIL
OU DE PANNE MÉCANIQUE. CONSULTEZ VOTRE
MÉDECIN OU VOTRE FOURNISSEUR AGRÉÉ AUSUJET DU TYPE DE SYSTÈME DE RÉSERVE À
UTILISER.
AVERTISSEMENT : CET APPAREIL DISPENSE UN
OXYGÈNE EXTRÊMEMENT PUR, SUSCEPTIBLE
DE S’ENFLAMMER TRÈS VITE. NE FUMEZ PAS À
PROXIMITÉ ET ÉVITEZ TOUTE FLAMME NUE À DEUX
(2) MÈTRES DE (1) CET APPAREIL, OU DE (2) TOUTAUTRE ACCESSOIRE DE TRANSPORT D’OXYGÈNE. LE
NON-RESPECT DE CET AVERTISSEMENT PEUT AVOIRPOUR CONSÉQUENCES UN INCENDIE, DES DÉGÂTS
MATÉRIELS ET/OU DES BLESSURES GRAVES VOIREMORTELLES.
AVERTISSEMENT : N’UTILISEZ PAS CET APPAREIL
DANS UN ESPACE FERMÉ OU CONFINÉ (PAR
EXEMPLE DANS UNE PETITE VALISE OU UN SAC)
OFFRANT UNE VENTILATION POTENTIELLEMENTINSUFFISANTE. CECI POURRAIT PROVOQUER UNE
SURCHAUFFE DU CONCENTRATEUR D’OXYGÈNE ET
UNE DÉGRADATION DE SES PERFORMANCES.
AVERTISSEMENT : LE CONCENTRATEURD’OXYGÈNE DOIT ÊTRE PLACÉ À L’ÉCARTDE TOUTE SOURCE DE FUMÉE, DE
POLLUTION OU DE VAPEURS.
AVERTISSEMENT : LE RECOURS À DES ACCESSOIRESD’ADMINISTRATION D’OXYGÈNE NON PRÉVUS
POUR UNE UTILISATION AVEC CE CONCENTRATEURD’OXYGÈNE EST SUSCEPTIBLE DE NUIRE AUXPERFORMANCES DE L’APPAREIL. LES RÉFÉRENCESDES ACCESSOIRES RECOMMANDÉS SONT
INDIQUÉES DANS CE MANUEL.
AVERTISSEMENT : EN CAS DE CHUTE, DE
DÉTÉRIORATION OU D’EXPOSITION À L’EAU DUCONCENTRATEUR D’OXYGÈNE, CONTACTEZ VOTRE
ORGANISME DE SOINS DE SANTÉ À DOMICILE AFIN
DE PROCÉDER À L’INSPECTION OU À L’ÉVENTUELLERÉPARATION DE L’APPAREIL. VEILLEZ À NE PASUTILISER LE CONCENTRATEUR D’OXYGÈNE SI
SA FICHE OU SON CORDON D’ALIMENTATION
ÉLECTRIQUE EST ENDOMMAGÉ(E).

24 - FREPN 15062781-C4 J Manuel d’utilisation

AVERTISSEMENT : NE LAISSEZ PAS LES ORIFICES
D’ADMISSION ET D’ÉVACUATION DE L’AIRS’OBSTRUER. NE FAITES PAS TOMBER L’APPAREIL
OU N’INSÉREZ AUCUN OBJET DANS SESOUVERTURES. CECI POURRAIT PROVOQUER UNE
SURCHAUFFE DU CONCENTRATEUR D’OXYGÈNE ET
UNE DÉGRADATION DE SES PERFORMANCES.
AVERTISSEMENT : NE REMPLISSEZ PAS DE FAÇON
EXCESSIVE L’HUMIDIFICATEUR OPTIONNEL.
REMPLISSEZ-LE AVEC DE L’EAU UNIQUEMENT,JUSQU’AU NIVEAU INDIQUÉ PAR LE FABRICANT DE
L’HUMIDIFICATEUR.
AVERTISSEMENT : LISEZ ET ASSUREZ-VOUS D’AVOIRCOMPRIS LES INFORMATIONS DE CE MANUELAVANT D’UTILISER CET APPAREIL. SI VOUS NE
COMPRENEZ PAS LES AVERTISSEMENTS ET LESINSTRUCTIONS, CONTACTEZ VOTRE FOURNISSEURAGRÉÉ AVANT D’ESSAYER D’UTILISER L’APPAREIL
AFIN D’ÉVITER TOUT RISQUE DE BLESSURE OU DEDÉGÂTS MATÉRIELS.
AVERTISSEMENT : SI VOUS VOUS SENTEZ MAL
À L’AISE OU EN CAS D’URGENCE MÉDICALE,SOLLICITEZ IMMÉDIATEMENT UNE ASSISTANCEMÉDICALE.
AVERTISSEMENT : L’UTILISATION DU
CONCENTRATEUR D’OXYGÈNE À UNE
TEMPÉRATURE NON CONFORME AUXCARACTÉRISTIQUES TECHNIQUES PEUT LIMITER
SA CAPACITÉ À DÉLIVRER LA CONCENTRATIONEN OXYGÈNE SPÉCIFIÉE. REPORTEZ-VOUS À LA
SECTION CARACTÉRISTIQUES TECHNIQUES DE
CE MANUEL POUR CONSULTER LES LIMITES DE
TEMPÉRATURE.
AVERTISSEMENT : VEILLEZ À NE PAS UTILISERD’HUILE, DE GRAISSE, DE PRODUITS À BASE
DE PÉTROLE OU D’AUTRES SUBSTANCESINFLAMMABLES AVEC LES ACCESSOIRESDE TRANSPORT D’OXYGÈNE OU AVEC LE
CONCENTRATEUR D’OXYGÈNE. SEULES DESLOTIONS ET DES CRÈMES À BASE D’EAU ET
COMPATIBLES AVEC LA PRÉSENCE D’OXYGÈNE
DOIVENT ÊTRE UTILISÉES. L’OXYGÈNE ACCÉLÈRE LACOMBUSTION DES SUBSTANCES INFLAMMABLES.
AVERTISSEMENT : LE CONCENTRATEUR D’OXYGÈNE
NE DOIT PAS ÊTRE UTILISÉ LORSQU’IL EST ACCOLÉÀ, OU EMPILÉ AVEC D’AUTRES ÉQUIPEMENTS. SI UN
TEL POSITIONNEMENT NE PEUT PAS ÊTRE ÉVITÉ,IL CONVIENT DE SURVEILLER L’APPAREIL POURS’ASSURER QU’IL FONCTIONNE CORRECTEMENT.

Companion 5
AVERTISSEMENT : EN CAS DE CHUTE, DE
DÉTÉRIORATION OU D’EXPOSITION À L’EAU DUCONCENTRATEUR D’OXYGÈNE, CONTACTEZ VOTRE
ORGANISME DE SOINS DE SANTÉ À DOMICILE AFIN
DE PROCÉDER À L’INSPECTION OU À L’ÉVENTUELLERÉPARATION DE L’APPAREIL. VEILLEZ À NE PASUTILISER LE CONCENTRATEUR D’OXYGÈNE SI
SA FICHE OU SON CORDON D’ALIMENTATION
ÉLECTRIQUE EST ENDOMMAGÉ(E).
AVERTISSEMENT : UTILISEZ UNIQUEMENTLES PIÈCES DE RECHANGE RECOMMANDÉESPAR LE FABRICANT POUR GARANTIR LE BON
FONCTIONNEMENT ET ÉVITER LES RISQUESD’INCENDIE ET DE BRÛLURES.
AVERTISSEMENT : NE LUBRIFIEZ PAS LESRACCORDS, LES CONNEXIONS, LA TUBULURE OU
D’AUTRES ACCESSOIRES DU CONCENTRATEURD’OXYGÈNE POUR ÉVITER LES RISQUES D’INCENDIE
ET DE BRÛLURES.
AVERTISSEMENT : FUMER EN UTILISANT DEL’OXYGÈNE CONSTITUE LA PREMIÈRE CAUSE DE
BLESSURES ET DE DÉCÈS LIÉS AUX INCENDIES.
VOUS DEVEZ IMPÉRATIVEMENT RESPECTER LESAVERTISSEMENTS DE SÉCURITÉ SUIVANTS :AVERTISSEMENT : NE LAISSEZ PERSONNE FUMERET N’UTILISEZ PAS DE BOUGIES OU DE FLAMMESNUES DANS LA MÊME PIÈCE QUE L’APPAREIL OUQUE LES ACCESSOIRES DE TRANSPORT D’OXYGÈNE.
AVERTISSEMENT : FUMER LORS DU PORT D’UNE
CANULE À OXYGÈNE PEUT PROVOQUER DESBRÛLURES AU VISAGE OU MÊME LA MORT.
AVERTISSEMENT : SI VOUS FUMEZ, CES 3 GESTESPEUVENT VOUS SAUVER LA VIE : ARRÊTEZ LE
CONCENTRATEUR D’OXYGÈNE, RETIREZ LA CANULEET QUITTEZ LA PIÈCE DANS LAQUELLE SE TROUVEL’APPAREIL.
AVERTISSEMENT : DES INDICATIONS« INTERDICTION DE FUMER - UTILISATIOND’OXYGÈNE » DOIVENT ÊTRE AFFICHÉES DEFAÇON BIEN VISIBLE À VOTRE DOMICILE OU DANSTOUT LIEU OÙ DE L’OXYGÈNE EST UTILISÉ. LESPATIENTS ET LE PERSONNEL SOIGNANT DOIVENTÊTRE INFORMÉS DES DANGERS LIÉS AU FAIT DE
FUMER EN PRÉSENCE OU PENDANT L’UTILISATIOND’OXYGÈNE MÉDICAL.
AVERTISSEMENT : N’UTILISEZ PAS VOTRE
CONCENTRATEUR D’OXYGÈNE EN PRÉSENCE DE
GAZ INFLAMMABLES. CELA PEUT PROVOQUERUN EMBRASEMENT RAPIDE SUSCEPTIBLE
D’ENTRAÎNER DES DÉGÂTS MATÉRIELS, DESBLESSURES OU LA MORT.
AVERTISSEMENT : LE FAIT DE RETIRER LA CANULEET DE LA PLACER SUR UN VÊTEMENT, UN DRAP,
UN SOFA OU TOUTE AUTRE MATIÈRE TEXTILE PEUTPROVOQUER UN EMBRASEMENT INSTANTANÉEN CAS D’EXPOSITION À UNE CIGARETTE, À UNE
SOURCE DE CHALEUR OU À UNE FLAMME.

AVERTISSEMENT : NE LAISSEZ JAMAIS UNECANULE NASALE SUR OU SOUS UN VÊTEMENT,UN DRAP OU UN COUSSIN. SI L’APPAREIL EST
ALLUMÉ SANS ÊTRE UTILISÉ, CES MATIÈRES
RISQUENT DE S’ENFLAMMER AU CONTACT DEL’OXYGÈNE. ÉTEIGNEZ SYSTÉMATIQUEMENT LE
CONCENTRATEUR D’OXYGÈNE LORSQU’IL N’ESTPAS UTILISÉ.
AVERTISSEMENT : NE TENTEZ PAS
DE RÉALISER DES OPÉRATIONS DE
MAINTENANCE AUTRES QUE LESSOLUTIONS DE DÉPANNAGE DÉTAILLÉESDANS CE MANUEL. NE RETIREZ PASLES CAPOTS. SEUL LE FOURNISSEUR DE
L’APPAREIL OU UN TECHNICIEN QUALIFIÉ
EST HABILITÉ À RETIRER LES CAPOTS OURÉPARER L’APPAREIL.
AVERTISSEMENT : UTILISEZ EXCLUSIVEMENT LESACCESSOIRES RECOMMANDÉS PAR LE FABRICANT.
L’UTILISATION D’AUTRES ACCESSOIRES POURRAITS’AVÉRER DANGEREUSE ET ENDOMMAGERGRAVEMENT VOTRE CONCENTRATEUR D’OXYGÈNE,ET PROVOQUERAIT L’ANNULATION DE LAGARANTIE.
MISE EN GARDE : déconnectez
toujours l’alimentation CA de la prisesecteur murale avant de déconnecter
l’alimentation CA du concentrateurd’oxygène.
MISE EN GARDE : disposez toujours la tubulured’alimentation en oxygène et les cordonsd’alimentation de façon à prévenir tout risquede chute.
MISE EN GARDE : suivez toutes les
instructions du fabricant afin d’éviter
l’annulation de la garantie.
Remarque : les équipements portables et mobiles decommunication par radiofréquences peuvent affecter lefonctionnement des équipements médicaux électriques.
AVERTISSEMENT: PLACEZ TOUJOURS LES TUBES
D’ALIMENTATION EN OXYGÈNE ET LES CORDONSD’ALIMENTATION DE MANIÈRE À ÉVITER LES RISQUES DE DÉCLENCHEMENT OU D’ÉTRANGLEMENT
ACCIDENTEL POSSIBLE.
AVERTISSEMENT : CE PRODUIT PEUT VOUS EXPOSER
À DES PRODUITS CHIMIQUES, NOTAMMENT LE
NICKEL, RECONNU DANS L’ÉTAT DE CALIFORNIECOMME CAUSE DE CANCER. POUR EN SAVOIR PLUS,
RENDEZ-VOUS SUR WWW.P65WARNINGS.CA.GOV.
AVERTISSEMENT: EN CAS D’INCIDENT GRAVESURVENANT AVEC CET APPAREIL, L’UTILISATEURDOIT IMMÉDIATEMENT SIGNALER L’INCIDENT AU
FOURNISSEUR ET / OU AU FABRICANT. UN INCIDENTGRAVE EST DÉFINI COMME UNE BLESSURE, LA
MORT OU LA POSSIBILITÉ DE CAUSER DES BLESSURES OU LA MORT EN CAS DE RÉAPPARITION
DE L’INCIDENT. L’UTILISATEUR PEUT ÉGALEMENTSIGNALER L’INCIDENT À L’AUTORITÉ COMPÉTENTEDU PAYS OÙ L’INCIDENT S’EST PRODUIT.
PN 15062781-C4 J | Manuel d’utilisation

25 - FRE

Companion 5
Caractéristiques techniques
Le tableau suivant fournit des informations importantes
concernant les environnements ou conditions defonctionnement recommandé(e)s pour une bonne
utilisation de l’appareil.

Caractéristiques techniques
DébitsConcentration
en O2
Dimensions
Poids
Niveau depression sonore
ConsommationélectriqueTempérature defonctionnement
Humidité defonctionnement
Température de
stockage
Humidité destockage
Caractéristiquesélectriques

0,5 à 5,0 l/min
90 % (+5,5 %/-3 %)21,5 x 12,5 x 13,5”(54,6 x 31,8 x 34,2 cm)
16,3 kg (36 livres)
50 dB(A) à 2,0 l/min

Choisissez un emplacement évitant à l’appareild’aspirer des fumées, des vapeurs et des polluants.
Le positionnement correct de l’appareil doit permettre
à l’air d’entrer à travers les trois emplacementsd’admission d’air situés dans la partie supérieurearrière et sous l’armoire et de sortir librement parl’orifice d’évacuation situé dans la partie inférieuregauche de l’appareil.

285 W à 2 l/min, 350 W maximum

Les admissions d’air se trouvent en bas et en hautà l’arrière du concentrateur. Positionnez l’appareilde manière à laisser un espace d’au moins 30 cm(12 pouces) entre le concentrateur et tous murs,
meubles, draperies et autres obstructions.

5 ºC à 40 ºC (41 ºF à 104 ºF)

Placez l’appareil de façon à pouvoir entendre lesalarmes.

15% - 90 % à un point de rosée de82,4 °F (28 °C)-13 ºF à 158 ºF (-25 ºC à 70 ºC)
0% - 90% sans condensation
N’utilisez pas de rallonges ni deprises secteur contrôlées par uninterrupteur.
-382 à 3011 mètres
(-1253 à 9879 pieds)
Jusqu’à 24 heures par jour

Altitude defonctionnement
Durée defonctionnement
La durée de vie estimée de l’appareil est d’au moins
cinq ans.
AVERTISSEMENT: L’UTILISATION D’UNDISPOSITIF HORS DES CONDITIONSD’EXPLOITATION SPÉCIFIÉES DEVRAIT
INFLUENCER DE FAÇON ADÉQUATE LEDÉBIT ET LE POURCENTAGE D’OXYGÈNE
ET, PAR CONSÉQUENT, LA QUALITÉ DU
TRAITEMENT.
Remarque : si le concentrateur d’oxygène a été stockéde façon prolongée à une température non conforme àsa plage de températures normales de fonctionnement,
il est impératif de laisser l’appareil reprendre unetempérature normale avant de le mettre en marche.
(Reportez-vous à la section Caractéristiques techniquesde ce manuel.)

26 - FRE

Positionnement approprié du systèmeCompanion 5

PN 15062781-C4 J Manuel d’utilisation

Positionnez le tube d’alimentation en oxygène de façon
à ce qu’il ne forme pas de coudes ou de plis.
AVERTISSEMENT : N’UTILISEZ PAS
VOTRE CONCENTRATEUR D’OXYGÈNE
EN PRÉSENCE DE GAZ INFLAMMABLES.
CELA PEUT PROVOQUER UNEMBRASEMENT RAPIDE SUSCEPTIBLE
D’ENTRAÎNER DES DÉGÂTS MATÉRIELS,DES BLESSURES OU LA MORT. VEILLEZ
À NE PAS UTILISER D’HUILE, DE
GRAISSE, DE PRODUITS À BASE DEPÉTROLE OU D’AUTRES SUBSTANCESINFLAMMABLES AVEC LES ACCESSOIRESDE TRANSPORT D’OXYGÈNE OU AVECLE CONCENTRATEUR D’OXYGÈNE.
SEULES DES LOTIONS ET DES CRÈMES
À BASE D’EAU ET COMPATIBLES AVECLA PRÉSENCE D’OXYGÈNE DOIVENTÊTRE UTILISÉES. L’OXYGÈNE ACCÉLÈRE
LA COMBUSTION DES SUBSTANCESINFLAMMABLES.
AVERTISSEMENT : CET APPAREIL DISPENSE UN
OXYGÈNE EXTRÊMEMENT PUR, SUSCEPTIBLE
DE S’ENFLAMMER TRÈS VITE. NE FUMEZ PAS
À PROXIMITÉ ET ÉVITEZ TOUTE FLAMME NUEDANS LA MÊME PIÈCE QUE (1) CET APPAREIL,OU QUE (2) TOUT AUTRE ACCESSOIRE DE
TRANSPORT D’OXYGÈNE. LE NON-RESPECTDE CET AVERTISSEMENT PEUT AVOIR POURCONSÉQUENCES UN INCENDIE, DES DÉGÂTS
MATÉRIELS ET/OU DES BLESSURES GRAVES VOIREMORTELLES.

Companion 5
Instructions d’utilisation
Avant utilisation

Ce manuel d’utilisation vous sert de référence pourvous aider à utiliser et à entretenir le dispositif. Pour
toute question ou préoccupation, veuillez contacter
votre organisme de soins de santé à domicile.
Important ! N’utilisez PAS le système Companion 5
avant d’avoir lu la section « Consignes de sécurité »de ce manuel. Veillez à respecter toutes les instructions
d’utilisation ainsi que les avertissements apposés
sur l’appareil ou figurant dans ce manuel. Afinde réduire les risques d’incendie, de blessures etd’endommagement grave du système Companion 5,
veillez à respecter toutes les précautions de sécurité.
Le système CAIRE Companion 5 ne produit pas son
propre oxygène. Il produit de l’oxygène hautement
concentré à partir de l’air ambiant et vous l’administre.
À son état naturel, l’air ambiant ne contient
approximativement que 21 % d’oxygène par volume.
L’air ambiant est aspiré dans le concentrateur par
un compresseur. L’air traverse ensuite un ensemblede filtres et le lit d’un tamis moléculaire qui absorbe
chimiquement les molécules d’azote. En conséquence,
l’air enrichi en oxygène qui sort du concentrateur
est administré à raison de 87 à 95 % d’oxygène parvolume.
AVERTISSEMENT : NETTOYEZ LE
BOÎTIER, LE PANNEAU DE COMMANDEET LE CORDON D’ALIMENTATIONUNIQUEMENT À L’AIDE D’UN
DÉTERGENT DOUX APPLIQUÉ AU MOYEN
D’UN CHIFFON OU D’UNE ÉPONGE
LÉGÈREMENT HUMIDE, PUIS ESSUYEZ
TOUTES LES SURFACES POUR LESSÉCHER. NE LAISSEZ AUCUN LIQUIDEPÉNÉTRER À L’INTÉRIEUR DE L’APPAREIL.
INSPECTEZ TOUT PARTICULIÈREMENTLA SORTIE D’OXYGÈNE AU NIVEAU DURACCORDEMENT DE LA CANULE AFIN
DE VÉRIFIER L’ABSENCE DE POUSSIÈRES,D’EAU OU AUTRES PARTICULES.
MISE EN GARDE : ne laissez pas lesorifices d’admission et d’évacuation del’air s’obstruer. Ceci pourrait provoquer
une surchauffe du concentrateurd’oxygène et une dégradation deses performances. n’utilisez pas leconcentrateur d’oxygène lorsque le filtred’admission d’air n’est pas en place.
Si un second filtre est fourni, mettezle en place avant de nettoyer le filtre
sale. Nettoyez le filtre encrassé à l’eausavonneuse tiède, puis laissez sécher
totalement avant de l’utiliser.

Étape 1 : positionnement de votre systèmeCompanion 5 en vue de son utilisation
Placez le système Companion 5 dans un lieu
correctement ventilé et éclairé. Vérifiez que lesorifices d’admission et d’évacuation de l’air ne sontpas obstrués. Positionnez le système Companion 5 defaçon à ce que les indicateurs ou les alarmes sonores etvisuels soient parfaitement visibles et audibles.
Branchez l’appareil sur une prise d’alimentation
secteur.
Étape 2 : si le flacon de l’humidificateur N’EST PAS
utilisé

Emplacementd’admission
d’air
Emplacementd’admission
d’air derefroidissement

Orificed’évacuation
Emplacementd’admission d’aira. C onnectez une canule nasale ou une tubulured’oxygène directement au raccord cranté de sortied’oxygène.

b. Passez à l’étape 4.
Étape 3 : si le flacon de l’humidificateur EST utilisé
a. Dévissez le couvercle du flacon d’humidificateur.
b. Remplissez le flacon d’eau distillée. Assurez-vousque le niveau d’eau se situe entre le niveau maximum
indiqué par le fabricant et le niveau minimum indiqué
sur le flacon.
c. Fixez le couvercle du flacon d’humidificateur.
Assurez-vous qu’il n’y a aucune fuite.
d. Connectez une canule nasale ou une tubulured’oxygène standard au raccord cranté sur le flacond’humidificateur.
Remarque : l’utilisation d’un humidificateur non
prévu pour ce concentrateur pourrait affecter ses
performances.
PN 15062781-C4 J | Manuel d’utilisation

27 - FRE

Companion 5
g. Placez le flacon à son emplacement désigné sur lesupport du flacon d’humidificateur.
h. Fixez le flacon sur le support à l’aide de la bandeélastique.

e. Connectez l’extrémité filetée du tube del’humidificateur au couvercle du flacond’humidificateur.

i. Passez à l’étape 4.
Étape 4 : mise en marche et préchauffagea. Branchez le cordon électrique du système CAIRE
Companion 5 sur une prise secteur.
AVERTISSEMENT : VÉRIFIEZ QUE LE
CORDON ÉLECTRIQUE NE PRÉSENTEAUCUNE TRACE D’ENDOMMAGEMENT
AVANT SON UTILISATION. SI LE CORDONEST ENDOMMAGÉ, NE LE BRANCHEZPAS SUR UNE PRISE ÉLECTRIQUE
ET N’ESSAYEZ PAS D’UTILISER LE
CONCENTRATEUR.

f. Connectez l’autre extrémité du tube del’humidificateur au raccord cranté de sortie dusystème CAIRE Companion 5.

b. Mettez l’interrupteur d’alimentation en position« MARCHE » (I). Lorsque le système Companion 5
est correctement allumé, un voyant vert s’allume
sur l’écran LED. Tous les voyants s’allument audémarrage. À la fin du cycle de préchauffage duconcentrateur, seul le voyant vert reste allumé.
Remarque : après avoir mis le dispositif en MARCHE
pour la première fois, attendez jusqu’à dix (10)minutes pour que celui-ci atteigne ses spécifications
de performances.

28 - FREPN 15062781-C4 J Manuel d’utilisation

Companion 5
AVERTISSEMENT : IL EST TRÈSIMPORTANT DE SÉLECTIONNER LE
DÉBIT D’OXYGÈNE QUI VOUS A ÉTÉPRESCRIT. NE MODIFIEZ PAS LE RÉGLAGE
DE DÉBIT SANS AVIS MÉDICAL. LE
CONCENTRATEUR D’OXYGÈNE PEUT
ÊTRE UTILISÉ PENDANT LE SOMMEIL DU
PATIENT SOUS LA RECOMMANDATION
D’UN CLINICIEN QUALIFIÉ.
Étape 6 : vérification du flux et respiration normale
Étape 5 : réglage du débit de contrôlea. Tournez le bouton de contrôle du débit sur le débit
d’oxygène (LPM) prescrit par le médecin.

a. Vérifiez que l’oxygène circule bien dans la canulenasale et que le tube de la canule n’est pas coudé,
plié ou obstrué.
Remarque : vérifiez que la canule est entièrement
insérée et convenablement positionnée. Vous devriez
entendre ou sentir le flux d’oxygène vers les branches
de la canule nasale.
b. Placez correctement la canule nasale et respirez
normalement à travers celle-ci.
Étape 7 : arrêt
a. Mettez l’interrupteur d’alimentation en position« ARRÊT » quand le système CAIRE Companion 5n’est plus utilisé.

Pour régler le débit :Tournez le bouton dans le sens contraire des aiguilles
d’une montre pour augmenter le débit.
Tournez le bouton dans le sens des aiguilles d’une
montre pour réduire le débit.
Le milieu de la bille indique le débit. L’image cidessous indique un débit de 4,0 l/min.

Correct

MISE EN GARDE : utilisez exclusivement
les accessoires recommandés parle fabricant. L’utilisation d’autres
accessoires pourrait s’avérer
dangereuse et endommager gravement
votre concentrateur d’oxygène, etprovoquerait l’annulation de la garantie.
MISE EN GARDE : n’utilisez pas de rallongeélectrique avec cet appareil et veillez à ne pas
brancher un trop grand nombre d’appareils sur lamême prise secteur. L’utilisation d’une rallongerisque d’affecter les performances de l’appareil. De
plus, si vous effectuez trop de branchements surune même prise, vous risquez de provoquer une
surcharge du tableau électrique et de déclencher ledisjoncteur ou de provoquer un incendie si celui-ci
ne fonctionne pas.
MISE EN GARDE : l’utilisation de câbles etd’adaptateurs autres que ceux spécifiés,à l’exception des câbles et adaptateurscommercialisés par le fabricant de l’équipementélectro-médical pour le remplacement decomposants internes, risque d’entraîner une
augmentation des émissions électromagnétiques
ou une diminution de l’immunitéélectromagnétique du concentrateur d’oxygène.

PN 15062781-C4 J | Manuel d’utilisation

29 - FRE

Companion 5
AVERTISSEMENT : NE LAISSEZ PASLES ORIFICES D’ADMISSION ET
D’ÉVACUATION DE L’AIR S’OBSTRUER.
CECI POURRAIT PROVOQUER UNE
SURCHAUFFE DU CONCENTRATEURD’OXYGÈNE ET UNE DÉGRADATION DE
SES PERFORMANCES.
Remarque: Pour le fournisseur d’équipement: Lesaccessoires d’administration d’oxygène suivants sontrecommandés pour une utilisation avec le Companion
5:• Canule nasale: référence CAIRE CU002-1• Tuyau adaptateur pour humidificateur: référence
CAIRE 20843882• Bouteille d’humidificateur: référence CAIRE HU003-1• Coupe-feu: référence CAIRE 20629671
Un coupe-feu est requis pour une utilisation avec
n’importe quelle canule.
• CAIRE propose un coupe-feu destiné à être utilisé
avec le concentrateur d’oxygène. Le coupe-feu est
un fusible thermique pour arrêter le flux de gaz dansle cas où la canule en aval ou le tube d’oxygène est
allumé et brûle jusqu’au coupe-feu. Il est placé enligne avec la canule nasale ou la tubulure d’oxygène
entre le patient et la sortie d’oxygène du Companion
5. Pour une utilisation correcte du coupe-feu, reportezvous toujours aux instructions du fabricant (incluses
avec chaque kit coupe-feu).
• Pour tout accessoire supplémentaire recommandé,veuillez consulter le catalogue d’accessoires (PN
MLLOX0010) disponible sur www.caireinc.com.

30 - FREPN 15062781-C4 J Manuel d’utilisation

Companion 5
Conditions d’alarme
Votre système CAIRE Companion 5 utilise une
combinaison de deux ou trois (pour les modèles dotésd’un indicateur d’état du concentrateur d’oxygène)
voyants et une alarme sonore pour vous alerter en cas
de dysfonctionnement du concentrateur.
1) Voyant vert (normal.

) – Indique un fonctionnement

2) Voyant rouge ( ) – Indique un dysfonctionnement
du système. Reportez-vous au tableau Conditions
d’alarme ci-dessous pour plus d’informations.

Vert

Rouge

Jaune

3) Voyant jaune ( ) – Indique une faible
concentration en oxygène. Reportez-vous autableau Conditions d’alarme ci-dessous pour plusd’informations.
Le tableau Conditions d’alarme, dans la page suivante,présente les différentes conditions d’alarme qui
peuvent être visibles sur le concentrateur. Il indiqueles causes possibles et les mesures à prendre en cas
d’alarme. En cas d’alarme, nous vous recommandonsde contacter aussitôt que possible votre organisme desoins de santé à domicile.

Ci-dessus : Companion 5

AVERTISSEMENT : N’IGNOREZ PAS LESALARMES.
AVERTISSEMENTS : LE FABRICANTRECOMMANDE DE DISPOSER D’UNE
AUTRE SOURCE D’ALIMENTATIONEN OXYGÈNE EN CAS DE COUPURE
DE COURANT, DE DÉCLENCHEMENTD’ALARME DE L’APPAREIL OU DEPANNE MÉCANIQUE. CONSULTEZ VOTRE
MÉDECIN OU VOTRE FOURNISSEURAGRÉÉ AU SUJET DU TYPE DE SYSTÈME
DE RÉSERVE À UTILISER.

PN 15062781-C4 J | Manuel d’utilisation

31 - FRE

Companion 5
Tableau Conditions d’alarme
AlarmesonoreVoyantdecouleur

Cause

ActionAucun

Le concentrateur
fonctionne correctement.

ÉteintVert

Intermittente
Rouge

Intermittente

Le concentrateurn’est plus alimenté
en courant électrique,mais l’interrupteurd’alimentation est toujours
en position « MARCHE ».

Vérifiez que le cordon électrique du concentrateur
est branché sur une prise et que cette dernière
est alimentée en courant. Essayez de brancher leconcentrateur sur une autre prise. Si le problème
persiste, connectez-vous à une source d’alimentation en
oxygène de secours, puis contactez votre organisme desoins de santé à domicile.

Dysfonctionnement dusystème

Mettez immédiatement l’interrupteur d’alimentation en
position « ARRÊT ». Déconnectez l’alimentation secteurde la prise murale. Patientez 5 minutes. Rebranchez
l’alimentation secteur sur la prise murale et remettez
le concentrateur en marche. Si l’alarme persiste, uneintervention est requise. Connectez-vous à une sourced’alimentation en oxygène de secours et contactez
immédiatement votre organisme de soins de santé àdomicile.

Débit du produit trop élevé
ou trop faible

1. Vérifiez que la canule n’est pas coudée ou obstruée.
Si vous utilisez un flacon d’humidificateur, vérifiez
qu’il est correctement rempli et ne provoque pas
d’obstruction.
2. Assurez-vous que le système Companion 5 est
correctement ventilé. Il doit être placé à au moins
30 cm (12 pouces) de toute surface afin d’éviter touteobstruction des orifices d’admission et d’évacuation.
3. Si le problème persiste, utilisez une autre sourced’oxygène et contactez votre organisme de soins desanté à domicile pour obtenir de l’aide.

Rouge

Intermittente
Rouge

Intermittente
Jaune

IntermittenteJaune

32 - FRE

Le système Companion 5 adétecté un faible niveaud’oxygène.

Le système Companion 5 adétecté un faible niveaud’oxygène.

PN 15062781-C4 J Manuel d’utilisation

1. Assurez-vous que le filtre d’admission d’air et lesorifices d’évacuation d’air ne sont pas bouchés ou
obstrués. 2. Assurez-vous que le Companion 5 se trouve
dans un endroit correctement ventilé. Assurez-vous quel’arrière et les côtés du système Companion se trouvent
à au moins 30 cm (12 pouces) de tout élément pouvant
provoquer une obstruction (meuble, rideau, etc.).
3. Si le problème persiste, utilisez une autre sourced’oxygène et contactez immédiatement votreorganisme de soins de santé à domicile.

Companion 5
Dépannage
Nous vous recommandons de contacter votreorganisme de soins de santé à domicile si votre
concentrateur affiche la moindre alarme ou si vouspensez qu’il ne fonctionne pas correctement. Vous
ne devez pas essayer de réparer le concentrateurvous-même. Seul un technicien qualifié est habilité
à intervenir sur l’appareil. Toutefois, vous pouvez

essayer quelques mesures de dépannage si desproblèmes surviennent lors de l’utilisation de votre
système CAIRE Companion 5. Ces mesures sontdécrites dans le tableau suivant. Vous pouvez suivre
les vérifications présentées dans ce tableau par ordre
numérique.

Tableau de dépannage
Problème

Le concentrateur
est allumé, mais
ne fonctionne pas.

Le débitd’oxygène est
faible, voire nul.

Cause possible

Action1. Le cordon électrique n’est pas
branché sur une prise.

1. Branchez le cordon électrique.

2. La prise électrique n’est pas
alimentée en courant.

2. Vérifiez le fusible et le circuit électrique de votre
domicile. Essayez de brancher le concentrateur sur
une autre prise.

3. Panne interne.

3. Connectez-vous à une source d’alimentation en
oxygène de secours. Contactez immédiatement votreorganisme de soins de santé à domicile.

1. Le réglage de contrôle du débit achangé.

1. Vérifiez que le bouton de contrôle du débit est réglé
sur le nombre de litres par minute (LPM) approprié
prescrit par votre médecin.

2. La tubulure a été déconnectée.

2. Vérifiez que la tubulure est fermement fixée àtoutes les connexions (raccord cranté de sortie, flacond’humidificateur, piège à eau, etc.). Refixez tous les
raccords, si nécessaire.

3. Les tubulures d’oxygène ou lacanule sont coudées.

3. Vérifiez la présence de coudes ou d’obstructions
dans les tubulures. Réparez si nécessaire.

4. La canule ou les tubuluresprésentent une fuite.

4. Vérifiez la présence de fuite sur toutes les tubulureset la canule. Pour ce faire, vous pouvez écouter ou
sentir l’échappement d’air. Remplacez les tubulures ou
la canule qui fuient.

5. La circulation d’air dans leconcentrateur est obstruée.

5. Vérifiez que rien ne bloque les orifices d’admission
du concentrateur. Vérifiez que le concentrateur se
trouve à 30 cm (12 pouces) au moins des meubles,
murs et autres obstructions.

6. Panne interne.

6. Connectez-vous à une source d’alimentation en
oxygène de secours. Contactez immédiatement votreorganisme de soins de santé à domicile.

Remarque : si le système Companion 5 émet toujours une alarme après que vous avez suivi les solutions décrites
dans les tableaux Conditions d’alarme et Dépannage, veuillez contacter le service technique au +1 800 482 2473
afin de lui communiquer le code d’alarme affiché sur l’écran LCD.

PN 15062781-C4 J | Manuel d’utilisation

33 - FRE

Companion 5
Nettoyage, entretien et maintenance régulière
Maintenance régulière

La maintenance régulière consiste à changer le filtre àgrosses particules de la poignée une fois par semaine(s’il est installé).

Entretien et nettoyage de l’appareil

Remplacement de la canule
Remplacez régulièrement votre tube d’alimentation etvotre canule, comme recommandé par votre organismede soins de santé à domicile. Votre médecin ou votreorganisme de soins de santé à domicile vous fournira
des informations concernant leur nettoyage, leurdésinfection et leur remplacement.
Remarque : suivez toujours les instructions dufabricant de la canule pour une utilisation correcte.
Remplacez la canule jetable conformément aux
recommandations du fabricant de la canule ou dufournisseur de l’équipement. D’autres fournitures sontdisponibles auprès de votre fournisseur.

Filtre d’admission d’air (cartouche)
Seul votre professionnel de la santé doit avoir accès au
filtre d’admission d’air ou le changer.
Filtre à grosses particules (en option) :Le filtre à particules grossières est situé tout en bas
à l’avant du Companion 5.

Filtre à grosses
particules

Vous devez l’inspecter et le nettoyer une fois par
semaine. Pour nettoyer le filtre, utilisez le processus
suivant:
1) Retirez le filtre à grosses particules.
2) Nettoyez le filtre dans de l’eau potable tiède en
utilisant une solution de détergent savonneux doux.
3) Rincez soigneusement le filtre à l’eau claire etcomprimez-le pour évacuer l’eau en excès.
4) Laissez le filtre sécher à l’air libre.
5) Réinstallez le filtre dans le boîtier.
34 - FREPN 15062781-C4 J Manuel d’utilisation

Remarque : le fabricant ne recommande pas destériliser cet équipement.
Remarque : n’utilisez pas le concentrateur d’oxygène
lorsque le filtre d’admission d’air n’est pas en place.

Flacon de l’humidificateur (le cas échéant)Vérifiez chaque jour votre flacon d’humidificateur pourvous assurer que son niveau d’eau se situe entre lesniveaux minimum et maximum spécifiés. Remplissez
le flacon d’eau distillée lorsque cela est nécessaire.
Nettoyez le flacon d’humidificateur conformément aux
instructions du fabricant.
Procédure de nettoyage :Arrêtez le système Companion 5 et déconnectez-lede l’alimentation secteur avant toute opération de
nettoyage ou de désinfection. NE vaporisez PAS
de liquides directement sur le boîtier extérieur.
Utilisez une éponge ou un chiffon légèrement
humide. Vaporisez un détergent doux sur l’épongeou sur le chiffon afin de nettoyer le boîtier et lecordon d’alimentation. Pour désinfecter le systèmeCompanion 5, utilisez le désinfectant Lysol® Brand II(ou un produit équivalent). Diluez-le comme indiquépar le fabricant du produit nettoyant, mais ne vaporisez
pas le liquide directement sur le système Companion 5.
AVERTISSEMENT : RISQUE DE CHOC
ÉLECTRIQUE. ARRÊTEZ L’APPAREILET DÉBRANCHEZ LE CORDON
D’ALIMENTATION DE LA PRISE
ÉLECTRIQUE AVANT DE NETTOYERL’APPAREIL AFIN D’ÉVITER TOUT CHOC
ÉLECTRIQUE ACCIDENTEL OU RISQUE
DE BRÛLURE. SEUL LE FOURNISSEUR DE
L’APPAREIL OU UN TECHNICIEN QUALIFIÉ
EST HABILITÉ À RETIRER LES CAPOTS OURÉPARER L’APPAREIL.
AVERTISSEMENT : RISQUE DE CHOC ÉLECTRIQUE.
ARRÊTEZ L’APPAREIL ET DÉBRANCHEZ LE CORDON
D’ALIMENTATION DE LA PRISE ÉLECTRIQUE AVANT
DE NETTOYER L’APPAREIL AFIN D’ÉVITER TOUT
CHOC ÉLECTRIQUE ACCIDENTEL OU RISQUE DEBRÛLURE. SEUL LE FOURNISSEUR DE L’APPAREIL OU
UN TECHNICIEN QUALIFIÉ EST HABILITÉ À RETIRERLES CAPOTS OU RÉPARER L’APPAREIL.

Companion 5
AVERTISSEMENT : RISQUE DE CHOC ÉLECTRIQUE.
DÉBRANCHEZ LE CORDON D’ALIMENTATION
DE LA PRISE ÉLECTRIQUE AVANT DE NETTOYERL’APPAREIL AFIN D’ÉVITER TOUT CHOC ÉLECTRIQUE
ACCIDENTEL OU RISQUE DE BRÛLURE. SEUL LE
FOURNISSEUR DE L’APPAREIL OU UN TECHNICIENQUALIFIÉ EST HABILITÉ À RETIRER LES CAPOTS OURÉPARER L’APPAREIL.
AVERTISSEMENT : PRENEZ TOUTES LESPRÉCAUTIONS NÉCESSAIRES POUR ÉVITER DE
MOUILLER LE CONCENTRATEUR D’OXYGÈNE ET
ASSUREZ-VOUS QU’AUCUN LIQUIDE NE PUISSEPÉNÉTRER À L’INTÉRIEUR DE L’APPAREIL, AFIN
D’ÉVITER TOUT DYSFONCTIONNEMENT OU ARRÊT
DE CELUI-CI, AINSI QU’UN RISQUE ACCRU DE CHOC
ÉLECTRIQUE OU DE BRÛLURE.
AVERTISSEMENT: IL PEUT ÊTRE DANGEREUXD’INTERCONNEXION DU COMPANION 5 AVECD’AUTRES ÉQUIPEMENTS.

AVERTISSEMENT : NETTOYEZ LE BOÎTIER, LE
PANNEAU DE COMMANDE ET LE CORDON
D’ALIMENTATION UNIQUEMENT À L’AIDE D’UN
DÉTERGENT DOUX APPLIQUÉ AU MOYEN D’UN
CHIFFON OU D’UNE ÉPONGE LÉGÈREMENTHUMIDE, PUIS ESSUYEZ TOUTES LES SURFACESPOUR LES SÉCHER. NE LAISSEZ AUCUN LIQUIDEPÉNÉTRER À L’INTÉRIEUR DE L’APPAREIL.
INSPECTEZ TOUT PARTICULIÈREMENT LA SORTIED’OXYGÈNE AU NIVEAU DU RACCORDEMENTDE LA CANULE AFIN DE VÉRIFIER L’ABSENCE DE
POUSSIÈRES, D’EAU OU AUTRES PARTICULES.
AVERTISSEMENT : VEILLEZ À NE PAS UTILISER
DE LIQUIDE DIRECTEMENT SUR L’APPAREIL. LA
LISTE DES AGENTS CHIMIQUES À ÉVITER INCLUT
DE FAÇON NON EXHAUSTIVE : LES PRODUITSALCOOLISÉS ET À BASE D’ALCOOL, LES PRODUITSÀ BASE DE CHLORE CONCENTRÉ (CHLORURE
D’ÉTHYLÈNE) ET LES PRODUITS À BASE D’HUILE(PINE-SOL®, LESTOIL®). CES PRODUITS NE DOIVENT
PAS ÊTRE UTILISÉS POUR NETTOYER LES PARTIESEN PLASTIQUE DU CONCENTRATEUR D’OXYGÈNEAU RISQUE DE LES ENDOMMAGER.

Élimination
Retournez toujours l’équipement Companion 5, ycompris tous les composants, à votre organisme desoins de santé pour qu’il soit correctement éliminé.
Vous pouvez également contacter vos services
municipaux locaux pour obtenir des instructions
concernant l’élimination appropriée de la batterie.

DEEE et RoHS
Ce symbole a pour objet d’inciter le propriétaire de l’équipement à le retourner à un centre derecyclage à la fin de son cycle de vie, conformément
à la directive relative aux déchets d’équipementsélectriques et électroniques (DEEE).

Nos produits sont conformes à la Directive sur lalimitation des substances dangereuses (RoHS). Ils ne
contiendront pas plus qu’une quantité infime de plombou d’autres substances dangereuses.
ATTENTION: Pour une élimination
appropriée, contactez votre fournisseurd’équipement ou une agence gouvernementale locale pour les instructionsd’élimination.

CAIRE et CAIRE Inc. sont des marques déposées de CAIRE Inc.
Veuillez visiter notre site Web ci-dessous pour obtenir une liste
complète des marques de commerce.
Marques: www.careinc.com/corporate/trademarks.
Copyright © 2020 CAIRE Inc.
CAIRE Inc. se réserve
le droit d’interrompre la commercialisation de ses produits ou demodifier les tarifs, les matériaux, les équipements, les niveaux dequalité, les descriptions, les caractéristiques et/ou les processus deses produits, et ce à tout moment, sans avertissement préalable etsans aucune autre obligation ni conséquence. Nous nous réservons
tous les droits non expressément stipulés dans le présent document,
selon les conditions applicables.
PN 15062781-C4 J | Manuel d’utilisation

35 - FRE

Companion 5
Steuerungselemente und Systemstatusanzeigen
Interne Symbole

02

Anwendungsteil des Typs BF(Schutzgrad gegen Stromschlag).
Reg.-Nr. 5333

Sauerstoffleistung

I/O Ein-/Ausschalter
Keine wartbaren Teile enthalten.
Abdeckung nicht öffnen.

Richtlinie 93/42/EWG des Rates für MedizinprodukteAutorisierter Vertreter in derEuropäischen Gemeinschaft
IEC 60417
Gerät der Klasse II, doppelt isoliert
Reg.-Nr. 5172

ISO 7000; graphische Symbole für die
Verwendung auf dem Gerät – Index und Übersicht
Lager-oder Betriebstemperaturbereich.
Reg.-Nr. 0632

ETL-Zertifizierung
Sicherheitsstandard CAN/CSA
C22.2 Nr. 601.1 M90 Medizinischeelektrische Geräte

Lagerung luftfeuchtigkeit bereich.
Reg.-Nr. 2620
Name und Adresse des Herstellers.
Reg.-Nr. 3082

FCC-Hinweis
21 CFR 801.15: Code of Federal Regulations, Titel 21

Katalognummer. Reg.-Nr. 2493

Nach den in den USA geltenden
Gesetzen darf dieses Gerät nur voneinem Arzt oder im Auftrag eines Arztes
verkauft werden.

Seriennummer. Reg.-Nr. 2498

Richtlinie 2012/19/EU des Rates für: Elektro- undElektronik-Altgeräte (WEEE-Richtlinie)

Herstellungsdatum. Reg.-Nr. 2497

Diese Seite nach oben. Reg.-Nr. 0623
Zerbrechlich, Vorsicht bei derHandhabung. Reg.-Nr. 0621
Vor Regen schützen, trocken
aufbewahren. Reg.-Nr. 0626
Lesen Sie vor der Verwendung dieseBedienungsanleitung. Reg.-Nr. 1641
Stapelgrenze nach Anzahl. Reg. Nr.
2403
ISO 7010: Graphische Symbole – Sicherheitsfarben
und Sicherheitszeichen – Registrierte
Sicherheitszeichen
Die Betriebsanleitung ist aufmerksam
durchzulesen. Reg.-Nr. M002
Warnung. Reg.-Nr. W001
Von Flammen, offenem Feuer undFunken fernhalten. Offene Zündquellen
und Rauchen verboten. Reg.-Nr. P003
Nicht in der Nähe des Geräts oderwährend dessen Betrieb rauchen.
Reg.-Nr. P002

36 - GER PN 15062781-C4 J | Bedienungsanleitung

WEEE-Richtlinie
EN 60601-1: Medizinische elektrische Geräte –Teil 1: Allgemeine Festlegungen für die
grundlegende Sicherheit und die wesentlichen
Leistungsmerkmale

IPX1

Tropfwassergeschütztes Gerät(Schutzart IPX1): Der Companion 5bietet Schutz gegen Schäden durch das
Eindringen von Flüssigkeiten. (IPX1
gemäß IEC 60529)

FCC-HINWEIS:

Dieses Gerät kann in Übereinstimmung mit Par.
15 der FCC-Regeln CYBLE-022001-00 enthalten,
einschließlich der Antenne 2450AT18B100
von Johnson Technology. Das Gerät erfüllt die
Anforderungen für die Zulassung modularer Sender,
wie im öffentlichen FCC-Hinweis DA00-1407
erläutert. Der Betrieb des Senders unterliegt den
folgenden beiden Bedingungen: (1) Dieses Gerät
kann keine störenden Interferenzen verursachen, und(2) Dieses Gerät muss alle empfangenen Störungentolerieren, einschließlich Störungen, die einen
unerwünschten Betrieb verursachen.
Dieses Produkt kann von einem oder mehreren USamerikanischen oder internationalen Patenten abgedeckt
sein. Auf unserer Webseite finden Sie eine Auflistung
der geltenden Patente.
Pat.: www.caireinc.com/corporate/patents/

Companion 5
Kurzanleitung
1. Auspacken des Companion 5

Companion 5

2. Kennenlernen des Companion 5
Lesen Sie sorgfältig alle Warnhinweise,
Vorsichtshinweise und zusätzlichen Informationen zum
Gerät im Rest dieser Anleitung. Machen Sie sich mit
den wichtigsten Komponenten des Companion 5 sowie
dem Bedienfeld vertraut.

VORDERSEITE
Außengehäuse

Netzschalter

LEDAnzeige

SauerstoffAuslassöffnung

Stundenzähler

Ständer für
Befeuchterflasche(Flasche ist
optional)

VolumenstrommessgerätRÜCKSEITE
Position desTragegriffs und
Lufteinlasses

Position desKühllufteinlasses

AbluftöffnungenPosition desLufteinlasses

PN 15062781-C4 J | Bedienungsanleitung

37 - GERCompanion 5
Komponenten des Companion 5
Außengehäuse

Netzschalter

Auslassöffnung

LED-Anzeige

Stundenzähler

Volumenstrommessgerät

Ständer für
Befeuchterflasche(Flasche ist
optional)

Stabiles Kunststoffgehäuse, dasdie internen Komponenten des
CAIRE Companion 5 schützt.
Ein-/Ausschalter zum Herstellen und
Unterbrechen der Stromversorgung
des Geräts. Das Symbol „I“bedeutet, dass das Gerät
eingeschaltet ist, das Symbol „O“,
dass es AUSGESCHALTET ist.
Hier verlässt der Sauerstoff
den Companion 5, nachdem
er gefiltert und konzentriert
wurde. Zur Versorgung desPatienten wird entweder eine
einlumige Nasenkanüle oder ein
standardmäßiger Sauerstoffschlauchangeschlossen. Der an den
Konzentrator angeschlossene
Sauerstoffschlauch ist maximal
15,2 m (50 ft) lang.
Verfügt über eine grüne, eine rote
und eine gelbe* LED-Leuchte.
Die grüne Leuchte steht für den
normalen Betrieb, die gelbe*
und rote Leuchte weisen auf
Alarmzustände hin. (weitere
Informationen finden Sie imAbschnitt „Alarmzustände“ indieser Anleitung).
Digitale Anzeige der verstrichenen
Betriebszeit des Konzentrators.
Zeigt auf eine zehntel Stunde genau
an und kann nicht zurückgesetzt
werden. Zeigt Diagnosealarmcodes
an, wenn der Konzentrator einen
Alarmzustand meldet.
Dient der Anpassung undAnzeige der Volumenstromrate
des zugeführten Sauerstoffs. Die
Volumenstromrate bewegt sichzwischen 0,5 und 5 Litern pro Minute(l/min) und lässt sich durch Drehen
des Knopfes anpassen.
Für das Anbringen einer
Befeuchterflasche. Der Ständer
umfasst ein Gummiband zur
Sicherung der Flasche im Ständer.
Eine Befeuchterflasche sorgt
für zusätzliche Feuchtigkeit im
Sauerstoffvolumenstrom. Die Flasche
ist mit destilliertem Wasser gefüllt
und die Flüssigkeit verhindert das
Austrocknen der Nasenschleimhaut.
*Nur Modelle mit Statusanzeige fürden Sauerstoffkonzentrator (Oxygen
Concentrator Status Indicator, OCSI)

38 - GER PN 15062781-C4 J | Bedienungsanleitung

3. Anbringen von Befeuchterflasche undKanüleWenn Sie eine Befeuchterflasche verwenden, befüllen
Sie diese zunächst bis zur vom Hersteller angezeigten
Höhe mit destilliertem Wasser. Setzen Sie die Flasche
in den dafür vorgesehenen Ständer ein und sichern
Sie sie mit dem Gummiband. Verbinden Sie dasEnde mit Gewindezapfen des Befeuchterschlauchs
mit der Flasche und das andere Ende mit der
Auslassöffnung des Companion 5. Schließen Sie den
Sauerstoffschlauch und/oder eine Nasenkanüle an derAuslassöffnung der Befeuchterflasche an.

Wird die Befeuchterflasche nicht verwendet, den
Sauerstoffschlauch und/oder die Nasenkanüle direkt an
der Auslassöffnung des Companion 5 anschließen.

Companion 5
4. Einschalten und Aufwärmen
Verbinden Sie das Wechselstrom-Kabel mit einer
Wechselstrom-Steckdose.

6. Einstellen des Volumenstroms auf dievorgeschriebenen WerteÜber den Volumenstromsteuerungsknopf können Sie
den vom Ihrem Arzt vorgeschriebenen Volumenstrom
einstellen.

Schalten Sie das Gerät mit dem Netzschalter ein.

Netzschalter

Wenn der Companion 5 korrekt eingeschaltet wird,
leuchtet die grüne Anzeige auf dem LED-Display.
Alle LED-Leuchten leuchten beim Einschalten auf.
Nachdem der Konzentrator aufgewärmt ist, leuchtet
nur noch die grüne Leuchte.
Hinweis: Nach dem ersten Einschalten leuchtet das
O2-Licht und „O2 UP“ wird auf dem Bildschirmangezeigt, bis die Konzentration die Spezifikationen
erreicht.

WARNUNG: DAS VORGESCHRIEBENESAUERSTOFFNIVEAU MUSS UNBEDINGTEINGEHALTEN WERDEN. EINSTELLUNGENDER FLUSSRATE DÜRFEN NUR
GEÄNDERT WERDEN, WENN DIES AUF
ANWEISUNG EINES APPROBIERTENARZTES ERFOLGT. DER GEBRAUCHDES SAUERSTOFFKONZENTRATORSWÄHREND DES SCHLAFS DARF NUR AUF
EMPFEHLUNG DURCH QUALIFIZIERTESKLINISCHES PERSONAL ERFOLGEN.
Atmen Sie normal über die Nasenkanüle.
So passen Sie die Volumenstromrate an:
Drehen Sie den Knopf gegen den Uhrzeigersinn, um
den Volumenstrom zu verstärken.
Drehen Sie ihn im Uhrzeigersinn, um den
Volumenstrom zu verringern.
Die Mitte der Kugel zeigt die Volumenstromrate an.
Siehe Bild unten.

StromanzeigeLED

Korrekt
PN 15062781-C4 J | Bedienungsanleitung

39 - GERCompanion 5
Wichtig!

Warn- und Sicherheitshinweise sind folgendermaßen
gekennzeichnet:
WARNUNG: WEIST AUF GEFAHREN HIN,DIE SCHWERE VERLETZUNGEN ZUR
FOLGE HABEN KÖNNEN.
VORSICHT: Wichtige Informationen
zur Vermeidung von Schäden am
Companion 5.
Hinweis: Besonders zu beachtende Informationen.

Gebrauchshinweise
Verwendungszweck

Der Companion 5 ist für die Verabreichung vonZusatzsauerstoff bestimmt. Das Gerät ist nicht fürlebenserhaltende Maßnahmen ausgelegt und bietet
keine Patientenüberwachung.
WARNUNG: DAS VORGESCHRIEBENESAUERSTOFFNIVEAU MUSS UNBEDINGTEINGEHALTEN WERDEN. EINSTELLUNGENDER FLUSSRATE DÜRFEN NUR GEÄNDERTWERDEN, WENN DIES AUF ANWEISUNGEINES APPROBIERTEN ARZTES ERFOLGT.
DER GEBRAUCH DES TRAGBARENSAUERSTOFFKONZENTRATORSWÄHREND DES SCHLAFS DARF NUR AUF
EMPFEHLUNG DURCH QUALIFIZIERTESKLINISCHES PERSONAL ERFOLGEN.
WARNUNG: DAMIT GEWÄHRLEISTET IST, DASS DIE
IHREM GESUNDHEITSZUSTAND ENTSPRECHENDETHERAPEUTISCHE SAUERSTOFFMENGE
VERABREICHT WIRD, DARF DER COMPANION 5
ERST EINGESETZT WERDEN, NACHDEM EINE ODERMEHRERE EINSTELLUNGEN INDIVIDUELL FÜR
SIE BEI IHREM SPEZIFISCHEN AKTIVITÄTSMASSBESTIMMT ODER VERSCHRIEBEN WURDEN. DER
COMPANION 5 DARF NUR IN VERBINDUNG MIT
DER SPEZIFISCHEN KOMBINATION BZW. MIT
DEN TEILEN UND ZUBEHÖRTEILEN VERWENDETWERDEN, DIE DEN SPEZIFIKATIONEN DESKONZENTRATORHERSTELLERS ENTSPRECHEN UND
DIE BEI BESTIMMUNG IHRER EINSTELLUNGEN
EINGESETZT WURDEN.
WARNUNG: DIESES GERÄT DARF NICHTFÜR LEBENSERHALTENDE MASSNAHMENEINGESETZT WERDEN. BEI GERIATRISCHEN,PÄDIATRISCHEN UND ANDEREN PATIENTEN,
DIE EIN UNWOHLSEIN BEIM GEBRAUCH DIESES
GERÄTS NICHT MITTEILEN KÖNNEN, SIND GGF.
ZUSÄTZLICHE ÜBERWACHUNGSMETHODEN ODEREIN DEZENTRALES ALARMSYSTEM NOTWENDIG,DAMIT INFORMATIONEN ZU UNWOHLSEIN
BZW. EINEM MEDIZINISCHEN NOTFALL AN
DAS VERANTWORTLICHE MEDIZINISCHEPERSONAL ÜBERMITTELT WERDEN KÖNNEN, UMSCHÄDIGUNGEN ZU VERMEIDEN. PATIENTEN MIT
HÖR- UND/ODER SEHBEHINDERUNG BENÖTIGEN
MÖGLICHERWEISE HILFE BEIM ÜBERWACHEN DER
ALARME.
40 - GER PN 15062781-C4 J | Bedienungsanleitung

WARNUNG: SCHWANGERE ODER STILLENDEFRAUEN SOLLTEN KEINE IN DIESEM HANDBUCH
EMPFOHLENEN ZUBEHÖRTEILE VERWENDEN, DA
SIE PHTHALATE ENTHALTEN KÖNNEN.
WARNUNG: LAUT US-GESETZEN DARFDIESES GERÄT NUR AUF ANWEISUNG EINESARZTES ODER EINES ANDEREN LIZENZIERTENGESUNDHEITSDIENSTLEISTERS VERKAUFT ODERVERLIEHEN WERDEN.

Gegenanzeigen
WARNUNG: DIE VERWENDUNGVON SAUERSTOFF OHNE ÄRZTLICHEVERSCHREIBUNG IST UNTER
UMSTÄNDEN GEFÄHRLICH. DAS GERÄTDARF NUR AUF VERSCHREIBUNG EINESARZTES VERWENDET WERDEN.
WARNUNG: DAS GERÄT DARF NICHT VERWENDETWERDEN, WENN BRENNBARE ANÄSTHETIKA
VORHANDEN SIND.
WARNUNG: WIE BEI JEDEM ELEKTRISCHEN
GERÄT KANN ES AUCH BEIM COMPANION 5 ZU
AUSFÄLLEN AUFGRUND VON UNTERBRECHUNGENDER STROMVERSORGUNG KOMMEN BZW. ES
NOTWENDIG SEIN, DAS GERÄT VON EINEMENTSPRECHEND GESCHULTEN TECHNIKER
WARTEN ZU LASSEN. DER COMPANION 5 IST
FÜR PATIENTEN, DIE AUFGRUND EINER SOLCHENVORÜBERGEHENDEN UNTERBRECHUNGGESUNDHEITLICHE SCHÄDEN DAVONTRAGEN
WÜRDEN, NICHT GEEIGNET.

Sicherheitsrichtlinien
WARNUNG: AN DIESEM GERÄT
DÜRFEN KEINE MODIFIKATIONEN
VORGENOMMEN WERDEN.
WARNUNG: FÜR DEN FALL EINES STROMAUSFALLS,ALARMS ODER MECHANISCHEN VERSAGENSEMPFIEHLT DER HERSTELLER EINE ALTERNATIVESAUERSTOFFZUFUHR. DEN ARZT ODERGERÄTEANBIETER FRAGEN, WELCHE ART VONRESERVESYSTEM BENÖTIGT WIRD.
WARNUNG: DIESES GERÄT LIEFERTHOCHKONZENTRIERTEN SAUERSTOFF, DER
SCHNELL ZU BRÄNDEN FÜHREN KANN. RAUCHENODER OFFENE FLAMMEN SIND INNERHALBVON ZWEI (2) METERN IM UMFELD VON(1) DIESEM GERÄT ODER (2) EINEM SAUERSTOFF
TRANSPORTIERENDEN ZUBEHÖRTEIL NICHTERLAUBT. DIE MISSACHTUNG DIESER WARNUNG
KANN SCHWERE BRÄNDE, SACH- BZW.
PERSONENSCHÄDEN ODER DEN TOD ZUR FOLGE
HABEN.

Companion 5
WARNUNG: BETREIBEN SIE DAS GERÄT NICHTIN EINEM EINGESCHRÄNKTEN ODER ENGEN
BEREICH (Z. B. EINEM KLEINEN KOFFER ODEREINER HANDTASCHE), IN DEM DIE BELÜFTUNG
UNZUREICHEND SEIN KÖNNTE. HIERDURCH KANNSICH DER SAUERSTOFFKONZENTRATOR ÜBERHITZENUND DIE LEISTUNG BEEINTRÄCHTIGT WERDEN.
WARNUNG: DER KONZENTRATOR DARFNICHT MIT RAUCH, SCHADSTOFFEN
ODER DÄMPFEN IN KONTAKT KOMMEN.
WARNUNG: DER EINSATZ EINIGER SAUERSTOFFFÜHRENDER ZUBEHÖRTEILE, DIE NICHTFÜR DIE VERWENDUNG MIT DIESEMSAUERSTOFFKONZENTRATOR ANGEGEBEN SIND,
KANN DESSEN LEISTUNG BEEINTRÄCHTIGEN.
AUF EMPFOHLENES ZUBEHÖR WIRD IN DIESERANLEITUNG VERWIESEN.
WARNUNG: IST DAS GERÄT HERUNTERGEFALLEN,WURDE ES BESCHÄDIGT ODER WASSERAUSGESETZT, SOLLTEN SIE SICH MIT IHREM
PFLEGEDIENST IN VERBINDUNG SETZEN, UM EINEINSPEKTION ODER EINE EVENTUELLE REPARATURVORNEHMEN ZU LASSEN. VERWENDEN SIE DENSAUERSTOFFKONZENTRATOR NICHT, WENN
STROMKABEL ODER STECKER BESCHÄDIGT SIND.
WARNUNG: STELLEN SIE SICHER, DASSLUFTEINLASS UND -AUSLASS NICHT BLOCKIERTWERDEN. FÜHREN SIE KEINE GEGENSTÄNDE IN DIEGERÄTEÖFFNUNGEN EIN UND LASSEN SIE NICHTSIN DIESE HINEINFALLEN. HIERDURCH KANN SICH
DER SAUERSTOFFKONZENTRATOR ÜBERHITZEN UND
DIE LEISTUNG BEEINTRÄCHTIGT WERDEN.
WARNUNG: DER OPTIONAL ERHÄLTLICHE
BEFEUCHTER DARF NICHT ÜBERFÜLLT WERDEN. DER
OPTIONAL ERHÄLTLICHE BEFEUCHTER DARF NUR
MIT WASSER BEFÜLLT WERDEN. DAS FÜLLNIVEAUDARF DIE VOM HERSTELLER DES BEFEUCHTERSANGEZEIGTE HÖHE NICHT ÜBERSCHREITEN.
WARNUNG: DAS GERÄT NUR NACH EINGEHENDEMSTUDIUM UND VERSTÄNDNIS DIESER ANLEITUNGVERWENDEN. SOLLTEN SIE DIE WARNUNGEN UND
ANWEISUNGEN NICHT VERSTEHEN, WENDEN SIE
SICH AN DEN BEREITSTELLER DIESES GERÄTES,BEVOR SIE VERSUCHEN, ES ZU VERWENDEN;
ANDERNFALLS KANN ES ZU VERLETZUNGEN ODERSCHÄDEN KOMMEN.
WARNUNG: BEI UNWOHLSEIN ODER EINEM
MEDIZINISCHEN NOTFALL SOFORT ÄRZTLICHE
HILFE ZURATE ZIEHEN.
WARNUNG: WIRD DER SAUERSTOFFKONZENTRATORAUSSERHALB DES ANGEGEBENENTEMPERATURBEREICHES VERWENDET,KANN DIES DIE FÄHIGKEIT DES GERÄTES
BEEINTRÄCHTIGEN, DIE ANGEGEBENENSAUERSTOFFKONZENTRATIONEN ZU ERZEUGEN.
FÜR TEMPERATURGRENZWERTE SIEHE DEN
ABSCHNITT „TECHNISCHE DATEN“ IN DIESERANLEITUNG.

WARNUNG: KEINE ÖL-, FETT-, ODERPETROLEUMBASIERTEN ODER ANDERENENTZÜNDLICHEN PRODUKTE MIT DEM
SAUERSTOFFKONZENTRATOR ODER DESSENSAUERSTOFF BEINHALTENDEM ZUBEHÖR
VERWENDEN. ES SOLLTEN AUSSCHLIESSLICHWASSERBASIERTE, SAUERSTOFFVERTRÄGLICHELOTIONEN ODER SALBEN VERWENDET WERDEN.
SAUERSTOFF BESCHLEUNIGT DIE VERBRENNUNG
ENTZÜNDLICHER SUBSTANZEN.
WARNUNG: DER SAUERSTOFFKONZENTRATORSOLLTE NICHT DIREKT NEBEN ANDEREN GERÄTENODER MIT ANDEREN GERÄTEN GESTAPELTVERWENDET WERDEN. IST EINE SOLCHE SITUATIONNICHT ZU VERMEIDEN, MUSS DAS GERÄT AUF
ORDNUNGSGEMÄSSEN BETRIEB HIN ÜBERWACHTWERDEN.
WARNUNG: FÜR DEN FALL EINESSTROMAUSFALLS, ALARMS ODERMECHANISCHEN VERSAGENS EMPFIEHLTDER HERSTELLER EINE ALTERNATIVESAUERSTOFFZUFUHR. DEN ARZT ODERGERÄTEANBIETER FRAGEN, WELCHE ART
VON RESERVESYSTEM BENÖTIGT WIRD.
WARNUNG: DIESES GERÄT LIEFERTHOCHKONZENTRIERTEN SAUERSTOFF, DER
SCHNELL ZU BRÄNDEN FÜHREN KANN. RAUCHENODER OFFENE FLAMMEN SIND IM RAUM, IN DEM
SICH (1) DAS GERÄT ODER (2) EIN SAUERSTOFF
TRANSPORTIERENDES ZUBEHÖRTEIL BEFINDEN,NICHT ERLAUBT. DIE MISSACHTUNG DIESERWARNUNG KANN SCHWERE BRÄNDE, SACH- BZW.
PERSONENSCHÄDEN ODER DEN TOD ZUR FOLGE
HABEN.
WARNUNG: DER EINSATZ EINIGER SAUERSTOFFFÜHRENDER ZUBEHÖRTEILE, DIE NICHTFÜR DIE VERWENDUNG MIT DIESEMSAUERSTOFFKONZENTRATOR ANGEGEBEN SIND,
KANN DESSEN LEISTUNG BEEINTRÄCHTIGEN.
AUF EMPFOHLENES ZUBEHÖR WIRD IN DIESERANLEITUNG VERWIESEN.
WARNUNG: IST DAS GERÄT HERUNTERGEFALLEN,WURDE ES BESCHÄDIGT ODER WASSERAUSGESETZT, SOLLTEN SIE SICH MIT IHREM
PFLEGEDIENST IN VERBINDUNG SETZEN, UM EINEINSPEKTION ODER EINE EVENTUELLE REPARATURVORNEHMEN ZU LASSEN. VERWENDEN SIE DENSAUERSTOFFKONZENTRATOR NICHT, WENN
STROMKABEL ODER STECKER BESCHÄDIGT SIND.
WARNUNG: VERWENDEN SIE AUSSCHLIESSLICH
VOM HERSTELLER EMPFOHLENE ERSATZTEILE,UM EINE ORDNUNGSGEMÄSSE FUNKTIONZU GEWÄHRLEISTEN UND BRAND- SOWIEVERBRENNUNGSGEFAHR ZU VERMEIDEN.
WARNUNG: SCHMIEREN SIE KEINE BESCHLÄGE,ANSCHLÜSSE, SCHLÄUCHE ODER ANDEREZUBEHÖRTEILE DES SAUERSTOFFKONZENTRATORS,
UM BRAND- SOWIE VERBRENNUNGSGEFAHR ZU
VERMEIDEN.

PN 15062781-C4 J | Bedienungsanleitung

41 - GERCompanion 5
WARNUNG: DAS RAUCHEN WÄHREND DER
SAUERSTOFFVERABREICHUNG IST DER
HAUPTGRUND FÜR BRANDVERLETZUNGEN UND-TODE. DIESE SICHERHEITSWARNUNGEN MÜSSEN
UNBEDINGT BEFOLGT WERDEN:
WARNUNG: RAUCHEN, KERZEN ODER OFFENE
FLAMMEN SIND IN DEM RAUM, IN DEM SICH DASGERÄT ODER EIN SAUERSTOFF BEFÖRDERNDESZUBEHÖRTEIL BEFINDEN, VERBOTEN.
WARNUNG: RAUCHEN WÄHREND DES
TRAGENS EINER SAUERSTOFFKANÜLE KANN ZUGESICHTSVERBRENNUNGEN UND ZUM TODEFÜHREN.
WARNUNG: WENN SIE RAUCHEN, KÖNNEN
DIESE 3 SCHRITTE IHR LEBEN RETTEN: DEN
SAUERSTOFFKONZENTRATOR AUSSCHALTEN, DIEKANÜLE ABNEHMEN UND DEN RAUM, IN DEM DASGERÄT SICH BEFINDET, VERLASSEN.
WARNUNG: SCHILDER MIT DER AUFSCHRIFT„RAUCHEN VERBOTEN – SAUERSTOFF INGEBRAUCH“ MÜSSEN IM HAUS ODER AN DEM
ORT, AN DEM DER SAUERSTOFF VERWENDETWIRD, DEUTLICH ERKENNBAR ANGEBRACHT SEIN.
PATIENTEN UND PFLEGEPERSONAL MÜSSEN ÜBERDIE GEFAHREN DES RAUCHENS IN DER NÄHE ODERWÄHREND DES GEBRAUCHS VON MEDIZINISCHEMSAUERSTOFF AUFGEKLÄRT WERDEN.
WARNUNG: DEN SAUERSTOFFKONZENTRATORNICHT BEI VORHANDENSEIN ENTZÜNDLICHER GASE
VERWENDEN. DIES KANN SCHNELL ZU BRÄNDEN
FÜHREN, DIE SACHSCHÄDEN, VERLETZUNGEN ODERDEN TOD NACH SICH ZIEHEN KÖNNEN.
WARNUNG: EIN ABNEHMEN DER KANÜLE UND
EINE LAGERUNG AUF KLEIDUNG, BETTWÄSCHE,SOFAS ODER ANDEREM POLSTERMATERIALFÜHRT BEI VORHANDENSEIN EINER ZIGARETTE,WÄRMEQUELLE ODER FLAMME ZU EINERVERPUFFUNG.
WARNUNG: LASSEN SIE EINE KANÜLE NICHT AUF
ODER UNTER KLEIDUNG, BETTWÄSCHE ODERSTUHLPOLSTERN LIEGEN. WENN DAS GERÄT
EINGESCHALTET IST, ABER NICHT BETRIEBEN WIRD,
KANN DER SAUERSTOFF ENTZÜNDLICH WIRKEN.
DAS GERÄT AUSSCHALTEN, WENN ES NICHT INGEBRAUCH IST.
WARNUNG: NICHT VERSUCHEN, DASGERÄT – ABGESEHEN VON DEN IN
DIESER ANLEITUNG AUFGEFÜHRTENSTÖRUNGSBEHEBUNGEN – ZU WARTEN.
ENTFERNEN SIE KEINE ABDECKUNG.
NUR DER BEREITSTELLER DESGERÄTES ODER EIN QUALIFIZIERTERWARTUNGSTECHNIKER DÜRFEN DIE
ABDECKUNGEN ENTFERNEN ODER DASGERÄT WARTEN.

42 - GER PN 15062781-C4 J | Bedienungsanleitung

WARNUNG: VERWENDEN SIE NUR VOM
HERSTELLER EMPFOHLENES ZUBEHÖR. DIE
VERWENDUNG ANDEREN ZUBEHÖRS KANNZU EINER SCHWEREN BESCHÄDIGUNG DESSAUERSTOFFKONZENTRATORS SOWIE ZUM
ERLÖSCHEN DER GARANTIE FÜHREN.
VORSICHT: Trennen Sie den
Wechselstrom-Stecker immer zuerst
von der Wandsteckdose, bevor
Sie das Wechselstrom-Kabel vom
Sauerstoffkonzentrator trennen.
VORSICHT: Stromkabel und Sauerstoffschläuche soplatzieren, dass eine Stolpergefahr vermieden wird.
VORSICHT: Alle Anweisungen desHerstellers befolgen, damit die Garantie
ihre Gültigkeit behält.
Hinweis: Tragbare und mobile HFKommunikationsausrüstung kann medizinischeelektrische Geräte beeinträchtigen.
WARNUNG: STELLEN SIE DIE SAUERSTOFFVERSORGUNGSROHREN UND STROMKABEL IMMERSO AUF, DASS EINE STOLPERGEFAHR ODER EINEMÖGLICHE VERSEHENTLICHE STRANGULATIONVERMIEDEN WIRD.
WARNUNG: DIESES PRODUKT EXPONIERT SIE
GEGENÜBER CHEMIKALIEN, WIE Z. B. NICKEL, DAS
IM STAAT KALIFORNIEN ALS URSACHE FÜR KREBSEINGESTUFT WIRD. WEITERE INFORMATIONEN DAZUFINDEN SIE UNTER WWW.P65WARNINGS.CA.GOV.
WARNUNG: WENN BEI DIESEM GERÄT EIN
SCHWERWIEGENDER VORFALL AUFTRITT, SOLLTE
DER BENUTZER DEN VORFALL UNVERZÜGLICH
DEM ANBIETER UND / ODER DEM HERSTELLERMELDEN. EIN SCHWERWIEGENDER VORFALL IST
DEFINIERT ALS EINE VERLETZUNG, DER TOD ODERDIE MÖGLICHKEIT, EINE VERLETZUNG / DEN TODZU VERURSACHEN, FALLS DER VORFALL ERNEUTAUFTRITT. DER BENUTZER KANN DEN VORFALL
AUCH DER ZUSTÄNDIGEN BEHÖRDE DES LANDESMELDEN, IN DEM DER VORFALL AUFGETRETEN IST.

Companion 5
Technische Daten
Die folgende Tabelle bietet zum ordnungsgemäßen
Gebrauch des Gerätes wichtige Informationen
bezüglich der empfohlenen Betriebsumgebungen oder
Betriebsbedingungen.

Technische Daten
Durchflussmengen
O2-Konzentration
Abmessungen0,5–5,0 l/min
90 % (+5,5 %/-3 %)54,6x31,8x34,2 cm(21,5x12,5x13,5”)
Gewicht
16,3 kg (36 Pfund)
Schalldruckpegel
50 dB(A) bei 2,0 l/min
Stromverbrauch
285 W bei 2 l/min,
350 W Maximum
Betriebstemperatur 5 °C bis 40 °C
Relative
15% - 90 % bei 28 °C
Luftfeuchtigkeit(82,4 °F) Taupunkttemperatur
Lagertemperatur-25 °C bis 70 °C
Lagerfeuchtigkeit 0% - 90 %, nicht
kondensierend
Stromversorgung
Verwenden Sie keine
Verlängerungskabel.
Verwenden Sie keine
Steckdosen, die über einen
Schalter ein- und ausgeschaltet
werden können.
Einsatzhöhe-382 bis 3.011 Meter
Betriebsdauer
bis zu 24 Stunden pro Tag
Die erwartete Lebensspanne des Geräts beträgt
mindestens 5 Jahre.
WARNUNG: DIE VERWENDUNGEINES GERÄTS AUSSERHALB DER
ANGEGEBENEN BETRIEBSBEDINGUNGENWIRD VORAUSSICHTLICH DIE
DURCHFLUSSRATE UND DEN
PROZENTSATZ DES SAUERSTOFFS UND
FOLGLICH DIE QUALITÄT DER THERAPIENEGATIV BEEINFLUSSEN.
Hinweis: Wurde der Sauerstoffkonzentrator füreinen längeren Zeitraum außerhalb seiner üblichen
Betriebstemperaturen gelagert, sollte dem Gerät vor
dem Anschalten Zeit gelassen werden, eine übliche
Betriebstemperatur anzunehmen (siehe den Abschnitt„Technische Daten“ in dieser Anleitung).

Companion 5

Wählen Sie den Standort für das Gerät so, dass weder
Rauch, Dämpfe noch Schadstoffe aufgenommen
werden können. Durch die korrekte Aufstellung
des Geräts sollte der Einlass von Luft durch diedrei Lufteinlass-Positionen oben an der Rückseite
und unterhalb des Gehäuses gewährleistet sein,
und außerdem sicherstellen, dass Abluft über die
Abluftöffnung unten links am Gerät nach außen
strömen kann.
Lufteinlässe befinden sich oben an der Rückseite undunten am Konzentrator. Stellen Sie das Gerät mit
einem Abstand von mindestens 30 cm zu Wänden,
Möbeln, Vorhängen oder anderen Gegenständen auf.
Platzieren Sie das Gerät so, dass die akustischen
Alarme hörbar sind.
Legen Sie den Sauerstoffversorgungsschlauch so, dassdieser nicht knicken oder blockiert werden kann.
WARNUNG: DEN
SAUERSTOFFKONZENTRATOR NICHTBEI VORHANDENSEIN ENTZÜNDLICHERGASE VERWENDEN. DIES KANNSCHNELL ZU BRÄNDEN FÜHREN, DIE
SACHSCHÄDEN, VERLETZUNGENODER DEN TOD NACH SICH ZIEHENKÖNNEN. KEINE ÖL-, FETT-, ODERPETROLEUMBASIERTEN ODER ANDERENENTZÜNDLICHEN PRODUKTE MIT DEM
SAUERSTOFFKONZENTRATOR ODERDESSEN SAUERSTOFF BEINHALTENDEM
ZUBEHÖR VERWENDEN. ES SOLLTENAUSSCHLIESSLICH WASSERBASIERTE,SAUERSTOFFVERTRÄGLICHE LOTIONENODER SALBEN VERWENDET WERDEN.
SAUERSTOFF BESCHLEUNIGT DIEVERBRENNUNG ENTZÜNDLICHERSUBSTANZEN.
WARNUNG: DIESES GERÄT LIEFERTHOCHKONZENTRIERTEN SAUERSTOFF, DER
SCHNELL ZU BRÄNDEN FÜHREN KANN. RAUCHENODER OFFENE FLAMMEN SIND IM RAUM, IN DEM
SICH (1) DAS GERÄT ODER (2) EIN SAUERSTOFF
TRANSPORTIERENDES ZUBEHÖRTEIL BEFINDEN,NICHT ERLAUBT. DIE MISSACHTUNG DIESERWARNUNG KANN SCHWERE BRÄNDE, SACH- BZW.
PERSONENSCHÄDEN ODER DEN TOD ZUR FOLGE
HABEN.

Ordnungsgemäße Aufstellung desPN 15062781-C4 J | Bedienungsanleitung

43 - GERCompanion 5
Gebrauchsanweisungen
Vor der Inbetriebnahme

Diese Bedienungsanleitung ist dazu bestimmt, Sie bei
der Verwendung und Wartung des Geräts zu unterstützen.
Sollten Sie Fragen oder Probleme haben, setzen Sie sichbitte mit Ihrem Pflegedienst in Verbindung.
Wichtig! Der Companion 5 darf ohne vorheriges
Lesen des Kapitels „Sicherheitsrichtlinien" in dieser
Anleitung NICHT betrieben werden. Befolgen Sie alle
Gebrauchshinweise. Beachten Sie dabei insbesondere
auch alle Warnhinweise, die auf dem Gerät und in derBedienungsanleitung zu finden sind. Zur Verringerungvon Brand- und Verletzungsgefahren sowie vonschweren Schäden am Companion 5 müssen zudem alle
Sicherheitsvorkehrungen beachtet werden.
Der CAIRE Companion 5 erzeugt nicht selbst
Sauerstoff. Er erzeugt hochkonzentrierten Sauerstoff
aus der umgebenden Raumluft und führt ihn dem
Patienten zu. Raumluft weist von Natur aus nur circa
21 Volumenprozent Sauerstoff auf.
Die Raumluft wird durch einen Kompressor in den
Konzentrator gesogen. Die Luft durchläuft anschließend
eine Reihe an Filtern und einen Molekular-Sieb-Filter,
der Stickstoffmoleküle chemisch absorbiert. Die
sauerstoffangereicherte Luft, die aus dem Konzentrator
austritt, enthält daher 87 bis 95 Volumenprozent
Sauerstoff.
WARNUNG: REINIGEN SIE GEHÄUSE,BEDIENFELD UND STROMKABEL NUR MIT
EINEM SANFTEN HAUSHALTSREINIGER,
DER MIT EINEM FEUCHTEN (NICHTNASSEN) TUCH ODER SCHWAMM
AUFGETRAGEN WIRD. WISCHEN SIE
ANSCHLIESSEND ALLE OBERFLÄCHEN
TROCKEN. KEINE FLÜSSIGKEITEN IN
DAS INNERE DES GERÄTES GELANGENLASSEN. ACHTEN SIE BESONDERSDARAUF, DASS DER SAUERSTOFFAUSLASSFÜR DEN ANSCHLUSS DER KANÜLE FREIVON STAUB, WASSER UND ANDEREN
PARTIKELN BLEIBT.
VORSICHT: Stellen Sie sicher, dass
Lufteinlass und -auslass nicht blockiert
werden. Hierdurch kann sich der
Sauerstoffkonzentrator überhitzen
und die Leistung beeinträchtigt
werden. Betreiben Sie den
Sauerstoffkonzentrator nicht ohne
eingesetzten Lufteinlassfilter. Sollte einzweiter Filter mitgeliefert worden sein,setzen Sie den Ersatzfilter ein, bevor Sie
den anderen Filter reinigen. Reinigen
Sie den verschmutzten Filter in warmem
Seifenwasser und trocknen Sie ihn vor
der Verwendung gründlich ab.
44 - GER PN 15062781-C4 J | Bedienungsanleitung

Schritt 1: Aufstellen des Companion 5 für die
Verwendung
Stellen Sie den Companion 5 in einer gut belüfteten undgut beleuchteten Umgebung auf. Stellen Sie sicher, dassdie Lufteinlass- und Abluftöffnungen nicht blockiertsind. Platzieren Sie den Companion 5 so, dass alleakustischen und visuellen Signale leicht wahrgenommen
werden können.
Stecken Sie das Netzkabel des Geräts in eine
Wechselstrom-Steckdose.
Schritt 2: Wird die Befeuchterflasche NICHT verwendet:

Position desLufteinlasses

Position desKühllufteinlasses

AbluftöffnungenPosition desLufteinlasses

a. Verbinden Sie eine Nasenkanüle odereinen Sauerstoffschlauch direkt mit der
Sauerstoffauslassöffnung.

b. Fahren Sie mit Schritt 4 fort.
Schritt 3: Wird die Befeuchterflasche verwendet:a. Schrauben Sie den Deckel von der Befeuchterflasche.
b. Befüllen Sie die Flasche mit destilliertem Wasser.
Achten Sie darauf, dass der Füllstand zwischen dem auf
der Flasche angegebenen Höchst- und Mindestniveau
liegt.
c. Schrauben Sie den Deckel auf die Befeuchterflasche.
Achten Sie darauf, dass kein Wasser austritt.
d. Schließen Sie eine Nasenkanüle oder einen
standardmäßigen Sauerstoffschlauch an die
Auslassöffnung der Befeuchterflasche an.
Hinweis: Der Einsatz eines Befeuchters, der nicht fürdie Verwendung mit diesem Konzentrator angegeben
ist, kann dessen Leistung beeinträchtigen.

Companion 5
g. Setzen Sie die Flasche in den dafür vorgesehenen
Ständer ein.
h. Sichern Sie die Flasche mit dem Gummiband im
Ständer.

e. Verbinden Sie das Ende mit Gewindezapfen
des Befeuchterschlauchs mit dem Deckel derBefeuchterflasche.

i. Fahren Sie mit Schritt 4 fort.
Schritt 4: Einschalten und Aufwärmen
a. Stecken Sie den Stecker des CAIRE Companion 5 ineine Wechselstrom-Steckdose.
WARNUNG: PRÜFEN SIE DASSTROMKABEL VOR DER VERWENDUNGAUF SCHÄDEN. SOLLTE DAS KABELBESCHÄDIGT SEIN, DARF ES NICHTIN EINE STECKDOSE GESTECKT ODERZUM BETRIEB DES KONZENTRATORSVERWENDET WERDEN.

f. Verbinden Sie das andere Ende desBefeuchterschlauchs mit der Auslassöffnung des
CAIRE Companion 5.

b. Schalten Sie den Netzschalter ein (Symbol l). Wenn
der Companion 5 korrekt eingeschaltet wird, leuchtet
eine grüne Anzeige auf dem LED-Display. Alle LEDLeuchten leuchten beim Einschalten auf. Nachdem
der Konzentrator aufgewärmt ist, leuchtet nur nochdie grüne Leuchte.
Hinweis: Warten Sie nach dem ersten Einschalten
bis zu zehn (10) Minuten, bis das Gerät seine
Betriebsbereitschaft erreicht hat.

PN 15062781-C4 J | Bedienungsanleitung

45 - GERCompanion 5
WARNUNG: DAS VORGESCHRIEBENESAUERSTOFFNIVEAU MUSS UNBEDINGTEINGEHALTEN WERDEN. EINSTELLUNGENDER FLUSSRATE DÜRFEN NUR
GEÄNDERT WERDEN, WENN DIES AUF
ANWEISUNG EINES APPROBIERTENARZTES ERFOLGT. DER GEBRAUCHDES SAUERSTOFFKONZENTRATORSWÄHREND DES SCHLAFS DARF NUR AUF
EMPFEHLUNG DURCH QUALIFIZIERTESKLINISCHES PERSONAL ERFOLGEN.
Schritt 5: Anpassen der Volumenstromratea. Drehen Sie den Volumenstromsteuerungsknopf
auf die von Ihrem Arzt vorgeschriebene
Sauerstoffvolumenstromrate (LPM).

Schritt 6: Überprüfen des Volumenstroms und
normales Atmena. Überprüfen Sie, ob tatsächlich Sauerstoff durch die
Nasenkanüle strömt und dass der Schlauch nicht
umgeknickt, verbogen oder verstopft ist.
Hinweis: Sicherstellen, dass die Kanüle vollständig
eingeführt ist und fest sitzt. Der Sauerstofffluss zu den
Zinken der Nasenkanüle sollte hörbar oder fühlbar sein.
b. Führen Sie die Nasenkanüle korrekt ein und atmen Sie
normal durch die Kanüle.
Schritt 7: Ausschalten
a. Schalten Sie den Netzschalter AUS (Symbol O), wenn
der CAIRE Companion 5 nicht mehr verwendet wird.

So passen Sie die Volumenstromrate an:
Drehen Sie den Knopf gegen den Uhrzeigersinn, um den
Volumenstrom zu verstärken.
Drehen Sie ihn im Uhrzeigersinn, um den Volumenstrom
zu verringern.
Die Mitte der Kugel zeigt die Volumenstromrate an. In
dem unten stehenden Bild ist eine Durchflussmenge von4,0 l/min angegeben.

VORSICHT: Verwenden Sie nur vom
Hersteller empfohlenes Zubehör. Die
Verwendung anderen Zubehörs kannzu einer schweren Beschädigung desSauerstoffkonzentrators sowie zum
Erlöschen der Garantie führen.
VORSICHT: Keine Verlängerungskabel für dieses
Gerät verwenden und nicht zu viele Stecker mit
derselben Steckdose verbinden. Die Verwendungeines Verlängerungskabels kann die Leistungdes Gerätes beeinträchtigen. Die Verbindungzu vieler Stecker mit einer Steckdose kann den
Schaltschrank überlasten und zum Auslösen vonTrennschalter/Sicherung oder bei Fehlfunktion vonTrennschalter/Sicherung zum Ausbruch eines Feuers
führen.
VORSICHT: Die Verwendung von anderen Kabeln
und Adaptern als den angegebenen, mit Ausnahme
der Kabel und Adapter, die vom Hersteller desmedizinischen elektrischen Geräts als Ersatzteile
für interne Komponenten verkauft werden, kannzu erhöhten Emissionen oder einer reduzierten
Störfestigkeit des Sauerstoffkonzentrators führen.

Korrekt

46 - GER PN 15062781-C4 J | Bedienungsanleitung

WARNUNG: STELLEN SIE SICHER, DASSLUFTEINLASS UND -AUSLASS NICHTBLOCKIERT WERDEN. HIERDURCH KANNSICH DER SAUERSTOFFKONZENTRATORÜBERHITZEN UND DIE LEISTUNG
BEEINTRÄCHTIGT WERDEN.

Companion 5
Hinweis: An den Geräteanbieter: Das folgende
Zubehör für die Sauerstoffverabreichung wird zur
Verwendung mit dem Companion 5 empfohlen:• Nasenkanüle: CAIRE-Teilenummer CU002-1• Luftbefeuchteradapterschlauch: CAIRE-Teilenummer
20843882• Luftbefeuchterflasche: CAIRE-Teilenummer HU003-1• Brandschutz: CAIRE-Teilenummer 20629671
Für die Verwendung mit einer Kanüle wird eine
Feuerpause benötigt.
• CAIRE bietet einen Brandschutz an, der in
Verbindung mit dem Sauerstoffkonzentrator
verwendet werden soll. Die Brandschutzklemmeist eine thermische Zündschnur, um den Gasfluss
zu stoppen, falls die nachgeschaltete Kanüle oderder Sauerstoffschlauch entzündet werden und zur
Brandschutzstelle verbrennen. Es wird in Linie mit
der Nasenkanüle oder dem Sauerstoffschlauch
zwischen dem Patienten und dem Sauerstoffauslassdes Companion 5 platziert. Beziehen Sie sich für die
ordnungsgemäße Verwendung des Brandschutzes
immer auf die Anweisungen des Herstellers (imLieferumfang jedes Brandschutz-Kits enthalten).
• Weitere empfohlene Zubehörteile finden Sie im
Zubehörkatalog (PN MLLOX0010) unter www.caireinc.
com.

PN 15062781-C4 J | Bedienungsanleitung

47 - GERCompanion 5
Warnzustände
Der CAIRE Companion 5 besitzt zwei oder drei (beiOCSI-Modellen) LED-Leuchten und einen akustischen
Alarm, um Sie auf Störungen des Konzentrators
hinzuweisen.
1) Grüne LED () – zeigt normalen Betrieb an.

2) Rote LED ( ) – zeigt eine Systemstörung an.
Genauere Informationen finden Sie in der unten
stehenden Tabelle zu den Alarmzuständen.
3) Gelbe LED ( ) – zeigt eine niedrige
Sauerstoffkonzentration an. Genauere
Informationen finden Sie in der unten stehenden
Tabelle zu den Alarmzuständen.
In der Tabelle zu den Alarmzuständen auf der nächsten
Seite finden Sie die unterschiedlichen Alarmzustände,die der Konzentrator anzeigen kann. Angegeben sindmögliche Ursachen sowie die im Falle eines Alarms zu
treffenden Maßnahmen. Sollte ein Alarm ausgegeben
werden, sollten Sie sich schnellstmöglich mit Ihrem
Gesundheitsdienstleister in Verbindung setzen.
WARNUNG: IGNORIEREN SIE KEINEN
ALARM
WARNHINWEISE: FÜR DEN FALL EINESSTROMAUSFALLS, ALARMS ODERMECHANISCHEN VERSAGENS EMPFIEHLTDER HERSTELLER EINE ALTERNATIVESAUERSTOFFZUFUHR. DEN ARZT ODERGERÄTEANBIETER FRAGEN, WELCHE ART
VON RESERVESYSTEM BENÖTIGT WIRD.

48 - GER PN 15062781-C4 J | Bedienungsanleitung

Grün

Rot

Gelb

Oben: Companion 5

Companion 5
Tabelle zu Alarmzuständen
Akustischer
Alarm

Farbige
LED

Aus
Grün

Wechselnd
ROT

Wechselnd

UrsacheDer Konzentrator
funktioniert
ordnungsgemäß.

Überprüfen Sie, ob das Kabel des Konzentrators
in eine stromführende Steckdose eingesteckt ist.
Probieren Sie eine andere Steckdose aus. Sollte
das Problem fortbestehen, schließen Sie eine
Reservesauerstoffversorgung an und wenden Sie sichan Ihren Gesundheitsdienstleister.

Systemstörung

Schalten Sie den Netzschalter sofort aus. Ziehen Sie
das Kabel aus der Steckdose. Warten Sie 5 Minuten.
Stecken Sie das Kabel wieder in die Steckdose
und schalten Sie den Konzentrator ein. Sollte der
Alarm weiterhin ausgegeben werden, benötigen Sie
Unterstützung vom technischen Dienst. Schließen Sieeine Reservesauerstoffversorgung an und setzen Sie
sich umgehend mit Ihrem Gesundheitsdienstleister
in Verbindung.

Volumenstromrate zu hochoder zu niedrig

1. Stellen Sie sicher, dass die Kanüle weder
umgeknickt noch verstopft ist. Bei Verwendung einer
Befeuchterflasche ist sicherzustellen, dass dieseordnungsgemäß befüllt ist und keine Blockierung
verursacht.
2. Stellen Sie sicher, dass der Companion 5
ordnungsgemäß belüftet wird. Das Gerät mussmindestens 30 cm (12 Zoll) von jeder Art vonOberfläche entfernt sein, um ein Blockieren derÖffnungen zu verhindern.
3. Sollte das Problem fortbestehen, wechseln Sie
zu einer alternativen Sauerstoffquelle und setzen
Sie sich mit Ihrem Gesundheitsdienstleister in
Verbindung.

ROT

Wechselnd
Gelb

Wechselnd
Gelb

Keine

Der Konzentrator hat an
Leistung verloren, derNetzschalter ist aber nocheingeschaltet.

ROT

Wechselnd

Maßnahme

Der Companion 5 hat ein
niedriges Sauerstoffniveau
festgestellt.

Der Companion 5 hat ein
niedriges Sauerstoffniveau
festgestellt.

1. Stellen Sie sicher, dass der Lufteinlassfilter und
Abluftpositionen nicht verstopft oder verengt ist.
2. Vergewissern Sie sich, dass der Companion 5
in einer gut belüfteten Umgebung aufgestellt ist.
Stellen Sie sicher, dass zwischen der Rückseite desCompanion und anderen Gegenständen (Möbel,Vorhänge etc.) ein Mindestabstand von 30 cm
eingehalten wird.
3. Sollte das Problem fortbestehen, wechseln Sie zu
einer alternativen Sauerstoffquelle und setzen Sie
sich umgehend mit Ihrem Gesundheitsdienstleister
in Verbindung.

PN 15062781-C4 J | Bedienungsanleitung

49 - GERCompanion 5
Fehlerbehebung
Wir empfehlen, dass Sie sich an Ihren
Gesundheitsdienstleister wenden, wenn Ihr
Konzentrator einen Alarm ausgibt oder wenn Sie
das Gefühl haben, dass er nicht ordnungsgemäß
funktioniert. Versuchen Sie nicht, das Gerät selbst zu
reparieren. Reparaturen dürfen nur von geschulten

Servicetechnikern vorgenommen werden. Sie können
bei Problemen mit Ihrem CAIRE Companion 5 jedochbestimmte Schritte zur Fehlerbehebung befolgen.
Diese finden Sie in der folgenden Tabelle. Führen
Sie die Ihrem Problem entsprechenden Schritte innumerischer Reihenfolge aus.

Störungsbehebungstabelle für den Benutzer
Problem

Der Konzentrator
ist eingeschaltet,
läuft jedoch nicht.

Es fließt nur
wenig oder kein
Sauerstoff.

Mögliche Ursache

Maßnahme

1. Das Netzkabel ist nicht mit einer
Steckdose verbunden.

1. Stecken Sie das Netzkabel ein.

2. Die Steckdose führt keinen Strom.

2. Überprüfen Sie die Sicherung und den Stromkreis.
Probieren Sie eine andere Steckdose aus.

3. Interner Fehler.

3. Schließen Sie eine Reservesauerstoffversorgung
an. Setzen Sie sich umgehend mit Ihrem
Gesundheitsdienstleister in Verbindung.

1. Die Volumenstromsteuerungseinstellung
wurde geändert.

1. Überprüfen Sie, ob derVolumenstromsteuerungsknopf auf die von Ihrem
Arzt vorgeschriebene LPM-Einstellung eingestellt ist.

2. Der Schlauch wurde getrennt.

2. Überprüfen Sie, ob der Schlauch anallen Anschlussstellen (Auslassöffnung,Befeuchterflasche, Wasserabscheider usw.) fest
angeschlossen ist. Stellen Sie die Verbindungen bei
Bedarf korrekt her.

3. Der Sauerstoffschlauch oder die Kanüle
ist umgeknickt.

3. Prüfen Sie den Schlauch auf Knicke oder
Verstopfungen. Bei Bedarf reparieren.

4. Die Kanüle oder der Schlauch ist
undicht.

4. Überprüfen Sie den Schlauch und die Kanüle
auf undichte Stellen. Dafür können Sie hören oderfühlen, ob Luft entweicht. Tauschen Sie undichte
Elemente aus.

5. Der Luftfluss in den Konzentrator wird
behindert.

5. Überprüfen Sie, ob die Einlassöffnungen desKonzentrators blockiert werden. Stellen Sie sicher,dass er sich in einem Abstand von mindestens
30 cm zu Möbeln, Wänden oder anderen
Gegenständen befindet.

6. Interner Fehler.

6. Schließen Sie eine Reservesauerstoffversorgung
an. Setzen Sie sich umgehend mit Ihrem
Gesundheitsdienstleister in Verbindung.

Hinweis: Sollte der Companion 5 auch nach Befolgung der Angaben in den Tabellen zu Alarmzuständen undFehlerbehebung einen Alarm ausgeben, wenden Sie sich unter der Nummer 1-800-482-2473 und unter Angabe desauf dem LCD-Display angezeigten Alarmcodes an den technischen Dienst.

50 - GER PN 15062781-C4 J | BedienungsanleitungCompanion 5
Reinigung, Pflege und regelmäßige Wartung
Regelmäßige Wartung

4) Lassen Sie den Filter an der Luft trocknen.

Die Routinewartung umfasst das Auswechseln desFilters für große Partikel einmal pro Woche (falls
eingebaut).

5) Setzen Sie den Filter wieder in das Gehäuse ein.

Reinigung und Pflege des Geräts durch
den Benutzer

Hinweis: Betreiben Sie den Sauerstoffkonzentrator
nicht ohne eingesetzten Lufteinlassfilter.

KanülenaustauschTauschen Sie den Versorgungsschlauch sowie die
Kanüle regelmäßig gemäß den Empfehlungen Ihres
Pflegediensts aus. Informationen zu Reinigung,Desinfektion und Austausch erhalten Sie von Ihrem
Arzt oder Pflegedienst.
Hinweis: Für einen ordnungsgemäßen Gebrauchstets den Anleitungen des Kanülenherstellers folgen.
Die Einweg-Kanüle gemäß den Empfehlungendes Kanülenherstellers oder des Geräteanbieters
ersetzen. Weiteres Zubehör erhalten Sie von Ihrem
Geräteanbieter.

Lufteinlassfilter (Kartusche)
Nur Ihr Gesundheitsdienstleister sollte Arbeiten am
Lufteinlassfilter vornehmen oder ihn auswechseln.
Filter für große Partikel (Optional):

Der Bruttopartikelfilter befindet sich ganz vorne am
Companion 5.

Hinweis: Der Hersteller empfiehlt, dieses Gerät nicht
zu sterilisieren.

Befeuchterflasche (falls zutreffend)
Sie sollten die Befeuchterflasche täglich überprüfen,um sicherzustellen, dass sich der Wasserstand zwischen
dem angegebenen Höchst- und Mindestniveau
befindet. Füllen Sie bei Bedarf destilliertes Wasser
nach. Reinigen Sie die Befeuchterflasche entsprechend
den Anweisungen des Herstellers.
Reinigungsverfahren:
Schalten Sie den Companion 5 vor dem Reinigen oder
Desinfizieren ab und trennen Sie das Gerät von derStromversorgung. Sprühen Sie das Gehäuse NICHT
direkt ein. Verwenden Sie ein feuchtes (nicht nasses)
Tuch oder einen Schwamm. Sprühen Sie das Tuchbzw. den Schwamm mit mildem Reinigungsmittel ein
und reinigen Sie dann das Gehäuse und das Netzkabel.
Verwenden Sie zur Desinfektion des Companion 5
Lysol® Brand II oder ein ähnliches Desinfektionsmittel.
Verdünnen Sie es entsprechend den Anweisungen desHerstellers, sprühen Sie es aber nicht direkt auf den
Companion 5.
WARNUNG: STROMSCHLAGGEFAHR.
SCHALTEN SIE DAS GERÄT AUS
UND ZIEHEN SIE DAS NETZKABELAUS DER STECKDOSE, BEVOR DASGERÄT GEREINIGT ODER GEWARTETWIRD, UM DIE GEFAHR EINESSTROMSCHLAGS AUSZUSCHLIESSEN.
NUR DER GERÄTEANBIETER ODER EINQUALIFIZIERTER WARTUNGSTECHNIKER
DÜRFEN DIE ABDECKUNGEN ENTFERNENODER DAS GERÄT WARTEN.

Filter für
große Partikel

Sie sollten es einmal wöchentlich überprüfen undreinigen. Gehen Sie wie folgt vor, um den Filter zu
reinigen:1) Entfernen Sie den Filter für große Partikel.
2) Waschen Sie den Filter unter warmen
Leitungswasser mit einem milden Spülmittel ab.

WARNUNG: STROMSCHLAGGEFAHR. SCHALTENSIE DAS GERÄT AUS UND ZIEHEN SIE DASNETZKABEL AUS DER STECKDOSE, BEVOR DASGERÄT GEREINIGT ODER GEWARTET WIRD, UM DIE
GEFAHR EINES STROMSCHLAGS AUSZUSCHLIESSEN.
NUR DER GERÄTEANBIETER ODER EINQUALIFIZIERTER WARTUNGSTECHNIKER DÜRFEN
DIE ABDECKUNGEN ENTFERNEN ODER DAS GERÄTWARTEN.

3) Spülen Sie den Filter gründlich mit klaren
Leitungswasser aus und pressen Sie dasüberschüssige Wasser heraus.
PN 15062781-C4 J | Bedienungsanleitung

51 - GERCompanion 5
WARNUNG: STROMSCHLAGGEFAHR. ZIEHEN SIE
DAS NETZKABEL AUS DER STECKDOSE, BEVOR DASGERÄT GEREINIGT WIRD, UM DIE GEFAHR EINESSTROMSCHLAGS ODER EINER VERBRENNUNGAUSZUSCHLIESSEN. NUR DER GERÄTEANBIETERODER EIN QUALIFIZIERTER WARTUNGSTECHNIKER
DÜRFEN DIE ABDECKUNGEN ENTFERNEN ODERDAS GERÄT WARTEN.
WARNUNG: ES MUSS DARAUF GEACHTET WERDEN,DASS DER SAUERSTOFFKONZENTRATOR STETSTROCKEN BLEIBT UND KEINERLEI FLÜSSIGKEIT INDAS GERÄT EINTRITT. ANDERNFALLS KANN ES ZU
EINER FEHLFUNKTION ODER ZUR ABSCHALTUNG
KOMMEN ODER ES BESTEHT EIN ERHÖHTESRISIKO EINES ELEKTRISCHEN SCHLAGS ODER VONVERBRENNUNGEN.
WARNUNG: ES KANN UNSICHER SEIN, DEN
COMPANION 5 MIT ANDEREN GERÄTEN ZU
VERBINDEN.

WARNUNG: REINIGEN SIE GEHÄUSE, BEDIENFELDUND STROMKABEL NUR MIT EINEM SANFTEN
HAUSHALTSREINIGER, DER MIT EINEM
FEUCHTEN (NICHT NASSEN) TUCH ODERSCHWAMM AUFGETRAGEN WIRD. WISCHENSIE ANSCHLIESSEND ALLE OBERFLÄCHEN
TROCKEN. KEINE FLÜSSIGKEITEN IN DASINNERE DES GERÄTES GELANGEN LASSEN.
ACHTEN SIE BESONDERS DARAUF, DASS DER
SAUERSTOFFAUSLASS FÜR DEN ANSCHLUSS DERKANÜLE FREI VON STAUB, WASSER UND ANDEREN
PARTIKELN BLEIBT.
WARNUNG: KEINE FLÜSSIGKEITEN DIREKTAUF DEM GERÄT VERWENDEN. ZU DEN NICHTVERTRÄGLICHEN CHEMISCHEN MITTELN
ZÄHLEN UNTER ANDEREM: ALKOHOL UND
PRODUKTE AUF ALKOHOLBASIS, KONZENTRIERTEPRODUKTE AUF CHLORBASIS (ETHYLCHLORID)
UND PRODUKTE AUF ÖLBASIS (PINE-SOL®,LESTOIL®). DIESE DÜRFEN NICHT VERWENDETWERDEN, UM DAS KUNSTSTOFFGEHÄUSE DESSAUERSTOFFKONZENTRATORS ZU REINIGEN, DA SIE
DEN KUNSTSTOFF BESCHÄDIGEN KÖNNTEN.

Entsorgung
Geben Sie den Companion 5 einschließlich aller
Komponenten stets zur Entsorgung an Ihren
medizinischen Leistungserbringer zurück. FürAnweisungen zur ordnungsgemäßen Entsorgung des
Akkus können Sie sich zudem an die zuständigen
Stellen Ihres Wohnortes wenden.

WEEE und RoHS

Unsere Produkte entsprechen der Richtlinie
zur Beschränkung der Verwendung bestimmter
gefährlicher Stoffe (RoHS). Sie enthalten Blei oderandere gefährliche Materialien höchstens in Spuren.
VORSICHT: Wenden Sie sich zur ordnungsgemäßen Entsorgung an Ihren
Ausrüstungsanbieter oder die örtliche
Behörde, um Anweisungen zur Entsorgung zu erhalten.

Das Symbol soll den Besitzer des Geräts
darauf hinweisen, dass das Gerät gemäß der Richtlinieüber Elektro- und Elektronik-Altgeräte am Ende seiner
Lebensdauer zu einer Recyclingstelle gebracht werden
muss.

CAIRE und CAIRE Inc. sind eingetragene Marken von CAIRE Inc.
Eine vollständige Liste der Marken finden Sie auf unserer Website.
Warenzeichen: www.careinc.com/corporate/trademarks.
Copyright © 2020 CAIRE Inc. CAIRE Inc. behält sich das
Recht vor, die Vermarktung seiner Produkte einzustellen bzw.
Preise, Werkstoffe, Ausrüstungsteile, Qualität, Beschreibungen,Spezifikationen und/oder Prozesse ohne Vorankündigung zu einem
beliebigen Zeitpunkt zu ändern, ohne dass hieraus irgendwelche
Verpflichtungen oder Rechtsfolgen entstehen. Alle hier nicht
ausdrücklich genannten Rechte bleiben im gesetzlich zulässigen
Rahmen CAIRE Inc. vorbehalten.
52 - GER PN 15062781-C4 J | BedienungsanleitungCompanion 5
Controles del usuario e indicadores de estado del sistema
Parte aplicada tipo BF (grado deprotección frente a descargas eléctricas).
Reg. n.º 5333

Símbolos internos

02

Salida de oxígeno

I/OInterruptor de Encendido/Apagado
No contiene piezas reparables en elinterior. No abra la cubierta.

Directiva del Consejo 93/42/CEE sobre
dispositivos médicos
Representante autorizado en la
Comunidad Europea
IEC 60417
Equipo de Clase II, Doble aislamiento,
Reg n.º. 5172

Norma ISO 7000: símbolos gráficos que deben
usarse en los equipos (índice y sinopsis)
Rango de temperatura de almacenamiento
u operación. Reg. n.º 0632

Certificación ETL
Agencia de seguridad para la norma
CAN/CSA C22.2 n.º 601.1 M90 paraequipos eléctricos médicos.

Rango de humedad de almacenamiento.
Reg. n.º 2620
Nombre y domicilio del fabricante.
Reg. n.º 3082
Fecha de fabricación. Reg. n.º 2497
Número de catálogo. Reg. n.º 2493
Número de serie. Reg. n.º 2498
Este lado hacia arriba. Reg. n.º 0623

Aviso de la FCC
21 CFR 801.15: Código de Normas Federales,
Título 21
La legislación federal limita la venta deeste dispositivo a médicos o por orden
médica.
Directiva del Consejo 2012/19/UE: Directiva
sobre residuos de aparatos eléctricos yelectrónicos (RAEE)

Frágil. Manipular con cuidado.
Reg. n.º 0621
Mantenerlo alejado de la lluvia.
Mantenerlo seco. Reg. n.º 0626
Lea el manual del usuario antes de usar elequipo. Reg. n.º 1641
Límite de apilamiento por número.
Reg. n.º 2403
ISO 7010: Símbolos gráficos: colores y símbolosde seguridad; símbolos de seguridad registrados
Debe leer el manual de instrucciones. Reg.
n.º M002
Advertencia. Reg. n.º W001
Mantener alejado de las llamas, del fuego
y de las chispas. No acercar a fuentes deignición abiertas. Se prohíbe fumar. Reg.
n.º P003
No fumar cerca de la unidad ni mientras laesté utilizando. Reg. n.º P002

RAEE
IEC 60601-1: Equipos médicos eléctricos, Parte 1:
requisitos generales para la seguridad básica y eldesempeño esencial.

IPX1

Equipo IPX1 a prueba de goteo: elCompanion 5 proporciona protección
contra los efectos dañinos a causadel ingreso de líquidos. (IPX1,
según CEI 60529)

AVISO DE LA FCC:

Este dispositivo puede contener CYBLE-022001-00,
incluida la antena 2450AT18B100 de Johnson
Technology, cumple con la parte 15 de las reglas dela FCC. El dispositivo cumple los requisitos parala aprobación del transmisor modular tal y como se
detalla en el aviso público de la FCC DA00-1407. El
funcionamiento del transmisor está sujeto a las dos
condiciones siguientes: (1) Este dispositivo no debe
causar interferencias dañinas y (2) Este dispositivo
debe aceptar cualquier interferencia recibida, incluidas
las interferencias que puedan causar un funcionamiento
no deseado.
Este producto puede estar cubierto por una o máspatentes, de EE. UU. o internacionales. Visite nuestro
sitio web para obtener el listado de patentes aplicables.
Pat.: patents.cairemedical.com.
PN 15062781-C4 J | Manual del usuario

53 - SPACompanion 5
Guía de inicio rápido
1 Desembale su Companion 5

Companion 5

2 Conozca su Companion 5
Revise todas las advertencias, las precauciones y la
información adicional del dispositivo en el resto de este
manual. Familiarícese con las nuevas funciones clave
del Companion 5 y con el panel de control del usuario.

PARTE DELANTERA
Carcasa
exterior

Interruptor
deencendido

Pantalla
LED

OxígenoPuertode salidaContadorhorario
Medidorde flujo
Pie desoporte parala botella dehumidificación(botella
opcional)

PARTE TRASERAAsa detransporte
y entradade aireEntradade aire frío

Entradade aire54 - SPAPN 15062781-C4 J | Manual del usuario

Aberturas
de escapeCompanion 5
Los componentes del Companion 5
Carcasa
exterior

Carcasa de plástico durable que cubre
y protege los componentes internos del
CAIRE Companion 5.
Interruptor de Interruptor de ENCENDIDO/APAGADO
encendidoque se usa para arrancar o detenerla fuente de alimentación de launidad. La posición de ENCENDIDO
se indica por el símbolo "I" y la
posición de APAGADO se indica por elsímbolo "O".
Puerto deEs por donde sale el oxígeno delsalidaCompanion 5 después de que se lo
filtra y concentra. Se conecta una
cánula nasal de lumen simple oun tubo de oxígeno estándar para
entregar el oxígeno al paciente. La
longitud máxima del tubo de oxígeno
conectado al concentrador es de15,2 metros (50 pies).
Pantalla LED Contiene una luz LED verde, rojay amarilla*. La luz verde indica
operación normal y la luz amarilla* yla roja indican condiciones de alarma(consulte la sección de Condiciones
de alarma de las instrucciones paraobtener más información).
Contador
Muestra, de forma digital, el tiempo
horariotranscurrido de funcionamiento delconcentrador. Muestra el décimo
más cercano de una hora y no puede
reiniciarse. Muestra los códigos dealarma de diagnóstico cuando elconcentrador experimenta un estado
de alarma.
MedidorSe usa para ajustar y mostrar el caudal
de flujo
de flujo de oxígeno entregado. Los
caudales de flujo oscilan entre 0,5 y
5 litros por minuto (LPM) y pueden
ajustarse girando la perilla.
Pie de soporte Ubicación para conectar la botella dehumidificación. El pie contiene una
para labanda elástica que se usa para sujetar
botella dehumidificación la botella en el pie. La botella dehumidificación proporciona humedad(botella
adicional al flujo de oxígeno. La
opcional)
botella se llena con agua destilada,
y la humedad se usa para prevenir el
secado de los tejidos nasales.
*Solo modelos con Indicador de Estadodel Concentrador de Oxígeno (OCSI)

3 Sujeción de la botella de humidificacióny la cánula
Si usa una botella de humidificación, primero llene labotella con agua destilada según el nivel especificado
por el fabricante. Coloque la botella en el pie de soportecorrespondiente y sujétela con la banda elástica.
Conecte el extremo del tubo humidificador a la botella,
y el otro al puerto de salida del Companion 5. Conecte
una cánula nasal o un tubo de oxígeno al puerto desalida de la botella de humidificación.

Si no va a utilizar una botella de humidificación,
conecte directamente la cánula nasal o el tubo deoxígeno al puerto de salida del Companion 5.

PN 15062781-C4 J | Manual del usuario

55 - SPACompanion 5
4 Encendido y calentamiento
Enchufe el cable de alimentación de CA a una toma
de CA.

6 Adecuar el ajuste de flujo al nivel
indicado
Use la perilla de control de flujo para seleccionar el
flujo indicado por su médico.

Coloque el Interruptor de Encendido en la posición deEncendido.

Interruptor
de encendido

Cuando el Companion 5 se encienda correctamente, se
encenderá un indicador verde en la pantalla LED. Todaslas luces LED se iluminarán al iniciar el dispositivo.
Una vez que el concentrador complete el ciclo decalentamiento, solo la luz verde permanecerá encendida.
Nota: Después de encenderse inicialmente, la luz de O2 seiluminará y “O2 UP” se mostrará en la pantalla hasta quela concentración alcance las especificaciones.

ADVERTENCIA: ES MUY IMPORTANTESELECCIONAR SOLO EL NIVEL DEOXÍGENO PRESCRITO. NO CAMBIE LA
SELECCIÓN DE FLUJO A MENOS QUE UN
PROFESIONAL MÉDICO ACREDITADOASÍ SE LO HAYA INDICADO. EL
CONCENTRADOR DE OXÍGENO PUEDE
USARSE CUANDO DUERME SI ASÍ SE LO
HA RECOMENDADO UN PROFESIONAL
CUALIFICADO.
Respire normalmente a través de la cánula nasal.
Para ajustar el caudal de flujo:
Gire en el sentido contrario a las agujas del reloj paraaumentar el flujo.
Gire en el sentido de las agujas del reloj para reducir
el flujo.
El punto medio de la esfera indica el caudal de flujo.
Consulte la imagen a continuación.

Indicador
LED deencendido

Correcto
56 - SPAPN 15062781-C4 J | Manual del usuario

Companion 5¡Importante!

Las instrucciones de seguridad se definen de lasiguiente manera:
ADVERTENCIA: INFORMACIÓN DE
SEGURIDAD IMPORTANTE SOBREPELIGROS QUE PUEDEN CAUSARLESIONES GRAVES.
PRECAUCIÓN: información importantesobre la prevención de daños en elsistema Companion 5.
Aclaración: información que requiere atención
especial.

Indicaciones de uso
Uso previsto

El sistema Companion 5 está destinado para laadministración de oxígeno adicional. El dispositivo
no está destinado para su uso como soporte vital ni
proporciona capacidad de supervisión del paciente.
ADVERTENCIA: ES MUY IMPORTANTESELECCIONAR SOLO EL NIVEL DEOXÍGENO PRESCRITO. NO CAMBIE LA
SELECCIÓN DE FLUJO A MENOS QUE UN
PROFESIONAL MÉDICO ACREDITADOASÍ SE LO HAYA INDICADO. EL
CONCENTRADOR DE OXÍGENO PORTÁTIL
PUEDE USARSE AL DORMIR SI SE LO
HA RECOMENDADO UN PROFESIONAL
CUALIFICADO.
ADVERTENCIA: PARA GARANTIZAR QUE ESTÁRECIBIENDO LA CANTIDAD TERAPÉUTICA DE
OXÍGENO CONFORME A SU ESTADO DE SALUD,
EL SISTEMA COMPANION 5 SOLO SE DEBE
USAR CUANDO SE HAYAN DETERMINADO OPRESCRITO INDIVIDUALMENTE UNO O MÁS
AJUSTES PARA SUS NIVELES ESPECÍFICOS DE
ACTIVIDAD. EL SISTEMA COMPANION 5 SE DEBE
USAR CON LA COMBINACIÓN ESPECÍFICA OLAS PIEZAS Y ACCESORIOS ADECUADOS PARALA ESPECIFICACIÓN DEL FABRICANTE DEL
CONCENTRADOR Y QUE SE USARON MIENTRAS SE
DETERMINABAN LOS AJUSTES.
ADVERTENCIA: LA UNIDAD NO DEBE USARSECOMO SOPORTE VITAL. LOS PACIENTESGERIÁTRICOS O PEDIÁTRICOS O QUE NO PUEDANCOMUNICAR MOLESTIAS PUEDEN NECESITARUNA SUPERVISIÓN ADICIONAL O UN SISTEMA
DE ALARMA DISTRIBUIDO QUE TRANSMITA LA
INFORMACIÓN ACERCA DE LAS MOLESTIAS O LA
URGENCIA MÉDICA A LA PERSONA RESPONSABLE
DE SU CUIDADO, A FIN DE EVITAR LOS DAÑOS.
LOS PACIENTES CON TRASTORNOS AUDITIVOS O
VISUALES PUEDEN NECESITAR ASISTENCIA PARACONTROLAR LAS ALARMAS.

ADVERTENCIA: LAS MUJERES EMBARAZADAS OLACTANTES NO DEBEN USAR LOS ACCESORIOSRECOMENDADOS EN ESTE MANUAL, YA QUEPUEDEN CONTENER FTALATOS.
ADVERTENCIA: LAS LEYES FEDERALES (DE LOSESTADOS UNIDOS) LIMITAN LA VENTA O ALQUILER
DE ESTE DISPOSITIVO POR PARTE DE UN MÉDICOU OTRO PROVEEDOR DE ATENCIÓN MÉDICA
ACREDITADO.

Contraindicaciones de uso
ADVERTENCIA: EN ALGUNASCIRCUNSTANCIAS, EL USO DE OXÍGENO
NO PRESCRITO PUEDE SER PELIGROSO.
ESTE DISPOSITIVO SOLO DEBERÁ
UTILIZARSE CUANDO LO PRESCRIBA UN
MÉDICO.
ADVERTENCIA: NO UTILIZAR EN PRESENCIA DE
ANESTÉSICOS INFLAMABLES.
ADVERTENCIA: AL IGUAL QUE CON CUALQUIEROTRO DISPOSITIVO ELÉCTRICO, EL USUARIO
PODRÍA EXPERIMENTAR PERÍODOS DE NO
FUNCIONAMIENTO DEBIDO A INTERRUPCIONESELÉCTRICAS O A LA NECESIDAD DE QUE EL
TÉCNICO CUALIFICADO REVISE EL SISTEMACOMPANION 5. EL SISTEMA COMPANION 5 NO ESADECUADO PARA PACIENTES QUE PUEDAN SUFRIRCONSECUENCIAS DAÑINAS PARA LA SALUD A
CAUSA DE DICHAS INTERRUPCIONES TEMPORALES.

Instrucciones de seguridad
ADVERTENCIA: NO SE PERMITEN
MODIFICACIONES A ESTE EQUIPO.
ADVERTENCIA: EL FABRICANTE RECOMIENDADISPONER DE UNA FUENTE ALTERNATIVA DEOXÍGENO COMPLEMENTARIO EN CASO DE
CORTE ELÉCTRICO, CONDICIÓN DE ALARMA O
FALLA MECÁNICA. CONSULTE A SU MÉDICO O AL
PROVEEDOR DEL EQUIPO PARA DETERMINAR EL
TIPO DE SISTEMA DE RESERVA QUE NECESITA.
ADVERTENCIA: ESTE DISPOSITIVO SUMINISTRAOXÍGENO DE ALTA CONCENTRACIÓN, LO QUELO HACE MUY INFLAMABLE. NO PERMITA QUE
SE FUME NI LA PRESENCIA DE LLAMAS VIVASA MENOS DE DOS (2) METROS DE (1) ESTEDISPOSITIVO O (2) ALGÚN ACCESORIO QUE
CONTENGA OXÍGENO. SI NO SE TIENE EN CUENTA
ESTA ADVERTENCIA, PUEDEN PRODUCIRSE UN
INCENDIO GRAVE, DAÑOS A LOS BIENES O HASTALESIONES FÍSICAS O LA MUERTE.

PN 15062781-C4 J | Manual del usuario

57 - SPACompanion 5
ADVERTENCIA: NO OPERE LA UNIDAD EN UN
ESPACIO RESTRINGIDO O CONFINADO (P. EJ., UNA
CAJA O BOLSO PEQUEÑO) DONDE LA VENTILACIÓN
PUEDE SER LIMITADA. ESTO PUEDE OCASIONARQUE EL CONCENTRADOR DE OXÍGENO SE
SOBRECALIENTE Y QUE SU RENDIMIENTO SE VEAPERJUDICADO.
ADVERTENCIA: EL CONCENTRADOR
DEBE UBICARSE DE FORMA TALQUE SE PUEDA EVITAR EL HUMO,LOS CONTAMINANTES O LOS GASES
TÓXICOS.
ADVERTENCIA: EL USO DE ACCESORIOS DE
ADMINISTRACIÓN DE OXÍGENO NO ESPECIFICADOSPARA ESTE CONCENTRADOR DE OXÍGENO PUEDEDISMINUIR SU RENDIMIENTO. LOS ACCESORIOSRECOMENDADOS SE MENCIONAN DENTRO DE
ESTE MANUAL.
ADVERTENCIA: SI EL CONCENTRADOR DE OXÍGENO
SE HA CAÍDO, HA SUFRIDO DAÑOS O SE HAEXPUESTO AL AGUA, PÓNGASE EN CONTACTO CON
EL PROVEEDOR DE ATENCIÓN SANITARIA PARA SUINSPECCIÓN O POSIBLE REPARACIÓN. NO USE EL
CONCENTRADOR DE OXÍGENO SI TIENE EL CABLE
DE ALIMENTACIÓN O EL ENCHUFE DAÑADOS.
ADVERTENCIA: NO PERMITA QUE SE BLOQUEEN LAENTRADA DE AIRE NI LAS ABERTURAS DE SALIDA
DE AIRE. NO DEJE CAER NI INSERTE OBJETOS ENNINGUNA ABERTURA DEL DISPOSITIVO. ESTO
PUEDE OCASIONAR QUE EL CONCENTRADORDE OXÍGENO SE SOBRECALIENTE Y QUE SU
RENDIMIENTO SE VEA PERJUDICADO.
ADVERTENCIA: NO LLENE DEMASIADO
EL HUMIDIFICADOR OPCIONAL. LLENE EL
HUMIDIFICADOR OPCIONAL CON AGUA SOLO
HASTA EL NIVEL INDICADO POR EL FABRICANTE
DEL HUMIDIFICADOR.
ADVERTENCIA: NO UTILICE ESTE EQUIPO SINANTES LEER Y COMPRENDER ESTE MANUAL. SI NO
COMPRENDE LAS ADVERTENCIAS E INSTRUCCIONES,CONTACTE A SU PROVEEDOR DEL EQUIPO ANTESDE INTENTAR USARLO; DE LO CONTRARIO, PUEDENPRODUCIRSE LESIONES O DAÑOS.
ADVERTENCIA: SI SIENTE MALESTAR O TIENEUNA EMERGENCIA MÉDICA, BUSQUE ASISTENCIAMÉDICA DE INMEDIATO.
ADVERTENCIA: LA OPERACIÓN DELCONCENTRADOR DE OXÍGENO SIN TENEREN CUENTA LAS ESPECIFICACIONES DE LA
TEMPERATURA DE FUNCIONAMIENTO PUEDE
LIMITAR LA CAPACIDAD DEL CONCENTRADOR
DE CUMPLIR CON LAS ESPECIFICACIONES DE
CONCENTRACIÓN DE OXÍGENO. CONSULTE LASECCIÓN DE ESPECIFICACIONES DE ESTE MANUALPARA CONOCER LOS LÍMITES DE TEMPERATURA.

58 - SPAPN 15062781-C4 J | Manual del usuario

ADVERTENCIA: NO USE ACEITE, GRASA,PRODUCTOS A BASE DE PETRÓLEO NI OTROSPRODUCTOS INFLAMABLES CON LOS ACCESORIOSDE TRANSPORTE DE OXÍGENO O CON EL
CONCENTRADOR DE OXÍGENO. SOLO DEBEN USARSELOCIONES O BÁLSAMOS COMPATIBLES CON ELOXÍGENO, A BASE DE AGUA. EL OXÍGENO ACELERALA COMBUSTIÓN DE LAS SUSTANCIAS INFLAMABLES.
ADVERTENCIA: EL CONCENTRADOR DE OXÍGENO
NO DEBE USARSE ADYACENTE A O APILADO CONOTROS EQUIPOS. SI EL USO ADYACENTE O APILADOES INEVITABLE, EL DISPOSITIVO DEBE OBSERVARSEPARA VERIFICAR SU FUNCIONAMIENTO NORMAL.
ADVERTENCIA: EL FABRICANTERECOMIENDA DISPONER DE UNAFUENTE ALTERNATIVA DE OXÍGENOCOMPLEMENTARIO EN CASO DE CORTEELÉCTRICO, CONDICIÓN DE ALARMAO FALLA MECÁNICA. CONSULTE A SUMÉDICO O AL PROVEEDOR DEL EQUIPOPARA DETERMINAR EL TIPO DE SISTEMA
DE RESERVA QUE NECESITA.
ADVERTENCIA: ESTE DISPOSITIVO SUMINISTRAOXÍGENO DE ALTA CONCENTRACIÓN, LO QUE LO
HACE MUY INFLAMABLE. NO PERMITA QUE SE
FUME NI LA PRESENCIA DE LLAMAS EXPUESTASDENTRO DE LA MISMA HABITACIÓN EN LA QUESE ENCUENTRE (1) ESTE DISPOSITIVO O (2) ALGÚNACCESORIO QUE CONTENGA OXÍGENO. SI NO SE
TIENE EN CUENTA ESTA ADVERTENCIA, PUEDENPRODUCIRSE UN INCENDIO GRAVE, DAÑOS A LOSBIENES O HASTA LESIONES FÍSICAS O LA MUERTE.
ADVERTENCIA: EL USO DE ACCESORIOS DE
ADMINISTRACIÓN DE OXÍGENO NO ESPECIFICADOSPARA ESTE CONCENTRADOR DE OXÍGENO PUEDEDISMINUIR SU RENDIMIENTO. LOS ACCESORIOSRECOMENDADOS SE MENCIONAN DENTRO DE
ESTE MANUAL.
ADVERTENCIA: SI EL CONCENTRADOR DE OXÍGENO
SE HA CAÍDO, HA SUFRIDO DAÑOS O SE HAEXPUESTO AL AGUA, PÓNGASE EN CONTACTO CON
EL PROVEEDOR DE ATENCIÓN SANITARIA PARA SUINSPECCIÓN O POSIBLE REPARACIÓN. NO USE EL
CONCENTRADOR DE OXÍGENO SI TIENE EL CABLE
DE ALIMENTACIÓN O EL ENCHUFE DAÑADOS.
ADVERTENCIA: UTILICE ÚNICAMENTE PIEZAS DE
REPUESTO RECOMENDADAS POR EL FABRICANTEPARA GARANTIZAR EL FUNCIONAMIENTOCORRECTO Y EVITAR EL RIESGO DE INCENDIOS YQUEMADURAS.
ADVERTENCIA: NO LUBRIQUE LOS CONECTORES, LAS
CONEXIONES, LOS TUBOS NI OTROS ACCESORIOSDEL CONCENTRADOR DE OXÍGENO PARA EVITAR EL
RIESGO DE INCENDIOS Y QUEMADURAS.

Companion 5
ADVERTENCIA: FUMAR MIENTRAS SE USAOXÍGENO ES LA CAUSA PRINCIPAL DE LESIONESPOR INCENDIOS Y MUERTES RELACIONADAS. DEBESEGUIR LAS ADVERTENCIAS DE SEGURIDAD:
ADVERTENCIA: NO PERMITA QUE HAYACIGARRILLOS ENCENDIDOS, VELAS O LLAMASEXPUESTAS EN LA MISMA HABITACIÓN EN QUE SE
ENCUENTREN EL DISPOSITIVO O LOS ACCESORIOSQUE TRANSPORTAN OXÍGENO.
ADVERTENCIA: FUMAR MIENTRAS SE USA
UNA CÁNULA DE OXÍGENO PUEDE CAUSARQUEMADURAS FACIALES Y POSIBLEMENTE LAMUERTE.
ADVERTENCIA: SI FUMA, ESTOS 3 PASOS PUEDENSALVAR SU VIDA: APAGUE EL CONCENTRADORDE OXÍGENO, RETIRE LA CÁNULA Y SALGA DE
LA HABITACIÓN DONDE SE ENCUENTRE ELDISPOSITIVO.
ADVERTENCIA: DEBEN EXHIBIRSE LETREROS DE"NO FUMAR: OXÍGENO EN USO" EN EL HOGAR
O EN EL SITIO DONDE SE USE EL OXÍGENO. LOSPACIENTES Y LOS CUIDADORES SANITARIOS DEBEN
RECIBIR INFORMACIÓN ACERCA DE LOS PELIGROS
DE FUMAR AL USAR OXÍGENO MEDICINAL O ENPRESENCIA DE ESTE.
ADVERTENCIA: NO USE SU CONCENTRADOR DEOXÍGENO EN PRESENCIA DE GASES INFLAMABLES.
ESTO PUEDE GENERAR UN FUEGO RÁPIDO QUE
PUEDE PROVOCAR DAÑOS A LOS BIENES, LESIONESCORPORALES O LA MUERTE.
ADVERTENCIA: RETIRAR LA CÁNULA Y
COLOCARLA SOBRE LA ROPA, LA CAMA, EL SOFÁU OTRO MATERIAL ACOLCHADO CAUSARÁ UNA
LLAMARADA SI SE LA EXPONE A UN CIGARRILLO,UNA FUENTE DE CALOR O UNA LLAMA.
ADVERTENCIA: NO DEJE UNA CÁNULA NASAL
SOBRE O DEBAJO DE LA ROPA, LA CAMA O LOSALMOHADONES DE UNA SILLA. SI LA UNIDAD
ESTÁ ENCENDIDA PERO NO EN USO, EL OXÍGENO
PODRÍA VOLVER INFLAMABLES ESOS MATERIALES.
COLOQUE EL INTERRUPTOR DE ENCENDIDO
I/0 EN LA POSICIÓN 0 (APAGADO) CUANDO EL
CONCENTRADOR DE OXÍGENO NO ESTÉ EN USO.
ADVERTENCIA: NO INTENTE REALIZARTAREAS DE MANTENIMIENTO APARTE
DE LAS SOLUCIONES POSIBLESENUMERADAS EN ESTE MANUAL.
NO RETIRE LAS CUBIERTAS: SOLO SUPROVEEDOR DEL EQUIPO O UN TÉCNICODE SERVICIO CUALIFICADO DEBENREVISAR O RETIRAR LAS CUBIERTAS DE
ESTE DISPOSITIVO.

ADVERTENCIA: USE SOLO ACCESORIOSRECOMENDADOS POR EL FABRICANTE. EL USO
DE CUALQUIER OTRA FUENTE DE ALIMENTACIÓN
PUEDE SER PELIGROSO, PROVOCAR DAÑOSGRAVES EN EL CONCENTRADOR DE OXÍGENOY ANULARÁ LA GARANTÍA.
PRECAUCIÓN: Siempre desconecte lafuente de alimentación de CA de lapared antes de desconectar la fuentede alimentación de CA del concentradorde oxígeno.
PRECAUCIÓN: siempre coloque el tubo desuministro de oxígeno y los cables de alimentaciónde forma tal que evite el riesgo de tropiezos.
PRECAUCIÓN: para evitar la anulaciónde la garantía, siga todas las
instrucciones del fabricante.
Aclaración: los equipos de comunicaciones portátiles
y móviles de radiofrecuencia pueden afectar a losequipos eléctricos médicos.

ADVERTENCIA: SIEMPRE COLOQUE EL CABLE
DE SUMINISTRO DE OXÍGENO Y LOS CABLES DE
ALIMENTACIÓN DE UNA MANERA QUE PREVENGAEL PELIGRO DE VIAJE O POSIBLE ESTRANGULACIÓN
ACCIDENTAL.
ADVERTENCIA: ESTE PRODUCTO PUEDE
EXPONERLE A PRODUCTOS QUÍMICOSCOMO EL NÍQUEL, CONOCIDO EN EL ESTADO
DE CALIFORNIA POR CAUSAR CÁNCER.
PARA MÁS INFORMACIÓN, VISITE
WWW.P65WARNINGS.CA.GOV.
ADVERTENCIA: EN EL CASO DE QUE OCURRA UN
INCIDENTE GRAVE CON ESTE DISPOSITIVO, EL USUARIO DEBE INFORMAR INMEDIATAMENTE EL INCIDENTE AL PROVEEDOR Y / O AL FABRICANTE. UN
INCIDENTE GRAVE SE DEFINE COMO UNA LESIÓN,MUERTE O EL POTENCIAL DE CAUSAR LESIONES /MUERTE EN CASO DE QUE VUELVA A OCURRIR EL
INCIDENTE. EL USUARIO TAMBIÉN PUEDE INFORMAR EL INCIDENTE A LA AUTORIDAD COMPETENTEEN EL PAÍS DONDE OCURRIÓ EL INCIDENTE.

PN 15062781-C4 J | Manual del usuario

59 - SPACompanion 5
Especificaciones
Para un uso correcto del dispositivo, el cuadro
siguiente incluye información importante con respecto
a las condiciones de operación o los entornos deoperación recomendados.

Especificaciones
Velocidades deflujo
Concentraciónde O2
Dimensiones
Peso
Nivel depresión acústica
Consumode energía
Temperatura defuncionamiento
Humedad defuncionamiento
Temperatura dealmacenamiento
Humedad dealmacenamiento
Especificaciones
eléctricas

0,5-5,0 LPM
90 % (+5,5 %/-3 %)54,6 x 31,8 x 34,2 cm(21,5 x 12,5 x 13,5”)
16,3 kg (36 lb)
50 dB(A) a 2,0 LPM
285 W a 2 LPM, 350 W máximo
De 5 °C a 40 °C (41 °F a 104 °F)
15% - 90 % en un punto decondensación de 28 °C (82,4 °F)
De -25 °C a 70 °C (-13 °F a 158 °F)
0% - 90 % sin condensación

No utilice cables de extensión.
No utilice tomas eléctricas
controladas por un interruptor.
Funcionamiento De -382 a 3011 metros
en altitud(de -1253 a 9879 pies)
Tiempo de
Hasta 24 horas al día
funcionamiento
La vida útil esperada del dispositivo es de un mínimo
de cinco años.
ADVERTENCIA: SE ESPERA QUE
EL USO DEL DISPOSITIVO FUERA
DE CONDICIONES OPERATIVASESPECIFICADAS AFECTA ADVERSAMENTELA TASA DE FLUJO Y EL PORCENTAJE DEOXÍGENO Y CONSECUENTEMENTE LACALIDAD DE LA TERAPIA.
Aclaración: si el concentrador de oxígeno se ha
almacenado por un período prolongado fuera de su
intervalo de temperatura de funcionamiento normal,
tendrá que darle tiempo a la unidad a que vuelva asu temperatura de funcionamiento normal antes deencenderla (consulte la sección de Especificaciones en
este manual).

Colocación correcta del Companion 5
60 - SPAPN 15062781-C4 J | Manual del usuario

Seleccione una ubicación para el dispositivo en la queno haya entrada de humo, gases ni contaminantes. La
correcta colocación del dispositivo debe permitir laentrada del aire a través de las tres entradas de aire en laparte superior trasera y en la parte inferior, además dedejar salir el aire de escape por la abertura indicada paraello situada en la parte inferior izquierda del dispositivo.
Las entradas de aire se ubican en la parte trasera
superior e inferior del concentrador. Ubique la unidad
de forma tal que queden al menos 30 cm (12 in)de espacio entre el concentrador y las paredes, losmuebles, las cortinas u otras obstrucciones.
Coloque el dispositivo de modo que puedan oírse las
alarmas.
Coloque el tubo de suministro de oxígeno de forma tal
que no se curve ni ocluya.
ADVERTENCIA: NO USE SUCONCENTRADOR DE OXÍGENO EN
PRESENCIA DE GASES INFLAMABLES.
ESTO PUEDE GENERAR UN FUEGORÁPIDO QUE PUEDE PROVOCAR DAÑOS
A LOS BIENES, LESIONES CORPORALESO LA MUERTE. NO USE ACEITE, GRASA,PRODUCTOS A BASE DE PETRÓLEO NIOTROS PRODUCTOS INFLAMABLES CONLOS ACCESORIOS DE TRANSPORTE DEOXÍGENO O CON EL CONCENTRADORDE OXÍGENO. SOLO DEBEN USARSELOCIONES O BÁLSAMOS COMPATIBLESCON EL OXÍGENO, A BASE DE AGUA. EL
OXÍGENO ACELERA LA COMBUSTIÓN DE
LAS SUSTANCIAS INFLAMABLES.
ADVERTENCIA: ESTE DISPOSITIVO SUMINISTRAOXÍGENO DE ALTA CONCENTRACIÓN, LO QUE LO
HACE MUY INFLAMABLE. NO PERMITA QUE SE
FUME NI LA PRESENCIA DE LLAMAS EXPUESTASDENTRO DE LA MISMA HABITACIÓN EN LA QUESE ENCUENTRE (1) ESTE DISPOSITIVO O (2) ALGÚNACCESORIO QUE CONTENGA OXÍGENO. SI NO SE
TIENE EN CUENTA ESTA ADVERTENCIA, PUEDENPRODUCIRSE UN INCENDIO GRAVE, DAÑOS A LOSBIENES O HASTA LESIONES FÍSICAS O LA MUERTE.

Companion 5
Instrucciones de funcionamiento
Antes de utilizar

Este Manual del usuario sirve a modo de referencia
para ayudarle a poner en funcionamiento el dispositivo
y llevar a cabo el mantenimiento. Si tiene alguna
pregunta o duda, llame al proveedor de atención
domiciliaria.
¡Importante! NO intente poner en funcionamiento
el sistema Companion 5 sin leer primero la sección
Lineamientos de seguridad de este manual. Siga todas
las instrucciones de funcionamiento. Observe todas las
Advertencias del dispositivo y del Manual del usuario.
Para reducir el riesgo de incendios, lesiones a laspersonas y daños graves en el sistema Companion 5,
preste atención a todas las precauciones de seguridad.
El sistema CAIRE Companion 5 no genera su propio
oxígeno. Produce oxígeno altamente concentrado
proveniente del aire ambiente que lo rodea y lo entrega al
paciente. En su estado natural, el aire ambiente contiene
aproximadamente solo un 21 % de oxígeno por volumen.
El aire ambiente se introduce al concentrador por
medio de un compresor. Luego el aire viaja por una
serie de filtros y un lecho de tamiz molecular que
absorbe químicamente las moléculas de nitrógeno.
Como resultado, el aire enriquecido con oxígeno quesale del concentrador se entrega con un 87-95 % deoxígeno por volumen.
ADVERTENCIA: LIMPIE EL GABINETE,
EL PANEL DE CONTROL Y EL CABLE
DE ALIMENTACIÓN SOLO CON UN
LIMPIADOR DOMÉSTICO APLICADO CON
UN PAÑO HÚMEDO (NO MOJADO) O UNAESPONJA, Y LUEGO SEQUE TODAS LASSUPERFICIES. NO PERMITA QUE NINGÚN
LÍQUIDO INGRESE AL DISPOSITIVO.
PRESTE ESPECIAL ATENCIÓN A LA SALIDADE OXÍGENO PARA LA CONEXIÓN DE LACÁNULA PARA ASEGURARSE DE QUE SE
MANTENGA LIBRE DE POLVO, AGUA YPARTÍCULAS.
PRECAUCIÓN: no permita que se
bloqueen la entrada de aire ni lasaberturas de salida de aire. Esto
puede ocasionar que el concentradorde oxígeno se sobrecaliente y que su
rendimiento se vea perjudicado. No
opere el concentrador de oxígeno sin
el filtro de entrada de aire colocado. Si
dispone de un segundo filtro, coloqueeste filtro de "reemplazo" antes delimpiar el filtro sucio. Limpie el filtro
sucio con una solución tibia de agua
y jabón, y luego seque por completoantes de usar.

Paso 1: Ubicación del Companion 5 para usar
Coloque el sistema Companion 5 en una zona bien
iluminada y con buena ventilación. Asegúrese de quela entrada de aire y las aberturas de escape no estén
obstruidas. Coloque el sistema Companion 5 de modoque los indicadores visuales y sonoros puedan verse yoírse fácilmente.
Enchufe el dispositivo en la toma de corriente CA.
Paso 2: Si la botella de humidificación NO se usara

Entradade aireEntradade aire frío
Aberturas
de escapeEntradade airea. Conecte una cánula nasal o un tubo de oxígeno
directamente al puerto de salida.

b. Continúe con el paso 4.
Paso 3: Si la botella de humidificación se usara
a. Desatornille la tapa de la botella de humidificación.
b. Llene la botella usando agua destilada. Asegúrese
de que el nivel de agua se encuentre entre los niveles
máximo y mínimo especificados por el fabricante, loscuales se muestran en la botella.
c. Sujete la tapa de la botella de humidificación.
Verifique que no haya fugas.
d. Conecte una cánula nasal o un tubo de oxígeno
estándar al puerto de salida de la botella dehumidificación.
Aclaración: el uso de un humidificador no especificado
para utilizar con este concentrador podría afectar su
rendimiento.

PN 15062781-C4 J | Manual del usuario

61 - SPACompanion 5
g. Coloque la botella en su espacio correspondiente enel pie de soporte para la botella de humidificación.
h. Sujete la botella en el pie con la banda elástica.

e. Conecte un extremo del tubo humidificador a la tapade la botella de humidificación.

i. Continúe con el paso 4.
Paso 4: Encendido y calentamiento
a. Enchufe el cable de alimentación eléctrica delsistema CAIRE Companion 5 en una toma decorriente CA.
ADVERTENCIA: INSPECCIONE EL CABLE
ELÉCTRICO PARA DETECTAR POSIBLESDAÑOS ANTES DE USAR. SI EL CABLEESTÁ DAÑADO, NO LO ENCHUFE EN UNATOMA ELÉCTRICA NI INTENTE UTILIZAR
EL CONCENTRADOR.

f. Conecte el otro extremo del tubo humidificador al
puerto de salida del sistema CAIRE Companion 5.

b. Presione el interruptor de encendido hasta la
posición de "ENCENDIDO" (I). Cuando elCompanion 5 se encienda correctamente, se
encenderá un indicador verde en la pantalla LED.
Todas las luces LED se iluminarán al iniciar eldispositivo. Una vez que el concentrador complete
el ciclo de calentamiento, solo la luz verde
permanecerá encendida.
Aclaración: Tras el encendido inicial del dispositivo,
deje que pase un período de hasta diez (10) minutos
para que el dispositivo alcance sus especificaciones
de rendimiento.
62 - SPAPN 15062781-C4 J | Manual del usuario

Companion 5
ADVERTENCIA: ES MUY IMPORTANTESELECCIONAR SOLO EL NIVEL DEOXÍGENO PRESCRITO. NO CAMBIE LA
SELECCIÓN DE FLUJO A MENOS QUE UN
PROFESIONAL MÉDICO ACREDITADOASÍ SE LO HAYA INDICADO. EL
CONCENTRADOR DE OXÍGENO PUEDE
USARSE CUANDO DUERME SI ASÍ SE LO
HA RECOMENDADO UN PROFESIONAL
CUALIFICADO.
Paso 5: Ajuste el caudal de control de flujo

Paso 6: Verifique el flujo y respire normalmente

a. Gire la perilla de control al caudal de flujo deoxígeno (LPM) indicado por su médico.

a. Verifique que fluya oxígeno por la cánula nasal y que
no haya presiones, pliegues o bloqueos en el tubo.
Aclaración: asegúrese de que la cánula esté bien
insertada y quede firme. Debe escuchar o sentir el flujo
de oxígeno en las puntas de la cánula nasal.
b. Coloque de forma correcta la cánula nasal y respire
normalmente por la cánula nasal.
Paso 7: Apagado
a. Presione el interruptor de encendido hasta la
posición de "APAGADO" (O) cuando ya no use elsistema CAIRE Companion 5.

Para ajustar el caudal de flujo:
Gire en el sentido contrario a las agujas del reloj paraaumentar el flujo.
Gire en el sentido de las agujas del reloj para reducir
el flujo.
El punto medio de la esfera indica el caudal de flujo.
La imagen que aparece a continuación índica un caudal
de flujo de 4,0 LPM.

PRECAUCIÓN: use solo accesorios
recomendados por el fabricante. El uso
de cualquier otra fuente de alimentación
puede ser peligroso, provocar daños
graves en el concentrador de oxígeno yanulará la garantía.
PRECAUCIÓN: no utilice cables de extensión enesta unidad ni conecte demasiados enchufes
a la misma toma eléctrica. El uso de cables deextensión podría afectar de forma adversa elrendimiento del dispositivo. Demasiados enchufes
en una toma pueden ocasionar una sobrecargadel panel eléctrico y hacer que el disyuntor/fusible
se active o dispare si el disyuntor o el fusible no
funcionan.
PRECAUCIÓN: el uso de cables y adaptadores
diferentes a los especificados, con excepción delos cables y de los adaptadores vendidos por el
fabricante del equipo médico eléctrico como piezasde repuesto para componentes internos, puede
derivar en una mayor emisión de menor inmunidaddel concentrador de oxígeno.

Correcto

PN 15062781-C4 J | Manual del usuario

63 - SPACompanion 5
ADVERTENCIA: NO PERMITA QUE SEBLOQUEEN LA ENTRADA DE AIRE
NI LAS ABERTURAS DE SALIDA DE
AIRE. ESTO PUEDE OCASIONAR QUE
EL CONCENTRADOR DE OXÍGENO
SE SOBRECALIENTE Y QUE SU
RENDIMIENTO SE VEA PERJUDICADO.
Nota: Para el proveedor del equipo: Se recomiendan
los siguientes accesorios de administración de oxígeno
para usar con el Companion 5:• Cánula nasal: Número de parte CAIRE CU002-1• Tubo adaptador de humidificador: CAIRE Número depieza 20843882• Botella del humidificador: Número de pieza CAIRE
HU003-1• Cortafuegos: Número de parte CAIRE 20629671
Se requiere un cortafuegos para usar con cualquiercánula.
• CAIRE ofrece un cortafuegos destinado a serutilizado junto con el concentrador de oxígeno. El
cortafuegos es un fusible térmico para detener elflujo de gas en caso de que se encienda la cánula oel tubo de oxígeno aguas abajo y se queme hasta elcortafuegos. Se coloca en línea con la cánula nasal
o el tubo de oxígeno entre el paciente y la salida deoxígeno del Companion 5. Para un uso adecuado delcortafuegos, consulte siempre las instrucciones delfabricante (incluidas con cada kit de cortafuegos).
• Para cualquier accesorio adicional recomendado,
consulte el Catálogo de accesorios (PN MLLOX0010)
disponible en www.caireinc.com.

64 - SPAPN 15062781-C4 J | Manual del usuario

Companion 5
Estados de alarma
Su sistema CAIRE Companion 5 usa una combinaciónde dos o tres (para modelos con OCSI) luces LED
y una alarma sonora para alertarlo cuando existe una
falla en su concentrador.
1) LED verde (es normal.

) – indica que el funcionamiento2) LED rojo ( ) – indica que hay una falla en elsistema. Consulte la Tabla de Condiciones dealarma que sigue para obtener más información.

Verde

Roja

Amarilla3) LED amarillo ( ) – indica que hay una baja
concentración de oxígeno. Consulte la Tabla deCondiciones de alarma que sigue para obtener más
información.
La Tabla de Condiciones de alarma en la página
siguiente muestra las diferentes condiciones de alarmaque pueden mostrarse en el concentrador. Describe
las causas posibles y las acciones que debe llevar acabo cuando ocurre una alarma. En caso de alarma,
se recomienda contactar a su proveedor de atención
sanitaria lo antes posible.

Arriba: Companion 5

ADVERTENCIA: NO IGNORE LAS
ALARMAS
ADVERTENCIAS: EL FABRICANTERECOMIENDA DISPONER DE UNAFUENTE ALTERNATIVA DE OXÍGENOCOMPLEMENTARIO EN CASO DE CORTEELÉCTRICO, CONDICIÓN DE ALARMAO FALLA MECÁNICA. CONSULTE A SUMÉDICO O AL PROVEEDOR DEL EQUIPOPARA DETERMINAR EL TIPO DE SISTEMA
DE RESERVA QUE NECESITA.

PN 15062781-C4 J | Manual del usuario

65 - SPACompanion 5
Tabla de Condiciones de alarma
AlarmasonoraLED de
color

Causa

AcciónNinguna

El concentrador funciona
correctamente.

Apagado
Verde

Intermitente
ROJO

Intermitente

El concentrador norecibe energía pero elinterruptor de encendido
aún está en la posiciónde "ENCENDIDO".

Falla del sistema

Coloque el interruptor de encendido en la posición de"APAGADO". Desconecte el cable de alimentación CA
de la toma de la pared. Espere 5 minutos. Conecte elcable de corriente CA nuevamente a la toma de pared yvuelva a encender el concentrador. Si la alarma continúa,
se requiere servicio. Conecte a una fuente de oxígeno derespaldo y contacte a su proveedor de atención sanitaria
de inmediato.

Caudal de flujo delproducto demasiado altoo demasiado bajo

1. Asegúrese de que la cánula no esté retorcida ni
bloqueada. Si está utilizando el sistema con una botellade humidificación, asegúrese de que la haya llenado
correctamente sin crear una obstrucción.
2. Asegúrese de que el sistema Companion 5 cuente
con la ventilación adecuada. Debe estar situado a por
lo menos 30 cm (12 in) de cualquier superficie para
asegurarse de que las aberturas de escape no esténbloqueadas.
3. Si el problema persiste, cambie a otra fuente deoxígeno alternativa y póngase en contacto con el
proveedor de atención sanitaria para obtener ayuda.

ROJO

Intermitente
ROJO

Intermitente
Amarillo

Intermitente
Amarillo

66 - SPA

Verifique que el cable eléctrico del concentrador esté
enchufado en una toma y que la toma tenga corriente.
Pruebe con una toma diferente. Si el problema persiste,conecte a una fuente de oxígeno de respaldo y contacte
a su proveedor de atención sanitaria.

El sistema Companion 5
ha detectado niveles
bajos de oxígeno.

El sistema Companion 5
ha detectado niveles
bajos de oxígeno.

PN 15062781-C4 J | Manual del usuario

1. Asegure que el filtro de entrada y salida no
estén bloqueados ni obstruidos. 2. Asegure que elCompanion 5 se encuentre en una zona bien ventilada.
Asegúrese de que exista una distancia de 30 cm (12 in)entre la parte trasera del sistema Companion y cualquierobstáculo posible (muebles, cortinas, etc.).
3. Si la condición persiste, cambie a otra fuente deoxígeno alternativa y póngase en contacto con el
proveedor de atención sanitaria de inmediato.

Companion 5
Resolución de problemas
Se recomienda contactar a su proveedor de atención
sanitaria si su concentrador presenta una alarma o si
siente que su concentrador no funciona correctamente.
No debe intentar reparar el concentrador. La
reparación solo debe ser realizada por un técnicode servicio capacitado. Sin embargo, hay ciertos

pasos de resolución de problemas que puede seguiren caso de que tenga inconvenientes con su sistema
CAIRE Companion 5. Dichos pasos se describen enla tabla que sigue. Puede usar este cuadro siguiendolas verificaciones de su problema en orden numérico.

Tabla de Resolución de problemas del usuario
Problema

El concentrador
está encendido
pero nofunciona.

Experimenta
un flujo deoxígeno bajoo directamente
no hay flujo deoxígeno.

Causa posible

Acción

1. El cable eléctrico no está enchufado
a una toma.

1. Enchufe el cable eléctrico.

2. La toma eléctrica no suministra
corriente.

2. Compruebe el fusible y el circuito de su hogar. Pruebe
con una toma diferente.

3. Falla interna.

3. Conecte a una fuente de oxígeno de respaldo.
Contacte a su proveedor de atención sanitaria deinmediato.

1. Se ha cambiado el ajuste de control
de flujo.

1. Verifique que la perilla de control de flujo se encuentre
en el ajuste apropiado de Litro Por Minuto (LPM)
indicado por su médico.

2. El tubo se ha desconectado.

2. Verifique que el tubo esté ajustado en todas lasconexiones (puerto de salida, botella del humidificador,
trampa de agua, etc.). Vuelva a ajustar las conexiones deser necesario.

3. El tubo de oxígeno o la cánula
están retorcidos.

3. Inspeccione para detectar que el tubo no esté
retorcido ni bloqueado. Repare según sea necesario.

4. Hay una fuga en su cánula o tubo.

4. Inspeccione todos los tubos y cánulas para detectarfugas. Puede hacerlo al escuchar o sentir el aire que
escapa. Reemplace los tubos o las cánulas con fugas.

5. El flujo de aire hacia elconcentrador está bloqueado.

5. Verifique que no haya obstrucciones en las entradasdel concentrador. Asegúrese de que esté a por lo menos
30 cm (12 in) de distancia de muebles, paredes u otras
obstrucciones.

6. Falla interna.

6. Conecte a una fuente de oxígeno de respaldo.
Contacte a su proveedor de atención sanitaria deinmediato.

Aclaración: si su Companion 5 aún presenta una alarma después de usar las tablas de Condiciones de alarmay Resolución de problemas, contacte al Servicio Técnico al 1-800-482-2473 con el código de alarma que apareceen la pantalla LCD.

PN 15062781-C4 J | Manual del usuario

67 - SPACompanion 5
Limpieza, cuidados y mantenimiento periódico
Mantenimiento periódico

El mantenimiento periódico consiste en cambiar elfiltro de partículas gruesas una vez a la semana (si está
instalado).

Cuidados y limpieza del dispositivo porparte del usuario

5) Vuelva a introducir el filtro en la caja.
Aclaración: el fabricante no recomienda laesterilización de este equipo.
Aclaración: no opere el concentrador de oxígeno sin el
filtro de entrada de aire colocado.

Reemplazo de la cánula
Reemplace la cánula y el tubo de suministro con
regularidad, tal como lo recomienda el proveedor deatención sanitaria. El médico o proveedor de atención
sanitaria le facilitarán información sobre la limpieza, ladesinfección y el reemplazo de las piezas.

Botella del humidificador (cuando proceda)
Debe comprobar la botella del humidificador a diariopara asegurarse de que el nivel de agua se encuentre
entre los niveles máximo y mínimo especificados.
Rellene la botella según sea necesario usando agua
destilada. Limpie la botella del humidificador deacuerdo con las instrucciones del fabricante.

Aclaración: Siempre siga las instrucciones delfabricante de la cánula para un uso correcto.
Reemplace la cánula desechable de acuerdo con las
recomendaciones del fabricante de la cánula o delproveedor de su equipo. El proveedor de su equipo
dispone de accesorios de repuesto.

Procedimiento de limpieza:
APAGUE el sistema Companion 5 y desconéctelo dela alimentación de CA antes de emprender cualquiertipo de actividad de limpieza o desinfección. NO
pulverice la carcasa exterior directamente. Utilice un
paño húmedo (no mojado) o una esponja. Pulverice latela o la esponja con una solución de detergente suave
para limpiar el gabinete y el cable de alimentación.
Para desinfectar el sistema Companion 5, utilice eldesinfectante Lysol® Brand II (o uno equivalente).
Diluya tal como lo indica el fabricante del productode limpieza, pero no pulverice el líquido directamente
sobre el Companion 5.

Filtro de entrada de aire (Cartucho)
Solo el proveedor de atención sanitaria debe manipularo cambiar el filtro de entrada de aire.
Filtro de partículas gruesas (Opcional):

El filtro de partículas gruesas se encuentra en laparte inferior frontal del Companion 5.

Filtro de partículas gruesas:

Debe inspeccionarlo y limpiarlo una vez por semana.
Para limpiar el filtro, use el siguiente proceso:
1) Extraiga el filtro de partículas gruesas.
2) Lave el filtro con agua del grifo tibia y una solución
jabonosa suave.
3) Aclare el filtro perfectamente con agua del grifo
limpia y apriete hasta que salga toda el agua.
4) Deje que el filtro se seque al aire.
68 - SPAPN 15062781-C4 J | Manual del usuario

ADVERTENCIA: RIESGO DE
DESCARGA ELÉCTRICA. APAGUE LA
UNIDAD Y DESCONECTE EL CABLE
DE ALIMENTACIÓN DE LA TOMAELÉCTRICA ANTES DE LIMPIAR LA
UNIDAD PARA EVITAR UNA DESCARGAELÉCTRICA ACCIDENTAL O EL RIESGODE QUEMADURAS. ÚNICAMENTE ELPROVEEDOR DE SU EQUIPO O UNTÉCNICO DE SERVICIO CUALIFICADODEBEN QUITAR LAS TAPAS O REALIZARLA INSPECCIÓN DE LA UNIDAD.
ADVERTENCIA: RIESGO DE DESCARGA ELÉCTRICA.
APAGUE LA UNIDAD Y DESCONECTE EL CABLE
DE ALIMENTACIÓN DE LA TOMA ELÉCTRICAANTES DE LIMPIAR LA UNIDAD PARA EVITAR UNA
DESCARGA ELÉCTRICA ACCIDENTAL O EL RIESGODE QUEMADURAS. ÚNICAMENTE EL PROVEEDOR
DE SU EQUIPO O UN TÉCNICO DE SERVICIOCUALIFICADO DEBEN QUITAR LAS TAPAS OREALIZAR LA INSPECCIÓN DE LA UNIDAD.

Companion 5
ADVERTENCIA: RIESGO DE DESCARGA ELÉCTRICA.
DESCONECTE EL CABLE DE ALIMENTACIÓN DE
LA TOMA ELÉCTRICA ANTES DE LIMPIAR LA
UNIDAD PARA EVITAR UNA DESCARGA ELÉCTRICA
ACCIDENTAL O RIESGOS DE QUEMADURAS.
ÚNICAMENTE EL PROVEEDOR DE SU EQUIPOO UN TÉCNICO DE SERVICIO CUALIFICADO DEBENQUITAR LAS TAPAS O REALIZAR LA INSPECCIÓN
DE LA UNIDAD.
ADVERTENCIA: HAY QUE TENER CUIDADO DE
EVITAR QUE EL CONCENTRADOR DE OXÍGENO SE
MOJE Y NO HAY QUE DEJAR QUE NINGÚN LÍQUIDOINGRESE A LA UNIDAD. ESTO PUEDE CAUSAR
UNA AVERÍA O DESACTIVACIÓN Y PRODUCIRUN MAYOR RIESGO DE DESCARGA ELÉCTRICAO QUEMADURAS.
ADVERTENCIA: PUEDE SER INSEGURO
INTERCONECTAR EL COMPAÑERO 5 CON OTROSEQUIPOS.

ADVERTENCIA: LIMPIE EL GABINETE, EL PANEL DE
CONTROL Y EL CABLE DE ALIMENTACIÓN SOLOCON UN LIMPIADOR DOMÉSTICO APLICADOCON UN PAÑO HÚMEDO (NO MOJADO) O UNAESPONJA, Y LUEGO SEQUE TODAS LAS SUPERFICIES.
NO PERMITA QUE NINGÚN LÍQUIDO INGRESEAL DISPOSITIVO. PRESTE ESPECIAL ATENCIÓN
A LA SALIDA DE OXÍGENO PARA LA CONEXIÓN
DE LA CÁNULA PARA ASEGURARSE DE QUE SE
MANTENGA LIBRE DE POLVO, AGUA Y PARTÍCULAS.
ADVERTENCIA: NO USE LÍQUIDO EN CONTACTODIRECTO CON LA UNIDAD. A CONTINUACIÓN,SE INCLUYE UNA LISTA DE ALGUNOS AGENTESQUÍMICOS NO DESEABLES, ENTRE OTROS: ALCOHOLY PRODUCTOS A BASE DE ALCOHOL, PRODUCTOSA BASE DE CLORO CONCENTRADO (CLORURO
DE ETILENO) Y PRODUCTOS A BASE DE ACEITE(PINE-SOL®, LESTOIL®). ESTOS NO DEBEN USARSEPARA LIMPIAR LA CARCASA DE PLÁSTICO DELCONCENTRADOR DE OXÍGENO, YA QUE PUEDEN
DAÑAR EL PLÁSTICO DE LA UNIDAD.

Eliminación
Devuelva siempre a su médico el Companion 5
junto con todos sus componentes para su correcta
eliminación. También puede ponerse en contacto conlas oficinas pertinentes de su ciudad para obtener
instrucciones sobre la correcta eliminación de labatería.

RAEE y RoHS

Nuestros productos cumplirán con la Restricción sobre
Sustancias Peligrosas (RoHS). No contendrán más que
trazas de plomo o de otros materiales peligrosos.
PRECAUCIÓN: Para una eliminación adecuada, comuníquese con su proveedor
de equipos o agencia gubernamental
local para obtener instrucciones deeliminación.

Este símbolo pretende recordar a los propietarios de los equipos que deben llevarlos a un centro
de reciclado al final de su vida útil, de conformidad
con la Directiva sobre residuos de aparatos eléctricos yelectrónicos (RAEE).

CAIRE and CAIRE Inc. are registered trademarks of CAIRE Inc.
Please visit our website below for a full listing of trademarks.
Trademarks: www.caireinc.com/corporate/trademarks/.
Copyright © 2020 CAIRE Inc. CAIRE Inc. se reserva el derecho
de suspender sus productos o modificar los precios, los materiales,el equipo, la calidad, las descripciones, las especificaciones y/o losprocesos de sus productos en cualquier momento, sin previo aviso ysin ningún compromiso ni ninguna consecuencia. Queda reservado
cualquier otro derecho que no se encuentre indicado explícitamente eneste documento, según corresponda.
PN 15062781-C4 J | Manual del usuario

69 - SPACompanion 5
Controlli utente e indicatori di stato del sistema
Parte applicata di tipo BF (grado diprotezione da scossa elettrica).
N. reg. 5333

Simboli interni

02

Uscita ossigeno

I/O Interruttore On/Off
Non contiene componenti riparabili.
Non aprire la copertura.

Direttiva del Consiglio 93/42/CEE relativa ai
dispositivi medici
Rappresentante autorizzato per la
Comunità Europea
IEC 60417

ISO 7000; Simboli grafici per l’utilizzo
sull’apparecchiatura - Indice e sinossi
Intervallo di temperatura diconservazione o di esercizio .N. reg.
0632

Apparecchio di Classe II, doppio
isolamento N. reg. 5172
Certificazione ETL
Agenzia per la sicurezza per
CAN/CSA C22.2 N. 601.1 M90 per
apparecchiature elettromedicali

Intervallo di umidità di stoccaggio. N.
reg. 2620
Nome e indirizzo del produttore.
N. reg. 3082
Data di produzione. N. reg. 2497
Numero di catalogo. N. reg. 2493
Numero di serie. N. reg. 2498
Lato alto. N. reg. 0623
Fragile, maneggiare con cura.
N. reg. 0621
Tenere lontano dalla pioggia e conservare
in luogo asciutto. N. reg. 0626
Leggere il Manuale dell’utente primadell’uso. N. reg. 1641
Limite di sovrapposizione per numero.
N. reg 2403
ISO 7010: Simboli grafici – Colori di sicurezzae simboli di sicurezza – Simboli di sicurezza
registrati
Leggere il manuale di istruzioni.
N. reg. M002
Avvertenza. N. reg. W001
Conservare lontano da fiamme libere,
fuoco e scintille. È vietato fumare eutilizzare fonti di accensione aperte.
N. reg. P003
Non fumare nei pressi dell’unità odurante il suo utilizzo. N. reg. P002

70 - ITLPN 15062781-C4 J Manuale dell’utente

Avviso FCC
21 CFR 801.15: Code of Federal Regulations, Title
21 (Codice delle normative federali USA, Titolo 21)
La legge federale degli Stati Uniti
limita la vendita del presente
dispositivo ai soli medici o dietro
prescrizione medica.
Direttiva del Consiglio 2012/19/UE: rifiuti diapparecchiature elettriche ed elettroniche (RAEE)RAEEIEC 60601-1: Apparecchi elettromedicali Parte 1
Prescrizioni generali per la sicurezza di base e leprestazioni essenziali

IPX1

Apparecchiatura IPX1 a tenuta stagna:
Companion 5 offre protezione contro
gli effetti dannosi causati dall’ingresso
di liquidi. (IPX1, in conformità allo
standard IEC 60529)

AVVISO FCC:

Questo dispositivo può contenere CYBLE-022001-00,
ivi compresa l’antenna 2450AT18B100 di Johnson
Technology, ed è conforme alla parte 15 dellenorme FCC. Il dispositivo soddisfa i requisiti diapprovazione del trasmettitore modulare, comedettagliato nell’Avviso pubblico FCC DA00-1407.
Il funzionamento del trasmettitore è soggetto alleseguenti due condizioni: (1) questo dispositivo nonpuò provocare interferenza pericolosa e (2) questo
dispositivo deve accettare l’interferenza ricevuta,
compresa l’interferenza che può provocare il
funzionamento non desiderato.
Questo prodotto può essere coperto da uno o più
brevetti, validi negli Stati Uniti e all’estero. Visitare il
nostro sito Web, sotto indicato, per l’elenco dei brevetti
applicabili. Brev.: www.caireinc.com/corporate/
patents/.

Companion 5
Guida di avvio rapido
1 Rimuovere dalla confezione il
dispositivo Companion 5
Companion 5

2 Conoscere il dispositivo Companion 5
Controllare tutte le avvertenze, gli avvisi e leinformazioni aggiuntive sul dispositivo contenute nel
resto del manuale. Acquisire dimestichezza con lefunzionalità principali del dispositivo Companion 5 e
del pannello di controllo utente.

PARTE ANTERIORE
CustodiaesternaInterruttore dialimentazione

Display
LED

Porta diGiuntodello
sfogo

Contaore

Flussometro
Supporto peril flacone
umidificatore(flacone
opzionale)

PARTE POSTERIORE
Maniglia peril trasporto
e posizione
dell’ingresso
dell’aria

Posizionedell’ingresso
dell’aria diraffreddamento

Posizionedell’ingresso
dell’aria

PN 15062781-C4 J | Manuale dell’utente

Sfiato discarico71 - ITLCompanion 5
Componenti del Companion 5
CustodiaesternaInterruttore dialimentazione

Giunto dello
sfogo

Display LED

Contaore

Flussometro

Supporto
per il flacone
umidificatore(flacone
opzionale)

72 - ITL

Una resistente custodia esterna inplastica che racchiude e protegge icomponenti interni del Companion 5
di CAIRE.
Interruttore di alimentazione ON/OFF
utilizzato per fornire o interrompere
l’alimentazione dell’unità. La
posizione ON viene indicata dalsimbolo “l”, e la posizione OFF è
indicata dal simbolo “O”.
L’ossigeno esce dal dispositivo
Companion 5 dopo che è stato
filtrato e concentrato. Per la fornitura
al paziente, può essere collegata
una cannula nasale a lume singoloo è possibile impiegare dei tubi perossigeno standard. La lunghezza
massima dei tubi per ossigeno
collegati al concentratore è di circa
15,2 m (50 piedi).
Contiene una luce LED verde, rossa e
gialla*. La luce verde indica il normale
funzionamento, mentre le luci gialla*
e rossa indicano condizioni di allarme.
(consultare la sezione Condizioni diallarme delle istruzioni per maggiori
informazioni).
Visualizzazione digitale del tempo
di funzionamento del concentratore
trascorso. Il tempo viene visualizzato
approssimando per eccesso al decimo
di ora, e non può essere reimpostato.
Visualizza i codici delle condizioni
di allarme quando il concentratore
registra una condizione di allarme.
Utilizzato per regolare e mostrarela portata dell’ossigeno erogato.
La portata oscilla tra 0,5 e 5 litri al
minuto (LPM), e può essere regolata
ruotando la manopola.
Posizione dove collegare un flacone
umidificatore. Il supporto comprende
un elastico da utilizzare per fissare
il flacone al supporto. Un flacone
umidificatore offre umidità aggiuntiva
al flusso di ossigeno. Il flacone viene
riempito con acqua distillata, el’umidità viene utilizzata per evitareche i tessuti nasali si asciughino.
*Solo per i modelli del concentratore
di ossigeno con indicatore distato (OCSI)PN 15062781-C4 J Manuale dell’utente

3 Collegamento del flacone umidificatore
e della cannula
Se si utilizza un flacone umidificatore, prima ditutto riempire la bottiglia con acqua distillata fino allivello indicato dal produttore. Posizionare il flacone
nel supporto predisposto e fissarlo con l’elastico.
Collegare l’estremità filettata del tubo umidificatore al
flacone, e l’altra al giunto di sfogo del Companion 5.
Collegare i tubi per ossigeno e/o una cannula nasale algiunto di sfiato sul flacone umidificatore.

Se non si utilizza un flacone umidificatore: collegare iltubo per ossigeno e/o una cannula nasale direttamente
al giunto di sfogo del Companion 5.

Companion 5
4 Accensione e riscaldamento
Collegare il cavo di alimentazione CA a una presa CA.

6 Regolare l’impostazione di flusso al
livello prescritto
Utilizzare la manopola di controllo del flusso perselezionare il flusso prescritto dal medico.

Spostare l’interruttore di alimentazione in posizione On.

Interruttore dialimentazione

Quando il dispositivo Companion 5 viene acceso
correttamente, si accenderà l’indicatore verde sul
display LED. Tutte le luci LED si illumineranno
all’avvio. Dopo che il concentratore avrà completato il
ciclo di riscaldamento, rimarrà accesa soltanto la luce
verde.
Nota: Dopo l’accensione iniziale, la luce O2 si illumina
e “O2 UP” viene visualizzato sullo schermo fino aquando la concentrazione raggiunge le specifiche.

AVVERTENZA: È MOLTO IMPORTANTESELEZIONARE SOLO IL LIVELLODI OSSIGENO PRESCRITTO. NON
MODIFICARE LA SELEZIONE DEL
FLUSSO SE NON IN CASO DI PRECISEISTRUZIONI DA PARTE DI UN MEDICOAUTORIZZATO. È POSSIBILE UTILIZZAREIL CONCENTRATORE DI OSSIGENO
DURANTE IL SONNO SOLO SURACCOMANDAZIONE DI UN MEDICOQUALIFICATO.
Respirare normalmente attraverso la cannula nasale.
Per regolare la portata:
Ruotare in senso antiorario per aumentare il flusso.
Ruotare in senso orario per diminuire il flusso.
La parte centrale della sfera indica la portata. Vedere
l’immagine di seguito.

LEDindicatore dialimentazione

Corretto

PN 15062781-C4 J | Manuale dell’utente

73 - ITLCompanion 5
Importante!

Le istruzioni per la sicurezza sono definite comeindicato di seguito:
AVVERTENZA: IMPORTANTIINFORMAZIONI PER LA SICUREZZARELATIVE AI PERICOLI CHE POSSONOCAUSARE GRAVI LESIONI.
ATTENZIONE: informazioni importanti
per evitare danni al dispositivo
Companion 5.
Nota: informazioni che richiedono particolare
attenzione.

Indicazioni per l’uso
Destinazione d’uso

Il dispositivo Companion 5 è progettato per lasomministrazione di ossigeno supplementare. Il
dispositivo non è progettato per il supporto vitale
artificiale, e non offre funzionalità di monitoraggio delpaziente.
AVVERTENZA: È MOLTO IMPORTANTESELEZIONARE SOLO IL LIVELLODI OSSIGENO PRESCRITTO. NON
MODIFICARE LA SELEZIONE DEL
FLUSSO SE NON IN CASO DI PRECISEISTRUZIONI DA PARTE DI UN MEDICOAUTORIZZATO. È POSSIBILE UTILIZZAREIL CONCENTRATORE PORTATILE DIOSSIGENO DURANTE IL SONNO SURACCOMANDAZIONE DI UN MEDICOQUALIFICATO.
AVVERTENZA: PER ASSICURARE LA RICEZIONE
DELL’APPORTO TERAPEUTICO DI OSSIGENO ADATTOALLA PROPRIA PATOLOGIA, COMPANION 5 DOVRÀ
ESSERE UTILIZZATO SOLTANTO IN SEGUITO ALLADETERMINAZIONE O ALLA PRESCRIZIONE DI UNA O
PIÙ IMPOSTAZIONI A LIVELLI DI ATTIVITÀ SPECIFICIPER L’UTILIZZATORE. COMPANION 5 DEVE ESSEREUTILIZZATO CON LA COMBINAZIONE SPECIFICA DI
PARTI E ACCESSORI CONFORMI ALLE SPECIFICHE
DEL PRODUTTORE DEL CONCENTRATORE E CHE
SIANO STATI UTILIZZATI AL MOMENTO DELLA
DETERMINAZIONE DELLE IMPOSTAZIONI.
AVVERTENZA: QUESTA UNITÀ NON DEVE ESSEREUTILIZZATA COME SUPPORTO SALVAVITA. I PAZIENTIGERIATRICI, PEDIATRICI O ALTRI PAZIENTI INCAPACIDI COMUNICARE I PROPRI DISTURBI POTREBBERO
NECESSITARE DI UN MONITORAGGIO AGGIUNTIVO
E/O DI UN SISTEMA DI ALLARME DISTRIBUITOCHE INVII LE INFORMAZIONI SUL DISTURBO
E/O SULL’EMERGENZA MEDICA ALL’ASSISTENTESANITARIO RESPONSABILE PER EVITAREINFORTUNI. I PAZIENTI CON DANNI ALL’UDITO
E/O ALLA VISTA POTREBBERO NECESSITARE DIASSISTENZA PER MONITORARE GLI ALLARMI.
74 - ITLPN 15062781-C4 J Manuale dell’utente

AVVERTENZA: LE DONNE INCINTE O CHE
ALLATTANO NON DEVONO USARE GLI ACCESSORICONSIGLIATI IN QUESTO MANUALE, POICHÉ QUESTIPOSSONO CONTENERE FTALATI.
AVVERTENZA: LA LEGGE FEDERALE (USA) LIMITA LAVENDITA O IL NOLEGGIO DI QUESTO DISPOSITIVODIETRO PRESCRIZIONE DI UN MEDICO O DI UN
ALTRO PROFESSIONISTA SANITARIO QUALIFICATO.

Controindicazioni all’uso
AVVERTENZA: IN ALCUNE CIRCOSTANZE,L’USO DI OSSIGENO NON PRESCRITTO
PUÒ ESSERE PERICOLOSO. QUESTODISPOSITIVO VA UTILIZZATO SOLODIETRO PRESCRIZIONE MEDICA.
AVVERTENZA: NON UTILIZZARE IN PRESENZA DI
ANESTETICI INFIAMMABILI.
AVVERTENZA: ANALOGAMENTE A QUALSIASIALTRO DISPOSITIVO ELETTRICO, POTREBBERO
VERIFICARSI PERIODI DI INATTIVITÀ DOVUTIALL’INTERRUZIONE DELL’ALIMENTAZIONEELETTRICA O ALLA NECESSITÀ DI FAR RIPARAREIL DISPOSITIVO COMPANION 5 DA UN TECNICOQUALIFICATO. IL DISPOSITIVO COMPANION 5 NON
È INDICATO SU PAZIENTI CHE POTREBBERO SUBIRECONSEGUENZE NEGATIVE SULLA PROPRIA SALUTE
A CAUSA DI TALE INTERRUZIONE TEMPORANEA.

Linee guida per la sicurezza
AVVERTENZA: NON È CONSENTITOAPPORTARE ALCUNA MODIFICA ALDISPOSITIVO
AVVERTENZA: IL PRODUTTORE CONSIGLIA DI
TENERE A DISPOSIZIONE UN’ALTRA SORGENTE
DI OSSIGENO SUPPLEMENTARE IN CASO DI
INTERRUZIONI DI CORRENTE, CONDIZIONI DI
ALLARME O GUASTI MECCANICI. PER CONOSCEREIL TIPO DI SISTEMA DI RISERVA NECESSARIO,RIVOLGERSI AL PROPRIO MEDICO O AL FORNITORE
DELL’APPARECCHIATURA.
AVVERTENZA: QUESTO DISPOSITIVO EROGAOSSIGENO A ELEVATA CONCENTRAZIONE CHE
FAVORISCE IL RAPIDO SVILUPPO DI USTIONI.
NON CONSENTIRE AL PAZIENTE DI FUMARE O
DI USARE FIAMME LIBERE A UNA DISTANZAINFERIORE A DUE (2) METRI DA (1) QUESTODISPOSITIVO O (2) QUALSIASI ALTRO ACCESSORIO
PER L’EROGAZIONE DELL’OSSIGENO. IL MANCATORISPETTO DI QUESTA AVVERTENZA PUÒ
COMPORTARE GRAVI INCENDI E DANNI MATERIALIE/O PROVOCARE LESIONI FISICHE O IL DECESSO.

Companion 5
AVVERTENZA: NON UTILIZZARE L’UNITÀ IN SPAZICONFINATI O ANGUSTI (AD ESEMPIO UNA BORSAO UNA CUSTODIA DI PICCOLE DIMENSIONI) IN
CUI LA VENTILAZIONE POSSA ESSERE LIMITATA. IN
QUESTO MODO IL CONCENTRATORE DI OSSIGENO
POTREBBE SURRISCALDARSI PREGIUDICANDO LE
PRESTAZIONI.
AVVERTENZA: IL CONCENTRATORE VA
POSIZIONATO IN MODO DA EVITARE
FUMO, AGENTI INQUINANTI O VAPORI.
AVVERTENZA: L’USO DI ALCUNI ACCESSORI PER
LA SOMMINISTRAZIONE DI OSSIGENO NONESPRESSAMENTE INDICATI PER L’USO CONQUESTO CONCENTRATORE DI OSSIGENO PUÒ
PREGIUDICARNE LE PRESTAZIONI. GLI ACCESSORIRACCOMANDATI SONO RIPORTATI ALL’INTERNO
DEL PRESENTE MANUALE.

AVVERTENZA: NON UTILIZZARE PRODOTTI A BASE
DI OLIO, GRASSO O PETROLIO O ALTRI PRODOTTIINFIAMMABILI CON GLI ACCESSORI PER IL
TRASPORTO DELL’OSSIGENO O IL CONCENTRATOREDI OSSIGENO. UTILIZZARE ESCLUSIVAMENTELOZIONI O UNGUENTI A BASE D’ACQUACOMPATIBILI CON L’OSSIGENO. L’OSSIGENOACCELERA LA COMBUSTIONE DELLE SOSTANZEINFIAMMABILI.
AVVERTENZA: NON UTILIZZARE IL CONCENTRATOREDI OSSIGENO ACCANTO O IMPILATO SU ALTRE
APPARECCHIATURE. SE NON È POSSIBILE EVITARETALE MODALITÀ DI UTILIZZO, OSSERVARE ILDISPOSITIVO PER VERIFICARNE IL NORMALEFUNZIONAMENTO.

AVVERTENZA: SE IL CONCENTRATORE DIOSSIGENO CADE, VIENE DANNEGGIATO O ESPOSTO
ALL’ACQUA, RIVOLGERSI AL FORNITORE DIASSISTENZA DOMICILIARE PER UN’ISPEZIONE O
UN’EVENTUALE RIPARAZIONE DEL DISPOSITIVO.
NON UTILIZZARE IL CONCENTRATORE DI OSSIGENOSE IL CAVO O LA SPINA DI ALIMENTAZIONE SONODANNEGGIATI.
AVVERTENZA: NON FAR BLOCCARE L’INGRESSO O
GLI SFIATI DI USCITA DELL’ARIA. NON FAR CADEREO INSERIRE ALCUN OGGETTO NELLE APERTURE
PRESENTI SUL DISPOSITIVO. IN QUESTO MODO
IL CONCENTRATORE DI OSSIGENO POTREBBESURRISCALDARSI PREGIUDICANDO LE PRESTAZIONI.
AVVERTENZA: NON RIEMPIRE ECCESSIVAMENTE
L’UMIDIFICATORE OPZIONALE. RIEMPIRE
L’UMIDIFICATORE OPZIONALE CON ACQUAESCLUSIVAMENTE FINO AL LIVELLO INDICATO DALPRODUTTORE.
AVVERTENZA: NON UTILIZZARE QUESTODISPOSITIVO SENZA PRIMA AVER LETTO ECOMPRESO IL PRESENTE MANUALE. IN CASO DI
DIFFICOLTÀ A COMPRENDERE LE AVVERTENZE
E LE ISTRUZIONI, RIVOLGERSI AL FORNITORE
DELL’APPARECCHIATURA PRIMA DI TENTARE DIUTILIZZARLA; IN CASO CONTRARIO, POSSONOVERIFICARSI DANNI O LESIONI.
AVVERTENZA: SE SI MANIFESTANO DISTURBI O
IN CASO DI EMERGENZA MEDICA, RIVOLGERSIIMMEDIATAMENTE A UN MEDICO.
AVVERTENZA: L’UTILIZZO DEL CONCENTRATOREDI OSSIGENO AL DI FUORI DELLE SPECIFICHE DITEMPERATURA DI ESERCIZIO PUÒ LIMITARE LACAPACITÀ DEL CONCENTRATORE DI SODDISFAREI VALORI SPECIFICATI PER LA CONCENTRAZIONE
DELL’OSSIGENO. PER I LIMITI DI TEMPERATURA,
FARE RIFERIMENTO ALLA SEZIONE DELLE
SPECIFICHE RIPORTATA ALL’INTERNO DEL PRESENTEMANUALE.

PN 15062781-C4 J | Manuale dell’utente

75 - ITLCompanion 5
AVVERTENZA: UTILIZZARE SOLTANTO PARTI DI
RICAMBIO CONSIGLIATE DAL PRODUTTORE PERASSICURARE UN CORRETTO FUNZIONAMENTOE PER EVITARE IL RISCHIO DI INCENDIO ESCOTTATURE.
AVVERTENZA: NON LUBRIFICARE RACCORDI,
COLLEGAMENTI, TUBATURE O ALTRI ACCESSORIDEL CONCENTRATORE DI OSSIGENO PER EVITARE IL
RISCHIO DI INCENDIO E SCOTTATURE.
AVVERTENZA: FUMARE DURANTE L’USO
DELL’OSSIGENO COSTITUISCE LA CAUSA NUMERO
UNO DI USTIONI E MORTI CORRELATE AGLIINCENDI. È NECESSARIO ATTENERSI ALLE SEGUENTIAVVERTENZE DI SICUREZZA:
AVVERTENZA: NON CONSENTIRE AL PAZIENTEDI FUMARE O DI USARE CANDELE O FIAMME
LIBERE ALL’INTERNO DELLA STANZA IN CUI SI
TROVA IL DISPOSITIVO OPPURE QUALSIASI ALTROACCESSORIO PER L’EROGAZIONE DELL’OSSIGENO.
AVVERTENZA: FUMARE MENTRE SI INDOSSA UNACANNULA PER L’OSSIGENO PUÒ PROVOCAREUSTIONI AL VISO E PROBABILMENTE LA MORTE.
AVVERTENZA: SE SI FUMA, QUESTI 3 PASSAGGIPOSSONO SALVARE LA VITA: SPEGNERE IL
CONCENTRATORE DI OSSIGENO, RIMUOVERE LACANNULA E LASCIARE LA STANZA IN CUI SI TROVAIL DISPOSITIVO.
AVVERTENZA: IN CASA O NEL LUOGO IN CUI ÈUTILIZZATO L’OSSIGENO OCCORRE AFFIGGERECARTELLI “NON FUMARE – OSSIGENO IN USO”
IN MODO CHE SIANO BEN VISIBILI. I PAZIENTI ECOLORO CHE PRESTANO ASSISTENZA DEVONOESSERE INFORMATI DEI PERICOLI DEL FUMO INPRESENZA O DURANTE L’UTILIZZO DI OSSIGENO
MEDICO.
AVVERTENZA: NON UTILIZZARE IL CONCENTRATOREDI OSSIGENO IN PRESENZA DI GAS INFIAMMABILI,POICHÉ POSSONO VERIFICARSI FENOMENIDI COMBUSTIONE RAPIDA, CHE POSSONOCOMPORTARE DANNI MATERIALI, LESIONICORPOREE O IL DECESSO.
AVVERTENZA: RIMUOVENDO LA CANNULA E
POSIZIONANDOLA SU INDUMENTI, BIANCHERIA,DIVANI O ALTRI MATERIALI IMBOTTITI SI POSSONOVERIFICARE LAMPI DI FUOCO IN CASO DIESPOSIZIONE A SIGARETTE, FONTI DI CALORE O
FIAMME.
AVVERTENZA: NON LASCIARE LA CANNULA
NASALE SOPRA O SOTTO INDUMENTI, COPERTE O
CUSCINI DI SEDIE. SE L’UNITÀ È ACCESA MA NON
IN USO, L’OSSIGENO PUÒ RENDERE INFIAMMABILE
IL MATERIALE. QUANDO IL CONCENTRATOREDI OSSIGENO NON È IN USO, IMPOSTARE
L’INTERRUTTORE DI ALIMENTAZIONE I/0 SULLAPOSIZIONE 0 (OFF).

76 - ITLPN 15062781-C4 J Manuale dell’utente

AVVERTENZA: NON TENTAREDI ESEGUIRE INTERVENTI DI
MANUTENZIONE, A ECCEZIONE DELLE
SOLUZIONI POSSIBILI ELENCATE
ALL’INTERNO DEL PRESENTE MANUALE.
NON RIMUOVERE LE COPERTURE, TALEOPERAZIONE O GLI INTERVENTI DI
ASSISTENZA DEVONO ESSERE ESEGUITIESCLUSIVAMENTE DAL FORNITORE
DELL’APPARECCHIATURA O DA UNTECNICO DI ASSISTENZA QUALIFICATO.
AVVERTENZA: IMPIEGARE ESCLUSIVAMENTEACCESSORI RACCOMANDATI DAL PRODUTTORE.
L’USO DI ALTRI ACCESSORI PUÒ ESSEREPERICOLOSO, CAUSARE GRAVI DANNI AL
CONCENTRATORE DI OSSIGENO E RENDERE NULLALA GARANZIA.
ATTENZIONE: scollegare sempre
l’alimentatore CA dalla presa di rete
prima di scollegarlo dal concentratore
di ossigeno.
ATTENZIONE: posizionare sempre il tubo dierogazione dell’ossigeno e i cavi di alimentazione
in modo che non vi sia pericolo di inciampare.
ATTENZIONE: per evitare l’annullamentodella garanzia, attenersi a tutte leistruzioni del produttore.
Nota: le apparecchiature mobili di comunicazione a RF
possono avere effetti sui dispositivi elettromedicali.
AVVERTENZA: POSIZIONARE SEMPRE IL TUBO DI
ALIMENTAZIONE OSSIGENO E I CAVI DI ALIMENTAZIONE IN UN MODO CHE PREVEDE IL PERICOLODI VIAGGIO O POSSIBILE STRANGOLAZIONEACCIDENTALE.
AVVERTENZA: QUESTO PRODOTTO PUÒ ESPORRE
L’UTENTE A SOSTANZE CHIMICHE COMPRENDENTIIL NICHEL, NOTO ALLO STATO DELLA CALIFORNIAPER LE SUE PROPRIETÀ CANCEROGENE. PER MAGGIORI INFORMAZIONI, VISITARE L’INDIRIZZO WWW.
P65WARNINGS.CA.GOV.
AVVERTENZA: NEL CASO IN CUI SI VERIFICHI UN
GRAVE INCIDENTE CON QUESTO DISPOSITIVO,
L’UTENTE DEVE SEGNALARE IMMEDIATAMENTEL’INCIDENTE AL FORNITORE E / O AL PRODUTTORE.
UN INCIDENTE GRAVE È DEFINITO COME UNA LESIONE, MORTE O POSSIBILITÀ DI CAUSARE LESIONI /MORTE IN CASO DI RIPETIZIONE DELL’INCIDENTE.
L’UTENTE PUÒ ANCHE SEGNALARE L’INCIDENTE
ALL’AUTORITÀ COMPETENTE NEL PAESE IN CUI SI È
VERIFICATO L’INCIDENTE.

Companion 5
Specifiche
Lo schema seguente fornisce informazioni importanti
sulle condizioni o sugli ambienti operativi consigliati
per un uso corretto del dispositivo.

SpecifichePortata
Da 0,5 a 5,0 l/min
Concentrazione O2 90% (+5,5%/-3%)
Dimensioni
54,6x31,8x34,2 cm(21,5x12,5x13,5”)
Peso
16,3 kg (36 lb)
Livello di
50 dB(A) a 2,0 l/min
pressione sonora
Consumo
285 W a 2 LPM, 350 W alenergetico
massimoTemperatura di da 5 °C a 40 °C
esercizio(da 41 °F a 104 °F)
Umidità di
15% - 90% a un punto di rugiada
eserciziodi 28 °C (82,4 °F)
Temperatura di
Da -25 ºC a 70 ºC
conservazione(da -13 ºF a 158 ºF)
Umidità di
0% - 90% senza condensa
conservazione
Specifiche
Non utilizzare prolunghe.
elettriche
Non utilizzare prese elettriche
controllate da un interruttore.
Altitudine dida -382 a 3011 metri
esercizio(da -1253 a 9879 piedi)
Durata di esercizio fino a 24 ore al giorno
La durata prevista dell’apparecchiatura è di almeno
cinque anni.
AVVERTENZA: L’USO DEL DISPOSITIVOFUORI DA CONDIZIONI OPERATIVESPECIFICATE È PREVISTO PER INFLUIRESULLA PORTATA E PERCENTUALE DIOSSIGENO E DI CONSEGUENZA SULLAQUALITÀ DELLA TERAPIA.
Nota: se il concentratore di ossigeno è rimasto
conservato per periodi prolungati al di fuori
dell’intervallo normale di temperatura di esercizio,
è necessario lasciare che raggiunga la normale
temperatura operativa prima di accenderlo (consultare
la sezione Specifiche tecniche del presente manuale).

Posizionamento corretto del Companion 5

Scegliere una posizione per il dispositivo in modo daevitare l’ingresso di fumo, vapori e agenti inquinanti.
Il corretto posizionamento del dispositivo deve
consentire l’entrata dell’aria dalle tre posizioni diingresso nella parte superiore posteriore e al di sottodell’alloggiamento, oltre che permettere all’aria discarico di fuoriuscire liberamente dallo sfiato di scarico
nella parte inferiore sinistra del dispositivo.
Gli ingressi dell’aria sono posizionati nella partesuperiore posteriore e nella parte inferiore delconcentratore. Posizionare l’unità in modo che ci
siano almeno 30 cm (12 pollici) di distanza fra il
concentratore ed eventuali pareti, accessori o altri
ostacoli.
Posizionare il dispositivo in modo da poter udire gli
allarmi.
Posizionare il tubo di erogazione dell’ossigeno inmodo che non si annodi e non sia ostruito.
AVVERTENZA: NON UTILIZZARE IL
CONCENTRATORE DI OSSIGENO INPRESENZA DI GAS INFIAMMABILI,POICHÉ POSSONO VERIFICARSIFENOMENI DI COMBUSTIONE RAPIDA,
CHE POSSONO COMPORTARE DANNIMATERIALI, LESIONI CORPOREE O IL
DECESSO. NON UTILIZZARE PRODOTTI
A BASE DI OLIO, GRASSO O PETROLIOO ALTRI PRODOTTI INFIAMMABILI CON
GLI ACCESSORI PER IL TRASPORTO
DELL’OSSIGENO O IL CONCENTRATOREDI OSSIGENO. UTILIZZAREESCLUSIVAMENTE LOZIONI O UNGUENTIA BASE D’ACQUA COMPATIBILI CON
L’OSSIGENO. L’OSSIGENO ACCELERALA COMBUSTIONE DELLE SOSTANZEINFIAMMABILI.
AVVERTENZA: QUESTO DISPOSITIVO EROGAOSSIGENO A ELEVATA CONCENTRAZIONE CHE
FAVORISCE IL RAPIDO SVILUPPO DI USTIONI.
NON CONSENTIRE AL PAZIENTE DI FUMARE O
DI USARE FIAMME LIBERE ALL’INTERNO DELLA
STANZA IN CUI SI TROVA (1) QUESTO DISPOSITIVO
O (2) QUALSIASI ALTRO ACCESSORIO PER IL
TRASPORTO DELL’OSSIGENO. IL MANCATORISPETTO DI QUESTA AVVERTENZA PUÒ
COMPORTARE GRAVI INCENDI E DANNI MATERIALIE/O PROVOCARE LESIONI FISICHE O IL DECESSO.

PN 15062781-C4 J | Manuale dell’utente

77 - ITLCompanion 5
Istruzioni per l’uso
Prima dell’uso

Il presente Manuale dell’utente va utilizzato comeriferimento per l’uso e la manutenzione del dispositivo.
In caso di dubbi o domande, rivolgersi al fornitore di
assistenza domiciliare.
Importante! NON tentare di utilizzare Companion
5 senza prima aver letto la sezione Linee guida disicurezza del presente manuale. Seguire tutte leistruzioni per l’uso. Osservare tutte le Avvertenze
indicate sul dispositivo e nel Manuale dell’utente. Per
ridurre il rischio di incendio, lesioni personali e gravi
danni al dispositivo Companion 5, osservare tutte leprecauzioni di sicurezza.
Companion 5 di CAIRE non crea autonomamente
il proprio ossigeno. Produce ossigeno altamente
concentrato traendolo dall’aria della stanza circostante,
e lo fornisce all’utilizzatore. Allo stato naturale, l’aria
dell’ambiente contiene in volume approssimativamente
il 21% di ossigeno.
L’aria dell’ambiente viene aspirata nel concentratore
grazie a un compressore. Dopo, l’aria si fa strada lungo
una serie di filtri e setacci molecolari che assorbono
chimicamente le molecole di azoto. Di conseguenza,
l’aria arricchita di ossigeno che viene espulsa dalconcentratore viene somministrata con un volume diossigeno compreso tra l’87 e il 95%.
AVVERTENZA: PULIRE
L’ALLOGGIAMENTO, IL PANNELLO
DI CONTROLLO E IL CAVO DI
ALIMENTAZIONE SOLO CONDETERGENTE DELICATO PER USO
DOMESTICO, APPLICANDOLO CON UNASPUGNA O UN PANNO UMIDO (NON
BAGNATO), E POI ASCIUGARE TUTTELE SUPERFICI. NON FAR PENETRARE
LIQUIDI NEL DISPOSITIVO. PRESTARE
PARTICOLARE ATTENZIONE ALL’USCITA
DELL’OSSIGENO PER IL COLLEGAMENTODELLA CANNULA E ACCERTARSI CHERESTI PRIVA DI POLVERE, ACQUA EPARTICELLE.
ATTENZIONE: non far bloccare l’ingressoo gli sfiati di uscita dell’aria. In questo
modo il concentratore di ossigeno
potrebbe surriscaldarsi pregiudicando
le prestazioni. non utilizzare il
concentratore di ossigeno senza il filtro
di ingresso dell’aria in posizione. Se è
disponibile un secondo filtro, inserire
il filtro “di ricambio” prima di pulire
quello sporco. Pulire il filtro sporco con
una soluzione tiepida di acqua e sapone
e asciugare a fondo prima dell’uso.

78 - ITLPN 15062781-C4 J Manuale dell’utenteFase 1: posizionare Companion 5 per l’utilizzo
Posizionare Companion 5 in un’area ben ventilata e
ben illuminata. Assicurarsi che l’ingresso dell’aria
e gli sfiati di scarico non siano ostruiti. Posizionare
il dispositivo Companion 5 in modo che tutti gli
indicatori o gli allarmi acustici e visivi possano essere
visti e uditi facilmente.
Collegare il dispositivo a una presa CA.
Fase 2: se il flacone umidificatore NON viene
utilizzato

Posizionedell’ingresso
dell’aria

Posizionedell’ingresso
dell’aria diraffreddamento

Sfiato discaricoPosizionedell’ingresso
dell’aria

a. Collegare una cannula nasale o un tubo per ossigeno
direttamente al giunto di sfogo dell’ossigeno.

b. Procedere alla fase 4.
Fase 3: se il flacone umidificatore VIENE utilizzato
a. Svitare il coperchio dal flacone umidificatore.
b. Riempire la bottiglia utilizzando acqua distillata.
Accertarsi che il livello dell’acqua si attesti tra ilmassimo indicato dal produttore e il minimo indicato
sul flacone.
c. Fissare il coperchio del flacone umidificatore.
Accertarsi che non siano presenti perdite.
d. Collegare una cannula nasale o tubi per ossigeno
standard al giunto di sfiato sul flacone umidificatore.
Nota: l’utilizzo di un umidificatore non indicato
per l’utilizzo con questo concentratore potrebbe
intaccarne le prestazioni.

Companion 5
g. Posizionare il flacone nella posizione designata sul
supporto per flacone umidificatore.
h. Fissare il flacone sul supporto utilizzando l’elastico.

e. Collegare l’estremità filettata del tubodell’umidificatore al coperchio del flacone.

i. Procedere alla fase 4.
Fase 4: accensione e riscaldamento
a. Collegare il cavo di alimentazione del Companion 5
di CAIRE a una presa CA.
AVVERTENZA: PRIMA DELL’USO,CONTROLLARE CHE IL CAVO DI
ALIMENTAZIONE NON PRESENTI DANNI.
SE IL CAVO RISULTA DANNEGGIATO,NON COLLEGARLO A UNA PRESAELETTRICA NÉ PROVARE A UTILIZZARE IL
CONCENTRATORE.

f. Collegare l’altra estremità del tubodell’umidificatore al giunto di sfiato delCompanion 5 di CAIRE.

b. Premere l’interruttore di alimentazione per portarloin posizione “ON” (I). Quando il dispositivo
Companion 5 viene collegato correttamente, si
accenderà l’indicatore verde sul display LED.
Tutte le luci LED si illumineranno all’avvio. Dopoche il concentratore avrà completato il ciclo di
riscaldamento, rimarrà accesa soltanto la luce verde.
Nota: una volta acceso, attendere fino a
dieci (10) minuti affinché il dispositivo possa
raggiungere le specifiche prestazionali.

PN 15062781-C4 J | Manuale dell’utente

79 - ITLCompanion 5
AVVERTENZA: È MOLTO IMPORTANTESELEZIONARE SOLO IL LIVELLODI OSSIGENO PRESCRITTO. NON
MODIFICARE LA SELEZIONE DEL
FLUSSO SE NON IN CASO DI PRECISEISTRUZIONI DA PARTE DI UN MEDICOAUTORIZZATO. È POSSIBILE UTILIZZAREIL CONCENTRATORE DI OSSIGENO
DURANTE IL SONNO SOLO SURACCOMANDAZIONE DI UN MEDICOQUALIFICATO.
Fase 5: regolare la portata di controllo del flusso

Fase 6: verificare il flusso e respirare normalmente

a. Ruotare la manopola di controllo del flusso fino alla
portata (LPM) prescritta dal medico.

a. Verificare che l’ossigeno stia effettivamente
affluendo attraverso la cannula nasale e che non ci
siano pieghe, curve o ostruzioni nei tubi.
Nota: accertarsi che la cannula sia completamente
inserita e salda. Si dovrebbe udire o avvertire il flusso
di ossigeno attraverso la biforcazione della cannula
nasale.
b. Posizionare adeguatamente la cannula nasale e
respirare normalmente attraverso essa.
Fase 7: spegnere il dispositivo
a. Premere il pulsante di alimentazione per spostarloin posizione “OFF” (O) quando non si utilizza più il
dispositivo Companion 5 di CAIRE.

Per regolare la portata:
Ruotare in senso antiorario per aumentare il flusso.
Ruotare in senso orario per diminuire il flusso.
La parte centrale della sfera indica la portata.
L’immagine riportata sotto indica una portata di
4,0 l/min.

Corretto

ATTENZIONE: utilizzare esclusivamente
gli accessori raccomandati dal
produttore. L’uso di altri accessori può
essere pericoloso, causare gravi danni
al concentratore di ossigeno e rendere
nulla la garanzia.
ATTENZIONE: non utilizzare prolunghe con questa
unità o evitare di collegare troppe spine alla stessa
presa elettrica. L’uso di prolunghe può avere
effetti avversi sulle prestazioni del dispositivo. Il
collegamento di troppe spine a un’unica presa può
provocare un sovraccarico del quadro elettrico,causando l’attivazione dell’interruttore/fusibileo un incendio in caso di mancato funzionamentodell’interruttore/fusibile.
ATTENZIONE: l’uso di cavi e adattatori diversi daquelli specificati, a eccezione di quelli venduti
dal produttore del dispositivo elettromedicale
come parti di ricambio per i componenti interni,può provocare un aumento delle emissioni e una
riduzione dell’immunità del concentratore diossigeno.
AVVERTENZA: NON FAR BLOCCAREL’INGRESSO O GLI SFIATI DI USCITA
DELL’ARIA. IN QUESTO MODO IL
CONCENTRATORE DI OSSIGENO
POTREBBE SURRISCALDARSIPREGIUDICANDO LE PRESTAZIONI.

80 - ITLPN 15062781-C4 J Manuale dell’utenteCompanion 5
Nota: al fornitore dell’apparecchiatura: si consiglia diutilizzare i seguenti accessori per la somministrazione
di ossigeno con il Companion 5:• Cannula nasale: numero parte CAIRE CU002-1• Tubo adattatore umidificatore: numero parte CAIRE
20843882• Bottiglia umidificatore: numero parte CAIRE HU0031• Firebreak: numero parte CAIRE 20629671
Un incendio è richiesto per l’uso con qualsiasi cannula.
• CAIRE offre un incendio progettato per essere
utilizzato insieme al concentratore di ossigeno.
L’incendio è un fusibile termico per arrestare il flusso
di gas nel caso in cui la cannula a valle o il tubodell’ossigeno vengano accesi e brucino all’incendio.
È posizionato in linea con la cannula nasale o il tubodell’ossigeno tra il paziente e l’uscita dell’ossigeno
di Companion 5. Per un uso corretto del focolaio,
consultare sempre le istruzioni del produttore (inclusein ciascun kit di focolaio).
• Per eventuali accessori aggiuntivi consigliati,
consultare il catalogo degli accessori (PN MLLOX0010)
disponibile su www.caireinc.com.

PN 15062781-C4 J | Manuale dell’utente

81 - ITLCompanion 5
Condizioni di allarme
Il dispositivo Companion 5 di CAIRE utilizza una
combinazione di due o tre (per i modelli con OCSI)
luci LED e un allarme sonoro per avvertire dellapresenza di un malfunzionamento nel concentratore.
1) LED verde () – Indica il normale
funzionamento.
2) LED rosso ( ) – Indica un malfunzionamento delsistema. Consultare la tabella delle Condizioni diallarme qui sotto per maggiori dettagli.

Verde

Rosso

Giallo3) LED giallo ( ) – Indica una bassa concentrazione
di ossigeno. Consultare la tabella delle Condizioni
di allarme qui sotto per maggiori dettagli.
La tabella delle Condizioni di allarme nella pagina
seguente mostra le differenti condizioni di allarmeche possono essere visualizzate dal concentratore.
Definisce le possibili cause e le azioni da intraprenderein caso di allarme. In caso di qualsiasi allarme, si
consiglia di contattare quanto prima possibile il proprio
fornitore di assistenza sanitaria.
AVVERTENZA: NON IGNORARE GLIALLARMI
AVVERTENZE: IL PRODUTTORECONSIGLIA DI TENERE A DISPOSIZIONEUN’ALTRA SORGENTE DI OSSIGENO
SUPPLEMENTARE IN CASO DI
INTERRUZIONI DI CORRENTE,
CONDIZIONI DI ALLARME O GUASTIMECCANICI. PER CONOSCERE IL TIPO
DI SISTEMA DI RISERVA NECESSARIO,RIVOLGERSI AL PROPRIO MEDICO O AL
FORNITORE DELL’APPARECCHIATURA.

82 - ITLPN 15062781-C4 J Manuale dell’utente

Sopra: Companion 5

Companion 5
Tabella Condizioni di allarme
AllarmeacusticoLEDcolorato

Causa

AzioneNessunoIl concentratore funziona
a dovere.

SpentoVerde

Intermittente
ROSSO

Intermittente

Il concentratore ha perso
l’alimentazione, mal’interruttore è ancora inposizione “ON”.

Verificare che il cavo di alimentazione delconcentratore sia collegato a una presa, e che lapresa disponga di alimentazione. Provare una presadifferente. Se il problema persiste, collegare a una
fonte di ossigeno alternativa e contattare il fornitore
di assistenza sanitaria.

Malfunzionamento delsistemaSpostare immediatamente l’interruttore dialimentazione in posizione “OFF”. Scollegare
l’alimentazione CA dalla presa a muro. Attendere
5 minuti. Collegare nuovamente l’alimentazione
CA alla presa a muro e accendere nuovamente ilconcentratore. Se l’allarme persiste, sarà necessariorichiedere assistenza. Collegare a una fonte diossigeno alternativa e contattare immediatamente ilfornitore di assistenza sanitaria.

Portata del prodotto
troppo alta o troppo bassa

1. Assicurarsi che la cannula non sia piegata obloccata. Se si utilizza un flacone umidificatore,
assicurarsi che sia riempito correttamente e non crei
un blocco.
2. Accertarsi che Companion 5 disponga di adeguata
ventilazione. È necessario che si trovi ad almeno acirca 30 cm (12 pollici) da qualsiasi superficie pergarantire che le ventole non siano bloccate.
3. Se il problema persiste, passare a una fonte diossigeno alternativa e rivolgersi al fornitore di
assistenza sanitaria.

ROSSO

Intermittente
ROSSO

Intermittente
GialloIntermittente
GialloIl dispositivo Companion
5 ha rilevato bassi livelli diossigeno.

Il dispositivo Companion
5 ha rilevato bassi livelli diossigeno.

1. Accertarsi che il filtro dell’ingresso dell’aria e leposizioni di sfiato non siano ostruite o ristrette.
2. Accertarsi che Companion 5 si trovi in un area
ben ventilata. Assicurarsi che vi siano almeno 30 cm
circa (12 pollici) tra il retro e i lati del dispositivo
Companion e qualsiasi tipo di ostruzione (mobili,tende, ecc.).
3. Se la condizione persiste, passare a una fonte diossigeno alternativa e rivolgersi immediatamente alfornitore di assistenza sanitaria.

PN 15062781-C4 J | Manuale dell’utente

83 - ITLCompanion 5
Risoluzione dei problemi
Si consiglia di contattare il proprio fornitore di
assistenza sanitaria se il display visualizza qualsiasi
tipo di allarme o se si pensa che il concentratore
non funzioni a dovere. Si sconsiglia di tentare
di riparare autonomamente il concentratore. Le
riparazioni devono essere eseguite esclusivamente da
un tecnico dell’assistenza appositamente addestrato.

Tuttavia, esistono alcune procedure di risoluzione dei
problemi che è possibile adottare se si riscontrano
problemi nel dispositivo Companion 5 di CAIRE.
Le procedure sono illustrate nella seguente tabella. È
possibile utilizzare questo schema seguendo in ordine
numerico le verifiche relative al proprio problema.

Tabella risoluzione dei problemi da parte dell’utente
ProblemaIl concentratoreè acceso,
ma non è infunzione.

Si riscontra un
basso flusso
di ossigeno ol’assenza diossigeno.

Possibile causa

Azione1. Il cavo di alimentazione non è statocollegato a una presa.

1. Collegare il cavo elettrico.

2. La presa della corrente non fornisce
alimentazione.

2. Controllare il fusibile e il circuito elettrico di casa.
Provare una presa differente.

3. Guasto interno.

3. Collegare a una fonte di ossigeno alternativa.
Contattare immediatamente il proprio fornitore di
assistenza sanitaria.

1. L’impostazione di controllo delflusso è stata cambiata.

1. Verificare che la manopola di controllo del flusso sia
impostata sul valore di litri al minuto (LPM) adeguato
prescritto dal medico.

2. I tubi sono stati scollegati.

2. Verificare che i tubi siano ben saldi a ogni
collegamento (giunto di uscita, flacone umidificatore,
sifone dell’acqua, ecc.) Fissare nuovamente i
collegamenti, se necessario.

3. I tubi per ossigeno o la cannula
risultano piegati.

3. Controllare che non siano presenti pieghe o ostruzioni
nei tubi. Riparare se necessario.

4. Si ha una perdita nella cannula onei tubi.

4. Ispezionare tutti i tubi e la cannula cercando perdite.
È trovarle ascoltando o avvertendo fisicamente lefuoriuscite d’aria. Sostituire il tubo o la cannula con la
perdita.

5. Il flusso d’aria nel concentratore
risulta bloccato.

5. Verificare che nulla ostruisca gli ingressi delconcentratore. Assicurarsi che il dispositivo sia lontano
almeno 30 cm (12 pollici) da accessori, pareti o altri
ostacoli.

6. Guasto interno.

6. Collegare a una fonte di ossigeno alternativa.
Contattare immediatamente il proprio fornitore di
assistenza sanitaria.

Nota: se Companion 5 continua a emettere allarmi dopo aver utilizzato le tabelle Condizioni di allarme e Risoluzione
dei problemi, contattare l’assistenza tecnica al numero 1-800-482-2473 e comunicare il codice di allarme
visualizzato sul display LCD.

84 - ITLPN 15062781-C4 J Manuale dell’utenteCompanion 5
Pulizia, cura e manutenzione ordinaria
Manutenzione ordinaria

La manutenzione ordinaria consiste nella sostituzionedel filtro antiparticolato della maniglia una volta asettimana (se installato).

Cura e pulizia del dispositivo da partedell’utilizzatore

Sostituzione della cannula
Sostituire regolarmente la cannula e il tubo dierogazione come raccomandato dal fornitore di
assistenza domiciliare. Il medico o il fornitore
di assistenza domiciliare fornirà le informazioni
riguardanti la pulizia, la disinfezione e la sostituzione.
Nota: Per l’utilizzo adeguato, attenersi sempre alleistruzioni del produttore della cannula. Sostituire lacannula monouso come consigliato dal produttoredella cannula o dal fornitore dell’apparecchiatura. Il
fornitore dell’apparecchiatura è in grado di fornire
ulteriori pezzi di ricambio.

Filtro di ingresso dell’aria (cartuccia)
Il filtro di ingresso dell’aria deve soltanto essere
trattato o sostituito dal fornitore dei servizi sanitari.
Filtro antiparticolato (opzionale):

Il filtro antiparticolato lordo si trova sul fondo
molto anteriore del Companion 5.

4) Far asciugare il filtro all’aria.
5) Reinserire il filtro nell’alloggiamento.
Nota: il produttore non raccomanda la sterilizzazione
di questo dispositivo.
Nota: non utilizzare il concentratore di ossigeno senza
il filtro di ingresso dell’aria in posizione.

Flacone umidificatore (se applicabile)
È necessario controllare il flacone umidificatoregiornalmente per assicurarsi che il livello dell’acqua
sia compreso tra il valore minimo e quello massimo.
Riempire la bottiglia quanto necessario utilizzando
acqua distillata. Pulire il flacone umidificatore secondo
le istruzioni del produttore.
Procedura di pulizia:
spegnere il dispositivo Companion 5 e scollegarlo
dall’alimentazione CA prima di qualsiasi attività dipulizia o disinfezione. NON spruzzare direttamente
alcun prodotto sul telaio esterno. Utilizzare una
spugna o un panno umido (non bagnato). Per pulirel’alloggiamento e il cavo di alimentazione, spruzzare
una soluzione detergente delicata sul panno o sulla
spugna. Per disinfettare il dispositivo Companion
5, utilizzare il disinfettante Lysol® Brand II (oequivalente). Diluire secondo quanto richiesto dal
produttore del prodotto detergente, ma non spruzzare
direttamente il liquido sul dispositivo Companion 5.
AVVERTENZA: PERICOLO DI SCOSSA
ELETTRICA. SPEGNERE L’UNITÀ E
SCOLLEGARE IL CAVO DI ALIMENTAZIONEDALLA PRESA ELETTRICA PRIMA DI
PULIRE L’UNITÀ PER PREVENIRE IL
PERICOLO DI SCOSSE ELETTRICHE E DI
USTIONI ACCIDENTALI. GLI INTERVENTIDI RIPARAZIONE E LA RIMOZIONE DELLE
COPERTURE DEVONO ESSERE ESEGUITIESCLUSIVAMENTE DAL FORNITORE
DELL’APPARECCHIATURA O DA UNTECNICO DI ASSISTENZA QUALIFICATO.

Filtroantiparticolato:È necessario ispezionarlo e pulirlo una volta alla
settimana. Per pulire il filtro, utilizzare la seguente
procedura:
1) Rimuovere il filtro antiparticolato.
2) Lavare il filtro in acqua di rubinetto tiepida
utilizzando una soluzione detergente con sapone
delicato.

AVVERTENZA: PERICOLO DI SCOSSA ELETTRICA.
SPEGNERE L’UNITÀ E SCOLLEGARE IL CAVODI ALIMENTAZIONE DALLA PRESA ELETTRICA
PRIMA DI PULIRE L’UNITÀ PER PREVENIRE IL
PERICOLO DI SCOSSE ELETTRICHE E DI USTIONIACCIDENTALI. GLI INTERVENTI DI RIPARAZIONE E
LA RIMOZIONE DELLE COPERTURE DEVONO ESSERE
ESEGUITI ESCLUSIVAMENTE DAL FORNITORE
DELL’APPARECCHIATURA O DA UN TECNICO DI
ASSISTENZA QUALIFICATO.

3) Sciacquare accuratamente il filtro con acqua dirubinetto pulita ed eliminare l’acqua in eccesso.
PN 15062781-C4 J | Manuale dell’utente

85 - ITLCompanion 5
AVVERTENZA: PERICOLO DI SCOSSA ELETTRICA.
SCOLLEGARE IL CAVO DI ALIMENTAZIONE DALLAPRESA ELETTRICA PRIMA DI PULIRE L’UNITÀ PER
PREVENIRE IL PERICOLO DI SCOSSE ELETTRICHEE USTIONI ACCIDENTALI. GLI INTERVENTI DI
RIPARAZIONE E LA RIMOZIONE DELLE COPERTURE
DEVONO ESSERE ESEGUITI ESCLUSIVAMENTE DALFORNITORE DELL’APPARECCHIATURA O DA UNTECNICO DI ASSISTENZA QUALIFICATO.
AVVERTENZA: PRESTARE ATTENZIONE A EVITARE
CHE IL CONCENTRATORE DI OSSIGENO SI BAGNIO CHE FLUIDI ENTRINO NELL’UNITÀ. CIÒ PUÒ
PROVOCARE UN MALFUNZIONAMENTO O UN
ARRESTO E UN AUMENTO DEL RISCHIO DI USTIONIO SCOSSE ELETTRICHE.
ATTENZIONE: PUO ‘ESSERE SICUROINTERCONNETTERE IL COMPANION 5 CON ALTRE
ATTREZZATURE.

AVVERTENZA: PULIRE L’ALLOGGIAMENTO,
IL PANNELLO DI CONTROLLO E IL CAVO DI
ALIMENTAZIONE SOLO CON DETERGENTE
DELICATO PER USO DOMESTICO, APPLICANDOLOCON UNA SPUGNA O UN PANNO UMIDO (NON
BAGNATO), E POI ASCIUGARE TUTTE LE SUPERFICI.
NON FAR PENETRARE LIQUIDI NEL DISPOSITIVO.
PRESTARE PARTICOLARE ATTENZIONE ALL’USCITA
DELL’OSSIGENO PER IL COLLEGAMENTO DELLACANNULA E ACCERTARSI CHE RESTI PRIVA DIPOLVERE, ACQUA E PARTICELLE.
AVVERTENZA: NON UTILIZZARE LIQUIDIDIRETTAMENTE SULL’UNITÀ. L’ELENCO DELLESOSTANZE CHIMICHE NON DESIDERATECOMPRENDE, IN VIA NON LIMITATIVA: ALCOLE PRODOTTI A BASE ALCOLICA, PRODOTTI
CONCENTRATI A BASE DI CLORO (CLORURODI ETILENE) E PRODOTTI A BASE OLEOSA(PINE-SOL®, LESTOIL®). QUESTE SOSTANZE NONVANNO UTILIZZATE PER PULIRE L’ALLOGGIAMENTODI PLASTICA SUL CONCENTRATORE DI OSSIGENO,
POICHÉ POSSONO DANNEGGIARE LE PARTI IN
PLASTICA DELL’UNITÀ.

Smaltimento
Restituire sempre Companion 5, inclusi tutti icomponenti, al fornitore di assistenza domiciliare peril corretto smaltimento. È inoltre possibile rivolgersi
agli uffici locali per istruzioni sul corretto smaltimentodella batteria.

RAEE e RoHS

I nostri prodotti sono conformi alle restrizioni dettate
dalla Direttiva sulle sostanze pericolose (RoHS). È
possibile che essi contengano tracce, ma non quantità
rilevanti, di piombo o altri materiali pericolosi.
ATTENZIONE: per uno smaltimento
adeguato, contattare il fornitore dell’attrezzatura o l’agenzia governativa locale
per le istruzioni sullo smaltimento.

Questo simbolo ha lo scopo di ricordare ai
proprietari dell’apparecchio di consegnarlo presso uncentro per il riciclaggio al termine della durata utile, inconformità alla Direttiva sui rifiuti di apparecchiature
elettriche ed elettroniche (RAEE).

CAIRE e CAIRE Inc. sono marchi registrati di CAIRE Inc. Visitare
il nostro sito Web di seguito per un elenco completo dei marchi.
Marchi: www.careinc.com/corporate/trademarks.
Copyright © 2020 CAIRE Inc.
CAIRE Inc. si riserva il diritto
di interrompere la produzione o di modificare prezzi, materiali,
apparecchiature, qualità, descrizioni, specifiche e/o processi dei
propri prodotti in qualsiasi momento, senza preavviso e senza
ulteriori obblighi o conseguenze. Tutti i diritti non espressamente
indicati nel presente documento sono riservati, laddove applicabili.
86 - ITLPN 15062781-C4 J Manuale dell’utenteCompanion 5
Controlos do utilizador e Indicadores de estado do sistema
Equipamento aplicado tipo BF (grau deproteção contra choques elétricos).
Reg. n.º 5333

Símbolos internos

02

Saída de Oxigénio

I/O Interruptor ligar/desligarNão existem peças reparáveis nointerior. Não abrir a cobertura.

Diretiva do Conselho 93/42/EEC; respeitante a
dispositivos médicos
Representante autorizado na União
Europeia
IEC 60417

ISO 7000; Símbolos gráficos para utilização noequipamento—Índice e resumo
Faixa de temperatura dearmazenamento ou operação. Reg. n.º0632

Equipamento de Classe II, com
Isolamento Duplo Reg. # 5172
Certificação ETLAgência de segurança da CAN/CSA
C22.2 Nº 601.1 M90 respeitante aequipamentos médicos elétricos

Faixa de umidade de armazenamento.
Reg. n.º 2620
Nome e morada do fabricante.
Reg. n.º 3082
Data de fabrico. Reg. n.º 2497
Número de catálogo. Reg. n.º 2493
Número de série. Reg. n.º 2498
Este lado para cima. Reg. n.º 0623

Nota da FCC
21 CFR 801.15: Código dos regulamentos federais
Título 21
A lei federal restringe a venda deste
dispositivo a ou por ordem de um
médico.
Diretiva do conselho 2012/19/EU: resíduos deequipamentos elétricos e eletrónicos (REEE)
WEEE

Frágil, manusear com cuidado.
Reg. n.º 0621
Manter afastado da chuva, manter seco.
Reg. n.º 0626
Ler o manual do utilizador antes deutilizar. Reg. n.º 1641
Limite de empilhamento por número.
Reg. n.º 2403
ISO 7010: Símbolos gráficos—Cores e sinalização
de segurança—Sinalização de segurança
registada
O manual de instruções tem de ser lido.
Reg. n.º M002
Aviso. Reg. n.º W001
Manter afastado de chamas e faíscas.
É proibido fumar e são proibidas fontes
de ignição abertas. Reg. n.º P003
Não fume perto da unidade ou quando
a estiver a utilizar. Reg. n.º P002

IEC 60601-1: Equipamento médico elétrico
Parte 1 Requisitos gerais de segurança básica e
desempenho essencial

IPX1

Equipamento IPX1 à prova degotejamento: O Companion 5 fornece
proteção contra os efeitos prejudiciais
da penetração de líquidos. (IPX1,
conforme IEC 60529)

NOTA da FCC:

Este dispositivo pode conter CYBLE-022001-00,
incluindo a antena 2450AT18B100 da Johnson
Technology, em conformidade com a parte 15 dos
regulamentos da FCC. O dispositivo cumpre os
requisitos de aprovação de transmissores modulares,
conforme descrito na Nota pública da FCC,
DA00-1407. O funcionamento do transmissor estásujeito às duas condições seguintes (1) Este dispositivo
não pode provocar interferências prejudiciais,
e (2) Este dispositivo tem de aceitar qualquerinterferência recebida, incluindo interferências que
possam causar um funcionamento indesejado.
Este produto pode estar abrangido por uma ou mais
patentes dos EUA e internacionais. Visite o nosso
website abaixo para obter uma lista das patentes
aplicáveis. Pat.: www.caireinc.com/corporate/patents/
PN 15062781-C4 J | Manual do utilizador

87 - POR

Companion 5
Guia de Iniciação Rápida
1 Desembale o Companion 5

Companion 5

2 Conheça o seu Companion 5
Reveja todos os Avisos, Precauções e outras
informações do dispositivo no restante manual.
Familiarize-se com as principais características
do Companion 5 e com o Painel de Controlo do
Utilizador.

PARTE FRONTAL
Estruturaexterior

Interruptor dealimentação

Ecrã
LED

Oxigénio
Saída
estriada

Contadorde horas

Fluxómetro
Suporteda garrafa
humidificadora(garrafa
opcional)

PARTE POSTERIOR
Localização
da Pega detransporte eda Entrada
de arLocalização
da entrada
de ar derefrigeração

Localização daentrada de ar

88 - PORPN 15062781-C4 J | Manual do utilizador

Saída deexaustãoCompanion 5
Os componentes do Companion 5
Estruturaexterior

Caixa em plástico resistente que
protege os componentes internos do
Companion 5 CAIRE.
Interruptor de Interruptor LIGAR/DESLIGAR usado
alimentação para iniciar ou interromper a
alimentação da unidade. A posição
LIGAR é indicada pelo símbolo “l” e
a posição DESLIGAR é indicada pelosímbolo “O”.
Saída estriada O oxigénio sai da unidade
Companion 5 aqui, após ter sido
filtrado e concentrado. É encaixada
uma cânula nasal de lúmen único
ou tubo de oxigénio padrão
para administrar no doente. O
comprimento máximo do tubo deoxigénio ligado ao concentrador é de15,2 m.
Ecrã LED
Contém uma luz LED verde, vermelha
e amarela*. A luz verde indica um
funcionamento normal e as luzes
amarela* e vermelha indicam
condições de alarme. (Para mais
informações, consulte a secção das
Condições de Alarme nas instruções.)Contador de Ecrã digital do tempo dehoras
funcionamento decorrido doconcentrador. Apresenta o valor
arredondado à decima de hora e nãopode ser redefinido. Exibe códigosde alarme de diagnóstico quando oconcentrador apresenta uma condição
de alarme.
Fluxómetro
Usado para ajustar e apresentar a
taxa de fluxo de oxigénio debitada.
As taxas de fluxo variam entre
0,5 - 5 litros por minuto (LPM) e
podem ser ajustadas rodando o botão.
Local para afixar a garrafa
Suportehumidificadora. O suporte contém
da garrafa
humidificadora uma banda elástica que mantém agarrafa segura no local. A garrafa(garrafa
humidificadora fornece humidade
opcional)
adicional ao fluxo de oxigénio. A
garrafa contém água destilada e ahumidade é usada para prevenir asecura dos tecidos nasais.
*Apenas modelos de Indicador
do Estado do Concentrador deOxigénio (OCSI)

3 Encaixe da Garrafa Humidificadora eCânulaSe pretender usar uma garrafa humidificadora, encha-aprimeiro com água destilada de acordo com o nível
especificado pelo fabricante. Coloque a garrafa nosuporte indicado e fixe-a com a banda elástica. Ligue a
extremidade roscada do tubo humidificador à garrafa e
a outra extremidade à saída estriada do Companion 5.
Ligue o tubo de oxigénio e/ou a cânula nasal à saída
estriada na garrafa humidificadora.

Se não for utilizada uma garrafa humidificadora, ligue
o tubo de oxigénio e/ou a cânula nasal diretamente à
saída estriada do Companion 5.

PN 15062781-C4 J | Manual do utilizador

89 - POR

Companion 5
4 Arranque e Aquecimento
Ligue o cabo de alimentação CA a uma tomada CA.

6 Ajuste a Configuração de Fluxo para oNível Prescrito
Use o Botão de Controlo do Fluxo para selecionar ofluxo prescrito pelo seu médico.

Mude o interruptor de alimentação para a posição de
Ligar (On).

Interruptor dealimentação

Quando o Companion 5 estiver corretamente ligado,o indicador de alimentação verde no ecrã LED ficará
iluminado. Todas as luzes LED ficarão iluminadas
durante o arranque. Após o concentrador concluir o ciclode aquecimento, apenas a luz verde se manterá acesa.
Nota: Após ligá-lo inicialmente, a luz do O2
acenderá e “O2 UP” será exibido na tela até que a
concentração atinja as especificações.

AVISO: É MUITO IMPORTANTESELECIONAR APENAS O NÍVEL DEOXIGÉNIO PRESCRITO. NÃO ALTERE
A SELEÇÃO DO FLUXO, EXCETO SE
TIVER RECEBIDO INDICAÇÕES NESSESENTIDO POR PARTE DE UM MÉDICOQUALIFICADO. O CONCENTRADOR DEOXIGÉNIO PODE SER USADO DURANTEO SONO SOB RECOMENDAÇÃO DE UM
MÉDICO QUALIFICADO.
Respire normalmente através da cânula nasal.
Para ajustar a taxa do fluxo:
Rode no sentido inverso ao dos ponteiros do relógio
para aumentar o fluxo.
Rode no sentido dos ponteiros do relógio para diminuiro fluxo.
O centro da bola indica a taxa de fluxo. Ver imagem
abaixo.

LEDindicador dealimentação
elétrica

Correto90 - PORPN 15062781-C4 J | Manual do utilizador

Companion 5
Importante!

As instruções de segurança estão definidas da
seguinte forma:
AVISO: INFORMAÇÕES DE SEGURANÇA
IMPORTANTES RELATIVAS A PERIGOSQUE PODEM CAUSAR LESÕES GRAVES.
ATENÇÃO: Informações importantes
para evitar danos no Companion 5.
Nota: Informações que necessitam de atenção especial.

Indicações de Utilização
Uso pretendido

O Companion 5 foi concebido para a administração
suplementar de oxigénio. O dispositivo não foi
concebido como aparelho de suporte de vida, nem
tão pouco possui quaisquer funcionalidades demonitorização do doente.
AVISO: É MUITO IMPORTANTESELECIONAR APENAS O NÍVEL DEOXIGÉNIO PRESCRITO. NÃO ALTERE
A SELEÇÃO DO FLUXO, EXCETO SE
TIVER RECEBIDO INDICAÇÕES NESSESENTIDO POR PARTE DE UM MÉDICOQUALIFICADO. O CONCENTRADOR DEOXIGÉNIO PORTÁTIL PODE SER USADO
DURANTE O SONO QUANDO TAL FOR
RECOMENDADO POR UM MÉDICOQUALIFICADO.
AVISO: PARA GARANTIR QUE RECEBE A
QUANTIDADE TERAPÊUTICA DE ADMINISTRAÇÃODE OXIGÉNIO ADEQUADA À SUA CONDIÇÃO
CLÍNICA, O COMPANION 5 SÓ DEVE SER USADODEPOIS DE UMA OU MAIS CONFIGURAÇÕES
TEREM SIDO DETERMINADAS INDIVIDUALMENTEOU PRESCRITAS PARA SI, DE ACORDO COM OS
SEUS NÍVEIS DE ATIVIDADE ESPECÍFICOS. O
COMPANION 5 TEM DE SER UTILIZADO COM A
COMBINAÇÃO ESPECÍFICA DE PEÇAS E ACESSÓRIOS
EM CONFORMIDADE COM AS ESPECIFICAÇÕES
DO FABRICANTE DO CONCENTRADOR E QUE FOI
USADA DURANTE A DETERMINAÇÃO DAS SUASCONFIGURAÇÕES.
AVISO: ESTA UNIDADE NÃO DEVE SER USADA
COMO SUPORTE DE VIDA. OS PACIENTESGERIÁTRICOS, PEDIÁTRICOS, OU OUTROSPACIENTES INCAPAZES DE COMUNICAR OSEU DESCONFORTO, PODERÃO EXIGIR UMA
MONITORIZAÇÃO ADICIONAL E/OU UM SISTEMA
DE ALARME DISTRIBUÍDO PARA TRANSMITIR A
INFORMAÇÃO ACERCA DO DESCONFORTO E/OU AURGÊNCIA MÉDICA AO PRESTADOR DE CUIDADOSRESPONSÁVEL, DE MODO A EVITAR DANOS. OS
DOENTES COM INSUFICIÊNCIA AUDITIVA E/OU
VISUAL PODEM NECESSITAR DE AJUDA COM OS
ALARMES DE MONITORIZAÇÃO.

AVISO: AS MULHERES GRÁVIDAS OU A
AMAMENTAR NÃO DEVEM UTILIZAR OS
ACESSÓRIOS RECOMENDADOS NESTE MANUAL,
UMA VEZ QUE PODEM CONTER FTALATOS.
AVISO: A LEI FEDERAL (EUA) RESTRINGE A VENDAOU ALUGUER DESTE DISPOSITIVO À CLASSEMÉDICA OU A OUTROS PROFISSIONAIS DE SAÚDEQUALIFICADOS.

Contraindicações de utilizaçãoAVISO: EM DETERMINADAS
CIRCUNSTÂNCIAS, A UTILIZAÇÃO DEOXIGÉNIO SEM PRESCRIÇÃO MÉDICA
PODE SER PERIGOSA. ESTE APARELHODEVE SER UTILIZADO APENAS QUANDOPRESCRITO POR UM MÉDICO.
AVISO: NÃO DEVE SER UTILIZADO NA PRESENÇA
DE ANESTÉSICOS INFLAMÁVEIS.
AVISO: TAL COMO QUALQUER OUTRO APARELHOELÉTRICO, O UTILIZADOR PODE EXPERIENCIAR
PERÍODOS DE NÃO FUNCIONAMENTO, COMORESULTADO DE UMA INTERRUPÇÃO NA
ALIMENTAÇÃO DE ENERGIA ELÉTRICA, OU A
NECESSIDADE DE PROCEDER À MANUTENÇÃO DO
COMPANION 5 POR UM TÉCNICO QUALIFICADO. O
COMPANION 5 NÃO É ADEQUADO PARA DOENTESQUE POSSAM VIR A SOFRER CONSEQUÊNCIASADVERSAS PARA A SAÚDE COMO RESULTADO DE
TAIS INTERRUPÇÕES TEMPORÁRIAS.

Diretrizes de Segurança
AVISO: É PROIBIDA QUALQUER
MODIFICAÇÃO A ESTE EQUIPAMENTO.
AVISO: O FABRICANTE RECOMENDA UMA FONTE
ALTERNATIVA DE OXIGÉNIO SUPLEMENTAR NA
EVENTUALIDADE DE UMA FALHA DE ENERGIA,CONDIÇÃO DE ALARME OU AVARIA MECÂNICA.
CONSULTE O SEU MÉDICO OU FORNECEDOR DO
EQUIPAMENTO PARA INFORMAÇÕES SOBRE O TIPO
DE SISTEMA DE RESERVA REQUERIDO.
AVISO: O DISPOSITIVO FORNECE OXIGÉNIO DE
ELEVADA CONCENTRAÇÃO QUE PROMOVE UMACOMBUSTÃO RÁPIDA. NÃO PERMITA QUE SE
FUME OU A PRESENÇA DE CHAMAS A MENOS
DE DOIS (2) METROS DESTE (1) DISPOSITIVO, OU
DE (2) QUALQUER ACESSÓRIO PORTADOR DEOXIGÉNIO. O NÃO CUMPRIMENTO DESTE AVISOPODE ORIGINAR UM INCÊNDIO GRAVE, DANOSMATERIAIS E/OU PROVOCAR LESÕES FÍSICAS OU
MORTE.

PN 15062781-C4 J | Manual do utilizador

91 - POR

Companion 5
AVISO: NÃO UTILIZE A UNIDADE NUM ESPAÇO
RESTRITO OU CONFINADO (OU SEJA, NUMACAIXA PEQUENA OU NUMA BOLSA) ONDE A
VENTILAÇÃO POSSA FICAR LIMITADA. ESTA AÇÃOPODE PROVOCAR O SOBREAQUECIMENTO DO
CONCENTRADOR DE OXIGÉNIO E PREJUDICAR OSEU DESEMPENHO.
AVISO: O CONCENTRADOR DEVESER COLOCADO NUM LOCAL ONDESEJA EVITADA A ENTRADA DE FUMO,POLUENTES OU VAPORES.
AVISO: A UTILIZAÇÃO DE ALGUNS ACESSÓRIOS
DE ADMINISTRAÇÃO DE OXIGÉNIO, NÃO
ESPECIFICADOS PARA ESTE CONCENTRADOR, PODEPREJUDICAR O SEU DESEMPENHO. OS ACESSÓRIOS
RECOMENDADOS ESTÃO MENCIONADOS NESTEMANUAL.
AVISO: SE O CONCENTRADOR DE OXIGÉNIO TIVERCAÍDO, SIDO DANIFICADO POR OU EXPOSTO
A ÁGUA, CONTACTE O SEU PRESTADOR DE
CUIDADOS AO DOMICÍLIO PARA UMA INSPEÇÃOOU POSSÍVEL REPARAÇÃO DO DISPOSITIVO. NÃO
USE O CONCENTRADOR DE OXIGÉNIO CASO ESTEAPRESENTE O CABO DE ALIMENTAÇÃO OU FICHADANIFICADOS.
AVISO: NÃO PERMITA QUE A ENTRADA E A SAÍDA
DE AR FIQUEM BLOQUEADAS. NÃO DEIXE CAIRNEM INSIRA QUAISQUER OBJETOS NAS ABERTURAS
DO DISPOSITIVO. ESTA AÇÃO PODE PROVOCARO SOBREAQUECIMENTO DO CONCENTRADOR DEOXIGÉNIO E PREJUDICAR O SEU DESEMPENHO.
AVISO: NÃO ENCHA EXCESSIVAMENTE OHUMIDIFICADOR OPCIONAL. ENCHA OHUMIDIFICADOR OPCIONAL COM ÁGUA APENASATÉ AO NÍVEL INDICADO PELO FABRICANTE DO
HUMIDIFICADOR.
AVISO: NÃO UTILIZE ESTE EQUIPAMENTO SEM
PRIMEIRO LER E COMPREENDER ESTE MANUAL.
CASO NÃO PERCEBA OS AVISOS E AS INSTRUÇÕES,CONTACTE O SEU FORNECEDOR DO EQUIPAMENTOANTES DE TENTAR USÁ-LO; CASO CONTRÁRIO,PODEM OCORRER LESÕES OU DANOS.
AVISO: SE SENTIR ALGUM DESCONFORTO OU SEENCONTRAR NUMA SITUAÇÃO DE EMERGÊNCIA
MÉDICA, PROCURE ASSISTÊNCIA MÉDICA DE
IMEDIATO.
AVISO: UTILIZAR O CONCENTRADOR DE OXIGÉNIOFORA DOS LIMITES DE TEMPERATURA DEFUNCIONAMENTO PODE LIMITAR A CAPACIDADE
DO CONCENTRADOR DE CUMPRIR COM A
ESPECIFICAÇÃO DA CONCENTRAÇÃO DE OXIGÉNIO.
CONSULTE A SECÇÃO RELATIVA A ESPECIFICAÇÕES
DESTE MANUAL PARA CONSULTAR OS LIMITES DETEMPERATURA.

92 - PORPN 15062781-C4 J | Manual do utilizador

AVISO: NÃO UTILIZE LUBRIFICANTES, GORDURAS,
PRODUTOS DERIVADOS DO PETRÓLEO OUINFLAMÁVEIS COM OS ACESSÓRIOS PORTADORESDE OXIGÉNIO, OU COM O CONCENTRADOR DEOXIGÉNIO. DEVEM SER USADAS APENAS LOÇÕESOU UNGUENTOS À BASE DE ÁGUA, COMPATÍVEISCOM OXIGÉNIO. O OXIGÉNIO ACELERA ACOMBUSTÃO DE SUBSTÂNCIAS INFLAMÁVEIS.
AVISO: O CONCENTRADOR DE OXIGÉNIO NÃO
DEVE SER ENCOSTADO OU EMPILHADO COM
OUTROS DISPOSITIVOS. SE NÃO FOR POSSÍVELEVITAR ENCOSTAR OU EMPILHAR O DISPOSITIVO,
ESTE DEVE SER SUPERVISIONADO A FIM DEGARANTIR O SEU NORMAL FUNCIONAMENTO.

AVISO: O FABRICANTE RECOMENDA
UMA FONTE ALTERNATIVA DE OXIGÉNIOSUPLEMENTAR NA EVENTUALIDADE
DE UMA FALHA DE ENERGIA,CONDIÇÃO DE ALARME OU AVARIA
MECÂNICA. CONSULTE O SEU MÉDICOOU FORNECEDOR DO EQUIPAMENTOPARA INFORMAÇÕES SOBRE O TIPO DE
SISTEMA DE RESERVA REQUERIDO.
AVISO: O DISPOSITIVO FORNECE OXIGÉNIO DE
ELEVADA CONCENTRAÇÃO QUE PROMOVE UMACOMBUSTÃO RÁPIDA. NÃO AUTORIZE QUE SE
FUME OU QUE SE FAÇAM CHAMAS DENTRO DAMESMA DIVISÃO (1) DESTE DISPOSITIVO OU (2) DEQUALQUER ACESSÓRIO PORTADOR DE OXIGÉNIO. O
NÃO CUMPRIMENTO DESTE AVISO PODE ORIGINAR
UM INCÊNDIO GRAVE, DANOS MATERIAIS E/OU
PROVOCAR LESÕES FÍSICAS OU MORTE.
AVISO: A UTILIZAÇÃO DE ALGUNS ACESSÓRIOS
DE ADMINISTRAÇÃO DE OXIGÉNIO, NÃO
ESPECIFICADOS PARA ESTE CONCENTRADOR, PODEPREJUDICAR O SEU DESEMPENHO. OS ACESSÓRIOS
RECOMENDADOS ESTÃO MENCIONADOS NESTEMANUAL.
AVISO: SE O CONCENTRADOR DE OXIGÉNIO TIVERCAÍDO, SIDO DANIFICADO POR OU EXPOSTO
A ÁGUA, CONTACTE O SEU PRESTADOR DE
CUIDADOS AO DOMICÍLIO PARA UMA INSPEÇÃOOU POSSÍVEL REPARAÇÃO DO DISPOSITIVO. NÃO
USE O CONCENTRADOR DE OXIGÉNIO CASO ESTEAPRESENTE O CABO DE ALIMENTAÇÃO OU FICHADANIFICADOS.
AVISO: USE APENAS PEÇAS SOBRESSELENTESRECOMENDADAS PELO FABRICANTE PARAGARANTIR UM FUNCIONAMENTO CORRETO EEVITAR O RISCO DE INCÊNDIO E DE QUEIMADURAS.
AVISO: NÃO LUBRIFIQUE ENCAIXES, LIGAÇÕES,TUBAGENS OU OUTROS ACESSÓRIOS DO
CONCENTRADOR DE OXIGÉNIO PARA EVITAR O
RISCO DE INCÊNDIO E DE QUEIMADURAS.

Companion 5
AVISO: FUMAR DURANTE A UTILIZAÇÃODE OXIGÉNIO É A PRIMEIRA CAUSA DELESÕES PROVOCADAS POR FOGO E MORTESRELACIONADAS. DEVERÁ CUMPRIR OS SEGUINTESAVISOS DE SEGURANÇA:AVISO: NÃO AUTORIZE QUE SE FUME, QUE SE
ACENDAM VELAS, OU SE FAÇAM CHAMASNA MESMA DIVISÃO ONDE SE ENCONTRA ODISPOSITIVO OU OS ACESSÓRIOS PORTADORES DEOXIGÉNIO.
AVISO: FUMAR DURANTE A UTILIZAÇÃO DE
UMA CÂNULA DE OXIGÉNIO PODE PROVOCARQUEIMADURAS FACIAIS E POSSIVELMENTE A
MORTE.
AVISO: CASO SEJA FUMADOR, OS 3 PASSOSSEGUINTES PODEM SALVAR-LHE A VIDA: DESLIGUEO CONCENTRADOR DE OXIGÉNIO, RETIRE
A CÂNULA E SAIA DA DIVISÃO NA QUAL SE
ENCONTRA O DISPOSITIVO.
AVISO: OS DÍSTICOS “PROIBIDO FUMAR –OXIGÉNIO EM UTILIZAÇÃO” DEVEM ESTARBEM VISÍVEIS NA CASA OU NO LOCAL ONDE OOXIGÉNIO ESTÁ A SER UTILIZADO. OS DOENTES EOS SEUS PRESTADORES DE CUIDADOS DEVEM SER
INFORMADOS ACERCA DOS PERIGOS DE FUMAR
NA PRESENÇA DE OU DURANTE A UTILIZAÇÃO DEOXIGÉNIO MEDICINAL.
AVISO: NÃO UTILIZE O CONCENTRADOR DEOXIGÉNIO NA PRESENÇA DE GASES INFLAMÁVEIS,SOB RISCO DE UMA COMBUSTÃO RÁPIDA PODERPROVOCAR DANOS MATERIAIS, LESÕES CORPORAISOU MORTE.
AVISO: REMOVER A CÂNULA E COLOCÁ-LA NO
VESTUÁRIO, ROUPA DE CAMA, SOFÁS OU OUTRO
MATERIAL ALMOFADADO IRÁ DEFLAGRAR UMINCÊNDIO QUANDO EXPOSTOS A UM CIGARRO,FONTE DE CALOR OU CHAMA.
AVISO: NUNCA DEIXE A CÂNULA NASAL EM
CIMA/BAIXO DE PEÇAS DE ROUPA, COLCHAS OU
ALMOFADAS DE CADEIRAS. SE A UNIDADE ESTIVER
LIGADA MAS NÃO ESTIVER EM UTILIZAÇÃO, O
OXIGÉNIO TORNARÁ O MATERIAL INFLAMÁVEL.
CONFIGURE O INTERRUPTOR DE ALIMENTAÇÃO I/0
PARA A POSIÇÃO 0 (DESLIGADO) SEMPRE QUE OCONCENTRADOR DE OXIGÉNIO NÃO ESTIVER EMUTILIZAÇÃO.

AVISO: UTILIZE APENAS ACESSÓRIOS
RECOMENDADOS PELO FABRICANTE. A UTILIZAÇÃODE OUTROS ACESSÓRIOS PODE SER PERIGOSA,CAUSAR DANOS GRAVES NO SEU CONCENTRADORDE OXIGÉNIO E ANULAR A GARANTIA.
ATENÇÃO: Desligue sempre a fonte dealimentação CA da parede antes dedesligar a fonte de alimentação CA doconcentrador de oxigénio.
ATENÇÃO: Coloque sempre os tubos deadministração de oxigénio e os cabos dealimentação de forma a evitar o perigo de quedas.
ATENÇÃO: Para evitar a anulação dagarantia, siga todas as instruções dos
fabricantes.
Nota: O equipamento de comunicação RF portátil ou
móvel pode afetar o equipamento elétrico médico.
AVISO: SEMPRE COLOQUE O TUBO DE FORNECIMENTO DE OXIGÊNIO E OS CABOS DE ALIMENTAÇÃO DE FORMA DE FORMA QUE IMPEDE OPERIGO DE VIAGEM OU O POSSÍVEL ESTRANGULAMENTO ACIDENTAL.
AVISO: ESTE PRODUTO PODE EXPÔ-LO A PRODUTOS
QUÍMICOS, INCLUINDO NÍQUEL, CONSIDERADO
COMO AGENTE CAUSADOR DE CANCRO PELO ESTADO DA CALIFÓRNIA. PARA MAIS INFORMAÇÕES,VISITE WWW.P65WARNINGS.CA.GOV.
AVISO: NO CASO DE OCORRER UM INCIDENTESÉRIO COM ESTE DISPOSITIVO, O USUÁRIO DEVE
RELATAR IMEDIATAMENTE O INCIDENTE AO FORNECEDOR E / OU AO FABRICANTE. UM INCIDENTEGRAVE É DEFINIDO COMO UMA LESÃO, MORTEOU POTENCIAL DE CAUSAR LESÃO / MORTE, CASOOCORRA UMA REINCIDÊNCIA. O USUÁRIO TAMBÉMPODE RELATAR O INCIDENTE À AUTORIDADE COMPETENTE DO PAÍS EM QUE O INCIDENTE OCORREU.

AVISO: NÃO PROCEDA A QUALQUER
TIPO DE MANUTENÇÃO PARA ALÉM DASSOLUÇÕES POSSÍVEIS APRESENTADASNESTE MANUAL. NÃO RETIRE AS
TAMPAS. APENAS O SEU FORNECEDORDO EQUIPAMENTO OU UM TÉCNICO DE
MANUTENÇÃO QUALIFICADO PODERÃOREMOVER AS TAMPAS OU PROCEDER À
MANUTENÇÃO DA UNIDADE.

PN 15062781-C4 J | Manual do utilizador

93 - POR

Companion 5
Especificações
Para garantir a correta utilização do equipamento,a tabela seguinte fornece informações importantes
acerca dos ambientes ou condições de funcionamento
recomendados.

Especificações
Taxas de fluxo
Concentração
de O2
Dimensões

0,5–5,0 LPM
90% (+5,5%/-3%)

54,6x31,8x34,2 cm(21,5X12,5x13,5”)
Peso
16,3 kg (36 lbs)
Nível de Pressão 50 dB(A) a 2,0 LPM
sonoraConsumo
285 W a 2 LPM, 350 W Máximo
energético
Temperatura de 5 °C a 40 °C (41 ºF a 104 ºF)
funcionamento
Humidade de
15% - 90% a um ponto de
funcionamento condensação de 28 °C (82,4 °F)
Temperatura de -25 ºC a 70 ºC (-13 ºF a 158 ºF)
armazenamento
Humidade de
0% - 90% sem condensação
armazenamento
ElétricaNão utilize extensões. Não utilize
tomadas elétricas controladas por
um interruptor.
Altitude de-382 - 3011 metros
funcionamento (-1253 - 9879 pés)Tempo deaté 24 horas por dia
funcionamento
A vida útil esperada deste equipamento é de, nomínimo, cinco anos.
AVISO: O USO DE DISPOSITIVO FORA
DE CONDIÇÕES OPERACIONAISESPECÍFICAS É ESPERADO PARA AFETARADVERSAMENTE A TAXA DE FLUXOE A PERCENTAGEM DE OXIGÊNIO ECONSEQÜENTEMENTE A QUALIDADE DATERAPIA.
Nota: Se o Concentrador de Oxigénio estiver
armazenado por um longo período de tempo
fora do seu intervalo normal de temperaturade funcionamento, a unidade deve voltar à sua
temperatura de funcionamento normal antes de serligada. (Consulte a secção de Especificações constantedeste manual.)

Posicionamento correto do Companion 5
94 - PORPN 15062781-C4 J | Manual do utilizador

Selecione uma localização para o aparelho onde seja
evitada a entrada de fumo, vapores e poluentes. O
posicionamento correto do dispositivo deverá permitir
a entrada de ar através dos três locais de entrada dear, na parte superior, traseira e inferior da caixa edeve permitir também a livre saída de ar pela saída deexaustão na parte inferior esquerda do dispositivo.
As entradas de ar estão localizadas na parte superior
traseira e inferior do concentrador. Posicione oaparelho a um mínimo de 30 cm de distância deparedes, mobiliário, cortinas ou outros obstáculos.
Posicione o equipamento de forma a que a os alarmes
sejam audíveis.
Posicione o tubo de fornecimento de oxigénio de modoa não ficar torcido ou obstruído.
AVISO: NÃO UTILIZE O CONCENTRADORDE OXIGÉNIO NA PRESENÇA DE
GASES INFLAMÁVEIS, SOB RISCO DE
UMA COMBUSTÃO RÁPIDA PODERPROVOCAR DANOS MATERIAIS,LESÕES CORPORAIS OU MORTE. NÃO
UTILIZE LUBRIFICANTES, GORDURAS,
PRODUTOS DERIVADOS DO PETRÓLEOOU INFLAMÁVEIS COM OS ACESSÓRIOS
PORTADORES DE OXIGÉNIO, OU COM
O CONCENTRADOR DE OXIGÉNIO.
DEVEM SER USADAS APENAS LOÇÕESOU UNGUENTOS À BASE DE ÁGUA,COMPATÍVEIS COM OXIGÉNIO. O
OXIGÉNIO ACELERA A COMBUSTÃO DE
SUBSTÂNCIAS INFLAMÁVEIS.
AVISO: O DISPOSITIVO FORNECE OXIGÉNIO DE
ELEVADA CONCENTRAÇÃO QUE PROMOVE UMACOMBUSTÃO RÁPIDA. NÃO AUTORIZE QUE SE
FUME OU QUE SE FAÇAM CHAMAS DENTRO DAMESMA DIVISÃO (1) DESTE DISPOSITIVO OU (2) DEQUALQUER ACESSÓRIO PORTADOR DE OXIGÉNIO. O
NÃO CUMPRIMENTO DESTE AVISO PODE ORIGINAR
UM INCÊNDIO GRAVE, DANOS MATERIAIS E/OU
PROVOCAR LESÕES FÍSICAS OU MORTE.

Companion 5
Instruções de utilização
Antes de utilizar

Este manual de instruções serve como referência para oajudar no funcionamento e manutenção do aparelho. Se
tiver dúvidas ou preocupações, entre em contacto com
o seu prestador de cuidados domiciliários.
Importante! NÃO tente operar o Companion 5 semprimeiro ler a secção de Diretivas de Segurança deste
manual. Siga todas as instruções de funcionamento.
Tenha em consideração todos os Avisos no aparelho e
no Manual de Instruções. De forma a reduzir o risco
de incêndio, ferimentos ou danos materiais graves
no Companion 5, tenha em consideração todas asprecauções de segurança.
O Companion 5 CAIRE não cria o seu próprio
oxigénio. Ao invés, produz oxigénio altamente
concentrado a partir do ar presente na divisão efornece-o a si. Uma vez que está naturalmente presente,o ar da divisão contém apenas aproximadamente 21%de oxigénio por volume.
O ar é levado para o interior do concentrador atravésde um compressor. O ar passa então por uma série defiltros e um filtro molecular que adsorve quimicamentemoléculas de azoto. Em resultado, o ar enriquecidode oxigénio que sai do concentrador é administrado a
87-95% por volume.
AVISO: LIMPE A CAIXA, O PAINEL DE
CONTROLO E O CABO DE ALIMENTAÇÃO
COM UM DETERGENTE SUAVE APLICADOCOM UM PANO OU ESPONJA HÚMIDOS(NÃO MOLHADOS) E DEPOIS SEQUE
BEM TODAS AS SUPERFÍCIES. NÃO
PERMITA QUE NENHUM LÍQUIDO ENTRE
PARA O INTERIOR DO DISPOSITIVO.
PRESTE ESPECIAL ATENÇÃO À SAÍDA DEOXIGÉNIO PARA A LIGAÇÃO DA CÂNULA
A FIM DE GARANTIR QUE SE MANTÉM
LIVRE DE PÓ, ÁGUA E PARTÍCULAS.
ATENÇÃO: Não permita que a entrada ea saída de ar fiquem bloqueadas. Estaação pode provocar o sobreaquecimento
do Concentrador de Oxigénio e
prejudicar o seu desempenho. Não
opere o Concentrador de Oxigénio sem
que o filtro de admissão de ar estejacorretamente posicionado. Se forfornecido um segundo filtro, introduza ofiltro de “substituição” antes de limpar
o filtro sujo. Limpe o filtro sujo numa
solução tépida de água e sabão e depois
seque muito bem antes de voltar a usar.

Passo 1: Posicionamento do Companion 5 para
utilização
Posicione o Companion 5 numa área bem ventilada
e iluminada. Certifique-se de que a entrada de ar e a
saída de exaustão não estão obstruídas. Posicione oCompanion 5 de forma a que todos os indicadoresvisuais e sonoros possam ser facilmente vistos e
ouvidos.
Ligue o dispositivo à tomada elétrica CA.
Passo 2: Se a Garrafa humidificadora NÃO FOR
utilizada

Localização
da entrada
de arLocalização
da entrada
de ar derefrigeração

Saída deexaustão
Localização daentrada de ara. Ligue uma cânula nasal ou tubo de oxigénio
diretamente à saída estriada de oxigénio.

b. Continue para o passo 4.
Passo 3: Se a Garrafa humidificadora FOR utilizada
a. Desenrosque a tampa da garrafa humidificadora.
b. Encha a garrafa com água destilada. Certifique-sede que o nível de água se encontra dentro dos limites
mínimo e máximo, visíveis na garrafa, especificados
pelo fabricante.
c. Fixe a tampa da garrafa humidificadora. Certifique-se de que não existem fugas.
d. Ligue a cânula nasal ou tubo de oxigénio padrão à
saída estriada na garrafa humidificadora.
Nota: A utilização de um humidificador não
especificado para utilização com este concentrador
pode afetar o seu desempenho.

PN 15062781-C4 J | Manual do utilizador

95 - POR

Companion 5
g. Coloque a garrafa no local destinado no Suporte da
Garrafa Humidificadora.
h. Fixe a garrafa com a banda elástica no suporte.

e. Ligue a extremidade roscada do tubo humidificador
à tampa da garrafa humidificadora.

i. Continue para o passo 4.
Passo 4: Arranque e Aquecimento
a. Ligue o cabo de alimentação do Companion 5
CAIRE à tomada CA.
AVISO: INSPECIONE O CABO ELÉTRICOPARA DETETAR EVENTUAIS DANOSANTES DA UTILIZAÇÃO. SE O CABO
ESTIVER DANIFICADO, NÃO O LIGUEA QUALQUER TOMADA NEM TENTECOLOCAR O CONCENTRADOR EM
FUNCIONAMENTO.

f. Ligue a outra extremidade do tubo humidificador à
saída estriada do Companion 5 da CAIRE.

b. Prima o interruptor de alimentação para aposição “ON” (I). Quando o Companion 5 estiver
corretamente ligado, o indicador verde no ecrãLED ficará iluminado. Todas as luzes LED ficarão
iluminadas durante o arranque. Após o concentrador
concluir o ciclo de aquecimento, apenas a luz verde
se manterá acesa.
Nota: Após LIGAR o aparelho, aguarde até dez(10) minutos para que o aparelho possa alcançar
as suas especificações de desempenho.

96 - PORPN 15062781-C4 J | Manual do utilizador

Companion 5
AVISO: É MUITO IMPORTANTESELECIONAR APENAS O NÍVEL DEOXIGÉNIO PRESCRITO. NÃO ALTERE
A SELEÇÃO DO FLUXO, EXCETO SE
TIVER RECEBIDO INDICAÇÕES NESSESENTIDO POR PARTE DE UM MÉDICOQUALIFICADO. O CONCENTRADOR DEOXIGÉNIO PODE SER USADO DURANTEO SONO SOB RECOMENDAÇÃO DE UM
MÉDICO QUALIFICADO.
Passo 5: Ajuste a Taxa de Controlo do Fluxo

Passo 6: Verifique o fluxo e respire normalmentea. Rode o botão de controlo do fluxo para selecionar ofluxo de oxigénio (LPM) prescrito pelo seu médico.

a. Verifique se o oxigénio está efetivamente a fluir
através da cânula nasal e de que não existem vincos,
dobras ou bloqueios no tubo.
Nota: Certifique-se de que a cânula está totalmente
inserida e fixa. Deverá conseguir ouvir ou sentir ofluxo de oxigénio para os tubos da cânula nasal.
b. Posicione corretamente a sua cânula nasal e respire
normalmente através da mesma.
Passo 7: Encerramento do Sistema
a. Prima o interruptor de alimentação para a posição
“OFF” (O) quando o Companion 5 CAIRE já não seencontrar em utilização.

Para ajustar a taxa do fluxo:
Rode no sentido inverso ao dos ponteiros do relógio
para aumentar o fluxo.
Rode no sentido dos ponteiros do relógio para diminuiro fluxo.
O centro da bola indica a taxa de fluxo. A imagem
abaixo indica uma taxa de fluxo de 4,0 LPM.

ATENÇÃO: Utilize apenas acessórios
recomendados pelo fabricante. A
utilização de outros acessórios podeser perigosa, causar danos graves noseu concentrador de oxigénio e anulara garantia.
ATENÇÃO: Não utilize extensões com esta unidadeou ligue várias fichas à mesma tomada elétrica. A
utilização de extensões pode afetar negativamenteo desempenho do dispositivo. A existência dedemasiadas fichas numa tomada pode resultar
numa sobrecarga do quadro elétrico, levando àativação do disjuntor/fusível ou a um incêndio casoo disjuntor ou o fusível deixem de funcionar.
ATENÇÃO: A utilização de cabos e adaptadores
que não os especificados, à exceção dos cabose adaptadores vendidos pelo fabricante do
equipamento enquanto peças de substituição para
componentes internos, pode resultar no aumento
das emissões ou numa diminuição da imunidade
do Concentrador de Oxigénio.

CorretoPN 15062781-C4 J | Manual do utilizador

97 - POR

Companion 5
AVISO: NÃO PERMITA QUE AENTRADA E A SAÍDA DE AR FIQUEMBLOQUEADAS. ESTA AÇÃO PODEPROVOCAR O SOBREAQUECIMENTO
DO CONCENTRADOR DE OXIGÉNIO E
PREJUDICAR O SEU DESEMPENHO.
Nota: Para o fornecedor do equipamento: Os
seguintes acessórios de administração de oxigênio são
recomendados para uso com o Companion 5:• Cânula nasal: Número de peça CAIRE CU002-1• Tubulação do adaptador do umidificador: CAIRE
Número da peça 20843882• Frasco do umidificador: CAIRE Número da peça
HU003-1• Firebreak: Número de peça CAIRE 20629671
Um quebra de fogo é necessário para uso com
qualquer cânula.
• O CAIRE oferece um combate a incêndio destinado
a ser usado em conjunto com o concentrador deoxigênio. O quebra-fogo é um fusível térmico parainterromper o fluxo de gás no caso de a cânula ajusante ou a tubulação de oxigênio ser inflamada equeimar no fogo. É colocado em linha com a cânula
nasal ou tubo de oxigênio entre o paciente e a saídade oxigênio do Companion 5. Para o uso adequado
do combate a incêndio, sempre consulte as instruções
do fabricante (incluídas em cada kit de combate aincêndio).
• Para quaisquer acessórios adicionais recomendados,
consulte o Catálogo de acessórios (PN MLLOX0010)
disponível em www.caireinc.com.

98 - PORPN 15062781-C4 J | Manual do utilizador

Companion 5
Condições de Alarme
O Companion 5 CAIRE usa uma combinação de duas
ou três (para modelos com OCSI) luzes LED e um
alarme sonoro para alertá-lo de que existe uma avaria
no seu equipamento.
1) LED verde () - indica funcionamento normal.

2) LED vermelho ( ) - indica avaria no sistema.
Ver Tabela de Condições de Alarme para mais
informações.
3) LED amarelo ( ) - indica baixa concentração deoxigénio. Ver Tabela de Condições de Alarme paramais informações.
A Tabela de Condições de Alarme na página seguinte
descreve as diferentes condições de alarme que podem
ser apresentadas no concentrador. Sublinha as possíveis
causas e as ações que deve tomar se o alarme soar.
Em caso de alarme, recomendamos contactar o seu
prestador de cuidados de saúde, assim que possível.

Verde

Vermelho

Amarelo

Em cima: Companion 5

AVISO: NÃO IGNORE OS ALARMESAVISOS: O FABRICANTE RECOMENDA
UMA FONTE ALTERNATIVA DE OXIGÉNIOSUPLEMENTAR NA EVENTUALIDADE
DE UMA FALHA DE ENERGIA,CONDIÇÃO DE ALARME OU AVARIA
MECÂNICA. CONSULTE O SEU MÉDICOOU FORNECEDOR DO EQUIPAMENTOPARA INFORMAÇÕES SOBRE O TIPO DE
SISTEMA DE RESERVA REQUERIDO.

PN 15062781-C4 J | Manual do utilizador

99 - POR

Companion 5
Tabela de Condições de Alarme
AlarmeSonoroCor do
LED

Causa

Ação
NenhumO concentrador está afuncionar corretamente.

Desligado
VerdeIntermitente
VERMELHOIntermitente

O concentrador
sofreu uma quebra na
alimentação mas ointerruptor de alimentação
continua na posição “ON”.

Verifique que o cabo elétrico está ligado a uma
tomada e de que a tomada tem energia elétrica. Tente
ligar numa tomada diferente. Se o problema persistir,
ligue a uma fonte de oxigénio de reserva e contacte oseu prestador de cuidados de saúde.

Avaria no sistema

Prima o interruptor de alimentação de imediato paraa posição “OFF”. Desligue a alimentação da tomada
de parede. Espere 5 minutos. Volte a ligar o cabode alimentação à tomada elétrica e volte a ligar oconcentrador. Se o alarme continuar, o equipamento
precisa de assistência. Ligue a uma fonte de oxigéniode reserva e contacte de imediato o seu prestador decuidados de saúde.

Taxa de fluxo demasiado
elevada ou demasiado
baixa

1. Certifique-se de que a cânula não está dobrada ou
obstruída. Se usado com uma garrafa humidificadora,
certifique-se de que esta está corretamente cheia enão provoca um bloqueio.
2. Certifique-se de que o Companion 5 tem aventilação adequada. Este deve estar, no mínimo, a
30 cm de qualquer superfície, de forma a garantir que
as saídas de ventilação não ficam obstruídas.
3. Se o problema persistir, mude para uma fonte
alternativa de oxigénio e contacte o seu prestador decuidados de saúde para obter assistência.

VERMELHOIntermitente
VERMELHOO Companion 5 detetou
níveis baixos de oxigénio.

Intermitente
Amarelo

O Companion 5 detetou
níveis baixos de oxigénio.

Intermitente
Amarelo

100 - PORPN 15062781-C4 J | Manual do utilizador

1. Certifique-se de que o filtro de admissão de ar e as
saídas de exaustão não se encontram obstruídos ou
restringidos. 2. Certifique-se de que o Companion 5
se encontra numa área bem ventilada. Certifique-sede que existe uma distância mínima de 30 cm entre aparte posterior e as laterais do Companion e quaisquerpotenciais obstruções (móveis, cortinas, etc.)
3. Se a situação persistir, mude para uma fonte
alternativa de oxigénio e contacte de imediato o seu
prestador de cuidados de saúde.

Companion 5
Resolução de Problemas
Recomendamos que contacte o seu prestador decuidados de saúde se o seu concentrador emitir
qualquer alarme ou se sentir que não está afuncionar adequadamente. Não deve tentar reparar oconcentrador sozinho. A reparação deve ser realizada
apenas por um técnico de manutenção qualificado.

Contudo, existem alguns passos de resolução deproblemas que pode seguir caso tiver algum problema
com o seu equipamento Companion 5 CAIRE. Estão
destacados na tabela seguinte. Pode usar esta tabela
seguindo as verificações do seu problema por ordem
numérica.

Tabela de Resolução de Problemas para o Utilizador
ProblemaO concentrador
está ligado mas
não funciona.

Verifica-se uma
taxa de fluxode oxigénio
baixa ouinexistente.

Causa PossívelAção

1. O cabo elétrico não está ligado à
tomada.

1. Ligue o cabo à tomada.

2. A tomada não está a passar energia
elétrica.

2. Verifique o quadro e o circuito elétrico de sua casa.
Tente ligar numa tomada diferente.

3. Avaria interna.

3. Ligue a uma fonte de oxigénio de reserva. Contacte deimediato o seu prestador de cuidados de saúde.

1. A definição de controlo do fluxo foi
alterada.

1. Verifique se o botão de controlo do fluxo estácorretamente posicionado na sua definição de Litro Por
Minuto (LPM) prescrita pela seu médico.

2. O tubo foi desconectado.

2. Verifique se o tubo está bem fixo em todas as ligações
(saída estriada, garrafa humidificadora, reservatório deágua, etc). Se necessário, volte a fixar as ligações.

3. O tubo ou cânula de oxigénio está
dobrado.

3. Verifique se existem dobras ou bloqueios no tubo. Se
necessário, proceda à reparação.

4. Existe uma fuga na cânula ou tubo.

4. Inspecione todos os tubos e a cânula a fim de detetar
eventuais fugas. Poderá fazê-lo ouvindo ou sentindo o ara sair. Substitua o tubo ou a cânula com fuga.

5. O fluxo de ar no concentrador é
impedido.

5. Certifique-se de que não existe nada a bloquear as
entradas do concentrador. Certifique-se também de que
se encontra a um mínimo de 30 cm (12 polegadas) dedistância de mobiliário, paredes ou outros obstáculos.

6. Avaria interna.

6. Ligue a uma fonte de oxigénio de reserva. Contacte deimediato o seu prestador de cuidados de saúde.

Nota: Se o seu Companion 5 continuar a emitir alarme após utilizar as tabelas das Condições de Alarme e da
Resolução de Problemas, contacte a Assistência Técnica através do nº. 1-800-482-2473 e forneça o código do alarmeapresentado no ecrã LCD.

PN 15062781-C4 J | Manual do utilizador

101 - POR

Companion 5
Limpeza, Conservação e Manutenção de Rotina
Manutenção de Rotina

A manutenção de rotina consiste em substituir o filtrode partículas volumosas da pega uma vez por semana(se estiver instalado).

Conservação e Limpeza do Aparelho pelo
Utilizador
Substituição da Cânula
Substitua o seu tubo de alimentação e a cânula com
regularidade, conforme indicado pelo seu prestador decuidados domiciliários. O seu médico ou prestador decuidados domiciliários irá fornecer-lhe as informações
de limpeza, desinfeção e substituição.
Nota: Siga sempre as instruções do fabricante dacânula para uma utilização correta. Substitua a cânula
descartável, conforme recomendado pelo fabricanteou fornecedor do equipamento. Estão disponíveis
outros acessórios através do Fornecedor do seu
equipamento.

Filtro de admissão de ar (Cartucho)
O Filtro de admissão de ar só deve ser acedido ou
substituído pelo seu prestador de cuidados de saúde.
Filtro de partículas volumosas (Opcional):

O filtro de partículas brutas está localizado na parteinferior frontal do Companion 5.

Filtro departículas
volumosas:Você deve inspecionar e limpar uma vez por semana. Para limpar o filtro, use o seguinte processo:
1) Remova o filtro de partículas volumosas.
2) Lave o filtro em água corrente morna, utilizando
uma solução de detergente moderada.
3) Enxague bem o filtro com água corrente limpa eremova a água em excesso.
4) Deixe o filtro secar ao ar.
5) Volte a colocar o filtro na caixa.
102 - PORPN 15062781-C4 J | Manual do utilizador

Nota: O fabricante não recomenda a esterilização
deste equipamento.
Nota: Não opere o Concentrador de Oxigénio sem
que o filtro de admissão de ar esteja corretamente
posicionado.
Garrafa humidificadora (se aplicável)Deverá inspecionar a sua garrafa humidificadoradiariamente a fim de garantir que o nível de águase encontra dentro dos limites mínimo e máximo
especificados. Volte a encher a garrafa com águadestilada, conforme necessário. Limpe a garrafa
humidificadora de acordo com as instruções dofabricante.
Procedimento de limpeza:
DESLIGUE o Companion 5 e desligue-o daalimentação CA antes de qualquer atividade de limpezaou desinfeção. NÃO pulverize diretamente a caixa
exterior. Utilize um pano ou esponja húmidos (nãomolhados). Pulverize o pano ou a esponja com uma
solução de detergente suave para limpar a caixa e ocabo de alimentação. Para desinfetar o Companion 5,
utilize o desinfetante Lysol® Brand II (ou equivalente).
Dilua o produto de limpeza de acordo com o indicado
pelo fabricante do produto, mas não pulverize
diretamente sobre o Companion 5.
AVISO: PERIGO DE CHOQUE ELÉTRICO.
DESLIGUE (OFF) A UNIDADE E RETIREO CABO DE ALIMENTAÇÃO DA
TOMADA ELÉTRICA ANTES DE LIMPARA UNIDADE, DE MODO A EVITARCHOQUES ELÉTRICOS ACIDENTAIS E OPERIGO DE QUEIMADURAS. APENAS O
SEU FORNECEDOR DO EQUIPAMENTOOU UM TÉCNICO DE MANUTENÇÃOQUALIFICADO DEVERÃO REMOVER ASTAMPAS OU EFETUAR A MANUTENÇÃO
DA UNIDADE.
AVISO: PERIGO DE CHOQUE ELÉTRICO. DESLIGUE(OFF) A UNIDADE E RETIRE O CABO DE
ALIMENTAÇÃO DA TOMADA ELÉTRICA ANTESDE LIMPAR A UNIDADE, DE MODO A EVITARCHOQUES ELÉTRICOS ACIDENTAIS E O PERIGO DE
QUEIMADURAS. APENAS O SEU FORNECEDOR DO
EQUIPAMENTO OU UM TÉCNICO DE MANUTENÇÃOQUALIFICADO DEVERÃO REMOVER AS TAMPAS OUEFETUAR A MANUTENÇÃO DA UNIDADE.

Companion 5
AVISO: PERIGO DE CHOQUE ELÉTRICO. DESLIGUEO CABO DE ALIMENTAÇÃO DA TOMADA ELÉTRICAANTES DE LIMPAR A UNIDADE, DE MODO A EVITARCHOQUES ELÉTRICOS ACIDENTAIS E PERIGO DE
QUEIMADURAS. APENAS O SEU FORNECEDOR DO
EQUIPAMENTO OU UM TÉCNICO DE MANUTENÇÃOQUALIFICADO DEVERÃO REMOVER AS TAMPAS OUEFETUAR A MANUTENÇÃO DA UNIDADE.
AVISO: DEVE TER-SE ESPECIAL CUIDADO A FIM
DE EVITAR QUE O CONCENTRADOR DE OXIGÉNIOFIQUE MOLHADO OU QUE ENTREM LÍQUIDOS PARAO INTERIOR DA UNIDADE. ISTO PODE CAUSAR UMA
AVARIA OU PARALISAÇÃO E RESULTAR NUM MAIOR
RISCO DE CHOQUES ELÉTRICOS OU QUEIMADURAS.
AVISO: NÃO É SEGURO INTERLIGAR O COMPANION
5 COM OUTROS EQUIPAMENTOS.

AVISO: LIMPE A CAIXA, O PAINEL DE CONTROLO EO CABO DE ALIMENTAÇÃO COM UM DETERGENTESUAVE APLICADO COM UM PANO OU ESPONJAHÚMIDOS (NÃO MOLHADOS) E DEPOIS SEQUE
BEM TODAS AS SUPERFÍCIES. NÃO PERMITA QUENENHUM LÍQUIDO ENTRE PARA O INTERIOR DO
DISPOSITIVO. PRESTE ESPECIAL ATENÇÃO À SAÍDADE OXIGÉNIO PARA A LIGAÇÃO DA CÂNULA A
FIM DE GARANTIR QUE SE MANTÉM LIVRE DE PÓ,ÁGUA E PARTÍCULAS.
AVISO: NÃO UTILIZE LÍQUIDOS DIRETAMENTENA UNIDADE. A LISTA DE AGENTES QUÍMICOSINDESEJÁVEIS INCLUI ENTRE OUTROS, OS
SEGUINTES: ÁLCOOL E PRODUTOS À BASE DEÁLCOOL, PRODUTOS CONCENTRADOS À BASE DE
CLORO (CLORETO DE ETILENO) E PRODUTOS À
BASE DE ÓLEOS (PINE-SOL®, LESTOIL®). ESTES NÃO
DEVEM SER USADOS PARA LIMPAR A ESTRUTURADE PLÁSTICO DO EQUIPAMENTO, SOB RISCO DE
DANIFICAR OS PLÁSTICOS.

Eliminação
Devolva sempre o Companion 5, incluindo todos oscomponentes, ao seu prestador de cuidados domiciliários
para garantir uma eliminação adequada. Pode também
entrar em contracto com as entidades locais para obter
instruções acerca da correta eliminação da bateria.

WEEE e RoHS
Este símbolo destina-se a recordar os proprietários do equipamento de que devem entregá-lo
a uma instalação de reciclagem no final da sua vidaútil, de acordo com a Diretiva relativa aos Resíduos deEquipamentos Elétricos e Eletrónicos (REEE).

Os nossos produtos estão em conformidade com adiretiva de Restrição de Substâncias Perigosas (RoHS).
Não contêm mais do que quantidades residuais dechumbo ou de outros materiais perigosos.
CUIDADO: Para descarte adequado,
entre em contato com o fornecedor do
equipamento ou a agência governamental local para obter instruções sobre odescarte.

CAIRE e CAIRE Inc. são marcas registradas da CAIRE Inc.
Visite nosso site abaixo para obter uma lista completa de marcas
comerciais.
Marcas comerciais: www.careinc.com/corporate/trademarks.
Copyright © 2020 CAIRE Inc. A CAIRE Inc. reserva o direito
de descontinuar os seus produtos ou alterar os preços, materiais,
equipamento, qualidade, descrições, especificações e/ou processos
dos seus produtos a qualquer momento e sem aviso prévio, semquaisquer obrigações ou consequências. Reservamos todos os
direitos não expressamente aqui declarados, conforme aplicável.
PN 15062781-C4 J | Manual do utilizador

103 - POR

Companion 5
Gebruikersbediening en indicatoren van de systeemstatus
Interne symbolen02

Toegepast onderdeel van type BF (matevan bescherming tegen elektrischeschokken). Reg.-nr. 5333

Zuurstofuitvoer

I/O Aan/uit-schakelaar
Dit apparaat bevat geen onderdelen
die door de gebruiker kunnen worden
onderhouden. Maak de kap niet open.

Richtlijn 93/42/EEG van de Raad inzake medische
instrumenten
Gevolmachtigde vertegenwoordiger inde Europese Gemeenschap
IEC 60417

ISO 7000; Grafische symbolen voor gebruik op deapparatuur - Index en samenvatting
Opslag- of bedrijfstemperatuurbereik.
Reg.-nr. 0632

Apparatuur van klasse II, dubbelgeïsoleerd reg.-nr. 5172
ETL-certificering

Luchtvochtigheid bij opslag.
Reg.-nr. 2620

Agentschap voor veiligheid voor
CAN/CSA C22.2 nr. 601.1 M90 voor
elektrische medische apparatuur

Naam en adres van de fabrikant.
Reg.-nr. 3082
Productiedatum. Reg.-nr. 2497
Catalogusnummer. Reg.-nr. 2493
Serienummer. Reg.-nr. 2498
Deze kant omhoog. Reg.-nr. 0623

FCC-kennisgeving
21 CFR 801.15: Code van federale regelgeving,
titel 21
Dit product mag op grond van defederale wetgeving uitsluitend worden
verkocht door of namens een arts.
Richtlijn 2012/19/EU van de Raad: afgedankteelektrische en elektronische apparatuur (AEEA)

Breekbaar, voorzichtig. Reg.-nr. 0621
Droog en uit de buurt van regenbewaren. Reg.-nr. 0626

AEEA
IEC 60601-1: Medische elektrische apparatuur,
deel 1 - Algemene basale veiligheidseisen en
essentiële werking

Lees de gebruikershandleiding voordat
u het systeem gebruikt. Reg.-nr. 1641
Stapellimiet op nummer. Reg.-nr. 2403
ISO 7010: Grafische symbolen - Veiligheidskleuren
en veiligheidssymbolen - Geregistreerde
veiligheidssymbolen
Het lezen van de instructiehandleiding
is verplicht. Reg.-nr. M002
Waarschuwing. Reg.-nr. W001
Uit de buurt van open vuur,
vlammen en vonken bewaren. Open
ontstekingsbronnen en roken verboden.
Reg.-nr. P003
Rook niet in de buurt van het apparaatof tijdens de bediening ervan.
Reg.-nr. P002
104 - DUT Onderdeelnr. 15062781-C4 J | Gebruikershandleiding

IPX1

Druipwaterdichte apparatuur (IPX1):De Companion 5 biedt bescherming
tegen de schadelijke effecten van
indringing van vloeistof. (IPX1,
overeenkomstig IEC 60529)

FCC-KENNISGEVING:

Dit apparaat omvat mogelijk CYBLE-022001-00,
waaronder de antenne 2450AT18B100 van
Johnson Technology, en voldoet aan deel 15 van de
FCC-regels. Het apparaat voldoet aan de vereistenvoor de goedkeuring van een modulaire zender,
zoals in de openbare FCC-kennisgeving DA00-1407
wordt omschreven. De werking van de zender isonderworpen aan de volgende twee voorwaarden(1) Dit apparaat mag geen schadelijke interferentie
veroorzaken en (2) Dit apparaat moet alle ontvangeninterferentie aanvaarden, waaronder interferentie die
een ongewenste werking kan veroorzaken.
Op dit product kunnen een of meer Amerikaanse eninternationale octrooien van toepassing zijn. Bezoek
onze website voor een lijst met toepasselijke octrooien.
Pat.: www.caireinc.com/corporate/patents/.

Companion 5
Snelstartgids
1 Uw Companion 5 uitpakken

Companion 5

2 Kennismaking met uw Companion 5
Lees alle waarschuwingen, voorzorgsmaatregelen en
aanvullende informatie in verband met het apparaat indeze handleiding. Leer de voornaamste functies vande Companion 5 en het gebruikersbedieningspaneel
kennen.

VOORZIJDE
Behuizing

Stroomschakelaar

Ledscherm

Zuurstofslangkoppelinguitgang

Urenteller

Debietmeter

Houder voor
luchtbevochtigerfles(fles optioneel)

ACHTERZIJDEDraaghandgreep enluchtinlaatlocatie

Locatie vankoelluchtinlaat

Uitlaatopening
Luchtinlaatlocatie

Onderdeelnr. 15062781-C4 J | Gebruikershandleiding 105 - DUT

Companion 5
De onderdelen van de Companion 5
Behuizing

Duurzame plastic behuizing diede interne onderdelen van deCompanion 5 van CAIRE omhult enbeschermt.
Stroomschakelaar AAN/UIT-stroomschakelaar
die wordt gebruikt om destroomtoevoer naar het apparaat
in of uit te schakelen. De AANstand wordt aangeduid door het'I'-symbool en de UIT-stand door het'O'-symbool.
Slangkoppeling
De gefilterde en geconcentreerde
uitgangzuurstof komt hier uit deCompanion 5. Er wordt ofwel
gebruik gemaakt van een
neuscanule met enkelvoudig lumenof van een standaardzuurstofslang
om zuurstof aan de patiënt toe tedienen. De zuurstofslang die aande concentrator wordt bevestigd,
mag maximaal 15,2 meter (50 ft.)lang zijn.
Ledscherm
Bevat een groen, rood en geel*
ledlampje. Het groene lampje staat
voor een normale werking en degele* en rode lampjes wijzen op
alarmsituaties. (Raadpleeg het deel
'Alarmsituaties' in de instructies
voor meer informatie.)
Urenteller
Digitale weergave van de verstreken
gebruiksduur van de concentrator.
Geeft het dichtstbijzijnde tiende
van een uur weer en kan niet
worden teruggesteld. Toont
diagnostische alarmcodes wanneer
de concentrator een alarmsituatie
ondervindt.
Debietmeter
Gebruikt om het toegediende
zuurstofdebiet aan te passen enweer te geven. Het debiet kan
worden ingesteld tussen 0,5 en
5 liter per minuut (l/min) door aan
de knop te draaien.
Plaats waar u een
Houder voor
luchtbevochtigerfles luchtbevochtigerfles kunt
vastmaken. De houder bestaat(fles optioneel)
uit een elastische band die wordt
gebruikt om de fles te bevestigen op
de houder. Een luchtbevochtigerfles
voorziet extra vocht in dezuurstofstroom. De fles is gevuld
met gedistilleerd water en het vocht
wordt gebruikt om de uitdroging
van het weefsel in de neus tevoorkomen.
*Alleen voor modellen met een statusindicatorvoor zuurstofconcentratie (IZC)

106 - DUT Onderdeelnr. 15062781-C4 J | Gebruikershandleiding

3 Bevestiging van deluchtbevochtigerfles en de canule
Bij het gebruik van een luchtbevochtigerfles, dient u
de fles eerst te vullen met gedistilleerd water tot het
niveau aangegeven door de fabrikant. Plaats de fles inde daartoe voorziene houder en maak hem vast met
de elastische band. Sluit het schroefdraaduiteinde vande luchtbevochtigerslang aan op de fles en het andere
uiteinde op de slangkoppeling van de uitgang vande Companion 5. Sluit een zuurstofslang en/of een
neuscanule aan op de slangkoppeling van de uitgang
van de luchtbevochtigerfles.

Als u geen luchtbevochtigerfles gebruikt, sluit dezuurstofslang en/of neuscanule dan rechtstreeks aan op
de slangkoppeling van de uitgang van de Companion 5.

Companion 5
4 Inschakelen en opwarmen

6 De debietinstelling regelen op het
voorgeschreven niveau

Sluit het wisselstroomsnoer aan op een
wisselstroomstopcontact.

Gebruik de debietregelknop om het debiet te kiezen dat
is voorgeschreven door uw arts.

Zet de stroomschakelaar in de aan-stand.

Stroomschakelaar

Wanneer de Companion 5 naar behoren is aangesloten
op het stroomnet, brandt het groene lampje op het
ledscherm. Alle ledlampjes branden tijdens het
opstarten. Nadat de concentrator de opwarmingscyclus
heeft voltooid, blijft alleen het groene lampje branden.
Opmerking: Nadat u het apparaat voor het eerst
hebt ingeschakeld, gaat het O2-lampje branden
en verschijnt “O2 UP” op het scherm totdat deconcentratie de specificaties bereikt.

WAARSCHUWING: HET IS ERG
BELANGRIJK DAT ALLEEN HETVOORGESCHREVEN ZUURSTOFNIVEAUWORDT GEKOZEN. WIJZIG DE
DEBIETSELECTIE NIET TENZIJ DIT WERDAANGEGEVEN DOOR EEN BEVOEGDECLINICUS. DE ZUURSTOFCONCENTRATORMAG WORDEN GEBRUIKT TIJDENS HETSLAPEN ALS DIT WERD AANBEVOLEN
DOOR EEN BEVOEGDE CLINICUS.
Adem normaal door de neuscanule.
Om het debiet aan te passen:
Draai linksom om het debiet te verhogen.
Draai rechtsom om het debiet te verlagen.
Het midden van het balletje geeft het debiet aan. Zie deafbeelding hieronder.

Ledlamp­indicatorvoor stroom

Correct

Onderdeelnr. 15062781-C4 J | Gebruikershandleiding 107 - DUT

Companion 5
Belangrijk!

Veiligheidsinstructies zijn als volgt vermeld:
WAARSCHUWING: BELANGRIJKE
VEILIGHEIDSINFORMATIE VOORGEVAREN DIE ERNSTIG LETSEL KUNNEN
VEROORZAKEN.
LET OP: belangrijke informatie ombeschadiging van de Companion 5 tevoorkomen.
Opmerking: deze informatie vereist extra aandacht.

Indicaties voor gebruik
Beoogd gebruik

De Companion 5 is ontworpen voor de toediening
van extra zuurstof. De apparatuur is niet bedoeld terondersteuning of instandhouding van de levensfuncties
en biedt ook geen voorzieningen voor het bewaken vanpatiënten.
WAARSCHUWING: HET IS ERG
BELANGRIJK DAT ALLEEN HETVOORGESCHREVEN ZUURSTOFNIVEAUWORDT GEKOZEN. WIJZIG DE
DEBIETSELECTIE NIET TENZIJ DITWERD AANGEGEVEN DOOR EEN
BEVOEGDE CLINICUS. DE DRAAGBAREZUURSTOFCONCENTRATOR MAGGEBRUIKT WORDEN TIJDENS HETSLAPEN ALS DIT WERD AANBEVOLEN
DOOR EEN BEVOEGDE CLINICUS.
WAARSCHUWING: OM TE WAARBORGEN DATU DE HOEVEELHEID ZUURSTOF TOEGEDIENDKRIJGT DIE PASSEND IS MET HET OOG OP UWMEDISCHE TOESTAND, MAG DE COMPANION 5 PAS
WORDEN GEBRUIKT NADAT EEN OF MEERDERE
INSTELLINGEN INDIVIDUEEL ZIJN BEPAALD OF
SPECIAAL VOOR U EN VOLGENS UW SPECIFIEKACTIVITEITSNIVEAU ZIJN VOORGESCHREVEN.
COMPANION 5 MOET WORDEN GEBRUIKT MET
DE SPECIFIEKE COMBINATIE OF ONDERDELEN
EN ACCESSOIRES DIE OVEREENSTEMMEN MET
DE AANBEVELING VAN DE FABRIKANT VAN DE
CONCENTRATOR EN DIE WERDEN GEBRUIKT BIJ
HET BEPALEN VAN UW INSTELLINGEN.
WAARSCHUWING: DIT APPARAAT MAG NIETALS LEVENSINSTANDHOUDINGSSYSTEEM
WORDEN GEBRUIKT. GERIATRISCHE,PEDIATRISCHE OF ANDERE PATIËNTEN DIE HUN
ONGEMAK NIET ONDER WOORDEN KUNNEN
BRENGEN, HEBBEN MOGELIJK BEHOEFTE AANEXTRA BEWAKING EN/OF EEN VERSPREID
ALARMSYSTEEM OM DE INFORMATIE OVER HETONGEMAK EN DE MEDISCHE URGENTIE AAN
DE VERANTWOORDELIJKE ZORGVERLENER TE
VERSTREKKEN EN ZO SCHADE TE VOORKOMEN.
PATIËNTEN MET AUDITIEVE EN/OF VISUELE
BEPERKING(EN) VEREISEN MOGELIJK HULP OP HETVLAK VAN ALARMBEWAKING.

108 - DUT Onderdeelnr. 15062781-C4 J | Gebruikershandleiding

WAARSCHUWING: ZWANGERE VROUWEN OF
VROUWEN DIE BORSTVOEDING GEVEN MOGEN DE
IN DEZE HANDLEIDING AANBEVOLEN ACCESSOIRESNIET GEBRUIKEN WEGENS DE MOGELIJKE
AANWEZIGHEID VAN FTALATEN.
WAARSCHUWING: VOLGENS DE AMERIKAANSEFEDERALE WETGEVING MAG DIT APPARAATALLEEN WORDEN VERKOCHT OF VERHUURD OP
VOORSCHRIFT VAN EEN ARTS OF EEN ANDEREERKENDE PROFESSIONELE ZORGVERLENER.

Contra-indicaties voor gebruik
WAARSCHUWING: IN BEPAALDEOMSTANDIGHEDEN KAN GEBRUIK VAN
NIET-VOORGESCHREVEN ZUURSTOFGEVAARLIJK ZIJN. DIT APPARAAT IS
ALLEEN BESTEMD VOOR GEBRUIK OPVOORSCHRIFT VAN EEN ARTS.
WAARSCHUWING: NIET BESTEMD VOORGEBRUIK IN DE NABIJHEID VAN ONTVLAMBARE
ANESTHETICA.
WAARSCHUWING: NET ZOALS BIJ ELK ELEKTRISCH
APPARAAT, KAN HET APPARAAT MOGELIJKTIJDELIJK NIET WERKEN TEN GEVOLGE VAN EEN
STROOMONDERBREKING OF DOOR DE BEHOEFTEAAN ONDERHOUD AAN DE COMPANION 5 DOOREEN BEVOEGDE MONTEUR. DE COMPANION 5 IS
NIET GESCHIKT VOOR PATIËNTEN DIE SCHADELIJKEGEVOLGEN KUNNEN ONDERVINDEN DOOR EEN
DERGELIJKE ONDERBREKING.

Veiligheidsrichtlijnen
WAARSCHUWING: ER MOGEN GEENWIJZIGINGEN WORDEN AANGEBRACHTAAN DEZE APPARATUUR.
WAARSCHUWING: DE FABRIKANT RAADT AAN OM
EEN ALTERNATIEVE BRON VAN EXTRA ZUURSTOFTE GEBRUIKEN IN GEVAL VAN EEN STROOMUITVAL,
ALARMSITUATIES OF MECHANISCHE STORINGEN.
RAADPLEEG UW ARTS OF DE LEVERANCIER VAN
UW APPARATUUR VOOR HET VEREISTE SOORTRESERVESYSTEEM.
WAARSCHUWING: DIT APPARAAT LEVERTHOOGZUIVERE ZUURSTOF DIE EEN SNELLEVERBRANDING BEVORDERT. ZORG ERVOORDAT MEN NIET ROOKT EN DAT ER ZICH GEEN
OPEN VLAMMEN BEVINDEN BINNEN TWEE
(2) METER VAN (1) DIT APPARAAT OF (2) ALLEZUURSTOFBEVATTENDE ACCESSOIRES. NIETNALEVING VAN DEZE WAARSCHUWING KANLEIDEN TOT ERNSTIGE BRAND, MATERIËLE SCHADE
EN/OF FYSIEK LETSEL OF DE DOOD.

Companion 5
WAARSCHUWING: GEBRUIK HET APPARAAT NIETIN EEN BEPERKTE OF BESLOTEN RUIMTE (BIJV. EEN
KLEINE KOFFER OF HANDTAS) MET ONVOLDOENDE
VENTILATIE. DIT KAN OVERVERHITTING VAN DEZUURSTOFCONCENTRATOR VEROORZAKEN EN DE
GOEDE WERKING ERVAN BEÏNVLOEDEN.
WAARSCHUWING: DE CONCENTRATOR
MOET WORDEN GEPLAATST OP EEN
LOCATIE ZONDER ROOK, VERVUILENDESTOFFEN OF DAMPEN.
WAARSCHUWING: BEPAALDEZUURSTOFTOEDIENINGSACCESSOIRES DIE NIETZIJN GENOEMD VOOR GEBRUIK IN COMBINATIE
MET DEZE ZUURSTOFCONCENTRATOR, KUNNEN
DE WERKING VAN HET APPARAAT NADELIG
BEÏNVLOEDEN. DE AANBEVOLEN ACCESSOIRESSTAAN VERMELD IN DEZE HANDLEIDING.
WAARSCHUWING: ALS DEZUURSTOFCONCENTRATOR IS GEVALLEN,BESCHADIGD OF BLOOTGESTELD AAN
WATER, NEEM DAN CONTACT OP MET UWTHUISZORGVERLENER VOOR CONTROLE EN
EVENTUELE REPARATIE VAN HET APPARAAT.
GEBRUIK DE ZUURSTOFCONCENTRATOR NIET ALSHET SNOER OF DE STEKKER IS BESCHADIGD.
WAARSCHUWING: BLOKKEER NOOIT DE
OPENINGEN VAN DE LUCHTINLAAT/-UITLAAT. STEEKGEEN VOORWERPEN IN DE OPENINGEN VAN HETAPPARAAT EN VOORKOM DAT ER VOORWERPENIN VALLEN. DIT KAN OVERVERHITTING VAN DEZUURSTOFCONCENTRATOR VEROORZAKEN EN DE
GOEDE WERKING ERVAN BEÏNVLOEDEN.
WAARSCHUWING: VUL DE OPTIONELE
LUCHTBEVOCHTIGER NIET OVERMATIG. VUL DEOPTIONELE LUCHTBEVOCHTIGER MET WATERUITSLUITEND TOT AAN HET PEIL DAT WORDTOPGEGEVEN DOOR DE FABRIKANT VAN DE
LUCHTBEVOCHTIGER.
WAARSCHUWING: LEES DEZE HANDLEIDING EN
GA NA OF U DE INHOUD BEGRIJPT VOORDATU DE APPARATUUR IN GEBRUIK NEEMT. ALS U
DE WAARSCHUWINGEN EN INSTRUCTIES NIETBEGRIJPT, DIENT U CONTACT OP TE NEMEN MET
DE LEVERANCIER VAN UW APPARATUUR ALVORENSU DEZE IN GEBRUIK NEEMT; DIT OM LETSEL OFSCHADE TE VERMIJDEN.
WAARSCHUWING: VRAAG ONMIDDELLIJKMEDISCHE HULP ALS U ONGEMAK OF EENMEDISCHE NOODTOESTAND ERVAART.
WAARSCHUWING: GEBRUIK DEZUURSTOFCONCENTRATOR NIET BIJ
TEMPERATUREN DIE BUITEN DE
BEDRIJFSSPECIFICATIES LIGGEN, DIT KAN HETVERMOGEN VAN HET APPARAAT BEÏNVLOEDEN OM
DE GESPECIFICEERDE ZUURSTOFCONCENTRATIE TE
BEREIKEN. RAADPLEEG HET DEEL 'SPECIFICATIES'IN DEZE HANDLEIDING VOOR MEER INFORMATIE
OVER HET TEMPERATUURBEREIK.

WAARSCHUWING: GEBRUIK GEEN OLIE, SMEER,
PRODUCTEN OP BASIS VAN PETROLEUM OF
ANDERE ONTVLAMBARE PRODUCTEN IN
COMBINATIE MET ZUURSTOFHOUDENDE
ACCESSOIRES OF DE ZUURSTOFCONCENTRATOR.
GEBRUIK ALLEEN LOTIONS OF ZALVEN OP BASIS
VAN WATER DIE COMPATIBEL ZIJN MET ZUURSTOF.
ZUURSTOF VERSNELT DE VERBRANDING VAN
ONTVLAMBARE STOFFEN.
WAARSCHUWING: DE ZUURSTOFCONCENTRATORMAG NIET WORDEN GEBRUIKT NAAST OFGESTAPELD OP ANDERE APPARATUUR.
ALS AANPALEND OF GESTAPELD GEBRUIKONVERMIJDELIJK IS, MOET WORDENGECONTROLEERD OF HET APPARAAT NORMAALWERKT.
WAARSCHUWING: DE FABRIKANTRAADT AAN OM EEN ALTERNATIEVE
BRON VAN EXTRA ZUURSTOF TEGEBRUIKEN IN GEVAL VAN EEN
STROOMUITVAL, ALARMSITUATIES OFMECHANISCHE STORINGEN. RAADPLEEGUW ARTS OF DE LEVERANCIER VAN
UW APPARATUUR VOOR HET VEREISTESOORT RESERVESYSTEEM.
WAARSCHUWING: DIT APPARAAT LEVERTHOOGZUIVERE ZUURSTOF DIE EEN SNELLEVERBRANDING BEVORDERT. ZORG ERVOOR DATMEN NIET ROOKT EN DAT ER ZICH GEEN OPEN
VLAMMEN BEVINDEN IN DEZELFDE RUIMTE ALS: (1)
DIT APPARAAT OF (2) ALLE ZUURSTOFBEVATTENDEACCESSOIRES. NIET-NALEVING VAN DEZEWAARSCHUWING KAN LEIDEN TOT ERNSTIGE
BRAND, MATERIËLE SCHADE EN/OF FYSIEK LETSELOF DE DOOD.
WAARSCHUWING: BEPAALDEZUURSTOFTOEDIENINGSACCESSOIRES DIE NIETZIJN GENOEMD VOOR GEBRUIK IN COMBINATIE
MET DEZE ZUURSTOFCONCENTRATOR, KUNNEN
DE WERKING VAN HET APPARAAT NADELIG
BEÏNVLOEDEN. DE AANBEVOLEN ACCESSOIRESSTAAN VERMELD IN DEZE HANDLEIDING.
WAARSCHUWING: ALS DEZUURSTOFCONCENTRATOR IS GEVALLEN,BESCHADIGD OF BLOOTGESTELD AAN
WATER, NEEM DAN CONTACT OP MET UWTHUISZORGVERLENER VOOR CONTROLE EN
EVENTUELE REPARATIE VAN HET APPARAAT.
GEBRUIK DE ZUURSTOFCONCENTRATOR NIET ALSHET SNOER OF DE STEKKER IS BESCHADIGD.

Onderdeelnr. 15062781-C4 J | Gebruikershandleiding 109 - DUT

Companion 5
WAARSCHUWING: GEBRUIK UITSLUITENDRESERVEONDERDELEN DIE DOOR DE FABRIKANTWORDEN AANBEVOLEN OM EEN CORRECTEWERKING TE WAARBORGEN EN BRAND EN
BRANDWONDEN TE VOORKOMEN.
WAARSCHUWING: U MAG DE HULPSTUKKEN,AANSLUITINGEN, SLANGEN OF ANDEREACCESSOIRES VAN DE ZUURSTOFCONCENTRATORNIET SMEREN OM BRAND EN BRANDWONDEN TEVOORKOMEN.
WAARSCHUWING: ROKEN TIJDENS GEBRUIKVAN ZUURSTOF IS DE VOORNAAMSTEOORZAAK VAN BRANDWONDEN EN DAARMEEGEPAARD GAAND OVERLIJDEN. VOLG DEZE
VEILIGHEIDSWAARSCHUWINGEN:WAARSCHUWING: STA NIET TOE DAT ER WORDTGEROOKT, DAT ER KAARSEN BRANDEN OF OPEN
VUUR IS IN DE KAMER WAAR HET APPARAAT OF
DE ZUURSTOFHOUDENDE ACCESSOIRES ZICH
BEVINDEN.
WAARSCHUWING: ROKEN TERWIJL U EENZUURSTOFCANULE DRAAGT, KAN LEIDEN TOT
BRANDWONDEN IN HET GELAAT OF ZELFS DE
DOOD.
WAARSCHUWING: ALS U ROOKT, KUNNEN DE
VOLGENDE 3 STAPPEN UW LEVEN REDDEN:
SCHAKEL DE ZUURSTOFCONCENTRATOR UIT,VERWIJDER DE CANULE EN VERLAAT DE KAMERWAAR HET APPARAAT ZICH BEVINDT.
WAARSCHUWING: PLAATS DUIDELIJKE BORDEN
MET 'ROKEN VERBODEN - GEBRUIK VAN
ZUURSTOF' IN DE WONING OF ELDERS WAARZUURSTOF WORDT GEBRUIKT. PATIËNTEN EN
HUN ZORGVERLENERS MOETEN INGELICHTWORDEN OVER DE GEVAREN DIE ROKEN IN
DE AANWEZIGHEID VAN OF BIJ GEBRUIK VAN
ZUURSTOF INHOUDT.
WAARSCHUWING: GEBRUIK UWZUURSTOFCONCENTRATOR NIET IN DE
AANWEZIGHEID VAN ONTVLAMBARE GASSEN.
DIT KAN SNEL BRAND VEROORZAKEN EN LEIDEN
TOT MATERIËLE SCHADE, LICHAMELIJK LETSEL OFOVERLIJDEN.
WAARSCHUWING: ALS U DE CANULE AFNEEMTEN PLAATST OP KLEDING, BEDLINNEN, EEN SOFAOF KUSSENS ONTSTAAT ER EEN STEEKVLAM BIJ
BLOOTSTELLING AAN EEN SIGARET, WARMTEBRONOF VLAM.
WAARSCHUWING: PLAATS GEEN NEUSCANULE OP
OF ONDER KLEDING, BEDLINNEN OF KUSSENS VAN
EEN STOEL. ALS HET APPARAAT IS INGESCHAKELD
MAAR NIET WORDT GEBRUIKT, ZAL DE ZUURSTOF
HET MATERIAAL ONTVLAMBAAR MAKEN. ZET
DE I/0-SCHAKELAAR OP 0 (UIT) WANNEER DEZUURSTOFCONCENTRATOR NIET WORDT GEBRUIKT.

110 - DUT Onderdeelnr. 15062781-C4 J | Gebruikershandleiding

WAARSCHUWING: VOER GEENONDERHOUD UIT DAT GEEN DEELUITMAAKT VAN DE MOGELIJKEOPLOSSINGEN IN DEZE HANDLEIDING.
VERWIJDER GEEN DEKSELS.
ALLEEN DE LEVERANCIER VAN UWAPPARATUUR OF EEN BEVOEGDEONDERHOUDSTECHNICUS MAG
DEKSELS VERWIJDEREN OF ONDERHOUDUITVOEREN AAN DIT APPARAAT.
WAARSCHUWING: GEBRUIK ALLEEN ACCESSOIRESDIE DOOR DE FABRIKANT ZIJN AANBEVOLEN.
GEBRUIK VAN ANDERE ACCESSOIRES KANGEVAARLIJK ZIJN, ERNSTIGE SCHADE TOEBRENGEN
AAN UW ZUURSTOFCONCENTRATOR EN DEGARANTIE ANNULEREN.
LET OP: koppel altijd dewisselstroomvoeding los van het
contact voor u de wisselstroomvoeding
loskoppelt van de zuurstofconcentrator.
LET OP: plaats de zuurstoftoevoerslang en
netsnoeren altijd zo dat ze geen gevaar voor
struikelen vormen.
LET OP: om te voorkomen dat degarantie zijn geldigheid verliest, dientu alle aanwijzingen van de fabrikant in
acht te nemen.
Opmerking: draagbare en mobiele RF-communicatieapparatuur kan elektrische medische apparatuurbeïnvloeden.
WAARSCHUWING: PLAATS ALTIJD DE ZUURSTOFTOEVOERBUIS EN DE STROOMKABELS OP EEN MANIERDIE TRIPGEVAAR OF MOGELIJKE ONGEVALLEN
VERSTORING VOORKOMT.
WAARSCHUWING: DIT PRODUCT KAN U BLOOTSTELLEN AAN CHEMICALIËN, WAARONDER NIKKEL,DAT DOOR DE STAAT CALIFORNIË ALS KANKERVERWEKKEND IS ERKEND. GA VOOR MEER INFORMATIE
NAAR WWW.P65WARNINGS.CA.GOV.
WAARSCHUWING: IN HET GEVAL ER EEN ERNSTIGINCIDENT MET DIT APPARAAT OPTREEDT, MOET DEGEBRUIKER HET INCIDENT ONMIDDELLIJK MELDENAAN DE AANBIEDER EN / OF DE FABRIKANT. EEN
ERNSTIG INCIDENT WORDT GEDEFINIEERD ALS
EEN LETSEL, OVERLIJDEN OF MOGELIJK LETSEL /OVERLIJDEN VEROORZAKEN ALS HET INCIDENTZICH OPNIEUW VOORDOET. DE GEBRUIKER KANHET INCIDENT OOK MELDEN AAN DE BEVOEGDEAUTORITEIT IN HET LAND WAAR HET INCIDENT
PLAATSVOND.

Companion 5
Specificaties
De volgende tabel geeft belangrijke informatie over deaanbevolen bedrijfsomgeving of bedrijfsvoorwaarden
voor correct gebruik van het apparaat.

Specificaties
DebietO2-concentratie
Afmetingen

0,5–5,0 l/min
90% (+5,5% /-3%)
54,6 x 31,8 x 34,2 cm(21,5 x 12,5 x 13,5 in.)
Gewicht
16,3 kg (36 lbs)
Geluidsdrukniveau 50 dB(A) bij 2,0 l/min
Stroomverbruik
285 W bij 2 l/min, max. 350 W
Bedrijfstemperatuur 5 °C tot 40 °C (41 °F tot 104 ºF )Bedrijfsvochtigheid 15% - 90% bij een dauwpunt van
28 °C (82,4 °F)
Opslagtemperatuur -25 °C tot 70 °C(-13 °F tot 158 °F)
Opslagvochtigheid 0% - 90% niet-condenserend
Elektriciteit
Gebruik geen verlengsnoer.
Gebruik geen elektrische
stopcontacten met een
schakelknop.
Hoogte bij gebruik -382-3011 meter (-1253-9879 ft.)
Bedrijfsduur
tot 24 uur per dag
De verwachte gebruiksduur van dit apparaat bedraagt
minimaal vijf jaar.
WAARSCHUWING: HETGEBRUIK VAN HET APPARAATBUITEN DE GESPECIFICEERDEBEDRIJFSOMSTANDIGHEDEN WORDTVERWACHT DAT HET DE STROOM- EN
PERCENTAGE VAN ZUURSTOF ENDAARVAN DE KWALITEIT VAN DE
THERAPIE NADER BEÏNVLOEDT.
Opmerking: als de zuurstofconcentrator voor lange
tijd buiten de normale bedrijfstemperatuur werd
opgeborgen, moet het apparaat eerst terug de normale
bedrijfstemperatuur bereiken alvorens opnieuw in teschakelen. (Raadpleeg het deel Specificaties in deze
handleiding.)

Correcte plaatsing van de Companion 5

Kies een locatie waar geen rook, dampen of
luchtverontreiniging is. Bij correcte plaatsing van het
apparaat is er opname van lucht mogelijk via de drieluchtinlaatlocaties bovenaan de achterzijde en aan deonderzijde van de behuizing en kan de uitlaatlucht
de uitlaatopening aan de linkerbenedenzijde van het
apparaat onbelemmerd verlaten.
De luchtinlaten bevinden zich bovenaan aan deachterzijde en aan de onderzijde van de concentrator.
Plaats het apparaat dusdanig dat de concentrator
minimaal 30 cm (12 in.) is verwijderd van muren,
meubels, gordijnen of andere voorwerpen.
Plaats het apparaat dusdanig dat de alarmen
hoorbaar zijn.
Plaats de zuurstoftoevoerslang dusdanig dat er geenplooien of blokkeringen zijn.
WAARSCHUWING: GEBRUIK UWZUURSTOFCONCENTRATOR NIET IN DE
AANWEZIGHEID VAN ONTVLAMBAREGASSEN. DIT KAN SNEL BRAND
VEROORZAKEN EN LEIDEN TOT
MATERIËLE SCHADE, LICHAMELIJKLETSEL OF OVERLIJDEN. GEBRUIK GEEN
OLIE, SMEER, PRODUCTEN OP BASIS VAN
PETROLEUM OF ANDERE ONTVLAMBARE
PRODUCTEN IN COMBINATIE METZUURSTOFHOUDENDE ACCESSOIRES OF
DE ZUURSTOFCONCENTRATOR. GEBRUIKALLEEN LOTIONS OF ZALVEN OP BASIS
VAN WATER DIE COMPATIBEL ZIJN METZUURSTOF. ZUURSTOF VERSNELT DEVERBRANDING VAN ONTVLAMBARE
STOFFEN.
WAARSCHUWING: DIT APPARAAT LEVERTHOOGZUIVERE ZUURSTOF DIE EEN SNELLEVERBRANDING BEVORDERT. ZORG ERVOORDAT MEN NIET ROOKT EN DAT ER ZICH GEEN
OPEN VLAMMEN BEVINDEN IN DEZELFDERUIMTE ALS: (1) DIT APPARAAT OF (2) ALLEZUURSTOFBEVATTENDE ACCESSOIRES. NIETNALEVING VAN DEZE WAARSCHUWING KANLEIDEN TOT ERNSTIGE BRAND, MATERIËLE SCHADE
EN/OF FYSIEK LETSEL OF DE DOOD.

Onderdeelnr. 15062781-C4 J | Gebruikershandleiding 111 - DUT

Companion 5
Gebruiksaanwijzing
Voor gebruik

Deze gebruikershandleiding dient als referentie om u tehelpen het apparaat te bedienen en te onderhouden. Alsu vragen of twijfels hebt, kunt u contact opnemen met
uw thuiszorgverlener.
Belangrijk! Probeer NIET om de Companion 5 te
gebruiken zonder eerst de Veiligheidsrichtlijnen
van deze handleiding te hebben gelezen. Volgalle gebruiksaanwijzingen. Houd rekening
met de Waarschuwingen op het apparaat
en in de gebruikershandleiding. Volg alleveiligheidsmaatregelen om gevaar voor brand,
persoonlijk letsel of ernstige beschadiging van deCompanion 5 te voorkomen.
De Companion 5 van CAIRE maakt niet zelf
zuurstof aan. Het apparaat produceert hoogzuivere
zuurstof uit de omgevingslucht in de ruimte en
dient deze vervolgens aan u toe. Lucht in de ruimte
bevat ongeveer slechts 21 volumeprocent natuurlijk
aanwezige zuurstof.
Door een compressor wordt lucht in de ruimte
opgenomen door de concentrator. De lucht reist dandoor een reeks filters en een moleculair zeefbed dat
stikstofmoleculen chemisch opneemt. Hierdoor wordt
de met zuurstof verrijkte lucht die uit de concentrator
komt, toegediend met 87-95 volumeprocent zuurstof.
WAARSCHUWING: MAAK DE BEHUIZING,HET BEDIENINGSPANEEL EN HETSTROOMSNOER ALLEEN SCHOON MET
EEN MILD SCHOONMAAKMIDDELEN EEN VOCHTIGE (GEEN NATTE)DOEK OF SPONS, VEEG VERVOLGENS
ALLE OPPERVLAKKEN DROOG. LAATGEEN VLOEISTOF IN HET APPARAATTERECHTKOMEN. BESTEED BIJZONDEREAANDACHT AAN DE ZUURSTOFUITGANGVOOR DE CANULEAANSLUITING OM
TE VERZEKEREN DAT DEZE VRIJ IS VAN
STOF, WATER EN ANDERE DEELTJES.
LET OP: blokkeer nooit de openingenvan de luchtinlaat/-uitlaat.
Dit kan oververhitting van dezuurstofconcentrator veroorzaken ende goede werking ervan beïnvloeden.
Gebruik de zuurstofconcentrator niet
als het luchtinlaatfilter niet is bevestigd.
Als een tweede filter is meegeleverd,
plaatst u het 'vervangingsfilter' voordat
u het vervuilde filter reinigt. Maak het
vuile filter schoon in een sopje en laat
het volledig drogen vóór gebruik.

112 - DUT Onderdeelnr. 15062781-C4 J | Gebruikershandleiding

Stap 1: uw Companion 5 plaatsen voor gebruik
Plaats de Companion 5 in een goed geventileerde,verlichte ruimte. Zorg ervoor dat de luchtinlaat- en-uitlaatopeningen niet worden geblokkeerd. Plaatsde Companion 5 dusdanig dat hoorbare en visueleindicatoren of alarmen zichtbaar en hoorbaar zijn.
Sluit het apparaat aan op een wisselstroomcontact.
Stap 2: Als u de luchtbevochtigerfles NIET gaat
gebruiken

Luchtinlaatlocatie

Locatie vankoelluchtinlaat

Uitlaatopening
Luchtinlaatlocatiea. Sluit een neuscanule of zuurstofslang rechtstreeks
aan op de slangkoppeling van de zuurstofuitgang.

b. Ga door naar stap 4.
Stap 3: Als u de luchtbevochtigerfles WEL gaat
gebruiken
a. Draai het deksel van de luchtbevochtigerfles open.
b. Vul de fles met gedistilleerd water. Zorg ervoor dat
het waterniveau het door de fabrikant aangegevenmaximale en minimale niveau dat zichtbaar is op defles niet overschrijdt.
c. Plaats het deksel terug op de luchtbevochtigerfles.
Zorg ervoor dat er geen lekkages zijn.
d. Sluit een neuscanule of standaardzuurstofslang
aan op de slangkoppeling van de uitgang van deluchtbevochtigerfles.
Opmerking: het gebruik van een luchtbevochtiger
die niet is genoemd voor gebruik in combinatie met
deze zuurstofconcentrator kan de werking ervanbeïnvloeden.

Companion 5
g. Plaats de fles in de daartoe voorziene ruimte van dehouder voor de luchtbevochtigerfles.
h. Bevestig de fles op de houder met behulp van deelastische band.

e. Sluit het schroefdraaduiteinde van deluchtbevochtigerslang aan op het deksel van deluchtbevochtigerfles.

i. Ga door naar stap 4.
Stap 4: inschakelen en opwarmen
a. Sluit het elektrische snoer van de Companion 5 van
CAIRE aan op een wisselstroomcontact.
WAARSCHUWING: CONTROLEER HET
ELEKTRISCHE SNOER OP SCHADE VOORU DIE GEBRUIKT. SLUIT HET SNOERNIET AAN OP EEN STOPCONTACT OF
PROBEER NIET OM DE CONCENTRATORTE GEBRUIKEN INDIEN HET SNOER ISBESCHADIGD.

f. Sluit het andere uiteinde van deluchtbevochtigerslang aan op de slangkoppeling vande uitgang van de Companion 5 van CAIRE.

b. Zet de stroomschakelaar in de 'AAN' (I)-stand.
Wanneer de Companion 5 naar behoren isaangesloten op het stroomnet, brandt het groene
lampje op het ledscherm. Alle ledlampjes branden
tijdens het opstarten. Nadat de concentrator de
opwarmingscyclus heeft voltooid, blijft alleen het
groene lampje branden.
Opmerking: na de eerste INSCHAKELING van het
apparaat dient u tien (10) minuten te wachten tot het
apparaat de prestatiespecificaties bereikt.

Onderdeelnr. 15062781-C4 J | Gebruikershandleiding 113 - DUT

Companion 5
WAARSCHUWING: HET IS ERG
BELANGRIJK DAT ALLEEN HETVOORGESCHREVEN ZUURSTOFNIVEAUWORDT GEKOZEN. WIJZIG DE
DEBIETSELECTIE NIET TENZIJ DIT WERDAANGEGEVEN DOOR EEN BEVOEGDECLINICUS. DE ZUURSTOFCONCENTRATORMAG WORDEN GEBRUIKT TIJDENS HETSLAPEN ALS DIT WERD AANBEVOLEN
DOOR EEN BEVOEGDE CLINICUS.
Stap 5: pas de debietregeling aan

Stap 6: controleer het debiet en adem normaal

a. Draai de debietregelknop naar het zuurstofdebiet(l/min) dat is voorgeschreven door uw arts.

a. Controleer of er wel degelijk zuurstof door deneuscanule stroomt en dat de slang geen plooien,knikken of blokkeringen vertoont.
Opmerking: controleer of de canule correct is geplaatst
en goed vastzit. U moet de zuurstofstroom naar devertakking van de neuscanule horen of voelen.
b. Plaats uw neuscanule zoals het moet en adem
normaal door de neuscanule.
Stap 7: uitschakelen
a. Zet de stroomschakelaar in de 'UIT' (O)-standwanneer de Companion 5 van CAIRE niet meer
wordt gebruikt.

Om het debiet aan te passen:
Draai linksom om het debiet te verhogen.
Draai rechtsom om het debiet te verlagen.
Het midden van het balletje geeft het debiet aan.
De onderstaande afbeelding geeft een debiet van
4,0 l/min. aan.

Correct

114 - DUT Onderdeelnr. 15062781-C4 J | Gebruikershandleiding

LET OP: gebruik alleen accessoires
die door de fabrikant zijn
aanbevolen. Gebruik van andere
accessoires kan gevaarlijk zijn,
ernstige schade toebrengen aan uw
zuurstofconcentrator en de garantie
annuleren.
LET OP: gebruik geen verlengsnoeren in combinatie
met dit apparaat en beperk het aantal apparaten
dat is aangesloten op hetzelfde stopcontact. Het
gebruik van verlengsnoeren kan de prestaties
van het apparaat negatief beïnvloeden. Indien
er toch te veel apparaten op een stopcontact
worden aangesloten, dan bestaat het risico op
overbelasting waardoor de zekeringen kunnen
springen. Als de zekeringen niet goed werken, kan
dit zelfs brand veroorzaken.
LET OP: gebruik van niet-gespecificeerde kabels
en adapters, met uitzondering van kabels en
adapters die worden verkocht door de fabrikant
van de medische elektrische apparatuur
als vervangende onderdelen voor internecomponenten, kan resulteren in een verhoogde
uitstoot of verminderde immuniteit van dezuurstofconcentrator.

Companion 5
WAARSCHUWING: BLOKKEER NOOIT
DE OPENINGEN VAN DE LUCHTINLAAT/UITLAAT. DIT KAN OVERVERHITTINGVAN DE ZUURSTOFCONCENTRATORVEROORZAKEN EN DE GOEDE WERKINGERVAN BEÏNVLOEDEN.
Opmerking: aan leverancier van apparatuur: De
volgende accessoires voor zuurstoftoediening worden
aanbevolen voor gebruik met de Companion 5:• Neuscanule: CAIRE-onderdeelnummer CU002-1• Ademslang voor luchtbevochtiger: CAIREonderdeelnummer 20843882• Luchtbevochtigerfles: CAIRE-onderdeelnummer
HU003-1• Firebreak: CAIRE-onderdeelnummer 20629671
Een brandgang wordt vereist voor gebruik met elke
canule.
• CAIRE biedt een brandgang aan die bedoeld
is om te gebruiken in combinatie met dezuurstofconcentrator. De brandgang is een thermischezekering om de gasstroom te stoppen in het geval dat
de stroomafwaartse canule of zuurstofslang wordt
ontstoken en brandt aan de brandgang. Het wordt
in lijn met de neuscanule of zuurstofslang tussende patiënt en de zuurstofuitlaat van de Companion
5 geplaatst. Raadpleeg voor een juist gebruik van
de brandgang altijd de instructies van de fabrikant(meegeleverd bij elke brandgangset).
• Zie voor alle extra aanbevolen accessoires deaccessoirecatalogus (PN MLLOX0010) die beschikbaar
is op www.caireinc.com.

Onderdeelnr. 15062781-C4 J | Gebruikershandleiding 115 - DUT

Companion 5
Alarmsituaties
Uw Companion 5 van CAIRE combineert twee of drie(voor modellen met IZC) ledlampjes en een hoorbaar
alarm om u te verwittigen als er een storing in uw
concentrator optreedt.
1) Groene ledlamp (werking.

Groen) – wijst op een normale

Rood2) Rode ledlamp ( ) – wijst op een systeemfout.
Raadpleeg de alarmsituatietabel hieronder voor
meer informatie.

Geel3) Gele ledlamp ( ) – wijst op een lage
zuurstofconcentratie. Raadpleeg dealarmsituatietabel hieronder voor meer informatie.
In de alarmsituatietabel op de volgende pagina kunt u
de verschillende alarmsituaties raadplegen die kunnen
worden weergegeven op uw concentrator. Er worden
mogelijke oorzaken en handelingen opgesomd die u
dient te ondernemen als het alarm zich voordoet. Het
is aanbevolen dat u zo snel mogelijk contact opneemt
met uw zorgverlener bij elk alarm.
WAARSCHUWING: NEGEER GEEN
ALARMENWAARSCHUWINGEN: DE FABRIKANTRAADT AAN OM EEN ALTERNATIEVE
BRON VAN EXTRA ZUURSTOF TEGEBRUIKEN IN GEVAL VAN EEN
STROOMUITVAL, ALARMSITUATIES OFMECHANISCHE STORINGEN. RAADPLEEGUW ARTS OF DE LEVERANCIER VAN
UW APPARATUUR VOOR HET VEREISTESOORT RESERVESYSTEEM.

116 - DUT Onderdeelnr. 15062781-C4 J | Gebruikershandleiding

Boven: Companion 5

Companion 5
Alarmsituatietabel
Hoorbaaralarm

Gekleurd
ledlampje

Oorzaak

Wat kunt u doen
Geen

De concentrator werkt
naar behoren.

Uit
groen

Periodieke
ROOD

Periodieke

De concentrator is zonderstroom gevallen, maar destroomschakelaar staat
nog steeds in de 'AAN'stand.

Controleer of het elektrische snoer van deconcentrator is aangesloten op een stopcontact
en of het stopcontact onder stroom staat.
Probeer een ander stopcontact. Sluit aan op een
reservezuurstoftoevoer en neem contact op met uw
zorgverlener als het probleem aanhoudt.

Systeemfout

Zet de stroomschakelaar onmiddellijk in de'UIT'-stand. Koppel de wisselstroomvoeding
los van het stopcontact. Wacht 5 minuten. Sluitde wisselstroomvoeding opnieuw aan op het
stopcontact en schakel de concentrator opnieuw
in. Er is onderhoud vereist als het alarm aanhoudt.
Sluit aan op een reservezuurstoftoevoer en neem
onmiddellijk contact op met uw zorgverlener.

Debiet van het product is
te hoog of te laag.

1. Controleer of de canule niet is geplooid ofgeblokkeerd. Controleer bij gebruik van een
luchtbevochtiger of de fles correct is gevuld en geen
blokkering veroorzaakt.
2. Controleer of de Companion 5 naar behoren
wordt geventileerd. Het apparaat moet minstens
30 cm (12 in.) van andere oppervlakken verwijderdzijn om de openingen niet te belemmeren.
3. Als het probleem blijft aanhouden, schakel danover naar een andere zuurstofbron of neem contact
op met uw zorgverlener voor ondersteuning.

ROOD

Periodieke
ROOD

Periodieke
geel

Periodiekegeel

De Companion 5 heeft
een laag zuurstofpeil
gedetecteerd.

De Companion 5 heeft
een laag zuurstofpeil
gedetecteerd.

1. Zorg dat het luchtinlaatfilter en de uitlaatlocaties
niet verstopt of belemmerd zijn. 2. Zorg dat deCompanion 5 zich in een goed geventileerde ruimte
bevindt. Controleer of er minstens 30 cm (12 in.)vrij is tussen de achterzijde en de zijkanten vande Companion 5 en andere voorwerpen (meubels,
gordijnen, enz.).
3. Als het probleem blijft aanhouden, schakel
dan over naar een andere zuurstofbron of neem
onmiddellijk contact op met uw zorgverlener.

Onderdeelnr. 15062781-C4 J | Gebruikershandleiding 117 - DUT

Companion 5
Probleemoplossing
Het is aanbevolen dat u contact opneemt met uw
zorgverlener bij elk alarm dat uw concentrator
weergeeft of als u denkt dat uw concentrator niet naarbehoren werkt. Probeer niet om de concentrator zelf terepareren. Alleen een opgeleide onderhoudstechnicus
dient het onderhoud uit te voeren. Nietteminzijn er enkele probleemoplossingsstappen die ukunt ondernemen als u problemen heeft met uw
Companion 5 van CAIRE. Deze worden opgesomd inde volgende tabel. U kunt deze tabel gebruiken doorde controles voor uw probleem te volgen in volgorde
waarin deze zijn vermeld.

Probleemoplossingstabel voor de gebruiker
Probleem

De concentrator
is ingeschakeld,
maar werkt niet.

U ervaart eenlage of geenzuurstofstroom.

Mogelijke oorzaak

Wat kunt u doen

1. Het elektrische snoer is niet
aangesloten op een stopcontact.

1. Sluit het elektrische snoer aan.

2. Het stopcontact staat niet onderstroom.

2. Controleer de zekering en de stroomkring bij u thuis.
Probeer een ander stopcontact.

3. Intern defect.

3. Sluit het apparaat aan op een reservezuurstoftoevoer.
Neem onmiddellijk contact op met uw zorgverlener.

1. De instelling van de debietregelingis veranderd.

1. Controleer of de debietregelknop is ingesteld op deaangewezen instelling (l/min) die is voorgeschreven door
uw arts.

2. De slang is losgekoppeld.

2. Controleer of de slang stevig is bevestigd op
alle aansluitingen (slangkoppeling van de uitgang,
luchtbevochtigerfles, waterafscheider, enz.). Maak deaansluitingen strakker vast indien nodig.

3. Er zit een knik in de zuurstofslangof de canule.

3. Controleer de slang op knikken of blokkeringen.
Repareer indien nodig.

4. Uw canule of slang vertoont een
lekkage.

4. Controleer alle slangen en de canule op lekkages.
U kunt dit doen door te controleren of u lucht hoort ofvoelt ontsnappen. Vervang de lekkende slang of canule.

5. De luchtstroom naar deconcentrator is verhinderd.

5. Controleer of er niets de ingangen van deconcentrator blokkeert. Zorg ervoor dat het apparaat
minimaal 30 cm (12 in.) is verwijderd van meubels,
muren of andere voorwerpen.

6. Intern defect.

6. Sluit het apparaat aan op een reservezuurstoftoevoer.
Neem onmiddellijk contact op met uw zorgverlener.

Opmerking: neem contact op met de technische ondersteuning op het nummer 1-800-482-2473 met de op het lcdscherm getoonde alarmcode, als uw Companion 5 nog altijd een alarmsignaal weergeeft nadat u de alarmsituatieen probleemoplossingstabel hebt gevolgd.

118 - DUT Onderdeelnr. 15062781-C4 J | Gebruikershandleiding

Companion 5
Reiniging, zorg en routineonderhoud
Routineonderhoud

Het routineonderhoud bestaat uit het wekelijks
verwisselen van het grovedeeltjesfilter in de handgreep(indien geïnstalleerd).

Zorg en reiniging van het apparaat doorde gebruiker
Vervanging van de canule
Vervang de toevoerslang en canule regelmatig
overeenkomstig de aanbevelingen van uw
thuiszorgverlener. Uw arts of thuiszorgverlener zal
u informatie geven over reiniging, ontsmetting envervanging.
Opmerking: volg altijd de instructies voor correct
gebruik van de fabrikant van de canule. Vervang
de wegwerpcanule volgens de instructies van defabrikant of van de leverancier van uw apparatuur.
Bijkomende benodigdheden zijn beschikbaar bij deleverancier van uw apparatuur.

Luchtinlaatfilter (patroon)
Het luchtinlaatfilter mag uitsluitend door uw
zorgverlener worden geopend of vervangen.
Grovedeeltjesfilter (optioneel):Het filter voor grove deeltjes bevindt zich helemaal
onderaan de Companion 5.

Grovedeeltjesfilter

Je moet het een keer per week inspecteren enschoonmaken. Gebruik het volgende proces om het
filter te reinigen:5) Plaats het filter terug in de behuizing.
Opmerking: de fabrikant raadt niet aan om dezeapparatuur te steriliseren.
Opmerking: gebruik de zuurstofconcentrator niet als
het luchtinlaatfilter niet is bevestigd.
Luchtbevochtigerfles (indien van toepassing)U dient uw luchtbevochtigerfles dagelijks tecontroleren om ervoor te zorgen dat het waterniveau
zich tussen de gespecificeerde minimale en maximale
niveaus bevindt. Vul de fles indien nodig opnieuw met
gedistilleerd water. Reinig de luchtbevochtigerfles
volgens de instructies van de leverancier.
Reinigingsprocedure:
Schakel de Companion 5 UIT en koppel het apparaat
los van de wisselstroomvoeding voordat u het apparaat
reinigt of ontsmet. Spuit NIET rechtstreeks op debehuizing. Gebruik een vochtige (geen natte) doekof spons. Spuit een milde detergentoplossing op dedoek of spons om de behuizing en stroomsnoer tereinigen. Gebruik Lysol® Brand II-ontsmettingsmiddel(of een gelijkaardig product) om de Companion 5 teontsmetten. Verdun het product volgens de instructies
van de fabrikant van het reinigingsmiddel, maar spuitde vloeistof niet rechtstreeks op de Companion 5.
WAARSCHUWING: RISICO OP
ELEKTRISCHE SCHOKKEN. SCHAKEL HETAPPARAAT UIT EN HAAL DE STEKKERUIT HET STOPCONTACT VOORDATU HET APPARAAT SCHOONMAAKT.
ZO VOORKOMT U MOGELIJKE
ELEKTRISCHE SCHOKKEN OF RISICOOP BRANDWONDEN. ALLEEN DE
LEVERANCIER VAN UW APPARATUUR OF
EEN BEVOEGD ONDERHOUDSTECHNICUS
MAG DE BEHUIZING VERWIJDEREN OF
HET APPARAAT ONDERHOUDEN.
WAARSCHUWING: RISICO OP ELEKTRISCHESCHOKKEN. SCHAKEL HET APPARAAT UIT EN HAAL
DE STEKKER UIT HET STOPCONTACT VOORDAT U
HET APPARAAT SCHOONMAAKT. ZO VOORKOMT U
MOGELIJKE ELEKTRISCHE SCHOKKEN OF RISICOOP BRANDWONDEN. ALLEEN DE LEVERANCIER
VAN UW APPARATUUR OF EEN BEVOEGD
ONDERHOUDSTECHNICUS MAG DE BEHUIZINGVERWIJDEREN OF HET APPARAAT ONDERHOUDEN.

1) Verwijder het grovedeeltjesfilter.
2) Was het filter in een warm sopje met een milde zeep.
3) Spoel het filter grondig met helder kraanwater en
knijp het overtollige water eruit.
4) Laat het filter aan de lucht drogen.
Onderdeelnr. 15062781-C4 J | Gebruikershandleiding 119 - DUT

Companion 5
WAARSCHUWING: RISICO OP ELEKTRISCHESCHOKKEN. VERWIJDER HET STROOMSNOER UIT
HET STOPCONTACT VOORDAT U HET APPARAATSCHOONMAAKT OM ELEKTRISCHE SCHOKKENEN BRANDGEVAAR TE VOORKOMEN. ALLEEN DE
LEVERANCIER VAN UW APPARATUUR OF EEN
BEVOEGD ONDERHOUDSTECHNICUS MAG DE
BEHUIZING VERWIJDEREN OF HET APPARAATONDERHOUDEN.
WAARSCHUWING: VOORKOM DAT DEZUURSTOFCONCENTRATOR NAT WORDT OFDAT ER VLOEISTOF BINNENDRINGT IN HETAPPARAAT. DIT KAN LEIDEN TOT EEN STORING
OF KAN HET APPARAAT UITSCHAKELEN EN EEN
VERHOOGD RISICO OP ELEKTRISCHE SCHOKKEN OF
BRANDWONDEN MET ZICH MEEBRENGEN.
WAARSCHUWING: HET KAN ONZEKER ZIJN OM
DE COMPANION 5 TE VERBINDEN MET ANDEREAPPARATUUR.

WAARSCHUWING: MAAK DE BEHUIZING, HET
BEDIENINGSPANEEL EN HET STROOMSNOER
ALLEEN SCHOON MET EEN MILDSCHOONMAAKMIDDEL EN EEN VOCHTIGE(GEEN NATTE) DOEK OF SPONS, VEEGVERVOLGENS ALLE OPPERVLAKKEN DROOG.
LAAT GEEN VLOEISTOF IN HET APPARAATTERECHTKOMEN. BESTEED BIJZONDEREAANDACHT AAN DE ZUURSTOFUITGANG VOOR
DE CANULEAANSLUITING OM TE VERZEKERENDAT DEZE VRIJ IS VAN STOF, WATER EN ANDERE
DEELTJES.
WAARSCHUWING: GEBRUIK GEEN VLOEISTOF
RECHTSTREEKS OP HET APPARAAT. DE LIJST MET
AFGERADEN CHEMISCHE STOFFEN OMVAT, MAAR
IS NIET BEPERKT TOT: ALCOHOL EN PRODUCTENOP ALCOHOLBASIS, PRODUCTEN OP BASIS VAN
GECONCENTREERD CHLOOR (ETHYLEENCHLORIDE)EN PRODUCTEN OP OLIEBASIS (PINE-SOL®,LESTOIL®). DEZE STOFFEN MOGEN NIET WORDENGEBRUIKT OM DE PLASTIC BEHUIZING VAN DEZUURSTOFCONCENTRATOR SCHOON TE MAKEN. ZE
KUNNEN HET PLASTIC BESCHADIGEN.

AfvoerLever de Companion 5, inclusief alle onderdelen,altijd in bij uw thuiszorgaanbieder voor een correcteafvoer. U kunt ook contact opnemen met de lokale
autoriteiten voor inlichtingen over de correcte afvoer
van de batterij.

AEEA en RoHS
Dit symbool herinnert gebruikers eraan om
de apparatuur aan het einde van de levensduur naar
een recyclingbedrijf te brengen, conform de AEEA-richtlijn (afgedankte elektrische en elektronische
apparatuur).

Onze producten voldoen aan de beperkingen vande richtlijn inzake gevaarlijke stoffen (RoHS). Ze
bevatten geen sporen van lood of andere gevaarlijke
stoffen.
VOORZICHTIG: Neem voor een juiste
verwijdering contact op met uw
leverancier van apparatuur of de lokale
overheidsinstantie voor instructies voor
verwijdering.

CAIRE en CAIRE Inc. zijn gedeponeerde handelsmerken van
CAIRE Inc. Bezoek onze onderstaande website voor een volledigelijst van handelsmerken.
Handelsmerken: www.careinc.com/corporate/trademarks.
Copyright © 2020 CAIRE Inc.   CAIRE Inc. behoudt zich het recht
voor om te allen tijde zonder voorafgaande kennisgeving en zonder
verdere verplichtingen of gevolgen producten uit het assortiment tehalen en om prijzen, materialen, uitrusting, kwaliteit, beschrijvingen,specificaties en/of processen van producten te wijzigen. Alle hierin
niet uitdrukkelijk genoemde rechten zijn, waar van toepassing,voorbehouden.
120 - DUT Onderdeelnr. 15062781-C4 J | Gebruikershandleiding

Companion 5
Brugerkontrolelementer og indikatorer for systemstatus
Interne symboler

02

Iltudtag

I/OTænd/sluk-knap
Der er ingen servicekrævende deleindeni. Åbn ikke låget.

ISO 7000; grafiske symboler til brug på udstyr –indeks og synopsis
Storage or operating temperature range.
Reg.nr. 0632

Type BF-del (beskyttelsesgrad modelektrisk stød). Reg.nr. 5333
Rådets direktiv 93/42/EØF om medicinske
anordninger
Autoriseret repræsentant i EU
IEC 60417
Klasse II-udstyr, dobbeltisoleret.
Reg.nr. 5172
ETL-certificering
Sikkerhedsagentur for CAN/CSA C22.2
nr. 601.1 M90 for elektromedicinsk
udstyr.

Opbevaring fugtighedsområde. Reg.
nr. 2620
Producentens navn og adresse.
Reg.nr. 3082
Dato for producenten. Reg.nr. 2497
Katalognummer. Reg.nr. 2493
Serienummer. Reg.nr. 2498
Denne side op. Reg.nr. 0623

FCC-erklæring
21 CFR 801.15: Code of Federal Regulations Title 21
Ifølge føderal amerikansk lovgivningmå dette apparat kun sælges af en lægeeller efter ordination af en læge.
Europa-Parlamentets og Rådets direktiv
2012/19/EU: om affald af elektrisk og elektronisk
udstyr (WEEE)
WEEE

Skrøbelig – skal håndteres forsigtigt.
Reg.nr. 0621
Må ikke udsættes for regn. Opbevares
tørt. Reg.nr. 0626
Læs brugervejledningen inden
betjening. Reg.nr. 1641
Stabelgrænse efter antal. Reg. nr. 2403.
ISO 7010: Grafiske symboler – sikkerhedsfarverog -skiltning – registreret sikkerhedsskiltning
Brugervejledningen skal læses.
Reg.nr. M002
Advarsel! Reg.nr. W001
Må ikke udsættes for åben ild, brandeller gnister. Åben antændelseskildeog rygning er ikke tilladt. Reg.nr. P003
Der må ikke ryges i nærheden af
enheden eller under drift. Reg.nr. P002

IEC 60601-1: Elektromedicinsk udstyr
Del 1 Generelle sikkerhedskrav og væsentlige
funktionskrav

IPX1

Dryptæt udstyr-IPX1: Companion 5yder beskyttelse mod de skadeligevirkninger ved indtrængning af væsker.
(IPX1, pr. IEC 60529)

FCC-ERKLÆRING:

Denne enhed kan indeholde CYBLE-022001-00,
herunder antennen 2450AT18B100 fra Johnson
Technology – overholder del 15 af FCC-reglerne.
Denne enhed opfylder kravene for godkendelse af
modulsendere i henhold til FCC's offentlige erklæring
DA00-1407. Betjening af senderen er underlagt
følgende to betingelser: (1) Denne enhed må ikkeforårsage skadelig interferens, og (2) Denne enhed skal
acceptere al interferens, den måtte modtage, herunderinterferens der kan påvirke betjeningen negativt.
Dette produkt er muligvis omfattet af ét eller flereamerikanske eller internationale patenter. Gå til vores
websted nedenfor for at se en liste over gældende
patenter. Pat.: www.caireinc.com/corporate/patents/

PN 15062781-C4 J | Brugervejledning

121 - DANCompanion 5
Startvejledning
1 Pak Companion 5 ud

Companion 5

2 Lær Companion 5 at kende
Gennemse alle advarsler, forsigtighedsforanstaltninger
og yderligere enhedsoplysninger i resten af denne
vejledning. Sæt dig ind i nøglefunktionerne
i Companion 5 og betjeningspanelet.

FORSIDE
KabinetAfbryderLEDdisplay

Iltudtagsstuds

Timemåler

Flowmåler
Holder til
befugterflaske(flaske valgfri)

BAGSIDEBærehåndtag
og placering af
luftindtag

Placering af
køleluftindtag

Udluftningshuller
Placering af
luftindtag

122 - DAN PN 15062781-C4 J | BrugervejledningCompanion 5
Companion 5-komponenterne
KabinetAfbryderUdtagsstuds

LED-display

Timemåler

Flowmåler

Holder til
befugterflaske(flaske valgfri)

Slidstærk plastickasse, som
omslutter og beskytter de indvendige
komponenter i CAIRE Companion 5.
Tænd/sluk-knappen bruges til atstarte eller stoppe strømforsyningen
til enheden. ON-position (tændt)er angivet med symbolet ”l”, og
OFF-position (slukket) er angivet
med symbolet ”O”.
Ilten forlader Companion 5 her,efter at den er blevet filtreretog koncentreret. Der påsættes
en enkeltlumenæsekanlyle eller
standardiltslange til at levere ilt tilpatienten. Den maksimale længde på
iltslanger monteret på koncentratoren
er 50 fod (15,2 m).
Omfatter en grøn, en rød og en gul*
LED-lysindikator. Den grønne lampe
angiver normal drift, og den gule*og den røde angiver alarmtilstande.
(Se afsnittet Alarmtilstande foryderligere oplysninger).
Digital visning af den forløbne
driftstid med koncentratoren.
Viser med en nøjagtighed på entiendedel time og kan ikke nulstilles.
Viser diagnosticeringskoder, når
koncentratoren gennemgår enalarmtilstand.
Bruges til at justere og viseflowhastigheden for den leverede ilt.
Flowhastigheden kan være 0,5-5 literpr. minut (l/min.) og kan indstilles ved
at dreje knappen.
Det sted, en befugterflaske anbringes.
Holderen er udstyret med en elastik,
som fastholder flasken i holderen.
En befugterflaske giver ekstra fugt
til iltflowet. Flasken fyldes med
destilleret vand, og fugten bruges
til at forhindre udtørring af vævet
i næsen.
*Kun modeller med
iltkoncentratorstatusindikator (OCSI)

3 Påsætning af befugterflaske og kanyleHvis der anvendes en befugterflaske, skal den først
fyldes med destilleret vand efter producentens
anvisninger. Anbring flasken i holderen, og fastgør
den med elastikken. Slut den gevindskårne ende af
befugterslangen til flasken og den anden ende til
Companion 5's udtagsstuds. Slut iltslangen og/elleren næsekanyle til udtagsstudsen på befugterflasken.

Hvis du ikke benytter en befugterflaske, skal duslutte iltslangen og/eller en næsekanyle direkte til
udtagsstudsen på Companion 5.

PN 15062781-C4 J | Brugervejledning

123 - DANCompanion 5
4 Tænd og opvarmning

6 Juster flowindstilling til ordineret niveau

Sæt lysnetledningen i en stikkontakt.

Brug flowkontrolknappen til at vælge det flow, lægen
har ordineret.

Sæt afbryderen på On.

AfbryderNår Companion 5 startes korrekt, tændes en grøn
indikator på LED-displayet. Alle LED-lamper
lyser ved start. Når koncentratoren har gennemgået
opvarmningscyklussen, lyser kun den grønne lampe.
Bemærk: Efter først at tænde lyser O2-lampen og ”O2
UP” vises på skærmen, indtil koncentrationen når
specifikationer.

ADVARSEL: DET ER MEGET VIGTIGT AT
VÆLGE DET ORDINEREDE ILTNIVEAU.
UNDGÅ AT ÆNDRE FLOWINDSTILLINGEN,
MEDMINDRE EN AUTORISERET KLINISKARBEJDENDE LÆGE HAR BEDT DIG OM
AT GØRE DET. ILTKONCENTRATOREN KAN
ANVENDES UNDER SØVN, HVIS DETTE
ANBEFALES AF EN KVALIFICERET KLINISKARBEJDENDE LÆGE.
Træk vejret normalt gennem næsekanylen.
Sådan justeres flowhastigheden:
Drej mod uret for at øge hastigheden.
Drej med uret for at sænke hastigheden.
Midten af kuglen angiver flowhastigheden. Se billede
nedenfor.

StrømLED

Rigtigt

124 - DAN PN 15062781-C4 J | BrugervejledningCompanion 5
Vigtigt!

Sikkerhedsinstruktionerne er defineret som følger:
ADVARSEL: VIGTIGESIKKERHEDSOPLYSNINGER IFORBINDELSE MED RISIKO FOR
ALVORLIG PERSONSKADE.
FORSIGTIG: Vigtige oplysninger
om forebyggelse af skader påCompanion 5.
Bemærk: Oplysninger, der kræver særlig
opmærksomhed.

Indikationer for anvendelse
Anvendelsesformål

Companion 5 er beregnet til administration
af supplerende ilt. Enheden er ikke beregnet
som respirator, og den har ingen yderligere
patientovervågningsfunktioner.
ADVARSEL: DET ER MEGET VIGTIGT AT
VÆLGE DET ORDINEREDE ILTNIVEAU.
UNDGÅ AT ÆNDRE FLOWINDSTILLINGEN,
MEDMINDRE EN AUTORISERET KLINISKARBEJDENDE LÆGE HAR BEDT DIG
OM AT GØRE DET. DEN BÆRBARE
ILTKONCENTRATOR KAN ANVENDESUNDER SØVN, HVIS DETTE ANBEFALES
AF EN KVALIFICERET LÆGE.
ADVARSEL: DET SKAL SIKRES, AT COMPANION 5
LEVERER DEN KORREKTE MÆNGDE ILT TIL
BEHANDLING AF DIN LIDELSE, OG DEN MÅ
DERFOR KUN ANVENDES, NÅR EN ELLER FLEREINDSTILLINGER ER FASTLAGT INDIVIDUELTELLER ORDINERET TIL DIG PÅ DINE SPECIFIKKEAKTIVITETSNIVEAUER. COMPANION 5 SKAL
BRUGES MED DEN SPECIFIKKE KOMBINATION AF
DELE OG TILBEHØR, SOM KONCENTRATORENSPRODUCENT HAR ANGIVET, OG SOM BLEV BRUGTTIL AT FASTLÆGGE DINE INDSTILLINGER.
ADVARSEL: DENNE ENHED MÅ IKKE ANVENDESSOM RESPIRATOR. ÆLDRE, BØRN ELLER ANDRE
PATIENTER, DER IKKE KAN KOMMUNIKEREUBEHAG, KAN KRÆVE YDERLIGERE OVERVÅGNING
OG/ELLER ET DISTRIBUERET ALARMSYSTEM, DER
KAN FORMIDLE OPLYSNINGER OM UBEHAG OG/
ELLER MEDICINSKE BEHOV TIL DEN ANSVARLIGE
PLEJER FOR AT UNDGÅ SKADE. PATIENTER MED
NEDSAT HØRELSE OG/ELLER SYN KAN HAVE BEHOVFOR HJÆLP TIL OVERVÅGNING AF ALARMER.

ADVARSEL: GRAVIDE OG AMMENDE KVINDERBØR IKKE BRUGE DET TILBEHØR, DER ANBEFALESI DENNE VEJLEDNING, DA DE KAN INDEHOLDEFTALATER.
ADVARSEL: IFØLGE NATIONAL LOVGIVNING I USA
MÅ DENNE ENHED KUN SÆLGES ELLER UDLEJESEFTER HENVISNING FRA EN LÆGE ELLER ANDET
AUTORISERET SUNDHEDSPERSONALE.

Kontraindikationer foranvendelse
ADVARSEL: UNDER VISSE FORHOLDKAN DET VÆRE FARLIGT AT ANVENDE
IKKE-ORDINERET ILT. ENHEDEN BØR KUN
ANVENDES EFTER ORDINATION FRA ENLÆGE.
ADVARSEL: MÅ IKKE ANVENDES I NÆRHEDEN AF
BRÆNDBARE ANÆSTESIMIDLER.
ADVARSEL: SOM DET ER TILFÆLDET MED
ALLE ELEKTRISKE ENHEDER, KAN BRUGERENOPLEVE PERIODER UDEN DRIFT SOM FØLGE
AF STRØMAFBRYDELSER ELLER BEHOV FOR, AT
COMPANION 5 SERVICERES AF EN KVALIFICERETTEKNIKER. COMPANION 5 ER IKKE VELEGNET
TIL PATIENTER, DER VILLE FÅ ALVORLIGE
SUNDHEDSMÆSSIGE FØLGER AF EN SÅDAN
MIDLERTIDIG STRØMAFBRYDELSE.

Sikkerhedsretningslinjer
ADVARSEL: DET ER IKKE TILLADT AT
MODIFICERE UDSTYRET.
ADVARSEL: PRODUCENTEN ANBEFALER, AT DU
HAR EN ALTERNATIV KILDE TIL SUPPLERENDE ILTI TILFÆLDE AF STRØMUDFALD, ALARMTILSTAND
ELLER MEKANISK FEJL. SPØRG DIN LÆGEELLER UDSTYRSLEVERANDØR, HVILKEN TYPERESERVESYSTEM DER ER PÅKRÆVET.
ADVARSEL: DENNE ENHED LEVERER EN HØJ
ILTKONCENTRATION, DER FREMMER HURTIGFORBRÆNDING. RYGNING ELLER BRUG AFÅBEN ILD ER IKKE TILLADT INDEN FOR TO(2) METERS AFSTAND AF (1) DENNE ENHED
ELLER (2) ILTBÆRENDE UDSTYR. MANGLENDEOVERHOLDELSE AF DENNE ADVARSEL KAN
MEDFØRE BRAND, MATERIEL SKADE OG/ELLERFYSISKE SKADER ELLER DØDSFALD.
ADVARSEL: ANVEND IKKE APPARATET I ET
AFLUKKET ELLER BEGRÆNSET RUM (DVS. EN
LILLE TASKE ELLER TASKE), HVOR VENTILATIONEN
KAN VÆRE DÅRLIG. DET KAN MEDFØREOVEROPHEDNING AF ILTKONCENTRATOREN,HVILKET FORRINGER YDEEVNEN.

PN 15062781-C4 J | Brugervejledning

125 - DANCompanion 5
ADVARSEL: KONCENTRATOREN SKAL
PLACERES PÅ GOD AFSTAND AF RØG,FORURENENDE STOFFER ELLER GASSER.
ADVARSEL: BRUGEN AF VISSE TYPERILTADMINISTRATIONSUDSTYR, DER IKKEER SPECIFICERET TIL BRUG MED DENNE
ILTKONCENTRATOR, KAN PÅVIRKE YDEEVNEN.
ANBEFALET TILBEHØR ER ANGIVET I NÆRVÆRENDEVEJLEDNING.
ADVARSEL: HVIS ILTKONCENTRATOREN BLIVERTABT, BESKADIGET ELLER UDSÆTTES FOR VAND,
SKAL DU KONTAKTE HJEMMEPLEJEN FOR AT FÅ
DEN UNDERSØGT OG EVENTUELT REPARERET.
UNDGÅ AT BRUGE ILTKONCENTRATOREN, HVISLEDNINGEN ELLER STIKKET ER BESKADIGET.
ADVARSEL: SØRG FOR, AT LUFTINDTAGET OG
LUFTDYSERNE IKKE BLIVER BLOKERET. UNDGÅ AT
TABE ELLER INDSÆTTE GENSTANDE I ENHEDENSÅBNINGER. DET KAN MEDFØRE OVEROPHEDNING
AF ILTKONCENTRATOREN, HVILKET FORRINGERYDEEVNEN.
ADVARSEL: UNDGÅ AT OVERFYLDE DET EKSTRABEFUGTNINGSAPPARAT. FYLD KUN VAND PÅ DET
EKSTRA BEFUGTNINGSAPPARAT TIL DET NIVEAU,DER ER ANGIVET AF PRODUCENTEN.
ADVARSEL: UNDGÅ AT BRUGE UDSTYRET
UDEN FØRST AT HAVE LÆST OG FORSTÅET
DENNE VEJLEDNING. HVIS DU IKKE FORSTÅRADVARSLERNE OG INSTRUKTIONERNE, SKAL DU
KONTAKTE LEVERANDØREN AF UDSTYRET, FØR DU
FORSØGER AT BRUGE DET. I MODSAT FALD KAN
DET MEDFØRE SKADER.
ADVARSEL: HVIS DU FØLER DIG UTILPAS ELLER
BEFINDER DIG I EN MEDICINSK NØDSITUATION,SKAL DU STRAKS SØGE LÆGE.
ADVARSEL: HVIS ILTKONCENTRATOREN BENYTTESUDEN FOR DE ANGIVNE DRIFTSTEMPERATURER,
KAN DET BEGRÆNSE KONCENTRATORENSEVNE TIL AT OPFYLDE SPECIFIKATIONERNEFOR ILTKONCENTRATION. DU KAN SETEMPERATURGRÆNSERNE I AFSNITTET OM
SPECIFIKATIONER I DENNE VEJLEDNING.
ADVARSEL: UNDGÅ AT BENYTTE OLIE,
FEDTSTOF ELLER PETROLEUMSBASEREDE
ELLER ANDRE BRÆNDBARE PRODUKTERSAMMEN MED ILTBÆRENDE TILBEHØR ELLER
ILTKONCENTRATOREN. KUN VANDBASEREDEPRODUKTER, ILTKOMPATIBLE LOTIONER ELLERSALVER BØR ANVENDES. ILT FREMSKYNDERFORBRÆNDING AF BRÆNDBARE STOFFER.
ADVARSEL: ILTKONCENTRATOREN BØR IKKEBRUGES VED SIDEN AF ELLER STABLET MED ANDETUDSTYR. HVIS DET ER NØDVENDIGT AT ANVENDE
DEN PLACERET VED SIDEN AF ELLER STABLET, SKAL
ENHEDEN OVERVÅGES FOR AT SIKRE KORREKTDRIFT.

126 - DAN PN 15062781-C4 J | Brugervejledning

ADVARSEL: PRODUCENTEN ANBEFALER,
AT DU HAR EN ALTERNATIV KILDETIL SUPPLERENDE ILT I TILFÆLDE AFSTRØMUDFALD, ALARMTILSTAND
ELLER MEKANISK FEJL. SPØRG DINLÆGE ELLER UDSTYRSLEVERANDØR,HVILKEN TYPE RESERVESYSTEM DER
ER PÅKRÆVET.
ADVARSEL: DENNE ENHED LEVERER EN HØJ
ILTKONCENTRATION, DER FREMMER HURTIGFORBRÆNDING. RYGNING ELLER BRUG AF ÅBENILD ER IKKE TILLADT I SAMME RUM SOM (1)DENNE ENHED ELLER (2) ILTBÆRENDE UDSTYR.
MANGLENDE OVERHOLDELSE AF DENNE ADVARSEL
KAN MEDFØRE BRAND, MATERIEL SKADE OG/ELLERFYSISKE SKADER ELLER DØDSFALD.
ADVARSEL: BRUGEN AF VISSE TYPERILTADMINISTRATIONSUDSTYR, DER IKKEER SPECIFICERET TIL BRUG MED DENNE
ILTKONCENTRATOR, KAN PÅVIRKE YDEEVNEN.
ANBEFALET TILBEHØR ER ANGIVET I NÆRVÆRENDEVEJLEDNING.
ADVARSEL: HVIS ILTKONCENTRATOREN BLIVERTABT, BESKADIGET ELLER UDSÆTTES FOR VAND,
SKAL DU KONTAKTE HJEMMEPLEJEN FOR AT FÅ
DEN UNDERSØGT OG EVENTUELT REPARERET.
UNDGÅ AT BRUGE ILTKONCENTRATOREN, HVISLEDNINGEN ELLER STIKKET ER BESKADIGET.
ADVARSEL: BRUG KUN RESERVEDELE, SOM ER
ANBEFALET AF PRODUCENTEN, FOR AT SIKRE
KORREKT DRIFT OG UNDGÅ RISIKO FOR BRAND
OG FORBRÆNDINGER.
ADVARSEL: UNDGÅ AT SMØRE FITTINGS,FORBINDELSER, SLANGER OG ANDET TILBEHØR TIL
ILTKONCENTRATOREN FOR AT UNDGÅ RISIKO FOR
BRAND OG FORBRÆNDINGER.

Companion 5
ADVARSEL: RYGNING UNDER BRUG AF ILT ER
DEN STØRSTE ÅRSAG TIL BRANDULYKKER OG
RELATEREDE DØDSFALD. DU SKAL OVERHOLDEFØLGENDE SIKKERHEDSADVARSLER:ADVARSEL: UNDGÅ RYGNING, LEVENDE LYS ELLERÅBEN ILD I SAMME RUM SOM ENHEDEN ELLERILTBÆRENDE TILBEHØR.
ADVARSEL: RYGNING UNDER BRUG AFNÆSEKANYLE KAN UDLØSE FORBRÆNDINGI ANSIGTET OG RISIKO FOR DØDSFALD.
ADVARSEL: HVIS DU RYGER, KAN FØLGENDE
3 TRIN MULIGVIS REDDE DIT LIV: SLUKILTKONCENTRATOREN, TAG KANYLEN AF, OG
FORLAD DET LOKALE, HVOR ENHEDEN ER
PLACERET.
ADVARSEL: DER SKAL VÆRE TYDELIG SKILTNING
MED ”RYGNING FORBUDT – ILTAPPARAT I BRUG”
I HJEM, HVOR DER ANVENDES ILTAPPARATER.
PATIENTER OG PLEJERE SKAL INFORMERES OM
FARERNE VED AT RYGE I NÆRHEDEN AF ELLER
UNDER BRUG AF MEDICINSK ORDINERET ILT.
ADVARSEL: UNDGÅ AT BENYTTE
ILTKONCENTRATOREN I NÆRHEDEN AF
BRÆNDBARE GASSER. DET KAN MEDFØRE BRAND
OG MEDFØLGENDE RISIKO FOR TINGSKADE,PERSONSKADE OG DØDSFALD.
ADVARSEL: HVIS DU TAGER NÆSEKANYLEN AF
OG ANBRINGER DEN PÅ TØJ, SENGETØJ, MØBLERELLER ANDET POLSTRET MATERIALE, UDLØSES ENEKSPLOSIONSAGTIG BRAND, HVIS DEN KOMMER
I FORBINDELSE MED CIGARETTER, VARMEKILDER
ELLER FLAMMER.
ADVARSEL: UNDGÅ AT EFTERLADE NÆSEKANYLEN
PÅ ELLER UNDER TØJ, SENGETØJ ELLER PUDER.
HVIS ENHEDEN TÆNDES UDEN AT VÆRE I BRUG,KAN ILTEN GØRE MATERIALET BRÆNDBART.
INDSTIL I/0-AFBRYDEREN PÅ 0 (SLUKKET), NÅR
ILTKONCENTRATOREN IKKE ER I BRUG.

ADVARSEL: KUN TILBEHØR, DER ER ANBEFALET
AF PRODUCENTEN. ENHVER ANDEN BRUG KAN
VÆRE FARLIG, MEDFØRE ALVORLIGE SKADER PÅ
ILTKONCENTRATOREN OG OPHÆVE GARANTIENSDÆKNING.
FORSIGTIG: Tag altidvekselstrømforsyningen ud
af stikkontakten, før du tager
vekselstrømforsyningen fra
iltkoncentratoren.
FORSIGTIG: Anbring altid iltslanger og ledninger på
en sådan måde, at man ikke kan falde i dem.
FORSIGTIG: Følg alle producentens
anvisninger for ikke at ophæve
garantien.
Bemærk: Bærbart og mobilt RF-kommunikationsudstyr
kan påvirke elektrisk medicinsk udstyr.
ADVARSEL: LÆS ALTID OXYGENFORSYNINGSLEDNINGEN OG STRØMKABLERNE I EN MAND, DER
FORHINDRER FARE FOR FARE ELLER MULIGVISTILFÆLDIG FORVIRRING.
ADVARSEL: DETTE PRODUKT KAN UDSÆTTE
DIG FOR KEMIKALIER, HERUNDER NIKKEL, DER I
DELSTATEN CALIFORNIEN ER KENDT FOR AT VÆREKRÆFTFREMKALDENDE. DU KAN FÅ YDERLIGERE
OPLYSNINGER PÅ WWW.P65WARNINGS.CA.GOV.
ADVARSEL: HVIS DER OPSTÅR EN ALVORLIGHÆNDELSE MED DENNE ENHED, SKAL BRUGERENSTRAKS RAPPORTERE HÆNDELSEN TIL UDBYDEREN OG / ELLER PRODUCENTEN. EN ALVORLIGHÆNDELSE DEFINERES SOM EN SKADE, DØD ELLERPOTENTIALE TIL AT FORÅRSAGE SKADE / DØD, HVISDER ER EN GENTAGELSE AF HÆNDELSEN. BRUGEREN KAN OGSÅ RAPPORTERE HÆNDELSEN TIL
DEN KOMPETENTE MYNDIGHED I DET LAND, HVOR
HÆNDELSEN FANDT STED.

ADVARSEL: UNDGÅ AT UDFØRE ANDEN
VEDLIGEHOLDELSE END ANGIVETI DENNE VEJLEDNING. UNDGÅ AT FJERNEDÆKSLER ELLER UDFØRE SERVICE PÅ
ENHEDEN. DET BØR KUN GØRES AF DINHJEMMEPLEJER ELLER EN KVALIFICERETSERVICETEKNIKER.

PN 15062781-C4 J | Brugervejledning

127 - DANCompanion 5
Specifikationer
Følgende diagram indeholder vigtige oplysninger omde anbefalede driftsmiljøer, eller driftsforhold, således
at enheden kan anvendes korrekt.

Specifikationer
FlowhastighederO2-koncentrationMål

0,5-5,0 l/min.
90 % (+5,5 %/-3 %)54,6x31,8x34,2 cm(21,5x12,5x13,5”)
Vægt
16,3 kg (36 lbs)
Lydtrykniveau
50 dB(A) ved 2,0 l/min.
Strømforbrug
285 W ved 2 l/min.,
350 W maks.
Driftstemperatur
41º F til 104º F (5 ° til 40 °C)
Driftsluftfugtighed
15% - 90 % ved et dugpunkt
på 28 °C (82,4 °F)
Opbevaringstemperatur -25 °C til 70 °C(-13 °F til 158 °F)
Fugtighed ved
0% - 90% ikkeopbevaring
kondenserende
ElektriskUndgå at brugeforlængerledninger. Undgå
at bruge elektriske udtag,
der styres af en kontakt.
Driftshøjde-1253-9879 fod(-382-3011 meter)
Driftstid
op til 24 timer i døgnet
Udstyrets forventede levetid er minimum fem år.
ADVARSEL: ANVENDELSE AF
ENHEDER UDEN FOR SPECIFICEREDE
DRIFTSBETINGELSER FORVENTES
AT PÅVIRKE STRØMMEN OG
PROCENTDELEN AF ILT OG FØLGELIGKVALITETEN AF TERAPIEN NEGATIVT.
Bemærk: Hvis iltkoncentratoren har været
opbevaret i længere tid uden for dens normale
driftstemperaturområde, skal enheden først opnånormal driftstemperatur, før den tændes. (Se afsnittet
Specifikationer i denne brugervejledning).

128 - DAN PN 15062781-C4 J | Brugervejledning

Korrekt placering af Companion 5Vælg en placering til enheden uden røg, dampe og
forurenende stoffer. Korrekt placering af enheden
bør muliggøre indsugning af luften igennem detre luftindtagssteder øverst på bagsiden og underkabinettet og gør det ligeledes muligt for luften frit atforlade udluftningshullerne på den nederste venstre delaf enheden.
Luftindtagene er placeret øverst bag på og nederst på
koncentratoren. Anbring enheden, så der er mindst
12 tommer (30 cm) afstand mellem koncentratorenog eventuelle vægge, møbler, gardiner eller andre
hindringer.
Anbring enheden, så det er muligt at høre alarmerne.
Anbring iltforsyningsslangerne, så de ikke bøjes eller
lukkes.
ADVARSEL: UNDGÅ AT BENYTTE
ILTKONCENTRATOREN I NÆRHEDEN
AF BRÆNDBARE GASSER. DET KAN
MEDFØRE BRAND OG MEDFØLGENDERISIKO FOR TINGSKADE, PERSONSKADE
OG DØDSFALD. UNDGÅ AT BENYTTE OLIE,
FEDTSTOF ELLER PETROLEUMSBASEREDE
ELLER ANDRE BRÆNDBARE PRODUKTERSAMMEN MED ILTBÆRENDE TILBEHØRELLER ILTKONCENTRATOREN. KUN
VANDBASEREDE PRODUKTER,
ILTKOMPATIBLE LOTIONER ELLER SALVER
BØR ANVENDES. ILT FREMSKYNDERFORBRÆNDING AF BRÆNDBARESTOFFER.
ADVARSEL: DENNE ENHED LEVERER EN HØJ
ILTKONCENTRATION, DER FREMMER HURTIGFORBRÆNDING. RYGNING ELLER BRUG AF ÅBENILD ER IKKE TILLADT I SAMME RUM SOM (1)DENNE ENHED ELLER (2) ILTBÆRENDE UDSTYR.
MANGLENDE OVERHOLDELSE AF DENNE ADVARSEL
KAN MEDFØRE BRAND, MATERIEL SKADE OG/ELLERFYSISKE SKADER ELLER DØDSFALD.

Companion 5
BrugsanvisningFør brug

Anvend denne brugervejledning som reference
i forbindelse med brug og vedligeholdelse af enheden.
Hvis du har spørgsmål eller problemer, skal du
kontakte hjemmeplejen.
Vigtigt! Forsøg IKKE at anvende Companion 5 uden
først at have læst sikkerhedsretningslinjerne i denne
brugervejledning. Følg alle driftsinstruktioner. Overholdalle advarsler på enheden og i brugervejledningen. Allesikkerhedsretningslinjer skal overholdes for at reducere
risikoen for brand, personskade og alvorlig beskadigelse
af Companion 5.
CAIRE Companion 5 producerer ikke selv ilt.
Den producerer stærkt koncentreret ilt ud fra den
omgivende luft og leverer den til dig. Omgivende
luft indeholder kun 21 % ilt.
Den omgivende luft trækkes ind i koncentratoren af
en kompressor. Luften bevæger sig derefter igennem
en række filtre og molekylsi, som kemisk absorbererkvælstofmolekyler. Derfor indeholder den iltberigede
luft, som forlader koncentratoren, 87-95 % ilt.

Trin 1: Anbringelse af Companion 5 til brugAnbring Companion 5 i et område med god
luftgennemstrømning og god belysning. Kontrollér,
at luftindtaget og udluftningshullerne ikke er blokeret.
Anbring Companion 5, så det er nemt at høre og sealle hørbare og synlige indikatorer eller alarmer.
Slut enheden til en stikkontakt.
Trin 2: Hvis befugterflasken IKKE vil være i brug

Placering af
luftindtag

Placering af
køleluftindtag
Udluftningshuller

Placering af
luftindtag

a. Slut en næsekanyle eller iltslange direkte til
iltudtagsstudsen.

ADVARSEL: RENGØR KUN KABINET,
BETJENINGSPANEL OG LEDNING MEDET MILDT RENGØRINGSMIDDEL PÅ EN
FUGTIG (IKKE VÅD) KLUD ELLER SVAMP,
OG AFTØR ALLE OVERFLADER. UNDGÅ,AT DER KOMMER VÆSKE I ENHEDEN.
VÆR ISÆR OPMÆRKSOM PÅ, AT
ILTUDTAGET TIL KANYLETILSLUTNINGENER FRI FOR STØV, VAND OG PARTIKLER.
FORSIGTIG: Sørg for, at luftindtaget
og luftdyserne ikke bliver blokeret.
Det kan medføre overophedningaf iltkoncentratoren, hvilket
forringer ydeevnen. Undgå atbruge iltkoncentratoren, uden at
luftindtagsfilteret er isat. Hvis der er
medleveret et ekstra filter, skal dette”reservefilter” isættes, før det snavsede
filter rengøres. Rengør det snavsede
filter med varmt vand og sæbe, og tør
det grundigt før brug.

b. Gå videre til trin 4.
Trin 3: Hvis befugterflasken VIL være i bruga. Skru låget af befugterflasken.
b. Fyld flasken med destilleret vand. Sørg for, atvandstanden er mellem producentens specificeredemaksimum og minimum som angivet på flasken.
c. Skru låget på befugterflasken. Kontrollér, at den ikke
er utæt.
d. Slut næsekanylen eller standardiltslangen til
udtagsstudsen på befugterflasken.
Bemærk: Hvis der anvendes en befugter, som ikke er
specificeret til brug med denne koncentrator, kan detpåvirke ydeevnen.

PN 15062781-C4 J | Brugervejledning

129 - DANCompanion 5g. Anbring flasken i befugterflaskeholderen.
h. Fastgør flasken i holderen med elastikken.

e. Slut den genvindskårne ende af befugterslagen til
befugterflaskens låg.

i. Gå videre til trin 4.
Trin 4: Tænd og opvarmninga. Sæt CAIRE Companion 5's elledning i en
stikkontakt.
ADVARSEL: EFTERSE ELLEDNINGEN FORSKADER FØR BRUG. HVIS LEDNINGEN ER
BESKADIGET, MÅ DEN IKKE SÆTTES I ENSTIKKONTAKT, OG DU MÅ IKKE FORSØGE
AT BENYTTE KONCENTRATOREN.

f. Slut den anden ende af befugterslangen til
udtagsstudsen på CAIRE Companion 5.
b. Sæt afbryderen på ”ON” (I). Når Companion 5
startes korrekt op, tændes en grøn indikatorpå LED-displayet. Alle LED-lamper lyser
ved start. Når koncentratoren har gennemgået
opvarmningscyklussen, lyser kun den grønne lampe.
Bemærk: Når enheden er tændt, skal den ståi op til ti (10) minutter, mens den registrerer
ydelsesspecifikationer.

130 - DAN PN 15062781-C4 J | BrugervejledningCompanion 5
ADVARSEL: DET ER MEGET VIGTIGT AT
VÆLGE DET ORDINEREDE ILTNIVEAU.
UNDGÅ AT ÆNDRE FLOWINDSTILLINGEN,
MEDMINDRE EN AUTORISERET KLINISKARBEJDENDE LÆGE HAR BEDT DIG OM
AT GØRE DET. ILTKONCENTRATOREN KAN
ANVENDES UNDER SØVN, HVIS DETTE
ANBEFALES AF EN KVALIFICERET KLINISKARBEJDENDE LÆGE.
Trin 6: Kontrollér flow, og træk vejret normalt
Trin 5: Juster flowkontrolhastighed
a. Drej flowkontrolknappen til den iltflowhastighed(l/min.), din læge har ordineret.

a. Kontrollér, at der kommer ilt gennem næsekanylen,og at der ikke er nogen knæk, bøjninger eller
blokeringer i slangen.
Bemærk: Sørg for, at kanylen er helt indsat og
sidder fast. Du skal kunne høre og føle iltflow pånæsekanylen.
b. Anbring næsekanylen korrekt, og træk vejretnormalt gennem næsekanylen.
Trin 7: Sluka. Sæt afbryderen på ”OFF” (O), når CAIRE
Companion 5 ikke længere er i brug.

Sådan justeres flowhastigheden:
Drej mod uret for at øge hastigheden.
Drej med uret for at sænke hastigheden.
Midten af kuglen angiver flowhastigheden. Billedet
nedenfor indikerer en gennemstrømningshastighed på
4,0 l/min.

FORSIGTIG: Anvend kun tilbehør, der eranbefalet af producenten. Enhver anden
brug kan være farlig, medføre alvorlige
skader på iltkoncentratoren og ophæve
garantiens dækning.
FORSIGTIG: Brug ikke forlængerledninger til
denne enhed, og tilslut ikke for mange stik i densamme stikkontakt. Brugen af f​ orlængerledninger
kan påvirke enhedens ydeevne. Alt for mange
stik i den samme stikkontakt kan resultere i en
overbelastning på el-panelet, så afbryderen/
sikringen aktiveres eller går i brand, hvis
afbryderen eller sikringen ikke fungerer.
FORSIGTIG: Brug af andre kabler og adaptereend dem, der er angivet, med undtagelse af
kabler og adaptere, som producenten af den
elektriske medicinske anordning sælger som
reservedele til interne komponenter, kan medføre
forøget emission af reduceret immunitet fra
iltkoncentratoren.

Rigtigt

ADVARSEL: SØRG FOR, AT
LUFTINDTAGET OG LUFTDYSERNEIKKE BLIVER BLOKERET. DET KAN
MEDFØRE OVEROPHEDNING AF
ILTKONCENTRATOREN, HVILKETFORRINGER YDEEVNEN.

PN 15062781-C4 J | Brugervejledning

131 - DANCompanion 5Bemærk: Til udstyrsudbyder: Følgende tilbehør til
iltadministration anbefales til brug sammen med
Companion 5:• Nasal kanyle: CAIRE Delnummer CU002-1• Luftfugteradapterrør: CAIRE Varenummer 20843882• Luftfugterflaske: CAIRE-delenummer HU003-1• Firebreak: CAIRE-delenummer 20629671
En brandbryd kræves til brug med enhver kanyle.
• CAIRE tilbyder en brandbryd beregnet til brugsammen med iltkoncentratoren. Firebreak er entermisk sikring for at stoppe strømmen af gas itilfælde af, at nedstrøms kanyle eller iltrør antændes
og brænder til ildstedet. Det anbringes på linje
med næsekanylen eller iltrøret mellem patienten
og iltudløbet fra Companion 5. For korrekt brugaf firebreak skal du altid henvise til producentens
instruktioner (inkluderet i hvert firebreak kit).
• For yderligere anbefalet tilbehør, seTilbehørskataloget (PN MLLOX0010) tilgængeligt påwww.caireinc.com.

132 - DAN PN 15062781-C4 J | BrugervejledningCompanion 5
Alarmtilstande
CAIRE Companion 5 bruger en kombination af toeller tre (for modeller med OCSI) LED-lamper og en
lydalarm for at fortælle dig, når der er funktionsfejl på
koncentratoren.
1) Grøn LED () – angiver normal drift.

2) Rød LED ( ) – angiver systemfejl. Se
Alarmtilstandstabel nedenfor for nærmere
oplysninger.
3) Gul LED ( ) – angiver lav iltkoncentration.
Se Alarmtilstandstabel nedenfor for nærmere
oplysninger.
Alarmtilstandstabellen på næste side viser deforskellige alarmtilstande, som kan vises af
koncentratoren. Den opridser de mulige årsager, og
hvad du skal gøre, hvis du får en alarm. Hvis du får enalarm, anbefales det, at du kontakter hjemmeplejen så
hurtigt som muligt.

Grøn

Rød

Gul

Ovenfor: Companion 5

ADVARSEL: IGNORER IKKE ALARMERADVARSLER: PRODUCENTEN ANBEFALER,
AT DU HAR EN ALTERNATIV KILDETIL SUPPLERENDE ILT I TILFÆLDE AFSTRØMUDFALD, ALARMTILSTAND ELLER
MEKANISK FEJL. SPØRG DIN LÆGE ELLERUDSTYRSLEVERANDØR, HVILKEN TYPERESERVESYSTEM DER ER PÅKRÆVET.

PN 15062781-C4 J | Brugervejledning

133 - DANCompanion 5
Alarmtilstandstabel
Lydalarm

Farvet
LED

Fra
Grøn

Årsag
Koncentratoren fungerer
korrekt.

Du skal gøre følgende
Intet

Koncentratoren har mistet
strøm, men afbryderen stårstadig på ”ON”.
RØDKontrollér, at koncentratorens elledning sidder i en
stikkontakt, og at der er strøm på stikkontakten. Prøv
en anden stikkontakt. Hvis problemet ikke er løst, skal
du koble dig på reserveiltforsyningen og kontakte
hjemmeplejen.

Funktionsfejl på systemetRØDSæt omgående afbryderen på ”OFF”. Tag ledningen
ud af stikkontakten. Vent 5 minutter. Sæt ledningeni stikkontakten igen, og tænd koncentratoren igen.
Hvis alarmen varer ved, skal der udføres service påenheden. Tilslut dig en reserveiltforsyning, og kontakt
hjemmeplejen omgående.

Produktflowhastighed
for høj eller for lav

1. Kontrollér, at kanylen ikke er krøllet eller blokeret. Hvis
der bruges en befugtningsflaske, skal du kontrollere, atden er korrekt påfyldt og ikke skaber en blokering.
2. Kontrollér, at Companion 5 har korrekt ventilation.
Der skal være mindst 30 cm (12 tommer) fra alle
overflader for at sikre, at udluftningen ikke er blokeret.
3. Hvis problemet varer ved, skal du skifte til en alternativ
iltkilde og kontakte hjemmeplejen for at få hjælp.

Sporadisk

Sporadisk

Sporadisk
RØDCompanion 5 harregistreret lave iltniveauer.

Sporadisk
Gul

Companion 5 harregistreret lave iltniveauer.

SporadiskGul

134 - DAN PN 15062781-C4 J | Brugervejledning

1. Sørg for, at luftindtagsfilteret og udsugningsstederne
ikke er tilstoppede eller begrænsede. 2. Sørg for, atCompanion 5 befinder sig i et velventileret område. Sørg
for, at der er mindst 30 cm (12 tommer) mellem bagsiden
af Companion og genstande (møbler, gardiner osv.).
3. Hvis der stadig opstår problemer, skal du skifte til enalternativ iltforsyning og straks kontakte hjemmeplejen.

Companion 5
Fejlfinding
Det anbefales, at du kontakter hjemmeplejen, hvis
koncentratoren viser nogen alarmer, eller hvis du ikke
synes, at koncentratoren fungerer korrekt. Du må ikke
forsøge at reparere koncentratoren selv. Reparationer
må kun udføres af en uddannet servicetekniker. Derer dog nogle fejlfindingstrin, du kan gennemgå, hvis
du har problemer med CAIRE Companion 5. De
er beskrevet i følgende tabel. Du kan bruge denne
tabel ved at følge kontrollerne for dit problem i dennummererede rækkefølge.

Tabel til brugerfejlfinding
Problem

Koncentratorentændes men
arbejder ikke.

Du opleverlavt eller
intet iltflow.

Mulig årsag

Du skal gøre følgende

1. Elledningen er ikke sat i en
stikkontakt.

1. Sæt elledningen i stikkontakten.

2. Elledningen giver ikke strøm.

2. Kontrollér sikringerne og HFI-relæet. Prøv en anden
stikkontakt.

3. Intern fejl.

3. Slut dig til en reserveiltforsyning. Kontakt
hjemmeplejen omgående.

1. Flowkontrolindstillingen er ændret.

1. Kontrollér, at flowkontrolknappen er indstillet til det
antal liter pr. minut (l/min.), din læge har ordineret.

2. Slangen er blevet koblet fra.

2. Kontroller, at slangen er tæt ved alle tilslutninger(udtagsstuds, befugterflaske, vandopsamler osv.). Fastgør
tilslutningerne, hvis det er nødvendigt.

3. Der er knæk på iltslangen eller
kanylen.

3. Kontrollér, om der er knæk eller blokeringer i slangen.
Reparer om nødvendigt.

4. Der er en utæthed i kanylen ellerslangen.

4. Efterse slange og kanyle for utætheder. Det kan dugøre ved at lytte eller mærke efter udsivende luft. Udskiftutæt slange eller kanyle.

5. Luftflowet til koncentratoren er
nedsat.

5. Kontrollér, at koncentratorens indtag ikke er blokeret.
Kontrollér, at der er mindst 12 tommer (30 cm) afstandfra møbler, vægge eller andre hindringer.

6. Intern fejl.

6. Slut dig til en reserveiltforsyning. Kontakt
hjemmeplejen omgående.

Bemærk: Hvis Companion 5 stadig er i alarmtilstand, efter at du har anvendt alarmtilstands- og fejlfindingstabellerne,skal du kontakte teknisk service på telefon 1-800-482-2473 og opgive den alarmkode, som er vist på LCD-displayet.

PN 15062781-C4 J | Brugervejledning

135 - DANCompanion 5
Rengøring, pasning og rutinevedligeholdelse
Rutinevedligeholdelse

Rutinevedligeholdelse består i at udskifte håndtagets
grovpartikelfilter én gang om ugen (hvis det er
installeret).

Brugervedligeholdelse og rengøring
af enheden
Udskiftning af kanyle
Udskift forsyningsslangen og kanylen jævnligt
efter hjemmeplejens anbefaling. Din læge eller
hjemmehjælper informerer dig om rengøring,desinfektion og udskiftning.

Bemærk: Følg altid kanyleproducentens vejledning forkorrekt brug. Udskift engangskanylen som anbefalet
af kanyleproducenten eller udstyrsleverandøren.
Reservedele kan bestilles hos udstyrsleverandøren.

Luftindsugningsfilter (kassette)
Luftindsugningsfilteret bør kun åbnes eller udskiftes af
din sundhedsplejeudbyder.
Grovpartikelfilter (ekstraudstyr):

Bruttopartikelfilteret er placeret helt på bunden af
Companion 5.

Grovpartikelfilter

Du bør inspicere og rengøre den en gang om ugen.
Brug følgende fremgangsmåde for at rengøre
filteret:1) Fjern grovpartikelfilteret.
2) Vask filteret i varmt vand fra hanen med en mildsæbeopløsning.
3) Skyl filteret grundigt med rent vand fra hanen, og
klem overskydende vand fra.
4) Lad filteret lufttørre.

136 - DAN PN 15062781-C4 J | Brugervejledning5) Indsæt filteret i kabinettet igen.
Bemærk: Producenten anbefaler ikke sterilisering af
udstyret.
Bemærk: Undgå at bruge iltkoncentratoren, uden at
luftindtagsfilteret er isat.
Befugterflaske (hvis det er relevant)Du skal kontrollere befugterflasken dagligt for atsikre, at vandstanden befinder sig mellem det angivne
minimums- og maksimumniveau. Fyld flasken
med destilleret vand, når det er nødvendigt. Rengør
befugterflasken efter producentens anvisninger.
Rengøringsprocedure:
SLUK Companion 5, og tag ledningen ud af
stikkontakten før rengøring eller desinfektion. Du
MÅ IKKE sprøjte direkte på kabinettet. Brug en
fugtig (ikke våd) klud eller svamp. Spray kluden
eller svampen med et mildt rengøringsmiddel
for at rengøre kabinettet og strømforsyningen.
Desinficer Companion 5 ved brug af Lysol® Brand
II (eller tilsvarende). Fortynd rengøringsmidlet efter
producentens anvisninger, men sprøjt ikke væske
direkte på Companion 5.
ADVARSEL: FARE FOR ELEKTRISKSTØD. SLUK FOR ENHEDEN, OG TRÆKSTRØMKABLET UD AF STIKKONTAKTEN,FØR DU RENGØR ENHEDEN, FOR AT
FORHINDRE UTILSIGTET ELEKTRISKSTØD OG FORBRÆNDINGER. FJERNELSE
AF DÆKSLER ELLER SERVICE PÅ
ENHEDEN MÅ KUN UDFØRES AF
UDSTYRSLEVERANDØREN ELLER ENKVALIFICERET SERVICETEKNIKER.
ADVARSEL: FARE FOR ELEKTRISK STØD. SLUKFOR ENHEDEN, OG TRÆK STRØMKABLET UD AFSTIKKONTAKTEN, FØR DU RENGØR ENHEDEN,
FOR AT FORHINDRE UTILSIGTET ELEKTRISK STØDOG FORBRÆNDINGER. FJERNELSE AF DÆKSLER
ELLER SERVICE PÅ ENHEDEN MÅ KUN UDFØRES AF
UDSTYRSLEVERANDØREN ELLER EN KVALIFICERETSERVICETEKNIKER.

Companion 5
ADVARSEL: FARE FOR ELEKTRISK STØD. TAG
LEDNINGEN UD AF STIKKONTAKTEN, FØR
DU RENGØR ENHEDEN, FOR AT FORHINDRERISIKOEN FOR ELEKTRISK STØD OG
FORBRÆNDING. FJERNELSE AF DÆKSLER ELLERSERVICE PÅ ENHEDEN MÅ KUN UDFØRES AF
UDSTYRSLEVERANDØREN ELLER EN KVALIFICERETSERVICETEKNIKER.

ADVARSEL: RENGØR KUN KABINET,
BETJENINGSPANEL OG LEDNING MED ET MILDTRENGØRINGSMIDDEL PÅ EN FUGTIG (IKKE VÅD)KLUD ELLER SVAMP, OG AFTØR ALLE OVERFLADER.
UNDGÅ, AT DER KOMMER VÆSKE I ENHEDEN.
VÆR ISÆR OPMÆRKSOM PÅ, AT ILTUDTAGET TIL
KANYLETILSLUTNINGEN ER FRI FOR STØV, VAND
OG PARTIKLER.

ADVARSEL: VÆR OMHYGGELIG MED AT UNDGÅ,AT ILTKONCENTRATOREN BLIVER VÅD, ELLER AT
DER KOMMER VÆSKE IND I ENHEDEN. DETTE KANMEDFØRE FUNKTIONSFEJL ELLER NEDLUKNINGSAMT ØGET RISIKO FOR ELEKTRISK STØD ELLERFORBRÆNDINGER.

ADVARSEL: UNDGÅ AT BRUGE VÆSKE DIREKTEPÅ ENHEDEN. LISTEN OVER UØNSKEDE KEMISKESTOFFER OMFATTER, MEN ER IKKE BEGRÆNSET
TIL, FØLGENDE: ALKOHOL OG ALKOHOLBASEREDEPRODUKTER, KONCENTREREDE KLORINBASEREDEPRODUKTER (ÆTYLENKLORID) SAMT OLIEBASEREDEPRODUKTER (PINE-SOL®, LESTOIL®). DISSE MÅ IKKEANVENDES TIL RENGØRING AF PLASTICKABINETTETPÅ ILTKONCENTRATOREN, DA DE KAN BESKADIGEENHEDENS PLASTICMATERIALE.

ADVARSEL: DET KAN VÆRE USIKKERT AT
SAMMENKOBLE SELSKAB 5 MED ANDET UDSTYR.

Bortskaffelse
Aflever altid Companion 5, inkl. alle komponenter,til din hjemmehjælper. Du kan også kontakte den
lokale myndighed angående instruktioner om korrektbortskaffelse af batteriet.

WEEE og RoHS

Vores produkter overholder begrænsningerne i RoHSdirektivet. De indeholder ikke mere end spor af bly
eller andre farlige stoffer.
FORSIGTIG: For korrekt bortskaffelse
skal du kontakte din udstyrsudbyder
eller det lokale myndigheds agentur for
instruktioner om bortskaffelse.

Dette symbol har til formål at minde ejere
af udstyr om at returnere det til et genvindingsanlæg,når det er udtjent, i henhold til direktivet om affald af
elektrisk og elektronisk udstyr (WEEE).

CAIRE og CAIRE Inc. er registrerede varemærker tilhørende
CAIRE Inc. Besøg vores websted herunder for en fuld liste overvaremærker.
Varemærker: www.careinc.com/corporate/trademarks.
Copyright © 2020 CAIRE Inc.
CAIRE Inc. forbeholder
sig retten til at indstille produktionen af sine produkter eller ændre
priser, materialer, udstyr, kvalitet, beskrivelser, specifikationer
og/eller processer i forhold til sine produkter til enhver tid uden
forudgående varsel og uden yderligere forpligtelse eller konsekvens.
Alle rettigheder, der ikke udtrykkeligt er angivet heri, er forbeholdtos, hvad der er relevant.
PN 15062781-C4 J | Brugervejledning

137 - DANCompanion 5
Brukerkontroller og indikatorer for systemstatus
Interne symboler

02

Anvendt del type BF (grad av
beskyttelse mot elektrisk støt).
Reg. nr. 5333

Oksygenutløp

I/O På/Av-bryter
Ingen innvendige deler som kan
repareres. Dekslet må ikke åpnes.

Rådsdirektiv 93/42/EØF vedrørende medisinsk
utstyrAutorisert representant i EU
IEC 60417
Klasse II utstyr, dobbeltisolert
reg. nr. 5172

ISO 7000; Grafiske symboler til bruk på spesielt
utstyr – indeks og sammendrag
Lagring eller driftstemperaturområde.
Reg. nr. 0632

ETL-sertifisering
Sikkerhetsbyrå for CAN/CSA
C22.2 nr. 601.1 M90 for medisinsk
elektrisk utstyr

Oppbevaring fuktighetsområde. Reg.
nr. 2620
Produsentens navn og adresse.
Reg. nr. 3082
Produksjonsdato. Reg. nr. 2497
Katalognummer. Reg. nr. 2493
Serienummer. Reg. nr. 2498

FCC-melding
21 CFR 801.15: Samling av offentlige forskrifter
seksjon 21
Amerikansk føderal lovgivningbegrenser denne enheten til salg fra
en lege eller foreskrevet av en lege.
Rådsdirektiv 2012/19/EF: avfall fra elektrisk og
elektronisk utstyr (WEEE)

Denne siden opp. Reg. nr. 0623
Glass – håndteres med forsiktighet.
Reg. nr. 0621

WEEE

Må holdes unna regn – oppbevares tørt.
Reg. nr. 0626

IEC 60601-1: Medisinsk elektrisk utstyr
Del 1 Generelle krav for grunnleggende
sikkerhet og nødvendig ytelse

Les bruksanvisningen før bruk.
Reg. nr. 1641

IPX1

Stabelgrense etter antall. Reg. nr. 2403.
ISO 7010: Grafiske symboler – Sikkerhetsfarger og
sikkerhetsskilter – Registrerte sikkerhetsskilt
Instruksjonsmanualen må leses.
Reg. nr. M002
Advarsel. Reg. nr. W001
Holdes unna åpen flamme, brann,
gnister. Åpne kilden til tenning og
røyking forbudt. Reg. nr. P003
Røyking forbudt i nærheten av enheten
eller mens enheten brukes.
Reg. nr. P002
138 - NOR PN 15062781-C4 J | Bruksanvisning

Dryppsikkert utstyr-IPX1:
Companion 5 beskytter mot skader
som følge av vanninntrenging.
(IPX1, i henhold til IEC 60529)

FCC-MELDING:

Denne enheten kan inneholde CYBLE-022001-00,
inkludert antennen 2450AT18B100 fra Johnson
Technology, som samsvarer med par. 15 i FCC‑reglene.
Enheten oppfyller kravene til godkjenning av modulære
sendere, som beskrevet i FCCs offentlige melding
DA00-1407. Drift av senderen er underlagt to følgendevilkår: (1) denne enheten kan ikke forårsake skadelig
interferens, og (2) denne enheten må tåle enhver mottatt
interferens, inkludert interferens som kan forårsake
uønsket drift.
Dette produktet kan være dekket av ett eller flereamerikanske eller internasjonale patenter. Gå til vår
hjemmeside nedenfor for en liste med gjeldende
patenter. Pat.: www.caireinc.com/corporate/patents/.

Companion 5
Hurtigstartveiledning
1 Pakk ut Companion 5

Companion 5

2 Bli kjent med din Companion 5
Se gjennom alle advarsler, forsiktighetsregler og annen
informasjon om enheten videre i denne veiledningen.
Gjør deg kjent med de viktigste funksjonene og
brukerkontrollpanelet til Companion 5.

FRONT
Det ytredekseletStrømbryter

LEDskjerm

OksygenOksygenuttak

Timeteller

Strømningsmåler
Stativ forluftfukterflaske(flaskevalgfritt)

BAKSIDEBærehåndtak og
luftinntakssted

Kjølende
luftinntakssted

Utluftingsventil
Luftinntakssted

PN 15062781-C4 J | Bruksanvisning

139 - NOR

Companion 5
Komponentene til Companion 5

3 Fest luftfukterflasken og kanylen

Det ytre dekselet Holdbart plastdeksel som omslutter
og beskytter de indre komponentene
til CAIRE Companion 5.
Strømbryter
PÅ/AV-bryteren brukes til å starteeller stoppe strømtilførselen til
enheten. Posisjonen PÅ er markert
med symbolet «I», og posisjonen AV
er markert med symbolet «O».
Oksygenuttak Oksygenet kommer ut av
Companion 5 her etter at det har
blitt filtrert og konsentrert. Det følger
enten med en enkellumen nesekanyleeller en standard oksygenslange, somskal leveres til pasienten. Maksimallengde på oksygenslangen som følger
med konsentratoren er ca. 15,2 meter(50 fot).
LED-skjerm
Inneholder et grønt, rødt og gult*
LED-lys. Det grønne lyset indikerer
normal drift, og det gule* og røde
lyset indikerer alarmtilstander.
(Se delen «Alarmtilstander»i brukerveiledningen for mer
informasjon.)
Timeteller
Digital visning av den tilbakelagte
driftstiden til konsentratoren.
Viser til nærmeste tiendedel av
en time og kan ikke tilbakestilles.
Viser diagnostiske alarmkodernår konsentratoren opplever en
alarmtilstand.
Strømningsmåler Brukes til å justere og vise
strømningshastigheten av oksygen
som blir levert. Strømningshastigheter
varierer fra 0,5 til 5 liter per minutt(LPM) og kan justeres ved å vri på
bryteren.
Sted til å feste en luftfukterflaske.
Stativ forluftfukterflaske På stativet henger det et elastisk(flaske valgfritt) bånd som brukes til å feste flasken
til stativet. Luftfukterflasken tilfører
ekstra fuktighet til oksygenstrømmen.
Flasken er fylt med destillert vannog fuktigheten brukes til å forhindre
uttørking av nesevevet.
*Kun for modellene av typen Oxygen
Concentrator Status Indicator (OCSI)

Hvis du bruker en luftfukterflaske, fyll først flasken
med destillert vann til det nivået produsenten har
spesifisert. Plasser flasken i det angitte stativet og
fest med det elastiske båndet. Koble den gjengede
enden av luftfukterslangen til flasken, og den andre tiloksygenuttaket på Companion 5. Koble oksygentuben
og/eller en nesekanyle til oksygenuttaket på
luftfukterflasken.

140 - NOR PN 15062781-C4 J | Bruksanvisning

Hvis du ikke bruker en luftfukterflaske, koble
oksygentuben og/eller en nesekanyle direkte tiloksygenuttaket til Companion 5.

Companion 5
4 Slå på og varm opp
Strømledningen (AC) kobles til en stikkontakt (AC).

6 Juster innstillingene foroksygentilførsel til foreskrevet nivåBruk kontrollbryteren for oksygentilførsel for å velge
strømningen foreskrevet av legen din.

Sett strømbryteren i posisjonen «på».

Strømbryter

Når Companion 5 er riktig tilkoblet lyser den grønne
indikatoren på LED-skjermen. Alle LED-lamper
lyser ved oppstart. Etter at konsentratoren har fullført
oppvarmingssyklusen, kommer bare det grønne lyset
til å være på.
Merk: Etter at du først har slått på, lyser O2-lampenog ”O2 UP” vises på skjermen til konsentrasjonen når
spesifikasjoner.

ADVARSEL: DET ER VELDIG VIKTIG Å KUN
VELGE DET FORESKREVNE NIVÅET AV
OKSYGEN. VALG AV OKSYGENTILFØRSELSKAL IKKE ENDRES MED MINDRE DU
HAR FÅTT BESKJED AV AUTORISERTHELSEPERSONELL OM Å GJØRE DET.
OKSYGENKONSENTRATOREN KANBRUKES NÅR MAN SOVER ETTERANBEFALING FRA KVALIFISERTHELSEPERSONELL.
Pust normalt gjennom nesekanylen.
For å justere strømningshastigheten av oksygen:
Vri mot klokken for å øke strømningshastigheten.
Vri med klokken for å redusere strømningshastigheten.
Midten på ballen indikerer strømningshastighet.
Se bildet nedenfor.

StrømindikatorLED

Korrekt

PN 15062781-C4 J | Bruksanvisning

141 - NOR

Companion 5
Viktig!

Sikkerhetsinstruksjoner defineres slik:
ADVARSEL: VIKTIGSIKKERHETSINFORMASJON OM FARER
SOM KAN FORÅRSAKE ALVORLIG SKADE.
FORSIKTIG: Viktig informasjon for å
forhindre skade på Companion 5.
Merk: Informasjon som trenger spesiell
oppmerksomhet.

Indikasjoner for bruk
Tiltenkt brukCompanion 5 er ment til å administrere ekstra
oksygen. Enheten er ikke beregnet for å værelivsopprettholdende, og har heller ikke mulighet tilå overvåke en pasient.
ADVARSEL: DET ER VELDIG VIKTIG Å KUN
VELGE DET FORESKREVNE NIVÅET AV
OKSYGEN. VALG AV OKSYGENTILFØRSELSKAL IKKE ENDRES MED MINDRE DU
HAR FÅTT BESKJED AV AUTORISERTHELSEPERSONELL OM Å GJØRE DET. DENBÆRBARE OKSYGENKONSENTRATOREN
KAN BRUKES NÅR MAN SOVER
ETTER ANBEFALING FRA KVALIFISERTHELSEPERSONELL.
ADVARSEL: FOR Å SIKRE AT DU MOTTARDEN TERAPEUTISKE OKSYGENTILFØRSELENI HENHOLD TIL DIN MEDISINSKE TILSTAND,
MÅ COMPANION 5 KUN BRUKES ETTER AT
EN ELLER FLERE INNSTILLINGER BESTEMMESINDIVIDUELT ELLER FORESKRIVES TIL DEG
FOR DINE SPESIFIKKE AKTIVITETSNIVÅER.
COMPANION 5 MÅ BRUKES MED DEN SPESIFIKKEKOMBINASJONEN ELLER DELENE OG TILBEHØRET
SOM ER I TRÅD MED SPESIFIKASJONEN TIL
KONSENTRATORPRODUSENTEN OG SOM BLEBRUKT MENS INNSTILLINGENE BLE BESTEMT.
ADVARSEL: DENNE ENHETEN SKAL IKKE BRUKES
SOM LIVSOPPRETTHOLDENDE. GERIATRISK,PEDIATRISK ELLER ANDRE TILFELLER MED
PASIENTER SOM IKKE KAN KOMMUNISEREUBEHAG, KAN KREVE TILLEGGSOVERVÅKNING
OG/ELLER ET DISTRIBUERT ALARMSYSTEM FOR
Å FORMIDLE INFORMASJONEN OM UBEHAG
OG/ELLER MEDISINSK HAST TIL DEN ANSVARLIGEOMSORGSGIVEREN FOR Å UNNGÅ SKADE.
PASIENTER MED NEDSATT HØRSEL OG/ELLER SYN
KAN HA BEHOV FOR HJELP MED Å OVERVÅKE
ALARMENE.

142 - NOR PN 15062781-C4 J | Bruksanvisning

ADVARSEL: GRAVIDE ELLER AMMENDE KVINNERSKAL IKKE BRUKE TILBEHØRENE I DENNEHÅNDBOKEN. DE KAN INNEHOLDE FTALATER.
ADVARSEL: FØDERAL LOV (I USA) BEGRENSER
DENNE ENHETEN TIL Å SELGES TIL ELLER LEIESUT TIL EN LEGE ELLER ANNEN AUTORISERTLEVERANDØR INNEN HELSESEKTOREN.

Kontraindikasjoner for bruk
ADVARSEL: I NOEN TILFELLER KANBRUK AV OKSYGEN SOM IKKE ER
FORESKREVET VÆRE FARLIG. DENNE
ENHETEN SKAL KUN BRUKES NÅR DET
ER FORESKREVET AV LEGE.
ADVARSEL: MÅ IKKE BRUKES I NÆRHETEN AV
BRANNFARLIGE ANESTESIMIDLER.
ADVARSEL: SOM MED ALT UTSTYR SOM DRIVESPÅ STRØM KAN BRUKEREN OPPLEVE PERIODER
MED DRIFTSAVBRUDD PÅ GRUNN AV BRUDD I
STRØMTILFØRSEL, ELLER BEHOV FOR VEDLIKEHOLDAV COMPANION 5 AV EN KVALIFISERT TEKNIKER.
COMPANION 5 ER IKKE EGNET FOR ENKELTEPASIENTER, SOM VIL KUNNE OPPLEVE NEGATIVEHELSEMESSIGE KONSEKVENSER SOM FØLGE AV
SLIKE MIDLERTIDIGE AVBRUDD.

Retningslinjer for sikkerhet
ADVARSEL: INGEN FORM FOR
MODIFISERING AV DETTE UTSTYRETER TILLATTADVARSEL: PRODUSENTEN ANBEFALER EN
ALTERNATIV OKSYGENKILDE I TILFELLER SOM
FOR EKSEMPEL STRØMUTFALL, ALARMTILSTAND
ELLER MEKANISK FEIL. RÅDFØR DEG MED LEGENDIN ELLER UTSTYRSLEVERANDØR VEDRØRENDE
NØDVENDIG RESERVEUTSTYR.
ADVARSEL: DENNE ENHETEN LEVERER OKSYGEN
AV HØY KONSENTRASJON SOM FREMMER RASKBRENNING. TILLAT ALDRI RØYKING ELLER BRUK AVÅPEN ILD INNENFOR TO (2) METER FRA (1) DENNE
ENHETEN, ELLER (2) NOE TILBEHØR MED OKSYGEN.
DERSOM DENNE ADVARSELEN IKKE FØLGES KAN
DET FØRE TIL ALVORLIG BRANN, MATERIELL SKADE
OG/ELLER PERSONSKADE ELLER DØD.
ADVARSEL: IKKE BRUK ENHETEN I ET AVGRENSET
ELLER TRANGT ROM (F.EKS. EN LITEN KOFFERT,ELLER EN VESKE) HVOR VENTILASJONEN KAN
VÆRE BEGRENSET. DETTE KAN FØRE TIL AT
OKSYGENKONSENTRATOREN BLIR OVEROPPHETETOG YTELSEN SVEKKES.

Companion 5
ADVARSEL: KONSENTRATOREN BØR
PLASSERES SLIK AT DEN IKKE BLIRUTSATT FOR RØYK, FORURENSNINGELLER GASSER.
ADVARSEL: BRUKEN AV NOEN TYPER TILBEHØR
FOR ADMINISTRERING AV OKSYGEN, SOM IKKE ER
SPESIFISERT FOR DENNE KONSENTRATOREN, KAN
SVEKKE DENS YTELSE. ANBEFALT TILBEHØR ER VIST
TIL I DENNE BRUKERVEILEDNINGEN.
ADVARSEL: HVIS OKSYGENKONSENTRATOREN HARFALT NED, BLITT SKADET ELLER VÆRT UTSATT FOR
VANN, KONTAKT HELSEPERSONELL FOR INSPEKSJONELLER EVENTUELL REPARASJON AV ENHETEN.
IKKE BRUK OKSYGENKONSENTRATOREN HVISSTRØMLEDNINGEN ELLER STØPSELET ER SKADET.
ADVARSEL: PASS PÅ AT HVERKEN LUFTINNTAKET
ELLER LUFTUTTAKENE BLOKKERES. IKKESLIPP ELLER FØR NOEN GJENSTANDER INN I
ÅPNINGENE PÅ UTSTYRET. DETTE KAN FØRE TIL AT
OKSYGENKONSENTRATOREN BLIR OVEROPPHETETOG YTELSEN SVEKKES.
ADVARSEL: IKKE OVERFYLL DEN VALGFRIELUFTFUKTEREN. FYLL DEN VALGFRIELUFTFUKTEREN MED VANN KUN OPP TIL DET
NIVÅET SOM PRODUSENTEN AV LUFTFUKTERENHAR ANGITT.
ADVARSEL: IKKE BRUK DETTE UTSTYRETUTEN FØRST Å HA LEST OG FORSTÅTT DENNE
VEILEDNINGEN. DERSOM DU IKKE FORSTÅR ALLEADVARSLENE OG INSTRUKSJONENE, KONTAKTUTSTYRSLEVERANDØREN FØR DU FORSØKER Å
BRUKE DETTE UTSTYRET. HVIS DU IGNORERER
DETTE KAN DET FØRE TIL PERSONSKADE ELLER
MATERIELL SKADE.
ADVARSEL: OPPSØK LEGEHJELP ØYEBLIKKELIG
HVIS DU FØLER UBEHAG ELLER OPPLEVER ENMEDISINSK NØDSITUASJON.
ADVARSEL: BRUK AV OKSYGENKONSENTRATORENVED TEMPERATURER UTENFOR DET SPESIFISERTETEMPERATUROMRÅDET KAN BEGRENSE
KONSENTRATORENS EVNE TIL Å AVGI DENSPESIFISERTE OKSYGENKONSENTRASJONEN. SE
ETTER TEMPERATURGRENSER I AVSNITTET FOR
SPESIFIKASJONER I DENNE BRUKERVEILEDNINGEN.

ADVARSEL: OKSYGENKONSENTRATOREN BØR IKKEBRUKES RETT VED SIDEN AV ELLER OPPÅ ANNETUTSTYR. HVIS DET IKKE ER MULIG Å UNNGÅ DETTE,BØR UTSTYRET OBSERVERES FOR Å SIKRE NORMALDRIFT.
ADVARSEL: PRODUSENTEN ANBEFALER
EN ALTERNATIV OKSYGENKILDE I
TILFELLER SOM FOR EKSEMPELSTRØMUTFALL, ALARMTILSTAND ELLER
MEKANISK FEIL. RÅDFØR DEG MEDLEGEN DIN ELLER UTSTYRSLEVERANDØR
VEDRØRENDE NØDVENDIG
RESERVEUTSTYR.
ADVARSEL: DENNE ENHETEN LEVERER OKSYGEN
AV HØY KONSENTRASJON SOM FREMMER RASKBRENNING. TILLAT ALDRI RØYKING ELLER BRUK AVÅPEN ILD I SAMME ROM SOM (1) DENNE ENHETEN,
ELLER (2) NOE TILBEHØR MED OKSYGEN. DERSOMDENNE ADVARSELEN IKKE FØLGES KAN DET FØRETIL ALVORLIG BRANN, MATERIELL SKADE OG/ELLERPERSONSKADE ELLER DØD.
ADVARSEL: BRUKEN AV NOEN TYPER TILBEHØR
FOR ADMINISTRERING AV OKSYGEN, SOM IKKE ER
SPESIFISERT FOR DENNE KONSENTRATOREN, KAN
SVEKKE DENS YTELSE. ANBEFALT TILBEHØR ER VIST
TIL I DENNE BRUKERVEILEDNINGEN.
ADVARSEL: HVIS OKSYGENKONSENTRATOREN HARFALT NED, BLITT SKADET ELLER VÆRT UTSATT FOR
VANN, KONTAKT HELSEPERSONELL FOR INSPEKSJONELLER EVENTUELL REPARASJON AV ENHETEN.
IKKE BRUK OKSYGENKONSENTRATOREN HVISSTRØMLEDNINGEN ELLER STØPSELET ER SKADET.
ADVARSEL: BRUK KUN RESERVEDELER SOM ER
ANBEFALT AV PRODUSENTEN FOR Å SIKRE RIKTIGFUNKSJON OG FOR Å UNNGÅ FARE FOR BRANN OG
BRANNSKADER.
ADVARSEL: SMØR IKKE BESLAG, TILKOBLINGER,SLANGER ELLER ANNET TILBEHØR TIL
OKSYGENKONSENTRATOREN FOR Å UNNGÅ FARE
FOR BRANN OG BRANNSKADER.

ADVARSEL: IKKE BRUK OLJE, FETT ELLERPETROLEUM-BASERTE, ELLER ANDRE
BRENNBARE PRODUKTER SAMMEN MEDOKSYGENKONSENTRATOREN ELLER TILBEHØRMED OKSYGEN. BARE VANNBASERTE, OKSYGENKOMPATIBLE KREMER ELLER SALVER SKAL BRUKES.
OKSYGEN AKSELERERER FORBRENNINGEN AV
BRENNBARE STOFFER.

PN 15062781-C4 J | Bruksanvisning

143 - NOR

Companion 5
ADVARSEL: RØYKING UNDER BRUK AV
OKSYGEN ER DEN MEST VANLIGE ÅRSAKEN TIL
BRANNSKADER OG RELATERTE DØDSFALL. DU
MÅ FØLGE DISSE SIKKERHETSADVARSLENE:ADVARSEL: TILLAT ALDRI RØYKING, STEARINLYS
ELLER ÅPEN ILD I SAMME ROM SOM DENNE
ENHETEN, ELLER NOE TILBEHØR MED OKSYGEN.
ADVARSEL: Å RØYKE UNDER BRUK AV
OKSYGENKANYLE KAN FORÅRSAKE BRANNSKADEI ANSIKTET OG MULIG DØD.
ADVARSEL: HVIS DU RØYKER KAN DISSE
3 TRINNENE REDDE LIVET DITT: SKRU AV
OKSYGENKONSENTRATOREN, TA UT KANYLEN,
OG GÅ UT AV ROMMET HVOR DENNE ENHETENER PLASSERT.
ADVARSEL: SKILT MED TEKSTEN «RØYKINGFORBUDT – OKSYGEN I BRUK» MÅ VÆRE TYDELIG
OG SYNLIG I HJEMMET, ELLER DER OKSYGENER I BRUK. PASIENTER OG DERES PLEIERE MÅVÆRE INFORMERT OM FARENE VED Å RØYKE
I NÆRHETEN AV MEDISINSK OKSYGEN, ELLER
NÅR MEDISINSK OKSYGEN ER I BRUK.
ADVARSEL: IKKE BRUK OKSYGENKONSENTRATOREN
I NÆRHETEN AV BRANNFARLIGE GASSER. DETTE
KAN RESULTERE I EKSPLOSIV BRANN SOM KAN
FØRE TIL MATERIELL SKADE, PERSONSKADE
ELLER DØD.
ADVARSEL: HVIS KANYLEN BLIR TATT UT OG
LAGT PÅ KLÆR, SOFAER ELLER ANDRE MYKE
MATERIALER NÅR DEN TAS UT, VIL DET KUNNE
FORÅRSAKE ØYEBLIKKELIG BRANN HVIS DETUTSETTES FOR EN SIGARETT, VARMEKILDE ELLERÅPEN ILD.
ADVARSEL: LEGG ALDRI FRA DEG EN NESEKANYLE
PÅ ELLER UNDER KLÆR, SENGETØY ELLER
STOLPUTER. HVIS ENHETEN ER SLÅTT PÅ, MEN
IKKE BRUKES, VIL OKSYGENET GJØRE MATERIALET
BRENNBART. SETT STRØMBRYTEREN I/0 TIL
POSISJON 0 (AV) NÅR OKSYGENKONSENTRATORENIKKE ER I BRUK.
ADVARSEL: FORSØK IKKE NOE ANNETVEDLIKEHOLD ENN DE MULIGELØSNINGENE SOM ER ANGITT I
DENNE BRUKERVEILEDNINGEN.
FJERN ALDRI DEKSLER. BAREUTSTYRSLEVERANDØREN ELLER ENKVALIFISERT SERVICETEKNIKER SKALFJERNE DEKSLENE ELLER REPARERE
DENNE ENHETEN.
ADVARSEL: KUN TILBEHØR ANBEFALT AV
PRODUSENTEN. BRUK AV NOE ANNET KAN
VÆRE FARLIG, FØRE TIL ALVORLIG SKADE PÅ
OKSYGENKONSENTRATOREN OG UGYLDIGGJØRGARANTIEN.

144 - NOR PN 15062781-C4 J | Bruksanvisning

FORSIKTIG: Koble alltid strømtilførselen
fra stikkontakten i veggen før
kontakten på oksygenkonsentratoren
kobles fra.
FORSIKTIG: Plasser alltid slangen for
oksygentilførsel og strømledninger slik at man
unngår å snuble i dem.
FORSIKTIG: Følg alle produsentens
anvisninger for at garantien ikke skal
bli ugyldig.
Merk: Bærbart og mobilt RF-kommunikasjonsutstyr kanpåvirke medisinsk elektrisk utstyr.
ADVARSEL: PLASSER ALLTID OKSYGENTILFØRSELSRØRENE OG STRØMKABLENE I EN MANN SOMFORHINDRER AT FAREN FARES ELLER ER MULIG
TILFELDIG TILFELDIG.
ADVARSEL: DETTE PRODUKTET KAN UTSETTE
DEG FOR KJEMIKALIER, INKLUDERT NIKKEL,
SOM ER KJENT FOR Å FORÅRSAKE KREFT, AV
CALIFORNIA. FOR MER INFORMASJON, BESØK
WWW.P65WARNINGS.CA.GOV.
ADVARSEL: I TILFELLE DET OPPSTÅR EN ALVORLIGHENDELSE MED DENNE ENHETEN, SKAL BRUKEREN
UMIDDELBART RAPPORTERE HENDELSEN TIL
LEVERANDØREN OG / ELLER PRODUSENTEN.
EN ALVORLIG HENDELSE ER DEFINERT SOM ENSKADE, DØD ELLER POTENSIELL FOR Å FORÅRSAKESKADE / DØD DERSOM DET SKULLE OPPSTÅ EN NY
HENDELSE. BRUKEREN KAN OGSÅ RAPPORTERE
HENDELSEN TIL DEN KOMPETENTE MYNDIGHETEN I
LANDET DER HENDELSEN SKJEDDE.

Companion 5
Spesifikasjoner
For riktig bruk av enheten, se følgende diagram med
viktig informasjon om de anbefalte driftsmiljøer og
gjeldende driftsforhold.

Spesifikasjoner
Strømningshastigheter
Oksygenkonsentrasjon
Dimensjoner

0,5–5,0 LPM
90 % (+5,5 %/-3 %)54,6 x 31,8 x 34,2 cm(21,5 x 12,5 x 13,5 in.)
Vekt
16,3 kg (36 lbs)
Lydtrykknivå
50 dB(A) ved 2,0 LPM
Strømforbruk
285 W ved 2 LPM, 350 W
maksimum
Driftstemperatur
5 °C til 40 °C(41 °F til 104 °F)
Luftfuktighet ved drift 15% - 90% ved et 28 °C(82,4 °F) duggpunkt
Oppbevaringstemperatur -25 °C til 70 °C(-13 °F til 158 °F)
Luftfuktighet ved
0% - 90% ikkeoppbevaring
kondenserende
Elektrisk
Ikke bruk skjøteledninger.
Ikke bruk stikkontakter
som kontrolleres med
bryter.
Høyde ved drift-382–3011 meter(-1253 – 9879 fot)
Driftstid
opp til 24 timer i døgnetForventet levetid på utstyret er minst fem år.
ADVARSEL: BRUK AV ENHET UTENFORSPESIFISERTE DRIFTSBETINGELSERFORVENTES Å PÅVIRKE STRØMMEN OG
PROSENTEN AV OKSYGEN OG FØLGELIGKVALITETEN PÅ TERAPIEN NEGATIVT.
Merk: Hvis oksygenkonsentratoren har vært lagret i enlengre periode utenfor området for driftstemperatur, gi
enheten tid til å gå tilbake til normal driftstemperatur
før den slås på. (Se avsnittet «spesifikasjoner» i denne
brukerveiledningen.)Riktig plassering av Companion 5

Velg en plassering av enheten slik at den unngår inntak
av røyk, gasser og forurensende stoffer. Riktig plassering
av enheten burde tillate oksygeninntak gjennom de tre
luftinntakene som er plasser øverst på baksiden og underkabinettet, og lar utslippsluften forlate utluftingsventilen
på venstre side av bunnen på enheten.
Luftinntakene er plassert øverst på baksiden og under
konsentratoren. Plasser enheten slik at det er minst
30 cm (12 in.) plass mellom konsentrator og vegger,
møbler, gardiner eller andre hindringer.
Plasser enheten slik at alarmene kan høres.
Posisjoner slangen for oksygentilførsel slik at den ikke
får knekk eller blokkeres.
ADVARSEL: IKKE BRUKOKSYGENKONSENTRATOREN I
NÆRHETEN AV BRANNFARLIGEGASSER. DETTE KAN RESULTERE IEKSPLOSIV BRANN SOM KAN FØRETIL MATERIELL SKADE, PERSONSKADE
ELLER DØD. IKKE BRUK OLJE, FETT ELLERPETROLEUM-BASERTE, ELLER ANDRE
BRENNBARE PRODUKTER SAMMENMED OKSYGENKONSENTRATOREN
ELLER TILBEHØR MED OKSYGEN.
BARE VANNBASERTE, OKSYGENKOMPATIBLE KREMER ELLER SALVER
SKAL BRUKES. OKSYGEN AKSELERERER
FORBRENNINGEN AV BRENNBARE
STOFFER.
ADVARSEL: DENNE ENHETEN TILFØRER OKSYGEN
AV HØY KONSENTRASJON SOM FREMMER RASKBRENNING. TILLAT ALDRI RØYKING ELLER BRUK AVÅPEN ILD I SAMME ROM SOM (1) DENNE ENHETEN,
ELLER (2) NOE TILBEHØR MED OKSYGEN. DERSOMDENNE ADVARSELEN IKKE FØLGES KAN DET FØRETIL ALVORLIG BRANN, MATERIELL SKADE OG/ELLERPERSONSKADE ELLER DØD.

PN 15062781-C4 J | Bruksanvisning

145 - NOR

Companion 5
Instruksjoner for brukFør drift

Denne brukerveiledningen tjener som din referanse forbruk og vedlikehold av utstyret. Hvis du har spørsmål
eller bekymringer, kontakt helsepersonell.
Viktig! IKKE forsøk å bruke Companion 5 uten først
å ha lest avsnittet om retningslinjer for sikkerhet
i denne veiledningen. Følg alle driftsinstruksjonene.
Gjør deg kjent med alle advarsler på enheten og
i brukerveiledningen. For å redusere risikoen forbrann, personskade og alvorlig skade på Companion 5,
skal du gjøre deg kjent med alle forholdsregler forsikkerhet.
CAIRE Companion 5 lager ikke oksygen. Den
produserer svært konsentrert oksygen fra den
omliggende luften i rommet og leverer den til deg.
Vanligvis inneholder luften i et rom bare et volum pårundt 21 % oksygen.
Romluften suges inn i konsentratoren av en
kompressor. Luften passerer så gjennom en serie
av filtre og en molekylsikt som kjemisk adsorberer
nitrogenmolekyler. Resultatet er oksygenrik luft som
kommer ut av konsentratoren og leveres med en verdi
på 87 % til 95 % oksygen i volum.
ADVARSEL: KABINETT,
KONTROLLPANELET OG STRØMKABELENSKAL KUN RENGJØRES MED ET MILDTRENGJØRINGSMIDDEL SOM PÅFØRES
MED EN FUKTIG (IKKE VÅT) KLUT ELLER
SVAMP, OG DERETTER TØRKES ALLEOVERFLATER. DET MÅ IKKE KOMMEVÆSKE INN I ENHETEN. VÆR SPESIELTNØYE MED OKSYGENUTLØPET FOR
KANYLETILKOBLINGEN SLIK AT DETTE
HOLDES FRI FOR STØV, VANN OG
PARTIKLER.
FORSIKTIG: Pass på at hverken
luftinntaket eller luftuttakene
blokkeres. Dette kan føre tilat oksygenkonsentratoren blir
overopphetet og ytelsen svekkes. Ikke
bruk oksygenkonsentratoren uten at
luftinntaksfilteret er på plass. Hvis
enheten er levert med et ekstra filter,sett inn dette «reservefilteret» før durengjør det skitne filteret. Rengjør det
skitne filteret i varmt vann med såpe,tørk deretter grundig før bruk.

146 - NOR PN 15062781-C4 J | Bruksanvisning

Trinn 1: Gjør Companion 5 klar for bruk
Plasser Companion 5 på et godt ventilert og godt
opplyst sted. Påse at luftinntaket og utluftingen ikke
er blokkert. Plasser Companion 5 slik at alle hørbareog synlige indikatorer eller alarmer lett kan bli sett og
hørt.
Koble enheten til et AC strømuttak.
Trinn 2: Hvis luftfukterflasken IKKE skal brukes

Luftinntakssted

Kjølende
luftinntakssted

Utluftingsventil

Luftinntakssted

a. Koble nesekanylen eller oksygenslangen direkte tiloksygenuttaket.

b. Gå videre til trinn 4.
Trinn 3: Hvis luftfukterflasken SKAL brukes
a. Skru lokket av luftfukterflasken.
b. Fyll flasken med destillert vann. Pass på at vannivået
er mellom produsentens oppgitte maksimum- og
minimumnivå som vist på flasken.
c. Skru lokket på luftfukterflasken godt til. Pass på atdet ikke er noen lekkasjer.
d. Koble en nesekanyle eller en standard
oksygenslange til uttaket på luftfukterflasken.
Merk: Bruk av en luftfukter som ikke er spesifikt lagettil å brukes med denne konsentratoren kan påvirke
dens ytelse.

Companion 5
g. Plasser flasken på det angitte stedet, i stativet for
luftfukterflasken.
h. Fest flasken på stativet med det elastiske båndet.

e. Koble den gjengede enden av luftfukterslangen til
lokket på luftfukterflasken.

i. Gå videre til trinn 4.
Trinn 4: Slå på og varm opp
a. Koble strømledningen til CAIRE Companion 5 tilen AC stikkontakt.
ADVARSEL: KONTROLLERSTRØMLEDNINGEN FOR SKADER FØR
BRUK. HVIS LEDNINGEN ER SKADET,MÅ DU IKKE KOBLE DEN TIL ENSTIKKONTAKT ELLER FORSØKE Å BRUKE
KONSENTRATOREN.

f. Koble den andre enden av luftfukterslangen til
oksygenuttaket på CAIRE Companion 5.

b. Trykk strømbryteren til posisjonen «PÅ» (I). NårCompanion 5 er riktig tilkoblet, lyser den grønne
indikatoren på LED-skjermen. Alle LED-lamper
lyser ved oppstart. Etter at konsentratoren har
fullført oppvarmingssyklusen, kommer bare det
grønne lyset til å være på.
Merk: Etter å ha slått på enheten, gi den ti (10)minutter til å nå sine ytelsesspesifikasjoner.

PN 15062781-C4 J | Bruksanvisning

147 - NOR

Companion 5
ADVARSEL: DET ER VELDIG VIKTIG Å KUN
VELGE DET FORESKREVNE NIVÅET AV
OKSYGEN. VALG AV OKSYGENTILFØRSELSKAL IKKE ENDRES MED MINDRE DU
HAR FÅTT BESKJED AV AUTORISERTHELSEPERSONELL OM Å GJØRE DET.
OKSYGENKONSENTRATOREN KANBRUKES NÅR MAN SOVER ETTERANBEFALING FRA KVALIFISERTHELSEPERSONELL.
Trinn 5: Juster strømningshastigheten av oksygen

Trinn 6: Kontroller strømmen og pust normalt

a. Vri kontrollbryteren for oksygentilførsel til den
strømningshastigheten av oksygen (LPM) som har
blitt foreskrevet av legen din.

a. Kontroller at oksygen faktisk flyter gjennom
nesekanylen, og at det ikke er knekk, bøy eller
blokkeringer i slangen.
Merk: Påse at kanylen sitter helt inne og godt fast. Du
skal høre eller føle oksygentilførselen til de ledendedelene av nesekanylen.
b. Plasser nesekanylen riktig og pust normalt gjennom
nesekanylen.
Trinn 7: Slå av
a. Sett strømbryteren i posisjonen «AV» (O) når
CAIRE Companion 5 ikke lenger er i bruk.

For å justere strømningshastigheten av oksygen:
Vri mot klokken for å øke strømningshastigheten.
Vri med klokken for å redusere strømningshastigheten.
Midten på ballen indikerer strømningshastighet. Bildet
nedenfor indikerer en strømningshastighet på 4,0 LPM.

FORSIKTIG: Bruk kun tilbehør anbefalt
av produsenten. Bruk av noe annet kanvære farlig, føre til alvorlig skade på
oksygenkonsentratoren og ugyldiggjør
garantien.
FORSIKTIG: Ikke bruk skjøteledning til denne
enheten. Koble heller ikke for mange støpsler
i den samme stikkontakten. Bruk av skjøteledningkan ha en negativ effekt på ytelsen til enheten.
For mange støpsler i én stikkontakt kan føre
til overbelastning på panelet som kan føre tilat sikringen går, eller brann hvis sikringen ikke
fungerer som den skal.
FORSIKTIG: Bruk av andre kabler og adaptereenn de som er spesifisert, med unntak av kablerog adaptere som selges av produsenten til det
medisinsk-elektriske utstyret som reservedeler forinterne komponenter, kan føre til økte utslipp av
redusert immunitet for oksygenkonsentratoren.

Korrekt

148 - NOR PN 15062781-C4 J | Bruksanvisning

ADVARSEL: PASS PÅ AT HVERKENLUFTINNTAKET ELLER LUFTUTTAKENEBLOKKERES. DETTE KAN FØRE TIL AT
OKSYGENKONSENTRATOREN BLIROVEROPPHETET OG YTELSEN SVEKKES.

Companion 5
Merk: Til utstyrsleverandør: Følgende tilbehør foroksygenadministrasjon anbefales for bruk sammen
med Companion 5:• Nasal kanyle: CAIRE-delenummer CU002-1• Luftfukteradapterrør: CAIRE Delenummer 20843882• Luftfukterflaske: CAIRE delenummer HU003-1• Brannbane: CAIRE-delenummer 20629671
Det kreves brannbrytning for bruk med eventuell
kanyle.
• CAIRE tilbyr et ildsted som skal brukes sammen
med oksygenkonsentratoren. Brannbrynet er entermisk sikring for å stoppe strømmen av gass itilfelle at nedstrøms kanyle eller oksygenrør tennes
og brenner for brannknappen. Den plasseres på linje
med nesekanylen eller oksygenrøret mellom pasientenog oksygenutløpet til Companion 5. For riktig bruk av
brannkjøringen, se alltid produsentens instruksjoner(følger med hvert brannkjøresett).
• For tilleggsutstyr som anbefales, seTilbehørskatalogen (PN MLLOX0010) tilgjengelig på
www.caireinc.com.

PN 15062781-C4 J | Bruksanvisning

149 - NOR

Companion 5
Alarmtilstander
CAIRE Companion 5 bruker en kombinasjon av toeller tre (for modeller med OCSI) LED-lys og en
hørbar alarm som varsler deg når det er en feil med
konsentratoren.
1) Grønn LED () – Indikerer normal drift.

Grønn

Rød

2) Rød LED ( ) – Indikerer systemfeil. Se tabellen
med alarmtilstander nedenfor for mer informasjon.
3) Gul LED () – Indikerer lav
oksygenkonsentrasjon. Se tabellen med
alarmtilstander nedenfor for mer informasjon.
Tabellen med alarmtilstander på neste side viser deulike alarmtilstandene som kan vises av konsentratoren.
Den skisserer mulige årsaker og hva du bør gjøre hvis
du opplever en slik alarm. Ved en eventuell alarm,
anbefales det at du tar kontakt med helsepersonell såsnart som mulig.
ADVARSEL: IKKE OVERSE ALARMERADVARSLER: PRODUSENTEN ANBEFALER
EN ALTERNATIV OKSYGENKILDEI TILFELLER SOM FOR EKSEMPELSTRØMUTFALL, ALARMTILSTAND ELLER
MEKANISK FEIL. RÅDFØR DEG MEDLEGEN DIN ELLER UTSTYRSLEVERANDØR
VEDRØRENDE NØDVENDIG
RESERVEUTSTYR.

150 - NOR PN 15062781-C4 J | Bruksanvisning

Gul

Over: Companion 5

Companion 5
Tabell for alarmtilstander
Hørbaralarm

Farge
på LED

Av
Grønn

Intermittent
RØDIntermittent

Årsak
Konsentratoren fungerersom den skal.

Kontroller at konsentratorens strømledning er koblet tilen stikkontakt og at det er strøm i stikkontakten. Prøv
en annen stikkontakt. Hvis problemet fortsetter, kan dukoble til en alternativ oksygenkilde og ta kontakt med
helsepersonell.

Systemfeil

Sett strømbryteren i posisjonen «AV» umiddelbart.
Ta ledningen ut av AC stikkontakten. Vent i 5 minutter.
Koble ledningen tilbake i AC stikkontakten og slåpå konsentratoren igjen. Hvis alarmen fortsetter
er vedlikehold nødvendig. Koble til en alternativ
oksygenkilde og ta kontakt med helsepersonell
umiddelbart.

Produktets
strømningshastighet
er for høy eller for lav

1. Sørg for at kanylen ikke er knekt eller blokkert. Hvis
luftfukterflaske brukes, sørg for at den er fylt riktig og
ikke skaper blokkeringer.
2. Sørg for at Companion 5 har tilstrekkelig ventilasjon.
Den må være minst 30 cm (12 in.) fra alle overflaterfor å sikre at luftehullene ikke er blokkert.
3. Hvis problemet vedvarer, bytt til en alternativ
oksygenkilde og kontakt helsepersonell for hjelp.

RØDIntermittent
Gul

Intermittent
Gul

Ingen

Konsentratoren har mistet
strøm, men strømbryteren
er fortsatt i posisjonen
«PÅ».

RØD

Intermittent

Nødvendige tiltak

Companion 5 har
oppdaget lave
oksygennivåer.

Companion 5 har
oppdaget lave
oksygennivåer.

1. Forsikre deg om at luftinntaksfilteret og
utluftingsventilen ikke er tilstoppet eller begrenset.
2. Forsikre deg om at Companion 5 er på et godt
ventilert sted. Sørg for at det er minst 30 cm (12 in.)mellom baksiden og sidene av Companion 5 og enhver
hindring (møbler, gardiner osv.).
3. Hvis tilstanden vedvarer, bytt til en alternativ
oksygenkilde og kontakt helsepersonell umiddelbart.

PN 15062781-C4 J | Bruksanvisning

151 - NOR

Companion 5
Feilsøking
Det anbefales at du tar kontakt med helsepersonellhvis konsentratoren viser eventuelle alarmer, ellerhvis du føler at konsentratoren ikke fungerer som den
skal. Du bør ikke forsøke å reparere konsentratoren
selv. Reparasjon skal kun utføres av en kvalifisert

servicetekniker. Det er imidlertid noen feilsøkingstrinn
du kan ta hvis du opplever problemer med CAIRE
Companion 5. De er beskrevet i den følgende tabellen.
Du kan bruke denne oversikten til å sjekke problemetditt i den fremstilte numeriske rekkefølgen.

Tabell for feilsøking
ProblemKonsentratoren
er skrudd
på, men den
fungerer ikke.

Du oppleverlav eller ingen
strøm av
oksygen.

Mulig årsak

Nødvendige tiltak

1. Strømledningen er ikke koblet tilen stikkontakt.

1. Koble til strømledningen.

2. Stikkontakten gir ikke strøm.

2. Sjekk sikringer og kretser i husstanden din. Prøv enannen stikkontakt.

3. Intern feil.

3. Koble til en alternativ oksygenkilde. Ta kontakt med
helsepersonell umiddelbart.

1. Innstillingene for strømningskontroll
har blitt endret.

1. Kontroller at kontrollbryteren for oksygentilførsel ersatt på riktig innstilling for liter per minutt (LPM) som
foreskrevet av legen din.

2. Slangen er koblet fra.

2. Kontroller at slangen er stram på alle tilkoblinger(oksygenuttaket, luftfukterflasken, vannlåsen osv.).
Sikre alle koblinger på nytt hvis nødvendig.

3. Oksygenslangen eller kanylen haren knekk.

3. Kontroller om det er knekk eller blokkeringer
i slangen. Reparer om nødvendig.

4. Det er en lekkasje i kanyleneller slangen.

4. Inspiser alle slanger og kanylen for lekkasjer. Du kan
gjøre dette enten ved å lytte eller å kjenne etter om luft
slipper ut. Bytt slangen eller kanylen om det er lekkasje.

5. Luftstrøm inn i konsentratoren
er forhindret.

5. Kontroller at ingenting blokkerer innløpene til
konsentratoren. Pass på at det er minst 30 cm (12 in.)mellomrom fra møbler, vegger eller andre hindringer.

6. Intern feil.

6. Koble til en alternativ oksygenkilde. Ta kontakt med
helsepersonell umiddelbart.

Merk: Hvis alarmen fortsatt går på din Companion 5 etter bruk av tabellene for alarmtilstander og feilsøking, kan dukontakte teknisk service på 1-800-482-2473 og oppgi alarmkoden som vises på LCD-skjermen.

152 - NOR PN 15062781-C4 J | Bruksanvisning

Companion 5
Rengjøring, pleie og rutinemessig vedlikehold
Rutinemessig vedlikehold

Rutinemessig vedlikehold innebærer å skifte ut
massepartikkelfilteret til håndtaket en gang i uken (hvisinstallert).

Brukerpleie og rengjøring av enheten

Merk: Produsenten anbefaler ikke sterilisering av
dette utstyret.
Merk: Ikke bruk oksygenkonsentratoren uten at
luftinntaksfilteret er på plass.

Bytte av kanyle
Bytt ut tilførselslange og kanyle regelmessig
som anbefalt av helsepersonell. Din lege eller
helsepersonell vil gi deg informasjon om rengjøring,desinfeksjon og utskifting.

Luftfukterflaske (hvis aktuelt)
Du bør sjekke din luftfukterflaske daglig for å sikre atdens vannstand er mellom det angitte minimums- og
maksimumsnivået. Fyll opp flasken med destillert vannom nødvendig. Rengjør luftfukterflasken i henhold til
produsentens instruksjoner.

Merk: Følg alltid kanyleprodusentens anvisninger
for riktig bruk. Bytt ut engangskanylen som anbefalt
fra kanyleprodusenten eller utstyrsleverandøren.
Ytterligere forsyninger er tilgjengelig hos
utstyrsleverandøren.

Rengjøringsprosedyre:Slå AV Companion 5 og koble fra AC strømmen før
noen form for rengjøring eller desinfisering utføres.
IKKE spray direkte på det ytre dekselet. Bruk en
fuktig (ikke våt) klut eller svamp. Spray på kluten
eller svampen med mildt såpevann for å rengjøre
kabinettet og strømledningen. Bruk Lysol® Brand II
desinfiseringsmiddel (eller likeverdig) til desinfisering
av Companion 5. Fortynn rengjøringsmiddelet som
anvist av produsenten, men ikke spray væske direktepå Companion 5.

Luftinntaksfilter (Patron)
Luftinntaksfilteret burde bare nås eller byttes av
helsepersonell.
Massepartikkelfilter (Valgfritt):

Bruttopartikkelfilteret er plassert helt foran i bunnen av Companion 5.

Massepartikkelfilter

Du bør inspisere og rengjøre den en gang ukentlig.
Slik rengjør du filteret:

ADVARSEL: FARE FOR ELEKTRISKSTØT. SLÅ AV ENHETEN OG KOBLESTRØMKABELEN FRA STIKKONTAKTENFØR DU RENGJØR ENHETEN
FOR Å FORHINDRE ELEKTRISKSTØT OG BRANNFARE. BAREUTSTYRSLEVERANDØREN ELLER ENKVALIFISERT SERVICETEKNIKER SKALFJERNE DEKSLENE ELLER REPARERE
DENNE ENHETEN.
ADVARSEL: FARE FOR ELEKTRISK STØT. SLÅ
AV ENHETEN OG KOBLE STRØMKABELEN FRASTIKKONTAKTEN FØR DU RENGJØR ENHETEN FOR Å
FORHINDRE ELEKTRISK STØT OG BRANNFARE. BAREUTSTYRSLEVERANDØREN ELLER EN KVALIFISERTSERVICETEKNIKER SKAL FJERNE DEKSLENE ELLER
REPARERE DENNE ENHETEN.

1) Fjern massepartikkelfilteret.
2) Vask filteret i varmt vann fra springen med et mildt
rengjøringsmiddel.
3) Skyll filteret grundig med rent vann fra springen og
klem ut overflødig vann.
4) La filteret lufttørke.
5) Sett filteret inn i kabinettet igjen.

PN 15062781-C4 J | Bruksanvisning

153 - NOR

Companion 5
ADVARSEL: FARE FOR ELEKTRISK STØT. KOBLESTRØMKABELEN FRA STIKKONTAKTEN FØR
DU RENGJØR ENHETEN FOR Å FORHINDRE
ELEKTRISK STØT OG BRANNFARE. BAREUTSTYRSLEVERANDØREN ELLER EN KVALIFISERTSERVICETEKNIKER SKAL FJERNE DEKSLENE ELLER
REPARERE DENNE ENHETEN.
ADVARSEL: TA HENSYN FOR Å FORHINDRE AT
OKSYGENKONSENTRATOREN BLIR VÅT. VÆSKER
MÅ IKKE KOMME INN I ENHETEN. DETTE KAN FØRETIL FUNKSJONSFEIL ELLER AT ENHETEN SLÅR SEG
AV, OG FØRER TIL ØKT RISIKO FOR ELEKTRISK STØTELLER BRANNSKADER.
ADVARSEL: DET KAN VÆRE USIKKERT Å
SAMMENKOBLE SELSKAP 5 MED ANNET UTSTYR.

ADVARSEL: KABINETT, KONTROLLPANELET OGSTRØMKABELEN SKAL KUN RENGJØRES MED
ET MILDT RENGJØRINGSMIDDEL SOM PÅFØRES
MED EN FUKTIG (IKKE VÅT) KLUT ELLER SVAMP,OG DERETTER TØRKES ALLE OVERFLATER. DET
MÅ IKKE KOMME VÆSKE INN I ENHETEN. VÆR
SPESIELT NØYE MED OKSYGENUTLØPET FOR
KANYLETILKOBLINGEN SLIK AT DETTE HOLDESFRITT FOR STØV, VANN OG PARTIKLER.
ADVARSEL: IKKE BRUK VÆSKE DIREKTE PÅENHETEN. EN LISTE OVER UØNSKEDE KJEMISKE
MIDLER INKLUDERER, MEN ER IKKE BEGRENSET
TIL, FØLGENDE: ALKOHOL OG ALKOHOLBASERTEPRODUKTER, KONSENTRERTE KLORBASERTEPRODUKTER (ETYLENKLORID), OG OLJEBASERTEPRODUKTER (PINE-SOL®, LESTOIL®). DISSE SKAL
IKKE BRUKES FOR Å RENGJØRE PLASTKABINETTETPÅ OKSYGENKONSENTRATOREN DA DE KAN SKADE
ENHETENS PLAST.

Avhending
Returner alltid Companion 5, med alle komponenter, tilhjemmepleieleverandøren din for riktig avhending. Du
kan også kontakte lokale myndigheter for anvisningom korrekt deponering av batteriet.

WEEE og RoHS

Dette symbolet skal påminne eierne av utstyret om at utstyret skal leveres til et gjenvinningsanlegg
når brukstiden er omme, i henhold til direktiv om
avfall fra elektrisk og elektronisk utstyr (WEEE-direktivet).

Produktene våre skal være i samsvar med begrensningav farlige stoffer (RoHS-direktivet). De vil ikke
inneholde mer enn spormengder av bly eller andre
farlige materialer.
FORSIKTIG: For korrekt avhending,kontakt utstyrsleverandøren din eller
lokale myndigheter for instruksjoner omavhending.

CAIRE og CAIRE Inc. er registrerte varemerker for CAIRE Inc.
Vennligst besøk vår hjemmeside nedenfor for en fullstendig oversikt
over varemerker.
Varemerker: www.careinc.com/corporate/trademarks.
Opphavsrett © 2020 CAIRE Inc.
CAIRE Inc. forbeholder seg
retten til å stanse produktene sine, eller endre priser, materialer, utstyr,kvalitet, beskrivelser, spesifikasjoner og/eller prosesser for produktene
sine når som helst, uten forvarsel og uten videre forpliktelser eller
konsekvenser. Alle rettigheter som ikke er uttrykkelig nevnt her, er
forbeholdt oss, som anvendelig.
154 - NOR PN 15062781-C4 J | Bruksanvisning

Companion 5
Kontrollpanel och indikatorer för systemstatus
Interna symboler

02

Syrgasutgång

I/O

Av-/På-brytare
Innehåller inga reparerbara delar.
Öppna inte höljet.

ISO 7000: Grafiska symboler för användningpå maskiner och utrustningar – register och
sammanställning
Lagring eller driftstemperaturområde.
Reg.nr 0632

Patientansluten del av typ BF (grad av
skydd mot elektrisk stöt). Reg.nr 5333
Rådets direktiv 93/42/EEG om medicintekniska
produkter
Auktoriserad representant i den
Europeiska gemenskapen
IEC 60417
Klass II-utrustning, dubbelisolerad reg.
nr 5172
ETL-certifiering

Lagringsfuktighetsområde. Reg.nr 2620

Säkerhetsbyrå för CAN/CSA C22.2
nr 601.1 M90 gällande elektrisk
utrustning för medicinskt bruk

Tillverkarens namn och adress.
Reg.nr 3082
Tillverkningsdatum. Reg.nr 2497
Katalognummer. Reg.nr 2493
Serienummer. Reg.nr 2498
Denna sida upp. Reg.nr 0623
Ömtåligt, hanteras varsamt.
Reg.nr 0621

FCC-meddelande
21 CFR 801.15: Code of Federal Regulations (CFR)
titel 21
Enligt amerikansk federal lag får denna
anordning endast säljas av läkare ellerpå läkares anmodan.
Rådets direktiv 2012/19/EU: avfall som utgörs
av eller innehåller elektrisk och elektronisk
utrustning (WEEE)
WEEE

Skydda den mot regn, håll den torr.
Reg.nr 0626
Läs bruksanvisningen före användning.
Reg.nr 1641
Staplingsgräns efter antal. Reg. nr.
2403.
ISO 7010: Grafiska symboler – säkerhetsfärger
och säkerhetssymboler – registreradesäkerhetssymboler
Bruksanvisningen måste läsas.
Reg.nr M002
Varning. Reg.nr W001
Håll på avstånd från öppna lågor, eld
och gnistor. Öppna antändningskällor
och rökning förbjudet. Reg.nr P003
Rök inte nära enheten eller när duanvänder den. Reg.nr P002

IEC 60601-1: Elektrisk utrustning för medicinskt
bruk – Del 1: Allmänna fordringar beträffandesäkerhet och väsentliga prestanda

IPX1

Droppsäker utrustning-IPX1:
Companion 5 ger skydd mot deskadliga effekterna av inträngande
vätskor. (IPX1, för IEC 60529)

FCC-MEDDELANDE:

Denna enhet kan innehålla CYBLE-022001-00,
inklusive antennen 2450AT18B100 från Johnson
Technology, och överensstämmer med del 15 av FCCreglerna. Enheten uppfyller kraven för godkännande av
modulära sändare enligt FCC:s offentliga meddelandeDA00-1407. Användning av sändaren gäller enligt
följande villkor: 1) Denna enhet får inte orsaka
skadliga störningar och 2) denna enhet måste klara
eventuell mottagen störning, inklusive störningar som
kan orsaka oönskad drift.
Den här produkten kan omfattas av ett eller flera
amerikanska eller internationella patent. Besök vår
webbplats nedan för att se en lista över tillämpliga
patent. Pat.: www.caireinc.com/corporate/patents/.
PN 15062781-C4 J | Bruksanvisning

155 - SWE

Companion 5
Snabbstartsguide1 Packa upp din Companion 5

Companion 5

2 Lär känna din Companion 5
Läs alla varningar, försiktighetsuppmaningar ochövrig information om enheten i resten av denna
bruksanvisning. Bekanta dig med nyckelfunktionerna
på Companion 5 och dess användarkontrollpanel.

FRAMSIDA
Ytterhölje

Strömbrytare

LEDdisplay

Syrgasuttagsnippel

Timmätare

Flödesmätare
Stödställ för
fuktfilterflaska(flaska valfritt
tillbehör)

BAKSIDA
Bärhandtag
och luftintag

Kylluftsintag

Luftutsläpp
Luftintag

156 - SWE

PN 15062781-C4 J | Bruksanvisning

Companion 5
Komponenterna på Companion 5
Ytterhölje

Strömbrytare

Uttagsnippel

LED-display

Timmätare

Flödesmätare

Stödställ för
fuktfilterflaska(flaska valfritt
tillbehör)Hållbart plasthölje som innesluter och
skyddar de inre komponenterna på
CAIRE Companion 5.
Av-/På-strömbrytare för att slå på
och stänga av strömförsörjningentill enheten. PÅ-positionen indikeras
med symbolen ”l” och AV-positionen
indikeras med symbolen ”O”.
Syrgasen kommer ut ur Companion 5
här efter att ha filtrerats och
koncentrerats. För att försörja
patienten med syrgas ansluts
antingen en enkel lumennäskanyl eller
standardsyrgasslangar. Den maximala
längden hos de koncentratoranslutna
syrgasslangarna är 15,2 meter (50 fot).
Har en grön, en röd och en gul*
LED-lampa. Grönt ljus indikerar
normal drift och det gula* och det
röda ljuset indikerar larmtillstånd.
(Se avsnittet Larmtillstånd i den
här bruksanvisningen för mer
information.)
Digital visning av koncentratorns
förbrukade driftstid. Visar till
närmaste tiondel av en timme och
kan inte återställas. Visar diagnostiska
felkoder när larmtillstånd uppstår på
koncentratorn.
Används för att reglera ochvisa flödeshastigheten hossyrgasförsörjningen. Flödeshastigheter
varierar mellan 0,5 och 5 liter per
minut (LPM) och kan regleras genom
att man vrider på knappen.
Plats för anslutning av en
fuktfilterflaska. Stället är försett med
en elastisk rem som ska användas
för att säkra flaskan på stället. En
fuktfilterflaska tillför syrgasflödet
extra fukt. Flaskan är fylld med
destillerat vatten och fukten används
för att förhindra uttorkning av
näsvävnad.
*Endast på
syrgaskoncentratormodeller med
statusindikator (OCSI)

3 Fäst fuktfilterflaskan och kanylen
Vid användning av fuktfilterflaska, fyll först flaskan
med destillerat vatten till den nivå som anvisas av
tillverkaren. Placera flaskan på det anvisade stödstället
och säkra den med den elastiska remmen. Anslut den
gängade änden av fuktfilterslangen till flaskan och
den andra till uttagsnippeln på Companion 5. Anslut
syrgasslangar och/eller en näskanyl till uttagsnippeln
på fuktfilterflaskan.

Om ingen fuktfilterflaska används ska syrgasslangarna
och/eller näskanylen anslutas direkt till uttagsnippeln
på Companion 5.

PN 15062781-C4 J | Bruksanvisning

157 - SWE

Companion 5
4 Sätt på enheten och låt den värma upp
Koppla in växelströmsladden i ett växelströmsuttag.

6 Justera flödesinställningar tillordinerad nivåAnvänd flödesregleringsknappen för att välja det flöde
som ordinerats av din läkare.

Slå över strömbrytaren till På-positionen.

Strömbrytare

VARNING: DET ÄR MYCKET VIKTIGTATT ENBART STÄLLA IN DEN
ORDINERADE NIVÅN AV SYRE. ÄNDRA
INTE FLÖDESVALET OM DU INTE
HAR BLIVIT UPPMANAD ATT GÖRA
DET AV EN KVALIFICERAD KLINIKER.
SYRGASKONCENTRATORN KAN
ANVÄNDAS NÄR MAN SOVER SÅLÄNGE DET SKER PÅ EN KVALIFICERADKLINIKERS REKOMMENDATION.

När Companion 5 har kommit igång tänds den gröna
strömindikatorn på LED-displayen. Alla LED-lamporlyser vid start. Efter det att koncentratorn har fullgjort
uppvärmningscykeln kommer bara den gröna lampan
att fortsätta lysa.

Andas normalt genom näskanylen.

Obs! O2-lampan tänds först när O2 UP-lampan tänds
och ”O2 UP” visas på skärmen tills koncentrationen
når specifikationerna.

För att reglera flödeshastighet:
Vrid motsols för att öka flödet.
Vrid medsols för att minska flödet.
Bollens mitt indikerar flödeshastigheten. Se bilden
nedan.

StrömindikatorLED

Korrekt

158 - SWE

PN 15062781-C4 J | Bruksanvisning

Companion 5
Viktigt!

Säkerhetsinstruktioner definieras enligt följande:
VARNING: VIKTIGSÄKERHETSINFORMATION FÖRFAROR SOM KAN ORSAKA ALLVARLIGPERSONSKADA.
VAR FÖRSIKTIG: Viktig information för
att förhindra skada på Companion 5.
Obs! Särskild uppmärksamhet bör ägnas åt
informationen.

Indikationer för användning
Avsedd användning

Companion 5 är avsedd för extra tillförsel av
syre. Enheten är inte avsedd för livsuppehållandeverksamhet och den fungerar inte sompatientövervakning.
VARNING: DET ÄR MYCKET VIKTIGTATT ENBART STÄLLA IN DEN
ORDINERADE NIVÅN AV SYRE. ÄNDRA
INTE FLÖDESVALET OM DU INTE HARBLIVIT UPPMANAD ATT GÖRA DET
AV EN KVALIFICERAD KLINIKER. DEN
PORTABLA SYRGASKONCENTRATORNKAN ANVÄNDAS NÄR MAN SOVER SÅLÄNGE DET SKER PÅ EN KVALIFICERADKLINIKERS REKOMMENDATION.
VARNING: FÖR ATT SÄKERSTÄLLA ATT DUERHÅLLER DEN TERAPEUTISKA MÄNGDENSYRGAS ENLIGT DITT MEDICINSKA TILLSTÅND FÅR
COMPANION 5 ENDAST ANVÄNDAS NÄR EN ELLER
FLERA INSTÄLLNINGAR HAR FASTSTÄLLTS ELLER
ORDINERATS INDIVIDUELLT FÖR DIG UTIFRÅN DINA
SPECIFIKA AKTIVITETSNIVÅER. COMPANION 5MÅSTE ANVÄNDAS MED DEN SPECIFIKA
KOMBINATIONEN AV DELAR OCH TILLBEHÖR
SOM ÖVERENSSTÄMMER MED SPECIFIKATIONENFRÅN TILLVERKAREN AV KONCENTRATORN OCH
SOM ANVÄNDES DÅ DINA INSTÄLLNINGARFASTSTÄLLDES.
VARNING: ENHETEN ÄR INTE AVSEDD FÖRLIVSUPPEHÅLLANDE BRUK. SMÅ BARN, ÄLDREELLER ÖVRIGA PATIENTER SOM INTE ÄR KAPABLA
ATT KOMMUNICERA OBEHAG KAN VARA IBEHOV AV YTTERLIGARE TILLSYN OCH/ELLER ETTLARMSYSTEM FÖR ATT FÖRMEDLA INFORMATIONOM OBEHAG OCH/ELLER BRÅDSKANDE BEHOVAV VÅRD TILL ANSVARIG VÅRDGIVARE FÖR ATTUNDVIKA SKADA. PATIENTER MED HÖRSEL- ELLERSYNNEDSÄTTNINGAR KAN BEHÖVA HJÄLP MED
ATT ÖVERVAKA LARM.

VARNING: GRAVIDA ELLER AMMANDEKVINNOR BÖR INTE ANVÄNDA TILLBEHÖR SOM
REKOMMENDERAS I DENNA HANDBOK EFTERSOMDE KAN INNEHÅLLA FTALATER.
VARNING: ENLIGT FEDERAL LAGSTIFTNING I
USA FÅR DENNA ENHET ENDAST SÄLJAS ELLERHYRAS UT PÅ LÄKARES ELLER ANNAN UTBILDADSJUKVÅRDSPERSONALS ORDINATION.

Kontraindikationer vid
användning
VARNING: UNDER VISSA
OMSTÄNDIGHETER KAN
ANVÄNDNINGEN AV ICKE-ORDINERATSYRE VARA FARLIG. DEN HÄR ENHETEN
BÖR ENDAST ANVÄNDAS VID
ORDINATION FRÅN EN LÄKARE.
VARNING: FÅR INTE ANVÄNDAS I NÄRHETEN AVLÄTTANTÄNDLIGA ANESTETIKA.
VARNING: ANVÄNDAREN KAN, SOM MED ALLAELEKTRISKT DRIVNA ENHETER, RÅKA UT FÖRPERIODER DÅ ENHETEN INTE FUNGERAR PÅ
GRUND AV STRÖMAVBROTT, ELLER BEHÖVATILLKALLA KVALIFICERAD TEKNIKER FÖR SERVICE
AV COMPANION 5. COMPANION 5 ÄR INTELÄMPLIG FÖR NÅGON PATIENT VARS HÄLSA
SKULLE PÅVERKAS NEGATIVT AV ETT SÅDANT
TEMPORÄRT AVBROTT.

Säkerhetsanvisningar
VARNING: DET ÄR INTE TILLÅTET ATT
MODIFIERA DENNA UTRUSTNINGVARNING: TILLVERKAREN REKOMMENDERAR EN
ALTERNATIV RESERVSYREKÄLLA I HÄNDELSE
AV STRÖMAVBROTT, LARMTILLSTÅND ELLER
MEKANISKT FEL. RÅDFRÅGA DIN LÄKAREELLER ÅTERFÖRSÄLJARE MED AVSEENDE PÅNÖDVÄNDIGT RESERVSYSTEM.
VARNING: DENNA PRODUKT LEVERERAR SYRE AV
HÖG KONCENTRATION SOM UNDERLÄTTAR SNABBFÖRBRÄNNING. TILLÅT INTE RÖKNING ELLERÖPPEN ELD INOM TVÅ (2) METER AV 1) DENNAENHET ELLER 2) SYREFYLLDA TILLBEHÖR. OM MAN
INTE TAR HÄNSYN TILL DENNA VARNING KAN DET
LEDA TILL ALLVARLIG BRAND, EGENDOMSSKADA,KROPPSSKADA OCH/ELLER DÖD.

PN 15062781-C4 J | Bruksanvisning

159 - SWE

Companion 5
VARNING: ANVÄND INTE ENHETEN I ETTBEGRÄNSAT UTRYMME (T.EX. ETT LITET SKÅPELLER EN VÄSKA) DÄR VENTILATIONEN ÄRBEGRÄNSAD. DETTA KAN LEDA TILL ATTSYRGASKONCENTRATORN ÖVERHETTAS OCH ATTPRESTANDAN FÖRSÄMRAS.
VARNING: KONCENTRATORN BÖRPLACERAS SÅ ATT DEN INTE UTSÄTTSFÖR RÖK, FÖRORENINGAR ELLER ÅNGA.
VARNING: ANVÄNDNING AV VISSA TILLBEHÖRFÖR SYRGASHANTERING SOM INTE ÄR DIREKTAVSEDDA ATT ANVÄNDAS MED DENNASYRGASKONCENTRATOR KAN PÅVERKA DESS
PRESTANDA NEGATIVT. DENNA BRUKSANVISNINGBESKRIVER VILKA TILLBEHÖR SOM
REKOMMENDERAS.
VARNING: VAR VÄNLIG KONTAKTA DINHEMVÅRDSLEVERANTÖR FÖR INSPEKTION OCH
MÖJLIG REPARATION AV DIN ENHET OM DEN
HAR TAPPATS, SKADATS ELLER EXPONERATS FÖRVATTEN. ANVÄND INTE SYRGASKONCENTRATORNOM NÄTSLADDEN ELLER KONTAKTEN ÄR SKADAD.
VARNING: SE TILL ATT VENTILERNA FÖR LUFTINTAG
OCH LUFTUTTAG INTE RISKERAR ATT BLOCKERAS.
TAPPA INTE OCH STICK INTE IN NÅGRA OBJEKTI NÅGRA AV ENHETENS ÖPPNINGAR. DETTA
KAN LEDA TILL ATT SYRGASKONCENTRATORNÖVERHETTAS OCH ATT PRESTANDAN FÖRSÄMRAS.
VARNING: ÖVERFYLL INTE FUKTFILTRET (TILLVAL).
FYLL ENDAST FUKTFILTRET (TILLVAL) MED VATTENTILL DEN NIVÅ SOM ANVISAS AV FUKTFILTRETSTILLVERKARE.
VARNING: ANVÄND INTE DENNA UTRUSTNINGUTAN ATT FÖRST LÄSA OCH FÖRSTÅ DENNA
BRUKSANVISNING. KONTAKTA ÅTERFÖRSÄLJAREN
INNAN DU FÖRSÖKER ANVÄNDA DEN OM DU INTEFÖRSTÅR VARNINGARNA OCH INSTRUKTIONERNA,ANNARS KAN FÖLJDEN BLI PERSON- ELLER
EGENDOMSSKADOR.
VARNING: OM DU KÄNNER OBEHAG ELLER
UPPLEVER EN MEDICINSK NÖDSITUATION SKA DUSÖKA LÄKARHJÄLP OMEDELBART.
VARNING: ANVÄNDNING AVSYRGASKONCENTRATORN UTANFÖR DET
SPECIFICERADE DRIFTSTEMPERATURINTERVALLET
KAN BEGRÄNSA KONCENTRATORNSMÖJLIGHET ATT UPPFYLLA SPECIFIKATIONENFÖR SYRGASKONCENTRATION. SE
SPECIFIKATIONSAVSNITTET I DENNA
BRUKSANVISNING FÖR INFORMATION OM
TEMPERATURGRÄNSERNA.

160 - SWE

PN 15062781-C4 J | Bruksanvisning

VARNING: ANVÄND INTE OLJE-, FETT-,PETROLEUMBASERADE ELLER ANDRALÄTTANTÄNDLIGA PRODUKTER TILLSAMMANS
MED SYREFYLLDA TILLBEHÖR ELLERSYRGASKONCENTRATORN. ANVÄND ENDASTVATTENBASERADE, SYREKOMPATIBLA LOTIONEROCH SALVOR. SYRE PÅSKYNDAR FÖRBRÄNNINGEN
AV LÄTTANTÄNDLIGA ÄMNEN.
VARNING: SYRGASKONCENTRATORN BÖR INTEANVÄNDAS DIREKT INTILL ELLER OVANPÅ ANNAN
UTRUSTNING. OM DETTA INTE GÅR ATT UNDVIKABÖR ENHETEN BEVAKAS FÖR ATT SÄKERSTÄLLANORMAL DRIFT.
VARNING: TILLVERKARENREKOMMENDERAR EN ALTERNATIV
RESERVSYREKÄLLA I HÄNDELSE AVSTRÖMAVBROTT, LARMTILLSTÅNDELLER MEKANISKT FEL. RÅDFRÅGA
DIN LÄKARE ELLER ÅTERFÖRSÄLJARE
MED AVSEENDE PÅ NÖDVÄNDIGTRESERVSYSTEM.
VARNING: DENNA PRODUKT LEVERERAR SYREAV HÖG KONCENTRATION SOM UNDERLÄTTARSNABB FÖRBRÄNNING. TILLÅT INTE RÖKNING
ELLER ÖPPEN ELD I SAMMA RUM SOM 1) DENNAENHET ELLER 2) SYREFYLLDA TILLBEHÖR. OM MAN
INTE TAR HÄNSYN TILL DENNA VARNING KAN DET
LEDA TILL ALLVARLIG BRAND, EGENDOMSSKADA,KROPPSSKADA OCH/ELLER DÖD.
VARNING: ANVÄNDNING AV VISSA TILLBEHÖRFÖR SYRGASHANTERING SOM INTE ÄR DIREKTAVSEDDA ATT ANVÄNDAS MED DENNASYRGASKONCENTRATOR KAN PÅVERKA DESS
PRESTANDA NEGATIVT. DENNA BRUKSANVISNINGBESKRIVER VILKA TILLBEHÖR SOM
REKOMMENDERAS.
VARNING: VAR VÄNLIG KONTAKTA DINHEMVÅRDSLEVERANTÖR FÖR INSPEKTION OCH
MÖJLIG REPARATION AV DIN ENHET OM DEN
HAR TAPPATS, SKADATS ELLER EXPONERATS FÖRVATTEN. ANVÄND INTE SYRGASKONCENTRATORNOM NÄTSLADDEN ELLER KONTAKTEN ÄR SKADAD.
VARNING: ANVÄND ENDAST RESERVDELAR SOM
REKOMMENDERAS AV TILLVERKAREN FÖR ATTSÄKERSTÄLLA KORREKT FUNKTION OCH FÖR ATTUNDVIKA RISK FÖR BRAND OCH BRÄNNSKADOR.
VARNING: SMÖRJ INTE KOPPLINGAR,ANSLUTNINGAR, RÖR ELLER ANDRA TILLBEHÖR
TILL SYRGASKONCENTRATORN FÖR ATT UNDVIKARISK FÖR BRAND OCH BRÄNNSKADOR.

Companion 5
VARNING: RÖKNING I SAMBAND MEDSYRGASANVÄNDNING ÄR DEN FRÄMSTAORSAKEN TILL BRANDSKADOR OCH DÄRTILL
RELATERADE DÖDSFALL. DU MÅSTE FÖLJA DESSA
SÄKERHETSVARNINGAR:VARNING: TILLÅT INTE RÖKNING, LJUSLÅGOR ELLERÖPPEN ELD I SAMMA RUM SOM DENNA ENHETELLER SYREFYLLDA TILLBEHÖR.
VARNING: RÖKNING I SAMBAND MED
ANVÄNDNING AV NÄSKANYL KAN ORSAKABRÄNNSKADOR I ANSIKTET OCH EVENTUELLTDÖDSFALL.
VARNING: DESSA 3 STEG KAN RÄDDA DITT LIV OM
DU RÖKER: STÄNG AV SYRGASKONCENTRATORN,TA AV NÄSKANYLEN OCH LÄMNA RUMMET DÄR
ENHETEN STÅR.
VARNING: SKYLTAR MED ”RÖKNING EJ TILLÅTEN –SYRGAS ANVÄNDS” MÅSTE VARA VÄL SYNLIGA I
HEMMET ELLER DÄR SYRGAS ANVÄNDS. PATIENTER
OCH DERAS VÅRDGIVARE MÅSTE INFORMERAS OM
FARORNA MED RÖKNING I NÄRHETEN AV, ELLER
UNDER ANVÄNDNING AV, MEDICINSK SYRGAS.
VARNING: ANVÄND INTE SYRGASKONCENTRATORNI NÄRHETEN AV LÄTTANTÄNDLIGA GASER. DET
KAN LEDA TILL SNABB BRANDUTVECKLING OCH
ORSAKA EGENDOMSSKADA, KROPPSSKADA ELLERDÖDSFALL.
VARNING: OM MAN TAR AV NÄSKANYLEN OCH
LÄGGER DEN PÅ KLÄDER, SÄNGKLÄDER, SOFFOR
ELLER ANDRA MJUKA MATERIAL KAN BRAND
PLÖTSLIGT UPPSTÅ OM NÄSKANYLEN UTSÄTTSFÖR CIGARETTGLÖD, EN VÄRMEKÄLLA ELLER LÅGA.
VARNING: LÄMNA INTE NÄSKANYLEN PÅ ELLER
UNDER KLÄDER, SÄNGKLÄDER ELLER STOLSDYNOR.
OM ENHETEN ÄR PÅSLAGEN MEN INTE ANVÄNDS
KAN SYRET GÖRA MATERIALET BRANDFARLIGT.
SÄTT I/0-STRÖMBRYTAREN I 0-LÄGET (AV) NÄRSYRGASKONCENTRATORN INTE ANVÄNDS.

VARNING: ANVÄND ENDAST TILLBEHÖR
SOM REKOMMENDERAS AV TILLVERKAREN.
ANVÄNDNING AV ANDRA TILLBEHÖR KAN
VARA FARLIGT, ORSAKA ALLVARLIG SKADA PÅSYRGASKONCENTRATORN OCH GÖR GARANTIN
OGILTIG.
VAR FÖRSIKTIG: Koppla alltid urväxelströmsadaptern från väggen
innan du kopplar ur den frånsyrgaskoncentratorn.
VAR FÖRSIKTIG: Placera alltid syrgasslangar ochelkablar på ett sätt som förhindrar snubbling.
VAR FÖRSIKTIG: Följ alla anvisningarfrån tillverkaren för att undvika att
garantin upphävs.
Obs! Portabel och mobil radiofrekvensutrustning för
kommunikation kan påverka medicinsk elektronisk
utrustning.
VARNING: SÄTT ALLT SÄTT AV TILLFÖRSÄLJNING AVOXYGENFÖRFÖRING OCH KRAFTKORTER I EN MANNER SOM FÖRHINDRAR SPRIDNINGSRISK ELLERMÖJLIG OAVSIKTLIG FÖRVIRRING.
VARNING: DENNA PRODUKT KAN EXPONERADIG FÖR KEMIKALIER, INKLUSIVE NICKEL SOM I
DELSTATEN KALIFORNIEN ANSES ORSAKA CANCER.
GÅ TILL WWW.P65WARNINGS.CA.GOV FÖR MER
INFORMATION.
VARNING: OM DET INTRÄFFAR EN ALLVARLIG HÄNDELSE MED DEN HÄR ENHETEN BÖR ANVÄNDARENOMEDELBART RAPPORTERA HÄNDELSEN TILL
LEVERANTÖREN OCH / ELLER TILLVERKAREN. EN
ALLVARLIG HÄNDELSE DEFINIERAS SOM EN SKADA,DÖDSFALL ELLER POTENTIAL ATT ORSAKA SKADA /DÖDSFALL OM HÄNDELSEN SKULLE ÅTERKOMMA.
ANVÄNDAREN KAN OCKSÅ RAPPORTERA HÄNDELSEN TILL DEN BEHÖRIGA MYNDIGHETEN I DETLAND DÄR HÄNDELSEN INTRÄFFADE.

VARNING: UTFÖR INGETUNDERHÅLLSARBETE ANNAT ÄN
DE MÖJLIGA LÖSNINGAR SOM TAS
UPP I DENNA BRUKSANVISNING. TA
INTE AV NÅGRA HÖLJEN. ENDASTÅTERFÖRSÄLJAREN ELLER ENKVALIFICERAD SERVICETEKNIKER FÅR TA
AV HÖLJEN ELLER UTFÖRA SERVICE PÅ
DENNA ENHET.

PN 15062781-C4 J | Bruksanvisning

161 - SWE

Companion 5
Specifikationer
Följande tabell visar information om rekommenderad
driftsmiljö och driftsförhållanden, som är viktig för
korrekt användning av enheten.

Specifikationer
Flödeshastigheter 0,5–5,0 liter/min (LPM)
Syrgaskoncentration 90 % (+5,5 %/−3 %)Mått54,6 × 31,8 × 34,2 cm(21,5 × 12,5 × 13,5 tum)
Vikt16,3 kg (36 lbs)
Ljudtrycksnivå
50 dB (A) vid 2,0 LPM
Strömförbrukning
285 W vid 2 LPM, max 350 W
Driftstemperatur
5 °C till 40 °C (41 °F till 104 °F)
Luftfuktighet
15% - 90% vid en
vid drift
daggpunktstemperatur på
28 °C (82,4 °F)
Förvaringstemperatur −25 ºC till 70 ºC(−13 ºF till 158 ºF)
Luftfuktighet vid
0% - 90% icke-kondenserande
förvaring
Elektronik
Använd inte
förlängningssladdar. Använd
inga elektriska uttag som styrs
av en brytare.
Drift vid höjd över -382 – 3 011 meterhavet(-1 253 – 9 879 fot)
Drifttid
Upp till 24 timmar per dag
Utrustningens förväntade livslängd är minst fem år.
VARNING: ANVÄNDNING AV
ENHET UTAN FÖR SPECIFICERADEDRIFTSFÖRFARANDE FÖRVÄNTAS
PÅVERKAS FLÖDESGRADEN OCH
PROCENTANDELEN AV SYRE OCHFÖLJAKTLIGEN KVALITETEN PÅ TERAPIN.
Obs! Om syrgaskoncentratorn har förvarats en längre
tid i en temperatur som är utanför det normala
driftstemperaturintervallet bör enheten få tid att återgå
till normal driftstemperatur innan man sätter på den.
(Se specifikationsavsnittet i denna manual.)

Korrekt placering av Companion 5

162 - SWE

PN 15062781-C4 J | Bruksanvisning

Välj en plats för enheten där du undviker intag avrök, ångor och luftföroreningar. Vid korrekt placeringska enheten kunna suga in luft genom de tre olika
luftintagen längst upp på baksidan och under höljet,
och ingenting får blockera luften som kommer ut ur
luftutsläppet längst ned till vänster på enheten.
Luftintagen sitter placerade upptill på koncentratorns
baksida och nedtill. Placera enheten så det är minst
30 centimeters (12 tums) frigång mellan koncentratornoch eventuella väggar, möbler, gardiner eller andra
föremål.
Placera enheten så att larmljud kan höras.
Placera syrgasslangar på ett sådant sätt att de inte kröks
eller täpps till.
VARNING: ANVÄND INTESYRGASKONCENTRATORN I
NÄRHETEN AV LÄTTANTÄNDLIGAGASER. DET KAN LEDA TILL SNABB
BRANDUTVECKLING OCH ORSAKAEGENDOMSSKADA, KROPPSSKADA
ELLER DÖDSFALL. ANVÄND INTE
OLJE-, FETT-, PETROLEUMBASERADE
ELLER ANDRA LÄTTANTÄNDLIGAPRODUKTER TILLSAMMANS MEDSYREFYLLDA TILLBEHÖR ELLERSYRGASKONCENTRATORN. ANVÄND
ENDAST VATTENBASERADE,SYREKOMPATIBLA LOTIONEROCH SALVOR. SYRE PÅSKYNDARFÖRBRÄNNINGEN AV LÄTTANTÄNDLIGAÄMNEN.
VARNING: DENNA PRODUKT LEVERERAR SYREAV HÖG KONCENTRATION SOM UNDERLÄTTARSNABB FÖRBRÄNNING. TILLÅT INTE RÖKNING
ELLER ÖPPEN ELD I SAMMA RUM SOM 1) DENNAENHET ELLER 2) SYREFYLLDA TILLBEHÖR. OM MAN
INTE TAR HÄNSYN TILL DENNA VARNING KAN DET
LEDA TILL ALLVARLIG BRAND, EGENDOMSSKADA,KROPPSSKADA OCH/ELLER DÖD.

Companion 5
Användarinstruktioner
Före användning

Den här bruksanvisningen fungerar som din referens
för att hjälpa dig att använda och underhålla enheten.
Kontakta din hemvårdsleverantör om du har några
frågor eller funderingar.
Viktigt! Använd INTE Companion 5 utan att
först ha läst sektionen med säkerhetsriktlinjer
i den här bruksanvisningen. Vänligen följ alla
användarinstruktioner. Observera alla varningar
på enheten och i denna bruksanvisning. Observera
alla säkerhetsföreskrifter för att reducera brandrisk,
personlig skada och allvarlig skada på Companion 5.
CAIRE Companion 5 tillverkar inte sitt eget syre. Den
producerar högkoncentrerat syre från den omgivande
rumsluften och levererar den till dig. I sitt naturliga
tillstånd består rumsluft av cirka 21 volymprocent syre.
Rumsluften sugs in i koncentratorn av en kompressor.
Luften rör sig sedan genom en rad olika filter och
genom en molekylsikt som på kemisk väg absorberar
kvävemolekyler. Som ett resultat levereras den
syreberikade luft som lämnar koncentratorn med
87-95 % volymprocent syre.
VARNING: RENGÖR HÖLJET,
KONTROLLPANELEN OCH
STRÖMSLADDEN MED ENBART ETTMILT HUSHÅLLSRENGÖRINGSMEDEL,
SOM PÅFÖRS MED EN FUKTIG (INTE
BLÖT) TRASA ELLER SVAMP. TORKA
DÄREFTER ALLA YTOR TORRA. SE
TILL ATT DET INTE HAMNAR NÅGON
VÄTSKA INNE I ENHETEN. VAR SPECIELLTUPPMÄRKSAM PÅ SYRGASUTTAGETFÖR KANYLANSLUTNINGEN SÅ ATT DETGARANTERAT ÄR FRITT FRÅN DAMM,VATTEN OCH PARTIKLAR.
VAR FÖRSIKTIG: Se till att ventilerna
för luftintag och luftuttag inte riskerar
att blockeras. Detta kan leda till att
syrgaskoncentratorn överhettas och
att prestandan försämras. Använd
inte syrgaskoncentratorn utan att
luftintagsfiltret sitter på plats. Om
ett andra filter tillhandahålls, sätt iersättningsfiltret innan du rengör det
smutsiga filtret. Rengör det smutsiga
filtret i en varm tvålvattenlösning och
torka det ordentligt innan det används.

Steg 1: Placera din Companion 5 för användning
Placera Companion 5 på en välventilerad, väl upplyst
plats. Se till att luftintag och luftuttag inte blockeras.
Placera Companion 5 så att alla hörbara och visuellaindikatorer eller larm enkelt kan ses eller höras.
Koppla in enheten i ett växelströmsuttag.
Steg 2: Om fuktfilterflaskan INTE SKA användas

Luftintag

Kylluftsintag
Luftutsläpp

Luftintag

a. Anslut en näskanyl eller syrgasslang direkt tillsyrgasuttagsnippeln.

b. Gå vidare till steg 4.
Steg 3: Om fuktfilterflaskan SKA användas
a. Skruva loss locket från fuktfilterflaskan.
b. Fyll flaskan med destillerat vatten. Se till att
vattennivån ligger mellan tillverkarens angivna maxoch miniminivå som visas på flaskan.
c. Skruva åt locket på fuktfilterflaskan. Se till att det
inte förekommer några läckor.
d. Anslut en näskanyl eller standardsyrgasslang till
uttagsnippeln på fuktfilterflaskan.
Obs! Användning av fuktfilter som inte är avsett för
användning tillsammans med denna koncentrator kan
innebär att prestandan påverkas.

PN 15062781-C4 J | Bruksanvisning

163 - SWE

Companion 5
g. Placera flaskan på det anvisade stödstället på
fuktfilterflaskan.
h. Säkra flaskan på stället med hjälp av den elastiska
remmen.

e. Anslut fuktfilterslangens gängade ände till locket på
fuktfilterflaskan.

i. Gå vidare till steg 4.
Steg 4: Sätt på enheten och låt den värma upp
a. Koppla strömsladden på CAIRE Companion 5 till
ett växelströmsuttag.
VARNING: INSPEKTERA STRÖMSLADDENMED AVSEENDE PÅ SKADOR FÖREANVÄNDNING. OM SLADDEN ÄR
SKADAD FÅR DU INTE KOPPLA IN DEN I
ETT ELUTTAG ELLER FÖRSÖKA ANVÄNDA
KONCENTRATORN.

f. Anslut den andra änden av fuktfilterslangen till
uttagsnippeln på CAIRE Companion 5.

b. Sätt strömbrytaren till ”PÅ”-position (I). När
Companion 5 har kommit igång tänds den gröna
indikatorn på LED-displayen. Alla LED-lamporlyser vid start. Efter det att koncentratorn har
fullgjort uppvärmningscykeln kommer bara den
gröna lampan att fortsätta lysa.
Obs! Låt enheten stå i upp till tio (10) minuter efter
att den slagits PÅ för första gången för att den ska nå
sina prestandaspecifikationer.

164 - SWE

PN 15062781-C4 J | Bruksanvisning

Companion 5
VARNING: DET ÄR MYCKET VIKTIGTATT ENBART STÄLLA IN DEN
ORDINERADE NIVÅN AV SYRE. ÄNDRA
INTE FLÖDESVALET OM DU INTE
HAR BLIVIT UPPMANAD ATT GÖRA
DET AV EN KVALIFICERAD KLINIKER.
SYRGASKONCENTRATORN KAN
ANVÄNDAS NÄR MAN SOVER SÅLÄNGE DET SKER PÅ EN KVALIFICERADKLINIKERS REKOMMENDATION.
Steg 5: Reglera flödeskontrollhastighet

Steg 6: Kontrollera flödet och andas normalt

a. Vrid flödeskontrollknappen till den flödeshastighet(LPM) som ordinerats av din läkare.

a. Kontrollera att syret flödar fritt genom näskanylenoch att det inte finns några fnurror, krökar eller
igensättningar i slangarna.
Obs! Se till att kanylen är helt instucken och ordentligt
fastsatt. Du ska kunna höra eller känna hur syre flödar
till näskanylens mynningar.
b. Placera din näskanyl korrekt och andas normalt
genom näskanylen.
Steg 7: Stäng av
a. Ställ strömbrytaren till ”AV”-position (O) när
CAIRE Companion 5 inte längre används.

För att reglera flödeshastighet:
Vrid motsols för att öka flödet.
Vrid medsols för att minska flödet.
Bollens mitt indikerar flödeshastigheten. Bilden nedan
visar en flödeshastighet på 4,0 LPM.

VAR FÖRSIKTIG: Använd endast tillbehör
som tillverkaren rekommenderar.
Användning av andra tillbehör kan
vara farligt, orsaka allvarlig skada på
syrgaskoncentratorn och gör garantin
ogiltig.
VAR FÖRSIKTIG: Använd inte förlängningssladdar
med denna enhet och undvik att ansluta förmånga stickproppar i samma eluttag. Användningav förlängningssladdar kan påverka produktens
prestanda negativt. För många stickproppar i ett
eluttag kan leda till en överbelastning i elcentralenvilket gör att proppen/säkringen löses ut eller
orsaka brand om proppen eller säkringen inte
fungerar.
VAR FÖRSIKTIG: Användning av andra kablar och
adaptrar än de specificerade, förutom sådana som
tillverkaren av den medicinska utrustningen säljer
som ersättningsdelar för ingående komponenter,kan orsaka ökad emission eller försämrad
immunitet hos syrgaskoncentratorn.

Korrekt

PN 15062781-C4 J | Bruksanvisning

165 - SWE

Companion 5
VARNING: SE TILL ATT VENTILERNAFÖR LUFTINTAG OCH LUFTUTTAG INTE
RISKERAR ATT BLOCKERAS. DETTA KAN
LEDA TILL ATT SYRGASKONCENTRATORNÖVERHETTAS OCH ATT PRESTANDAN
FÖRSÄMRAS.
Obs: Till utrustningsleverantören: Följande syretillbehör
rekommenderas för användning med Companion 5:• Nasal kanyl: CAIRE artikelnummer CU002-1• Luftfuktningsadapterrör: CAIRE Artikelnummer
20843882• Luftfuktare: CAIRE artikelnummer HU003-1• Firebreak: CAIRE artikelnummer 20629671
En eldbrytning krävs för användning med kanyler.
• CAIRE erbjuder en eldbrytare avsedd att användas
i samband med syrekoncentratorn. Firebreak är entermisk säkring för att stoppa gasflödet i händelse
av att nedströms kanylen eller syrgasröret antänds
och bränns till firebreak. Den placeras i linje med
näsekanylen eller syrgasröret mellan patienten
och syreutloppet från Companion 5. För korrekt
användning av brandbrytaren, se alltid tillverkarens
anvisningar (medföljer varje brandbrytarkit).
• För ytterligare rekommenderade tillbehör, setillbehörskatalogen (PN MLLOX0010) som finns på
www.caireinc.com.

166 - SWE

PN 15062781-C4 J | Bruksanvisning

Companion 5
Larmtillstånd
Din CAIRE Companion 5 använder en kombination av
två eller tre (för modeller med OCSI) LED-lampor och
ett larmande ljud för att uppmärksamma dig på fel hos
din koncentrator.
1) Grön LED-lampa () - påvisar normal drift.

Grön

Röd

2) Röd LED-lampa ( ) - påvisar systemfel. Se
tabellen över larmtillstånd nedan för mer info.
3) Gul LED () - påvisar låg syrekoncentration. Se
tabellen över larmtillstånd nedan för mer info.
I tabellen över larmtillstånd på nästa sida visas deolika larmtillstånd som kan visas av koncentratorn.
Den innehåller möjliga orsaker och åtgärder som bör
vidtas om larmet lyser/ljuder. Vid eventuella larm
rekommenderar vi att du omedelbart kontaktar dinvårdleverantör.

Gul

Ovan: Companion 5

VARNING: IGNORERA INTE LARMVARNINGAR: TILLVERKARENREKOMMENDERAR EN ALTERNATIV
RESERVSYREKÄLLA I HÄNDELSE AVSTRÖMAVBROTT, LARMTILLSTÅNDELLER MEKANISKT FEL. RÅDFRÅGA
DIN LÄKARE ELLER ÅTERFÖRSÄLJARE
MED AVSEENDE PÅ NÖDVÄNDIGTRESERVSYSTEM.

PN 15062781-C4 J | Bruksanvisning

167 - SWE

Companion 5
Tabell över larmtillstånd
Ljudandelarm

FärgadLEDlampa

Av
grönt

Orsak

Koncentratorn fungerar
korrekt.

Koncentratorn har förlorat
ström men strömbrytarenstår fortfarande på ”PÅ”.

Intermittent

Din åtgärd
Inga

Kontrollera att koncentratorns strömsladd ärinkopplad i ett uttag och att uttaget har ström. Testa
ett annat uttag. Om problemet kvarstår, anslut till en
reservsyrgaskälla och kontakta din vårdleverantör.

RÖTTIntermittent

Systemfel

Ställ strömbrytaren omedelbart i AVSTÄNGT läge.
Dra ut växelströmsladden ur vägguttaget. Vänta i 5
minuter. Koppla in växelströmsladden i vägguttaget
igen och sätt igång koncentratorn igen. Om larmet
ljuder behövs service. Anslut till en reservsyrgaskälla
och kontakta omedelbart din vårdleverantör.

Produktflödet för högt
eller för lågt1. Se till att kanylen inte är vikt eller blockerad. Om
en fuktfilterflaska används, försäkra dig om att den är
korrekt fylld och inte orsakar en blockering.
2. Försäkra dig om att Companion 5 har ordentlig
ventilation. Den måste stå minst 30 cm (12 tum) frånalla ytor för att säkerställa att ventilationsöppningarna
inte är blockerade.
3. Växla till en alternativ syrgaskälla och kontaktavårdleverantör för assistans om problemet kvarstår.

RÖTTIntermittent
RÖTT

Companion 5 har upptäcktlåga syrgasnivåer.

Intermittent
gult

Companion 5 har upptäcktlåga syrgasnivåer.

Intermittent
gult

168 - SWE

PN 15062781-C4 J | Bruksanvisning

1. Se till att luftintagsfiltret och utsläppsöppningarna
inte är igensatta eller blockerade. 2. Se till att
Companion 5 står i ett välventilerat utrymme. Försäkra
dig om att det är minst 30 cm (12 tum) mellan baksidan
på Companion och eventuella hinder (möbler, gardinerosv.).
3. Växla till en alternativ syrgaskälla och kontakta dinvårdleverantör omedelbart om problemet kvarstår.

Companion 5
Felsökning
Vi rekommenderar att du kontaktar din vårdleverantör
om din koncentrator aktiverar larm eller om dumärker att din koncentrator inte fungerar som denska. Försök inte utföra reparationer på koncentratorn
på egen hand. Reparationer får bara utföras av en
utbildad servicetekniker. Det finns emellertid ett antal

felsökningssteg du kan vidta om du upplever att du har
problem med din CAIRE Companion 5. De beskrivs iföljande tabell. Du kan använda denna tabell genom att
följa de numrerade stegen.

Felsökningstabell
Problem
Koncentratornär påslagen,men den är inteigång.

Syrgasflödetär lågt eller
obefintligt.

Möjlig orsak

Din åtgärd

1. Strömsladden är inte inkopplad i
ett uttag.

1. Koppla in strömsladden.

2. Eluttaget ger ingen ström.

2. Kontrollera hushållssäkringar och kretsar. Testa ett
annat uttag.

3. Internfel.

3. Anslut till en reservsyrgaskälla. Kontakta dinvårdleverantör omedelbart.

1. Flödeskontrollinställningarna harändrats.

1. Försäkra dig om att flödeskontrollknappen är
inställd på korrekt liter per minut-inställning (LPM) som
ordinerats av din läkare.

2. Slangarna har kopplats ur.

2. Kontrollera att slangarna är täta vid alla anslutningar(uttagsnippel, fuktfilterflaska, vattenavskiljare osv.).
Säkra om anslutningarna vid behov.

3. Syrgasslangen eller -kanylen ärkrökt.

3. Kontrollera slangarna med avseende på krökningar
eller blockeringar. Reparera vid behov.

4. Det finns en läcka i kanylen ellerslangen.

4. Inspektera alla slangar och kanylen med avseendepå läckor. Detta kan du göra genom att antingen lyssna
eller känna efter utsipprande luft. Byt ut läckande slangeller kanyl.

5. Luftflödet in i koncentratorn är
försämrat.

5. Kontrollera att inget blockerar inloppen på
koncentratorn. Se till att den står placerad minst 30 cm(12 tum) från möbler, väggar eller andra hinder.

6. Internfel.

6. Anslut till en reservsyrgaskälla. Kontakta dinvårdleverantör omedelbart.

Obs! Om din Companion 5 fortfarande larmar efter att du vidtagit åtgärderna i tabellerna över larmtillstånd och
felsökning, kontakta teknisk service på 1-800-482-2473 med den larmkod som visas på din LCD-display.

PN 15062781-C4 J | Bruksanvisning

169 - SWE

Companion 5
Rengöring, skötsel och rutinunderhåll
Rutinunderhåll

Rutinunderhåll består av att byta grovpartikelfiltret i
handtaget en gång i veckan (om ett sådant är monterat).

Användarens skötsel och rengöring av
enheten

Byte av kanyl
Ersätt syrgasslangar och kanyler regelbundet enligthemvårdsleverantörens rekommendationer. Din
läkare eller hemvårdsleverantör kommer att ge dig
information om rengöring, desinfektion och ersättning.
Obs! Följ alltid kanyltillverkarens anvisningar
för korrekt användning. Byt ut engångskanylen
enligt kanyltillverkarens eller återförsäljarens
rekommendationer. Du kan beställa nya från dinåterförsäljare.

Luftintagsfilter (kassett)Luftintagsfiltret får endast hanteras eller bytas av
vårdleverantören.

Obs! Tillverkaren rekommenderar inte att utrustningensteriliseras.
Obs! Använd inte syrgaskoncentratorn utan att
luftintagsfiltret sitter på plats.
Fuktfilterflaska (om tillämpligt)
Kontrollera fuktfilterflaskan dagligen för att försäkra
dig om att dess vattennivå ligger mellan angivna minoch maxnivåer. Fyll på flaskan med destillerat vatten
vid behov. Rengör fuktfilterflaskan enligt tillverkarens
anvisningar.
Rengöringsprocedur:
Stäng AV Companion 5 och koppla ur den frånväxelströmsförsörjningen före all rengöring eller
desinficering. Spreja INTE direkt på ytterhöljet.
Använd en fuktig (inte blöt) trasa eller svamp. Spreja
trasan eller svampen med ett milt rengöringsmedel
för att rengöra hölje och strömsladd. Använd
rengöringsmedel Lysol® Fabrikat II (eller motsvarande)
för desinficering av Companion 5. Späd enligt
rengöringsmedelstillverkarens anvisningar, men spreja
inte vätska direkt på Companion 5.

Grovpartikelfilter (tillval):

Bruttopartikelfiltret är beläget på framsidan av
Companion 5.

Grovpartikelfilter:

Du bör inspektera och rengöra den en gång iveckan. För att rengöra filtret använder du följande
process:
1) Ta bort grovpartikelfiltret.
2) Tvätta filtret i varmt kranvatten med en mildtvålbaserad tvättlösning.
3) Skölj filtret noggrant med rent kranvatten och krama
ur överblivet vatten.
4) Låt filtret lufttorka.
5) Sätt tillbaka filtret i höljet.

170 - SWE

PN 15062781-C4 J | Bruksanvisning

VARNING: RISK FÖR ELEKTRISKASTÖTAR. STÄNG AV ENHETEN OCH DRAUR STRÖMSLADDEN FRÅN ELUTTAGETINNAN DU RENGÖR ENHETEN FÖRATT FÖRHINDRA OAVSIKTLIGAELEKTRISKA STÖTAR OCH BRAND.
ENDAST DIN ÅTERFÖRSÄLJARE ELLEREN KVALIFICERAD SERVICETEKNIKER
FÅR TA BORT HÖLJET ELLER REPARERAENHETEN.
VARNING: RISK FÖR ELEKTRISKA STÖTAR. STÄNG
AV ENHETEN OCH DRA UR STRÖMSLADDEN FRÅNELUTTAGET INNAN DU RENGÖR ENHETEN FÖR ATTFÖRHINDRA OAVSIKTLIGA ELEKTRISKA STÖTAR
OCH BRAND. ENDAST DIN ÅTERFÖRSÄLJARE ELLEREN KVALIFICERAD SERVICETEKNIKER FÅR TA BORTHÖLJET ELLER REPARERA ENHETEN.

Companion 5
VARNING: RISK FÖR ELEKTRISKA STÖTAR. KOPPLAIFRÅN NÄTSLADDEN FRÅN ELUTTAGET INNAN DURENGÖR APPARATEN FÖR ATT UNDVIKA RISKENFÖR ELSTÖTAR OCH BRÄNNSKADOR. ENDASTDIN ÅTERFÖRSÄLJARE ELLER EN KVALIFICERADSERVICETEKNIKER FÅR TA BORT HÖLJET ELLER
REPARERA ENHETEN.
VARNING: SÄKERSTÄLL ATTSYRGASKONCENTRATORN INTE BLIR BLÖT OCH ATTVÄTSKOR INTE KOMMER IN I ENHETEN. ANNARS
KAN DET LEDA TILL FEL ELLER ATT ENHETENSTÄNGS AV OCH MEDFÖRA EN RISK FÖR ELCHOCKELLER BRÄNNSKADOR.
VARNING: DET KAN VARA OSÄKER ATT
SAMMANKOPPLA KOMPANIONEN 5 MED ANNATUTRUSTNING.

VARNING: RENGÖR HÖLJET, KONTROLLPANELENOCH STRÖMSLADDEN MED ENBART ETT MILTHUSHÅLLSRENGÖRINGSMEDEL, SOM PÅFÖRS
MED EN FUKTIG (INTE BLÖT) TRASA ELLER SVAMP.
TORKA DÄREFTER ALLA YTOR TORRA. SE TILLATT DET INTE HAMNAR NÅGON VÄTSKA INNE
I ENHETEN. VAR SPECIELLT UPPMÄRKSAM PÅSYRGASUTTAGET FÖR KANYLANSLUTNINGEN SÅ
ATT DET GARANTERAT ÄR FRITT FRÅN DAMM,VATTEN OCH PARTIKLAR.
VARNING: ANVÄND INTE VÄTSKA DIREKT PÅ
ENHETEN. EN LISTA ÖVER OLÄMPLIGA KEMISKA
ÄMNEN INKLUDERAR, MEN ÄR INTE BEGRÄNSAD
TILL, FÖLJANDE: ALKOHOL OCH ALKOHOLBASERADEPRODUKTER, KONCENTRERADE KLORBASERADEPRODUKTER (ETYLENKLORID) OCH OLJEBASERADEPRODUKTER (PINE-SOL®, LESTOIL®). DESSA FÅR
INTE ANVÄNDAS FÖR ATT RENGÖRA PLASTHÖLJET
PÅ SYRGASKONCENTRATORN EFTERSOM DE KANSKADA PLASTEN PÅ ENHETEN.

BortskaffandeÅterlämna alltid Companion 5 inklusive alla
komponenter till din hemsjukvårdsgivare för korrekt
avfallshantering. Du kan även kontakta ditt lokala
förvaltnings- eller kommunkontor för instruktioner om
korrekt avfallshantering av batteriet.

WEEE och RoHS
Syftet med denna symbol är att påminnaägaren av utrustningen att lämna tillbaka den till enåtervinningsanläggning när den ska kasseras, enligt
WEEE-direktivet om kassering av elektriskt avfall(Waste Electrical and Electronic Equipment).

Våra produkter uppfyller kraven i RoHS-direktivet(begränsning av användningen av vissa farliga ämnen
i elektriska och elektroniska produkter). De innehåller
inte mer än spårelement av bly och andra farliga
material.
FÖRSIKTIGHET: För korrekt bortskaffande, kontakta din utrustningsleverantör eller lokala myndigheter för
instruktioner om bortskaffande.

CAIRE och CAIRE Inc. är registrerade varumärken som tillhör
CAIRE Inc. Besök vår webbplats nedan för en fullständig lista övervarumärken.
Varumärken: www.careinc.com/corporate/trademarks.
Copyright © 2020 CAIRE Inc.   CAIRE Inc. förbehåller sig rätten
att när som helst utan förvarning upphöra med försäljning av sina
produkter eller ändra produkternas priser, material, utrustning,kvalitet, beskrivningar, specifikationer och/eller processer utan
vidare förpliktelser eller följder. Alla rättigheter som inte uttryckligen
anges häri är reserverade av oss, enligt tillämplighet.
PN 15062781-C4 J | Bruksanvisning

171 - SWE

Companion 5
Säätimet ja järjestelmän tilan merkkivalot
Sisäiset symbolit

02

Tuottaa happea

I/O

Katkaisija
Ei sisällä huollettavia osia. Älä avaa
kantta.

ISO 7000; laitteessa käytettävät graafiset
symbolit – hakemisto ja yhteenveto

Tyypin BF potilaskosketuksessa
oleva osa (sähköiskusuojauksen taso).
Sääntö 5333
Neuvoston direktiivi 93/42/ETY lääkinnällisistä
laitteista
Valtuutettu edustaja Euroopan
yhteisössä
IEC 60417

Varastointi- tai käyttölämpötila-alue.
Sääntö 0632
Varastointi- tai käyttölämpötila-alue.
Sääntö 2620

Luokan II laite, kaksinkertaisesti
eristetty. Sääntö 5172
ETL-sertifiointi

Valmistajan nimi ja osoite. Sääntö 3082

Turvallisuusvirasto CAN/CSA C22.2
nro 601.1 M90 lääketieteellisiä laitteita
vartenValmistuspäivämäärä. Sääntö 2497
Luettelonumero. Sääntö 2493
Sarjanumero. Sääntö 2498
Tämä puoli ylös. Sääntö 0623

FCC:n ilmoitus
21 CFR 801.15: Code of Federal Regulations
Title 21 (Yhdysvallat)
Yhdysvaltojen liittovaltion laki sallii
tämän laitteen myynnin vain lääkärilletai lääkärin määräyksestä.
Neuvoston direktiivi 2012/19/EU: sähkö- ja
elektroniikkalaiteromu (WEEE)

Särkyvää, käsittele varoen. Sääntö 0621
WEEE
Säilytettävä kuivassa ja pidettävä
kuivana. Sääntö 0626
Lue käyttöopas ennen käyttöä.
Sääntö 1641
Pinoamisraja lukumäärän mukaan.
Sääntö 2403
ISO 7010: Graafiset symbolit—Turvavärit ja
turvamerkit—Rekisteröidyt turvamerkit
Käyttöohjeet on luettava. Sääntö M002
Varoitus. Sääntö W001
Pidettävä kaukana avotulesta, tulesta ja
kipinöistä. Avoimet syttymislähteet ja
tupakointi kielletty. Sääntö P003
Älä tupakoi yksikön lähellä tai kun
käytät sitä. Sääntö P002

172 - FIN PN 15062781-C4 J | Käyttöopa

IEC 60601-1: Sähkökäyttöiset lääkintälaitteet Osa 1: Yleiset vaatimukset turvallisuudelle jaolennaiselle suorituskyvylle

IPX1

Tippuvedenpitävä laite IPX1:
Companion 5 -laite on suojattu
nesteiden haitallista sisäänpääsyä
vastaan. (IPX1, IEC 60529)

FCC:N ILMOITUS:

Tämä laite saattaa sisältää Johnson Technology -yhtiön
CYBLE-022001-00-komponentin, mukaan lukien
antenni 2450AT18B100. Nämä ovat FCC-sääntöjen
osan 15 vaatimusten mukaisia. Tämä laite täyttää
FCC:n julkisen ilmoituksen DA00-1407 modulaarisia
lähettimiä koskevat vaatimukset. Lähettimen toimintaa
koskevat seuraavat ehdot: (1) tämä laite ei saa
aiheuttaa haitallisia häiriöitä, ja (2) tämän laitteen
on vastaanotettava kaikki häiriöt, mukaan lukien
mahdollisesti ei-toivottua toimintaa aiheuttavat häiriöt.
Tämä tuote saattaa olla suojattu yhdellä tai useammalla
patentilla Yhdysvalloissa ja muissa maissa. Alla
olevalla verkkosivullamme on lueteltu kaikki
sovellettavat patentit. Pat.: www.caireinc.com/
corporate/patents/

Companion 5
Pika-aloitusopas
1 Poista Companion 5 pakkauksesta

Companion 5

2 Tutustuminen Companion 5 -laitteeseen
Lue kaikki tässä käyttöoppaassa olevat varoitukset,
muistutukset ja laitteen lisätiedot. Tutustu
Companion 5 -laitteen ja käyttöohjauspaneelin
tärkeimpiin ominaisuuksiin.

EDESTÄ
KoteloVirtakytkin

LEDnäyttö

Hapen
ulostuloaukko

Tuntimittari

Virtausmittari
Kostutinpullon teline(pullovalinnainen)

TAKAA
Kantokahva ja
ilmanottoaukko

Jäähdytysilman
tuloaukko

Ilmanpoistoaukko
Ilmanottoaukko

PN 15062781-C4 J | Käyttöopa

173 - FIN

Companion 5
Companion 5 -laitteen osat
KoteloVirtakytkin

Ulostuloaukko

LED-näyttö

Tuntimittari

Virtausmittari

Kostutinpullon
teline (pullo
valinnainen)

Kestävä muovikotelo, joka suojaa
sisällään olevia CAIRE Companion 5-laitteen osia.
Katkaisija, jonka avulla käynnistetään
tai lopetetaan virran syöttö
laitteeseen. Virransyöttö päällä
I-asennossa ja katkaistu O-asennossa.
Kun happi on suodatettu jakonsentroitu, se poistuu Companion 5-laitteesta tämän aukon kautta.
Potilasta varten aukkoon on liitettyjoko yksiputkinen nenäkanyyli
tai tavallinen happiputki.
Konsentraattoriin liitetyn happiputken
enimmäispituus on 15,2 m (50 jalkaa).
Sisältää vihreän, punaisen ja
keltaisen* LED-valon. Vihreä valo
tarkoittaa normaalia toimintaa.
Keltainen* ja punainen valo
ilmaisevat hälytystiloja. (Lisätietoja on
käyttöoppaan Hälytystilat-osassa.)
Konsentraattorin kuluneen käyttöajan
näyttö. Aikana näytetään lähin tunnin
kymmenesosa. Lukua ei voi nollata.
Näyttää diagnostiset hälytyskoodit,
kun konsentraattori havaitsee
hälytystilan.
Käytetään annostellun hapen
virtausnopeuden säätämiseen janäyttämiseen. Virtausnopeutta
voidaan säätää välillä 0,5–5 litraa
minuutissa (LPM). Säätäminentapahtuu säädintä kääntämällä.
Teline kostutinpulloa varten.
Telineeseen kuuluu joustava
kiinnityslenkki, jonka avulla
pullo pysyy vakaasti jalustalla.
Kostutinpullosta saadaan lisää
kosteutta happivirtaukseen. Pullo
täytetään tislatulla vedellä, ja siitä
saatava kosteus estää nenäkudosten
kuivumista.
*Vain happirikastimen tilailmaisimella
varustetuissa malleissa (OCSI-mallit)

174 - FIN PN 15062781-C4 J | Käyttöopa

3 Kostutinpullon ja kanyylin
kiinnittäminen
Jos käytät kostutinpulloa, täytä pullo ensin tislatulla
vedellä valmistajan määrittämään tasoon asti.
Aseta pullo sille tarkoitetulle jalustalle ja kiinnitä
joustavalla kiinnityslenkillä. Kiinnitä kostutinputken
kierteinen pää pulloon ja toinen Companion 5-laitteen ulostuloaukkoon. Kiinnitä happiputki ja/tai
nenäkanyyli kostutinpullon ulostuloaukkoon.

Jos ei käytetä kostutinpulloa, kiinnitä happiputki
ja/tai nenäkanyyli suoraan Companion 5 -laitteen
ulostuloaukkoon.

Companion 5
4 Käynnistäminen ja lämmitys
Kytke vaihtovirtajohto vaihtovirtapistorasiaan.

6 Säädä virtausasetus määräyksen
mukaiselle tasolle
Valitse lääkärin määräämä virtaus kääntämällä
virtausnopeuden säädintä.

Paina virtakytkin I-asentoon.

Virtakytkin

Kun Companion 5 käynnistetään oikein, LEDnäytölle ilmestyy vihreä virran merkkivalo. Kaikki
LED-valot syttyvät käynnistymisen yhteydessä. Kun
konsentraattori on lämmittänyt itsensä, vain vihreä
valo jää palamaan.
Huomautus: O2-merkkivalo syttyy ja näyttöön ilmestyy
O2-valo, kun konsentraatio saavuttaa tekniset tiedot.

VAROITUS: ON TÄRKEÄÄ, ETTÄNOUDATAT LÄÄKÄRIN SINULLE
MÄÄRÄÄMÄÄ HAPPITASOA. ÄLÄ MUUTAVIRTAUSASETUSTA ILMAN LÄÄKÄRIN
MÄÄRÄYSTÄ. HAPPIRIKASTINTA VOIKÄYTTÄÄ NUKKUESSA, JOS PÄTEVÄ
LÄÄKÄRI ON ANTANUT TÄSTÄSUOSITUKSEN.
Hengitä normaalisti nenäkanyylin kautta.
Virtausnopeuden säätäminen:
Käännä vastapäivään, jos haluat lisätä virtausta.
Käännä myötäpäivään, jos haluat vähentää virtausta.
Virtausnopeus katsotaan pallon puolivälin kohdalta.
Ks. alla oleva kuva.

Virranmerkkivalo

OikeinPN 15062781-C4 J | Käyttöopa

175 - FIN

Companion 5
Tärkeää!

Turvaohjeet on merkitty seuraavasti:
VAROITUS: TÄRKEITÄ TURVATIETOJAVAAROISTA, JOTKA VOIVAT AIHEUTTAAVAKAVIA VAMMOJA.
TÄRKEÄ HUOMAUTUS: Tärkeää
tietoa Companion 5 -laitteen
vahingoittumisen estämisestä.
Huomautus: nämä tiedot vaativat erityistä huomiota.

Käyttöindikaatiot
Käyttötarkoitus

Companion 5 on tarkoitettu varahapen annostelun
hallintaan. Tätä laitetta ei ole tarkoitettu elämää
ylläpitäviin sovelluksiin eikä siinä ole potilaan
monitorointiominaisuutta.
VAROITUS: ON TÄRKEÄÄ, ETTÄNOUDATAT LÄÄKÄRIN SINULLE
MÄÄRÄÄMÄÄ HAPPITASOA. ÄLÄ
MUUTA VIRTAUSASETUSTA ILMAN
LÄÄKÄRIN MÄÄRÄYSTÄ. KANNETTAVAAHAPPIRIKASTINTA VOI KÄYTTÄÄNUKKUMISEN AIKANA LÄÄKÄRINSUOSITUKSESTA.
VAROITUS: JOTTA VARMASTI SAISITTERVEYDENTILAASI VASTAAVAN HOITOANNOKSEN
HAPPEA, COMPANION 5 -LAITETTA SAA KÄYTTÄÄ
VASTA SITTEN, KUN SINULLE ON MÄÄRÄTTYTAI MÄÄRITETTY VÄHINTÄÄN YKSI LAITTEENASETUS, JOKA VASTAA HENKILÖKOHTAISTAAKTIIVISUUSTASOASI. COMPANION 5 -LAITTEENKANSSA ON KÄYTETTÄVÄ NIMENOMAAN SITÄOSA- JA LISÄVARUSTEKOKOONPANOA, JOKA
VASTAA HAPPIRIKASTIMEN VALMISTAJAN
MÄÄRITYKSIÄ JA JOTA KÄYTETTIIN SINULLESOVELTUVIA ASETUKSIA MÄÄRITETTÄESSÄ.
VAROITUS: TÄTÄ YKSIKKÖÄ EI OLE TARKOITETTUELÄMÄÄ YLLÄPITÄVIIN SOVELLUKSIIN. VANHUKSET,LAPSET TAI MUUT POTILAAT, JOTKA EIVÄT PYSTYKERTOMAAN EPÄMUKAVASTA OLOSTAAN, VOIVATEDELLYTTÄÄ TAVALLISTA TARKEMPAA SEURANTAATAI HÄLYTYSJÄRJESTELMÄN KÄYTTÖÄ, JOTTAHOITAJA SAISI TIEDON EPÄMUKAVASTA OLOSTATAI LÄÄKETIETEELLISESTÄ HÄTÄTILANTEESTA JA
VAMMOILTA VOITAISIIN VÄLTTYÄ. KUULO- JA/TAINÄKÖVAMMAISET POTILAAT VOIVAT TARVITA APUAHÄLYTYSTEN SEURANNASSA.

VAROITUS: RASKAANA OLEVAT TAI IMETTÄVÄT
NAISET EIVÄT SAA KÄYTTÄÄ TÄSSÄ OHJEKIRJASSASUOSITELTUJA VARUSTEITA, KOSKA NE VOIVATSISÄLTÄÄ FTALAATTEJA.
VAROITUS: YHDYSVALTAIN LIITTOVALTIONLAIN MUKAISESTI TÄTÄ LAITETTA SAAOSTAA TAI VUOKRATA VAIN LÄÄKÄRIN TAIMUUN LISENSOIDUN TERVEYDENHUOLLONPALVELUNTARJOAJAN MÄÄRÄYKSESTÄ.

Käytön vasta-aiheet
VAROITUS: JOISSAKIN OLOSUHTEISSAVOI OLLA VAARALLISTA ANNOSTELLALISÄHAPPEA ILMAN LÄÄKÄRIN
MÄÄRÄYSTÄ. TÄTÄ LAITETTATULEE KÄYTTÄÄ VAIN LÄÄKÄRIN
MÄÄRÄYKSESTÄ.
VAROITUS: EI SAA KÄYTTÄÄ HELPOSTI SYTTYVIENANESTESIA-AINEIDEN LÄHELLÄ.
VAROITUS: KUTEN KAIKKIEN SÄHKÖKÄYTTÖISTENLAITTEIDEN KANSSA, KÄYTTÄJÄ VOI KOKEA
SÄHKÖHÄIRIÖISTÄ JOHTUVIA KÄYTTÖKATKOKSIA,
TAI COMPANION 5 -LAITE SAATTAA OLLA TARPEEN
HUOLLATTAA VALTUUTETULLA TEKNIKOLLA.
COMPANION 5 EI SOVELLU POTILAILLE, JOILLETÄMÄNKALTAINEN TILAPÄINEN KÄYTTÖKATKOSAIHEUTTAA HAITALLISIA TERVEYDELLISIÄSEURAAMUKSIA.

Turvallisuusohjeet
VAROITUS: TÄTÄ LAITETTA EI SAA
MUOKATA.

VAROITUS: VALMISTAJA SUOSITTELEE
VARAHAPPIJÄRJESTELMÄÄ SÄHKÖKATKON,LAITTEEN HÄLYTYSTILAN TAI MEKAANISEN VIAN
VARALTA. PYYDÄ LISÄTIETOJA TARVITTAVASTAVARAJÄRJESTELMÄSTÄ LÄÄKÄRILTÄSI TAILAITETOIMITTAJALTASI.
VAROITUS: LAITE ANNOSTELEE ERITTÄIN
HAPPIPITOISTA ILMAA, JOKA KIIHDYTTÄÄ
PALAMISTA. TUPAKOINTI JA AVOTULENKÄYTTÄMINEN ON KIELLETTYÄ KAHTA (2) METRIÄ
LÄHEMPÄNÄ (1) TÄTÄ LAITETTA TAI (2) MUUTA
HAPPILISÄVARUSTETTA. TÄMÄN VAROITUKSENLAIMINLYÖMINEN VOI AIHEUTTAA VAKAVANTULIPALON, OMAISUUSVAHINKOJA, FYYSISIÄ
VAMMOJA TAI KUOLEMAN.
VAROITUS: ÄLÄ KÄYTÄ LAITETTA AHTAASSATAI SULJETUSSA TILASSA (KUTEN PIENESSÄLAATIKOSSA TAI KÄSILAUKUSSA), JOSSAILMANVAIHTO ON HEIKKO. SE VOI AIHEUTTAA
HAPPIRIKASTIMEN YLIKUUMENEMISEN JA TEHONHEIKKENEMISEN.

176 - FIN PN 15062781-C4 J | Käyttöopa

Companion 5
VAROITUS: HAPPIRIKASTINTA EI SAA
ALTISTAA SAVULLE, ILMANSAASTEILLEEIKÄ HÖYRYILLE.
VAROITUS: JOS KÄYTETÄÄN HAPPILISÄVARUSTEITA,JOITA EI OLE SUOSITELTU TÄLLE
HAPPIRIKASTIMELLE, LAITTEEN TEHO VOIHEIKENTYÄ. SUOSITELLUT LISÄVARUSTEET ON
MAINITTU TÄSSÄ KÄYTTÖOPPAASSA.
VAROITUS: JOS HAPPIRIKASTIN ON PUDONNUT,
VAHINGOITTUNUT TAI ALTISTUNUT VEDELLE,OTA YHTEYTTÄ KOTIHOIDON AMMATTILAISEENLAITTEEN TARKASTUSTA TAI MAHDOLLISTA
KORJAAMISTA VARTEN. ÄLÄ KÄYTÄ
HAPPIRIKASTINTA, JOS SEN VIRTAJOHTO TAIPISTOKE ON VAHINGOITTUNUT.
VAROITUS: ÄLÄ TUKI TULO- TAI POISTOILMAAUKKOJA. ÄLÄ PUDOTA TAI ASETA MITÄÄN
ESINEITÄ LAITTEEN AUKKOIHIN. SE VOI AIHEUTTAA
HAPPIRIKASTIMEN YLIKUUMENEMISEN JA TEHONHEIKKENEMISEN.
VAROITUS: ÄLÄ YLITÄYTÄ ILMANKOSTUTINTA(LISÄVARUSTE). TÄYTÄ ILMANKOSTUTIN(LISÄVARUSTE) VEDELLÄ ILMANKOSTUTTIMENVALMISTAJAN OSOITTAMAAN TASOON ASTI.
VAROITUS: ÄLÄ KÄYTÄ TÄTÄ LAITETTA, ENNEN
KUIN OLET LUKENUT TÄMÄN KÄYTTÖOPPAAN
JA YMMÄRTÄNYT, MITÄ SIINÄ SANOTAAN. JOSET YMMÄRRÄ VAROITUKSIA JA OHJEITA, OTA
YHTEYTTÄ LAITETOIMITTAJAAN ENNEN LAITTEENKÄYTTÄMISTÄ. MUUTOIN SEURAUKSENA VOI OLLAHENKILÖVAHINKO TAI LAITEVAURIO.
VAROITUS: JOS OLOSI ON EPÄMUKAVA TAIKOHTAAT LÄÄKETIETEELLISEN HÄTÄTILAN,
HAKEUDU VÄLITTÖMÄSTI LÄÄKÄRINHOITOON.
VAROITUS: JOS HAPPIRIKASTINTA KÄYTETÄÄNSUOSITELTUJEN KÄYTTÖLÄMPÖTILOJENULKOPUOLELLA, HAPPIRIKASTIMEN TOIMINTA
EI EHKÄ VASTAA TEKNISISSÄ TIEDOISSAILMOITETTUA. TARKISTA LÄMPÖTILAT TÄMÄN
KÄYTTÖOPPAAN TEKNISISTÄ TIEDOISTA.
VAROITUS: ÄLÄ KÄYTÄ ÖLJYÄ, RASVAA TAIMAAÖLJYPOHJAISIA TAI MUITA HELPOSTISYTTYVIÄ TUOTTEITA HAPPILISÄVARUSTEIDENTAI HAPPIRIKASTIMEN KANSSA. KÄYTÄ VAIN
VESIPOHJAISIA, HAPEN KANSSA YHTEENSOPIVIAKOSTEUSEMULSIOITA TAI VOITEITA. HAPPIKIIHDYTTÄÄ HELPOSTI SYTTYVIEN AINEIDEN
PALAMISTA.

VAROITUS: HAPPIRIKASTINTA EI SAA KÄYTTÄÄMUIDEN LAITTEIDEN VIERESSÄ TAI PINOTTUNAMUIDEN LAITTEIDEN KANSSA. JOS LAITETTAJOUDUTAAN KÄYTTÄMÄÄN MUIDEN LAITTEIDEN
VIERESSÄ TAI PINOTTUNA NIIDEN KANSSA,LAITETTA PITÄÄ VALVOA NORMAALIN TOIMINNANVARMISTAMISEKSI.
VAROITUS: VALMISTAJA SUOSITTELEE
VARAHAPPIJÄRJESTELMÄÄ
SÄHKÖKATKON, LAITTEEN HÄLYTYSTILAN
TAI MEKAANISEN VIAN VARALTA.
PYYDÄ LISÄTIETOJA TARVITTAVASTAVARAJÄRJESTELMÄSTÄ LÄÄKÄRILTÄSI
TAI LAITETOIMITTAJALTASI.
VAROITUS: LAITE ANNOSTELEE ERITTÄIN
HAPPIPITOISTA ILMAA, JOKA KIIHDYTTÄÄ
PALAMISTA. TUPAKOINTI JA AVOTULENKÄYTTÄMINEN ON KIELLETTYÄ SAMASSAHUONEESSA (1) TÄMÄN LAITTEEN KANSSA TAI (2)MUUN HAPPILISÄVARUSTEEN KANSSA. TÄMÄN
VAROITUKSEN LAIMINLYÖMINEN VOI AIHEUTTAAVAKAVAN TULIPALON, OMAISUUSVAHINKOJA,FYYSISIÄ VAMMOJA TAI KUOLEMAN.
VAROITUS: JOS KÄYTETÄÄN HAPPILISÄVARUSTEITA,JOITA EI OLE SUOSITELTU TÄLLE
HAPPIRIKASTIMELLE, LAITTEEN TEHO VOIHEIKENTYÄ. SUOSITELLUT LISÄVARUSTEET ON
MAINITTU TÄSSÄ KÄYTTÖOPPAASSA.
VAROITUS: JOS HAPPIRIKASTIN ON PUDONNUT,
VAHINGOITTUNUT TAI ALTISTUNUT VEDELLE,OTA YHTEYTTÄ KOTIHOIDON AMMATTILAISEENLAITTEEN TARKASTUSTA TAI MAHDOLLISTA
KORJAAMISTA VARTEN. ÄLÄ KÄYTÄ
HAPPIRIKASTINTA, JOS SEN VIRTAJOHTO TAIPISTOKE ON VAHINGOITTUNUT.
VAROITUS: KÄYTTÄMÄLLÄ AINOASTAAN
VALMISTAJAN SUOSITTELEMIA VARAOSIAVARMISTAT, ETTÄ LAITE TOIMII ASIANMUKAISESTIJA VÄLTÄT TULIPALON JA PALOVAMMOJEN
VAARAN.
VAROITUS: TULIPALON JA PALOVAMMOJEN
VAARAN VÄLTTÄMISEKSI HAPPIRIKASTIMENLIITÄNTÖJÄ, LIITTIMIÄ, LETKUJA TAI MUITALISÄVARUSTEITA EI SAA VOIDELLA.

PN 15062781-C4 J | Käyttöopa

177 - FIN

Companion 5
VAROITUS: TUPAKOINTI HAPPILAITTEENKÄYTTÄMISEN YHTEYDESSÄ ON YLEISIN SYY
TULIPALOIHIN JA NIIHIN LIITTYVIIN KUOLEMIIN.
NOUDATA NÄITÄ TURVAVAROITUKSIA:
VAROITUS: TUPAKOINTI, KYNTTILÖIDEN
POLTTAMINEN JA AVOTULEN KÄYTTÄMINENON KIELLETTYÄ SAMASSA HUONEESSA TÄMÄN
LAITTEEN TAI HAPPILISÄVARUSTEIDEN KANSSA.
VAROITUS: TUPAKOINTI HAPPIKANYYLIN
KÄYTTÄMISEN AIKANA VOI AIHEUTTAAPALOVAMMOJA KASVOILLE JA MAHDOLLISESTIKUOLEMAN.
VAROITUS: JOS POLTAT, NÄMÄ 3 VAIHETTA VOIVATPELASTAA HENKESI: SAMMUTA HAPPIRIKASTIN,
IRROTA KANYYLI JA POISTU HUONEESTA, JOSSALAITE ON.
VAROITUS: "TUPAKOINTI KIELLETTY – HAPPILAITE
KÄYTÖSSÄ" -KYLTIN ON OLTAVA NÄKYVÄSTI
ESILLÄ KOTONA TAI MUUSSA TILASSA, MISSÄ
HAPPILAITETTA KÄYTETÄÄN. POTILAILLE
JA HEIDÄN HOITAJILLEEN ON KERROTTAVATUPAKOINNIN VAAROISTA LÄÄKKEELLISEN HAPEN
LÄHEISYYDESSÄ JA SITÄ KÄYTETTÄESSÄ.
VAROITUS: ÄLÄ KÄYTÄ HAPPIRIKASTINTA HELPOSTISYTTYVIEN KAASUJEN LÄHELLÄ. SE VOI AIHEUTTAANOPEAN PALON, MISTÄ VOI OLLA SEURAUKSENAOMAISUUSVAHINKOJA, FYYSISIÄ VAMMOJA TAIKUOLEMA.
VAROITUS: IRROTETUN KANYYLIN ASETTAMINENVAATTEIDEN, VUODEVAATTEIDEN, SOHVANTAI MUUN TEKSTIILIMATERIAALIN PÄÄLLE VOIAIHEUTTAA RÄJÄHTÄVÄN PALON, JOS KANYYLIALTISTUU SAVUKKEELLE, LÄMMÖNLÄHTEELLE TAIAVOTULELLE.
VAROITUS: ÄLÄ JÄTÄ NENÄKANYYLIA VAATTEIDEN,
VUODEVAATTEIDEN TAI ISTUINTYYNYJEN PÄÄLLETAI ALLE. JOS LAITTEEN VIRTA ON KYTKETTY MUTTA
SE EI OLE KÄYTÖSSÄ, HAPPI TEKEE MATERIAALISTAHELPOSTI SYTTYVÄN. ASETA I/0-VIRTAKYTKIN
0 (OFF) -ASENTOON, KUN HAPPIRIKASTINTA EIKÄYTETÄ.
VAROITUS: ÄLÄ KOHDISTA LAITTEESEENMITÄÄN HUOLTOTOIMIA LUKUUNOTTAMATTA TÄSSÄ KÄYTTÖOPPAASSA
MAINITTUJA RATKAISUJA. ÄLÄ POISTAPANEELEJA. VAIN LAITETOIMITTAJA TAIPÄTEVÄ HUOLTOTEKNIKKO SAA POISTAALAITTEEN PANEELIT TAI HUOLTAALAITETTA.

178 - FIN PN 15062781-C4 J | Käyttöopa

VAROITUS: KÄYTÄ VAIN VALMISTAJANSUOSITTELEMIA LISÄVARUSTEITA. MUIDEN
LISÄVARUSTEIDEN KÄYTTÄMISESTÄ VOI AIHEUTUAVAARAA, HAPPIRIKASTIN VOI VAHINGOITTUAVAKAVASTI TAI TAKUU VOI KUMOUTUA.
TÄRKEÄ HUOMAUTUS: Irrota
vaihtovirtalähde seinäpistorasiasta aina
ennen kuin irrotat vaihtovirtalähteen
happirikastimesta.
TÄRKEÄ HUOMAUTUS: Reititä happiletku
ja virtajohdot aina niin, etteivät ne aiheuta
kompastumisvaaraa.
TÄRKEÄ HUOMAUTUS: Noudata
valmistajan ohjeita, jotta takuu ei
kumoudu.
Huomautus: kannettavat ja siirrettävät radiotaajuiset
viestintävälineet voivat häiritä sähkökäyttöisiä
lääkintälaitteita.
VAROITUS: SIJOITA AINA HAPPEA SISÄLTÄVÄT
PUTKET JA VIRTAJOHDOT SEINÄSSÄ, JOKA ESTÄÄ
MATKAN VAARAN TAI MAHDOLLISEN VAHINGOSSATAPAHTUVAN VAHINGON.
VAROITUS: TÄMÄ TUOTE VOI ALTISTAA SINUTKEMIKAALEILLE, MUKAAN LUKIEN NIKKELI, JOTKA
KALIFORNIAN OSAVALTION MUKAAN VOIVATAIHEUTTAA SYÖPÄÄ. LISÄTIETOJA ON OSOITTEESSA
WWW.P65WARNINGS.CA.GOV.
VAROITUS: JOS LAITTEESSA TAPAHTUU VAKAVAVAARATILANNE, KÄYTTÄJÄN TULEE ILMOITTAA TAPAUKSESTA VÄLITTÖMÄSTI PALVELUNTARJOAJALLE
JA / TAI VALMISTAJALLE. VAKAVANA TAPAHTUMANA
MÄÄRITELLÄÄN LOUKKAANTUMINEN, KUOLEMA TAI MAHDOLLISUUS AIHEUTTAA VAMMOJA /KUOLEMIA, JOS TAPAHTUMA TOISTUU. KÄYTTÄJÄ
VOI MYÖS ILMOITTAA TAPAHTUMASTA SEN MAAN
TOIMIVALTAISELLE VIRANOMAISELLE, JOSSA TAPAHTUMA TAPAHTUI.

Companion 5
Tekniset tiedot
Seuraava taulukko sisältää tärkeitä tietoja suositelluista
käyttöympäristöistä ja käyttöolosuhteista, jotka
takaavat laitteen asianmukaisen toiminnan.

Tekniset tiedot
Virtausnopeudet
O2-pitoisuus
Mitat

0,5–5,0 LPM
90 % (+5,5 %/-3 %)54,6 x 31,8 x 34,2 cm(21,5 x 12,5 x 13,5”)
Paino
16,3 kg (36 lbs)
Äänenpainetaso 50 dB(A) / 2,0 LPM
Virrankulutus
285 W / 2 LPM, enintään 350 W
Käyttölämpötilat 5–40 °C (41 ºF–104 ºF)
Käyttöympäristön 15% - 90% 28 °C:n (82,4 °F)
kosteuskastepistelämpötilassa
Säilytyslämpötila -25–70 °C (-13 ºF–158 ºF)
Säilytystilan
0% - 90% tiivistymätön
kosteus
Sähkö
Älä käytä jatkojohtoa. Älä
käytä kytkimellä varustettua
sähköpistorasiaa.
Käyttökorkeus -382–3 011 metriä(-1 253–9 879 jalkaa)
Käyttöaika
jopa 24 tuntia vuorokaudessa
Laitteen odotettavissa oleva käyttöikä on vähintään
viisi vuotta.
VAROITUS: LAITTEEN KÄYTTÖ
TIETTYJEN KÄYTTÖOLOSUHTEIDENULKOPUOLELLA ODOTETAANVAIKUTTAVAN HAITALLISESTI HAPPEASISÄLTÄVÄN VIRTAUKSEN MÄÄRÄÄN
JA PROSENTTIMÄÄRÄÄN JA SENSEURAUKSENA TERAPIAN LAATUUN.
Huomautus: Jos happirikastinta on säilytetty pitkään
sen normaalista käyttölämpötilasta poikkeavassa
lämpötilassa, anna laitteen palautua normaaliin
käyttölämpötilaan, ennen kuin käynnistät sen. (Katso
lisätietoja tämän käyttöoppaan teknisistä tiedoista.)

Companion 5 -laitteen sijoittaminen

Sijoita laite paikkaan, jossa sen sisään ei pääse
savua, kaasuja ja epäpuhtauksia. Laitteen oikean
sijoituksen on mahdollistettava ilman tulo kolmen
ilmanottokohdan kautta kotelon ylä- ja alapuolella sekäpoistoilman vapaa poistuminen laitteen vasemmassa
alakulmassa olevan ilmanpoistoaukon kautta.
Ilmanottoaukot ovat konsentraattorin takana ylhäällä
ja alhaalla. Sijoita laite niin, että konsentraattorin ja
seinien, huonekalujen, verhojen tai muiden esteiden
väliin jää vähintään 30 cm tilaa.
Sijoita laite niin, että hälytykset kuuluvat.
Reititä happiletku niin, ettei se mene mutkalle eikä
pääse tukkeutumaan.
VAROITUS: ÄLÄ KÄYTÄ
HAPPIRIKASTINTA HELPOSTISYTTYVIEN KAASUJEN LÄHELLÄ. SE
VOI AIHEUTTAA NOPEAN PALON,MISTÄ VOI OLLA SEURAUKSENAOMAISUUSVAHINKOJA, FYYSISIÄ
VAMMOJA TAI KUOLEMA. ÄLÄ KÄYTÄ
ÖLJYÄ, RASVAA TAI MAAÖLJYPOHJAISIATAI MUITA HELPOSTI SYTTYVIÄ
TUOTTEITA HAPPILISÄVARUSTEIDENTAI HAPPIRIKASTIMEN KANSSA. KÄYTÄ
VAIN VESIPOHJAISIA, HAPEN KANSSAYHTEENSOPIVIA KOSTEUSEMULSIOITATAI VOITEITA. HAPPI KIIHDYTTÄÄHELPOSTI SYTTYVIEN AINEIDEN
PALAMISTA.
VAROITUS: LAITE ANNOSTELEE ERITTÄIN
HAPPIPITOISTA ILMAA, JOKA KIIHDYTTÄÄ
PALAMISTA. TUPAKOINTI JA AVOTULENKÄYTTÄMINEN ON KIELLETTYÄ SAMASSAHUONEESSA (1) TÄMÄN LAITTEEN KANSSA TAI(2) MUUN HAPPILISÄVARUSTEEN KANSSA. TÄMÄN
VAROITUKSEN LAIMINLYÖMINEN VOI AIHEUTTAAVAKAVAN TULIPALON, OMAISUUSVAHINKOJA,FYYSISIÄ VAMMOJA TAI KUOLEMAN.

PN 15062781-C4 J | Käyttöopa

179 - FIN

Companion 5
Käyttöohjeet
Ennen käyttöä

Tämä käyttöopas toimii apunasi laitteen käyttöä
ja huoltoa koskevissa kysymyksissä. Jos sinulla
on kysymyksiä tai ongelmia, ota yhteyttä
kotihoitopalveluusi.
Tärkeää! ÄLÄ yritä käyttää Companion 5 -laitetta
ennen kuin olet ensin lukenut tämän käyttöoppaan
Turvallisuusohjeet-kappaleen. Noudata kaikkia
käyttöohjeita. Huomioi kaikki laitteessa ja
käyttöoppaassa annetut vaarailmoitukset. Noudata
kaikkia turvallisuusvarotoimia vähentääksesi tulipalon,
henkilövahinkojen ja Companion 5 -laitteelle
aiheutuvien vakavien vahinkojen vaaraa.
CAIRE Companion 5 ei luo happea. Se tuottaa
potilaalle korkeapitoista happea ympäröivästä
huoneilmasta. Huoneilman luonnollinen happipitoisuus
on noin 21 %.
Konsentraattori vetää itseensä huoneilmaa
kompressorin avulla. Ilma siirtyy tämän jälkeen
useiden suodattimien ja typpimolekyylejä adsorboivan
molekyylisihdin läpi. Näin konsentraattorista lähtevän
hapen happipitoisuudeksi saadaan 87–95 %.
VAROITUS: PUHDISTA LAITTEEN KOTELO,OHJAUSPANEELI JA VIRTAJOHTO VAINMIEDOLLA KOTITALOUSKÄYTTÖÖNTARKOITETULLA PESUAINEELLA.
KÄYTÄ KOSTEAA (EI MÄRKÄÄ)
LIINAA TAI SIENTÄ JA PYYHI LOPUKSIPINNAT KUIVIKSI. LAITTEEN SISÄÄN
EI SAA PÄÄSTÄÄ MITÄÄN NESTETTÄ.
KIINNITÄ ERITYISTÄ HUOMIOTA
HAPEN ULOSTULOPORTISSA OLEVAAN
KANYYLIN KIINNITYSKOHTAAN, JOTTA
SIINÄ EI OLE PÖLYÄ, VETTÄ EIKÄPIENHIUKKASIA.
TÄRKEÄ HUOMAUTUS: Älä tuki tulo- taipoistoilma-aukkoja. Se voi aiheuttaa
happirikastimen ylikuumenemisen
ja tehon heikkenemisen. Älä käytä
happirikastinta, jos tuloilmasuodatin ei
ole paikallaan. Jos mukana on toinen
suodatin, aseta varasuodatin, ennen
kuin puhdistat likaisen suodattimen.
Puhdista likainen suodatin lämpimällä
saippuavesiliuoksella ja anna sen kuivua
kokonaan ennen käyttöä.

180 - FIN PN 15062781-C4 J | Käyttöopa

Vaihe 1: Companion 5 -laitteen sijoittaminen
käyttöä varten
Aseta Companion 5 paikkaan, jossa ilmanvaihto ja
valaistus ovat hyvät. Huolehdi, että mikään ei tuki
ilmanotto- ja ilmanpoistoaukkoja. Sijoita Companion 5
paikkaan, josta kaikki äänimerkit, visuaaliset merkit ja
hälytykset näkyvät ja kuuluvat hyvin.
Kytke laite vaihtovirtapistorasiaan.
Vaihe 2: Jos kostutinpulloa EI KÄYTETÄIlmanottoaukko

Jäähdytysilman
tuloaukko
Ilmanpoistoaukko
Ilmanottoaukko

a. Kiinnitä nenäkanyyli tai happiputki suoraan hapen
ulostuloaukkoon.

b. Siirry vaiheeseen 4.
Vaihe 3: Jos kostutinpulloa KÄYTETÄÄN
a. Irrota kostutinpullon kannen ruuvit ja irrota kansi.
b. Täytä pullo tislatulla vedellä. Varmista, että vedentaso on valmistajan määrittämien enimmäis- javähimmäistasojen välillä.
c. Kiinnitä kostutinpullon kansi. Varmista, että vuotoja
ei ole.
d. Kiinnitä nenäkanyyli tai tavallinen happiputki
kostutinpullon ulostuloaukkoon.
Huomautus: Konsentraattorin toiminta voi häiriintyä,
jos sen kanssa käytetään muuta kuin sen kanssa
käytettäväksi määritettyä kosteutinta.

Companion 5g. Aseta kostutinpullo sille tarkoitettuun telineeseen.
h. Kiinnitä pullo joustavan kiinnityslenkin avulla.

e. Kiinnitä kostutinputken kierteinen pää
kostutinpullon kanteen.

i. Siirry vaiheeseen 4.
Vaihe 4: Käynnistäminen ja lämmitys
a. Kytke CAIRE Companion 5 -laitteen sähköjohto
vaihtovirtapistorasiaan.
VAROITUS: VARMISTA ENNEN
SÄHKÖJOHDON KÄYTTÖÄ, ETTEISIINÄ OLE VAURIOITA. JOS SE ON
VAURIOITUNUT, ÄLÄ KYTKE SITÄSÄHKÖPISTORASIAAN ÄLÄKÄ YRITÄ
KÄYTTÄÄ KONSENTRAATTORIA.

f. Kiinnitä kostutinputken toinen pää CAIRE
Companion 5 -laitteen ulostuloaukkoon.

b. Paina virtakytkin "Päällä" (I) -asentoon. Kun
Companion 5 käynnistetään oikein, LED-näytölle
ilmestyy vihreä merkkivalo. Kaikki LEDvalot syttyvät käynnistymisen yhteydessä. Kun
konsentraattori on lämmittänyt itsensä, vain vihreä
valo jää palamaan.
Huomautus: Kun laite on ensimmäisen kerran kytketty
päälle, odota kymmenen (10) minuuttia, kunnes laite
on saavuttanut toimintavaatimukset täyttävän tilan.

PN 15062781-C4 J | Käyttöopa

181 - FIN

Companion 5
VAROITUS: ON TÄRKEÄÄ, ETTÄNOUDATAT LÄÄKÄRIN SINULLE
MÄÄRÄÄMÄÄ HAPPITASOA. ÄLÄ MUUTAVIRTAUSASETUSTA ILMAN LÄÄKÄRIN
MÄÄRÄYSTÄ. HAPPIRIKASTINTA VOIKÄYTTÄÄ NUKKUESSA, JOS PÄTEVÄ
LÄÄKÄRI ON ANTANUT TÄSTÄSUOSITUKSEN.
Vaihe 6: Virtauksen tarkistaminen ja normaali
hengitys
Vaihe 5: Virtausnopeuden säätäminen
a. Säädä hapen virtausnopeus (LPM) lääkärisi
määräyksen mukaiseksi kääntämällä
virtausnopeuden säädintä.

a. Tarkista, että happi todella virtaa nenäkanyylin läpi
ja ettei putki ole taittunut tai tukossa.
Huomautus: Varmista, että kanyyli on asetettu
kokonaan ja pitävästi. Kun kanyyli on kiinnitetty
oikein, happivirtauksen nenäkanyylin haarakkeisiin
pystyy kuulemaan tai tuntemaan.
b. Aseta nenäkanyyli asianmukaisesti ja hengitä
normaalisti nenäkanyylin kautta.
Vaihe 7: Sammutus
a. Paina virtakytkin "POIS" (0) -asentoon, jolloin
CAIRE Companion 5 ei enää ole käytössä.

Virtausnopeuden säätäminen:
Käännä vastapäivään, jos haluat lisätä virtausta.
Käännä myötäpäivään, jos haluat vähentää virtausta.
Virtausnopeus katsotaan pallon puolivälin kohdalta.
Alla olevassa kuvassa virtausnopeus on 4,0 LPM.

TÄRKEÄ HUOMAUTUS: Käytä vain
valmistajan suosittelemia lisävarusteita.
Muiden lisävarusteiden käyttämisestä
voi aiheutua vaaraa, happirikastin voi
vahingoittua vakavasti tai takuu voi
kumoutua.
TÄRKEÄ HUOMAUTUS: Älä käytä jatkojohtoja
tämän laitteen kanssa äläkä kytke liikaa
sähkölaitteita samaan pistorasiaan. Jatkojohtojen
käyttäminen voi heikentää laitteen tehoa
huomattavasti. Jos yhteen pistorasiaan kytketään
liikaa sähkölaitteita, sähkötaulu voi kuormittua,
mikä voi laukaista katkaisimen/sulakkeen tai
aiheuttaa tulipalon, jos katkaisin/sulake ei toimi.
TÄRKEÄ HUOMAUTUS: Muiden kuin määritettyjen
kaapeleiden tai sovittimien käyttäminen, lukuun
ottamatta lääkinnällisen sähkölaitteen valmistajan
toimittamia kaapeleita ja sovittimia, jotka ontarkoitettu sisäisten osien varaosiksi, voi lisätä
happirikastimen päästöjä tai vähentää sen
suojausta.

Oikein

182 - FIN PN 15062781-C4 J | Käyttöopa

Companion 5
VAROITUS: ÄLÄ TUKI TULO- TAI
POISTOILMA-AUKKOJA. SE VOIAIHEUTTAA HAPPIRIKASTIMENYLIKUUMENEMISEN JA TEHONHEIKKENEMISEN.
Huomautus: Laitevalmistajalle: Seuraavia hapen
annostelun lisävarusteita suositellaan käytettäväksi
Companion 5 kanssa:• nenänkanyyli: CAIRE-osanumero CU002-1• Kostutinsovittimen letkut: CAIRE Tuotenumero
20843882• Kostutinpullo: CAIRE-osanumero HU003-1• Tulipalo: CAIRE-osanumero 20629671
Tulipalo vaaditaan käytettäväksi minkä tahansa
kanyylin kanssa.
• CAIRE tarjoaa palomuurin, joka on tarkoitettu
käytettäväksi happikonsentraattorin kanssa. Tulipalo
on lämpövaroke, joka estää kaasun virtauksen, josalavirran kanyyli tai happiputki sytytetään ja palaa
paloputkeen. Se sijoitetaan linjaan nenän kanyylin tai
happiputken kanssa potilaan ja Companion 5 happea
poistoaukon väliin. Tietoja palomuurin oikeasta
käytöstä on aina valmistajan ohjeissa (joka sisältyy
jokaiseen palopalorasiaan).
• Lisätietoja mahdollisista lisävarusteista on
lisävarusteluettelossa (PN MLLOX0010), joka on
saatavana osoitteessa www.caireinc.com.

PN 15062781-C4 J | Käyttöopa

183 - FIN

Companion 5
Hälytystilat
CAIRE Companion 5 varoittaa konsentraattorin
toimintahäiriöistä kahdella tai kolmella (OCSImalleissa) LED-merkkivalolla ja äänihälytyksellä.
1) Vihreä merkkivalo (Vihreä

) – normaali toiminta.

2) Punainen merkkivalo ( ) – järjestelmän
toimintahäiriö. Lisätietoja on alla olevassa
hälytystilataulukossa.
3) Keltainen merkkivalo () – alhainen
happirikasteen happipitoisuus. Lisätietoja on alla
olevassa hälytystilataulukossa.
Seuraavalla sivulla olevassa hälytystilataulukossa
näkyvät eri hälytystilat, joita voidaan näyttää
konsentraattorissa. Taulukossa kerrotaan hälytysten
mahdollisia syitä sekä toimenpiteitä ongelmien
poistamiseksi. Hälytystilanteessa on suositeltavaa
ottaa mahdollisimman pian yhteyttä terveydenhuollon
ammattilaiseen.
VAROITUS: ÄLÄ SIVUUTA HÄLYTYKSIÄ
VAARAILMOITUKSET: VALMISTAJASUOSITTELEE VARAHAPPIJÄRJESTELMÄÄ
SÄHKÖKATKON, LAITTEEN HÄLYTYSTILAN
TAI MEKAANISEN VIAN VARALTA.
PYYDÄ LISÄTIETOJA TARVITTAVASTAVARAJÄRJESTELMÄSTÄ LÄÄKÄRILTÄSI
TAI LAITETOIMITTAJALTASI.

184 - FIN PN 15062781-C4 J | Käyttöopa

Punainen

Keltainen

Yllä: Companion 5

Companion 5
HälytystilataulukkoÄänihälytys

Merkkivalo

Poisvihreä

Ajoittainen
punainen

Ajoittainen

Syy
Konsentraattori toimii
oikein.
Konsentraattori ei saa
sähkövirtaa, mutta
virtakytkin on edelleen"Päällä"-asennossa.

Varmista, että konsentraattorin sähköjohto on
kytketty pistorasiaan ja että pistorasia antaa
virtaa. Kokeile toista pistorasiaa. Jos ongelma
jatkuu, yhdistä varahappilähteeseen ja ota yhteyttä
terveydenhuollon ammattilaiseen.

Järjestelmän
toimintahäiriö

Käännä virtakytkin välittömästi "Pois"-asentoon.
Irrota vaihtovirtajohto pistorasiasta. Odota
5 minuuttia. Kytke vaihtovirtajohto takaisin
pistorasiaan ja käynnistä konsentraattori
uudelleen. Jos laite hälyttää edelleen, sevaatii huoltoa. Yhdistä varahappilähteeseen
ja ota välittömästi yhteyttä terveydenhuollon
ammattilaiseen.

Laitteen virtausnopeus
on liian suuri tai pieni.

1. Tarkista, että kanyyli ei ole kiertynyt tai
tukkeutunut. Jos ilmankostuttimen säiliö on
kiinnitetty, varmista, että se on täytetty oikein ja
että se ei aiheuta tukosta.
2. Varmista, että Companion 5 -laitteen
ilmanvaihto on asianmukainen. Sen on oltava
vähintään 30 cm (12 tuumaa) irti pinnoista, jotta
ilma-aukot eivät tukkiudu.
3. Jos ongelma ei häviä, ota käyttöön toinenhappilähde ja pyydä apua terveydenhuollon
ammattilaiselta.

punainen

Ajoittainen
punainen

Companion 5 on
havainnut alhaisia
happitasoja.

Ajoittainen
keltainen

Companion 5 on
havainnut alhaisia
happitasoja.

Ajoittainen

Miten toimit
Ei mitään.

1. Varmista, että tuloilmasuodatin ja poistoilman
kohdat eivät ole tukossa tai niiden edessä ei
ole esteitä. 2. Varmista, että Companion 5 onpaikassa, jossa ilmanvaihto on hyvä. Varmista, ettäCompanion 5 -laitteen takaosan ja mahdollisten
esteiden (huonekalujen, verhojen jne.) välissä onvähintään 30 cm.
3. Jos ongelma ei häviä, ota käyttöön toinenhappilähde ja ota välittömästi yhteyttä
terveydenhuollon ammattilaiseen.

keltainen

PN 15062781-C4 J | Käyttöopa

185 - FIN

Companion 5
Vianmääritys
On suositeltavaa ottaa yhteyttä terveydenhuollon
ammattilaiseen, jos konsentraattori näyttää hälytyksiä
tai jos olet sitä mieltä, että se ei toimi oikein. Älä
yritä korjata konsentraattoria itse. Vain koulutetulla
huoltoteknikolla on lupa suorittaa korjaustyö. On
kuitenkin joitakin vianmääritysvaiheita, joita voit

suorittaa omatoimisesti CAIRE Companion 5 -laitteen
ongelmatilanteissa. Vaiheet on kerrottu seuraavassa
taulukossa. Suorita ongelmaan liittyvät tarkistukset
numerojärjestyksessä.

Käyttäjän vianetsintätaulukko
OngelmaKonsentraattori
on päällä,mutta se ei ole
käynnissä.

Happivirtaus
on hiljainen
tai sitä ei ole
lainkaan.

Mahdollinen syy

Miten toimit

1. Sähköjohtoa ei ole kytketty
pistorasiaan.

1. Kytke sähköjohto.

2. Sähköpistorasia ei anna virtaa.

2. Tarkista kodin sulake ja virtapiiri. Kokeile toista
pistorasiaa.

3. Sisäinen virhe.

3. Yhdistä varahappilähteeseen. Ota välittömästi
yhteyttä terveydenhuollon ammattilaiseen.

1. Virtausnopeusasetusta on muutettu.

1. Varmista, että virtausnopeuden säätimellä on valittu
lääkärin määräämä virtausnopeus (litraa minuutissa, LPM).

2. Putki on irti.

2. Varmista, että putkiliitännät ovat tiukat kaikkialla(ulostuloaukko, kostutinpullo, vesilukko jne.). Kytke
liitännät uudelleen tarpeen mukaan.

3. Happiputki tai kanyyli on mutkalla.

3. Tarkasta, onko putkissa mutkia tai tukoksia. Korjaa
tarpeen mukaan.

4. Kanyylissa tai putkessa on vuoto.

4. Etsi vuotoja putkista ja kanyylista kuuntelemalla tai
tunnustelemalla, karkaako ilmaa. Vaihda vuotava putki
tai kanyyli.

5. Ilman virtaus konsentraattoriin on
estynyt.

5. Varmista, että mikään ei peitä konsentraattorin
ilmanottoaukkoja. Varmista, että konsentraattori on
vähintään 30 cm:n päässä huonekaluista, seinistä tai
muista esteistä.

6. Sisäinen virhe.

6. Yhdistä varahappilähteeseen. Ota välittömästi
yhteyttä terveydenhuollon ammattilaiseen.

Huomautus: Jos Companion 5 hälyttää senkin jälkeen, kun olet kokeillut hälytystila- ja vianmääritystaulukoissa
annettuja ratkaisuja, ota yhteyttä tekniseen tukeen soittamalla numeroon 1-800-482-2473 ja varaudu kertomaan
LCD-näytössä näkyvä hälytyskoodi.

186 - FIN PN 15062781-C4 J | Käyttöopa

Companion 5
Puhdistus, ylläpito ja huolto
HuoltoHuolto koostuu kahvan karkeasuodattimen
vaihtamisesta kerran viikossa (jos asennettu).

Käyttäjän toimet ja laitteen puhdistus

Kanyylin vaihtaminen
Vaihda syöttöletku ja kanyyli säännöllisin väliajoin
kotihoidon ammattilaiselta saamiesi suositusten
mukaan. Lääkärisi tai kotihoitopalvelusi antaa sinulle
puhdistus-, desinfiointi- ja vaihto-ohjeet.
Huomautus: Noudata aina kanyylin valmistajan
käyttöohjeita. Vaihda kertakäyttöinen kanyyli kanyylin
valmistajan tai laitetoimittajasi suositusten mukaisesti.
Lisätarvikkeita on saatavana laitetoimittajalta.

Tuloilmasuodatin (patruuna)
Vain hoitohenkilökunta saa käyttää tai vaihtaa
tuloilmasuodatinta.
Karkeasuodatin (valinnainen):

Bruttohiukkassuodatin sijaitsee Companion 5: naivan alaosassa.

Huomautus: Valmistaja ei suosittele tämän laitteen
sterilointia.
Huomautus: Älä käytä happirikastinta, jostuloilmasuodatin ei ole paikallaan.
Kostutinpullo (jos käytettävissä)
Tarkista kostutinpullo päivittäin ja varmista, että
sen veden taso on määritettyjen vähimmäis- ja
enimmäistasojen välillä. Täytä pullo tarpeen mukaan
tislatulla vedellä. Puhdista kostutinpullo valmistajan
ohjeiden mukaisesti.
Puhdistus:
Sammuta Companion 5 -laitteen virta ja irrota
vaihtovirtalähteestä ennen puhdistus- taidesinfiointitoimenpiteitä. ÄLÄ suihkuta nesteitä
suoraan ulkokoteloon. Käytä kosteaa (ei märkää) liinaa
tai sientä. Suihkuta liinaan tai pesusieneen mietoa
puhdistusliuosta ja puhdista kotelo ja virtajohto.
Käytä Companion 5 -laitteen desinfiointiin Lysol®
Brand II -desinfiointiainetta (tai vastaavaa). Laimenna
puhdistustuotteen valmistajan ohjeiden mukaisesti,
mutta älä suihkuta nestettä suoraan Companion 5-laitteeseen.
VAROITUS: SÄHKÖISKUN VAARA.
SAMMUTA LAITE JA IRROTA VIRTAJOHTOSÄHKÖLIITÄNNÄSTÄ, ENNEN KUIN
PUHDISTAT LAITTEEN. MUUTOINSEURAUKSENA ON SÄHKÖISKUNJA PALOVAMMAN VAARA. VAINLAITETOIMITTAJA TAI PÄTEVÄHUOLTOTEKNIKKO SAA POISTAALAITTEEN PANEELIT TAI HUOLTAALAITETTA.

Karkeasuodatin

Tarkista ja puhdista se kerran viikossa. Puhdista
suodatin seuraavalla tavalla:

VAROITUS: SÄHKÖISKUN VAARA. SAMMUTA LAITE
JA IRROTA VIRTAJOHTO SÄHKÖLIITÄNNÄSTÄ,ENNEN KUIN PUHDISTAT LAITTEEN. MUUTOINSEURAUKSENA ON SÄHKÖISKUN JA PALOVAMMAN
VAARA. VAIN LAITETOIMITTAJA TAI PÄTEVÄHUOLTOTEKNIKKO SAA POISTAA LAITTEENPANEELIT TAI HUOLTAA LAITETTA.

1. Irrota karkeasuodatin.
2. Pese suodatin lämpimällä vesijohtovedellä ja
miedolla saippualiuoksella.
3. Huuhtele suodatin huolellisesti pelkällä
vesijohtovedellä ja purista ylimääräinen vesi pois.
4. Anna suodattimen ilmakuivua.
5. Aseta suodatin takaisin koteloon.

PN 15062781-C4 J | Käyttöopa

187 - FIN

Companion 5
VAROITUS: SÄHKÖISKUN VAARA. IRROTAVIRTAJOHTO SÄHKÖLIITÄNNÄSTÄ, ENNEN KUIN
PUHDISTAT LAITTEEN. MUUTOIN SEURAUKSENAON SÄHKÖISKUN JA PALOVAMMAN VAARA. VAINLAITETOIMITTAJA TAI PÄTEVÄ HUOLTOTEKNIKKOSAA POISTAA LAITTEEN PANEELIT TAI HUOLTAALAITETTA.
VAROITUS: HUOLEHDI, ETTÄ HAPPIRIKASTIN
EI PÄÄSE KASTUMAAN JA ETTÄ LAITTEENSISÄÄN EI PÄÄSE NESTEITÄ. TÄSTÄ VOI SEURATA
TOIMINTAHÄIRIÖ TAI LAITTEEN SAMMUMINEN,
JA SÄHKÖISKUN TAI PALOVAMMOJEN RISKI VOILISÄÄNTYÄ.
VAROITUS: YRITYSTEN 5 KYTKENTÄMINEN MUIDENLAITTEIDEN KANSSA VOI TYÖTYÖTÖMYYTÖN.

VAROITUS: PUHDISTA LAITTEEN KOTELO,OHJAUSPANEELI JA VIRTAJOHTO VAIN MIEDOLLA
KOTITALOUSKÄYTTÖÖN TARKOITETULLAPESUAINEELLA. KÄYTÄ KOSTEAA (EI MÄRKÄÄ)
LIINAA TAI SIENTÄ JA PYYHI LOPUKSI PINNATKUIVIKSI. LAITTEEN SISÄÄN EI SAA PÄÄSTÄÄ
MITÄÄN NESTETTÄ. KIINNITÄ ERITYISTÄHUOMIOTA HAPEN ULOSTULOPORTISSA OLEVAAN
KANYYLIN KIINNITYSKOHTAAN, JOTTA SIINÄ EI OLEPÖLYÄ, VETTÄ EIKÄ PIENHIUKKASIA.
VAROITUS: ÄLÄ ALTISTA LAITETTA NESTEILLE. VÄLTÄ
NÄITÄ KEMIALLISIA AINEITA (EI TÄYDELLINENLUETTELO): ALKOHOLI JA ALKOHOLIPOHJAISETTUOTTEET, TIIVISTETYT KLOORIPOHJAISETTUOTTEET (ETYLEENIKLORIDI) SEKÄ ÖLJYPOHJAISETTUOTTEET (PINE-SOL®, LESTOIL®). NÄITÄ EI SAAKÄYTTÄÄ HAPPIRIKASTIMEN MUOVIKOTELONPUHDISTAMISEEN, SILLÄ NE VOIVAT VAURIOITTAALAITTEEN MUOVIOSIA.

Hävittäminen
Palauta Companion 5 kaikkine osineen hävitettäväksi
kotihoitoon. Voit myös ottaa yhteyttä paikalliseenkaupungin tai kunnan viranomaiseen ja kysyä ohjeita
akun asianmukaisesta hävittämisestä.

WEEE ja RoHS

Tuotteemme täyttävät vaarallisia aineita koskevan
RoHS-direktiivin vaatimukset. Tuotteissamme on vain
vähäisiä jäämiä lyijystä ja muista vaarallisista aineista.
VAROITUS: Jotta hävittäminen olisi
asianmukaista, ota yhteys laitteen toimittajaan tai paikallishallinnon virastoon
hävittämisohjeita varten.

Tämä symboli muistuttaa laitteen omistajaa
siitä, että elinkaarensa lopussa laite on kierrätettävä
sähkö- ja elektroniikkalaiteromudirektiivin (WEEE)
mukaisesti.

CAIRE ja CAIRE Inc. ovat CAIRE Inc: n rekisteröityjä
tavaramerkkejä. Katso täydellinen luettelo tavaramerkeistä alla
olevalta verkkosivustoltamme.
Tavaramerkit: www.careinc.com/corporate/trademarks.
Tekijänoikeus © 2020 CAIRE Inc.
CAIRE Inc. pidättää
oikeuden lakkauttaa tuotteitaan tai muuttaa hintoja, materiaaleja,
laitteita, laatua, nimityksiä, teknisiä tietoja ja/tai tuotteiden
valmistustapoja milloin tahansa ilman ennakkoilmoitusta ja ilman,
että tästä aiheutuu sille lisävelvoitteita tai seuraamuksia. Kaikki
oikeudet, joita tässä ei ilmaista suoraan, pidätetään soveltuvissa
määrin.
188 - FIN PN 15062781-C4 J | Käyttöopa

Companion 5Στοιχεία ελέγχου χρήστη και ενδείξεις κατάστασης συστήματος
Εσωτερικά σύμβολα02Έξοδος οξυγόνου

I/OΔιακόπτης Ενεργό/Ανενεργό
Δεν υπάρχουν μέρη που επιδέχονται
σέρβις στο εσωτερικό. Μην ανοίγετε τοκάλυμμα.

ISO 7000· Γραφικά σύμβολα για χρήση στον εξοπλισμό —Ευρετήριο και σύνοψη

Οδηγία του Συμβουλίου 93/42/ΕΟΚ για τα ιατροτεχνολογικά
προϊόνταΕξουσιοδοτημένος αντιπρόσωπος στηνΕυρωπαϊκή Κοινότητα
IEC 60417
Εξοπλισμός κλάσης II, διπλά
μονωμένος Reg. # 5172
Πιστοποίηση ETL

Περιοχή θερμοκρασίας αποθήκευσης ήλειτουργίας. Reg. # 0632

Οργανισμός για την Ασφάλεια γιαCAN/CSA C22.2 Αρ. 601.1 M90 γιαιατρικό ηλεκτρικό εξοπλισμό

Εύρος υγρασίας αποθήκευσης. Reg. #2620Όνομα και διεύθυνση τουκατασκευαστή. Reg. # 3082
Ημερομηνία κατασκευής. Reg. # 2497
Αριθμός καταλόγου. Reg. # 2493
Σειριακός αριθμός. Reg. # 2498
Με αυτή την όψη προς τα πάνω.
Reg. # 0623

Ανακοίνωση FCC
21 CFR 801.15: Κώδικας Ομοσπονδιακών Κανονισμών (Codeof Federal Regulations) των ΗΠΑ, Τίτλος 21Η ομοσπονδιακή νομοθεσία τωνΗΠΑ επιτρέπει την πώληση αυτής τηςσυσκευής μόνο από ιατρό ή κατόπιν
εντολής ιατρού.
Οδηγία του Συμβουλίου 2012/19/ΕΕ: οδηγία για τα απόβληταηλεκτρικού και ηλεκτρονικού εξοπλισμού (ΑΗΗΕ)Εύθραυστο, χειριστείτε με προσοχή.
Reg. # 0621
Φυλάσσετε μακριά από τη βροχή,διατηρείτε στεγνό. Reg. # 0626
Διαβάστε το εγχειρίδιο χρήστη πριν απότη λειτουργία. Reg. # 1641
Όριο οριοθέτησης κατά αριθμό. Reg.
# 2403.
ISO 7010: Γραφικά σύμβολα — Χρώματα ασφάλειας καισήματα ασφάλειας — Καταχωρημένα σήματα ασφάλειαςΑπαιτείται να διαβαστεί το εγχειρίδιο
οδηγιών. Reg. # M002
Προειδοποίηση. Reg. # W001
Μακριά από γυμνές φλόγες, φωτιά,σπίθες. Απαγορεύονται οι ανοιχτές πηγές
ανάφλεξης και το κάπνισμα.
Reg. # P003
Μην καπνίζετε κοντά στη μονάδα ήκατά τη διάρκεια της λειτουργίας τηςμονάδας. Reg. # P002
Εφαρμοζόμενο μέρος τύπου BF (βαθμός
προστασίας από ηλεκτροπληξία). Reg.
# 5333

ΑΗΗΕ
IEC 60601-1: Ιατρικός ηλεκτρικός εξοπλισμός Μέρος 1 Γενικές απαιτήσεις για βασική ασφάλεια και ουσιαστική
απόδοσηIPX1Εξοπλισμός με προστασία από κάθετη
πτώση σταγόνων - IPX1: Το Companion
5 παρέχει προστασία από τις επιβλαβείς
επιδράσεις της διείσδυσης υγρών.
(IPX1, σύμφωνα με IEC 60529)

ΑΝΑΚΟΙΝΩΣΗ FCC:Αυτή η συσκευή μπορεί να περιέχει
CYBLE-022001-00, συμπεριλαμβανομένης τηςκεραίας 2450AT18B100 από την Johnson Technology,και συμμορφώνεται με την παρ. 15 των κανόνων
FCC. Η συσκευή πληροί τις απαιτήσεις για έγκριση
αρθρωτού πομπού όπως περιγράφονται στη δημόσια
Ανακοίνωση FCC DA00-1407. Η λειτουργία τουπομπού υπόκειται στις ακόλουθες δύο προϋποθέσεις:(1) Αυτή η συσκευή δεν πρέπει να προκαλεί
επιβλαβείς παρεμβολές και (2) Αυτή η συσκευή πρέπει
να δέχεται οποιαδήποτε λαμβανόμενη παρεμβολή,συμπεριλαμβανομένης παρεμβολής που μπορεί ναπροκαλέσει ανεπιθύμητη λειτουργία.
Αυτό το προϊόν μπορεί να καλύπτεται από ένα ήπερισσότερα διπλώματα ευρεσιτεχνίας, των ΗΠΑ καιδιεθνή. Παρακαλούμε επισκεφθείτε τον δικτυακό τόποτης εταιρείας μας παρακάτω για τον κατάλογο τωνεφαρμοστέων διπλωμάτων ευρεσιτεχνίας. Διπλώματα
ευρεσιτεχνίας: www.caireinc.com/corporate/patents/.
PN 15062781-C4 J | Εγχειρίδιο χρήστη

189 - GRE

Companion 5Σύντομος οδηγός εκκίνησης
1 Αποσυσκευασία του Companion 5

Companion 5

2 Γνωριμία με το Companion 5Ανασκοπήστε όλες τις προειδοποιήσεις, τις συστάσεις
προσοχής και τις πρόσθετες πληροφορίες συσκευήςστο υπόλοιπο του παρόντος εγχειριδίου. Εξοικειωθείτε
με τα βασικά χαρακτηριστικά του Companion 5 καιτου πίνακα ελέγχου χρήστη.

ΠΡΟΣΟΨΗΕξωτερικό
περίβλημαΔιακόπτης
λειτουργίας

Οθόνη
LED

Θύρα
εξόδου
οξυγόνουΩρομετρητής

ΡοόμετροΒάσηστήριξης
φιάλης
υγραντήρα
(φιάλη
προαιρετική)ΠΙΣΩ ΟΨΗΛαβήμεταφοράς
και θέση
εισαγωγής
αέρα

Θέση
εισαγωγής
αέρα ψύξης

Θέση εισαγωγής
αέρα

190 - GREPN 15062781-C4 J | Εγχειρίδιο χρήστη

ΆνοιγμαεξαερισμούCompanion 5Τα εξαρτήματα του Companion 5Εξωτερικό
περίβλημαΑνθεκτικό πλαστικό περίβλημα που περιβάλλει
και προστατεύει τα εσωτερικά εξαρτήματα του
CAIRE Companion 5.
ΔιακόπτηςΔιακόπτης λειτουργίας ΕΝΕΡΓΟ/ΑΝΕΝΕΡΓΟλειτουργίαςπου χρησιμοποιείται για την εκκίνηση ή τηδιακοπή της παροχής τροφοδοσίας ρεύματοςστη μονάδα. Η θέση ΕΝΕΡΓΟ υποδεικνύεταιαπό το σύμβολο «l» και η θέση ΑΝΕΝΕΡΓΟυποδεικνύεται από το σύμβολο «O».
Θύρα εξόδουΤο οξυγόνο εξέρχεται από το Companion 5οξυγόνουστο σημείο αυτό αφού έχει φιλτραριστεί καισυμπυκνωθεί. Για τη χορήγηση στον ασθενή,συνδέεται είτε ρινική κάνουλα μονού αυλού είτετυπική σωλήνωση οξυγόνου. Το μέγιστο μήκοςτης σωλήνωσης οξυγόνου που συνδέεται στοσυμπυκνωτή είναι 15,2 m (50 ft).
Οθόνη LEDΠεριέχει πράσινη, κόκκινη και κίτρινη λυχνία*LED. Η πράσινη λυχνία υποδεικνύει κανονική
λειτουργία, ενώ η κίτρινη* και η κόκκινη λυχνία
υποδεικνύουν καταστάσεις συναγερμού. (Βλ.
την ενότητα «Καταστάσεις συναγερμού» στιςοδηγίες χρήσης για περισσότερες πληροφορίες.)Ωρομετρητής
Ψηφιακή ένδειξη του χρόνου λειτουργίας τουσυμπυκνωτή που έχει παρέλθει. Εμφανίζειτο κοντινότερο δέκατο της ώρας και δενείναι δυνατή η επαναφορά της. Εμφανίζει
κωδικούς διαγνωστικών συναγερμών ότανο συμπυκνωτής παρουσιάζει κατάσταση
συναγερμού.
ΡοόμετροΧρησιμοποιείται για την προσαρμογή και τηνεμφάνιση του ρυθμού ροής του χορηγούμενουοξυγόνου. Οι ρυθμοί ροής κυμαίνονται από
0,5–5 λίτρα ανά λεπτό (LPM) και μπορούν ναρυθμιστούν περιστρέφοντας το περιστροφικό
κουμπί.
Θέση για τη σύνδεση φιάλης υγραντήρα.
Βάση στήριξης
φιάλης υγραντήρα Η βάση περιλαμβάνει μια ελαστική ταινία που(φιάλη προαιρετική) χρησιμοποιείται για τη στερέωση της φιάληςστη βάση. Μια φιάλη υγραντήρα παρέχει
πρόσθετη υγρασία στη ροή του οξυγόνου.
Η φιάλη γεμίζει με απεσταγμένο νερό και ηυγρασία χρησιμοποιείται για την πρόληψη τηςξήρανσης των ρινικών ιστών.
*Μοντέλα με δείκτη κατάστασης
συμπυκνωτή οξυγόνου (OCSI) μόνο3 Σύνδεση φιάλης υγραντήρα και κάνουλαςΕάν χρησιμοποιείτε φιάλη υγραντήρα, γεμίστε πρώτατη φιάλη με απεσταγμένο νερό μέχρι τη στάθμη πουκαθορίζεται από τον κατασκευαστή. Τοποθετήστετη φιάλη στην προβλεπόμενη βάση στήριξης και
ασφαλίστε την με την ελαστική ταινία. Συνδέστε τοκοχλιωτό άκρο του σωλήνα υγραντήρα στη φιάλη καιτο άλλο στη θύρα εξόδου του Companion 5. Συνδέστε
τη σωλήνωση οξυγόνου ή/και μια ρινική κάνουλα στηθύρα εξόδου που βρίσκεται στη φιάλη υγραντήρα.

Εάν δεν χρησιμοποιείτε φιάλη υγραντήρα, συνδέστε
τη σωλήνωση οξυγόνου ή/και μια ρινική κάνουλααπευθείας στη θύρα εξόδου του Companion 5.

PN 15062781-C4 J | Εγχειρίδιο χρήστη

191 - GRE

Companion 5
4 Ενεργοποίηση και προθέρμανσηΣυνδέστε το καλώδιο τροφοδοσίας εναλλασσόμενουρεύματος (AC) σε μια πρίζα AC.

6 Προσαρμογή της ρύθμισης ροής στοσυνταγογραφημένο επίπεδο
Χρησιμοποιήστε το περιστροφικό κουμπί ελέγχου ροήςγια να επιλέξετε τη ροή που έχει συνταγογραφηθεί απότο γιατρό σας.

Θέστε το διακόπτη λειτουργίας στη θέση Ενεργό.

Διακόπτης
λειτουργίας

Όταν το Companion 5 ενεργοποιηθεί σωστά, θαανάψει η πράσινη ένδειξη τροφοδοσίας ρεύματος στηνοθόνη LED. Όλες οι λυχνίες LED θα ανάψουν κατάτην εκκίνηση. Αφού ο συμπυκνωτής ολοκληρώσει
τον κύκλο προθέρμανσης, μόνο η πράσινη λυχνία θαπαραμείνει αναμμένη.
Σημείωση: Μετά την αρχική ενεργοποίηση, η λυχνία O2 θα ανάψει
και η ένδειξη «O2 UP» θα εμφανιστεί στην οθόνη μέχρι να φτάσει ησυγκέντρωση στις προδιαγραφές.

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΊΝΑΙ ΠΟΛΎ ΣΗΜΑΝΤΙΚΌ ΝΑΕΠΙΛΈΓΕΤΕ ΜΌΝΟ ΤΟ ΣΥΝΤΑΓΟΓΡΑΦΗΜΈΝΟ
ΕΠΊΠΕΔΟ ΟΞΥΓΌΝΟΥ. ΜΗΝ ΑΛΛΆΖΕΤΕ
ΤΗΝ ΕΠΙΛΟΓΉ ΡΟΉΣ ΕΚΤΌΣ ΕΆΝ ΣΑΣ ΤΟΥΠΈΔΕΙΞΕ ΑΔΕΙΟΎΧΟΣ ΚΛΙΝΙΚΌΣ ΓΙΑΤΡΌΣ.
Ο ΣΥΜΠΥΚΝΩΤΉΣ ΟΞΥΓΌΝΟΥ ΜΠΟΡΕΊ ΝΑΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ ΚΑΤΆ ΤΗ ΔΙΆΡΚΕΙΑ ΤΟΥ ΎΠΝΟΥΥΠΌ ΤΗΝ ΚΑΘΟΔΉΓΗΣΗ ΕΞΕΙΔΙΚΕΥΜΈΝΟΥ
ΚΛΙΝΙΚΟΎ ΓΙΑΤΡΟΎ.
Αναπνέετε κανονικά διαμέσου της ρινικής κάνουλας.
Για να ρυθμίσετε το ρυθμό ροής:Περιστρέψτε αριστερόστροφα για να αυξήσετε τη ροή.
Περιστρέψτε δεξιόστροφα για να μειώσετε τη ροή.
Το μέσον της σφαίρας υποδεικνύει το ρυθμό ροής.
Βλ. την εικόνα παρακάτω.

LED ένδειξης
τροφοδοσίας
ρεύματος

Σωστό

192 - GREPN 15062781-C4 J | Εγχειρίδιο χρήστηCompanion 5Σημαντικό!

Οι οδηγίες ασφαλείας καθορίζονται ως εξής:ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΣΗΜΑΝΤΙΚΕΣ ΠΛΗΡΟΦΟΡΙΕΣ
ΑΣΦΑΛΕΙΑΣ ΓΙΑ ΚΙΝΔΥΝΟΥΣ ΠΟΥ ΜΠΟΡΕΙ ΝΑΠΡΟΚΑΛΕΣΟΥΝ ΣΟΒΑΡΟ ΤΡΑΥΜΑΤΙΣΜΟ.
ΠΡΟΣΟΧΗ: Σημαντικές πληροφορίες για τηνπρόληψη ζημιάς στο Companion 5.
Σημείωση: Πληροφορίες που χρήζουν ιδιαίτερης προσοχής.

Ενδείξεις χρήσης
Προβλεπόμενη χρήση

Το Companion 5 προορίζεται για τη χορήγηση
συμπληρωματικού οξυγόνου. Η συσκευή δενπροορίζεται για ζωτική υποστήριξη, ούτε παρέχει
οποιεσδήποτε δυνατότητες παρακολούθησης
ασθενούς.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΊΝΑΙ ΠΟΛΎ ΣΗΜΑΝΤΙΚΌ ΝΑΕΠΙΛΈΓΕΤΕ ΜΌΝΟ ΤΟ ΣΥΝΤΑΓΟΓΡΑΦΗΜΈΝΟ
ΕΠΊΠΕΔΟ ΟΞΥΓΌΝΟΥ. ΜΗΝ ΑΛΛΆΖΕΤΕ ΤΗΝΕΠΙΛΟΓΉ ΡΟΉΣ ΕΚΤΌΣ ΕΆΝ ΣΑΣ ΤΟ ΥΠΈΔΕΙΞΕΑΔΕΙΟΎΧΟΣ ΚΛΙΝΙΚΌΣ ΓΙΑΤΡΌΣ. Ο ΦΟΡΗΤΌΣ
ΣΥΜΠΥΚΝΩΤΉΣ ΟΞΥΓΌΝΟΥ ΜΠΟΡΕΊ ΝΑΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ ΚΑΤΆ ΤΗ ΔΙΆΡΚΕΙΑ ΤΟΥ ΎΠΝΟΥΥΠΌ ΤΗΝ ΚΑΘΟΔΉΓΗΣΗ ΕΞΕΙΔΙΚΕΥΜΈΝΟΥ
ΚΛΙΝΙΚΟΎ ΓΙΑΤΡΟΎ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΓΙΑ ΝΑ ΔΙΑΣΦΑΛΙΣΤΕΊ ΌΤΙ ΛΑΜΒΆΝΕΤΕΤΗ ΘΕΡΑΠΕΥΤΙΚΉ ΠΟΣΌΤΗΤΑ ΧΟΡΗΓΟΎΜΕΝΟΥ ΟΞΥΓΌΝΟΥ
ΣΎΜΦΩΝΑ ΜΕ ΤΗΝ ΙΑΤΡΙΚΉ ΠΆΘΗΣΉ ΣΑΣ, ΤΟ COMPANION 5ΠΡΈΠΕΙ ΝΑ ΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ ΜΌΝΟ ΑΦΟΎ ΜΊΑ ΉΠΕΡΙΣΣΌΤΕΡΕΣ ΡΥΘΜΊΣΕΙΣ ΈΧΟΥΝ ΠΡΟΣΔΙΟΡΙΣΤΕΊ
ΕΞΑΤΟΜΙΚΕΥΜΈΝΑ Ή ΣΥΝΤΑΓΟΓΡΑΦΗΘΕΊ ΓΙΑ ΕΣΆΣ ΣΤΑΣΥΓΚΕΚΡΙΜΈΝΑ ΕΠΊΠΕΔΑ ΔΡΑΣΤΗΡΙΌΤΗΤΆΣ ΣΑΣ. ΤΟ
COMPANION 5 ΠΡΈΠΕΙ ΝΑ ΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ ΜΕ ΤΟΝΕΙΔΙΚΌ ΣΥΝΔΥΑΣΜΌ Ή ΜΈΡΗ ΚΑΙ ΠΑΡΕΛΚΌΜΕΝΑ ΠΟΥ ΕΊΝΑΙ
ΣΎΜΦΩΝΑ ΜΕ ΤΙΣ ΠΡΟΔΙΑΓΡΑΦΈΣ ΤΟΥ ΚΑΤΑΣΚΕΥΑΣΤΉ ΤΟΥΣΥΜΠΥΚΝΩΤΉ ΚΑΙ ΠΟΥ ΧΡΗΣΙΜΟΠΟΙΉΘΗΚΑΝ ΚΑΤΆ ΤΟΝΠΡΟΣΔΙΟΡΙΣΜΌ ΤΩΝ ΡΥΘΜΊΣΕΏΝ ΣΑΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΥΤΉ Η ΜΟΝΆΔΑ ΔΕΝ ΠΡΟΟΡΊΖΕΤΑΙ ΝΑΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ ΓΙΑ ΖΩΤΙΚΉ ΥΠΟΣΤΉΡΙΞΗ. ΗΛΙΚΙΩΜΈΝΟΙ,ΠΑΙΔΙΑΤΡΙΚΟΊ Ή ΟΠΟΙΟΙΔΉΠΟΤΕ ΆΛΛΟΙ ΑΣΘΕΝΕΊΣ ΠΟΥ ΔΕΝΕΊΝΑΙ ΣΕ ΘΈΣΗ ΝΑ ΓΝΩΣΤΟΠΟΙΉΣΟΥΝ ΤΥΧΌΝ ΔΥΣΦΟΡΊΑ
ΜΠΟΡΕΊ ΝΑ ΑΠΑΙΤΟΎΝ ΠΡΌΣΘΕΤΗ ΠΑΡΑΚΟΛΟΎΘΗΣΗ
Ή/ΚΑΙ ΈΝΑ ΚΑΤΑΝΕΜΗΜΈΝΟ ΣΎΣΤΗΜΑ ΣΥΝΑΓΕΡΜΟΎ
ΓΙΑ ΤΗ ΜΕΤΆΔΟΣΗ ΤΩΝ ΠΛΗΡΟΦΟΡΙΏΝ ΣΧΕΤΙΚΆ ΜΕΤΥΧΌΝ ΔΥΣΦΟΡΊΑ Ή/ΚΑΙ ΕΠΕΊΓΟΥΣΑ ΙΑΤΡΙΚΉ ΚΑΤΆΣΤΑΣΗΣΤΟΝ ΥΠΕΎΘΥΝΟ ΦΡΟΝΤΙΣΤΉ ΜΕ ΣΚΟΠΌ ΤΗΝ ΑΠΟΦΥΓΉ
ΕΠΙΒΛΑΒΏΝ ΣΥΝΕΠΕΙΏΝ. ΑΣΘΕΝΕΊΣ ΜΕ ΜΕΙΩΜΈΝΗ ΑΚΟΉΉ/ΚΑΙ ΌΡΑΣΗ ΜΠΟΡΕΊ ΝΑ ΑΠΑΙΤΟΎΝ ΒΟΉΘΕΙΑ ΜΕ ΤΗΝΠΑΡΑΚΟΛΟΎΘΗΣΗ ΤΩΝ ΣΥΝΑΓΕΡΜΏΝ.

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΟΙ ΈΓΚΥΕΣ ΚΑΙ ΘΗΛΆΖΟΥΣΕΣ ΓΥΝΑΊΚΕΣ
ΔΕΝ ΠΡΈΠΕΙ ΝΑ ΧΡΗΣΙΜΟΠΟΙΟΎΝ ΤΑ ΠΑΡΕΛΚΌΜΕΝΑ ΠΟΥΣΥΝΙΣΤΏΝΤΑΙ ΣΕ ΑΥΤΌ ΤΟ ΕΓΧΕΙΡΊΔΙΟ, ΚΑΘΏΣ ΜΠΟΡΕΊ ΝΑΠΕΡΙΈΧΟΥΝ ΦΘΑΛΙΚΈΣ ΟΥΣΊΕΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΟΜΟΣΠΟΝΔΙΑΚΉ ΝΟΜΟΘΕΣΊΑ (ΗΠΑ)ΠΕΡΙΟΡΊΖΕΙ ΤΗΝ ΠΏΛΗΣΗ Ή ΜΊΣΘΩΣΗ ΑΥΤΉΣ ΤΗΣ ΣΥΣΚΕΥΉΣΑΠΌ Ή ΚΑΤ' ΕΝΤΟΛΉ ΓΙΑΤΡΟΎ Ή ΆΛΛΟΥ ΑΔΕΙΟΎΧΟΥ ΠΑΡΌΧΟΥ
ΥΓΕΙΟΝΟΜΙΚΉΣ ΠΕΡΊΘΑΛΨΗΣ.

Αντενδείξεις χρήσης
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΣΕ ΟΡΙΣΜΕΝΕΣ ΠΕΡΙΠΤΩΣΕΙΣ,Η ΧΡΗΣΗ ΜΗ ΣΥΝΤΑΓΟΓΡΑΦΗΜΕΝΟΥ ΟΞΥΓΟΝΟΥΜΠΟΡΕΙ ΝΑ ΕΙΝΑΙ ΕΠΙΚΙΝΔΥΝΗ. ΑΥΤΗ Η ΣΥΣΚΕΥΗΠΡΕΠΕΙ ΝΑ ΧΡΗΣΙΜΟΠΟΙΕΙΤΑΙ ΜΟΝΟ ΟΤΑΝ ΕΧΕΙΣΥΝΤΑΓΟΓΡΑΦΗΘΕΙ ΑΠΟ ΓΙΑΤΡΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΔΕΝ ΠΡΟΟΡΙΖΕΤΑΙ ΓΙΑ ΧΡΗΣΗ ΥΠΟ ΤΗΝΠΑΡΟΥΣΙΑ ΕΥΦΛΕΚΤΩΝ ΑΝΑΙΣΘΗΤΙΚΩΝ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΟΠΩΣ ΚΑΙ ΜΕ ΚΑΘΕ ΗΛΕΚΤΡΙΚΑ
ΤΡΟΦΟΔΟΤΟΥΜΕΝΗ ΣΥΣΚΕΥΗ, Ο ΧΡΗΣΤΗΣ ΜΠΟΡΕΙ ΝΑΑΝΤΙΜΕΤΩΠΙΣΕΙ ΠΕΡΙΟΔΟΥΣ ΕΚΤΟΣ ΛΕΙΤΟΥΡΓΙΑΣ ΩΣΑΠΟΤΕΛΕΣΜΑ ΔΙΑΚΟΠΗΣ ΗΛΕΚΤΡΙΚΟΥ ΡΕΥΜΑΤΟΣ, Ή ΤΗΝΑΝΑΓΚΗ ΝΑ ΑΝΑΘΕΣΕΙ ΤΟ ΣΕΡΒΙΣ ΤΟΥ COMPANION 5 ΣΕΕΞΕΙΔΙΚΕΥΜΕΝΟ ΤΕΧΝΙΚΟ. ΤΟ COMPANION 5 ΔΕΝ ΕΙΝΑΙΚΑΤΑΛΛΗΛΟ ΓΙΑ ΟΠΟΙΟΝΔΗΠΟΤΕ ΑΣΘΕΝΗ Ο ΟΠΟΙΟΣ ΘΑΜΠΟΡΟΥΣΕ ΝΑ ΠΑΡΟΥΣΙΑΣΕΙ ΑΝΕΠΙΘΥΜΗΤΕΣ ΕΠΙΠΤΩΣΕΙΣΣΤΗΝ ΥΓΕΙΑ ΩΣ ΑΠΟΤΕΛΕΣΜΑ ΤΕΤΟΙΑΣ ΠΡΟΣΩΡΙΝΗΣ
ΔΙΑΚΟΠΗΣ.

Οδηγίες ασφαλείας
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΔΕΝ ΕΠΙΤΡΈΠΕΤΑΙ ΚΑΜΊΑ
ΤΡΟΠΟΠΟΊΗΣΗ ΑΥΤΟΎ ΤΟΥ ΕΞΟΠΛΙΣΜΟΎ
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΚΑΤΑΣΚΕΥΑΣΤΉΣ ΣΥΝΙΣΤΆ ΜΙΑΕΝΑΛΛΑΚΤΙΚΉ ΠΗΓΉ ΣΥΜΠΛΗΡΩΜΑΤΙΚΟΎ ΟΞΥΓΌΝΟΥΣΕ ΠΕΡΊΠΤΩΣΗ ΔΙΑΚΟΠΉΣ ΡΕΎΜΑΤΟΣ, ΚΑΤΆΣΤΑΣΗΣΣΥΝΑΓΕΡΜΟΎ Ή ΜΗΧΑΝΙΚΉΣ ΒΛΆΒΗΣ. ΣΥΜΒΟΥΛΕΥΤΕΊΤΕ ΤΟΝΓΙΑΤΡΌ Ή ΤΟΝ ΠΆΡΟΧΟ ΤΟΥ ΕΞΟΠΛΙΣΜΟΎ ΓΙΑ ΤΟΝ ΤΎΠΟ ΤΟΥΕΦΕΔΡΙΚΟΎ ΣΥΣΤΉΜΑΤΟΣ ΠΟΥ ΑΠΑΙΤΕΊΤΑΙ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΥΤΉ Η ΣΥΣΚΕΥΉ ΠΑΡΈΧΕΙ ΟΞΥΓΌΝΟ
ΥΨΗΛΉΣ ΣΥΓΚΈΝΤΡΩΣΗΣ, ΤΟ ΟΠΟΊΟ ΠΡΟΆΓΕΙ ΤΗΝ ΤΑΧΕΊΑ
ΚΑΎΣΗ. ΜΗΝ ΕΠΙΤΡΈΠΕΤΕ ΤΟ ΚΆΠΝΙΣΜΑ Ή ΤΙΣ ΑΝΟΙΧΤΈΣ
ΦΛΌΓΕΣ ΕΝΤΌΣ ΔΎΟ (2) ΜΈΤΡΩΝ ΑΠΌ (1) ΑΥΤΉΝ ΤΗ ΣΥΣΚΕΥΉ,Ή (2) ΟΠΟΙΟΔΉΠΟΤΕ ΠΑΡΕΛΚΌΜΕΝΟ ΠΟΥ ΦΈΡΕΙ ΟΞΥΓΌΝΟ.
ΑΠΟΤΥΧΊΑ ΤΉΡΗΣΗΣ ΑΥΤΉΣ ΤΗΣ ΠΡΟΕΙΔΟΠΟΊΗΣΗΣ ΜΠΟΡΕΊ
ΝΑ ΟΔΗΓΉΣΕΙ ΣΕ ΠΥΡΚΑΓΙΆ, ΚΑΤΑΣΤΡΟΦΉ ΙΔΙΟΚΤΗΣΊΑΣ
Ή/ΚΑΙ ΝΑ ΠΡΟΚΑΛΈΣΕΙ ΦΥΣΙΚΌ ΤΡΑΥΜΑΤΙΣΜΌ Ή ΘΆΝΑΤΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΛΕΙΤΟΥΡΓΕΊΤΕ ΤΗ ΜΟΝΆΔΑ ΣΕΠΕΡΙΟΡΙΣΜΈΝΟ Ή ΣΤΕΝΌ ΧΏΡΟ (ΔΗΛ. ΜΙΚΡΉ ΤΣΆΝΤΑ ΉΤΣΆΝΤΑ ΧΕΙΡΌΣ) ΌΠΟΥ Ο ΑΕΡΙΣΜΌΣ ΜΠΟΡΕΊ ΝΑ ΕΊΝΑΙ
ΠΕΡΙΟΡΙΣΜΈΝΟΣ. ΚΆΤΙ ΤΈΤΟΙΟ ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΕΙ
ΥΠΕΡΘΈΡΜΑΝΣΗ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΚΑΙ ΝΑΕΠΗΡΕΆΣΕΙ ΑΡΝΗΤΙΚΆ ΤΗΝ ΑΠΌΔΟΣΗ.

PN 15062781-C4 J | Εγχειρίδιο χρήστη

193 - GRE

Companion 5ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΣΥΜΠΥΚΝΩΤΉΣ ΠΡΈΠΕΙΝΑ ΤΟΠΟΘΕΤΕΊΤΑΙ ΜΕ ΤΈΤΟΙΟΝ ΤΡΌΠΟ ΏΣΤΕ ΝΑΑΠΟΦΕΎΓΕΤΑΙ Ο ΚΑΠΝΌΣ, ΟΙ ΡΥΠΑΝΤΈΣ Ή ΟΙΑΝΑΘΥΜΙΆΣΕΙΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΧΡΉΣΗ ΟΡΙΣΜΈΝΩΝ ΠΑΡΕΛΚΟΜΈΝΩΝ
ΧΟΡΉΓΗΣΗΣ ΟΞΥΓΌΝΟΥ ΠΟΥ ΔΕΝ ΚΑΘΟΡΊΖΟΝΤΑΙ ΓΙΑ ΑΥΤΌΝ
ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΜΠΟΡΕΊ ΝΑ ΕΠΗΡΕΆΣΕΙ
ΑΡΝΗΤΙΚΆ ΤΗΝ ΑΠΌΔΟΣΗ. ΠΑΡΑΠΟΜΠΈΣ ΣΤΑ ΣΥΝΙΣΤΏΜΕΝΑ
ΠΑΡΕΛΚΌΜΕΝΑ ΠΑΡΈΧΟΝΤΑΙ ΣΕ ΑΥΤΌ ΤΟ ΕΓΧΕΙΡΊΔΙΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΆΝ Ο ΣΥΜΠΥΚΝΩΤΉΣ ΟΞΥΓΌΝΟΥ ΈΧΕΙΠΈΣΕΙ, ΥΠΟΣΤΕΊ ΖΗΜΙΆ Ή ΕΚΤΕΘΕΊ ΣΕ ΝΕΡΌ, ΠΑΡΑΚΑΛΟΎΜΕ
ΕΠΙΚΟΙΝΩΝΉΣΤΕ ΜΕ ΤΟΝ ΠΆΡΟΧΟ ΚΑΤ' ΟΊΚΟΝ ΦΡΟΝΤΊΔΑΣ
ΣΑΣ ΓΙΑ ΈΛΕΓΧΟ Ή ΠΙΘΑΝΉ ΕΠΙΣΚΕΥΉ ΤΗΣ ΣΥΣΚΕΥΉΣ. ΜΗΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΕΆΝ ΈΧΕΙΥΠΟΣΤΕΊ ΖΗΜΙΆ ΤΟ ΚΑΛΏΔΙΟ Ή ΤΟ ΦΙΣ ΡΕΎΜΑΤΟΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΑΦΉΝΕΤΕ ΝΑ ΦΡΆΞΟΥΝ ΤΑΑΝΟΊΓΜΑΤΑ ΕΙΣΑΓΩΓΉΣ Ή ΕΞΌΔΟΥ ΑΈΡΑ. ΜΗΝ ΑΦΉΝΕΤΕΝΑ ΠΈΣΟΥΝ ΚΑΙ ΜΗΝ ΕΙΣΆΓΕΤΕ ΑΝΤΙΚΕΊΜΕΝΑ ΣΕ ΚΑΝΈΝΑ
ΆΝΟΙΓΜΑ ΤΗΣ ΣΥΣΚΕΥΉΣ. ΚΆΤΙ ΤΈΤΟΙΟ ΜΠΟΡΕΊ ΝΑΠΡΟΚΑΛΈΣΕΙ ΥΠΕΡΘΈΡΜΑΝΣΗ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ
ΚΑΙ ΝΑ ΕΠΗΡΕΆΣΕΙ ΑΡΝΗΤΙΚΆ ΤΗΝ ΑΠΌΔΟΣΗ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΚΆΝΕΤΕ ΥΠΕΡΠΛΉΡΩΣΗ ΤΟΥΠΡΟΑΙΡΕΤΙΚΟΎ ΥΓΡΑΝΤΉΡΑ. ΓΕΜΊΖΕΤΕ ΤΟΝ ΠΡΟΑΙΡΕΤΙΚΌ
ΥΓΡΑΝΤΉΡΑ ΜΕ ΝΕΡΌ ΜΌΝΟ ΜΈΧΡΙ ΤΗ ΣΤΆΘΜΗ ΠΟΥΥΠΟΔΕΙΚΝΎΕΤΑΙ ΑΠΌ ΤΟΝ ΚΑΤΑΣΚΕΥΑΣΤΉ ΤΟΥ ΥΓΡΑΝΤΉΡΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΛΕΙΤΟΥΡΓΕΙΤΕ ΑΥΤΟΝ ΤΟΝ ΕΞΟΠΛΙΣΜΟΧΩΡΙΣ ΝΑ ΕΧΕΤΕ ΠΡΩΤΑ ΔΙΑΒΑΣΕΙ ΚΑΙ ΚΑΤΑΝΟΗΣΕΙ ΤΟ ΠΑΡΟΝΕΓΧΕΙΡΙΔΙΟ. ΕΑΝ ΔΕΝ ΜΠΟΡΕΙΤΕ ΝΑ ΚΑΤΑΝΟΗΣΕΤΕ ΤΙΣΠΡΟΕΙΔΟΠΟΙΗΣΕΙΣ ΚΑΙ ΤΙΣ ΟΔΗΓΙΕΣ, ΕΠΙΚΟΙΝΩΝΗΣΤΕ ΜΕ ΤΟΝΠΑΡΟΧΟ ΤΟΥ ΕΞΟΠΛΙΣΜΟΥ ΣΑΣ ΠΡΟΤΟΥ ΕΠΙΧΕΙΡΗΣΕΤΕ ΝΑΧΡΗΣΙΜΟΠΟΙΗΣΕΤΕ ΑΥΤΟΝ ΤΟΝ ΕΞΟΠΛΙΣΜΟ· ΣΕ ΑΝΤΙΘΕΤΗΠΕΡΙΠΤΩΣΗ, ΜΠΟΡΕΙ ΝΑ ΠΡΟΚΛΗΘΕΙ ΤΡΑΥΜΑΤΙΣΜΟΣ ΉΥΛΙΚΗ ΖΗΜΙΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΆΝ ΑΙΣΘΆΝΕΣΤΕ ΔΥΣΦΟΡΊΑ ΉΠΑΡΟΥΣΙΆΖΕΤΕ ΕΠΕΊΓΟΥΣΑ ΙΑΤΡΙΚΉ ΚΑΤΆΣΤΑΣΗ,ΑΝΑΖΗΤΉΣΤΕ ΙΑΤΡΙΚΉ ΒΟΉΘΕΙΑ ΑΜΈΣΩΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΛΕΙΤΟΥΡΓΊΑ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ
ΟΞΥΓΌΝΟΥ ΕΚΤΌΣ ΤΩΝ ΠΡΟΔΙΑΓΡΑΦΏΝ ΘΕΡΜΟΚΡΑΣΊΑΣ
ΛΕΙΤΟΥΡΓΊΑΣ ΜΠΟΡΕΊ ΝΑ ΠΕΡΙΟΡΊΣΕΙ ΤΗ ΔΥΝΑΤΌΤΗΤΑ ΤΟΥΣΥΜΠΥΚΝΩΤΉ ΝΑ ΙΚΑΝΟΠΟΙΕΊ ΤΗΝ ΠΡΟΔΙΑΓΡΑΦΉ ΤΟΥΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ. ΑΝΑΤΡΈΞΤΕ ΣΤΗΝ ΕΝΌΤΗΤΑ
ΠΡΟΔΙΑΓΡΑΦΏΝ ΤΟΥ ΠΑΡΌΝΤΟΣ ΕΓΧΕΙΡΙΔΊΟΥ ΓΙΑ ΤΑ ΌΡΙΑΘΕΡΜΟΚΡΑΣΊΑΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΛΆΔΙ, ΓΡΆΣΟΉ ΠΡΟΪΌΝΤΑ ΜΕ ΒΆΣΗ ΠΕΤΡΈΛΑΙΟ Ή ΆΛΛΑ ΕΎΦΛΕΚΤΑ
ΠΡΟΪΌΝΤΑ ΜΕ ΤΑ ΠΑΡΕΛΚΌΜΕΝΑ ΠΟΥ ΦΈΡΟΥΝ ΟΞΥΓΌΝΟΉ ΜΕ ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ. ΠΡΈΠΕΙ ΝΑΧΡΗΣΙΜΟΠΟΙΟΎΝΤΑΙ ΜΌΝΟ ΛΟΣΙΌΝ Ή ΑΛΟΙΦΈΣ ΜΕ ΒΆΣΗ ΤΟΝΕΡΌ Ή ΣΥΜΒΑΤΈΣ ΜΕ ΤΟ ΟΞΥΓΌΝΟ. ΤΟ ΟΞΥΓΌΝΟ ΕΠΙΤΑΧΎΝΕΙ
ΤΗΝ ΚΑΎΣΗ ΤΩΝ ΕΎΦΛΕΚΤΩΝ ΟΥΣΙΏΝ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΣΥΜΠΥΚΝΩΤΉΣ ΟΞΥΓΌΝΟΥ ΔΕΝ ΠΡΈΠΕΙΝΑ ΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ ΠΑΡΑΚΕΊΜΕΝΑ ΣΕ Ή ΣΕ ΔΙΆΤΑΞΗ
ΣΤΟΊΒΑΣ ΜΕ ΆΛΛΟ ΕΞΟΠΛΙΣΜΌ. ΕΆΝ Η ΧΡΉΣΗ ΠΑΡΑΚΕΊΜΕΝΑΉ ΣΕ ΔΙΆΤΑΞΗ ΣΤΟΊΒΑΣ ΕΊΝΑΙ ΑΝΑΠΌΦΕΥΚΤΗ, Η ΣΥΣΚΕΥΉ
ΠΡΈΠΕΙ ΝΑ ΠΑΡΑΚΟΛΟΥΘΕΊΤΑΙ ΓΙΑ ΝΑ ΕΠΑΛΗΘΕΎΕΤΑΙ ΗΚΑΝΟΝΙΚΉ ΛΕΙΤΟΥΡΓΊΑ.

194 - GREPN 15062781-C4 J | Εγχειρίδιο χρήστη

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΚΑΤΑΣΚΕΥΑΣΤΉΣ ΣΥΝΙΣΤΆ
ΜΙΑ ΕΝΑΛΛΑΚΤΙΚΉ ΠΗΓΉ ΣΥΜΠΛΗΡΩΜΑΤΙΚΟΎ
ΟΞΥΓΌΝΟΥ ΣΕ ΠΕΡΊΠΤΩΣΗ ΔΙΑΚΟΠΉΣ
ΡΕΎΜΑΤΟΣ, ΚΑΤΆΣΤΑΣΗΣ ΣΥΝΑΓΕΡΜΟΎ ΉΜΗΧΑΝΙΚΉΣ ΒΛΆΒΗΣ. ΣΥΜΒΟΥΛΕΥΤΕΊΤΕ ΤΟΝΓΙΑΤΡΌ Ή ΤΟΝ ΠΆΡΟΧΟ ΤΟΥ ΕΞΟΠΛΙΣΜΟΎ ΓΙΑΤΟΝ ΤΎΠΟ ΤΟΥ ΕΦΕΔΡΙΚΟΎ ΣΥΣΤΉΜΑΤΟΣ ΠΟΥΑΠΑΙΤΕΊΤΑΙ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΥΤΉ Η ΣΥΣΚΕΥΉ ΠΑΡΈΧΕΙ ΟΞΥΓΌΝΟ
ΥΨΗΛΉΣ ΣΥΓΚΈΝΤΡΩΣΗΣ, ΤΟ ΟΠΟΊΟ ΠΡΟΆΓΕΙ ΤΗΝ ΤΑΧΕΊΑ
ΚΑΎΣΗ. ΜΗΝ ΕΠΙΤΡΈΠΕΤΕ ΤΟ ΚΆΠΝΙΣΜΑ Ή ΤΙΣ ΑΝΟΙΧΤΈΣ
ΦΛΌΓΕΣ ΣΤΟΝ ΊΔΙΟ ΧΏΡΟ ΜΕ (1) ΑΥΤΉΝ ΤΗ ΣΥΣΚΕΥΉ, Ή(2) ΟΠΟΙΟΔΉΠΟΤΕ ΠΑΡΕΛΚΌΜΕΝΟ ΠΟΥ ΦΈΡΕΙ ΟΞΥΓΌΝΟ.
ΑΠΟΤΥΧΊΑ ΤΉΡΗΣΗΣ ΑΥΤΉΣ ΤΗΣ ΠΡΟΕΙΔΟΠΟΊΗΣΗΣ ΜΠΟΡΕΊ
ΝΑ ΟΔΗΓΉΣΕΙ ΣΕ ΠΥΡΚΑΓΙΆ, ΚΑΤΑΣΤΡΟΦΉ ΙΔΙΟΚΤΗΣΊΑΣ
Ή/ΚΑΙ ΝΑ ΠΡΟΚΑΛΈΣΕΙ ΦΥΣΙΚΌ ΤΡΑΥΜΑΤΙΣΜΌ Ή ΘΆΝΑΤΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΧΡΉΣΗ ΟΡΙΣΜΈΝΩΝ ΠΑΡΕΛΚΟΜΈΝΩΝ
ΧΟΡΉΓΗΣΗΣ ΟΞΥΓΌΝΟΥ ΠΟΥ ΔΕΝ ΚΑΘΟΡΊΖΟΝΤΑΙ ΓΙΑ ΑΥΤΌΝ
ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΜΠΟΡΕΊ ΝΑ ΕΠΗΡΕΆΣΕΙ
ΑΡΝΗΤΙΚΆ ΤΗΝ ΑΠΌΔΟΣΗ. ΠΑΡΑΠΟΜΠΈΣ ΣΤΑ ΣΥΝΙΣΤΏΜΕΝΑ
ΠΑΡΕΛΚΌΜΕΝΑ ΠΑΡΈΧΟΝΤΑΙ ΣΕ ΑΥΤΌ ΤΟ ΕΓΧΕΙΡΊΔΙΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΆΝ Ο ΣΥΜΠΥΚΝΩΤΉΣ ΟΞΥΓΌΝΟΥ ΈΧΕΙΠΈΣΕΙ, ΥΠΟΣΤΕΊ ΖΗΜΙΆ Ή ΕΚΤΕΘΕΊ ΣΕ ΝΕΡΌ, ΠΑΡΑΚΑΛΟΎΜΕ
ΕΠΙΚΟΙΝΩΝΉΣΤΕ ΜΕ ΤΟΝ ΠΆΡΟΧΟ ΚΑΤ' ΟΊΚΟΝ ΦΡΟΝΤΊΔΑΣ
ΣΑΣ ΓΙΑ ΈΛΕΓΧΟ Ή ΠΙΘΑΝΉ ΕΠΙΣΚΕΥΉ ΤΗΣ ΣΥΣΚΕΥΉΣ. ΜΗΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΕΆΝ ΈΧΕΙΥΠΟΣΤΕΊ ΖΗΜΙΆ ΤΟ ΚΑΛΏΔΙΟ Ή ΤΟ ΦΙΣ ΡΕΎΜΑΤΟΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΜΌΝΟ ΤΑ ΑΝΤΑΛΛΑΚΤΙΚΆ
ΠΟΥ ΣΥΝΙΣΤΏΝΤΑΙ ΑΠΌ ΤΟΝ ΚΑΤΑΣΚΕΥΑΣΤΉ ΓΙΑ ΝΑΔΙΑΣΦΑΛΊΣΕΤΕ ΤΗ ΣΩΣΤΉ ΛΕΙΤΟΥΡΓΊΑ ΚΑΙ ΝΑ ΑΠΟΦΎΓΕΤΕ
ΤΟΝ ΚΊΝΔΥΝΟ ΠΥΡΚΑΓΙΆΣ ΚΑΙ ΕΓΚΑΥΜΆΤΩΝ.
ΠΡΟΕΙΔΟΠΟΊΗΣΗ: ΜΗ ΛΙΠΑΊΝΕΤΕ ΤΑ ΕΞΑΡΤΉΜΑΤΑ, ΤΙΣΣΥΝΔΈΣΕΙΣ, ΤΗ ΣΩΛΉΝΩΣΗ Ή ΆΛΛΑ ΠΑΡΕΛΚΌΜΕΝΑ ΤΟΥΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΓΙΑ ΝΑ ΑΠΟΦΎΓΕΤΕ ΤΟΝ ΚΊΝΔΥΝΟΠΥΡΚΑΓΙΆΣ ΚΑΙ ΕΓΚΑΥΜΆΤΩΝ.

Companion 5ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΟ ΚΆΠΝΙΣΜΑ ΕΝΏ ΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ
ΟΞΥΓΌΝΟ ΕΊΝΑΙ Η ΠΡΏΤΗ ΑΙΤΊΑ ΤΡΑΥΜΑΤΙΣΜΏΝ ΑΠΌΠΥΡΚΑΓΙΆ ΚΑΙ ΣΧΕΤΙΖΌΜΕΝΩΝ ΘΑΝΆΤΩΝ. ΠΡΈΠΕΙ ΝΑΤΗΡΕΊΤΕ ΤΙΣ ΑΚΌΛΟΥΘΕΣ ΠΡΟΕΙΔΟΠΟΙΉΣΕΙΣ ΑΣΦΑΛΕΊΑΣ:ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΕΠΙΤΡΈΠΕΤΕ ΤΟ ΚΆΠΝΙΣΜΑ, ΤΑ ΚΕΡΙΆΉ ΤΙΣ ΑΝΟΙΧΤΈΣ ΦΛΌΓΕΣ ΣΤΟΝ ΊΔΙΟ ΧΏΡΟ ΜΕ ΤΗ ΣΥΣΚΕΥΉ ΉΤΑ ΠΑΡΕΛΚΌΜΕΝΑ ΠΟΥ ΦΈΡΟΥΝ ΟΞΥΓΌΝΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΟ ΚΆΠΝΙΣΜΑ ΕΝΏ ΦΟΡΆΤΕ ΚΆΝΟΥΛΑ
ΟΞΥΓΌΝΟΥ ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΕΙ ΕΓΚΑΎΜΑΤΑ ΤΟΥΠΡΟΣΏΠΟΥ ΚΑΙ ΠΙΘΑΝΏΣ ΘΆΝΑΤΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΆΝ ΚΑΠΝΊΖΕΤΕ, ΤΑ ΑΚΌΛΟΥΘΑ 3 ΒΉΜΑΤΑ
ΜΠΟΡΕΊ ΝΑ ΣΏΣΟΥΝ ΤΗ ΖΩΉ ΣΑΣ: ΑΠΕΝΕΡΓΟΠΟΙΉΣΤΕ ΤΟΝΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ, ΑΦΑΙΡΈΣΤΕ ΤΗΝ ΚΆΝΟΥΛΑ ΚΑΙΒΓΕΊΤΕ ΑΠΌ ΤΟΝ ΧΏΡΟ ΌΠΟΥ ΒΡΊΣΚΕΤΑΙ Η ΣΥΣΚΕΥΉ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΣΉΜΑΤΑ «ΑΠΑΓΟΡΕΎΕΤΑΙ ΤΟ ΚΆΠΝΙΣΜΑ –ΧΡΉΣΗ ΟΞΥΓΌΝΟΥ» ΠΡΈΠΕΙ ΝΑ ΑΝΑΡΤΏΝΤΑΙ ΣΕ ΠΕΡΊΟΠΤΗ
ΘΈΣΗ ΣΤΟ ΣΠΊΤΙ Ή ΣΤΟΝ ΧΏΡΟ ΌΠΟΥ ΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ
ΟΞΥΓΌΝΟ. ΟΙ ΑΣΘΕΝΕΊΣ ΚΑΙ ΟΙ ΦΡΟΝΤΙΣΤΈΣ ΤΟΥΣ ΠΡΈΠΕΙΝΑ ΕΝΗΜΕΡΏΝΟΝΤΑΙ ΣΧΕΤΙΚΆ ΜΕ ΤΟΥΣ ΚΙΝΔΎΝΟΥΣ ΤΟΥΚΑΠΝΊΣΜΑΤΟΣ ΥΠΌ ΤΗΝ ΠΑΡΟΥΣΊΑ, Ή ΚΑΤΆ ΤΗ ΔΙΆΡΚΕΙΑ ΤΗΣΧΡΉΣΗΣ, ΙΑΤΡΙΚΟΎ ΟΞΥΓΌΝΟΥ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ
ΟΞΥΓΌΝΟΥ ΥΠΌ ΤΗΝ ΠΑΡΟΥΣΊΑ ΕΎΦΛΕΚΤΩΝ ΑΕΡΊΩΝ. ΑΥΤΌΜΠΟΡΕΊ ΝΑ ΟΔΗΓΉΣΕΙ ΣΕ ΤΑΧΕΊΑ ΚΑΎΣΗ ΠΡΟΚΑΛΏΝΤΑΣ
ΚΑΤΑΣΤΡΟΦΉ ΙΔΙΟΚΤΗΣΊΑΣ, ΣΩΜΑΤΙΚΟΎΣ ΤΡΑΥΜΑΤΙΣΜΟΎΣΉ ΘΆΝΑΤΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΑΦΑΊΡΕΣΗ ΤΗΣ ΚΆΝΟΥΛΑΣ ΚΑΙ ΗΑΠΌΘΕΣΉ ΤΗΣ ΕΠΆΝΩ ΣΕ ΕΝΔΎΜΑΤΑ, ΚΛΙΝΟΣΚΕΠΆΣΜΑΤΑ,ΚΑΝΑΠΈΔΕΣ, ΜΑΞΙΛΆΡΙΑ Ή ΠΑΡΌΜΟΙΑ ΥΛΙΚΆ ΘΑ ΠΡΟΚΑΛΈΣΕΙ
ΚΑΤΆΚΑΥΣΗ ΑΕΡΊΟΥ ΝΈΦΟΥΣ ΌΤΑΝ ΕΚΤΕΘΕΊ ΣΕ ΤΣΙΓΆΡΟ,ΠΗΓΉ ΘΕΡΜΌΤΗΤΑΣ Ή ΦΛΌΓΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΑΦΉΝΕΤΕ ΤΗ ΡΙΝΙΚΉ ΚΆΝΟΥΛΑ
ΕΠΆΝΩ Ή ΚΆΤΩ ΑΠΌ ΕΝΔΎΜΑΤΑ, ΚΛΙΝΟΣΚΕΠΆΣΜΑΤΑ ΉΜΑΞΙΛΆΡΙΑ ΚΑΡΈΚΛΑΣ. ΕΆΝ Η ΜΟΝΆΔΑ ΕΝΕΡΓΟΠΟΙΗΘΕΊ
ΑΛΛΆ ΔΕΝ ΧΡΗΣΙΜΟΠΟΙΗΘΕΊ, ΤΟ ΟΞΥΓΌΝΟ ΘΑ ΚΑΤΑΣΤΉΣΕΙ ΤΟΥΛΙΚΌ ΕΎΦΛΕΚΤΟ. ΘΈΣΤΕ ΤΟ ΔΙΑΚΌΠΤΗ ΛΕΙΤΟΥΡΓΊΑΣ I/Ο ΣΤΗΘΈΣΗ Ο (ΑΝΕΝΕΡΓΌ) ΌΤΑΝ Ο ΣΥΜΠΥΚΝΩΤΉΣ ΟΞΥΓΌΝΟΥ ΔΕΝΒΡΊΣΚΕΤΑΙ ΣΕ ΧΡΉΣΗ.

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΜΌΝΟ ΠΑΡΕΛΚΌΜΕΝΑ
ΠΟΥ ΣΥΝΙΣΤΏΝΤΑΙ ΑΠΌ ΤΟΝ ΚΑΤΑΣΚΕΥΑΣΤΉ. Η ΧΡΉΣΗ
ΟΠΟΙΩΝΔΉΠΟΤΕ ΆΛΛΩΝ ΜΠΟΡΕΊ ΝΑ ΕΊΝΑΙ ΕΠΙΚΊΝΔΥΝΗ, ΝΑΠΡΟΚΑΛΈΣΕΙ ΣΟΒΑΡΉ ΖΗΜΙΆ ΣΤΟ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ
ΚΑΙ ΝΑ ΚΑΤΑΣΤΉΣΕΙ ΤΗΝ ΕΓΓΎΗΣΗ ΜΗ ΈΓΚΥΡΗ.
ΠΡΟΣΟΧΗ: Αποσυνδέετε πάντα την παροχή
τροφοδοσίας AC από τον τοίχο πριν τηναποσύνδεση της τροφοδοσίας AC από τονσυμπυκνωτή οξυγόνου.
ΠΡΟΣΟΧΗ: Πάντα να τοποθετείτε τη σωλήνωση παροχής
οξυγόνου και τα καλώδια ρεύματος με τέτοιον τρόπο ώστε νααποφεύγεται ο κίνδυνος να σκοντάψει κάποιος σε αυτά.
ΠΡΟΣΟΧΗ: Προκειμένου να αποφύγετε τηνακύρωση της εγγύησης, ακολουθείτε όλες τιςοδηγίες του κατασκευαστή.
Σημείωση: Φορητός και κινητός εξοπλισμός επικοινωνιών μεραδιοσυχνότητες (RF) μπορεί να επηρεάσει τον ιατρικό ηλεκτρικό
εξοπλισμό.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΟΠΟΘΕΤΗΣΤΕ ΠΑΝΤΑ ΤΟ ΣΩΛΗΝΑΠΡΟΜΗΘΕΙΩΝ ΟΞΥΓΟΝΟΥ ΚΑΙ ΤΑ ΕΞΑΡΤΗΜΑΤΑ ΤΡΟΦΟΔΟΣΙΑΣΠΟΥ ΠΡΟΒΛΕΠΟΝΤΑΙ ΑΠΟ ΤΟΝ ΚΙΝΔΥΝΟ ΚΑΙ ΤΗΝ ΠΙΘΑΝΗΑΝΤΙΔΡΑΣΤΙΚΗ ΑΛΛΑΓΗ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΥΤΟ ΤΟ ΠΡΟΪΟΝ ΜΠΟΡΕΙ ΝΑ ΣΑΣ ΕΚΘΕΣΕΙ ΣΕΧΗΜΙΚΕΣ ΟΥΣΙΕΣ ΣΥΜΠΕΡΙΛΑΜΒΑΝΟΜΕΝΟΥ ΤΟΥ ΝΙΚΕΛΙΟΥ, ΤΟΟΠΟΙΟ ΕΙΝΑΙ ΓΝΩΣΤΟ ΣΤΗΝ ΠΟΛΙΤΕΙΑ ΤΗΣ ΚΑΛΙΦΟΡΝΙΑ ΟΤΙ ΠΡΟΚΑΛΕΙ ΚΑΡΚΙΝΟ. ΓΙΑ ΠΕΡΙΣΣΟΤΕΡΕΣ ΠΛΗΡΟΦΟΡΙΕΣ, ΜΕΤΑΒΕΙΤΕ
ΣΤΟ WWW.P65WARNINGS.CA.GOV.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΣΕ ΠΕΡΊΠΤΩΣΗ ΠΟΥ ΣΥΜΒΕΊ ΚΆΠΟΙΟ
ΣΟΒΑΡΌ ΣΥΜΒΆΝ ΜΕ ΑΥΤΉΝ ΤΗ ΣΥΣΚΕΥΉ, Ο ΧΡΉΣΤΗΣ ΘΑΠΡΈΠΕΙ ΝΑ ΑΝΑΦΈΡΕΙ ΑΜΈΣΩΣ ΤΟ ΣΥΜΒΆΝ ΣΤΟΝ ΠΑΡΟΧΈΑ Ή/ ΚΑΙ ΣΤΟΝ ΚΑΤΑΣΚΕΥΑΣΤΉ. ΈΝΑ ΣΟΒΑΡΌ ΣΥΜΒΆΝ ΟΡΊΖΕΤΑΙΩΣ ΤΡΑΥΜΑΤΙΣΜΌΣ, ΘΆΝΑΤΟΣ Ή ΠΙΘΑΝΌΤΗΤΑ ΠΡΌΚΛΗΣΗΣΤΡΑΥΜΑΤΙΣΜΟΎ / ΘΑΝΆΤΟΥ ΣΕ ΠΕΡΊΠΤΩΣΗ ΕΠΑΝΆΛΗΨΗΣ ΤΟΥΣΥΜΒΆΝΤΟΣ. Ο ΧΡΉΣΤΗΣ ΜΠΟΡΕΊ ΕΠΊΣΗΣ ΝΑ ΑΝΑΦΈΡΕΙ ΤΟΠΕΡΙΣΤΑΤΙΚΌ ΣΤΗΝ ΑΡΜΌΔΙΑ ΑΡΧΉ ΤΗΣ ΧΏΡΑΣ ΌΠΟΥ ΣΥΝΈΒΗΤΟ ΠΕΡΙΣΤΑΤΙΚΌ.

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΕΠΙΧΕΙΡΉΣΕΤΕ
ΟΠΟΙΑΔΉΠΟΤΕ ΕΡΓΑΣΊΑ ΣΥΝΤΉΡΗΣΗΣ ΕΚΤΌΣΑΠΌ ΤΙΣ ΠΙΘΑΝΈΣ ΛΎΣΕΙΣ ΠΟΥ ΠΑΡΑΤΊΘΕΝΤΑΙΣΤΟ ΠΑΡΌΝ ΕΓΧΕΙΡΊΔΙΟ. ΜΗΝ ΑΦΑΙΡΕΊΤΕΤΑ ΚΑΛΎΜΜΑΤΑ· ΜΌΝΟ Ο ΠΆΡΟΧΟΣ ΤΟΥΕΞΟΠΛΙΣΜΟΎ ΣΑΣ Ή ΈΝΑΣ ΕΞΕΙΔΙΚΕΥΜΈΝΟΣ
ΤΕΧΝΙΚΌΣ ΣΈΡΒΙΣ ΕΠΙΤΡΈΠΕΤΑΙ ΝΑ ΑΦΑΙΡΈΣΕΙΤΑ ΚΑΛΎΜΜΑΤΑ Ή ΝΑ ΣΥΝΤΗΡΉΣΕΙ ΑΥΤΉΝ ΤΗΣΥΣΚΕΥΉ.

PN 15062781-C4 J | Εγχειρίδιο χρήστη

195 - GRE

Companion 5Προδιαγραφές
Για τη σωστή χρήση της συσκευής, το ακόλουθοδιάγραμμα παρέχει σημαντικές πληροφορίες όσοναφορά το συνιστώμενο περιβάλλον λειτουργίας ή τιςσυνθήκες λειτουργίας.

Προδιαγραφές
Ρυθμοί ροήςΣυγκέντρωση O2Διαστάσεις
ΒάροςΣτάθμη ηχητικής
πίεσης
Κατανάλωση
ρεύματος
Θερμοκρασία
λειτουργίας
Υγρασία
λειτουργίας
Θερμοκρασία
αποθήκευσης
Υγρασία
αποθήκευσης
Ηλεκτρικές

0,5–5,0 LPM
90% (+5,5%/-3%)
54,6 x 31,8 x 34,2 cm (21,5 x 12,5 x 13,5 in.)
16,3 kg (36 lb)
50 dB(A) στα 2,0 LPM
285 W στα 2 LPM, 350 W μέγιστο
5 °C έως 40 °C (41 °F έως 104 °F)
15% - 95% σε σημείο δρόσου 28 ° C (82,4 °F)-25 °C έως 70 °C (-13 °F έως 158 °F)
0% - 95% χωρίς συμπύκνωση υδρατμώνΜη χρησιμοποιείτε καλώδια επέκτασης.
Μη χρησιμοποιείτε ηλεκτρικές παροχές πουελέγχονται από διακόπτη.
-382–3.011 m (-1.253–9.879 ft)Υψόμετρολειτουργίας
Χρόνος
μέχρι 24 ώρες την ημέρα
λειτουργίαςΗ ελάχιστη αναμενόμενη ωφέλιμη διάρκεια ζωής τουεξοπλισμού είναι πέντε χρόνια.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΧΡΗΣΗ ΤΗΣ ΣΥΣΚΕΥΗΣΕΚΤΟΣ ΤΩΝ ΕΙΔΙΚΩΝ ΛΕΙΤΟΥΡΓΙΚΩΝ ΣΥΝΘΗΚΩΝΑΝΑΜΕΝΕΙ ΝΑ ΕΠΗΡΕΆΣΕΙ ΑΝΕΠΙΘΎΜΗΤΑ ΤΗΝΤΑΧΎΤΗΤΑ ΡΟΉΣ ΚΑΙ ΤΟ ΠΟΣΟΣΤΌ ΤΟΥ ΟΞΥΓΌΝΟΥ
ΚΑΙ ΣΥΝΕΠΏΣ ΤΗΝ ΠΟΙΌΤΗΤΑ ΤΗΣ ΘΕΡΑΠΕΊΑΣ.
Σημείωση: Εάν ο συμπυκνωτής οξυγόνου έχει αποθηκευτεί γιαπαρατεταμένη χρονική περίοδο εκτός του κανονικού εύρους
θερμοκρασίας λειτουργίας, η μονάδα θα πρέπει να αφεθεί ναεπιστρέψει στην κανονική θερμοκρασία λειτουργίας προτού
ενεργοποιηθεί. (Ανατρέξτε στην ενότητα «Προδιαγραφές» στο παρόν
εγχειρίδιο.)196 - GREPN 15062781-C4 J | Εγχειρίδιο χρήστη

Σωστή τοποθέτηση του Companion 5Επιλέξτε μια τοποθεσία για τη συσκευή όπουαποτρέπεται η εισαγωγή καπνού, αναθυμιάσεων καιρυπαντών. Η σωστή τοποθέτηση της συσκευής πρέπει
να επιτρέπει την εισαγωγή αέρα μέσω των τριών
θέσεων εισαγωγής αέρα στο επάνω πίσω μέρος τουπεριβλήματος και κάτω από αυτό, και να επιτρέπει
επίσης την ελεύθερη έξοδο του αέρα εξαερισμού απότο άνοιγμα εξαερισμού στο κάτω αριστερό μέρος τηςσυσκευής.
Οι θέσεις εισαγωγής αέρα βρίσκονται στο επάνω
πίσω μέρος και στο κάτω μέρος του συμπυκνωτή.
Τοποθετήστε τη μονάδα με τέτοιον τρόπο ώστενα υπάρχει απόσταση τουλάχιστον 30 cm (12 in.)ανάμεσα στο συμπυκνωτή και τους τοίχους, τα έπιπλα,τις κουρτίνες ή άλλα εμπόδια.
Τοποθετήστε τη συσκευή με τέτοιον τρόπο ώστε ναμπορούν να ακούγονται οι συναγερμοί.
Τοποθετήστε τη σωλήνωση παροχής οξυγόνου μετέτοιον τρόπο ώστε να μη συστρέφεται ή φράσσεται.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ
ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΥΠΌ ΤΗΝΠΑΡΟΥΣΊΑ ΕΎΦΛΕΚΤΩΝ ΑΕΡΊΩΝ. ΑΥΤΌΜΠΟΡΕΊ ΝΑ ΟΔΗΓΉΣΕΙ ΣΕ ΤΑΧΕΊΑ ΚΑΎΣΗ
ΠΡΟΚΑΛΏΝΤΑΣ ΚΑΤΑΣΤΡΟΦΉ ΙΔΙΟΚΤΗΣΊΑΣ,ΣΩΜΑΤΙΚΟΎΣ ΤΡΑΥΜΑΤΙΣΜΟΎΣ Ή ΘΆΝΑΤΟ. ΜΗΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΛΆΔΙ, ΓΡΆΣΟ Ή ΠΡΟΪΌΝΤΑ ΜΕΒΆΣΗ ΠΕΤΡΈΛΑΙΟ Ή ΆΛΛΑ ΕΎΦΛΕΚΤΑ ΠΡΟΪΌΝΤΑΜΕ ΤΑ ΠΑΡΕΛΚΌΜΕΝΑ ΠΟΥ ΦΈΡΟΥΝ ΟΞΥΓΌΝΟΉ ΜΕ ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ. ΠΡΈΠΕΙ ΝΑΧΡΗΣΙΜΟΠΟΙΟΎΝΤΑΙ ΜΌΝΟ ΛΟΣΙΌΝ Ή ΑΛΟΙΦΈΣΜΕ ΒΆΣΗ ΤΟ ΝΕΡΌ Ή ΣΥΜΒΑΤΈΣ ΜΕ ΤΟ ΟΞΥΓΌΝΟ.
ΤΟ ΟΞΥΓΌΝΟ ΕΠΙΤΑΧΎΝΕΙ ΤΗΝ ΚΑΎΣΗ ΤΩΝΕΎΦΛΕΚΤΩΝ ΟΥΣΙΏΝ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΥΤΉ Η ΣΥΣΚΕΥΉ ΠΑΡΈΧΕΙ ΟΞΥΓΌΝΟ
ΥΨΗΛΉΣ ΣΥΓΚΈΝΤΡΩΣΗΣ, ΤΟ ΟΠΟΊΟ ΠΡΟΆΓΕΙ ΤΗΝ ΤΑΧΕΊΑ
ΚΑΎΣΗ. ΜΗΝ ΕΠΙΤΡΈΠΕΤΕ ΤΟ ΚΆΠΝΙΣΜΑ Ή ΤΙΣ ΑΝΟΙΧΤΈΣ
ΦΛΌΓΕΣ ΣΤΟΝ ΊΔΙΟ ΧΏΡΟ ΜΕ (1) ΑΥΤΉΝ ΤΗ ΣΥΣΚΕΥΉ, Ή(2) ΟΠΟΙΟΔΉΠΟΤΕ ΠΑΡΕΛΚΌΜΕΝΟ ΠΟΥ ΦΈΡΕΙ ΟΞΥΓΌΝΟ.
ΑΠΟΤΥΧΊΑ ΤΉΡΗΣΗΣ ΑΥΤΉΣ ΤΗΣ ΠΡΟΕΙΔΟΠΟΊΗΣΗΣ ΜΠΟΡΕΊ
ΝΑ ΟΔΗΓΉΣΕΙ ΣΕ ΠΥΡΚΑΓΙΆ, ΚΑΤΑΣΤΡΟΦΉ ΙΔΙΟΚΤΗΣΊΑΣ
Ή/ΚΑΙ ΝΑ ΠΡΟΚΑΛΈΣΕΙ ΦΥΣΙΚΌ ΤΡΑΥΜΑΤΙΣΜΌ Ή ΘΆΝΑΤΟ.

Companion 5Οδηγίες λειτουργίας
Πριν από τη λειτουργία

Το παρόν εγχειρίδιο χρήστη χρησιμεύει ως αναφορά
για να σας βοηθήσει με τη λειτουργία και τησυντήρηση της συσκευής. Εάν έχετε οποιεσδήποτε
ερωτήσεις ή αμφιβολίες, παρακαλούμε καλέστε τονπάροχο κατ' οίκον φροντίδας σας.
Σημαντικό! ΜΗΝ επιχειρήσετε να λειτουργήσετε
το Companion 5 χωρίς να έχετε διαβάσει πρώτα
την ενότητα «Οδηγίες ασφαλείας» του παρόντος
εγχειριδίου. Παρακαλούμε ακολουθείτε όλες τιςοδηγίες λειτουργίας. Παρακαλούμε τηρείτε όλες τιςπροειδοποιήσεις στη συσκευή και στο εγχειρίδιο
χρήστη. Για να μειωθεί ο κίνδυνος πυρκαγιάς,προσωπικού τραυματισμού και σοβαρής ζημιάς στοCompanion 5, τηρείτε όλες τις προφυλάξεις ασφαλείας.
Το CAIRE Companion 5 δεν δημιουργεί το δικό τουοξυγόνο. Παράγει οξυγόνο υψηλής συγκέντρωσης απότον αέρα του γύρω χώρου και το χορηγεί σε εσάς. Όπωςυπάρχει στη φυσική του μορφή, ο αέρας του δωματίου
περιέχει μόνο περίπου 21% οξυγόνο κατ' όγκο.
Ο αέρας του δωματίου αναρροφάται μέσα στοσυμπυκνωτή μέσω ενός συμπιεστή. Ο αέρας στησυνέχεια διέρχεται από μια σειρά από φίλτρα και έναστρώμα μοριακού κόσκινου που απορροφά χημικά
τα μόρια αζώτου. Ως αποτέλεσμα, ο εμπλουτισμένος
με οξυγόνο αέρας που εξέρχεται από το συμπυκνωτή
χορηγείται σε αναλογία 87–95% οξυγόνου κατ' όγκο.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΚΑΘΑΡΊΣΤΕ ΤΟ ΠΕΡΊΒΛΗΜΑ,ΤΟΝ ΠΊΝΑΚΑ ΕΛΈΓΧΟΥ ΚΑΙ ΤΟ ΚΑΛΏΔΙΟ
ΡΕΎΜΑΤΟΣ ΜΌΝΟ ΜΕ ΈΝΑ ΉΠΙΟ ΟΙΚΙΑΚΌ
ΚΑΘΑΡΙΣΤΙΚΌ ΠΟΥ ΑΠΛΏΝΕΤΕ ΜΕ ΥΓΡΑΜΈΝΟ
(ΌΧΙ ΜΟΥΣΚΕΜΈΝΟ) ΠΑΝΊ Ή ΣΦΟΥΓΓΆΡΙ, ΚΑΙ ΣΤΗΣΥΝΈΧΕΙΑ ΣΤΕΓΝΏΣΤΕ ΣΚΟΥΠΊΖΟΝΤΑΣ ΌΛΕΣ ΤΙΣΕΠΙΦΆΝΕΙΕΣ. ΜΗΝ ΕΠΙΤΡΈΠΕΤΕ ΤΗ ΔΙΕΊΣΔΥΣΗΟΠΟΙΟΥΔΉΠΟΤΕ ΥΓΡΟΎ ΣΤΟ ΕΣΩΤΕΡΙΚΌ ΤΗΣΣΥΣΚΕΥΉΣ. ΔΏΣΤΕ ΙΔΙΑΊΤΕΡΗ ΠΡΟΣΟΧΉ
ΣΤΗΝ ΈΞΟΔΟ ΟΞΥΓΌΝΟΥ ΓΙΑ ΤΗ ΣΎΝΔΕΣΗ ΤΗΣΚΆΝΟΥΛΑΣ ΠΡΟΚΕΙΜΈΝΟΥ ΝΑ ΔΙΑΣΦΑΛΊΣΕΤΕ ΌΤΙΠΑΡΑΜΈΝΕΙ ΑΠΑΛΛΑΓΜΈΝΗ ΑΠΌ ΣΚΌΝΗ, ΝΕΡΌΚΑΙ ΣΩΜΑΤΊΔΙΑ.
ΠΡΟΣΟΧΗ: Μην αφήνετε να φράξουν ταανοίγματα εισαγωγής ή εξόδου αέρα. Κάτιτέτοιο μπορεί να προκαλέσει υπερθέρμανση
του συμπυκνωτή οξυγόνου και να επηρεάσει
αρνητικά την απόδοση. Μη λειτουργείτε
τον συμπυκνωτή οξυγόνου χωρίς να είναι
τοποθετημένο το φίλτρο εισαγωγής αέρα. Εάνπαρέχεται ένα δεύτερο φίλτρο, τοποθετήστε το«ανταλλακτικό» φίλτρο πριν να καθαρίσετε τοβρώμικο φίλτρο. Καθαρίστε το βρώμικο φίλτρο
με ζεστό διάλυμα σαπουνιού και νερού και στησυνέχεια στεγνώστε τελείως πριν από τη χρήση.

Βήμα 1: Τοποθέτηση του Companion 5 για χρήση
Τοποθετήστε το Companion 5 σε έναν καλάαεριζόμενο, καλά φωτισμένο χώρο. Βεβαιωθείτεότι τα ανοίγματα εισόδου αέρα και εξαερισμού δενεμποδίζονται. Τοποθετήστε το Companion 5 με τέτοιον
τρόπο ώστε όλες οι ηχητικές και οπτικές ενδείξεις ή οισυναγερμοί να είναι εύκολα ακουστοί και ορατοί.
Συνδέστε τη συσκευή σε μια πρίζα AC.
Βήμα 2: Εάν ΔΕΝ ΠΡΟΚΕΙΤΑΙ να χρησιμοποιηθεί φιάλη υγραντήρα

Θέσηεισαγωγής
αέρα

Θέση
εισαγωγής
αέρα ψύξης

Άνοιγμαεξαερισμού
Θέσηεισαγωγής
αέραα. Συνδέστε μια ρινική κάνουλα ή σωλήνωση
οξυγόνου απευθείας στη θύρα εξόδου οξυγόνου.

β. Προχωρήστε στο βήμα 4.
Βήμα 3: Εάν ΠΡΟΚΕΙΤΑΙ να χρησιμοποιηθεί φιάλη υγραντήρα
α. Ξεβιδώστε το καπάκι από τη φιάλη υγραντήρα.
β. Γεμίστε τη φιάλη χρησιμοποιώντας απεσταγμένο
νερό. Βεβαιωθείτε ότι η στάθμη του νερού βρίσκεται
ανάμεσα στην καθορισμένη από τον κατασκευαστή
μέγιστη και ελάχιστη στάθμη που φαίνεται στη φιάλη.
γ. Ασφαλίστε το καπάκι της φιάλης υγραντήρα.
Βεβαιωθείτε ότι δεν υπάρχουν διαρροές.
δ. Συνδέστε μια ρινική κάνουλα ή τυπική σωλήνωση
οξυγόνου στη θύρα εξόδου που βρίσκεται στη φιάλη
υγραντήρα.
Σημείωση: Η χρήση υγραντήρα που δεν καθορίζεται για χρήση μεαυτόν το συμπυκνωτή μπορεί να επηρεάσει την απόδοσή του.

PN 15062781-C4 J | Εγχειρίδιο χρήστη

197 - GRE

Companion 5
ζ. Τοποθετήστε τη φιάλη στην προβλεπόμενη θέση στηβάση στήριξης φιάλης υγραντήρα.
η. Ασφαλίστε τη φιάλη στη βάση με χρήση τηςελαστικής ταινίας.

ε. Συνδέστε το κοχλιωτό άκρο του σωλήνα υγραντήραστο καπάκι της φιάλης υγραντήρα.

θ. Προχωρήστε στο βήμα 4.
Βήμα 4: Ενεργοποίηση και προθέρμανσηα. Συνδέστε το ηλεκτρικό καλώδιο του CAIRE
Companion 5 σε μια πρίζα AC.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΠΙΘΕΩΡΉΣΤΕ ΤΟ ΗΛΕΚΤΡΙΚΌ
ΚΑΛΏΔΙΟ ΓΙΑ ΤΥΧΌΝ ΖΗΜΙΈΣ ΠΡΙΝ ΑΠΌ ΤΗΧΡΉΣΗ. ΕΆΝ ΤΟ ΚΑΛΏΔΙΟ ΈΧΕΙ ΥΠΟΣΤΕΊ ΖΗΜΙΆ,ΜΗΝ ΤΟ ΣΥΝΔΈΣΕΤΕ ΣΕ ΗΛΕΚΤΡΙΚΉ ΠΑΡΟΧΉ
ΚΑΙ ΜΗΝ ΕΠΙΧΕΙΡΉΣΕΤΕ ΝΑ ΛΕΙΤΟΥΡΓΉΣΕΤΕ ΤΟΣΥΜΠΥΚΝΩΤΉ.

στ. Σ υνδέστε το άλλο άκρο του σωλήνα υγραντήραστη θύρα εξόδου του CAIRE Companion 5.

β. Πιέστε το διακόπτη λειτουργίας στη θέση«ΕΝΕΡΓΟ» (I). Όταν το Companion 5ενεργοποιηθεί σωστά, θα ανάψει η πράσινη
ένδειξη στην οθόνη LED. Όλες οι λυχνίες LED θαανάψουν κατά την εκκίνηση. Αφού ο συμπυκνωτής
ολοκληρώσει τον κύκλο προθέρμανσης, μόνο ηπράσινη λυχνία θα παραμείνει αναμμένη.
Σημείωση: Μετά από την αρχική ΕΝΕΡΓΟΠΟΙΗΣΗ της συσκευής,παρακαλούμε αφήστε να περάσουν έως δέκα (10) λεπτά για ναφθάσει η συσκευή στις προδιαγραφές απόδοσής της.

198 - GREPN 15062781-C4 J | Εγχειρίδιο χρήστηCompanion 5ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΊΝΑΙ ΠΟΛΎ ΣΗΜΑΝΤΙΚΌ ΝΑΕΠΙΛΈΓΕΤΕ ΜΌΝΟ ΤΟ ΣΥΝΤΑΓΟΓΡΑΦΗΜΈΝΟ
ΕΠΊΠΕΔΟ ΟΞΥΓΌΝΟΥ. ΜΗΝ ΑΛΛΆΖΕΤΕ
ΤΗΝ ΕΠΙΛΟΓΉ ΡΟΉΣ ΕΚΤΌΣ ΕΆΝ ΣΑΣ ΤΟΥΠΈΔΕΙΞΕ ΑΔΕΙΟΎΧΟΣ ΚΛΙΝΙΚΌΣ ΓΙΑΤΡΌΣ.
Ο ΣΥΜΠΥΚΝΩΤΉΣ ΟΞΥΓΌΝΟΥ ΜΠΟΡΕΊ ΝΑΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ ΚΑΤΆ ΤΗ ΔΙΆΡΚΕΙΑ ΤΟΥ ΎΠΝΟΥΥΠΌ ΤΗΝ ΚΑΘΟΔΉΓΗΣΗ ΕΞΕΙΔΙΚΕΥΜΈΝΟΥ
ΚΛΙΝΙΚΟΎ ΓΙΑΤΡΟΎ.
Βήμα 6: Επαλήθευση της ροής και κανονική αναπνοή
Βήμα 5: Ρύθμιση του ρυθμού ελέγχου ροήςα. Περιστρέψτε το περιστροφικό κουμπί ελέγχου
ροής στο ρυθμό ροής οξυγόνου (LPM) πουσυνταγογραφήθηκε από το γιατρό σας.

α. Ε παληθεύστε ότι ρέει πράγματι οξυγόνο διαμέσουτης ρινικής κάνουλας και ότι δεν υπάρχουν
συστροφές, τσακίσματα ή εμφράξεις στη σωλήνωση.
Σημείωση: Διασφαλίστε ότι η κάνουλα έχει εισαχθεί και στερεωθεί
πλήρως. Θα πρέπει να ακούσετε ή να αισθανθείτε τη ροή τουοξυγόνου στα ρινικά άκρα της κάνουλας.
β. Τοποθετήστε σωστά τη ρινική σας κάνουλα καιαναπνέετε κανονικά διαμέσου της ρινικής κάνουλας.
Βήμα 7: Απενεργοποίηση
α. Πιέστε το διακόπτη λειτουργίας στη θέση«ΑΝΕΝΕΡΓΟ» (O) όταν το CAIRE Companion 5δεν χρησιμοποιείται πλέον.

Για να ρυθμίσετε το ρυθμό ροής:Περιστρέψτε αριστερόστροφα για να αυξήσετε τη ροή.
Περιστρέψτε δεξιόστροφα για να μειώσετε τη ροή.
Το μέσον της σφαίρας υποδεικνύει το ρυθμό ροής. Ηεικόνα παρακάτω υποδεικνύει ρυθμό ροής 4,0 LPM.

ΠΡΟΣΟΧΗ: Χρησιμοποιείτε μόνο παρελκόμενα
που συνιστώνται από τον κατασκευαστή. Η χρήση
οποιωνδήποτε άλλων μπορεί να είναι επικίνδυνη,να προκαλέσει σοβαρή ζημιά στο συμπυκνωτή
οξυγόνου και να καταστήσει την εγγύηση μηέγκυρη.
ΠΡΟΣΟΧΗ: Μη χρησιμοποιείτε καλώδια επέκτασης με αυτήντη μονάδα και μη συνδέετε υπερβολικό αριθμό βυσμάτων
στην ίδια παροχή ρεύματος. Η χρήση καλωδίων επέκτασης
θα μπορούσε να επηρεάσει αρνητικά την απόδοση τηςσυσκευής. Υπερβολικός αριθμός βυσμάτων σε μία παροχή
μπορεί να οδηγήσει σε υπερφόρτωση του ηλεκτρικού πίνακα
προκαλώντας την ενεργοποίηση του ασφαλειοδιακόπτη/της ασφάλειας, ή πυρκαγιά σε περίπτωση που οασφαλειοδιακόπτης/ η ασφάλεια δεν λειτουργήσουν.
ΠΡΟΣΟΧΗ: Η χρήση καλωδίων και προσαρμογέων εκτόςαπό τα προβλεπόμενα, με την εξαίρεση των καλωδίων καιπροσαρμογέων που πωλούνται από τον κατασκευαστή
του ιατρικού ηλεκτρικού εξοπλισμού ως ανταλλακτικά γιαεσωτερικά εξαρτήματα, μπορεί να οδηγήσει σε αυξημένες
εκπομπές ή μειωμένη ατρωσία του συμπυκνωτή οξυγόνου.

Σωστό

PN 15062781-C4 J | Εγχειρίδιο χρήστη

199 - GRE

Companion 5ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΑΦΉΝΕΤΕ ΝΑ ΦΡΆΞΟΥΝΤΑ ΑΝΟΊΓΜΑΤΑ ΕΙΣΑΓΩΓΉΣ Ή ΕΞΌΔΟΥ ΑΈΡΑ.
ΚΆΤΙ ΤΈΤΟΙΟ ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΕΙ
ΥΠΕΡΘΈΡΜΑΝΣΗ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ
ΚΑΙ ΝΑ ΕΠΗΡΕΆΣΕΙ ΑΡΝΗΤΙΚΆ ΤΗΝ ΑΠΌΔΟΣΗ.
Σημείωση: Για τον παροχέα εξοπλισμού: Τα παρακάτω εξαρτήματα
χορήγησης οξυγόνου συνιστώνται για χρήση με το Companion 5:• Ρινική κάνουλα: Αριθμός CAIRE CU002-1• Σωληνάριο προσαρμογέα υγραντήρα: CAIRE Κωδικός είδους
20843882• Μπουκάλι υγραντήρα: Αριθ. Εξαρτήματος CAIRE HU003-1• Firebreak: CAIRE Αριθμός Αρ. 20629671
απαιτείται συνύπαρξη για χρήση με οποιονδήποτε σωληνίσκο.
• Το CAIRE προσφέρει πυροσβέστη που προορίζεται ναχρησιμοποιηθεί σε συνδυασμό με τον συμπυκνωτή οξυγόνου. Ηπυροσβεστική είναι μια θερμική ασφάλεια για να σταματήσει ηροή του αερίου σε περίπτωση που ο σωληνίσκος ή το σωληνάριο
οξυγόνου που βρίσκεται κάτω από τη ροή καύσης αναφλέγεταικαι καίγεται στην πυρκαγιά. Τοποθετείται σε σειρά με τη ρινική
κάνουλα ή τη σωλήνωση οξυγόνου μεταξύ του ασθενούς και τηςεξόδου οξυγόνου του Companion 5. Για τη σωστή χρήση τουπυροσβέστη, ανατρέξτε πάντα στις οδηγίες του κατασκευαστή (πουπεριλαμβάνονται σε κάθε κιτ πυροπροστασίας).
• Για οποιαδήποτε πρόσθετα συνιστώμενα εξαρτήματα, ανατρέξτε
στον κατάλογο εξαρτημάτων (PN MLLOX0010) που διατίθεται στηδιεύθυνση www.caireinc.com.

200 - GREPN 15062781-C4 J | Εγχειρίδιο χρήστηCompanion 5Καταστάσεις συναγερμού
Το CAIRE Companion 5 χρησιμοποιεί ένα συνδυασμόδύο ή τριών (για μοντέλα με OCSI) λυχνιών LED καιενός ηχητικού συναγερμού για να σας ειδοποιεί ότανυπάρχει δυσλειτουργία με το συμπυκνωτή.
1) Πράσινο LED (λειτουργία.

) - Υποδεικνύει κανονική

2) Κόκκινο LED ( ) - Υποδεικνύει δυσλειτουργία
του συστήματος. Βλ. τον πίνακα καταστάσεων
συναγερμού παρακάτω για περισσότερες
λεπτομέρειες.

ΠράσινοΚόκκινοΚίτρινο3) Κίτρινο LED () - Υποδεικνύει χαμηλή
συγκέντρωση οξυγόνου. Βλ. τον πίνακα
καταστάσεων συναγερμού παρακάτω γιαπερισσότερες λεπτομέρειες.
Ο πίνακας καταστάσεων συναγερμού στην επόμενη
σελίδα παρουσιάζει τις διαφορετικές καταστάσεις
συναγερμού που μπορεί να εμφανίζονται απότο συμπυκνωτή. Περιγράφει τις πιθανές αιτίες
και τις ενέργειες που πρέπει να εκτελέσετε σεπερίπτωση εκδήλωσης συναγερμού. Σε περίπτωσηεκδήλωσης οποιουδήποτε συναγερμού, συνιστάται
να επικοινωνήσετε με τον πάροχο υγειονομικής
περίθαλψής σας το συντομότερο δυνατόν.

Επάνω: Companion 5ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΑΓΝΟΕΊΤΕ ΤΟΥΣΣΥΝΑΓΕΡΜΟΎΣ
ΠΡΟΕΙΔΟΠΟΙΉΣΕΙΣ: Ο ΚΑΤΑΣΚΕΥΑΣΤΉΣ ΣΥΝΙΣΤΆ
ΜΙΑ ΕΝΑΛΛΑΚΤΙΚΉ ΠΗΓΉ ΣΥΜΠΛΗΡΩΜΑΤΙΚΟΎ
ΟΞΥΓΌΝΟΥ ΣΕ ΠΕΡΊΠΤΩΣΗ ΔΙΑΚΟΠΉΣ
ΡΕΎΜΑΤΟΣ, ΚΑΤΆΣΤΑΣΗΣ ΣΥΝΑΓΕΡΜΟΎ ΉΜΗΧΑΝΙΚΉΣ ΒΛΆΒΗΣ. ΣΥΜΒΟΥΛΕΥΤΕΊΤΕ ΤΟΝΓΙΑΤΡΌ Ή ΤΟΝ ΠΆΡΟΧΟ ΤΟΥ ΕΞΟΠΛΙΣΜΟΎ ΓΙΑΤΟΝ ΤΎΠΟ ΤΟΥ ΕΦΕΔΡΙΚΟΎ ΣΥΣΤΉΜΑΤΟΣ ΠΟΥΑΠΑΙΤΕΊΤΑΙ.

PN 15062781-C4 J | Εγχειρίδιο χρήστη

201 - GRE

Companion 5Πίνακας καταστάσεων συναγερμού
Ηχητικός
συναγερμός

ΈγχρωμοLEDΑπενεργοποιημένο
πράσινοΔιακοπτόμενη
ΚΟΚΚΙΝΟΔιακοπτόμενηΑιτίαΟ συμπυκνωτής λειτουργεί
κανονικά.

Επαληθεύστε ότι το ηλεκτρικό καλώδιο του συμπυκνωτή είναι
συνδεδεμένο σε μια πρίζα και ότι η πρίζα παρέχει ρεύμα. Δοκιμάστε
διαφορετική πρίζα. Εάν το πρόβλημα συνεχίζεται, συνδεθείτε σε
μια εφεδρική παροχή οξυγόνου και επικοινωνήστε με τον πάροχο
υγειονομικής περίθαλψής σας.

Δυσλειτουργία του συστήματοςΘέστε το διακόπτη λειτουργίας στη θέση «ΑΝΕΝΕΡΓΟ» αμέσως.
Αποσυνδέστε την τροφοδοσία AC από την επιτοίχια πρίζα. Περιμένετε
για 5 λεπτά. Επανασυνδέστε την τροφοδοσία AC στην επιτοίχια πρίζα
και επανενεργοποιήστε το συμπυκνωτή. Εάν ο συναγερμός επιμένει,απαιτείται σέρβις. Συνδεθείτε σε μια εφεδρική παροχή οξυγόνου καιεπικοινωνήστε με τον πάροχο υγειονομικής περίθαλψής σας.

Ο ρυθμός ροής του συστήματοςείναι πολύ υψηλός ή πολύχαμηλός

1. Διασφαλίστε ότι η κάνουλα δεν έχει συστραφεί ή φράξει. Εάν τοσύστημα χρησιμοποιείται με φιάλη υγραντήρα, διασφαλίστε ότι αυτήείναι γεμισμένη σωστά και δεν δημιουργεί έμφραξη.
2. Διασφαλίστε ότι το Companion 5 έχει σωστό αερισμό. Πρέπει ναβρίσκεται σε απόσταση τουλάχιστον 30 cm (12 in.) από οποιαδήποτε
επιφάνεια προκειμένου να διασφαλιστεί ότι τα ανοίγματα δενφράσσονται.
3. Εάν το πρόβλημα επιμένει, αλλάξτε σε εναλλακτική πηγή οξυγόνου
και επικοινωνήστε με τον πάροχο υγειονομικής περίθαλψης γιαβοήθεια.

ΚΟΚΚΙΝΟΤο Companion 5 ανίχνευσε
χαμηλά επίπεδα οξυγόνου.

Διακοπτόμενη
κίτρινοΤο Companion 5 ανίχνευσε
χαμηλά επίπεδα οξυγόνου.

Διακοπτόμενη
κίτρινο202 - GREΚαμίαΈχει διακοπεί η ηλεκτρική
τροφοδοσία του συμπυκνωτή,αλλά ο διακόπτης λειτουργίας
εξακολουθεί να βρίσκεται στηθέση «ΕΝΕΡΓΟ».

ΚΟΚΚΙΝΟΔιακοπτόμενη

ΕνέργειαPN 15062781-C4 J | Εγχειρίδιο χρήστη

1. Διασφαλίστε ότι οι θέσεις φίλτρου εισαγωγής αέρα και εξαερισμού
δεν είναι φραγμένες ούτε εμποδίζονται. 2. Διασφαλίστε ότι τοCompanion 5 βρίσκεται σε καλά αεριζόμενο χώρο. Βεβαιωθείτε ότιυπάρχει απόσταση τουλάχιστον 30 cm (12 in.) ανάμεσα στο πίσωμέρος και τις πλευρές του Companion και σε οποιαδήποτε εμπόδια
(έπιπλα, κουρτίνες κ.λπ.).
3. Εάν η κατάσταση επιμένει, αλλάξτε σε εναλλακτική πηγή οξυγόνου
και επικοινωνήστε με τον πάροχο υγειονομικής περίθαλψής σαςαμέσως.

Companion 5Αντιμετώπιση προβλημάτων
Συνιστάται να επικοινωνήσετε με τον πάροχο
υγειονομικής περίθαλψής σας εάν ο συμπυκνωτής
εκδηλώνει οποιουσδήποτε συναγερμούς ή εάν νομίζετεότι ο συμπυκνωτής δεν λειτουργεί σωστά. Δεν πρέπει
να επιχειρήσετε να επισκευάσετε το συμπυκνωτή εσείςοι ίδιοι. Η επισκευή πρέπει να πραγματοποιείται μόνοαπό εκπαιδευμένο τεχνικό σέρβις. Ωστόσο, υπάρχουν

ορισμένα βήματα αντιμετώπισης προβλημάτων
που μπορείτε να εκτελέσετε εάν αντιμετωπίσετε
προβλήματα με το CAIRE Companion 5. Αυτάπεριγράφονται στον ακόλουθο πίνακα. Μπορείτε ναχρησιμοποιήσετε αυτό το διάγραμμα ακολουθώντας
τους ελέγχους για το πρόβλημά σας με αριθμητική
σειρά.

Πίνακας αντιμετώπισης προβλημάτων από το χρήστη
Πρόβλημα
Ο συμπυκνωτήςείναιενεργοποιημένος,αλλά δενλειτουργεί.

Η ροή οξυγόνου
είναι χαμηλή ήανύπαρκτη.

Πιθανή αιτία

Ενέργεια1. Το ηλεκτρικό καλώδιο δεν έχει συνδεθεί
σε πρίζα.

1. Συνδέστε το ηλεκτρικό καλώδιο.

2. Η ηλεκτρική παροχή δεν παρέχει ρεύμα.

2. Ελέγξτε την ασφάλεια και το κύκλωμα της οικίας σας. Δοκιμάστε
διαφορετική πρίζα.

3. Εσωτερική βλάβη.

3. Συνδεθείτε σε εφεδρική παροχή οξυγόνου. Επικοινωνήστε με τονπάροχο υγειονομικής περίθαλψής σας αμέσως.

1. Η ρύθμιση ελέγχου ροής έχει αλλάξει.

1. Επαληθεύστε ότι το περιστροφικό κουμπί ελέγχου ροής βρίσκεταιστην κατάλληλη ρύθμιση λίτρων ανά λεπτό (LPM) που έχεισυνταγογραφηθεί από το γιατρό σας.

2. Η σωλήνωση έχει αποσυνδεθεί.

2. Επαληθεύστε ότι η σωλήνωση είναι συνδεδεμένη σφιχτά σε όλεςτις συνδέσεις (θύρα εξόδου, φιάλη υγραντήρα, υδατοπαγίδα κ.λπ.).
Ασφαλίστε εκ νέου οποιεσδήποτε συνδέσεις εάν είναι απαραίτητο.

3. Η σωλήνωση οξυγόνου ή η κάνουλα έχεισυστραφεί.

3. Ελέγξτε για συστροφές ή εμφράξεις στη σωλήνωση. Επανορθώστε
εάν είναι απαραίτητο.

4. Υπάρχει διαρροή στην κάνουλα ή στησωλήνωση.

4. Ελέγξτε όλη τη σωλήνωση και την κάνουλα για διαρροές. Για τοσκοπό αυτό, μπορείτε να προσπαθήσετε να ακούσετε ή να αισθανθείτετη διαφυγή αέρα. Αντικαταστήστε τη σωλήνωση ή την κάνουλα πουπαρουσιάζει τη διαρροή.

5. Η ροή αέρα προς το συμπυκνωτή εμποδίζεται.

5. Επαληθεύστε ότι τίποτα δεν φράσσει τις εισόδους του συμπυκνωτή.
Βεβαιωθείτε ότι υπάρχει απόσταση τουλάχιστον 30 cm (12 in.) απόέπιπλα, τοίχους ή άλλα εμπόδια.

6. Εσωτερική βλάβη.

6. Συνδεθείτε σε εφεδρική παροχή οξυγόνου. Επικοινωνήστε με τονπάροχο υγειονομικής περίθαλψής σας αμέσως.

Σημείωση: Εάν το Companion 5 εξακολουθεί να εκδηλώνει συναγερμό αφού χρησιμοποιήσετε τους πίνακες καταστάσεων συναγερμού καιαντιμετώπισης προβλημάτων, παρακαλούμε επικοινωνήστε με το Τμήμα Τεχνικής Υποστήριξης στο 1-800-482-2473 παρέχοντας τον κωδικό
συναγερμού που εμφανίζεται στην οθόνη LCD.

PN 15062781-C4 J | Εγχειρίδιο χρήστη

203 - GRE

Companion 5Καθαρισμός, φροντίδα και συντήρηση ρουτίνας
Συντήρηση ρουτίναςΗ συντήρηση ρουτίνας αφορά την αλλαγή τουφίλτρου χονδρών σωματιδίων στη λαβή μία φορά τηνεβδομάδα (εάν είναι εγκατεστημένο).

Φροντίδα και καθαρισμός της συσκευής από τοχρήστη

Αντικατάσταση της κάνουλας
Αντικαθιστάτε τη σωλήνωση παροχής και την κάνουλα
σε τακτική βάση όπως συνιστάται από τον πάροχο
κατ' οίκον φροντίδας σας. Ο γιατρός ή ο πάροχος κατ'οίκον φροντίδας σας θα σας παράσχουν πληροφορίες
καθαρισμού, απολύμανσης και αντικατάστασης.
Σημείωση: Πάντα να ακολουθείτε τις οδηγίες του κατασκευαστήτης κάνουλας για σωστή χρήση. Αντικαταστήστε την αναλώσιμη
κάνουλα όπως συνιστάται από τον κατασκευαστή της κάνουλας ή τονπάροχο του εξοπλισμού σας. Πρόσθετα αναλώσιμα είναι διαθέσιμααπό τον πάροχο του εξοπλισμού σας.

Φίλτρο εισαγωγής αέρα (κασέτα)Η πρόσβαση ή η αλλαγή του φίλτρου εισαγωγής αέραπρέπει να γίνεται μόνο από τον πάροχο υγειονομικής
περίθαλψής σας.
Φίλτρο χονδρών σωματιδίων (προαιρετικό):Το μεικτό φίλτρο σωματιδίων βρίσκεται στον πολύμπροστινό πυθμένα του Companion 5.

περίσσιο νερό.
4) Αφήστε το φίλτρο να στεγνώσει με τον αέρα.
5) Επανατοποθετήστε το φίλτρο στο περίβλημα.
Σημείωση: Ο κατασκευαστής δεν συνιστά την αποστείρωση αυτού
του εξοπλισμού.
Σημείωση: Μη λειτουργείτε τον συμπυκνωτή οξυγόνου χωρίς ναείναι τοποθετημένο το φίλτρο εισαγωγής αέρα.
Φιάλη υγραντήρα (εάν εφαρμόζεται)Πρέπει να ελέγχετε τη φιάλη υγραντήρα καθημερινά
προκειμένου να διασφαλίσετε ότι η στάθμη του νερού
βρίσκεται ανάμεσα στην καθορισμένη ελάχιστη
και μέγιστη στάθμη. Αναπληρώνετε τη φιάλη όπωςαπαιτείται χρησιμοποιώντας απεσταγμένο νερό.
Καθαρίζετε τη φιάλη υγραντήρα σύμφωνα με τιςοδηγίες του κατασκευαστή.
Διαδικασία καθαρισμού:ΑΠΕΝΕΡΓΟΠΟΙΗΣΤΕ το Companion 5 καιαποσυνδέστε το από την τροφοδοσία AC πριν απόοποιαδήποτε διαδικασία καθαρισμού ή απολύμανσης.
ΜΗΝ ψεκάζετε απευθείας στο εξωτερικό περίβλημα.
Χρησιμοποιήστε ένα υγραμένο (όχι μουσκεμένο) πανίή σφουγγάρι. Ψεκάστε το πανί ή το σφουγγάρι με έναδιάλυμα ήπιου απορρυπαντικού για να καθαρίσετε τοερμάριο και το καλώδιο ρεύματος. Για να απολυμάνετε
το Companion 5, χρησιμοποιήστε απολυμαντικό
Lysol® Brand II (ή ισοδύναμο). Αραιώστε σύμφωναμε τις οδηγίες του κατασκευαστή του προϊόντος
καθαρισμού, αλλά μην ψεκάζετε υγρό απευθείας στοCompanion 5.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΚΊΝΔΥΝΟΣ ΗΛΕΚΤΡΟΠΛΗΞΊΑΣ.
ΑΠΕΝΕΡΓΟΠΟΙΗΣΤΕ ΤΗ ΜΟΝΆΔΑ ΚΑΙΑΠΟΣΥΝΔΈΣΤΕ ΤΟ ΚΑΛΏΔΙΟ ΡΕΎΜΑΤΟΣ ΑΠΌ ΤΗΝΗΛΕΚΤΡΙΚΉ ΠΑΡΟΧΉ ΠΡΙΝ ΤΟΝ ΚΑΘΑΡΙΣΜΌ ΤΗΣΜΟΝΆΔΑΣ ΠΡΟΚΕΙΜΈΝΟΥ ΝΑ ΑΠΟΦΎΓΕΤΕ ΤΥΧΌΝ
ΗΛΕΚΤΡΟΠΛΗΞΊΑ ΚΑΙ ΚΊΝΔΥΝΟ ΕΓΚΑΥΜΆΤΩΝ.
ΜΌΝΟ Ο ΠΆΡΟΧΟΣ ΤΟΥ ΕΞΟΠΛΙΣΜΟΎ ΣΑΣΉ ΈΝΑΣ ΕΙΔΙΚΕΥΜΈΝΟΣ ΤΕΧΝΙΚΌΣ ΣΈΡΒΙΣ
ΕΠΙΤΡΈΠΕΤΑΙ ΝΑ ΑΦΑΙΡΈΣΕΙ ΤΑ ΚΑΛΎΜΜΑΤΑ ΉΝΑ ΣΥΝΤΗΡΉΣΕΙ ΤΗ ΜΟΝΆΔΑ.

Φίλτρο χονδρών σωματιδίων

Θα πρέπει να το επιθεωρείτε και να το καθαρίζετε
μία φορά την εβδομάδα. Για να καθαρίσετε τοφίλτρο, χρησιμοποιήστε την ακόλουθη διαδικασία:1) Αφαιρέστε το φίλτρο χονδρών σωματιδίων.
2) Πλύνετε το φίλτρο σε ζεστό νερό της βρύσης
χρησιμοποιώντας απορρυπαντικό διάλυμα ήπιου
σαπουνιού.
3) Ξεπλύνετε το φίλτρο σχολαστικά με καθαρό νερότης βρύσης και συμπιέστε το για να εξωθήσετε το
204 - GREPN 15062781-C4 J | Εγχειρίδιο χρήστη

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΚΊΝΔΥΝΟΣ ΗΛΕΚΤΡΟΠΛΗΞΊΑΣ.
ΑΠΕΝΕΡΓΟΠΟΙΗΣΤΕ ΤΗ ΜΟΝΆΔΑ ΚΑΙ ΑΠΟΣΥΝΔΈΣΤΕ ΤΟΚΑΛΏΔΙΟ ΡΕΎΜΑΤΟΣ ΑΠΌ ΤΗΝ ΗΛΕΚΤΡΙΚΉ ΠΑΡΟΧΉ ΠΡΙΝ ΤΟΝΚΑΘΑΡΙΣΜΌ ΤΗΣ ΜΟΝΆΔΑΣ ΠΡΟΚΕΙΜΈΝΟΥ ΝΑ ΑΠΟΦΎΓΕΤΕ
ΤΥΧΌΝ ΗΛΕΚΤΡΟΠΛΗΞΊΑ ΚΑΙ ΚΊΝΔΥΝΟ ΕΓΚΑΥΜΆΤΩΝ. ΜΌΝΟΟ ΠΆΡΟΧΟΣ ΤΟΥ ΕΞΟΠΛΙΣΜΟΎ ΣΑΣ Ή ΈΝΑΣ ΕΙΔΙΚΕΥΜΈΝΟΣ
ΤΕΧΝΙΚΌΣ ΣΈΡΒΙΣ ΕΠΙΤΡΈΠΕΤΑΙ ΝΑ ΑΦΑΙΡΈΣΕΙ ΤΑΚΑΛΎΜΜΑΤΑ Ή ΝΑ ΣΥΝΤΗΡΉΣΕΙ ΤΗ ΜΟΝΆΔΑ.

Companion 5ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΚΊΝΔΥΝΟΣ ΗΛΕΚΤΡΟΠΛΗΞΊΑΣ.
ΑΠΟΣΥΝΔΈΣΤΕ ΤΟ ΚΑΛΏΔΙΟ ΡΕΎΜΑΤΟΣ ΑΠΌ ΤΗΝΗΛΕΚΤΡΙΚΉ ΠΑΡΟΧΉ ΠΡΙΝ ΤΟΝ ΚΑΘΑΡΙΣΜΌ ΤΗΣ ΜΟΝΆΔΑΣ
ΠΡΟΚΕΙΜΈΝΟΥ ΝΑ ΑΠΟΦΎΓΕΤΕ ΤΥΧΌΝ ΗΛΕΚΤΡΟΠΛΗΞΊΑ
ΚΑΙ ΚΊΝΔΥΝΟ ΕΓΚΑΥΜΆΤΩΝ. ΜΌΝΟ Ο ΠΆΡΟΧΟΣ ΤΟΥΕΞΟΠΛΙΣΜΟΎ ΣΑΣ Ή ΈΝΑΣ ΕΙΔΙΚΕΥΜΈΝΟΣ ΤΕΧΝΙΚΌΣ
ΣΈΡΒΙΣ ΕΠΙΤΡΈΠΕΤΑΙ ΝΑ ΑΦΑΙΡΈΣΕΙ ΤΑ ΚΑΛΎΜΜΑΤΑ Ή ΝΑΣΥΝΤΗΡΉΣΕΙ ΤΗ ΜΟΝΆΔΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΠΡΈΠΕΙ ΝΑ ΛΑΜΒΆΝΕΤΑΙ ΜΈΡΙΜΝΑ
ΏΣΤΕ ΝΑ ΜΗ ΒΡΑΧΕΊ Ο ΣΥΜΠΥΚΝΩΤΉΣ ΟΞΥΓΌΝΟΥ Ή ΝΑΜΗ ΔΙΕΙΣΔΎΣΟΥΝ ΥΓΡΆ ΣΤΗ ΜΟΝΆΔΑ. ΚΆΤΙ ΤΈΤΟΙΟ ΜΠΟΡΕΊΝΑ ΠΡΟΚΑΛΈΣΕΙ ΔΥΣΛΕΙΤΟΥΡΓΊΑ Ή ΑΠΕΝΕΡΓΟΠΟΊΗΣΗ,ΚΑΘΏΣ ΚΑΙ ΑΥΞΗΜΈΝΟ ΚΊΝΔΥΝΟ ΗΛΕΚΤΡΟΠΛΗΞΊΑΣ ΉΕΓΚΑΥΜΆΤΩΝ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΔΕΝ ΜΠΟΡΕΙ ΝΑ ΕΙΝΑΙ ΑΣΦΑΛΗ ΓΙΑ ΤΗΣΥΝΔΕΣΗ ΤΗΣ ΕΤΑΙΡΕΙΑΣ 5 ΜΕ ΑΛΛΟ ΕΞΟΠΛΙΣΜΟ.

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΚΑΘΑΡΊΣΤΕ ΤΟ ΠΕΡΊΒΛΗΜΑ, ΤΟΝ ΠΊΝΑΚΑ
ΕΛΈΓΧΟΥ ΚΑΙ ΤΟ ΚΑΛΏΔΙΟ ΡΕΎΜΑΤΟΣ ΜΌΝΟ ΜΕ ΈΝΑ ΉΠΙΟΟΙΚΙΑΚΌ ΚΑΘΑΡΙΣΤΙΚΌ ΠΟΥ ΑΠΛΏΝΕΤΕ ΜΕ ΥΓΡΑΜΈΝΟ (ΌΧΙΜΟΥΣΚΕΜΈΝΟ) ΠΑΝΊ Ή ΣΦΟΥΓΓΆΡΙ, ΚΑΙ ΣΤΗ ΣΥΝΈΧΕΙΑ
ΣΤΕΓΝΏΣΤΕ ΣΚΟΥΠΊΖΟΝΤΑΣ ΌΛΕΣ ΤΙΣ ΕΠΙΦΆΝΕΙΕΣ. ΜΗΝΕΠΙΤΡΈΠΕΤΕ ΤΗ ΔΙΕΊΣΔΥΣΗ ΟΠΟΙΟΥΔΉΠΟΤΕ ΥΓΡΟΎ ΣΤΟΕΣΩΤΕΡΙΚΌ ΤΗΣ ΣΥΣΚΕΥΉΣ. ΔΏΣΤΕ ΙΔΙΑΊΤΕΡΗ ΠΡΟΣΟΧΉ
ΣΤΗΝ ΈΞΟΔΟ ΟΞΥΓΌΝΟΥ ΓΙΑ ΤΗ ΣΎΝΔΕΣΗ ΤΗΣ ΚΆΝΟΥΛΑΣΠΡΟΚΕΙΜΈΝΟΥ ΝΑ ΔΙΑΣΦΑΛΊΣΕΤΕ ΌΤΙ ΠΑΡΑΜΈΝΕΙ
ΑΠΑΛΛΑΓΜΈΝΗ ΑΠΌ ΣΚΌΝΗ, ΝΕΡΌ ΚΑΙ ΣΩΜΑΤΊΔΙΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΥΓΡΌ ΑΠΕΥΘΕΊΑΣ
ΕΠΆΝΩ ΣΤΗ ΜΟΝΆΔΑ. ΈΝΑΣ ΚΑΤΆΛΟΓΟΣ ΑΝΕΠΙΘΎΜΗΤΩΝΧΗΜΙΚΏΝ ΟΥΣΙΏΝ ΠΕΡΙΛΑΜΒΆΝΕΙ, ΑΛΛΆ ΌΧΙΠΕΡΙΟΡΙΣΤΙΚΆ, ΤΑ ΑΚΌΛΟΥΘΑ: ΑΛΚΟΌΛΗ ΚΑΙ ΠΡΟΪΌΝΤΑΜΕ ΒΆΣΗ ΑΛΚΟΌΛΗ, ΣΥΜΠΥΚΝΩΜΈΝΑ ΠΡΟΪΌΝΤΑ ΜΕΒΆΣΗ ΧΛΏΡΙΟ (ΧΛΩΡΙΟΎΧΟ ΑΙΘΥΛΈΝΙΟ) ΚΑΙ ΠΡΟΪΌΝΤΑ ΜΕΒΆΣΗ ΛΆΔΙ (PINE-SOL®, LESTOIL®). ΑΥΤΆ ΔΕΝ ΠΡΈΠΕΙ ΝΑΧΡΗΣΙΜΟΠΟΙΟΎΝΤΑΙ ΓΙΑ ΤΟΝ ΚΑΘΑΡΙΣΜΌ ΤΟΥ ΠΛΑΣΤΙΚΟΎ
ΠΕΡΙΒΛΉΜΑΤΟΣ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ, ΚΑΘΏΣ
ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΟΥΝ ΖΗΜΙΆ ΣΤΟ ΠΛΑΣΤΙΚΌ ΤΗΣΜΟΝΆΔΑΣ.

ΑπόρριψηΠάντοτε να επιστρέφετε το Companion 5,συμπεριλαμβανομένων όλων των εξαρτημάτων,στον πάροχο κατ’ οίκον φροντίδας σας για σωστή
απόρριψη. Μπορείτε επίσης να επικοινωνήσετε με τατοπικά γραφεία της πόλης σας για οδηγίες σχετικά μετη σωστή απόρριψη της μπαταρίας.

ΑΗΗΕ και RoHSΤα προϊόντα μας συμμορφώνονται με την οδηγία γιατον περιορισμό των επικίνδυνων ουσιών (RoHS). Δενπεριέχουν περισσότερο από ιχνοποσότητες μολύβδου ήάλλων επικίνδυνων υλικών.
ΠΡΟΣΟΧΗ: Για σωστή διάθεση, επικοινωνήστε με τονπαροχέα εξοπλισμού ή με την τοπική κυβερνητική
υπηρεσία για οδηγίες απόρριψης.

Αυτό το σύμβολο υπενθυμίζει στους
κατόχους του εξοπλισμού να τον επιστρέψουν σεεγκαταστάσεις ανακύκλωσης στο τέλος της ωφέλιμηςζωής του, σύμφωνα με την Οδηγία για τα ΑπόβληταΗλεκτρικού και Ηλεκτρονικού Εξοπλισμού (ΑΗΗΕ).

Η CAIRE και η CAIRE Inc. είναι σήματα κατατεθέντα της CAIRE
Inc. Επισκεφθείτε την ιστοσελίδα μας παρακάτω για μια πλήρη
λίστα εμπορικών σημάτων.
Εμπορικά σήματα: www.careinc.com/corporate/trademarks.
Copyright © 2020 CAIRE Inc.
Η CAIRE Inc. διατηρείτο δικαίωμα να διακόψει τα προϊόντα της, ή να αλλάξει τιςτιμές, τα υλικά, τον εξοπλισμό, την ποιότητα, τις περιγραφές,τις προδιαγραφές ή/και τις διαδικασίες για τα προϊόντα της ανάπάσα στιγμή χωρίς προηγούμενη ειδοποίηση και χωρίς περαιτέρω
υποχρέωση ή συνέπεια. Όλα τα δικαιώματα που δεν αναφέρονται
ρητά στο παρόν διατηρούνται από την εταιρεία μας, ανάλογα με τηνπερίπτωση.
PN 15062781-C4 J | Εγχειρίδιο χρήστη

205 - GRE

Companion 5
Kullanıcı Kontrolleri ve Sistem Durumu Göstergeleri
Dahili Semboller

BF Tipi uygulamalı parça (elektrikçarpmasına karşı koruma derecesi).
5333 sayılı yönetmelik

02

Oksijen Çıkışı

I/O

Açma/Kapama Düğmesi

Tıbbi cihazlara ilişkin Konsey Direktifi 93/42/EEC

İçinde servise uygun parça yoktur.
Kapağını açmayın.

IEC 60417

Avrupa Topluluğundaki yetkili temsilci

ISO 7000; Ekipmanda kullanılan grafik sembolleri - Dizinve özetDepolama veya çalışma sıcaklığı aralığı.
0632 sayılı yönetmelik

Sınıf II Ekipman, İki Kat Yalıtımlı 5172
sayılı yönetmelik
ETL Sertifikası

Depolama nem aralığı. 2620 sayılı
yönetmelik

Tıbbi elektrikli ekipmana yönelik CAN/CSA C22.2 No. 601.1 M90 için güvenlik
ajansı

Üreticinin adı ve adresi.
3082 sayılı yönetmelikÜretim tarihi. 2497 sayılı yönetmelik

FCC Bildirimi

Katalog Numarası.
2493 sayılı yönetmelik

21 CFR 801.15: Federal Düzenlemeler Kanunu Başlık 21
Federal yasalar uyarınca bu cihaz yalnızca
bir hekim tarafından veya hekimin
talimatıyla satılabilir.

Seri Numarası. 2498 sayılı yönetmelikÜst kısmıdır. 0623 sayılı yönetmelik

Konsey Direktifi 2012/19/EU: atık elektrikli ve elektronik
ekipmanlar (WEEE)
WEEE

Dikkat, kırılır. 0621 sayılı yönetmelikYağmurdan koruyun, kuru tutun.
0626 sayılı yönetmelikÇalıştırmadan önce kullanıcı el kitabını
okuyun. 1641 sayılı yönetmelik
Sayıya göre istifleme limiti. 2403 sayılı
yönetmelik
ISO 7010: Grafik sembolleri - Güvenlik renkleri ve güvenlikişaretleri - Kayıtlı güvenlik işaretleri
Kullanıcı el kitabı okunmalıdır.
M002 sayılı yönetmelik
Uyarı. W001 sayılı yönetmelik
Açık alev, yangın ve kıvılcımdan uzak
tutun. Yakınında açık alev kaynağı
bulundurmak ve sigara içmek yasaktır.
P003 sayılı yönetmelikÜnitenin yakınındayken veya üniteyiçalıştırırken sigara içmeyin.
P002 sayılı yönetmelik

206 - TUR

PN 15062781-C4 J | Kullanıcı El Kitab

IEC 60601-1: Tıbbi elektrikli ekipman Bölüm 1 Temel
güvenlik ve temel performans için genel gereksinimler

IPX1

Damlamaya karşı Korumalı EkipmanIPX1: Companion 5 sıvı girişinin zararlı
etkilerine karşı koruma sağlar. (IPX1,
IEC 60529 uyarınca)FCC BİLDİRİMİ:Bu cihaz CYBLE-022001-00 (Johnson Technology
firmasının 2450AT18B100 no’lu anteni dahil) içerebilirve FCC Kuralları'nın 15. bölümü ile uyumludur. Cihaz,
DA00-1407 sayılı FCC kamu Bilgilendirmesinde
ayrıntılarıyla açıklanan modüler verici onayına ilişkin
gereksinimleri karşılamaktadır. Vericinin çalışması şuiki koşula tabidir: (1) Bu cihaz, zararlı parazite neden
olamaz ve (2) Bu cihaz, çalışmasını istenmeyen birşekilde etkileyebilecek parazitler dahil aldığı tüm
parazitleri kabul etmelidir.
Bu ürün, ABD’de ve tüm dünyada geçerli bir veya
daha fazla patent kapsamında olabilir. Geçerli
patentlerin listesini görmek için lütfen aşağıdaki
web sitemizi ziyaret edin. Pat.: www.caireinc.com/
corporate/patents/

Companion 5
Hızlı Başlangıç Kılavuzu
1 Companion 5'inizi Ambalajdan Çıkarma

Companion 5

2 Companion 5'inizi Tanıma
Bu el kitabının geri kalanındaki tüm Uyarı, İkaz ve
ek cihaz bilgilerini gözden geçirin. Companion 5 ve
Kullanıcı Kontrol Panelinin temel özelliklerine aşinalık
kazanın.

ÖNDışMuhafaza

Güç
Anahtarı

LED
Ekran

Oksijen
Çıkış
Ucu

Saat
Ölçer

Akış
Ölçer
NemlendiriciŞişe Destek
Altlığı (Şişeisteğe bağlı)ARKATaşımaSapı ve
Hava Giriş
KonumuSoğutmaHavası Giriş
KonumuHava Giriş
KonumuPN 15062781-C4 J | Kullanıcı El Kitab

Çıkış
Ağzı

207 - TUR

Companion 5
Companion 5 Bileşenleri
Dış Muhafaza

CAIRE Companion 5'in dahili bileşenlerini saranve koruyan, dayanıklı plastik muhafaza.
Güç AnahtarıÜniteye gelen güç beslemesini başlatma
veya durdurmada kullanılan AÇMA/KAPAMA
güç düğmesi. AÇIK konum “l” sembolü ile
gösterilirken, KAPALI konum ise “O” sembolü
ile gösterilir.
Çıkış Ucu
Oksijen, filtrelendikten ve yoğunlaştırıldıktan
sonra Companion 5'ten bu noktada çıkar.
Oksijeni hastaya iletmek için ya tek lumenli
burun kanülü ya da standart oksijen
hortumu takılır. Yoğunlaştırıcıya takılacak
oksijen hortumunun maksimum uzunluğu
15,2 metredir (50 fit).
LED EkranYeşil, kırmızı ve sarı* LED ışığı içerir. Yeşil ışıknormal çalışmayı gösterirken sarı* ve kırmızıışıklar alarm durumlarını gösterir. (Dahafazla bilgi için talimatların Alarm Durumları
bölümüne bakın.)
Saat Ölçer
Yoğunlaştırıcının çalışmada geçen süresini
gösteren dijital ekran. Bir saatin en yakın
onda birlik dilimini gösterir ve sıfırlanamaz.
Yoğunlaştırıcı bir alarm durumuyla
karşılaştığında tanılayıcı alarm kodlarını
görüntüler.
Akış Ölçer
İletilen oksijenin akış oranını ayarlama ve
görüntülemede kullanılır. Akış hızları dakikada
0,5 - 5 litre (LPM) arasındadır ve döner düğme
çevrilerek ayarlanabilir.
Nemlendirici Şişe Nemlendirici şişenin takılacağı yer. Altlık,
Destek Altlığı (Şişe şişenin altlık üzerinde sabitlenmesi için elastik
bir bant içerir. Nemlendirici şişe, oksijenisteğe bağlı)akışına ek nem kazandırır. Şişe damıtılmış
su ile doldurulur ve nem burun dokularının
kurumasını engellemede kullanılır.
*Yalnızca Oksijen Yoğunlaştırıcı Durum
Göstergesi (OCSI) modelleri

208 - TUR

PN 15062781-C4 J | Kullanıcı El Kitab

3 Nemlendirici Şişesi ve Kanülün Takılması
Eğer nemlendirici şişe kullanılıyorsa önce şişeyi
damıtılmış su ile üreticinin belirttiği seviyeye kadar
doldurun. Şişeyi belirtilen destek altlığına yerleştirin
ve elastik bant ile yerine sabitleyin. Nemlendirici
hortumunun dişli ucunu şişeye, diğer ucunu ise
Companion 5'in çıkış ucuna takın. Nemlendirici şişesi
üzerindeki çıkış ucuna oksijen hortumunu ve/veya
burun kanülünü takın.

Nemlendirici şişesi kullanmıyorsanız oksijen
hortumunu ve/veya burun kanülünü doğrudan
Companion 5 çıkış ucuna takın.

Companion 5
4 Açma ve Isıtma

6 Akış Ayarını Reçete Edilmiş Olan Düzeye
Ayarlama

AC güç kordonunu AC prizine takın.

Hekiminiz tarafından reçete edilen akışı seçmek içinAkış Kontrolü Döner Düğmesini kullanın.

Güç Anahtarını Açık konuma getirin.

Güç
AnahtarıCompanion 5 doğru şekilde açıldığında LED
Ekranında yeşil bir güç göstergesi ışığı yanar.
Tüm LED ışıklar cihaz açıldığı zaman yanar.
Yoğunlaştırıcının ısınma çevrimini tamamlamasından
sonra, sadece yeşil ışık yanık durumda kalır.
Not: İlk açıldıktan sonra, O2 ışığı yanacak ve konsantrasyonözelliklerine ulaşana kadar ekranda “O2 UP” görüntülenecektir.

UYARI: YALNIZCA REÇETE EDILEN OKSIJEN
DÜZEYININ SEÇILMESI BÜYÜK ÖNEM TAŞIR.
LISANSLI BIR KLINISYEN TARAFINDAN BU YÖNDETALIMAT VERILMEDIĞI SÜRECE AKIŞ SEÇIMINIDEĞIŞTIRMEYIN. OKSIJEN YOĞUNLAŞTIRICI UYKUSIRASINDA, YETKILI KLINISYENIN TAVSIYESINETABI OLACAK ŞEKILDE KULLANILABILIR.
Burun kanülü içerisinden normal biçimde nefes alın.
Akış oranını ayarlamak için:Akışı artırmak için saat yönünün tersine çevirin.
Akışı azaltmak için saat yönünde çevirin.
Topun orta kısmı akış oranını belirtir. Aşağıdaki resme
bakın.

Güç
Göstergesi
LED'i

Doğru

PN 15062781-C4 J | Kullanıcı El Kitab

209 - TUR

Companion 5Önemli!

Güvenlik Talimatları aşağıdaki gibi tanımlanmıştır:UYARI: CİDDİ YARALANMAYA NEDEN OLABİLECEKTEHLİKELERE DAİR ÖNEMLİ GÜVENLİK BİLGİLERİ.

DİKKAT: Companion 5'te hasar oluşmasınıönlemeye yönelik önemli bilgiler.
Not: Özellikle dikkat edilmesi gereken bilgiler.

Kullanım Endikasyonları
Kullanım amacı

Companion 5 tamamlayıcı nitelikteki oksijenin hastaya
verilmesine yöneliktir. Bu cihaz yaşam desteğine
yönelik olmadığı gibi, hiçbir hasta izleme yetisi desağlamaz.
UYARI: YALNIZCA REÇETE EDILEN OKSIJEN
DÜZEYININ SEÇILMESI BÜYÜK ÖNEM TAŞIR.
LISANSLI BIR KLINISYEN TARAFINDAN BUYÖNDE TALIMAT VERILMEDIĞI SÜRECE
AKIŞ SEÇIMINI DEĞIŞTIRMEYIN. OKSIJEN
YOĞUNLAŞTIRICI, YETKILI BIR KLINISYENINÖNERISI DOĞRULTUSUNDA UYKU SIRASINDAKULLANILABILIR.
UYARI: SAĞLIK DURUMUNUZA UYGUN MIKTARDA TERAPIAMAÇLI OKSIJENIN ALINDIĞINDAN EMIN OLMAK IÇINCOMPANION 5, ANCAK BIR VEYA DAHA FAZLA AYARSIZIN ETKINLIK DÜZEYLERINIZE GÖRE AYRI OLARAKBELIRLENIP REÇETE EDILDIKTEN SONRA KULLANILMALIDIR.
COMPANION 5, BELIRLI BIR KOMBINASYONLA VEYAYOĞUNLAŞTIRICI ÜRETICISININ SPESIFIKASYONLARIYLAUYUMLU VE AYARLARINIZ BELIRLENIRKEN KULLANILMIŞ
OLAN VEYA PARÇA VE AKSESUARLARLA BIRLIKTEKULLANILMALIDIR.
UYARI: BU ÜNITE YAŞAM DESTEK ÜNITESI OLARAKKULLANILAMAZ. ZARAR GÖRÜLMESINI ENGELLEMEKIÇIN YAŞLILAR, ÇOCUKLAR VEYA RAHATSIZLIKLARINI
BELIRTEMEYEN TÜM HASTALARIN DAHA YAKINDANIZLENMESI VE/VEYA RAHATSIZLIKLARI VE/VEYA TIBBI ACIL
DURUMLARI HAKKINDA SORUMLU HASTA BAKICIYA BILGIILETEBILECEKLERI DAĞITILMIŞ BIR ALARM SISTEMININ
KULLANILMASI GEREKLI OLABILIR. İŞITME VE/VEYAGÖRME BOZUKLUĞU/BOZUKLUKLARI OLAN HASTALARDAALARMLARIN IZLENMESI KONUSUNDA YARDIM GEREKLIOLABILIR.

210 - TUR

PN 15062781-C4 J | Kullanıcı El Kitab

UYARI: HAMILE VEYA EMZIREN KADINLAR, FTALATLAR
IÇERDIĞINDEN DOLAYI BU EL KITABINDA ÖNERILENAKSESUARLARI KULLANMAMALIDIR.
UYARI: FEDERAL (ABD) YASALAR, BU CIHAZIN YALNIZCA
BIR HEKIM VEYA LISANSLI BAŞKA BIR SAĞLIK HIZMETISAĞLAYICISI TARAFINDAN YA DA BU KIŞILERIN TALIMATIYLASATILMASINI VEYA KIRALANMASINI GEREKLI KILAR.

Kullanım Kontrendikasyonları
UYARI: BELİRLİ DURUMLARDA, REÇETELENMİŞOLANIN DIŞINDA OKSİJEN KULLANIMITEHLİKELİ OLABİLİR. BU CİHAZ YALNIZCA BİRHEKİM TARAFINDAN REÇETE EDİLDİĞİNDE
KULLANILMALIDIR.
UYARI: YANICI ANESTETİKLERİN BULUNDUĞU YERLERDEKULLANILMAMALIDIR.
UYARI: ELEKTRİKLE ÇALIŞAN HER CİHAZDA OLDUĞU GİBİ,KULLANICI ELEKTRİK GÜCÜ KESİNTİSİ NEDENİYLE CİHAZIN
ÇALIŞMADIĞI DÖNEMLER YAŞAYABİLİR YA DA COMPANION
5'IN VASIFLI BİR TEKNİSYEN TARAFINDAN SERVİS İŞLEMİNE
TABİ TUTULMASI GEREKLİ OLABİLİR. COMPANION 5 BÖYLESİ
GEÇİCİ BİR KESİNTİ SONUCUNDA SAĞLIK ÜZERİNDE TERSETKİLER YAŞAYABİLECEK OLAN HİÇBİR HASTA İÇİN UYGUN
DEĞİLDİR.

Güvenlik Kılavuzları
UYARI: BU EKIPMAN ÜZERINDE HERHANGI BIRDEĞIŞIKLIK YAPILMASI YASAKTIRUYARI: ÜRETICI, ELEKTRIK KESINTISI, ALARM DURUMU
VEYA MEKANIK ARIZA IHTIMALINE KARŞI ALTERNATIFBIR TAMAMLAYICI OKSIJEN KAYNAĞININ HAZIRBULUNDURULMASINI ÖNERIR. GEREKLI OLAN YEDEKSISTEMIN TÜRÜNÜ ÖĞRENMEK IÇIN HEKIMINIZE YA DAEKIPMAN SAĞLAYICINIZA DANIŞIN.
UYARI: BU CIHAZ, HIZLI YANMAYI KOLAYLAŞTIRAN YÜKSEK
YOĞUNLUKLU OKSIJEN SAĞLAMAKTADIR. (1) BU CIHAZIN
VEYA (2) OKSIJEN TAŞIYAN HERHANGI BIR AKSESUARIN IKI(2) METRE YAKININDA SIGARA IÇILMESINE VEYA AÇIK ALEVOLMASINA IZIN VERMEYIN. BU UYARIYA UYULMAMASICIDDI YANGINLARA, MADDI HASARA YOL AÇABILIR VE/VEYAFIZIKSEL YARALANMA YA DA ÖLÜME SEBEBIYET VEREBILIR.

Companion 5
UYARI: ÜNITEYI, HAVALANDIRMANIN SINIRLI OLABILECEĞIKISITLI VEYA SIKIŞIK BIR ALANDA (ÖR. KÜÇÜK BIR KILIF VEYAEL ÇANTASI IÇINDE) ÇALIŞTIRMAYIN. BU DURUM, OKSIJEN
YOĞUNLAŞTIRICININ AŞIRI ISINMASINA VE PERFORMANSININ
DÜŞMESINE NEDEN OLABILIR.
UYARI: YOĞUNLAŞTIRICININ, DUMAN,
KIRLETICI MADDE VEYA ISTEN UZAK BIR YERDEKONUMLANDIRILMASI GEREKIR.
UYARI: BU OKSIJEN YOĞUNLAŞTIRICIDA KULLANIM
IÇIN BELIRTILMEMIŞ OLAN OKSIJEN UYGULAMAAKSESUARLARININ KULLANILMASI, ÜNITE PERFORMANSININ
DÜŞMESINE YOL AÇABILIR. ÖNERILEN AKSESUARLAR BU ELKITABINDA BELIRTILMIŞTIR.
UYARI: OKSIJEN YOĞUNLAŞTIRICI DÜŞÜRÜLDÜYSE, HASAR
GÖRDÜYSE YA DA SUYA MARUZ KALDIYSA LÜTFEN CIHAZIN
INCELENMESI VEYA OLASI ONARIMI IÇIN EVDE BAKIMSAĞLAYICINIZ ILE ILETIŞIME GEÇIN. GÜÇ KABLOSU YA DA FIŞIHASARLIYSA OKSIJEN YOĞUNLAŞTIRICIYI KULLANMAYIN.
UYARI: HAVA GIRIŞI YA DA HAVA ÇIKIŞI DELIKLERININ
ENGELLENMESINE IZIN VERMEYIN. CIHAZ ÜZERINDEKI
AÇIKLIKLARDAN HIÇBIRININ IÇINE NESNE DÜŞÜRMEYİN
VEYA SOKMAYIN. BU DURUM, OKSIJEN YOĞUNLAŞTIRICININ
AŞIRI ISINMASINA VE PERFORMANSININ DÜŞMESINE NEDEN
OLABILIR.
UYARI: İSTEĞE BAĞLI NEMLENDIRICIYI AŞIRI DERECEDEDOLDURMAYIN. İSTEĞE BAĞLI NEMLENDIRICIYI SADECENEMLENDIRICININ ÜRETICISI TARAFINDAN GÖSTERILENSEVIYEYE KADAR SUYLA DOLDURUN.
UYARI: BU EKİPMANI, ÖNCELİKLE BU EL KİTABINI OKUYUPANLADIĞINIZDAN EMİN OLDUKTAN SONRA ÇALIŞTIRIN. UYARIVE TALİMATLARI ANLAMAKTA SIKINTI YAŞIYORSANIZ BUEKİPMANI KULLANMADAN ÖNCE EKİPMAN SAĞLAYICINIZLAİLETİŞİME GEÇİN; AKSİ HALDE YARALANMA VEYA HASARMEYDANA GELEBİLİR.
UYARI: HERHANGI BIR RAHATSIZLIĞINIZ YA DA ACIL
ILGILENILMESI GEREKEN BIR HASTALIĞINIZ VARSA VAKIT
KAYBETMEDEN TIBBI DESTEK IÇIN BAŞVURUN.
UYARI: OKSIJEN YOĞUNLAŞTIRICININ, ÇALIŞTIRMA SICAKLIĞITEKNIK ÖZELLIKLERINDE BELIRTILENIN DIŞINDA BIR
SICAKLIKTA ÇALIŞTIRILMASI, YOĞUNLAŞTIRICININ OKSIJEN
YOĞUNLUĞU TEKNIK ÖZELLIKLERINE UYGUN ÇALIŞMA
BECERISINI SINIRLAYABILIR. SICAKLIK SINIRLARI IÇIN BU ELKITABININ TEKNIK ÖZELLIKLER BÖLÜMÜNE BAKIN.

UYARI: OKSIJEN YOĞUNLAŞTIRICI, BAŞKA
EKIPMANLARLA BITIŞIK VEYA ÜST ÜSTE KONMUŞ OLARAKKULLANILMAMALIDIR. BAŞKA EKIPMANLARLA BITIŞIK YA DAÜST ÜSTE KONMUŞ OLARAK KULLANILMASI GEREKIYORSACIHAZIN NORMAL ŞEKILDE ÇALIŞIP ÇALIŞMADIĞIGÖZLEMLENMELIDIR.
UYARI: ÜRETICI, ELEKTRIK KESINTISI, ALARM
DURUMU VEYA MEKANIK ARIZA IHTIMALINEKARŞI ALTERNATIF BIR TAMAMLAYICI OKSIJEN
KAYNAĞININ HAZIR BULUNDURULMASINIÖNERIR. GEREKLI OLAN YEDEK SISTEMINTÜRÜNÜ ÖĞRENMEK IÇIN HEKIMINIZE YA DAEKIPMAN SAĞLAYICINIZA DANIŞIN.
UYARI: BU CIHAZ, HIZLI YANMAYI KOLAYLAŞTIRAN YÜKSEK
YOĞUNLUKLU OKSIJEN SAĞLAMAKTADIR. (1) BU CIHAZIN
VEYA (2) OKSIJEN TAŞIYAN HERHANGI BIR AKSESUARINBULUNDUĞU ODADA SIGARA IÇILMESINE VEYA AÇIK ATEŞOLMASINA IZIN VERMEYIN. BU UYARIYA UYULMAMASICIDDI YANGINLARA, MADDI HASARA YOL AÇABILIR VE/VEYAFIZIKSEL YARALANMA YA DA ÖLÜME SEBEBIYET VEREBILIR.
UYARI: BU OKSIJEN YOĞUNLAŞTIRICIDA KULLANIM
IÇIN BELIRTILMEMIŞ OLAN OKSIJEN UYGULAMAAKSESUARLARININ KULLANILMASI, ÜNITE PERFORMANSININ
DÜŞMESINE YOL AÇABILIR. ÖNERILEN AKSESUARLAR BU ELKITABINDA BELIRTILMIŞTIR.
UYARI: OKSIJEN YOĞUNLAŞTIRICI DÜŞÜRÜLDÜYSE, HASAR
GÖRDÜYSE YA DA SUYA MARUZ KALDIYSA LÜTFEN CIHAZIN
INCELENMESI VEYA OLASI ONARIMI IÇIN EVDE BAKIMSAĞLAYICINIZ ILE ILETIŞIME GEÇIN. GÜÇ KABLOSU YA DA FIŞIHASARLIYSA OKSIJEN YOĞUNLAŞTIRICIYI KULLANMAYIN.
UYARI: CIHAZIN DÜZGÜN ÇALIŞMASINI SAĞLAMAK, YANGINVE YANIK OLUŞMA RISKINI ÖNLEMEK IÇIN SADECE ÜRETICITARAFINDAN ÖNERILEN YEDEK PARÇALARI KULLANIN.
UYARI: YANGIN VE YANIK OLUŞMA RISKINI ÖNLEMEKIÇIN OKSIJEN YOĞUNLAŞTIRICININ EKLEME PARÇALARI,
BAĞLANTILARI, HORTUMLARI VEYA DIĞER AKSESUARLARINIYAĞLAMAYIN.

UYARI: OKSIJEN TAŞINAN AKSESUARLAR VEYA OKSIJEN
YOĞUNLAŞTIRICI ILE YAĞ, GRES YA DA PETROL BAZLI VEYADIĞER YANICI ÜRÜNLERI KULLANMAYIN. YALNIZCA SUBAZLI, OKSIJEN ILE UYUMLU LOSYON YA DA MERHEMLERKULLANILMALIDIR. OKSIJEN, YANICI MADDELERINTUTUŞMASINI HIZLANDIRIR.

PN 15062781-C4 J | Kullanıcı El Kitab

211 - TUR

Companion 5
UYARI: OKSIJEN KULLANIRKEN SIGARA IÇMEK, YANGINDAN
KAYNAKLANAN YARALANMA VE BUNA BAĞLI ÖLÜMLERINEN YAYGIN NEDENIDIR. ŞU GÜVENLIK UYARILARINIIZLEMELISINIZ:UYARI: CIHAZIN YA DA OKSIJEN TAŞINAN AKSESUARLARINBULUNDUĞU ODADA SIGARA IÇILMESINE, MUM VEYA AÇIKALEV BULUNMASINA IZIN VERMEYIN.
UYARI: OKSIJEN KANÜLÜ TAKILIYKEN SIGARA IÇMEK, YÜZDEYANIKLARA NEDEN OLARAK ÖLÜMLE SONUÇLANABILIR.
UYARI: SIGARA IÇIYORSANIZ BU 3 ADIM HAYATINIZIKURTARABILIR: OKSIJEN YOĞUNLAŞTIRICIYI KAPATIN,KANÜLÜ ÇIKARIN VE CIHAZIN BULUNDUĞU ODADAN ÇIKIN.
UYARI: EVDE YA DA OKSIJENIN KULLANILDIĞI HERHANGI BIR
YERDE HER ZAMAN "SIGARA İÇILMEZ - OKSIJEN KAYNAĞI"BENZERI LEVHALAR BULUNMALIDIR. HASTALAR VE HASTABAKICILAR, TIBBI AMAÇLI OKSIJENIN BULUNDUĞU YERDEVEYA KULLANILDIĞI SIRADA SIGARA IÇMENIN TEHLIKELERI
KONUSUNDA BILGILENDIRILMELIDIR.

UYARI: SADECE ÜRETICI TARAFINDAN TAVSIYE EDILENAKSESUARLARI KULLANIN. BAŞKA HERHANGI BIRAKSESUARIN KULLANILMASI OKSIJEN YOĞUNLAŞTIRICINIZ
IÇIN ZARARLI OLABILIR, CIDDI HASARA YOL AÇABILIR VEGARANTIYI GEÇERSIZ KILAR.
DİKKAT: Oksijen yoğunlaştırıcının AC güç
kaynağını kesmeden önce, AC güç kaynağını
mutlaka prizden çekin.
DİKKAT: Oksijen kaynağı hortumlarını ve güç kablolarını her
zaman, takılarak düşme tehlikesini engelleyecek şekilde
yerleştirin.
DİKKAT: Garantinin geçersiz hale gelmesiniönlemek için üreticinin tüm talimatlarını izleyin.
Not: Taşınabilir ve mobil RF haberleşme ekipmanı tıbbi elektrikli
ekipmanı etkileyebilir.

UYARI: OKSIJEN YOĞUNLAŞTIRICINIZI YANICI GAZLARINBULUNDUĞU ORTAMDA KULLANMAYIN. BU DURUMHIZLA YANMAYA YOL AÇARAK MADDI HASARA, VÜCUTTAYARALANMALARA VEYA ÖLÜME SEBEBIYET VEREBILIR.

UYARI: HER ZAMAN OKSİJEN TEDARİK BORU VE GÜÇ KABLOLARINIYOLU TEHLİKEYİ VEYA KAZA KAZA GÜÇLENDİRMEYİ ÖNLEYEN BİR
YERE YERLEŞTİRİN.

UYARI: KANÜLÜN ÇIKARILMASI VE GIYSILERIN, YATAĞIN,KANEPENIN YA DA YASTIK BENZERI DIĞER EŞYALARIN
ÜZERINE KONULMASI, SIGARA, ISI KAYNAĞI VEYA ALEVEMARUZ KALDIĞINDA PARLAYARAK ALEV ALMASINA NEDENOLACAKTIR.

UYARI: BU ÜRÜN, CALIFORNIA EYALETİ’NCE KANSERE YOL AÇTIĞI
BİLİNEN NİKEL GİBİ KİMYASAL MADDELERE MARUZ KALMANIZANEDEN OLABİLİR. DAHA FAZLA BİLGİ İÇİN WWW.P65WARNINGS.
CA.GOV ADRESİNİ ZİYARET EDİN.

UYARI: BURUN KANÜLÜNÜ GIYSILERIN, YATAK ÖRTÜLERININ
VEYA SANDALYE MINDERLERININ ÜZERINDE YA DA ALTINDABIRAKMAYIN. ÜNITE AÇIK OLDUĞU HALDE KULLANIMDADEĞILSE OKSIJEN BU EŞYALARIN YANICI OLMASINA SEBEBIYETVERIR. OKSIJEN YOĞUNLAŞTIRICI KULLANIMDA OLMADIĞIZAMAN I/0 GÜÇ ANAHTARINI 0 (KAPALI) KONUMA GETIRIN.
UYARI: BU EL KITABINDA BELIRTILEN OLASIÇÖZÜMLER DIŞINDA HERHANGI BIR BAKIMIŞLEMINI DENEMEYIN. KAPAKLARI ÇIKARMAYIN;
SADECE EKIPMAN SAĞLAYICINIZ YA DAVASIFLI BIR SERVIS TEKNISYENI BU CIHAZINKAPAKLARINI ÇIKARMALI YA DA SERVIS IŞLEMI
GERÇEKLEŞTIRMELIDIR.

212 - TUR

PN 15062781-C4 J | Kullanıcı El Kitab

UYARI: BU CIHAZLA CIDDI BIR OLAY OLMASI DURUMUNDA,
KULLANICI OLAYI DERHAL SAĞLAYICIYA VE / VEYA ÜRETICIYEBILDIRMELIDIR. CIDDI BIR OLAY, OLAYIN TEKRARLANMASIDURUMUNDA YARALANMA, ÖLÜM VEYA YARALANMA / ÖLÜME
NEDEN OLMA POTANSIYELI OLARAK TANIMLANIR. KULLANICIAYRICA OLAYI, OLAYIN MEYDANA GELDIĞI ÜLKEDEKI YETKILI
MAKAMA RAPOR EDEBILIR.

Companion 5
Teknik Özellikler
Cihazın gereken şekilde kullanılması için aşağıdaki
tabloda tavsiye edilen çalıştırma ortamları veya çalışma
sıcaklıkları ile ilgili önemli bilgiler sağlanmaktadır.

Teknik ÖzelliklerAkış Hızları
0,5-5,0 LPM
Oksijen Yoğunluğu %90 (+%5,5%/-%3)Boyutlar
54,6x31,8x34,2 cm (21,5x12,5x13,5 in.)Ağırlık
16,3 kg (36 lb)
Ses Basınç Seviyesi 2,0 LPM’de 50 dB(A)Güç Tüketimi
2 LPM’de 285 W, 350 W MaksimumÇalışma Sıcaklığı 5 °C ila 40 °C (41 °F ila 104 °F)Çalışma Nemliliği 28 ° C (82.4 ° F) çiğ noktasında% 15 -% 95
Depolama-25°C ila 70 °C (-13 °F ila 158 °F)
Sıcaklığı
Depolama Nemi % 0 -% 95 yoğuşmasız
Elektrik
Uzatma kablosu kullanmayın. Anahtar ile
kontrol edilen elektrik prizi kullanmayın.
Çalışma İrtifası -382–3011 metre (-1253–9879 fit)Çalışma Süresi günde en fazla 24 saat
Ekipmanın öngörülen kullanım ömrü en az beş yıldır.
UYARI: BELİRTİLMİŞ ÇALIŞMA ŞARTLARININDIŞINDAN CİHAZIN KULLANILMASI, OKSİJENİN
AKIŞ HIZINI VE YÜZEYİNİ VE TEDAVİSİN
KALİTESİNİ GEÇERLİĞİNDEKİ ETKİSİNDİR.
Not: Oksijen Yoğunlaştırıcı uzun bir süre boyunca normal çalışma
sıcaklığı aralığının dışında saklanmışsa, ünite açılmadan öncenormal çalışma sıcaklığına dönmesi için beklenmelidir. (Bu el
kitabındaki Teknik Özellikler bölümüne bakın.)

Cihazı duman, is ve kirletici maddelerin girişinden
uzak olan bir yere koyun. Cihazın doğru şekilde
yerleştirilmesi, üst arka taraftaki ve kabinin altındaki
üç hava giriş konumuyla hava girişi sağlar ve çıkış
havasının cihazın sol alt tarafındaki çıkış ağzından
serbest bir şekilde çıkmasına imkan tanır.
Hava girişleri, yoğunlaştırıcının üst ve alt arka
bölümlerinde yer almaktadır. Üniteyi, yoğunlaştırıcı ile
duvar, mobilya, perde veya engelleyici diğer unsurlar
arasında en az 30 cm (12 in.) mesafe kalacak şekilde
yerleştirin.
Cihazı, alarmların duyulabileceği biçimde
konumlandırın.
Oksijen besleme hortumlarını, bükülmeyecek veya
tıkanmayacak şekilde konumlandırın.
UYARI: OKSIJEN YOĞUNLAŞTIRICINIZIYANICI GAZLARIN BULUNDUĞU ORTAMDAKULLANMAYIN. BU DURUM HIZLA YANMAYAYOL AÇARAK MADDI HASARA, VÜCUTTAYARALANMALARA VEYA ÖLÜME SEBEBIYET
VEREBILIR. OKSIJEN TAŞINAN AKSESUARLAR
VEYA OKSIJEN YOĞUNLAŞTIRICI ILE YAĞ, GRES YADA PETROL BAZLI VEYA DIĞER YANICI ÜRÜNLERIKULLANMAYIN. YALNIZCA SU BAZLI, OKSIJENILE UYUMLU LOSYON YA DA MERHEMLERKULLANILMALIDIR. OKSIJEN, YANICIMADDELERIN TUTUŞMASINI HIZLANDIRIR.
UYARI: BU CIHAZ, HIZLI YANMAYI KOLAYLAŞTIRAN YÜKSEK
YOĞUNLUKLU OKSIJEN SAĞLAMAKTADIR. (1) BU CIHAZIN
VEYA (2) OKSIJEN TAŞIYAN HERHANGI BIR AKSESUARINBULUNDUĞU ODADA SIGARA IÇILMESINE VEYA AÇIK ATEŞOLMASINA IZIN VERMEYIN. BU UYARIYA UYULMAMASICIDDI YANGINLARA, MADDI HASARA YOL AÇABILIR VE/VEYAFIZIKSEL YARALANMA YA DA ÖLÜME SEBEBIYET VEREBILIR.

Companion 5'in Doğru Konumlandırılması

PN 15062781-C4 J | Kullanıcı El Kitab

213 - TUR

Companion 5Çalıştırma TalimatlarıÇalıştırmadan ÖnceBu el kitabı, cihazı çalıştırmanız ve bakımını
yapmanızda başvuru kaynağınız işlevini görür.
Sorularınız veya kaygılarınız var ise lütfen evde bakım
sağlayıcınızı arayın.
Önemli! Companion 5'i öncelikle bu el kitabının
Güvenlik Kılavuzları bölümünü okumadan çalıştırmayı
denemeyin. Lütfen tüm çalıştırma talimatlarını
izleyin. Lütfen cihazdaki ve Kullanıcı El Kitabındaki
tüm Uyarılara uygun hareket edin. Yangın, kişisel
yaralanma ve Companion 5'te ciddi hasar riskini
azaltmak için, lütfen tüm güvenlik önlemlerine uygun
hareket edin.
CAIRE Companion 5 kendi oksijenini oluşturmaz.
Ortamdaki oda havasından son derece yoğunlaştırılmış
oksijen üreterek bunu size iletir. Oda havası, doğal
koşullar altında, yaklaşık olarak hacminin sadece %21'ioranında oksijen içerir.
Oda havası kompresör yoluyla yoğunlaştırıcı içineçekilir. Ardından, bir dizi filtreden ve nitrojen
moleküllerini kimyasal olarak eleyen moleküler elekyatağından geçer. Bunun sonucunda, yoğunlaştırıcıdançıkan oksijen açısından zengin hava, hacmine göre
%87-95 oranında oksijenle iletilir.

Adım 1: Companion 5'inizi Kullanım için KonumlandırmaCompanion 5'i iyi havalandırılan, iyi aydınlatılan
bir alana yerleştirin. Hava girişi ve egzoz ağızlarının
tıkalı olmadığından emin olun. Companion 5'i
tüm işitsel ve görsel göstergeler veya alarmlar
kolaylıkla görülebilecek veya işitilebilecek şekilde
konumlandırın.
Cihazı Dalgalı Akım Güç prizine takın.
Adım 2: Nemlendirici Şişesinin KULLANILMAMASI durumunda

Hava Giriş
KonumuSoğutmaHavası Giriş
KonumuÇıkış AğzıHava Giriş
Konumua. Burun kanülü veya oksijen hortumlarını doğrudanoksijen çıkışı ucuna bağlayın.

UYARI: KABINI, KONTROL PANELINI VE GÜÇ
KABLOSUNU YALNIZCA EV TIPI YUMUŞAK
BIR TEMIZLIK ÜRÜNÜNÜN UYGULANDIĞI
NEMLI (ISLAK OLMAYAN) BIR BEZ VEYASÜNGER ILE TEMIZLEYIN VE ARDINDAN TÜMYÜZEYLERI SILEREK KURULAYIN. CIHAZIN
IÇINE SIVI GIRMESINE IZIN VERMEYIN. KANÜL
BAĞLANTISININ OKSIJEN ÇIKIŞININ TOZ, SU
VE PARTIKÜLLERDEN ARINMIŞ OLDUĞUNDAN
EMIN OLMAK IÇIN BU ÇIKIŞA ÖZELLIKLE ÖZENGÖSTERIN.

b. Adım 4'e ilerleyin.

DİKKAT: Hava girişi ya da hava çıkışı deliklerinin
engellenmesine izin vermeyin. Bu durum,
Oksijen Yoğunlaştırıcının aşırı ısınmasına ve
performansının düşmesine neden olabilir.
Oksijen Yoğunlaştırıcıyı, hava girişi filtresi
takılı olmadan çalıştırmayın. İkinci bir filtresağlanmışsa, kirli filtreyi temizlemeden önce"yedek" filtreyi takın. Kirli filtreyi ılık su ve sabunçözeltisinde temizleyin, ardından kullanmadanönce tamamen kurutun.

Adım 3: Nemlendirici Şişesinin KULLANILMASI durumunda
a. Kapağı nemlendirici şişesinden çevirerek çıkarınb. Şişeyi damıtılmış su kullanarak doldurun. Su
seviyesinin, üreticinin, şişe üzerinde gösterilen, endüşük ve en yüksek seviyeleri arasında olduğundanemin olun.
c. Nemlendirici şişesinin kapağını sabitleyin. Sızıntı
olmadığından emin olun.
d. Nemlendirici şişesi üzerindeki çıkış ucuna bir burun
kanülü veya standart oksijen hortumu takın.
Not: Bu yoğunlaştırıcıyla kullanım için belirtilmemiş olan bir
nemlendiricinin kullanılması onun performansını etkileyebilir.

214 - TUR

PN 15062781-C4 J | Kullanıcı El Kitab

Companion 5g. Şişeyi, Nemlendirici Şişe Destek Altlığındaki özelolarak belirtilen yerine yerleştirin.
h. Şişeyi altlık üzerinde elastik bant ile sabitleyin.

e. Nemlendirici hortumunun dişli ucunu nemlendiricişişesinin kapağına bağlayın.

i. Adım 4'e ilerleyin.
Adım 4: Açma ve Isıtma
a. CAIRE Companion 5'in elektrik kablosunu bir
Dalgalı Akım prizine takın.
UYARI: ELEKTRIK KABLOSUNU KULLANIM
ÖNCESINDE HASAR AÇISINDAN INCELEYIN.
EĞER KABLO HASARLI ISE ONU BIR ELEKTRIKPRIZINE TAKMAYIN VEYA YOĞUNLAŞTIRICIYIÇALIŞTIRMAYI DENEMEYIN.

f. Nemlendirici hortumunun diğer ucunu CAIRE
Companion 5'in çıkış ucuna bağlayın.

b. Güç anahtarına basarak “AÇIK” (I) konuma getirin.
Companion 5 doğru şekilde açıldığında LED
Ekranında yeşil bir gösterge ışığı yanar. Tüm LEDışıklar cihaz açıldığı zaman yanar. Yoğunlaştırıcınınısınma çevrimini tamamlamasından sonra, sadeceyeşil ışık yanık durumda kalır.
Not: Cihazı ilk olarak AÇIK konuma getirdikten sonra lütfen cihazın
performans özelliklerine ulaşması için en fazla on (10) dakika
bekleyin.

PN 15062781-C4 J | Kullanıcı El Kitab

215 - TUR

Companion 5
UYARI: YALNIZCA REÇETE EDILEN OKSIJEN
DÜZEYININ SEÇILMESI BÜYÜK ÖNEM TAŞIR.
LISANSLI BIR KLINISYEN TARAFINDAN BU YÖNDETALIMAT VERILMEDIĞI SÜRECE AKIŞ SEÇIMINIDEĞIŞTIRMEYIN. OKSIJEN YOĞUNLAŞTIRICI UYKUSIRASINDA, YETKILI KLINISYENIN TAVSIYESINETABI OLACAK ŞEKILDE KULLANILABILIR.
Adım 6: Akışı Doğrulama ve Normal Nefes Alıp Verme

Adım 5: Akış Kontrol Hızını Ayarlama
a. Akış kontrolü döner düğmesini hekiminiz tarafından
reçete edilen oksijen akış oranına (LPM) çevirin.

a. Oksijenin burun kanülü içinden gerçek anlamdaaktığını ve hortumlarda hiçbir kıvrılma, bükülme
veya tıkanma olmadığını doğrulayın.
Not: Kanülün tamamen takılı olduğundan ve sabitlendiğinden
emin olun. Burun kanülünün sivri uçlarına oksijen akışı olduğunu
duymanız ya da hissetmeniz gerekir.
b. Burun kanülünüzü gereken şekilde konumlandırınve burun kanülü içinden normal şekilde nefes alın.
Adım 7: Kapatma
a. CAIRE Companion 5 artık kullanımda olmadığında
güç anahtarına basarak “KAPALI” (O) konuma
getirin.
DİKKAT: Sadece üretici tarafından tavsiye
edilen aksesuarları kullanın. Başka herhangi biraksesuarın kullanılması oksijen yoğunlaştırıcınız
için zararlı olabilir, ciddi hasara yol açabilir vegarantiyi geçersiz kılar.

Akış oranını ayarlamak için:Akışı artırmak için saat yönünün tersine çevirin.
Akışı azaltmak için saat yönünde çevirin.
Topun orta kısmı akış oranını belirtir. Aşağıdaki
resimde, 4,0 LPM'lik akış oranı gösterilmiştir.

DİKKAT: Bu üniteyi uzatma kablosuyla kullanmayın veya
aynı elektrik prizine çok sayıda fiş takmayın. Uzatma
kablosuyla kullanılması cihazın performansını olumsuz yönde
etkileyebilir. Tek bir prize çok sayıda fiş takılması, elektrik
panelinde aşırı yüke neden olarak devre kesicinin/sigortanındevreye girmesine ya da devre kesici veya sigorta çalışmazsa
yangına sebep olabilir.
DİKKAT: Tıbbi elektrikli ekipmanın üreticisi tarafından
iç bileşenlerin yedek parçaları olarak satılan kablolar ve
adaptörler hariç olmak üzere, belirtilenlerin dışındaki kablo ve
adaptörlerin kullanılması, Oksijen Yoğunlaştırıcının bağışıklık
sisteminin zayıflayarak daha fazla emisyon salmasına
sebebiyet verebilir.

Doğru

216 - TUR

PN 15062781-C4 J | Kullanıcı El Kitab

Companion 5
UYARI: HAVA GIRIŞI YA DA HAVA ÇIKIŞIDELIKLERININ ENGELLENMESINE IZIN VERMEYIN.
BU DURUM, OKSIJEN YOĞUNLAŞTIRICININ AŞIRI
ISINMASINA VE PERFORMANSININ DÜŞMESINENEDEN OLABILIR.
Not: Ekipman Sağlayıcıya: Companion 5 ile kullanım için aşağıdaki
oksijen uygulama aksesuarları önerilir:• Burun Kanülü: CAIRE Parça Numarası CU002-1• Nemlendirici Adaptör Hortumu: CAIRE Parça numarası 20843882• Nemlendirici Şişe: CAIRE Parça Numarası HU003-1• Ateş Çukuru: CAIRE Parça Numarası 20629671
Herhangi bir kanül ile kullanım için bir ateş kesilmesi gereklidir.
• CAIRE, oksijen konsantratörü ile birlikte kullanılması amaçlanan
bir ateş kesmesi sunar. Firebreak, aşağı akış kanülünün veya oksijen
tüpünün tutuşması ve ateş bariyerine yanması durumunda gazakışını durdurmak için termal bir sigortadır. Hasta ve Companion
5 oksijen çıkışı arasındaki burun kanülü veya oksijen tüpü ileaynı hizadadır. Firebreak’in doğru kullanımı için daima üreticinin
talimatlarına bakın (her firebreak kitiyle birlikte verilir).
• Önerilen diğer aksesuarlar için lütfen www.caireinc.com adresindebulunan Aksesuar Kataloğu’na (PN MLLOX0010) bakın.

PN 15062781-C4 J | Kullanıcı El Kitab

217 - TUR

Companion 5
Alarm Durumları
CAIRE Companion 5'inizde, yoğunlaştırıcınızda arıza
olması halinde uyarılmanız için, iki veya üç (OCSI
modellerinde) LED ışığı ve bir sesli alarm bileşimi
kullanılır.
1) Yeşil LED () - Normal çalışmayı gösterir.

2) Kırmızı LED ( ) - Sistemde arıza olduğunu
gösterir. Ayrıntılı bilgi için aşağıdaki AlarmKoşulları Tablosuna bakın.
3) Sarı LED () - Düşük oksijen yoğunluğunu
gösterir. Ayrıntılı bilgi için aşağıdaki AlarmKoşulları Tablosuna bakın.
Sonraki sayfada yer alan Alarm Koşulları Tablosu,yoğunlaştırıcıda gösterilebilecek olan farklı alarmkoşullarını göstermektedir. Alarmın yaşanması
halinde olası nedenlerini ve gerçekleştirmeniz gerekenişlemleri ortaya koyar. Alarm olması halinde, mümkün
olduğunca kısa sürede sağlık uzmanınız ile iletişime
geçmeniz tavsiye edilir.
UYARI: ALARMLARI İHMAL ETMEYINUYARILAR: ÜRETICI, ELEKTRIK KESINTISI, ALARM
DURUMU VEYA MEKANIK ARIZA IHTIMALINEKARŞI ALTERNATIF BIR TAMAMLAYICI OKSIJEN
KAYNAĞININ HAZIR BULUNDURULMASINIÖNERIR. GEREKLI OLAN YEDEK SISTEMINTÜRÜNÜ ÖĞRENMEK IÇIN HEKIMINIZE YA DAEKIPMAN SAĞLAYICINIZA DANIŞIN.

218 - TUR

PN 15062781-C4 J | Kullanıcı El Kitab

Yeşil

KırmızıSarı

Yukarıda: Companion 5

Companion 5
Alarm Koşulları Tablosu
Sesli Alarm

Renkli
LED

Kapalı
Yeşil

NedeniYoğunlaştırıcı doğru şekildeçalışmaktadır.

Yoğunlaştırıcının gücü kesilmiştir,ancak güç anahtarı hala “AÇIK”
konumdadır.

Aralıklı

Yapacağınız İşlem
YokYoğunlaştırıcının elektrik kablosunun prize takılı olduğunu ve prize
güç geldiğini doğrulayın. Farklı bir priz deneyin. Sorun devam ederseyedek oksijen kaynağı bağlayın ve sağlık uzmanınız ile iletişime geçin.

KIRMIZI

Aralıklı

Sistem Arızası

Güç anahtarını hemen “KAPALI” konuma getirin. Dalgalı Akım Gücübağlantısını duvardaki prizden kesin. 5 dakika bekleyin. Dalgalı Akım
Gücünü tekrar duvardaki prizden bağlayın ve yoğunlaştırıcıyı yenidenaçın. Alarm devam ederse servis gerekli demektir. Yedek oksijen
kaynağı bağlayın ve vakit geçirmeden sağlık uzmanınız ile iletişime
geçin.

Ürün Akış Oranı Çok Yüksek veya
Çok Düşük

1. Kanülün bükülmediğini veya tıkalı olmadığını doğrulayın.
Nemlendirici şişesi kullanıyorsanız gereken şekilde dolu olduğunu ve
tıkanma olmadığını doğrulayın.
2. Companion 5'da havalandırmanın gereken şekilde olduğunu
doğrulayın. Ağızların tıkanmamasını sağlamak için herhangi bir
yüzeyden en az 30 cm (12 in.) uzakta olmalıdır.
3. Sorun devam ederse, başka bir oksijen kaynağına geçin ve yardım
için sağlık uzmanı ile iletişim kurun.

KIRMIZI

Aralıklı
KIRMIZI

Companion 5 düşük seviyelerdeoksijen tespit etti.

Aralıklı
Sarı

Companion 5 düşük seviyelerdeoksijen tespit etti.

Aralıklı

1. Hava giriş filtresinin ve çıkış konumlarının tıkanmadığından veya
engellenmediğinden emin olun. 2. Companion 5'in iyi havalandırılan
bir alanda olduğundan emin olun. Companion'un arka bölümü ile her
türlü engel (mobilya, perde vs.) arasında en az 30 cm (12 in.) mesafeolduğundan emin olun.
3. Eğer sorun giderilemezse başka bir oksijen kaynağına geçin veya
hemen sağlık uzmanınız ile iletişime geçin.

Sarı

PN 15062781-C4 J | Kullanıcı El Kitab

219 - TUR

Companion 5
Sorun GidermeEğer yoğunlaştırıcınız herhangi bir alarm görüntülerse
ya da yoğunlaştırıcınızın gerektiği gibi çalışmadığını
hissederseniz sağlık uzmanınız ile iletişime geçmeniz
tavsiye edilir. Yoğunlaştırıcınızı kendiniz onarmagirişiminde bulunmamalısınız. Onarım sadece eğitimli
bir servis teknisyenince yerine getirilmelidir. Bununla

birlikte, CAIRE Companion 5'iniz ile sorunlaryaşamanız halinde yerine getirebileceğiniz bazı sorun
giderme adımları bulunmaktadır. Bunlar aşağıdaki
tabloda ortaya konmuştur. Bu tabloyu, sorununuz
için olan kontrolleri numara sırasına göre izleyerek
kullanabilirsiniz.

Kullanıcı Sorun Giderme Tablosu
Sorun

Olası Neden

Yapacağınız İşlem

1. Elektrik kablosu bir prize takılı değil.

1. Elektrik kablosunu prize takın.

Yoğunlaştırıcı açık
ama çalışmıyor.

2. Elektrik kablosu güç sağlamıyor.

2. Evinizin sigortasını ve devresini kontrol edin. Farklı bir priz deneyin.

3. Dahili arıza.

3. Yedek bir oksijen kaynağına bağlayın. Hemen sağlık uzmanınızla
iletişime geçin.

1. Akış kontrolü ayarı değiştirilmiş.

1. Akış kontrol döner düğmesinin, doktorunuz tarafından reçete edilen
ilgili Dakika başına Litre (LPM) ayarına göre ayarlanmış olduğunu
doğrulayın.

2. Hortumların bağlantısı kesilmiş.

2. Hortumların tüm bağlantı yerlerinde sıkı olduğunu doğrulayın (çıkış
ucu, nemlendirici şişesi, su tutucu vs.). Gerekirse bağlantıları yenidensıkılaştırın.

3. Oksijen hortumları veya kanülü bükülmüş.

3. Hortumlarınızda bükülme veya tıkanma kontrolü yapın. Gerekirse
onarın.

4. Kanülünüz veya hortumunuzda sızıntı var.

4. Tüm hortumlarda ve kanülde sızıntı incelemesi yapın. Bunu havakaçağı olup olmadığını dinleyerek veya yoklayarak yapabilirsiniz.
Sızıntılı hortumları veya kanülü değiştirin.

5. Yoğunlaştırıcıya olan hava akışı
engellenmiştir.

5. Yoğunlaştırıcının girişlerini engelleyen bir şey olmadığını
doğrulayın. Mobilya, duvar veya diğer engellerden en az 30 cm (12 fit)uzakta olduğundan emin olun.

6. Dahili arıza.

6. Yedek bir oksijen kaynağına bağlayın. Hemen sağlık uzmanınızla
iletişime geçin.

Düşük oksijen akışı
yaşıyorsunuz veya
oksijen akışı yok.

Not: Companion 5'iniz Alarm Koşulları ve Sorun Giderme tablolarına başvurduktan sonra hala alarm veriyor ise lütfen LCD Ekranındaki alarm
kodu ile birlikte 1-800-482-2473 numaralı telefondan Teknik Servis ile iletişime geçin.

220 - TUR

PN 15062781-C4 J | Kullanıcı El Kitab

Companion 5
Temizlik, Özen ve Düzenli Bakım
Düzenli Bakım

Düzenli bakım kapsamında haftada bir kez büyük
partikül filtresi (takılıysa) değiştirilir.

Cihazın Kullanıcı tarafından Bakımı ve
Temizlenmesi

Kanülü Değiştirme
Tedarik hortumlarınızı ve kanülünüzü, evde bakım
sağlayıcınız tarafından tavsiye edildiği üzere düzenli
olarak değiştirin. Hekiminiz ya da evde bakım
sağlayıcınız size temizlik, dezenfeksiyon ve değiştirmebilgilerini sunacaktır.
Not: Doğru kullanım için daima kanül üreticisinin talimatlarını
izleyin. Atılabilir kanülü, kanül üreticisi veya Ekipman Sağlayıcınız
tarafından önerilen şekilde değiştirin. Ek malzemeler EkipmanSağlayıcınızdan temin edilebilir.

Hava Giriş Filtresi (Kartuş)Yalnızca sağlık uzmanınız hava giriş filtresine erişebilir
veya bu filtreyi değiştirebilir.
Büyük Partikül Filtresi (İsteğe Bağlı):Brüt parçacık filtresi, Companion 5’in en ön
tabanında bulunur.

Büyük
Partikül
Filtresi

Haftada bir kez kontrol etmeli ve temizlemelisiniz.
Filtreyi temizlemek için aşağıdaki işlemi kullanın:Not: Üretici bu ekipmanın sterilize edilmesini önermemektedir.
Not: Oksijen Yoğunlaştırıcıyı, hava girişi filtresi takılı olmadançalıştırmayın.
Nemlendirici Şişesi (Varsa)
Nemlendirici şişenizi her gün kontrol ederek, su
seviyesinin belirtilen en düşük ve en yüksek seviyelerarasında olduğundan emin olmalısınız. Şişeyi,damıtılmış su kullanarak gerektiğinde yeniden
doldurun. Nemlendirici şişesini üretici talimatlarınauygun şekilde temizleyin.
Temizlik Prosedürü:
Her türlü temizlik veya dezenfeksiyon işlemi
öncesinde Companion 5'i KAPALI konuma getirinve AC gücüyle bağlantısını kesin. Dış muhafazayadoğrudan ürün PÜSKÜRTMEYİN. Nemli (ıslak
olmayan) bir bez veya sünger kullanın. Kabini ve güç
kablosunu temizlemek için beze veya süngere hafif bir
deterjanlı çözelti püskürtün. Companion 5'i dezenfekte
etmek için Lysol® Brand II dezenfektan (veya muadili)kullanın. Temizlik ürününü üreticisi tarafından
belirtilen şekilde seyreltin, ancak sıvıyı doğrudan
Companion 5 üzerine püskürtmeyin.
UYARI: ELEKTRIK ÇARPMASI TEHLIKESI. ÜNITEYITEMIZLEMEDEN ÖNCE, KAZARA ELEKTRIKÇARPMASI VE YANMA TEHLIKESINI ÖNLEMEKIÇIN ÜNITEYI KAPALI KONUMA GETIRIN VE GÜÇ
KABLOSUNU ELEKTRIK PRIZINDEN ÇIKARIN.
YALNIZCA EKIPMAN SAĞLAYICINIZ VEYA YETKILIBIR SERVIS TEKNISYENI ÜNITENIN KAPAKLARINIÇIKARMALI YA DA ÜNITE ÜZERINDE SERVISIŞLEMI GERÇEKLEŞTIRMELIDIR.
UYARI: ELEKTRIK ÇARPMASI TEHLIKESI. ÜNITEYITEMIZLEMEDEN ÖNCE, KAZARA ELEKTRIK ÇARPMASI VE
YANMA TEHLIKESINI ÖNLEMEK IÇIN ÜNITEYI KAPALI KONUMAGETIRIN VE GÜÇ KABLOSUNU ELEKTRIK PRIZINDEN ÇIKARIN.
YALNIZCA EKIPMAN SAĞLAYICINIZ VEYA YETKILI BIR SERVISTEKNISYENI ÜNITENIN KAPAKLARINI ÇIKARMALI YA DA ÜNITE
ÜZERINDE SERVIS IŞLEMI GERÇEKLEŞTIRMELIDIR.

1) Büyük partikül filtresini çıkarın.
2) Filtreyi, ılık çeşme suyunda yumuşak bir sabun
solüsyonu kullanarak yıkayın.
3) Filtreyi temiz çeşme suyuyla tamamen durulayın ve
fazla suyu sıkın.
4) Filtreyi kurutmak için havalandırın.
5) Filtreyi kabine geri takın.

PN 15062781-C4 J | Kullanıcı El Kitab

221 - TUR

Companion 5
UYARI: ELEKTRIK ÇARPMASI TEHLIKESI. ÜNITEYITEMIZLEMEDEN ÖNCE, KAZARA ELEKTRIK ÇARPMASI VE
YANMA TEHLIKESINI ÖNLEMEK IÇIN GÜÇ KORDONUNU
ELEKTRIK PRIZINDEN ÇIKARIN. YALNIZCA EKIPMAN
SAĞLAYICINIZ VEYA YETKILI BIR SERVIS TEKNISYENI ÜNITENINKAPAKLARINI ÇIKARMALI YA DA ÜNITE ÜZERINDE SERVISIŞLEMI GERÇEKLEŞTIRMELIDIR.
UYARI: OKSIJEN YOĞUNLAŞTIRICININ ISLANMASININ VEYAÜNITEYE SIVI GIRMESININ ENGELLENMESINI SAĞLAMAYAÖZEN GÖSTERILMELIDIR. BU DURUM ARIZAYA YA DAKAPANMAYA YOL AÇABILIR VE ELEKTRIK ÇARPMASI VEYAYANMA RISKINDE ARTIŞA SEBEBIYET VEREBILIR.
UYARI: FİRMA 5’İ DİĞER EKİPMANLARLA KESİNTİSİZ
BAĞLAMAK GÜVENLİ OLABİLİR.

UYARI: KABINI, KONTROL PANELINI VE GÜÇ KABLOSUNUYALNIZCA EV TIPI YUMUŞAK BIR TEMIZLIK ÜRÜNÜNÜNUYGULANDIĞI NEMLI (ISLAK OLMAYAN) BIR BEZ VEYA SÜNGER
ILE TEMIZLEYIN VE ARDINDAN TÜM YÜZEYLERI SILEREKKURULAYIN. CIHAZIN IÇINE SIVI GIRMESINE IZIN VERMEYIN.
KANÜL BAĞLANTISININ OKSIJEN ÇIKIŞININ TOZ, SU VE
PARTIKÜLLERDEN ARINMIŞ OLDUĞUNDAN EMIN OLMAK IÇINBU ÇIKIŞA ÖZELLIKLE ÖZEN GÖSTERIN.
UYARI: DOĞRUDAN ÜNITE ÜZERINDE SIVI KULLANMAYIN.
İSTENMEYEN KIMYASAL MADDELERIN LISTESI AŞAĞIDAKILERIIÇERIR, ANCAK BUNLARLA SINIRLI DEĞILDIR: ALKOL VE
ALKOL BAZLI ÜRÜNLER, KONSANTRE KLOR BAZLI ÜRÜNLER(ETILEN KLORÜR) VE YAĞ BAZLI ÜRÜNLER (PINE-SOL®,LESTOIL®). BUNLAR ÜNITENIN PLASTIK MALZEMESINE ZARAR
VEREBILECEĞINDEN, OKSIJEN YOĞUNLAŞTIRICININ PLASTIKMUHAFAZASINI TEMIZLEMEDE KULLANILMAMALIDIR.

Elden çıkarma
Companion 5 ve tüm bileşenlerini, atılmak üzere
mutlaka evde bakım hizmeti sağlayıcınıza iade edin.
Pilin doğru bir biçimde imha edilmesi için şehrinizin
yerel idareleriyle de iletişime geçebilirsiniz.

WEEE ve RoHSÜrünlerimiz, Tehlikeli Maddelerin Kısıtlanması(RoHS) direktifine uyar. Eser miktardan fazla kurşun
veya başka tehlikeli malzeme içermezler.
DİKKAT: Düzgün imha için, imha talimatları için ekipmansağlayıcınıza veya yerel resmi kurumunuza başvurun.

Bu sembol, ekipmanın sahiplerine kullanımömrü sonunda ekipmanı Atık Elektrikli ve Elektronik
Ekipmanlar (WEEE) Direktifine uygun olarak bir
geri dönüşüm tesisine iade etmelerini hatırlatmaya
yöneliktir.

CAIRE ve CAIRE Inc., CAIRE Inc.’in tescilli ticari markalarıdır.
Ticari markaların tam listesi için lütfen web sitemizi ziyaret edin.
Ticari markalar: www.careinc.com/corporate/trademarks.
Telif hakkı © 2020 CAIRE Inc.   CAIRE Inc., önceden habervermeden ve hiçbir sorumluluk altına girmeden dilediği zamanürünlerin üretimini durdurma veya ürünlerinin fiyatlarını,
malzemelerini, ekipmanlarını, kalitesini, tanımlarını, özelliklerini ve/veya prosedürlerini değiştirme hakkını saklı tutar. Bu belgede açıkça
ifade edilmeyen tüm haklar, geçerli olduğu üzere tarafımızca saklıdır.
222 - TUR

PN 15062781-C4 J | Kullanıcı El Kitab

Companion 5Uživatelské ovládací prvky a ukazatele stavu systémuInterní symboly

02Výstup kyslíku

I/O

Vypínač zap./vyp.

Směrnice Rady 93/42/EHS o zdravotnických prostředcích
Zplnomocněný zástupce v Evropském
společenství
IEC 60417

Neobsahuje žádné opravitelné části.
Neotevírejte kryt.
ISO 7000: Grafické značky pro použití na zařízeních – rejstřík
a přehledZařízení třídy II s dvojitou izolací. Reg.
č. 5172
Certifikace ETL

Rozsah skladovacích nebo provozních
teplot. Reg. č. 0632

Bezpečnostní agentura pro normu
CAN/CSA C22.2, č. 601.1 M90 pro
zdravotnické elektrické přístroje

Rozsah vlhkosti při skladování. Reg.
č. 2620
Název a adresa výrobce. Reg. č. 3082
Datum výroby. Reg. č. 2497

Vyhláška FCC
21 CFR 801.15: Sbírka federálních předpisů, hlava 21
Tato zařízení smí být podle federálních
zákonů prodávána pouze lékaři nebo na
jejich příkaz.

Katalogové číslo. Reg. č. 2493
Sériové číslo. Reg. č. 2498

Směrnice Rady 2012/19/EU: o odpadních elektrických
a elektronických zařízeních (OEEZ)

Touto stranou nahoru. Reg. č. 0623
Křehké, manipulujte opatrně.
Reg. č. 0621
Chraňte před deštěm, uchovávejte
v suchu. Reg. č. 0626
Před použitím si přečtěte uživatelskou
příručku. Reg. č. 1641
Limit stohování podle počtu. Reg. č.
2403
ISO 7010: Grafické značky – bezpečnostní barvy
a bezpečnostní značky – registrované bezpečnostní značky
Je nezbytné přečíst příručku k obsluze.
Reg. č. M002
Varování. Reg. č. W001
Chraňte před otevřeným plamenem,ohněm, jiskrami. Zákaz kouření
a manipulace s plamenem. Reg. č. P003
V blízkosti jednotky ani při manipulacis ní nekuřte. Reg. č. P002Příložná část typu BF (stupeň ochrany
proti úrazu elektrickým proudem).
Reg. č. 5333

OEEZ
IEC 60601-1: Zdravotnické elektrické přístroje, část 1:
Všeobecné požadavky na základní bezpečnost a nezbytnou
funkčnost

IPX1Zařízení s úrovní ochrany proti kapajícívodě – IPX1: Přístroj Companion 5
poskytuje ochranu proti škodlivým
účinkům vniknutí kapalin. (IPX1, podle
IEC 60529)

VYHLÁŠKA FCC:

Tento přístroj může obsahovat modul
CYBLE-022001-00, včetně antény 2450AT18B100
od společnosti Johnson Technology, splňuje část 15
předpisů FCC. Tento přístroj splňuje požadavky pro
schválení modulárního vysílače, jak je uvedeno veveřejné vyhlášce FCC DA00-1407. Provoz vysílače \podléhá následujícím dvěma podmínkám: (1) Tentopřístroj nesmí způsobovat škodlivé rušení a (2) musísnést jakékoli přijaté rušení, včetně rušení, které můžezpůsobit nežádoucí provoz.
Na tento výrobek se může v USA i mezinárodně
vztahovat jeden nebo více patentů. Seznam platných
patentů najdete na níže uvedené webové stránce.
Pat.: www.caireinc.com/corporate/patents/

PN 15062781-C4 J | Uživatelská příručka

223 – CZHCompanion 5
Stručný úvodní návod
1 Vybalení přístroje Companion 5

Companion 5

2 Seznámení s přístrojem Companion 5
Projděte si všechna varování, upozornění a další
informace o přístroji ve zbytku této příručky. Seznamte
se s hlavními prvky přístroje Companion 5 a jehouživatelským ovládacím panelem.

PŘEDNÍ STRANAVnější
kryt

Hlavní
vypínačLED
displejVýstupprokyslíkPočitadlohodinPrůtokoměr
Podstavec
pro láhev
zvlhčovače(láhev jako
volitelnépříslušenství)ZADNÍ STRANA
Rukojeť
pro nošení
a místopřívoduvzduchuMístopřívoduchladicího
vzduchuMísto přívodu
vzduchu

224 – CZH PN 15062781-C4 J | Uživatelská příručka

Otvor
vyfukovaného
vzduchuCompanion 5
Součásti přístroje Companion 5Vnější kryt

Hlavní vypínačVýstupLED displej

Počitadlo hodinPrůtokoměr

Podstavec pro
láhev zvlhčovače(láhev je volitelnépříslušenství)

Kryt z odolného plastu, ve kterém jsou uzavřeny
a který chrání vnitřní součásti přístroje
Companion 5 společnosti CAIRE.
Hlavní vypínač ZAP./VYP. používaný k vypnutínebo zapnutí napájení jednotky. Poloha ZAP.
je označena symbolem „l“ a poloha VYP. jeoznačena symbolem „O“.
Zde kyslík vychází z přístroje Companion 5
po filtraci a koncentrování. Pro přívod kyslíku
k pacientovi se používá buď jednolumenovánosní kanyla, nebo standardní kyslíková
hadička. Maximální délka kyslíkové hadičkypřipojené ke koncentrátoru je 15,2 m (50 ft.).
Obsahuje zelenou, červenou a žlutou* LED
kontrolku. Zelená kontrolka znamená normální
provoz a žlutá* a červená kontrolka znamenají
stavy alarmů. (Více informací najdete v oddílu
Stavy alarmů.)Digitální zobrazení uplynulé doby provozu
koncentrátoru. Ukazuje čas na nejbližší desetinu
hodiny a nelze ho resetovat. Zobrazuje kódy
diagnostických alarmů, když v koncentrátoru
dojde ke stavu alarmu.
Používá se k nastavení a zobrazení průtokové
rychlosti dodávaného kyslíku. Rozsahprůtokových rychlostí je 0,5–5 litrů za minutu(l/min) a je možné ho upravit otočením
knoflíku.
Místo pro připevnění láhve zvlhčovače.
Podstavec obsahuje gumičku používanou
k zajištění láhve na podstavci. Láhev zvlhčovače
dodává dodatečnou vlhkost do proudu kyslíku.
Láhev je naplněna destilovanou vodou a vlhkost
se používá k tomu, aby se zabránilo vysychání
nosních tkání.
*Pouze modely s ukazatelem stavu kyslíkového
koncentrátoru (OCSI – Oxygen Concentrator
Status Indicator)

3 Připojení láhve zvlhčovače a kanyly
Jestliže používáte láhev zvlhčovače, nejprve láhevnaplňte destilovanou vodou na úroveň stanovenouvýrobcem. Umístěte láhev na určený podstavec
a zajistěte ji gumičkou. Připojte závitový konec
hadičky zvlhčovače k láhvi a druhý konec k výstupu
na přístroji Companion 5. Připojte kyslíkovou hadičku
a/nebo nosní kanylu k výstupu na láhvi zvlhčovače.

Jestliže láhev zvlhčovače nepoužíváte, připojte
kyslíkovou hadičku a/nebo nosní kanylu přímo
k výstupu přístroje Companion 5.

PN 15062781-C4 J | Uživatelská příručka

225 – CZHCompanion 54 Zapnutí a zahřívání
Zapojte napájecí kabel střídavého proudu do zásuvky
střídavého proudu.

6 Úprava nastavení průtoku na předepsanouúroveň
Použijte regulační knoflík průtoku ke zvolení průtoku,který předepíše váš lékař.

Přepněte hlavní vypínač do polohy Zap.

Hlavní
vypínačKdyž je přístroj Companion 5 řádně zapnutý, rozsvítí
se na LED displeji zelený indikátor napájení. Po
spuštění se rozsvítí všechny LED kontrolky. Kdyžkoncentrátor dokončí cyklus zahřívání, zůstanerozsvícená pouze zelená kontrolka.
Poznámka: Po počátečním zapnutí se rozsvítí kontrolka O2 ana displeji se zobrazí „O2 UP“, dokud koncentrace nedosáhne
specifikací.

VAROVÁNÍ: JE VELMI DŮLEŽITÉ ZVOLIT POUZEPŘEDEPSANOU ÚROVEŇ KYSLÍKU. NEMĚŇTE
ÚROVEŇ PRŮTOKU, POKUD VÁM TO NEBYLO
NAŘÍZENO OŠETŘUJÍCÍM LÉKAŘEM. PODLEDOPORUČENÍ KVALIFIKOVANÉHO LÉKAŘE LZEKYSLÍKOVÝ KONCENTRÁTOR POUŽÍVAT I BĚHEM
SPÁNKU.
Normálně dýchejte nosní kanylou.
Nastavení průtokové rychlosti:
Otáčením proti směru hodinových ručiček se průtok
zvyšuje.
Otáčením ve směru hodinových ručiček se průtok
snižuje.
Střed kuličky označuje průtokovou rychlost. Víz
obrázek níže.

LED
indikátor
napájení

Správně

226 – CZH PN 15062781-C4 J | Uživatelská příručka

Companion 5Důležité!

Bezpečnostní pokyny jsou určeny následujícím způsobem:
VAROVÁNÍ: DŮLEŽITÉ BEZPEČNOSTNÍ INFORMACEO NEBEZPEČÍCH, KTERÁ MOHOU ZPŮSOBIT VÁŽNÁ
ZRANĚNÍ.
UPOZORNĚNÍ: Důležité informace, jak zabránit
poškození přístroje Companion 5.
Poznámka: Informace, které vyžadují zvláštní pozornost.

Indikace pro použití
Zamýšlené použití

Přístroj Companion 5 je určen pro podávání
doplňkového kyslíku. Přístroj není určen k podpoře
života ani neposkytuje žádné možnosti sledovánípacienta.
VAROVÁNÍ: JE VELMI DŮLEŽITÉ ZVOLIT POUZEPŘEDEPSANOU ÚROVEŇ KYSLÍKU. NEMĚŇTE
ÚROVEŇ PRŮTOKU, POKUD VÁM TO NEBYLO
NAŘÍZENO OŠETŘUJÍCÍM LÉKAŘEM. PODLEDOPORUČENÍ KVALIFIKOVANÉHO LÉKAŘE
LZE PŘENOSNÝ KYSLÍKOVÝ KONCENTRÁTOR
POUŽÍVAT I BĚHEM SPÁNKU.
VAROVÁNÍ: ABY BYLO ZAJIŠTĚNO DODÁVÁNÍ TERAPEUTICKÉHOMNOŽSTVÍ KYSLÍKU PODLE VAŠEHO ZDRAVOTNÍHO STAVU,PŘÍSTROJ COMPANION 5 SE MUSÍ POUŽÍVAT AŽ POTÉ, COVÁM BYLO INDIVIDUÁLNĚ STANOVENO NEBO PŘEDEPSÁNOJEDNO NEBO VÍCE NASTAVENÍ PRO VAŠE SPECIFICKÉ ÚROVNĚ
AKTIVITY. PŘÍSTROJ COMPANION 5 SE MUSÍ POUŽÍVAT SE
SPECIFICKOU KOMBINACÍ SOUČÁSTÍ A PŘÍSLUŠENSTVÍ,KTERÉ JSOU V SOULADU SE SPECIFIKACÍ OD VÝROBCE
KONCENTRÁTORU A KTERÉ BYLY POUŽITY, KDYŽ BYLAURČOVÁNA VAŠE NASTAVENÍ.
VAROVÁNÍ: TATO JEDNOTKA NENÍ URČENA K POUŽITÍ PRO
PODPORU ŽIVOTA. U GERIATRICKÝCH, PEDIATRICKÝCH
NEBO JAKÝCHKOLI JINÝCH PACIENTŮ, KTEŘÍ NEDOKÁŽOU
INFORMOVAT O POTÍŽÍCH, MŮŽE BÝT ZAPOTŘEBÍ DALŠÍ
MONITOROVÁNÍ A/NEBO DISTRIBUOVANÝ ALARMOVÝ SYSTÉMPRO SDĚLENÍ INFORMACÍ O POTÍŽÍCH ANEBO KRITICKÉMZDRAVOTNÍM STAVU ODPOVĚDNÉ PEČUJÍCÍ OSOBĚ, ABY SEPŘEDEŠLO POŠKOZENÍ ZDRAVÍ. PACIENTI S POŠKOZENÝM
SLUCHEM A/NEBO ZRAKEM MOHOU POTŘEBOVAT POMOC PŘI
SLEDOVÁNÍ ALARMŮ.

VAROVÁNÍ: TĚHOTNÉ A KOJÍCÍ ŽENY NESMÍ POUŽÍVAT
PŘÍSLUŠENSTVÍ DOPORUČOVANÉ V TÉTO PŘÍRUČCE –PŘÍSLUŠENSTVÍ MŮŽE OBSAHOVAT FTALÁTY.
VAROVÁNÍ: V USA SMÍ BÝT PODLE FEDERÁLNÍCH ZÁKONŮ
TATO ZAŘÍZENÍ PRODÁVÁNA A PRONAJÍMÁNA POUZE LÉKAŘIČI JINÝMI LICENCOVANÝMI POSKYTOVATELI ZDRAVOTNÍ PÉČENEBO NA JEJICH PŘÍKAZ.

Kontraindikace pro použití
VAROVÁNÍ: ZA URČITÝCH OKOLNOSTÍ MŮŽE BÝTPOUŽITÍ NEPŘEDEPSANÉHO KYSLÍKU NEBEZPEČNÉ.
TOTO ZAŘÍZENÍ BY MĚLO BÝT POUŽÍVÁNO POUZETEHDY, POKUD BYLO PŘEDEPSÁNO LÉKAŘEM.
VAROVÁNÍ: NENÍ URČENO K POUŽITÍ V PŘÍTOMNOSTIHOŘLAVÝCH ANESTETIK.
VAROVÁNÍ: STEJNĚ JAKO U JINÝCH ELEKTRICKY NAPÁJENÝCH
ZAŘÍZENÍ MŮŽE DOJÍT K SITUACI, KDY PŘÍSTROJ NEBUDEMOCI FUNGOVAT KVŮLI PŘERUŠENÍ NAPÁJENÍ, PŘÍPADNĚ
BUDE NUTNÉ PROVÉST SERVIS PŘÍSTROJE COMPANION 5KVALIFIKOVANÝM TECHNIKEM. PŘÍSTROJ COMPANION 5NENÍ VHODNÝ PRO ŽÁDNÉHO PACIENTA, JENŽ BY BYL KVŮLI
TOMUTO DOČASNÉHO PŘERUŠENÍ VYSTAVEN NEPŘÍZNIVÝMZDRAVOTNÍM DŮSLEDKŮM.

Bezpečnostní pokyny
VAROVÁNÍ: NEJSOU POVOLENY ŽÁDNÉ ÚPRAVYTOHOTO ZAŘÍZENÍ.
VAROVÁNÍ: VÝROBCE DOPORUČUJE ALTERNATIVNÍ ZDROJDOPLŇKOVÉHO KYSLÍKU PRO PŘÍPADY VÝPADKU PROUDU
NEBO STAVU ALARMU ČI MECHANICKÉ ZÁVADY PŘÍSTROJE.
O TYPU POŽADOVANÉHO ZÁLOŽNÍHO SYSTÉMU SE PORAĎTE SESVÝM LÉKAŘEM NEBO DODAVATELEM PŘÍSTROJE.
VAROVÁNÍ: TENTO PŘÍSTROJ DODÁVÁ VYSOCE
KONCENTROVANÝ KYSLÍK, KTERÝ ZNAČNĚ PODPORUJEHOŘENÍ. VYVARUJTE SE KOUŘENÍ NEBO OTEVŘENÉHO OHNĚDO DVOU (2) METRŮ OD (1) TOHOTO PŘÍSTROJE NEBO (2)JAKÉHOKOLI PŘÍSLUŠENSTVÍ PŘENÁŠEJÍCÍHO KYSLÍK.
NEDODRŽENÍ TOHOTO VAROVÁNÍ MŮŽE VÉST K VÁŽNÉMU
POŽÁRU A ŠKODÁM NA MAJETKU A/NEBO ZPŮSOBIT ZRANĚNÍ
ČI SMRT.

PN 15062781-C4 J | Uživatelská příručka

227 – CZHCompanion 5
VAROVÁNÍ: NEPOUŽÍVEJTE JEDNOTKU V OMEZENÉM NEBO
STÍSNĚNÉM PROSTORU (TJ. V MALÉM POUZDRU NEBO RUČNÍTAŠCE), V NĚMŽ MŮŽE BÝT OMEZENO VĚTRÁNÍ. MOHLO BYDOJÍT K PŘEHŘÁTÍ KYSLÍKOVÉHO KONCENTRÁTORU A SNÍŽENÍJEHO VÝKONU.
VAROVÁNÍ: KONCENTRÁTOR BY MĚL BÝTUMÍSTĚN TAK, ABY BYL CHRÁNĚN PŘED KOUŘEM,ZNEČIŠTĚNÍM ČI VÝPARY.
VAROVÁNÍ: POUŽITÍ URČITÝCH PŘÍSLUŠENSTVÍ PRO VEDENÍ
KYSLÍKU, KTERÁ NEJSOU URČENA PRO TENTO KYSLÍKOVÝ
KONCENTRÁTOR, MŮŽE VÉST KE SNÍŽENÍ JEHO VÝKONU.
DOPORUČENÁ PŘÍSLUŠENSTVÍ JSOU UVEDENA V TÉTOPŘÍRUČCE.
VAROVÁNÍ: POKUD KYSLÍKOVÝ KONCENTRÁTOR SPADL, BYL
POŠKOZEN NEBO SE DOSTAL DO KONTAKTU S VODOU, OBRAŤTE
SE NA SVÉHO POSKYTOVATELE DOMÁCÍ ZDRAVOTNÍ PÉČE, ABY
PROVEDL KONTROLU NEBO PŘÍPADNOU OPRAVU ZAŘÍZENÍ.
NEPOUŽÍVEJTE KYSLÍKOVÝ KONCENTRÁTOR, POKUD MÁ
POŠKOZENÝ NAPÁJECÍ KABEL NEBO ZÁSTRČKU.
VAROVÁNÍ: ZABRAŇTE TOMU, ABY DOŠLO K ZABLOKOVÁNÍ
PŘÍVODU VZDUCHU A VÝSTUPU VYFUKOVANÉHO VZDUCHU.
NEUPUSŤTE ANI NEVKLÁDEJTE ŽÁDNÉ PŘEDMĚTY DO OTVORŮ
PŘÍSTROJE. MOHLO BY DOJÍT K PŘEHŘÁTÍ KYSLÍKOVÉHOKONCENTRÁTORU A SNÍŽENÍ JEHO VÝKONU.
VAROVÁNÍ: NEPŘEPLŇUJTE VOLITELNÝ ZVLHČOVAČ. NAPLŇTEVOLITELNÝ ZVLHČOVAČ VODOU POUZE NA ÚROVEŇ, KTEROUUKAZUJE VÝROBCE ZVLHČOVAČE.
VAROVÁNÍ: TENTO PŘÍSTROJ NEPOUŽÍVEJTE, DOKUD SINEPŘEČTETE A NEPOCHOPÍTE TUTO PŘÍRUČKU. POKUDNEROZUMÍTE VAROVÁNÍM A POKYNŮM, KONTAKTUJTEDODAVATELE ZAŘÍZENÍ, DŘÍVE NEŽ PŘÍSTROJ POUŽIJETE,JINAK MŮŽE DOJÍT KE ZRANĚNÍ NEBO ŠKODÁM.
VAROVÁNÍ: POKUD CÍTÍTE BOLESTI NEBO SE NACHÁZÍTE
V NALÉHAVÉM ZDRAVOTNÍM STAVU, OKAMŽITĚ VYHLEDEJTE
LÉKAŘSKOU POMOC.
VAROVÁNÍ: PROVOZ KYSLÍKOVÉHO KONCENTRÁTORU MIMOURČENÝ ROZSAH PROVOZNÍCH TEPLOT MŮŽE OMEZITSCHOPNOST KONCENTRÁTORU SPLNIT SPECIFIKACE
KYSLÍKOVÉHO KONCENTRÁTORU. MEZNÍ HODNOTY TEPLOTY
NAJDETE V ČÁSTI TÉTO PŘÍRUČKY VĚNOVANÉ SPECIFIKACÍM.
VAROVÁNÍ: NA PŘÍSLUŠENSTVÍ PŘENÁŠEJÍCÍ KYSLÍK A NAKYSLÍKOVÝ KONCENTRÁTOR NEPOUŽÍVEJTE OLEJE, MAZIVA,
MATERIÁLY NA BÁZI ROPY ANI JINÉ HOŘLAVÉ VÝROBKY.
POUŽÍVAT BY SE MĚLY POUZE ROZTOKY A BALZÁMY NA BÁZIVODY, KTERÉ JSOU KOMPATIBILNÍ S KYSLÍKEM. KYSLÍK
URYCHLUJE SPALOVÁNÍ HOŘLAVÝCH LÁTEK.

228 – CZH PN 15062781-C4 J | Uživatelská příručka

VAROVÁNÍ: KYSLÍKOVÝ KONCENTRÁTOR BY SE NEMĚL
POUŽÍVAT V BLÍZKOSTI JINÝCH ZAŘÍZENÍ ANI BÝT ULOŽEN
NA NICH ČI POD NIMI. POKUD SE TAKOVÉMU POUŽITÍ NELZEVYHNOUT, ZAŘÍZENÍ BY MĚLO BÝT SLEDOVÁNO, ABY SE OVĚŘILJEHO BĚŽNÝ PROVOZ.
VAROVÁNÍ: VÝROBCE DOPORUČUJE
ALTERNATIVNÍ ZDROJ DOPLŇKOVÉHO KYSLÍKU
PRO PŘÍPADY VÝPADKU PROUDU NEBO STAVU
ALARMU ČI MECHANICKÉ ZÁVADY PŘÍSTROJE.
O TYPU POŽADOVANÉHO ZÁLOŽNÍHO SYSTÉMUSE PORAĎTE SE SVÝM LÉKAŘEM NEBO
DODAVATELEM PŘÍSTROJE.
VAROVÁNÍ: TENTO PŘÍSTROJ DODÁVÁ VYSOCE
KONCENTROVANÝ KYSLÍK, KTERÝ ZNAČNĚ PODPORUJEHOŘENÍ. VYVARUJTE SE KOUŘENÍ NEBO OTEVŘENÝCH
PLAMENŮ VE STEJNÉ MÍSTNOSTI, KDE SE NACHÁZÍ (1) TENTO
PŘÍSTROJ NEBO (2) JAKÉKOLI PŘÍSLUŠENSTVÍ PŘENÁŠEJÍCÍ
KYSLÍK. NEDODRŽENÍ TOHOTO VAROVÁNÍ MŮŽE VÉSTK VÁŽNÉMU POŽÁRU A ŠKODÁM NA MAJETKU A/NEBOZPŮSOBIT ZRANĚNÍ ČI SMRT.
VAROVÁNÍ: POUŽITÍ URČITÝCH PŘÍSLUŠENSTVÍ PRO VEDENÍ
KYSLÍKU, KTERÁ NEJSOU URČENA PRO TENTO KYSLÍKOVÝ
KONCENTRÁTOR, MŮŽE VÉST KE SNÍŽENÍ JEHO VÝKONU.
DOPORUČENÁ PŘÍSLUŠENSTVÍ JSOU UVEDENA V TÉTOPŘÍRUČCE.
VAROVÁNÍ: POKUD KYSLÍKOVÝ KONCENTRÁTOR SPADL, BYL
POŠKOZEN NEBO SE DOSTAL DO KONTAKTU S VODOU, OBRAŤTE
SE NA SVÉHO POSKYTOVATELE DOMÁCÍ ZDRAVOTNÍ PÉČE, ABY
PROVEDL KONTROLU NEBO PŘÍPADNOU OPRAVU ZAŘÍZENÍ.
NEPOUŽÍVEJTE KYSLÍKOVÝ KONCENTRÁTOR, POKUD MÁ
POŠKOZENÝ NAPÁJECÍ KABEL NEBO ZÁSTRČKU.
VAROVÁNÍ: POUŽÍVEJTE POUZE NÁHRADNÍ SOUČÁSTKYDOPORUČENÉ VÝROBCEM, ABY BYLO ZAJIŠTĚNO ŘÁDNÉFUNGOVÁNÍ A ABY SE PŘEDEŠLO RIZIKU POŽÁRU
A POPÁLENIN.
VAROVÁNÍ: NEPROMAZÁVEJTE ARMATURY, PŘÍPOJE, HADIČKY
NEBO DALŠÍ PŘÍSLUŠENSTVÍ KYSLÍKOVÉHO KONCENTRÁTORU,ABY SE ZABRÁNILO NEBEZPEČÍ OHNĚ A POPÁLENIN.

Companion 5
VAROVÁNÍ: KOUŘENÍ PŘI POUŽÍVÁNÍ KYSLÍKU JE HLAVNÍ
PŘÍČINOU ZRANĚNÍ ZPŮSOBENÝCH OHNĚM A S TÍMSOUVISEJÍCÍCH ÚMRTÍ. MUSÍTE DODRŽOVAT NÁSLEDUJÍCÍ
BEZPEČNOSTNÍ VAROVÁNÍ:VAROVÁNÍ: VYVARUJTE SE KOUŘENÍ A POUŽITÍ SVÍČEK ČIOTEVŘENÉHO OHNĚ VE STEJNÉ MÍSTNOSTI, KDE SE NACHÁZÍ
PŘÍSTROJ NEBO PŘÍSLUŠENSTVÍ PŘENÁŠEJÍCÍ KYSLÍK.
VAROVÁNÍ: KOUŘENÍ PŘI NOŠENÍ KYSLÍKOVÉ KANYLY MŮŽEZPŮSOBIT POPÁLENINY OBLIČEJE A MOŽNÁ I SMRT.
VAROVÁNÍ: POKUD KOUŘÍTE, MOHOU VÁM NÁSLEDUJÍCÍ
3 KROKY ZACHRÁNIT ŽIVOT: VYPNĚTE KYSLÍKOVÝ
KONCENTRÁTOR, SEJMĚTE KANYLU A OPUSŤTE MÍSTNOST,V NÍŽ SE TENTO PŘÍSTROJ NACHÁZÍ.
VAROVÁNÍ: V DOMĚ NEBO NA JINÉM MÍSTĚ, KDE SE POUŽÍVÁ
KYSLÍK, MUSÍ BÝT NÁPADNĚ VYSTAVENY CEDULE „ZÁKAZ
KOUŘENÍ – POUŽÍVÁNÍ KYSLÍKU“. PACIENTI A JEJICH
PEČOVATELÉ MUSÍ BÝT INFORMOVÁNI O NEBEZPEČÍCH
KOUŘENÍ V PŘÍTOMNOSTI LÉKAŘSKÉHO KYSLÍKU NEBO PŘIJEHO POUŽÍVÁNÍ.
VAROVÁNÍ: NEPOUŽÍVEJTE KYSLÍKOVÝ KONCENTRÁTOR
V PŘÍTOMNOSTI HOŘLAVÝCH PLYNŮ. TO MŮŽE VÉSTK RYCHLÉMU HOŘENÍ, KTERÉ ZPŮSOBÍ ŠKODY NA MAJETKU,TĚLESNÁ ZRANĚNÍ NEBO SMRT.
VAROVÁNÍ: ODSTRANĚNÍ KANYLY A JEJÍ ULOŽENÍ NAOBLEČENÍ, POVLEČENÍ, POHOVKU NEBO JINÝ VÝPLŇOVÝ
MATERIÁL ZPŮSOBÍ PO KONTAKTU S CIGARETOU, ZDROJEMTEPLA NEBO PLAMENEM VÝBUCH OHNĚ.
VAROVÁNÍ: NOSNÍ KANYLU NENECHÁVEJTE NA OBLEČENÍ,POKRÝVCE POSTELE ČI POLSTROVÁNÍ KŘESLA ANI POD
NIMI. POKUD JE JEDNOTKA ZAPNUTÁ, ALE NEPOUŽÍVÁ SE,
KYSLÍK UČINÍ Z TĚCHTO MATERIÁLŮ HOŘLAVÉ LÁTKY. KDYŽ SE
KYSLÍKOVÝ KONCENTRÁTOR NEPOUŽÍVÁ, NASTAVTE HLAVNÍ
VYPÍNAČ I/0 DO POLOHY 0 (VYP.).

VAROVÁNÍ: POUŽÍVEJTE POUZE PŘÍSLUŠENSTVÍ DOPORUČENÉ
VÝROBCEM. POUŽITÍ JAKÉHOKOLI JINÉHO PŘÍSLUŠENSTVÍ
MŮŽE BÝT NEBEZPEČNÉ A MŮŽE ZPŮSOBIT VÁŽNÉ POŠKOZENÍ
VAŠEHO KYSLÍKOVÉHO KONCENTRÁTORU A POVEDEK ZNEPLATNĚNÍ ZÁRUKY.
UPOZORNĚNÍ: Před odpojením napájecího zdrojestřídavého proudu od kyslíkového koncentrátoru
vždy nejdříve odpojte napájecí zdroj od sítě.
UPOZORNĚNÍ: Přívodní hadice kyslíku a napájecí kabelyvždy umístěte takovým způsobem, aby se zabránilo riziku
zakopnutí.
UPOZORNĚNÍ: Abyste zabránili zneplatnění
záruky, dodržujte všechny pokyny výrobce.
Poznámka: Přenosná a mobilní radiofrekvenční komunikační zařízení
mohou mít vliv na zdravotnické elektrické přístroje.
VÝSTRAHA: VŽDY UMÍSTĚJTE DÁVKOVACÍ KYSELINY A DODÁVKY
KYSELINY V RÁMCI, KTERÝ ZABRÁŇU NEBEZPEČÍ VÝKONU NEBO
MOŽNÉHO NÁHODNÉHO ROZMNOŽENÍ.
VAROVÁNÍ: TENTO VÝROBEK VÁS MŮŽE VYSTAVIT CHEMICKÝM
LÁTKÁM, VČETNĚ NIKLU, KTERÝ JE VE STÁTĚ KALIFORNIE POVAŽOVÁN ZA KARCINOGEN. PODROBNĚJŠÍ INFORMACE NAJDETE NA
ADRESE WWW.P65WARNINGS.CA.GOV.
VAROVÁNÍ: V PŘÍPADĚ VÁŽNÉHO INCIDENTU S TÍMTO
ZAŘÍZENÍM BY MĚL UŽIVATEL OKAMŽITĚ NAHLÁSIT INCIDENTPOSKYTOVATELI A / NEBO VÝROBCI. VÁŽNÝ INCIDENT JE
DEFINOVÁN JAKO ZRANĚNÍ, SMRT NEBO POTENCIÁL ZPŮSOBITZRANĚNÍ / SMRT V PŘÍPADĚ OPAKOVANÉHO VÝSKYTU INCIDENTU. UŽIVATEL MŮŽE TAKÉ NAHLÁSIT INCIDENT PŘÍSLUŠNÉMU
ORGÁNU V ZEMI, KDE K INCIDENTU DOŠLO.

VAROVÁNÍ: NEPOKOUŠEJTE SE O ŽÁDNOU
ÚDRŽBU S VÝJIMKOU MOŽNÝCH ŘEŠENÍ
UVEDENÝCH V TÉTO PŘÍRUČCE. NESNÍMEJTEKRYTY. SNÍMAT KRYTY A PROVÁDĚT SERVISTOHOTO PŘÍSTROJE SMÍ POUZE DODAVATELZAŘÍZENÍ NEBO KVALIFIKOVANÝ SERVISNÍTECHNIK.

PN 15062781-C4 J | Uživatelská příručka

229 – CZHCompanion 5
Specifikace
Pro řádné používání přístroje poskytuje následující
tabulka důležité informace týkající se doporučeného
provozního prostředí nebo provozních podmínek.

SpecifikacePrůtokové
rychlosti
Koncentrace O2Rozměry
Hmotnost
Hladinaakustického tlakuPříkonProvozní teplota
Provozní vlhkost

0,5–5,0 l/min
90 % (+5,5 % / –3 %)54,6 x 31,8 x 34,2 cm (21,5 x 12,5 x 13,5 in)
16,3 kg (36 lb)
50 dB(A) při 2,0 l/min

285 W při 2 l/min, 350 W maximum
5 až 40 °C (41 až 104 °F)
15% - 95% při kondenzačním bodu 28 ° C(82,4 ° F)
Skladovací teplota –25 až 70 °C (–13 až 158 °F)
Skladovací vlhkost 0% - 95% bez kondenzace
Elektrické
Nepoužívejte prodlužovací kabely.
vybavení
Nepoužívejte elektrické zásuvky ovládané
spínačem.
Provozní–382–3011 metrů (–1253–9879 stop)
nadmořská výška
Provozní doba až 24 hodin denně
Očekávaná životnost zařízení je minimálně pět let.
VAROVÁNÍ: POUŽÍVÁNÍ ZAŘÍZENÍ MIMOSPECIFIKOVANÝCH PROVOZNÍCH PODMÍNEK
SE OČEKÁVÁ NEŽÁDOUCÍ ÚČINKY PRŮTOKU APERCENTU OXYGENU A NÁSLEDNĚ KVALITYTERAPIE.
Poznámka: Jestliže byl kyslíkový koncentrátor delší dobu skladován
mimo rozsah své normální provozní teploty, před zapnutím se musíjednotce umožnit, aby se vrátila k normální provozní teplotě. (Vizoddíl Specifikace v této příručce.)Správné umístění přístroje Companion 5

230 – CZH PN 15062781-C4 J | Uživatelská příručka

Vyberte pro přístroj místo, kde nebude docházetk nasávání kouře, výparů a znečišťujících látek.
Správné umístění přístroje musí umožnit přívod
vzduchu skrze tři místa přívodu vzduchu vzadu nahoře
a pod skříní přístroje a také umožnit, aby vyfukovaný
vzduch volně odcházel z otvoru vyfukovaného
vzduchu v levé dolní části přístroje.
Přívody vzduchu jsou umístěny na zadní straně nahoře
a na spodní straně koncentrátoru. Umístěte jednotku
tak, aby mezi koncentrátorem a jakoukoli stěnou,nábytkem, závěsy nebo jinými překážkami byl prostor
minimálně 30 cm (12 palců).
Umístěte přístroj tak, aby bylo slyšet alarmy.
Umístěte přívodní hadičku kyslíku tak, aby u nínedocházelo k zalomení nebo okluzi.
VAROVÁNÍ: NEPOUŽÍVEJTE KYSLÍKOVÝ
KONCENTRÁTOR V PŘÍTOMNOSTI HOŘLAVÝCH
PLYNŮ. TO MŮŽE VÉST K RYCHLÉMU HOŘENÍ,KTERÉ ZPŮSOBÍ ŠKODY NA MAJETKU, TĚLESNÁ
ZRANĚNÍ NEBO SMRT. NA PŘÍSLUŠENSTVÍ
PŘENÁŠEJÍCÍ KYSLÍK A NA KYSLÍKOVÝ
KONCENTRÁTOR NEPOUŽÍVEJTE OLEJE, MAZIVA,
MATERIÁLY NA BÁZI ROPY ANI JINÉ HOŘLAVÉ
VÝROBKY. POUŽÍVAT BY SE MĚLY POUZEROZTOKY A BALZÁMY NA BÁZI VODY, KTERÉ JSOU
KOMPATIBILNÍ S KYSLÍKEM. KYSLÍK URYCHLUJESPALOVÁNÍ HOŘLAVÝCH LÁTEK.
VAROVÁNÍ: TENTO PŘÍSTROJ DODÁVÁ VYSOCE
KONCENTROVANÝ KYSLÍK, KTERÝ ZNAČNĚ PODPORUJEHOŘENÍ. VYVARUJTE SE KOUŘENÍ NEBO OTEVŘENÝCH
PLAMENŮ VE STEJNÉ MÍSTNOSTI, KDE SE NACHÁZÍ (1) TENTO
PŘÍSTROJ NEBO (2) JAKÉKOLI PŘÍSLUŠENSTVÍ PŘENÁŠEJÍCÍ
KYSLÍK. NEDODRŽENÍ TOHOTO VAROVÁNÍ MŮŽE VÉSTK VÁŽNÉMU POŽÁRU A ŠKODÁM NA MAJETKU A/NEBOZPŮSOBIT ZRANĚNÍ ČI SMRT.

Companion 5
Provozní pokynyPřed provozem

Tato uživatelská příručka vám slouží k vyhledávání
informací, které vám pomáhají při obsluze a údržbě
přístroje. Máte-li jakékoli dotazy nebo obavy, zavolejtesvému poskytovateli domácí zdravotní péče.
Důležité! NEPOKOUŠEJTE SE používat přístroj
Companion 5, dokud si nepřečtete oddíl této příručky
s názvem Bezpečnostní pokyny. Dodržujte prosím
všechny provozní pokyny. Přečtěte si prosím všechna
varování na přístroji i v uživatelské příručce. Aby sesnížilo nebezpečí požáru, poranění osob a vážného
poškození přístroje Companion 5, dodržujte prosím
všechna preventivní bezpečnostní opatření.
Přístroj CAIRE Companion 5 nevytváří svůj vlastní
kyslík. Produkuje vysoce koncentrovaný kyslík
z okolního vzduchu, který vám poté dodává. Vzduch
v místnosti v přirozeném stavu obsahuje přibližně
pouze 21 procent kyslíku.
Vzduch v místnosti je do koncentrátoru nasáván
kompresorem. Vzduch pak prochází sérií filtrů
a vrstvou materiálu zvaného molekulové síto, který
chemicky absorbuje molekuly dusíku. Výsledkemje, že vzduch obohacený kyslíkem, který vycházíz koncentrátoru, obsahuje 87–95 procent kyslíku.
VAROVÁNÍ: SKŘÍŇ PŘÍSTROJE, OVLÁDACÍ PANEL
A NAPÁJECÍ KABEL OČISTĚTE POUZE POMOCÍJEMNÉHO DOMÁCÍHO ČISTICÍHO PROSTŘEDKU
NANESENÉHO VLHKÝM (NIKOLI MOKRÝM)HADŘÍKEM NEBO HOUBIČKOU A NÁSLEDNĚ
VŠECHNY POVRCHY VYTŘETE DOSUCHA.
ZABRAŇTE VNIKNUTÍ JAKÉKOLI KAPALINYDOVNITŘ PŘÍSTROJE. VĚNUJTE ZVLÁŠTNÍ
POZORNOST VÝSTUPU KYSLÍKU URČENÉMU
K PŘIPOJENÍ KANYLY, UJISTĚTE SE, ŽE NA NĚMNENÍ ZACHYCEN PRACH, VODA ANI JINÉ ČÁSTICE.
UPOZORNĚNÍ: Zabraňte tomu, aby došlok zablokování přívodu vzduchu a výstupu
vyfukovaného vzduchu. Mohlo by dojít k přehřátí
kyslíkového koncentrátoru a snížení jehovýkonu. Nepoužívejte kyslíkový koncentrátor
bez nainstalovaného filtru na přívodu vzduchu.
Je-li k dispozici druhý filtr, vložte do přístroje„náhradní“ filtr před vyčištěním zaneseného
filtru. Zanesený filtr vyčistěte v teplé mýdlovévodě a před použitím jej řádně vysušte.

Krok 1: Umístění vašeho přístroje Companion 5 pro použití
Umístěte přístroj Companion 5 na dobře větraném
a dobře osvětleném místě. Ujistěte se, že přívody
vzduchu a otvory vyfukovaného vzduchu nejsou
blokovány. Umístěte přístroj Companion 5 tak, abyvšechny zvukové a vizuální ukazatele nebo alarmy
byly snadno vidět a slyšet.
Zapojte přístroj do zásuvky střídavého proudu.
Krok 2: Jestliže se láhev zvlhčovače NEBUDE používat

MístopřívoduvzduchuMístopřívoduchladicího
vzduchu

Otvor
vyfukovaného
vzduchuMísto přívodu
vzduchua. Připojte nosní kanylu nebo kyslíkovou hadičku
přímo k výstupu kyslíku.

b. Přejděte ke kroku 4.
Krok 3: Jestliže se láhev zvlhčovače BUDE používat
a. Odšroubujte víko láhve zvlhčovače.
b. Naplňte láhev s použitím destilované vody. Ujistěte
se, že hladina vody je mezi maximem a minimem
stanovenými výrobcem a označenými na láhvi.
c. Připevněte víko láhve zvlhčovače. Ujistěte se, ženikde nejsou netěsnosti.
d. Připojte nosní kanylu nebo standardní kyslíkovou
hadičku k výstupu na láhvi zvlhčovače.
Poznámka: Použití zvlhčovače, který není určen pro použití s tímto
koncentrátorem, může negativně ovlivnit jeho výkon.

PN 15062781-C4 J | Uživatelská příručka

231 – CZHCompanion 5
g. Umístěte láhev na její určené místo na podstavci pro
láhev zvlhčovače.
h. Láhev na podstavci zajistěte gumičkou.

e. Připojte závitový konec hadičky zvlhčovače k víku
láhve zvlhčovače.

i. Přejděte ke kroku 4.
Krok 4: Zapnutí a zahřívání
a. Zapojte elektrický kabel přístroje CAIRE
Companion 5 do zásuvky střídavého proudu.
VAROVÁNÍ: PŘED POUŽITÍM ELEKTRICKÝ KABELZKONTROLUJTE, ZDA NENÍ POŠKOZENÝ. POKUDJE KABEL POŠKOZENÝ, NEZAPOJUJTE HO DO
ELEKTRICKÉ ZÁSUVKY ANI SE NEPOKOUŠEJTE
POUŽÍVAT KONCENTRÁTOR.

f. Připojte druhý konec hadičky zvlhčovače k výstupu
na přístroji CAIRE Companion 5.

b. Přepněte hlavní vypínač do polohy „ZAP.“ (I). Kdyžje přístroj Companion 5 řádně zapnutý, rozsvítí sena LED displeji zelený indikátor. Po spuštění serozsvítí všechny LED kontrolky. Když koncentrátor
dokončí cyklus zahřívání, zůstane rozsvícená pouze
zelená kontrolka.
Poznámka: Po počátečním zapnutí přístroje počkejte prosím až
deset (10) minut, aby přístroj mohl dosáhnout svých specifikacívýkonu.

232 – CZH PN 15062781-C4 J | Uživatelská příručka

Companion 5
VAROVÁNÍ: JE VELMI DŮLEŽITÉ ZVOLIT POUZEPŘEDEPSANOU ÚROVEŇ KYSLÍKU. NEMĚŇTE
ÚROVEŇ PRŮTOKU, POKUD VÁM TO NEBYLO
NAŘÍZENO OŠETŘUJÍCÍM LÉKAŘEM. PODLEDOPORUČENÍ KVALIFIKOVANÉHO LÉKAŘE LZEKYSLÍKOVÝ KONCENTRÁTOR POUŽÍVAT I BĚHEM
SPÁNKU.
Krok 6: Zkontrolujte průtok a normálně dýchejte

Krok 5: Nastavení průtokové rychlosti
a. Otočte regulačním knoflíkem průtoku na průtokovou
rychlost kyslíku (l/min) předepsanou vašímlékařem.

a. Zkontrolujte, zda nosní kanylou skutečně proudí
kyslík a že hadička není nikde zalomená, ohnutá
nebo zablokovaná.
Poznámka: Přesvědčte se, že je kanyla zcela zasunutá a zajištěná.
Měli byste slyšet nebo cítit kyslík proudící do hrotů nosní kanyly.
b. Nosní kanylu řádně umístěte a normálně dýchejte
nosní kanylou.
Krok 7: Vypnutía. Když se přístroj CAIRE Companion 5 už nepoužívá,přepněte hlavní vypínač do polohy „VYP.“ (O).
UPOZORNĚNÍ: Používejte pouze příslušenství
doporučené výrobcem. Použití jakéhokoli jinéhopříslušenství může být nebezpečné a můžezpůsobit vážné poškození vašeho kyslíkového
koncentrátoru a povede k zneplatnění záruky.

Nastavení průtokové rychlosti:
Otáčením proti směru hodinových ručiček se průtok
zvyšuje.
Otáčením ve směru hodinových ručiček se průtok snižuje.
Střed kuličky označuje průtokovou rychlost. Nížeuvedený obrázek ukazuje průtokovou rychlost 4,0 l/min.

UPOZORNĚNÍ: V kombinaci s tímto výrobkem nepoužívejte
prodlužovací kabely ani nezapojujte příliš mnoho zástrček
do stejné elektrické zásuvky. Použití prodlužovacích kabelů
nepříznivě ovlivňuje výkon zařízení. Příliš mnoho zástrček
v jedné zásuvce může vést k přetížení elektrického panelu, cožzpůsobí aktivaci jističe/pojistky nebo dokonce vznícení, pokud
se jistič/pojistka neuvede do provozu.
UPOZORNĚNÍ: Použití jiných než k tomu určených kabelů
a adaptérů, s výjimkou kabelů a adaptérů prodávaných
výrobcem zdravotnického elektrického přístroje jako náhradnídíly za vnitřní součásti, může vést ke snížení odolnosti
kyslíkového koncentrátoru.

Správně

PN 15062781-C4 J | Uživatelská příručka

233 – CZHCompanion 5
VAROVÁNÍ: ZABRAŇTE TOMU, ABY DOŠLOK ZABLOKOVÁNÍ PŘÍVODU VZDUCHU A VÝSTUPUVYFUKOVANÉHO VZDUCHU. MOHLO BY DOJÍT
K PŘEHŘÁTÍ KYSLÍKOVÉHO KONCENTRÁTORUA SNÍŽENÍ JEHO VÝKONU.
Poznámka: Poskytovateli zařízení: K použití s Companion 5 sedoporučuje následující příslušenství pro podávání kyslíku:• Nasální kanyla: CAIRE Číslo dílu CU002-1• Hadice adaptéru zvlhčovače: CAIRE Číslo dílu 20843882• Zvlhčovací láhev: CAIRE Číslo dílu HU003-1• Firebreak: CAIRE Číslo dílu 20629671
Pro použití s jakoukoli kanylou se vyžaduje požární štěrk.
• CAIRE nabízí požár určený k použití ve spojení s kyslíkovým
koncentrátorem. Firebreak je tepelná pojistka, která zastaví tok
plynu v případě, že se zapálí kanyla nebo kyslíková trubka po
proudu a spálí se do firebreak. Je umístěna in-line s nosní kanylou
nebo kyslíkovou hadičkou mezi pacientem a výstupem kyslíku zCompanion 5. Pro správné použití firebreaku se vždy řiďte pokynyvýrobce (součástí každé sady firebreak).
• Další doporučené příslušenství najdete v katalogu příslušenství(PN MLLOX0010), který je k dispozici na adrese www.caireinc.com.

234 – CZH PN 15062781-C4 J | Uživatelská příručka

Companion 5
Stavy alarmů
Váš přístroj CAIRE Companion 5 používá kombinaci
dvou nebo tří (pro modely s OCSI) LED kontrolek
a zvukového alarmu, aby vás upozornil, že došlo
k poruše koncentrátoru.
1) Zelená LED kontrolka (provoz.

) – Označuje normální2) Červená LED kontrolka ( ) – Označuje
systémovou poruchu. Podrobnosti najdete v nížeuvedené tabulce Stavy alarmů.

Zelená

ČervenáŽlutá3) Žlutá LED kontrolka () – Označuje nízkou
koncentraci kyslíku. Podrobnosti najdete v nížeuvedené tabulce Stavy alarmů.
Tabulka Stavy alarmů na následující stránce ukazujerůzné stavy alarmů, které může koncentrátor zobrazit.
Stručně popisuje možné příčiny a opatření, která bysteměli učinit, pokud dojde k určitému alarmu. V případě
jakéhokoli alarmu doporučujeme, abyste co nejdříve
kontaktovali svého poskytovatele zdravotní péče.

Nahoře: Companion 5

VAROVÁNÍ: NEIGNORUJTE ALARMYVAROVÁNÍ: VÝROBCE DOPORUČUJE
ALTERNATIVNÍ ZDROJ DOPLŇKOVÉHO KYSLÍKU
PRO PŘÍPADY VÝPADKU PROUDU NEBO STAVU
ALARMU ČI MECHANICKÉ ZÁVADY PŘÍSTROJE.
O TYPU POŽADOVANÉHO ZÁLOŽNÍHO SYSTÉMUSE PORAĎTE SE SVÝM LÉKAŘEM NEBO
DODAVATELEM PŘÍSTROJE.

PN 15062781-C4 J | Uživatelská příručka

235 – CZHCompanion 5
Tabulka Stavy alarmů
Zvukový
alarm

BarevnáLED
kontrolkaPříčinaOpatřeníŽádnéVypnutý

Koncentrátor pracuje správně.
Zelená

PřerušovanéČERVENÁ

Přerušované

Koncentrátor není napájen, alehlavní vypínač je stále v poloze„ZAP.“.

Zkontrolujte, zda je elektrický kabel koncentrátoru zapojený do
zásuvky a že v zásuvce je proud. Zkuste jinou zásuvku. Pokud
problém přetrvává, použijte záložní zdroj kyslíku a kontaktujte svého
poskytovatele zdravotní péče.

Systémová porucha

Ihned přepněte hlavní vypínač do polohy „VYP.“. Odpojte napájení
střídavým proudem z nástěnné zásuvky. Počkejte 5 minut. Zapojte
napájení střídavým proudem zpět do nástěnné zásuvky a znovu zapněte
koncentrátor. Pokud alarm dále trvá, je nutný servis. Použijte záložní
zdroj kyslíku a ihned kontaktujte svého poskytovatele zdravotní péče.

Průtoková rychlost výrobku jepříliš vysoká nebo příliš nízká

1. Ujistěte se, že kanyla není zalomená nebo zablokovaná. Pokud sepřístroj používá s láhví zvlhčovače, ujistěte se, že je správně naplněná
a nezpůsobuje zablokování.
2. Ujistěte se, že přístroj Companion 5 má řádné větrání. Musí být
minimálně 30 cm (12 palců) od jakéhokoli povrchu, aby se zajistilo, že
otvory nebudou blokovány.
3. Pokud problém přetrvává, použijte náhradní zdroj kyslíku
a požádejte o pomoc svého poskytovatele zdravotní péče.

ČERVENÁ

PřerušovanéČERVENÁ

Přístroj Companion 5 zjistil nízké
hladiny kyslíku.

PřerušovanéŽlutáPřístroj Companion 5 zjistil nízké
hladiny kyslíku.

PřerušovanéŽlutá

236 – CZH PN 15062781-C4 J | Uživatelská příručka

1. Ujistěte se, že filtry na přívodu vzduchu a místa vyfukovaného
vzduchu nejsou ucpané nebo jinak omezené. 2. Zajistěte, aby bylpřístroj Companion 5 na dobře větraném místě. Ujistěte se, že prostor
mezi zadní stranou a bočními stranami přístroje Companion a jakoukolipřekážkou (nábytek, závěsy atd.) je minimálně 30 cm (12 palců).
3. Pokud stav přetrvává, použijte náhradní zdroj kyslíku a ihned
kontaktujte svého poskytovatele zdravotní péče.

Companion 5
Odstraňování problémů
Jestliže váš koncentrátor zobrazí jakékoli alarmynebo jestliže cítíte, že váš koncentrátor nepracuje
správně, doporučujeme, abyste kontaktovali svého
poskytovatele zdravotní péče. Koncentrátor se
nepokoušejte sami opravit. Opravu může provést
pouze vyškolený servisní technik. Existují však některé

kroky pro řešení problémů, které můžete podniknout,
pokud se vyskytnou problémy s vaším přístrojem
CAIRE Companion 5. Jsou uvedeny v následující
tabulce. Tabulku můžete použít tak, že budete postupně
provádět kontroly související s vaším problém podlečíselného pořadí.

Tabulka Odstraňování problémů pro uživatele
Problém

Možná příčina

Opatření

1. Elektrický kabel není zapojený do zásuvky.

1. Zapojte elektrický kabel.

Koncentrátor
je zapnutý, alenepracuje.

2. Elektrická zásuvka nedodává elektrický
proud.

2. Zkontrolujte jističe a obvod. Zkuste jinou zásuvku.

Nízký nebo žádný
průtok kyslíku.

3. Interní porucha.

3. Použijte náhradní zdroj kyslíku. Ihned kontaktujte svého
poskytovatele zdravotní péče.

1. Bylo změněno nastavení průtoku.

1. Zkontrolujte, zda je regulační knoflík průtoku nastaven na vašipříslušnou hodnotu nastavení litrů za minutu (l/min) předepsanou
vaším lékařem.

2. Odpojila se hadička.

2. Zkontrolujte, zda je hadička pevně připojena na všech spojích(výstup, láhev zvlhčovače, odlučovač vody atd.). Podle potřeby znovu
zajistěte všechny spoje.

3. Kyslíková hadička nebo kanyla je zalomená.

3. Zkontrolujte, zda na hadičce nejsou zalomená nebo zablokovaná
místa. V případě potřeby je opravte.

4. Na kanyle nebo hadičce je netěsnost.

4. Zkontrolujte všechny hadičky a kanylu, zda na nich nejsounetěsnosti. To můžete udělat tak, že unikající vzduch lokalizujete buďposlechem, nebo hmatem. Netěsnící hadičku nebo kanylu vyměňte.

5. Něco brání proudění vzduchu dokoncentrátoru.

5. Zkontrolujte, že nic neblokuje přívody koncentrátoru. Ujistěte se, že
koncentrátor je minimálně 30 cm (12 palců) od nábytku, stěn nebojiných překážek.

6. Interní porucha.

6. Použijte náhradní zdroj kyslíku. Ihned kontaktujte svého
poskytovatele zdravotní péče.

Poznámka: Jestliže váš přístroj Companion 5 stále vysílá alarm i po použití řešení v tabulkách Stavy alarmů a Odstraňování problémů, kontaktujte
prosím technický servis na čísle +1 800 482 2473 a nahlaste kód alarmu zobrazeným na LCD displeji.

PN 15062781-C4 J | Uživatelská příručka

237 – CZHCompanion 5Čištění, péče a pravidelná údržba
Pravidelná údržba

Pravidelná údržba sestává z výměny filtru hrubých
částic umístěného v rukojeti, která se provádí jednoutýdně (je-li nainstalován).

Péče a čištění přístroje prováděné uživatelemVýměna kanylyPřívodní hadičku a kanylu vyměňujte pravidelně,jak doporučí váš poskytovatel domácí zdravotnípéče. Informace o čištění, dezinfekci a výměně vám
poskytne váš lékař nebo poskytovatel domácí zdravotní
péče.
Poznámka: Vždy postupujte podle pokynů ke správnému použití odvýrobce kanyl. Vyměňujte jednorázovou kanylu podle doporučenívýrobce kanyl nebo vašeho dodavatele přístroje. Další produkty jsou
k dostání u vašeho dodavatele přístroje.

Filtr na přívodu vzduchu (vložka)Filtr na přívodu vzduchu může kontrolovat a měnit
pouze váš poskytovatel zdravotní péče.
Filtr hrubých částic (volitelné vybavení):

Filtr hrubých částic je umístěn na samém předním
spodku Companion 5.

FiltrhrubýchčásticMěli byste ji jednou týdně zkontrolovat a vyčistit.
K čištění filtru použijte následující postup:1) Vyjměte filtr hrubých částic.
2) Vyperte filtr v teplé vodě z vodovodu s použitím
roztoku jemného mýdlového čisticího prostředku.
3) Filtr důkladně vymáchejte v čisté vodě z vodovodu
a přebytečnou vodu vymačkejte.
4) Nechte filtr uschnout na vzduchu.
5) Vložte filtr zpět do skříně přístroje.

238 – CZH PN 15062781-C4 J | Uživatelská příručka

Poznámka: Výrobce nedoporučuje sterilizaci tohoto zařízení.
Poznámka: Nepoužívejte kyslíkový koncentrátor bez
nainstalovaného filtru na přívodu vzduchu.
Láhev zvlhčovače (je-li používána)
Láhev zvlhčovače musíte kontrolovat každý den,abyste se ujistili, že hladina vody v láhvi je mezi
stanoveným maximem a minimem. Podle potřeby
láhev znovu naplňte s použitím destilované vody.
Láhev zvlhčovače čistěte podle pokynů výrobce.
Postup čištění:Před jakýmkoli čištěním nebo dezinfekcí přístroj
Companion 5 VYPNĚTE a odpojte od napájení
střídavým proudem. Na vnější kryt přímo
NESTŘÍKEJTE. Používejte vlhký (nikoli mokrý)hadřík nebo houbičku. Za účelem vyčištění krytu
a napájecího kabelu postříkejte hadřík nebo
houbičku roztokem jemného čisticího prostředku.
K dezinfikování přístroje Companion 5 použijte
dezinfekční prostředek Lysol® Brand II (nebo jeho
ekvivalent). Čisticí přípravek nařeďte podle pokynů
výrobce, ale nestříkejte kapalinu přímo na přístroj
Companion 5.
VAROVÁNÍ: NEBEZPEČÍ ÚRAZU ELEKTRICKÝMPROUDEM. PŘED ČIŠTĚNÍM PŘÍSTROJ VYPNĚTEA ODPOJTE NAPÁJECÍ KABEL ZE SÍŤOVÉ
ZÁSUVKY, ABYSTE ZABRÁNILI NEBEZPEČÍ
NÁHODNÉHO ÚRAZU ELEKTRICKÝM PROUDEM
A VZNIKU POPÁLENIN. ODSTRAŇOVAT KRYTYNEBO PROVÁDĚT ÚDRŽBU JEDNOTKY BY MĚLPOUZE VÁŠ DODAVATEL PŘÍSTROJE NEBOKVALIFIKOVANÝ SERVISNÍ TECHNIK.
VAROVÁNÍ: NEBEZPEČÍ ÚRAZU ELEKTRICKÝM PROUDEM. PŘEDČIŠTĚNÍM PŘÍSTROJ VYPNĚTE A ODPOJTE NAPÁJECÍ KABEL ZE
SÍŤOVÉ ZÁSUVKY, ABYSTE ZABRÁNILI NEBEZPEČÍ NÁHODNÉHOÚRAZU ELEKTRICKÝM PROUDEM A VZNIKU POPÁLENIN.
ODSTRAŇOVAT KRYTY NEBO PROVÁDĚT ÚDRŽBU JEDNOTKY BY
MĚL POUZE VÁŠ DODAVATEL PŘÍSTROJE NEBO KVALIFIKOVANÝSERVISNÍ TECHNIK.

Companion 5
VAROVÁNÍ: NEBEZPEČÍ ÚRAZU ELEKTRICKÝM PROUDEM.
PŘED ČIŠTĚNÍM ODPOJTE NAPÁJECÍ KABEL ZE SÍŤOVÉ
ZÁSUVKY, ABYSTE ZABRÁNILI NEBEZPEČÍ NÁHODNÉHOÚRAZU ELEKTRICKÝM PROUDEM A VZNIKU POPÁLENIN.
ODSTRAŇOVAT KRYTY NEBO PROVÁDĚT ÚDRŽBU JEDNOTKY BY
MĚL POUZE VÁŠ DODAVATEL PŘÍSTROJE NEBO KVALIFIKOVANÝSERVISNÍ TECHNIK.
VAROVÁNÍ: JE TŘEBA DÁVAT POZOR, ABY SE ZABRÁNILO
NAMOČENÍ KYSLÍKOVÉHO KONCENTRÁTORU NEBO
PRONIKNUTÍ KAPALIN DO JEDNOTKY. MOHLO BY TO VÉSTK NESPRÁVNÉMU FUNGOVÁNÍ NEBO ZASTAVENÍ PŘÍSTROJE
A ZVÝŠIT NEBEZPEČÍ ÚRAZU ELEKTRICKÝM PROUDEM NEBOVZNIKU POPÁLENIN.
VAROVÁNÍ: NESMÍ BÝT NEBEZPEČNÉ PROPOJENÍ SPOLEČNOSTI
5 S JINÝMI ZAŘÍZENÍMI.

VAROVÁNÍ: SKŘÍŇ PŘÍSTROJE, OVLÁDACÍ PANEL A NAPÁJECÍKABEL OČISTĚTE POUZE POMOCÍ JEMNÉHO DOMÁCÍHO
ČISTICÍHO PROSTŘEDKU NANESENÉHO VLHKÝM (NIKOLIMOKRÝM) HADŘÍKEM NEBO HOUBIČKOU A NÁSLEDNĚ
VŠECHNY POVRCHY VYTŘETE DOSUCHA. ZABRAŇTE VNIKNUTÍJAKÉKOLI KAPALINY DOVNITŘ PŘÍSTROJE. VĚNUJTE ZVLÁŠTNÍ
POZORNOST VÝSTUPU KYSLÍKU URČENÉMU K PŘIPOJENÍ
KANYLY, UJISTĚTE SE, ŽE NA NĚM NENÍ ZACHYCEN PRACH,VODA ANI JINÉ ČÁSTICE.
VAROVÁNÍ: NEPOUŽÍVEJTE KAPALINY PŘÍMO NA JEDNOTCE.
SEZNAM NEŽÁDOUCÍCH CHEMICKÝCH LÁTEK ZAHRNUJEMIMO JINÉ TYTO LÁTKY: ALKOHOL A PRODUKTY NA BÁZI
ALKOHOLU, KONCENTROVANÉ PRODUKTY NA BÁZI CHLORU(ETHYLENCHLORID) A PRODUKTY NA BÁZI OLEJE (PINE-SOL®,LESTOIL®). TYTO LÁTKY SE NESMÍ POUŽÍVAT K ČIŠTĚNÍ
PLASTOVÝCH KRYTŮ NA KYSLÍKOVÉM KONCENTRÁTORU,
PROTOŽE MOHOU POŠKODIT PLAST JEDNOTKY.

LikvidaceZa účelem řádné likvidace vždy vraťte zásobník
Companion 5, včetně všech součástí, poskytovateli
domácí zdravotní péče. Můžete také kontaktovat místní
městský úřad a požádat o pokyny k řádné likvidaci
baterie.

Směrnice OEEZ a RoHS

Naše výrobky odpovídají směrnici o omezenípoužívání některých nebezpečných látek v elektrických
a elektronických zařízeních (RoHS). Neobsahují vícenež stopové množství olova ani jiných nebezpečných
materiálů.
UPOZORNĚNÍ: Pro správnou likvidaci kontaktujte svého
dodavatele zařízení nebo místní vládní agenturu, kde
získáte pokyny k likvidaci.

Tento symbol má majitelům zařízení připomenout, že jej mají podle směrnice o odpadních elektrických a elektronických zařízeních (OEEZ) na konciživotnosti odevzdat sběrnému recyklačnímu zařízení.

CAIRE a CAIRE Inc. jsou registrované ochranné známky společnosti CAIRE Inc. Úplný seznam ochranných známek naleznete na nížeuvedené webové stránce.
Ochranné známky: www.careinc.com/corporate/trademarks.
Copyright © 2020 CAIRE Inc. CAIRE Inc. si vyhrazuje právo
kdykoli bez předchozího upozornění a bez dalších závazků přerušitvýrobu svých výrobků nebo měnit ceny, materiály, vybavení, kvalitu,
popisy, specifikace a / nebo procesy svých produktů bez předchozího
upozornění. nebo následek. Všechna práva, která zde nejsouvýslovně uvedena, si vyhrazujeme, pokud to bude možné.
PN 15062781-C4 J | Uživatelská příručka

239 – CZHCompanion 5
Elementy sterowania i wskaźniki stanu systemu
Symbole wewnętrzne

02Wyjście tlenu

I/OPrzełącznik Wł./Wył.
Wewnątrz nie ma części podlegających
naprawie. Nie otwierać pokrywy.

ISO 7000: Symbole graficzne stosowane w urządzeniach —indeks i streszczenie
Zakres temperatur przechowywania lub
pracy. Reg. # 0632

Część klasy BF (stopień ochrony przedporażeniem elektrycznym). Reg. # 5333
Dyrektywa Rady 93/42/EWG dotycząca wyrobów medycznych
Autoryzowany przedstawiciel
w Europie
IEC 60417
Urządzenie klasy II, z podwójną
izolacją Reg. # 5172
Certyfikat ETL

Zakres wilgotności podczas
przechowywania. Reg. # 2620

Agencja bezpieczeństwa
CAN/CSA C22.2 No. 601.1 M90 dla
elektrycznych urządzeń medycznych.

Nazwa i adres producenta. Reg. # 3082
Data produkcji. Reg. # 2497
Numer katalogowy. Reg. # 2493
Numer seryjny. Reg. # 2498
Góra. Reg. # 0623

Informacja FCC
21 CFR 801.15: Kodeks przepisów federalnych, tytuł 21
Prawo federalne dopuszcza sprzedaż
tego urządzenia wyłącznie przez
lekarza lub na jego zlecenie.
Dyrektywa Rady 2012/19/UE w sprawie zużytego sprzętu
elektrycznego i elektronicznego (WEEE)

Ostrożnie. Reg. # 0621
Chronić przed deszczem.
Przechowywać w suchym miejscu.
Reg. # 0626
Przed rozpoczęciem użytkowania
przeczytać instrukcję obsługi.
Reg. # 1641
Limit układania według liczb. Reg. #2403.
ISO 7010: Symbole graficzne — kolory i znaki ostrzegawcze —zarejestrowane znaki ostrzegawcze
Przeczytać instrukcję obsługi.
Reg. # M002
Ostrzeżenie. Reg. # W001
Przechowywać z dala od nieosłoniętego
płomienia, ognia i źródeł iskrzenia.
Chronić przed źródłem zapłonu. Zakaz
palenia. Reg. # P003
Nie palić w pobliżu urządzenia ani
podczas jego obsługi. Reg. # P002

240 - POLPN 15062781-C4 J | Instrukcja obsługi

WEEE
IEC 60601-1: Medyczne urządzenia elektryczne —część 1: Wymagania ogólne dotyczące bezpieczeństwa
podstawowego oraz funkcjonowania zasadniczego

IPX1Sprzęt kroploszczelny — IPX1:
Companion 5 zapewnia ochronę przed
szkodliwym wpływem wnikania
płynów. (IPX1, zgodnie z IEC 60529)

INFORMACJA FCC:

To urządzenie może zawierać CYBLE-022001-00,łącznie z anteną 2450AT18B100 firmy Johnson
Technology, spełnia wymagania par. 15 przepisów
FCC. Urządzenie spełnia wymagania dotyczące
homologacji nadajnika modułowego zgodnie
z informacją publiczną FCC DA00-1407. Działanie
nadajnika podlega następującym dwóm warunkom(1): Urządzenie nie może powodować szkodliwych
zakłóceń oraz (2) Urządzenie musi przyjmować
wszelkie odbierane zakłócenia, w tym zakłócenia,
które mogą powodować niepożądane działanie.
Ten produkt może być chroniony jednym lub wieloma
patentami w Stanach Zjednoczonych i innych
krajach. Odpowiednie patenty wyszczególnione sąw naszej witrynie pod adresem podanym poniżej.
Pat.: www.caireinc.com/corporate/patents/.

Companion 5
Skrócona instrukcja obsługi
1 Rozpakowanie urządzenia Companion 5

Companion 5

2 Poznawanie urządzenia Companion 5
Zapoznać się ze wszystkimi ostrzeżeniami,
przestrogami i dodatkowymi informacjami
o urządzeniu zawartymi w pozostałej części niniejszej
instrukcji. Poniżej opisano najważniejsze funkcjeurządzenia Companion 5 oraz panel sterowaniaużytkownika.
PRZÓDObudowazewnętrzna

Przełącznik
zasilaniaWyświetlacz
LED

Tlen
Króciec
wylotowy
tlenu

Licznik
godzinPrzepływomierz
Stojak butli
nawilżacza
(butla jest
wyposażeniem
opcjonalnym)

TYŁ
Uchwyti wlotpowietrza

Wlot
powietrza
chłodzącego

Wlot
powietrza

PN 15062781-C4 J | Instrukcja obsługi

Wylot
powietrza

241 - POL

Companion 5
Komponenty urządzenia Companion 5
Obudowazewnętrzna

Wytrzymała plastikowa obudowa, która chroniwewnętrzne elementy urządzenia CAIRE
Companion 5.
Przełącznik zasilania Przełącznik zasilania służący do włączania
lub wyłączania zasilania urządzenia. Pozycjawłączenia jest oznaczona symbolem „l”, a pozycjawyłączenia jest oznaczona symbolem „O”.
Króciec wylotowy W tym miejscu tlen wydostaje się z urządzenia
tlenuCompanion 5 po tym, jak zostaje przefiltrowany
i skoncentrowany. W celu dostarczenia gopacjentowi dołączone są pojedyncze wąsytlenowe lub standardowe przewody tlenowe.
Maksymalna długość przewodów tlenowychpodłączonych do koncentratora wynosi 15,2 m(50 stóp).
Wyświetlacz LED Mieści zieloną, czerwoną i żółtą* lampkę LED.
Zielona lampka wskazuje normalne działanie,a żółta* i czerwona – stany alarmowe. (Więcej
informacji na ten temat znajduje się w sekcji„Stany alarmowe” w instrukcji).
Licznik godzin
Cyfrowy wyświetlacz wskazujący
dotychczasowy czas pracy koncentratora.
Wyświetla czas z dokładnością do dziesiątej
części godziny i nie można go zresetować.
Wyświetla kody alarmów diagnostycznych, gdy
koncentrator znajduje się w stanie alarmu.
Przepływomierz Służy do regulacji i wyświetlania natężenia
przepływu dostarczanego tlenu. Przepływmieści się w zakresie od 0,5 do 5 litrów na
minutę (l/min) i można go regulować za
pomocą pokrętła.
Miejsce, do którego mocuje się butlę
Stojak butli
nawilżacza (butla nawilżacza. Stojak zawiera elastyczną
jest wyposażeniem opaskę służącą do mocowania butli. Butla
nawilżacza zapewnia dodatkowe nawilżenie
opcjonalnym)
tlenu w układzie. Butla jest wypełniona wodądestylowaną, a wilgoć pomaga zapobiegać
wysychaniu tkanek nosa.
* Tylko modele ze wskaźnikiem
stanu stężenia tlenu (OCSI)

242 - POLPN 15062781-C4 J | Instrukcja obsługi

3 Podłączanie butli nawilżacza i wąsów tlenowych
W przypadku stosowania butli nawilżacza należy
najpierw napełnić ją wodą destylowaną do poziomuokreślonego przez producenta. Umieścić butlę na
przeznaczonym do tego stojaku i zabezpieczyć
elastyczną opaską. Podłączyć gwintowaną końcówkę
rury nawilżacza do butli, a drugą do króćca
wylotowego urządzenia Companion 5. Podłączyć
przewody tlenowe i/lub wąsy tlenowe do króćca
wylotowego na butli nawilżacza.

Jeśli butla nawilżacza nie jest używana, podłączyć
przewody i/lub wąsy tlenowe bezpośrednio do króćca
wylotowego urządzenia Companion 5.

Companion 5
4 Włączenie zasilania i rozgrzaniePodłączyć przewód zasilania do gniazdka sieciowego.

6 Dostosowanie ustawienia przepływu dozalecanego poziomuUżyć pokrętła regulacji przepływu, aby wybrać
przepływ zalecany przez lekarza.

Przełączyć przełącznik zasilania do pozycji Wł.

Przełącznik
zasilaniaJeżeli urządzenie Companion 5 jest włączone
prawidłowo, zaświeci zielony wskaźnik zasilania nawyświetlaczu LED. Wszystkie diody LED zaświecą
po uruchomieniu. Po zakończeniu przez koncentrator
cyklu rozgrzewania zapala się tylko zielona lampka.
Uwaga: Po pierwszym włączeniu zapali się lampka O2, a na ekraniewyświetli się „O2 UP”, dopóki stężenie nie osiągnie specyfikacji.

OSTRZEŻENIE: ZAWSZE USTAWIAĆ PRZEPISANYPRZEPŁYW TLENU. USTAWIENIE PRZEPŁYWU
MOŻNA ZMIENIAĆ TYLKO NA POLECENIE
LEKARZA. KONCENTRATORA TLENU MOŻNA
UŻYWAĆ PODCZAS SNU ZGODNIE Z ZALECENIAMILEKARZA.
Oddychać normalnie przez wąsy tlenowe.
Aby wyregulować natężenie przepływu:Obrócić w kierunku przeciwnym do ruchu wskazówek
zegara, aby zwiększyć przepływ.
Obracać zgodnie z ruchem wskazówek zegara, aby
zmniejszyć przepływ.
Środek kulki wskazuje natężenie przepływu. Patrz
rysunek poniżej.

Dioda LEDwskaźnika
zasilania

Prawidłowe

PN 15062781-C4 J | Instrukcja obsługi

243 - POL

Companion 5Ważne!

Definicje instrukcji bezpieczeństwa:
OSTRZEŻENIE: WAŻNA INFORMACJA DOTYCZĄCAZAGROŻEŃ, KTÓRE MOGĄ SPOWODOWAĆ
POWAŻNE OBRAŻENIA.
PRZESTROGA: Ważna informacja pozwalająca
uniknąć uszkodzenia urządzenia Companion 5.
Uwaga: Informacja wymagająca szczególnej uwagi.

Wskazania do zastosowania
PrzeznaczenieUrządzenie Companion 5 jest przeznaczone do
podawania tlenu. Urządzenie nie jest przeznaczone
do podtrzymywania życia. Nie zapewnia też żadnychnarzędzi do monitorowania pacjenta.
OSTRZEŻENIE: ZAWSZE USTAWIAĆ PRZEPISANYPRZEPŁYW TLENU. USTAWIENIE PRZEPŁYWU
MOŻNA ZMIENIAĆ TYLKO NA POLECENIE
LEKARZA. PRZENOŚNEGO KONCENTRATORATLENU MOŻNA UŻYWAĆ PODCZAS SNU ZGODNIEZ ALECENIAMI LEKARZA.
OSTRZEŻENIE: ABY MIEĆ PEWNOŚĆ, ŻE PODAWANA
DAWKA TLENU JEST ODPOWIEDNIA DLA STANU ZDROWIAUŻYTKOWNIKA, NALEŻY UŻYWAĆ URZĄDZENIA COMPANION
5 TYLKO Z USTAWIENIAMI PRZEPISANYMI DLA DANEGOPOZIOMU AKTYWNOŚCI. UŻYWANE KOMPONENTY MUSZĄ BYĆ
ZGODNE ZE SPECYFIKACJAMI PODANYMI PRZEZ PRODUCENTA
KONCENTRATORA ORAZ TAKIE SAME, JAK PODCZASKONFIGUROWANIA USTAWIEŃ.
OSTRZEŻENIE: URZĄDZENIE TO NIE JEST SYSTEMEMPODTRZYMYWANIA ŻYCIA. OSOBY W PODESZŁYM WIEKU,
DZIECI ORAZ INNI PACJENCI, KTÓRZY NIE SĄ W STANIEINFORMOWAĆ O DYSKOMFORCIE, MOGĄ WYMAGAĆ
DODATKOWEGO MONITOROWANIA I/LUB ROZMIESZCZONEGOSYSTEMU ALARMOWEGO W CELU PRZEKAZANIA INFORMACJI
O DYSKOMFORCIE I PILNEJ POTRZEBIE MEDYCZNEJODPOWIEDZIALNEMU OPIEKUNOWI W CELU ZAPEWNIENIABEZPIECZEŃSTWA. PACJENCI Z WADAMI SŁUCHU LUB WZROKUMOGĄ WYMAGAĆ POMOCY W MONITOROWANIU ALARMÓW.

244 - POLPN 15062781-C4 J | Instrukcja obsługi

OSTRZEŻENIE: KOBIETY W CIĄŻY ANI KARMIĄCE NIE POWINNYSTOSOWAĆ AKCESORIÓW ZALECANYCH W TEJ INSTRUKCJI,
PONIEWAŻ MOGĄ ONE ZAWIERAĆ FTALANY.
OSTRZEŻENIE: PRAWO FEDERALNE (STANY ZJEDNOCZONE)
DOPUSZCZA SPRZEDAŻ TEGO URZĄDZENIA WYŁĄCZNIE PRZEZLEKARZA LUB PRACOWNIKA SŁUŻBY ZDROWIA ALBO NA JEGOZLECENIE.

Przeciwwskazania dozastosowania
OSTRZEŻENIE: W NIEKTÓRYCH OKOLICZNOŚCIACHSTOSOWANIE TLENU BEZ ZALECENIA LEKARZAMOŻE BYĆ NIEBEZPIECZNE. UŻYWAĆ URZĄDZENIA,
TYLKO JEŚLI ZOSTAŁO PRZEPISANE PRZEZ LEKARZA.
OSTRZEŻENIE: NIE STOSOWAĆ W OBECNOŚCI ŁATWOPALNYCHŚRODKÓW ZNIECZULAJĄCYCH.
OSTRZEŻENIE: COMPANION 5 JEST URZĄDZENIEM
ELEKTRYCZNYM, W ZWIĄZKU Z CZYM NIE DZIAŁA PODCZASPRZERW W ZASILANIU. MOŻE TEŻ WYMAGAĆ SERWISOWANIAPRZEZ WYKWALIFIKOWANEGO SPECJALISTĘ. NIESTOSOWAĆ URZĄDZENIA COMPANION 5 U UŻYTKOWNIKÓW,W PRZYPADKU KTÓRYCH TEGO RODZAJU PRZERWYW DZIAŁANIU MOGŁYBY DOPROWADZIĆ DO NIEPOŻĄDANYCHSKUTKÓW ZDROWOTNYCH.

Wytyczne dotyczące
bezpieczeństwa
OSTRZEŻENIE: MODYFIKOWANIE TEGOURZĄDZENIA JEST ZABRONIONE.
OSTRZEŻENIE: PRODUCENT ZALECA, ABY W RAZIE PRZERWYW ZASILANIU, SYTUACJI ALARMOWEJ LUB AWARIIMECHANICZNEJ DOSTĘPNE BYŁO ALTERNATYWNE ŹRÓDŁO
TLENU. ABY WYBRAĆ RODZAJ SYSTEMU REZERWOWEGO,NALEŻY SKONSULTOWAĆ SIĘ Z DOSTAWCĄ SPRZĘTU.
OSTRZEŻENIE: TO URZĄDZENIE DOSTARCZA MIESZANKĘ
O WYSOKIM STĘŻENIU TLENU, KTÓRA POWODUJE SZYBKIESPALANIE. W ODLEGŁOŚCI DWÓCH (2) METRÓW OD (1)URZĄDZENIA LUB (2) AKCESORIÓW ZAWIERAJĄCYCH
TLEN NIE WOLNO PALIĆ ANI UŻYWAĆ OTWARTEGO OGNIA.
ZIGNOROWANIE TEGO OSTRZEŻENIA MOŻE DOPROWADZIĆ DOPOŻARU, USZKODZENIA MIENIA, OBRAŻEŃ CIAŁA, A NAWET
ŚMIERCI.

Companion 5
OSTRZEŻENIE: NIE UŻYTKOWAĆ URZĄDZENIAW OGRANICZONEJ LUB ZAMKNIĘTEJ PRZESTRZENI(NP. MAŁEJ WALIZCE LUB TOREBCE), GDZIE WENTYLACJA JESTUTRUDNIONA. MOŻE TO SPOWODOWAĆ PRZEGRZEWANIESIĘ KONCENTRATORA TLENU I MIEĆ NEGATYWNY WPŁYW NAJEGO PRACĘ.
OSTRZEŻENIE: USTAWIAJĄC KONCENTRATOR,UNIKAĆ DYMU, ZANIECZYSZCZEŃ POWIETRZAI OPARÓW.
OSTRZEŻENIE: KORZYSTANIE Z NIEKTÓRYCH AKCESORIÓW
DO PODAWANIA TLENU, KTÓRYCH NIE WYSZCZEGÓLNIONO
WŚRÓD KOMPONENTÓW DO STOSOWANIA Z TYM
URZĄDZENIEM, MOŻE MIEĆ NEGATYWNY WPŁYW NAJEGO DZIAŁANIE. ZALECANE AKCESORIA WYMIENIONE SĄW NINIEJSZEJ INSTRUKCJI.
OSTRZEŻENIE: JEŚLI DOSZŁO DO UPUSZCZENIA, USZKODZENIALUB ZAMOCZENIA KONCENTRATORA TLENU, SKONTAKTOWAĆ
SIĘ Z PRACOWNIKIEM OPIEKI ZDROWOTNEJ W CELUSPRAWDZENIA I W RAZIE POTRZEBY NAPRAWY. NIE UŻYWAĆ
KONCENTRATORA TLENU, JEŚLI MA USZKODZONE PRZEWÓD
LUB WTYCZKĘ ZASILANIA.
OSTRZEŻENIE: UWAŻAĆ, ABY NIE ZABLOKOWAĆ WLOTÓW
ANI WYLOTU POWIETRZA. NIE WRZUCAĆ ANI NIEWKŁADAĆ PRZEDMIOTÓW DO OTWORÓW W OBUDOWIEURZĄDZENIA. MOŻE TO SPOWODOWAĆ PRZEGRZEWANIE SIĘ
KONCENTRATORA TLENU I MIEĆ NEGATYWNY WPŁYW NA JEGOPRACĘ.
OSTRZEŻENIE: NIE PRZEPEŁNIAĆ OPCJONALNEGONAWILŻACZA. OPCJONALNY NAWILŻACZ NALEŻY NAPEŁNIĆ
WODĄ TYLKO DO POZIOMU PODANEGO PRZEZ PRODUCENTA
NAWILŻACZA.
OSTRZEŻENIE: PRZED ROZPOCZĘCIEM OBSŁUGI URZĄDZENIABEZWZGLĘDNIE PRZECZYTAĆ I ZROZUMIEĆ INSTRUKCJĘ
OBSŁUGI. JEŚLI UŻYTKOWNIK NIE JEST W STANIE ZROZUMIEĆ
OSTRZEŻEŃ I INSTRUKCJI, PRZED ROZPOCZĘCIEM
UŻYTKOWANIA SPRZĘTU NALEŻY SKONTAKTOWAĆ SIĘ
Z DOSTAWCĄ. W PRZECIWNYM RAZIE MOŻE DOJŚĆ DOOBRAŻEŃ CIAŁA LUB USZKODZENIA URZĄDZENIA.
OSTRZEŻENIE: W RAZIE ODCZUWANIA DYSKOMFORTU
LUB WYSTĄPIENIA ZAGROŻENIA ZDROWIA NATYCHMIASTSKORZYSTAĆ Z POMOCY LEKARSKIEJ.

OSTRZEŻENIE: NIE USTAWIAĆ KONCENTRATORA TLENUW POBLIŻU INNEGO SPRZĘTU ANI NA NIM. JEŚLI NIE MA INNEJMOŻLIWOŚCI, MONITOROWAĆ PRAWIDŁOWE DZIAŁANIE
URZĄDZENIA.

OSTRZEŻENIE: PRODUCENT ZALECA, ABY W RAZIEPRZERWY W ZASILANIU, SYTUACJI ALARMOWEJ
LUB AWARII MECHANICZNEJ DOSTĘPNE BYŁOALTERNATYWNE ŹRÓDŁO TLENU. ABY WYBRAĆRODZAJ SYSTEMU REZERWOWEGO, NALEŻYSKONSULTOWAĆ SIĘ Z DOSTAWCĄ SPRZĘTU.
OSTRZEŻENIE: TO URZĄDZENIE DOSTARCZA MIESZANKĘ
O WYSOKIM STĘŻENIU TLENU, KTÓRA POWODUJE SZYBKIESPALANIE. W POMIESZCZENIACH, W KTÓRYCH JEST (1) TOURZĄDZENIE LUB ZNAJDUJĄ SIĘ (2) AKCESORIA ZAWIERAJĄCE
TLEN, NIE WOLNO PALIĆ ANI UŻYWAĆ OTWARTEGO OGNIA.
ZIGNOROWANIE TEGO OSTRZEŻENIA MOŻE DOPROWADZIĆ DOPOŻARU, USZKODZENIA MIENIA, OBRAŻEŃ CIAŁA, A NAWET
ŚMIERCI.
OSTRZEŻENIE: KORZYSTANIE Z NIEKTÓRYCH AKCESORIÓW
DO PODAWANIA TLENU, KTÓRYCH NIE WYSZCZEGÓLNIONO
WŚRÓD KOMPONENTÓW DO STOSOWANIA Z TYM
URZĄDZENIEM, MOŻE MIEĆ NEGATYWNY WPŁYW NAJEGO DZIAŁANIE. ZALECANE AKCESORIA WYMIENIONE SĄW NINIEJSZEJ INSTRUKCJI.
OSTRZEŻENIE: JEŚLI DOSZŁO DO UPUSZCZENIA, USZKODZENIALUB ZAMOCZENIA KONCENTRATORA TLENU, SKONTAKTOWAĆ
SIĘ Z PRACOWNIKIEM OPIEKI ZDROWOTNEJ W CELUSPRAWDZENIA I W RAZIE POTRZEBY NAPRAWY. NIE UŻYWAĆ
KONCENTRATORA TLENU, JEŚLI MA USZKODZONE PRZEWÓD
LUB WTYCZKĘ ZASILANIA.
OSTRZEŻENIE: STOSOWAĆ WYŁĄCZNIE CZĘŚCI ZAMIENNEZALECANE PRZEZ PRODUCENTA W CELU ZAPEWNIENIAPRAWIDŁOWEGO DZIAŁANIA URZĄDZENIA ORAZ UNIKNIĘCIARYZYKA POŻARU I OPARZEŃ.
OSTRZEŻENIE: NIE SMAROWAĆ KOŃCÓWEK, ZŁĄCZY,PRZEWODÓW ANI INNYCH AKCESORIÓW KONCENTRATORATLENU, PONIEWAŻ STWARZA TO ZAGROŻENIE POŻAREMI POPARZENIAMI.

OSTRZEŻENIE: JEŚLI KONCENTRATOR TLENU BĘDZIE
UŻYWANY W WARUNKACH WYKRACZAJĄCYCH POZA ZAKRESTEMPERATUR ROBOCZYCH, GENEROWANE STĘŻENIE
TLENU MOŻE BYĆ NIŻSZE NIŻ W DANYCH TECHNICZNYCH.
INFORMACJE NA TEMAT LIMITÓW TEMPERATUR DOSTĘPNE
SĄ W TEJ INSTRUKCJI, W CZĘŚCI ZAWIERAJĄCEJ DANETECHNICZNE.
OSTRZEŻENIE: DO KONSERWACJI AKCESORIÓWZAWIERAJĄCYCH TLEN ANI DO KONCENTRATORA TLENU
NIE UŻYWAĆ OLEJU, SMARU, PRODUKTÓW OPARTYCH NA
ROPIE NAFTOWEJ ANI INNYCH ŚRODKÓW ŁATWOPALNYCH.
WYKORZYSTYWAĆ TYLKO BALSAMY I MAŚCI NA BAZIE WODY,KTÓRE MOŻNA STOSOWAĆ W OBECNOŚCI TLENU. TLEN
PRZYŚPIESZA SPALANIE ŁATWOPALNYCH SUBSTANCJI.

PN 15062781-C4 J | Instrukcja obsługi

245 - POL

Companion 5
OSTRZEŻENIE: PALENIE PODCZAS UŻYWANIA TLENU JEST
GŁÓWNĄ PRZYCZYNĄ OBRAŻEŃ I ŚMIERCI SPOWODOWANYCHPRZEZ OGIEŃ. NALEŻY BEZWZGLĘDNIE PRZESTRZEGAĆ
PONIŻSZYCH OSTRZEŻEŃ BEZPIECZEŃSTWA:
OSTRZEŻENIE: W POMIESZCZENIACH, W KTÓRYCH JEST TOURZĄDZENIE LUB ZNAJDUJĄ SIĘ AKCESORIA ZAWIERAJĄCE
TLEN, NIE WOLNO PALIĆ, UŻYWAĆ ŚWIECZEK ANI OTWARTEGO
OGNIA.
OSTRZEŻENIE: PALENIE PODCZAS STOSOWANIA WĄSÓW
TLENOWYCH MOŻE DOPROWADZIĆ DO OPARZEŃ TWARZY,A NAWET ŚMIERCI.
OSTRZEŻENIE: PALĄCY UŻYTKOWNICY POWINNI POSTĘPOWAĆZGODNIE Z NASTĘPUJĄCĄ PROCEDURĄ, KTÓRA MOŻE IM
URATOWAĆ ŻYCIE: WYŁĄCZYĆ KONCENTRATOR TLENU, WYJĄĆ
WĄSY I OPUŚCIĆ POMIESZCZENIE, W KTÓRYM ZNAJDUJE SIĘURZĄDZENIE.
OSTRZEŻENIE: W DOMU LUB W MIEJSCU PODAWANIA
TLENU W WIDOCZNYCH MIEJSCACH UMIEŚCIĆ OSTRZEŻENIA„ZAKAZ PALENIA — PODAWANIE TLENU”. POINFORMOWAĆ
PACJENTÓW I ICH OPIEKUNÓW O ZAGROŻENIACH
ZWIĄZANYCH Z PALENIEM W OBECNOŚCI LUB PODCZASPODAWANIA TLENU MEDYCZNEGO.
OSTRZEŻENIE: NIE UŻYWAĆ KONCENTRATORA TLENUW OBECNOŚCI ŁATWOPALNYCH GAZÓW. MOŻE TO
SPOWODOWAĆ NAGŁY ZAPŁON, KTÓREGO EFEKTEM BĘDZIEUSZKODZENIE MIENIA, OBRAŻENIE CIAŁA LUB ŚMIERĆ.
OSTRZEŻENIE: JEŚLI WĄSY ZOSTANĄ WYJĘTE Z NOSAI UMIESZCZONE NA UBRANIU, POŚCIELI, SOFIE LUB INNYM
MIĘKKIM MATERIALE, A W POBLIŻU BĘDZIE ZNAJDOWAĆ SIĘ
PAPIEROS, ŹRÓDŁO CIEPŁA LUB PŁOMIEŃ, MOŻE DOJŚĆ DO
WYBUCHU POŻARU.
OSTRZEŻENIE: NIE ZOSTAWIAĆ WĄSÓW TLENOWYCH NAUBRANIU, POŚCIELI CZY PODUSZKACH ANI POD NIMI. GDYWŁĄCZONE URZĄDZENIE NIE JEST UŻYWANE, OBECNOŚĆ
TLENU ZWIĘKSZA RYZYKO ZAPALENIA SIĘ MATERIAŁU.
GDY KONCENTRATOR TLENU NIE JEST UŻYWANY, USTAWIĆ
PRZEŁĄCZNIK ZASILANIA I/0 W POZYCJI 0 (WYŁ.).
OSTRZEŻENIE: NIE PODEJMOWAĆ PRÓBSERWISOWANIA URZĄDZENIA, Z WYJĄTKIEM
POTENCJALNYCH ROZWIĄZAŃ PROBLEMÓWWYMIENIONYCH W TEJ INSTRUKCJI. NIE
ZDEJMOWAĆ POKRYW. SERWISOWAĆ
URZĄDZENIE I ZDEJMOWAĆ POKRYWYMOŻE WYŁĄCZNIE DOSTAWCA SPRZĘTU LUB
WYKWALIFIKOWANY TECHNIK.

246 - POLPN 15062781-C4 J | Instrukcja obsługi

OSTRZEŻENIE: UŻYWAĆ TYLKO AKCESORIÓW ZALECANYCHPRZEZ PRODUCENTA. STOSOWANIE INNYCH MOŻE BYĆ
NIEBEZPIECZNE, SPOWODOWAĆ POWAŻNE USZKODZENIEKONCENTRATORA TLENU I DOPROWADZIĆ DO UNIEWAŻNIENIAGWARANCJI.
PRZESTROGA: Zawsze odłączać zasilacz AC
najpierw od gniazda sieciowego, a dopieropóźniej od koncentratora tlenu.
PRZESTROGA: Przewody doprowadzania tlenu i zasilaniaumieścić w taki sposób, aby nie stwarzały zagrożenia
potknięcia się.
PRZESTROGA: Przestrzegać instrukcji producenta.
W przeciwnym razie gwarancja będzie nieważna.
Uwaga: Przenośny sprzęt komunikacyjny wykorzystujący
częstotliwości radiowe może oddziaływać na elektryczny sprzęt
medyczny.
OSTRZEŻENIE: ZAWSZE PRZEDŁUŻAĆ PRZEWÓD ZASILAJĄCY
TLENEM I KORBY ZASILAJĄCE W SPOSÓB, KTÓRY ZAPOBIEGAZAGROŻENIU PRZEWRÓCENIA LUB MOŻLIWEMU PRZYPADKOWEMU ZANIECZYSZCZENIU.
OSTRZEŻENIE: STOSOWANIE TEGO PRODUKTU MOŻE NARAŻAĆ NA
KONTAKT Z NIKLEM, KTÓRY W STANIE KALIFORNIA UZNAWANYJEST ZA SUBSTANCJĘ RAKOTWÓRCZĄ. DODATKOWE INFORMACJE
DOSTĘPNE SĄ NA STRONIE WWW.P65WARNINGS.CA.GOV.
OSTRZEŻENIE: W PRZYPADKU WYSTĄPIENIA POWAŻNEGO
INCYDENTU Z TYM URZĄDZENIEM, UŻYTKOWNIK POWINIENNATYCHMIAST ZGŁOSIĆ INCYDENT DOSTAWCY I / LUB PRODUCENTOWI. POWAŻNY INCYDENT JEST DEFINIOWANY JAKO
URAZ, ŚMIERĆ LUB MOŻLIWOŚĆ SPOWODOWANIA OBRAŻEŃ
/ ŚMIERCI W PRZYPADKU PONOWNEGO WYSTĄPIENIAZDARZENIA. UŻYTKOWNIK MOŻE RÓWNIEŻ ZGŁOSIĆ INCYDENTWŁAŚCIWEMU ORGANOWI W KRAJU, W KTÓRYM ZDARZENIEMIAŁO MIEJSCE.

Companion 5
Dane techniczne
W poniższej tabeli znajdują się ważne informacje
dotyczące zalecanego środowiska pracy lub warunków
pracy urządzenia w celu jego prawidłowegoużytkowania.

Dane techniczneNatężenia
przepływu
Stężenie O2
Wymiary
Masa
Poziom ciśnienia
akustycznegoZużycie energii
Dopuszczalna
temperatura
podczas pracy
Dopuszczalna
wilgotność
podczas pracy
Temperatura
przechowywania
Wilgotność
podczasprzechowywaniaPołączenia
elektryczne
Dopuszczalna
wysokość pracy
Czas pracy

0,5–5,0 l/min
90% (+5,5%/-3%)
54,6x31,8x34,2 cm (21,5x12,5x13,5 cala)
16,3 kg (36 funtów)
50 dB(A) przy 2,0 l/min
285 W przy 2 l/min, maks. 350 W
5°–40°C (41°–104°F)

15% - 95% w punkcie rosy 28 ° C (82,4 ° F)

-25°–70ºC (-13°F–158ºF)
0% - 95% bez kondensacji

Nie używać przedłużaczy. Nie używać gniazdek
elektrycznych z włącznikami.
-382–3011 m (-1253–9879 stóp)
do 24 godzin na dobęPrzewidywany okres eksploatacji urządzenia toprzynajmniej pięć lat.
VAROVÁNÍ: POUŽÍVÁNÍ ZAŘÍZENÍ MIMOSPECIFIKOVANÝCH PROVOZNÍCH PODMÍNEK
SE OČEKÁVÁ NEŽÁDOUCÍ ÚČINKY PRŮTOKU APERCENTU OXYGENU A NÁSLEDNĚ KVALITYTERAPIE.

szafą, a także umożliwiać swobodny wylot powietrza
wylotowego w dolnej lewej części urządzenia.
Wloty powietrza znajdują się w górnej części tylnej
i dolnej koncentratora. Urządzenie ustawić w taki
sposób, aby między koncentratorem a ścianami,
meblami, zasłonami lub innymi przeszkodami
pozostawał odstęp co najmniej 30 cm (12 cali).
Umieścić urządzenie w taki sposób, aby alarmy byłysłyszalne.
Przewody doprowadzające tlen należy ułożyć w taki
sposób, aby nie załamywały się ani nie zaciskały.
OSTRZEŻENIE: NIE UŻYWAĆ KONCENTRATORATLENU W OBECNOŚCI ŁATWOPALNYCH GAZÓW.
MOŻE TO SPOWODOWAĆ NAGŁY ZAPŁON,KTÓREGO EFEKTEM BĘDZIE USZKODZENIEMIENIA, OBRAŻENIE CIAŁA LUB ŚMIERĆ. DO
KONSERWACJI AKCESORIÓW ZAWIERAJĄCYCH
TLEN ANI DO KONCENTRATORA TLENU NIEUŻYWAĆ OLEJU, SMARU, PRODUKTÓW OPARTYCH
NA ROPIE NAFTOWEJ ANI INNYCH ŚRODKÓW
ŁATWOPALNYCH. WYKORZYSTYWAĆ TYLKOBALSAMY I MAŚCI NA BAZIE WODY, KTÓRE
MOŻNA STOSOWAĆ W OBECNOŚCI TLENU.
TLEN PRZYŚPIESZA SPALANIE ŁATWOPALNYCHSUBSTANCJI.
OSTRZEŻENIE: TO URZĄDZENIE DOSTARCZA MIESZANKĘ
O WYSOKIM STĘŻENIU TLENU, KTÓRA POWODUJE SZYBKIESPALANIE. W POMIESZCZENIACH, W KTÓRYCH JEST (1) TOURZĄDZENIE LUB ZNAJDUJĄ SIĘ (2) AKCESORIA ZAWIERAJĄCE
TLEN, NIE WOLNO PALIĆ ANI UŻYWAĆ OTWARTEGO OGNIA.
ZIGNOROWANIE TEGO OSTRZEŻENIA MOŻE DOPROWADZIĆ DOPOŻARU, USZKODZENIA MIENIA, OBRAŻEŃ CIAŁA, A NAWET
ŚMIERCI.

Uwaga: Jeśli koncentrator tlenu był przez dłuższy okres
przechowywany w warunkach wykraczających poza normalne
przedziały temperatur, przed włączeniem go należy poczekać, aż
uzyska standardową temperaturę pracy. (Patrz część instrukcji
zawierająca dane techniczne).

Właściwe umieszczenie Companion 5Urządzenie ustawić w miejscu, w którym nie ma dymu,
oparów i zanieczyszczeń. Prawidłowe umiejscowienieurządzenia powinno umożliwiać przepływ powietrzaprzez trzy wloty powietrza w górnej części tylnej i pod

PN 15062781-C4 J | Instrukcja obsługi

247 - POL

Companion 5
Instrukcja obsługi
Przed rozpoczęciem używania

Niniejsza instrukcja obsługi służy za pomoc podczas
obsługi i konserwacji urządzenia. W razie pytań lubwątpliwości prosimy zwrócić się do dostawcy usług
opieki domowej.
Ważne! NIE należy obsługiwać urządzenia Companion 5
bez uprzedniego zapoznania się z rozdziałem Wytyczne
dotyczące bezpieczeństwa w niniejszej instrukcji. Należy
przestrzegać wszystkich instrukcji obsługi. Należy
przestrzegać wszystkich ostrzeżeń umieszczonych
na urządzeniu i w instrukcji obsługi. Aby zmniejszyć
ryzyko pożaru, obrażeń ciała i poważnego uszkodzeniaurządzenia Companion 5, należy przestrzegać wszystkich
podanych środków ostrożności.
Urządzenie CAIRE Companion 5 nie wytwarza tlenu.
Pobiera jedynie tlen z powietrza w pomieszczeniu
i dostarcza go użytkownikowi w wysoce stężonej
postaci. Powietrze w pomieszczeniu zawiera
w przybliżeniu tylko 21% tlenu.
Powietrze z pomieszczenia jest zasysane dokoncentratora przez sprężarkę. Następnie przepływa
przez kilka filtrów i sito molekularne, które
chemicznie adsorbuje cząsteczki azotu. W rezultacie
powietrze wzbogacone tlenem, które wydostaje sięz koncentratora, zawiera 87–95% tlenu.

Krok 1: Ustawienie urządzenia Companion 5 w celu rozpoczęcia
użytkowaniaUmieścić urządzenie Companion 5 w dobrze
wentylowanym i dobrze oświetlonym miejscu.
Upewnić się, że wloty i wyloty powietrza nie sązasłonięte. Urządzenie Companion 5 należy ustawić
tak, aby wszystkie wskaźniki oraz alarmy dźwiękowe
i graficzne były dobrze widoczne i słyszalne.
Podłączyć urządzenie do gniazdka zasilania AC.
Krok 2: Jeśli butla nawilżacza NIE BĘDZIE używana:

Wlot
powietrza

Wlot
powietrza
chłodzącego

Wylot
powietrza
Wlot
powietrza

a. Podłączyć wąsy lub przewody tlenowe do króćca
wylotowego tlenu.

OSTRZEŻENIE: DO CZYSZCZENIA OBUDOWY,
PANELU STEROWANIA I PRZEWODU ZASILANIA
UŻYWAĆ WYŁĄCZNIE ŁAGODNEGO DOMOWEGOŚRODKA CZYSZCZĄCEGO I WILGOTNEJ SZMATKI(NIE MOKREJ) LUB GĄBKI. PO CZYSZCZENIUWYTRZEĆ DO SUCHA. UWAŻAĆ, ABY ŻADEN PŁYNNIE DOSTAŁ SIĘ DO WNĘTRZA URZĄDZENIA.
SZCZEGÓLNĄ UWAGĘ ZWRÓCIĆ NA ZŁĄCZE
WYLOTOWE TLENU SŁUŻĄCE DO PODŁĄCZANIAWĄSÓW — UPEWNIĆ SIĘ, ŻE NIE MA NA NIM
KURZU, WODY ANI INNYCH ZANIECZYSZCZEŃ.
PRZESTROGA: Uważać, aby nie zablokować
wlotów ani wylotu powietrza. Może tospowodować przegrzewanie się koncentratora
tlenu i mieć negatywny wpływ na jego pracę.
Nie używać koncentratora tlenu bez założonego
filtra wlotu powietrza. Jeśli dostarczono filtr
zamienny, założyć go przed przystąpieniem do
czyszczenia dotychczas używanego filtra. Wymyć
zanieczyszczony filtr w ciepłym roztworze mydła
w wodzie, a następnie przed zamontowaniem
wysuszyć.

b. Przejść do kroku 4.
Krok 3: Jeśli butla nawilżacza BĘDZIE używanaa. Odkręcić pokrywę z butli nawilżacza.
b. Napełnij butlę wodą destylowaną. Upewnić się, żepoziom wody znajduje się pomiędzy maksymalnym
i minimalnym wskazaniem podanym przez producentana butli.
c. Zakręcić pokrywę butli nawilżacza. Sprawdzić, czyjest szczelna.
d. Podłączyć wąsy lub przewody tlenowe do króćca
wylotowego na butli nawilżacza.
Uwaga: Używanie nawilżacza, który nie jest przeznaczony dostosowania z tym koncentratorem, może mieć negatywny wpływ
na jego działanie.

248 - POLPN 15062781-C4 J | Instrukcja obsługi

Companion 5
g. Umieścić butlę w wyznaczonym miejscu na stojaku
butli nawilżacza.
h. Zamocować butlę na stojaku za pomocą elastycznej
opaski.

e. Podłączyć gwintowaną końcówkę przewodu
nawilżacza do pokrywy butli nawilżacza.
i. Przejść do kroku 4.
Krok 4: Włączenie zasilania i rozgrzanie
a. Podłączyć przewód elektryczny urządzenia CAIRE
Companion 5 do gniazdka zasilania.
OSTRZEŻENIE: PRZED UŻYCIEM SPRAWDZIĆ, CZYPRZEWÓD ELEKTRYCZNY NIE JEST USZKODZONY.
JEŚLI PRZEWÓD JEST USZKODZONY, NIE NALEŻY
GO PODŁĄCZAĆ DO GNIAZDKA ELEKTRYCZNEGOANI PRÓBOWAĆ OBSŁUGIWAĆ KONCENTRATORA.

f. Podłączyć drugi koniec przewodu nawilżacza
do króćca wylotowego urządzenia CAIRE
Companion 5.

b. Przełączyć przełącznik zasilania do pozycji
Wł (I). Jeżeli urządzenie Companion 5 jestwłączone prawidłowo, zaświeci zielony wskaźnik
na wyświetlaczu LED. Wszystkie diody LEDzaświecą po uruchomieniu. Po zakończeniu przez
koncentrator cyklu rozgrzewania zapala się tylko
zielona lampka.
Uwaga: Po pierwszym włączeniu zasilania urządzenia, należy
odczekać do dziesięciu (10) minut, aby urządzenie osiągnęło
odpowiednią wydajność.

PN 15062781-C4 J | Instrukcja obsługi

249 - POL

Companion 5
Krok 6: Sprawdzenie przepływu i rozpoczęcie oddychaniaa. Sprawdzić, czy tlen przepływa przez wąsy oraz czy
przewód nie jest zgięty ani zatkany.
Uwaga: Zadbać, aby wąsy były prawidłowo zamontowane.
Powinien być słyszalny przepływ tlenu do końcówki nosowej.
b. Umieścić wąsy w nosie i zacząć oddychać
normalnie.
Krok 5: Regulacja przepływu
a. Obrócić pokrętło regulacji przepływu, aby wybrać
przepływ zalecany przez lekarza.

Krok 7: Wyłączenie urządzenia
a. Ustawić przełącznik zasilania w pozycji Wył. (O),gdy urządzenie CAIRE Companion 5 nie jest jużużywane.
PRZESTROGA: Używać tylko akcesoriów
zalecanych przez producenta. Stosowanie innychmoże być niebezpieczne, spowodować poważne
uszkodzenie koncentratora tlenu i doprowadzić
do unieważnienia gwarancji.
PRZESTROGA: Nie podłączać urządzenia za pomocą
przedłużaczy. Nie podłączać zbyt wielu urządzeń do tegosamego gniazda zasilania. Zastosowanie przedłużaczy
może mieć negatywny wpływ na działanie koncentratora.
Podłączenie zbyt wielu urządzeń do jednego gniazda możespowodować przeciążenie skrzynki rozdzielczej i w efekcie
aktywację wyłącznika/bezpiecznika lub pożar, jeśli
wyłącznik/bezpiecznik nie zadziała.

Aby wyregulować natężenie przepływu:Obrócić w kierunku przeciwnym do ruchu wskazówek
zegara, aby zwiększyć przepływ.
Obracać zgodnie z ruchem wskazówek zegara, aby
zmniejszyć przepływ.
Środek kulki wskazuje natężenie przepływu. Naponiższym rysunku ukazano przepływ wynoszący
4,0 l/min.

Prawidłowe

250 - POL

OSTRZEŻENIE: ZAWSZE USTAWIAĆ PRZEPISANYPRZEPŁYW TLENU. USTAWIENIE PRZEPŁYWU
MOŻNA ZMIENIAĆ TYLKO NA POLECENIE
LEKARZA. KONCENTRATORA TLENU MOŻNA
UŻYWAĆ PODCZAS SNU ZGODNIE Z ZALECENIAMIPN 15062781-C4
LEKARZA. J | Instrukcja obsługi

PRZESTROGA: Korzystanie z przewodów i adapterów
innych niż wskazane, z wyjątkiem przewodów i adapterów
sprzedawanych przez producenta medycznego urządzenia
elektrycznego, jako części zamiennych dla komponentówwewnętrznych może spowodować zwiększenie emisji lubobniżenie odporności koncentratora tlenu na zakłócenia.
OSTRZEŻENIE: UWAŻAĆ, ABY NIE ZABLOKOWAĆ
WLOTÓW ANI WYLOTU POWIETRZA. MOŻETO SPOWODOWAĆ PRZEGRZEWANIE SIĘ
KONCENTRATORA TLENU I MIEĆ NEGATYWNYWPŁYW NA JEGO PRACĘ.

Companion 5
Uwaga: do dostawcy sprzętu: Do Companion 5 zalecane sąnastępujące akcesoria do podawania tlenu:• Kaniula nosowa: Numer katalogowy CAIRE CU002-1• Rurki adaptera nawilżacza: CAIRE Numer części 20843882• Butelka z nawilżaczem: CAIRE Numer części HU003-1• Firebreak: Numer katalogowy CAIRE 20629671
Firebreak jest wymagany do użycia z każdą kaniulą.
• CAIRE oferuje ochronę przeciwpożarową przeznaczoną do użycia
w połączeniu z koncentratorem tlenu. Firebreak to bezpiecznik
termiczny, który zatrzymuje przepływ gazu w przypadku zapalenianastępnej kaniuli lub rurki tlenowej i zapalenia się do wybuchupożaru. Jest on umieszczony w linii z kaniulą nosową lub rurką
tlenową między pacjentem a wylotem tlenu w Companion 5. W
celu prawidłowego użycia osłony przeciwpożarowej należy zawszepostępować zgodnie z instrukcjami producenta (dołączonymi dokażdego zestawu przeciwpożarowego).
• Dodatkowe zalecane akcesoria można znaleźć w katalogu
akcesoriów (PN MLLOX0010) dostępnym na stronie www.caireinc.
com.

PN 15062781-C4 J | Instrukcja obsługi

251 - POL

Companion 5
Stany alarmoweUrządzenie CAIRE Companion 5 używa kombinacjidwóch lub trzech (w przypadku modeli z OCSI)
lampek LED i alarmu dźwiękowego, za pomocą
których ostrzega o wystąpieniu usterki koncentratora.
1) Zielona dioda LED (działanie.

) — sygnalizuje normalne

2) Czerwona dioda LED ( ) — sygnalizuje awarię
systemu. Więcej informacji na ten temat można
znaleźć w poniższej tabeli stanów alarmowych.

Zielony

CzerwonyŻółty3) Żółta dioda LED () — sygnalizuje niskiestężenie tlenu. Więcej informacji na ten tematmożna znaleźć w poniższej tabeli stanów
alarmowych.
W tabeli stanów warunków alarmowych na następnej
stronie przedstawiono różne stany alarmowe, któremogą być wyświetlane przez koncentrator. Tabela
przedstawia możliwe przyczyny i działania, którenależy podjąć w przypadku wystąpienia alarmu.
W przypadku wystąpienia alarmu zaleca się jaknajszybsze skontaktowanie się z pracownikiem służby
zdrowia.
OSTRZEŻENIE: NIE WOLNO IGNOROWAĆ
ALARMÓW
OSTRZEŻENIA: PRODUCENT ZALECA, ABY W RAZIEPRZERWY W ZASILANIU, SYTUACJI ALARMOWEJ
LUB AWARII MECHANICZNEJ DOSTĘPNE BYŁOALTERNATYWNE ŹRÓDŁO TLENU. ABY WYBRAĆRODZAJ SYSTEMU REZERWOWEGO, NALEŻYSKONSULTOWAĆ SIĘ Z DOSTAWCĄ SPRZĘTU.

252 - POLPN 15062781-C4 J | Instrukcja obsługiPowyżej: Companion 5

Companion 5
Tabela stanów alarmowych
Alarmdźwiękowy

Kolorowa
dioda LED

Przyczyna

Wymagane działanie
Brak

Koncentrator działa
prawidłowo.

Wył.
Zielona

Brak zasilania, ale wyłącznik
zasilania jest nadal
w pozycji Wł.

Przerywany

Sprawdzić, czy przewód elektryczny koncentratora jest podłączony
do gniazdka i czy w gniazdku jest zasilanie. Spróbować użyć innegogniazdka. Jeśli problem nie ustąpi, należy skorzystać z zapasowegoźródła tlenu i skontaktować się z pracownikiem służby zdrowia.

Czerwona

Przerywany

Usterka systemu.

Natychmiast obrócić przełącznik zasilania do pozycji Wył. Odłączyć
przewód zasilania od gniazdka. Odczekać 5 minut. Podłączyć przewód
zasilania z powrotem do gniazdka ściennego i ponownie włączyć
koncentrator. Jeśli alarm nie wyłączy się, konieczne jest wezwanie
serwisu. Skorzystać z zapasowego źródła tlenu i natychmiast
skontaktować się z pracownikiem służby zdrowia.

Zbyt wysoka lub zbyt niskaprędkość przepływu.

1. Upewnić się, że wąsy tlenowe nie są zagięte lub zablokowane.
W przypadku stosowania z butlą nawilżacza należy upewnić się, że jest
prawidłowo napełniona i nie powoduje niedrożności.
2. Zadbać, aby urządzenie Companion 5 posiadało odpowiednią
wentylację. Musi znajdować się co najmniej 30 cm (12 cali)
od dowolnej powierzchni, aby otwory wentylacyjne nie byłyzablokowane.
3. Jeśli problem nie ustąpi, należy skorzystać z innego źródła tlenu
i skontaktować się z pracownikiem służy zdrowia w celu uzyskania
pomocy.

Czerwona

Przerywany
CzerwonaUrządzenie Companion 5
wykryło niski poziom tlenu.

Przerywany
ŻółtaUrządzenie Companion 5
wykryło niski poziom tlenu.

Przerywany

1. Upewnić się, że filtry wlotów powietrza i wyloty nie są niedrożne
lub zasłonięte. 2. Upewnić się, że urządzenie Companion 5 znajduje
się w dobrze wentylowanym i dobrze oświetlonym miejscu. Upewnićsię, że pomiędzy tyłem i bokami urządzenia Companion a wszelkimi
przeszkodami (meble, zasłony, itp.) pozostaje odstęp co najmniej
30 cm.
3. Jeśli problem nie ustąpi, należy skorzystać z innego źródła tlenu
i niezwłocznie skontaktować się z pracownikiem służy zdrowia.

Żółta

PN 15062781-C4 J | Instrukcja obsługi

253 - POL

Companion 5
Rozwiązywanie problemów
Zaleca się skontaktowanie się z pracownikiem służby
zdrowia, jeśli koncentrator wyświetla jakiekolwiek
alarmy lub wydaje się nie działać prawidłowo.
Nie należy próbować samodzielnie naprawiać
koncentratora. Naprawa może być wykonywanawyłącznie przez wykwalifikowanego serwisanta.
Jednakże istnieją pewne kroki, które można podjąć

w przypadku problemów z urządzeniem CAIRE
Companion 5. Zostały one przedstawione w poniższej
tabeli. W celu rozwiązania problemu należy wykonać
opisane czynności w podanej kolejności.

Tabela rozwiązywania problemów
Problem

Koncentrator jestwłączony, ale nie
działa.

Przepływ tlenu jest
niski lub zerowy.

Możliwa przyczyna

Wymagane działanie

1. Przewód elektryczny nie jest podłączony dogniazdka.

1. Podłączyć przewód elektryczny.

2. W gniazdu nie ma zasilania.

2. Sprawdzić bezpieczniki i instalację. Spróbować użyć innegogniazdka.

3. Usterka wewnętrzna.

3. Skorzystać z zapasowego źródła tlenu. Natychmiast skontaktowaćsię z pracownikiem służby zdrowia.

1. Zmieniono ustawienie regulacji przepływu.

1. Sprawdzić, czy pokrętło regulacji przepływu jest ustawione na
odpowiednią wartość litrów na minutę (l/min) przepisaną przez
lekarza.

2. Przewody zostały odłączone.

2. Sprawdzić, czy przewody są szczelne na wszystkich połączeniach(króciec wylotowy, butla nawilżacza, tacka na wodę itp.). W razie
potrzeby ponownie dokręcić wszelkie połączenia.

3. Przewód lub wąsy tlenowe są zagięte.

3. Sprawdzić czy przewody nie są zagięte lub niedrożne. W razie
potrzeby naprawić.

4. Wąsy lub przewody przeciekają.

4. Sprawdzić wszystkie przewody i wąsy pod kątem nieszczelności.
Uciekające powietrze można usłyszeć lub wyczuć. Nieszczelne
przewody lub wąsy wymienić.

5. Dopływ powietrza do koncentratora jest
utrudniony.

5. Sprawdzić, czy nic nie zasłania wlotów koncentratora. Koncentrator
powinien być oddalony przynajmniej o 30 cm (12 cali) od ścian lub
innych przeszkód.

6. Usterka wewnętrzna.

6. Skorzystać z zapasowego źródła tlenu. Natychmiast skontaktowaćsię z pracownikiem służby zdrowia.

Uwaga: Jeśli mimo skorzystania z tabeli stanów alarmowych i rozwiązywania problemów urządzenie Companion 5 nadal sygnalizuje alarm,należy skontaktować się z serwisem technicznym pod numerem 1-800-482-2473, podając kod alarmu widoczny na wyświetlaczu LCD.

254 - POLPN 15062781-C4 J | Instrukcja obsługi

Companion 5
Czyszczenie, pielęgnacja i rutynowe czynności konserwacyjne
Rutynowe czynności konserwacyjne

Rutynowe czynności konserwacyjne polegają na
wymianie filtr dużych cząstek raz w tygodniu (jeśli jest
zainstalowany).

Pielęgnacja i czyszczenie urządzenia przezużytkownika

Wymiana wąsów
Regularnie wymieniać przewody zasilające i wąsytlenowe zgodnie z zaleceniami dostawcy usług
opieki domowej. Informacje na temat czyszczenia,
dezynfekcji i wymiany powinien dostarczyć lekarz lub
dostawca usług opieki domowej.
Uwaga: Bezwzględnie przestrzegać instrukcji prawidłowegoużytkowania dostarczonej przez producenta wąsów. Wymieniaćwąsy zgodnie z zaleceniami producenta lub dostawcy sprzętu.
Części do urządzenia dostępne są u dostawcy sprzętu.

Filtr wlotu powietrza (wkład)
Filtr wlotu powietrza może być wymieniany wyłącznie
przez pracownika służby zdrowia.

Uwaga: Producent nie zaleca sterylizowania tego sprzętu.
Uwaga: Nie używać koncentratora tlenu bez założonego filtra wlotu
powietrza.

Butla nawilżacza (jeśli dotyczy)Należy codziennie sprawdzać butlę nawilżacza,
aby upewnić się, że poziom wody mieści siępomiędzy określonym minimalnym i maksymalnym
zakresem. W razie potrzeby napełnić butlę wodądestylowaną. Butlę nawilżacza należy czyścić zgodnie
z instrukcjami producenta.
Procedura czyszczenia:
Przed rozpoczęciem czyszczenia lub dezynfekcji
wyłączyć urządzenie Companion 5 i odłączyć je odzasilania. NIE spryskiwać obudowy zewnętrznej.
Stosować wilgotną (nie mokrą) szmatkę lubgąbkę. Spryskać szmatkę lub gąbkę łagodnym
roztworem detergentu, aby wyczyścić obudowę
i przewód zasilania. Do zdezynfekowania urządzenia
Companion 5 należy użyć środka Lysol® Brand II(lub podobnego). Rozcieńczyć zgodnie z zaleceniami
producenta środka czyszczącego, ale nie rozpylać
cieczy bezpośrednio na urządzenie Companion 5.

Filtr dużych cząstek (opcjonalny):

Filtr cząstek stałych znajduje się na bardzo przednim dole Companion 5.

FiltrCząstek
StałychOSTRZEŻENIE: NIEBEZPIECZEŃSTWO PORAŻENIA
PRĄDEM. ABY UNIKNĄĆ PRZYPADKOWEGOPORAŻENIA PRĄDEM LUB POPARZENIA,
PRZED CZYSZCZENIEM URZĄDZENIA NALEŻY
JE WYŁĄCZYĆ I WYJĄĆ PRZEWÓD ZASILANIAZ GNIAZDA. SERWISOWAĆ KONCENTRATORI ZDEJMOWAĆ POKRYWY MOŻE WYŁĄCZNIEDOSTAWCA SPRZĘTU LUB WYKWALIFIKOWANYTECHNIK.
OSTRZEŻENIE: NIEBEZPIECZEŃSTWO PORAŻENIA PRĄDEM.
ABY UNIKNĄĆ PRZYPADKOWEGO PORAŻENIA PRĄDEM LUB
POPARZENIA, PRZED CZYSZCZENIEM URZĄDZENIA NALEŻY
JE WYŁĄCZYĆ I WYJĄĆ PRZEWÓD ZASILANIA Z GNIAZDA.
SERWISOWAĆ KONCENTRATOR I ZDEJMOWAĆ POKRYWY MOŻEWYŁĄCZNIE DOSTAWCA SPRZĘTU LUB WYKWALIFIKOWANYTECHNIK.

Raz w tygodniu należy go sprawdzać i czyścić. Aby
wyczyścić filtr, użyj następującego procesu:
1) Zdjąć filtr dużych cząstek.
2) Umyć filtr w ciepłej wodzie kranowej przy użyciu
łagodnego roztworu mydła.
3) Dokładnie przepłukać filtr czystą wodą z kranu
i wycisnąć nadmiar wody.
4) Pozostawić filtr do wyschnięcia na powietrzu.
5) Włożyć filtr z powrotem do obudowy.
PN 15062781-C4 J | Instrukcja obsługi

255 - POL

Companion 5
OSTRZEŻENIE: NIEBEZPIECZEŃSTWO PORAŻENIA PRĄDEM.
ABY UNIKNĄĆ PRZYPADKOWEGO PORAŻENIA PRĄDEM LUB
POPARZENIA, PRZED CZYSZCZENIEM URZĄDZENIA NALEŻY
WYJĄĆ PRZEWÓD ZASILANIA Z GNIAZDA. SERWISOWAĆ
KONCENTRATOR I ZDEJMOWAĆ POKRYWY MOŻE WYŁĄCZNIEDOSTAWCA SPRZĘTU LUB WYKWALIFIKOWANY TECHNIK.
OSTRZEŻENIE: UWAŻAĆ, ABY NIE ZAMOCZYĆ KONCENTRATORATLENU I ABY DO WEWNĄTRZ NIE DOSTAŁ SIĘ ŻADEN PŁYN.
MOGŁOBY TO SPOWODOWAĆ USZKODZENIE LUB WYŁĄCZENIEURZĄDZENIA I ZWIĘKSZYĆ RYZYKO PORAŻENIA PRĄDEMI POPARZEŃ.
OSTRZEŻENIE: MOŻE BYĆ NIEBEZPIECZNE POŁĄCZENIEKOMPUTERA 5 Z INNYM SPRZĘTEM.

OSTRZEŻENIE: DO CZYSZCZENIA OBUDOWY, PANELUSTEROWANIA I PRZEWODU ZASILANIA UŻYWAĆ WYŁĄCZNIEŁAGODNEGO DOMOWEGO ŚRODKA CZYSZCZĄCEGO
I WILGOTNEJ SZMATKI (NIE MOKREJ) LUB GĄBKI. PO
CZYSZCZENIU WYTRZEĆ DO SUCHA. UWAŻAĆ, ABY ŻADEN PŁYNNIE DOSTAŁ SIĘ DO WNĘTRZA URZĄDZENIA. SZCZEGÓLNĄ
UWAGĘ ZWRÓCIĆ NA ZŁĄCZE WYLOTOWE TLENU SŁUŻĄCE DOPODŁĄCZANIA WĄSÓW — UPEWNIĆ SIĘ, ŻE NIE MA NA NIM
KURZU, WODY ANI INNYCH ZANIECZYSZCZEŃ.
OSTRZEŻENIE: NIE STOSOWAĆ PŁYNÓW BEZPOŚREDNIO NAURZĄDZENIU. NIEKTÓRE ŚRODKI CHEMICZNE, KTÓRYCH
NIE NALEŻY STOSOWAĆ: ALKOHOL I PRODUKTY NA BAZIEALKOHOLU, KONCENTRATY NA BAZIE CHLORU (CHLOREKETYLENU) ORAZ PRODUKTY NA BAZIE OLEJU (PINE-SOL®,LESTOIL®). ŚRODKÓW TYCH NIE NALEŻY UŻYWAĆ DO
CZYSZCZENIA PLASTIKOWEJ OBUDOWY KONCENTRATORATLENU, PONIEWAŻ MOGĄ USZKODZIĆ POWIERZCHNIĘ.

Utylizacja
W celu prawidłowej utylizacji urządzenie Companion
5 wraz z wszystkimi komponentami należy oddać
do placówki opieki. Można też skontaktować sięz urzędem gminy lub miasta, aby dowiedzieć się, jak
prawidłowo zutylizować akumulator.

Dyrektywy WEEE i RoHS

Nasze produkty są zgodne z Dyrektywą w sprawie
ograniczenia stosowania niektórych niebezpiecznych
substancji w sprzęcie elektrycznym i elektronicznym(RoHS). Zawierają jedynie śladowe ilości ołowiu
i innych niebezpiecznych materiałów.
UWAGA: W celu prawidłowej utylizacji skontaktuj się z
dostawcą sprzętu lub lokalną agencją rządową w celu
uzyskania instrukcji utylizacji.

Ten symbol przypomina właścicielomsprzętu, że zgodnie z Dyrektywą w sprawie zużytegosprzętu elektrycznego i elektronicznego (WEEE) pozakończeniu eksploatacji należy go oddać do zakładu
utylizacji.

CAIRE i CAIRE Inc. są zastrzeżonymi znakami towarowymi firmy
CAIRE Inc. Aby uzyskać pełną listę znaków towarowych, odwiedź
naszą stronę internetową poniżej.
Znaki towarowe: www.careinc.com/corporate/trademarks.
Prawa autorskie © 2020 CAIRE Inc. CAIRE Inc. zastrzega sobie
prawo do zaprzestania produkcji swoich produktów lub zmiany cen,materiałów, wyposażenia, jakości, opisów, specyfikacji i / lub procesów swoich produktów w dowolnym momencie bez uprzedniego
powiadomienia i bez dalszych zobowiązań. lub konsekwencja. Zastrzegamy sobie wszelkie prawa, które nie zostały wyraźnie określone
w niniejszym dokumencie, o ile to możliwe.
256 - POLPN 15062781-C4 J | Instrukcja obsługi

Companion 5
Felhasználói vezérlők és a rendszer állapotjelzői
Belső szimbólumok

02

Oxigénkimenet

I/OFőkapcsoló gomb
A berendezés belsejében nincsenek
javítható alkatrészek. Ne nyissa fel aborítólemezt.

ISO 7000-es szabvány szerinti, a készülék használatára
vonatkozó grafikus szimbólumok – Tárgymutató és összegzésAz orvostechnikai eszközökről szóló 93/42/EGK tanácsi
irányelv
Meghatalmazott képviselő az Európai
Közösségben.
IEC 60417
II. osztályú berendezés, kettős
szigetelésű. Reg. sz.: 5172
ETL igazolás

Tárolási vagy működési hőmérsékleti
tartomány Reg. sz.: 0632

CAN/CSA biztonsági hivatala szerinti
C22.2 No. 601.1 M90 a gyógyászati
villamos készülékekre vonatkozóan.

Tárolási páratartalom. Reg. sz.: 2620
A gyártó neve és címe. Reg. sz.: 3082

FCC-tájékoztató

Gyártás dátuma. Reg. sz.: 2497

21 CFR 801.15: Szövetségi rendeletek kódexe – 21. cím
A szövetségi törvény jelen készülékértékesítését orvos által vagy orvosi
rendelvényre korlátozza.

Katalógusszám. Reg. sz.: 2493
Sorozatszám. Reg. sz.: 2498

A 2012/19/EU tanácsi irányelv az elektromos és elektronikus
berendezések hulladékairólÁlló helyzetet jelölő nyilak.
Reg. sz.: 0623
Vigyázat, törékeny! Reg. sz.: 0621
Óvja az esőtől, tartsa szárazon.
Reg. sz.: 0626
Üzemeltetés előtt olvassa el a felhasználói
kézikönyvet. Reg. sz.: 1641
Halmozási határ szám szerint. Reg. sz.:
2403.
ISO 7010-es szabvány: Grafikus szimbólumok – Biztonsági
színek és biztonsági jelzések – Regisztrált biztonsági jelzések
Olvassa el a használati útmutatót.
Reg. sz.: M002
Figyelmeztetés. Reg. sz.: W001
Tartsa távol nyílt lángtól, tűztől,szikrától. A dohányzás és a nyílt lánghasználata tilos! Reg. sz.: P003
Az egység közelében vagy annaküzemeltetésekor tilos a dohányzás!
Reg. sz.: P002
BF típusú alkalmazott alkatrész(elektromos áramütés elleni védelem
fokozata). Reg. sz.: 5333

WEEE
IEC 60601-1-es szabvány: Gyógyászati villamos készülékek.
1. rész: Az alapvető biztonságra és a lényeges működésre
vonatkozó általános követelmények

IPX1

Csepegő víz ellen védett készülék –IPX1: A Companion 5 készülék
védelmet biztosít a folyadékok
behatolásának káros hatásaival szemben.
(IPX1, az IEC 60529 szabvány szerint)

FCC-TÁJÉKOZTATÓ:A készülék CYBLE-022001-00 félvezetőt tartalmazhat,a Johnson Technology által gyártott 2450AT18B100
számú antennát is ideértve; megfelel az FCC
szabályzat 15. paragrafusában foglaltaknak. A készülék
megfelel a moduláris adókészülékekre vonatkozó,
az FCC DA00-1407 jelű nyilvános közleményében
leírt jóváhagyási követelményeknek. Az adókészülék
az alábbi feltételek teljesülése esetén működtethető:(1) a készülék nem okozhat káros interferenciát, és(2) a készüléknek el kell viselnie minden interferenciát,
még a működését nem kívánatos módon befolyásoló
interferenciát is.
Ez a termék egy vagy több USA-beli és nemzetközi
szabadalom oltalma alatt állhat. A vonatkozó
szabadalmak listájának megtekintéséhez, kérjük,
látogasson el weboldalunkra az alábbi címen.
Szabadalmak: www.caireinc.com/corporate/patents/.
PN 15062781-C4 J | Felhasználói kézikönyv

257 - HUN

Companion 5
Gyors üzembehelyezési útmutató
1 A Companion 5 kicsomagolása

Companion 5

2 A Companion 5 készülék megismerése
Olvassa el a figyelmeztetéseket és óvintézkedéseket,
valamint a jelen kézikönyv további részeiben
ismertetett egyéb információkat. Ismerje meg alaposana Companion 5 fontosabb funkcióit és a felhasználói
vezérlőpultot.

ELÜLSŐ OLDALKülsőburkolatFőkapcsoló

LEDkijelző

OxigénkimeneticsőcsatlakozóÓraszámlálóÁramlásmérő

Párásítópalacktartóállványa(opcionális
palack)HÁTOLDAL
Szállító
fogantyú és
levegőbemenetHűtőlevegőbemenet

Szellőzőnyílás
Levegőbemenet

258 - HUNPN 15062781-C4 J | Felhasználói kézikönyv

Companion 5
A Companion 5 alkotóelemeiKülső burkolat

Tartós műanyag burkolat, amely körbezárjaés védi a CAIRE Companion 5 készülék belső
alkotóelemeit.
FőkapcsolóFőkapcsoló gomb, amely az egységáramellátásának bekapcsolására, illetvemegszakítására szolgál. A bekapcsolt állást az„I” szimbólum, a kikapcsolt állást pedig az „O”
szimbólum jelöli.
Kimeneti
Itt lép ki a szűrt és koncentrált oxigén acsőcsatlakozó
Companion 5 készülékből. A csatlakoztatottegylumenű orrkanül vagy szabványos
oxigéncső szállítja az oxigént a betegnek.
A koncentrátorhoz csatlakoztatott oxigéncső
maximális hossza 15,2 m (50 láb).
LED kijelző
Egy zöld, egy piros és egy sárga* LED lámpából
áll. A zöld lámpa a normál üzemmódot,a sárga* és a piros lámpa pedig riasztási
körülményeket jelez. (További információkért
olvassa el a használati útmutató Riasztási
körülmények című részét.)Óraszámláló
A koncentrátor eltelt működési idejének
digitális kijelzője. A legközelebbi tizedórát
jelzi ki, és nem állítható vissza. A koncentrátor
riasztási állapotában diagnosztikus riasztási
kódokat jelenít meg.
Áramlásmérő
A leadott oxigén áramlási sebességének
beállítására és megjelenítésére szolgál. Az
áramlási sebesség a 0,5–5 liter per perc (LPM)
tartományban állítható a gomb elforgatásával.
A párásítópalack rögzítésének helye. Az
Párásítópalackállványhoz tartozik egy rugalmas heveder,
tartóállványa(opcionális palack) amellyel a palack az állványhoz rögzíthető.
A párásítópalack az oxigénáram további
nedvesítését biztosítja. A palackot desztillált
vízzel kell feltölteni; a nedvesség az orr
szöveteinek kiszáradását hivatott meggátolni.
*Csak oxigénkoncentrátor állapotjelző (OCSI)
típusok esetében.

3 A párásítópalack és a kanül csatlakoztatása
Párásítópalack használata esetén először töltse fel
a palackot desztillált vízzel a gyártó által megadottszintig. Helyezze a palackot az erre a célra szolgáló
tartóállványra, és rögzítse a rugalmas hevederrel.
Csatlakoztassa a párásítócső menetes végét apalackhoz, a másik végét pedig a Companion 5
kimeneti csőcsatlakozójához. Csatlakoztassa az
oxigéncsövet és/vagy az orrkanült a párásítópalackon
található kimeneti csőcsatlakozóhoz.

Ha nem használnak párásítópalackot, csatlakoztassaaz oxigéncsövet és/vagy az orrkanült közvetlenül aCompanion 5 készülék kimeneti csőcsatlakozójához.

PN 15062781-C4 J | Felhasználói kézikönyv

259 - HUN

Companion 5
4 Bekapcsolás és bemelegedés

6 Az áramlás beállítása az előírt szintre

Csatlakoztassa a váltóáramú tápkábelt a váltóáramú
csatlakozóaljzathoz.

Az áramlásszabályozó gomb segítségével válassza ki
az orvosa által előírt áramlási sebességet.

A főkapcsolót állítsa bekapcsolt állásba.

Főkapcsoló

FIGYELMEZTETÉS: NAGYON FONTOS, HOGY CSAKA FELÍRT OXIGÉNSZINTET ÁLLÍTSA BE. CSAKAKKOR MÓDOSÍTSA A KIVÁLASZTOTT ÁRAMLÁSISEBESSÉGET, HA EGY ILLETÉKES ORVOSTÓL ERREKAP UTASÍTÁST. AZ OXIGÉNKONCENTRÁTORT
ALVÁS KÖZBEN IS SZABAD HASZNÁLNI, HA AZTSZAKORVOS JAVASOLJA.
Lélegezzen normálisan az orrkanülön keresztül.

Amikor a Companion 5 készüléket megfelelően
kapcsolják be, a zöld tápellátás jelzőlámpa kigyullada LED kijelzőn. Elindításkor az összes LED lámpa
kigyullad. A bemelegedési ciklus befejeződése után
csak a zöld lámpa világít.

Az áramlási sebesség beállítása:
Az áramlás fokozásához forgassa el a gombot azóramutató járásával ellentétes irányba.
Az áramlás csökkentéséhez forgassa el a gombot azóramutató járásával megegyező irányba.

Megjegyzés: Az első bekapcsolás után az O2 jelzőfény világít, és az„O2 UP” felirat jelenik meg a képernyőn, amíg a koncentráció eléria specifikációkat.

A gömb közepe jelzi az áramlási sebességet. Lásd az
alábbi képet.

Tápfeszültségjelző LED

Helyes260 - HUNPN 15062781-C4 J | Felhasználói kézikönyv

Companion 5
Fontos!

A következő biztonsági figyelmeztetések vannak érvényben:
FIGYELMEZTETÉS: FONTOS BIZTONSÁGI
INFORMÁCIÓK OLYAN VESZÉLYEKKELKAPCSOLATBAN, AMELYEK SÚLYOSEGÉSZSÉGKÁROSODÁST OKOZHATNAK.
VIGYÁZAT: Fontos tudnivalók, hogy elkerülhessea Companion 5 megrongálódását.

FIGYELMEZTETÉS: TERHES ÉS SZOPTATÓ NŐKNEK TILOS AKÉZIKÖNYVBEN JAVASOLT TARTOZÉKOKAT HASZNÁLNI,MIVEL AZOK FTALÁTOKAT TARTALMAZHATNAK.
FIGYELMEZTETÉS: AZ AMERIKAI EGYESÜLT ÁLLAMOKSZÖVETSÉGI TÖRVÉNYEI A JELEN BERENDEZÉSFORGALMAZÁSÁT ÉS BÉRLETÉT CSAK ORVOS VAGYMÁS EGÉSZSÉGÜGYI SZAKEMBER RENDELVÉNYÉRE
ENGEDÉLYEZIK.

A használat ellenjavallatai
Megjegyzés: Különös figyelmet igénylő információk.

Alkalmazási terület
Rendeltetésszerű használat

A Companion 5 oxigénpótlás leadására szolgál.
A készülék nem alkalmas életmentő kezelésre, ésbetegmonitorozó funkciókkal sem rendelkezik.
FIGYELMEZTETÉS: NAGYON FONTOS, HOGY CSAKA FELÍRT OXIGÉNSZINTET ÁLLÍTSA BE. CSAKAKKOR MÓDOSÍTSA A KIVÁLASZTOTT ÁRAMLÁSISEBESSÉGET, HA EGY ILLETÉKES ORVOSTÓL
ERRE KAP UTASÍTÁST. A HORDOZHATÓOXIGÉNKONCENTRÁTORT ALVÁS KÖZBEN ISSZABAD HASZNÁLNI, HA AZT SZAKORVOS
JAVASOLJA.

FIGYELMEZTETÉS: A NEM ORVOS ÁLTAL
ELRENDELT OXIGÉNHASZNÁLAT BIZONYOSKÖRÜLMÉNYEK KÖZÖTT VESZÉLYES IS LEHET.
A KÉSZÜLÉKET CSAK ORVOS ÁLTAL ELRENDELTESETBEN SZABAD HASZNÁLNI.
FIGYELMEZTETÉS: NE HASZNÁLJA A KÉSZÜLÉKET GYÚLÉKONYALTATÓSZEREK JELENLÉTÉBEN!
FIGYELMEZTETÉS: MINT MINDEN ELEKTROMOS
KÉSZÜLÉK ESETÉBEN A FELHASZNÁLÓ TAPASZTALHATMŰKÖDÉSKIESÉSSEL JÁRÓ IDŐSZAKOKAT ÁRAMKIMARADÁSMIATT, ILLETVE SZÜKSÉGESSÉ VÁLHAT A COMPANION 5SZAKKÉPZETT TECHNIKUS ÁLTALI SZERVIZELÉSE. A
COMPANION 5 KÉSZÜLÉK NEM MEGFELELŐ OLYAN BETEG
SZÁMÁRA, AKINEK AZ ILYEN ÁTMENETI MŰKÖDÉSKIESÉSEKNEMKÍVÁNATOS EGÉSZSÉGÜGYI KÖVETKEZMÉNYEKKEL
JÁRNÁNAK.

FIGYELMEZTETÉS: A BETEGSÉGÉNEK MEGFELELŐ TERÁPIÁSOXIGÉNMENNYISÉG BEJUTTATÁSÁNAK BIZTOSÍTÁSAÉRDEKÉBEN A COMPANION 5 KÉSZÜLÉKET KIZÁRÓLAGAKKOR SZABAD HASZNÁLNI, HA MEGTÖRTÉNT EGY VAGYTÖBB BEÁLLÍTÁSI LEHETŐSÉG AZ ÖN ADOTT AKTIVITÁSISZINTJÉNEK MEGFELELŐ ÉRTÉK SZERINTI SZEMÉLYRESZABÁSA VAGY ELŐÍRÁSA. A COMPANION 5 KÉSZÜLÉKETA KONCENTRÁTOR GYÁRTÓJA ÁLTAL KÉSZÍTETT MŰSZAKILEÍRÁSBAN SZEREPLŐ OLYAN SPECIÁLIS ALKATRÉSZEKKEL ÉSTARTOZÉKOKKAL SZABAD CSAK HASZNÁLNI, AMELYEKET AZÖNRE VONATKOZÓ BEÁLLÍTÁSOK MEGHATÁROZÁSA SORÁNIS HASZNÁLTAK.

Biztonsági irányelvek

FIGYELMEZTETÉS: A KÉSZÜLÉKET NEM SZABAD
ÉLETFENNTARTÁSRA HASZNÁLNI. IDŐS, GYERMEKKORÚ
VAGY MÁS OLYAN BETEG ESETÉBEN, AKI NEM KÉPES JELEZNIDISZKOMFORTÉRZÉSÉT, KIEGÉSZÍTŐ MONITOROZÁSRAÉS/VAGY ELOSZTOTT RIASZTÁSI RENDSZERRELEHET SZÜKSÉG, AMELY A DISZKOMFORTÉRZÉSSELKAPCSOLATOS INFORMÁCIÓKAT VAGY SÜRGŐSSÉGIÁLLAPOTOT JELEZHET A FELELŐS GONDOZÓ FELÉ A BETEGEGÉSZSÉGKÁROSODÁSÁNAK ELKERÜLÉSE ÉRDEKÉBEN.
A HALLÁS- ÉS/VAGY LÁTÁSSÉRÜLT BETEGEK SEGÍTSÉGRESZORULHATNAK A RIASZTÁSOK FIGYELÉSÉBEN.

FIGYELMEZTETÉS: EZ A KÉSZÜLÉK NAGY KONCENTRÁCIÓJÚOXIGÉNT AD LE, AMI ELŐSEGÍTI A GYORS ÉGÉST. NE ENGEDJE
MEG A DOHÁNYZÁST VAGY NYÍLT LÁNG HASZNÁLATÁT
(1) A KÉSZÜLÉK VAGY (2) BÁRMILYEN OXIGÉNT SZÁLLÍTÓTARTOZÉK KÉT (2) MÉTERES KÖRZETÉBEN. ENNEK A
FIGYELMEZTETÉSNEK A FIGYELMEN KÍVÜL HAGYÁSA SÚLYOS
TÜZET, ANYAGI KÁROKAT ÉS/VAGY FIZIKAI SÉRÜLÉST VAGYHALÁLT OKOZHAT.

FIGYELMEZTETÉS: A KÉSZÜLÉKEN TILOS
MÓDOSÍTÁST VÉGEZNI!
FIGYELMEZTETÉS: A GYÁRTÓ AZT JAVASOLJA, HOGY TARTSON
MAGÁNÁL KIEGÉSZÍTŐ OXIGÉNFORRÁST ARRA AZ ESETRE, HAÁRAMSZÜNET LENNE, ILLETVE RIASZTÁSI KÖRÜLMÉNY VAGYMŰSZAKI HIBA LÉPNE FEL. ÉRDEKLŐDJÖN ORVOSÁNÁL VAGY
A BERENDEZÉS FORGALMAZÓJÁNÁL, HOGY MILYEN TÍPUSÚTARTALÉKRENDSZERRE VAN SZÜKSÉGE.

FIGYELMEZTETÉS: NE MŰKÖDTESSE AZ EGYSÉGETKISMÉRETŰ VAGY ZÁRT TEREKBEN (PL. KIS TOKBANVAGY KÉZITÁSKÁBAN), AHOL KORLÁTOZOTT A LÉGCSERELEHETŐSÉGE. EZ UGYANIS AZ OXIGÉNKONCENTRÁTOR
TÚLHEVÜLÉSÉT ÉS A TELJESÍTMÉNYÉNEK ROMLÁSÁT
OKOZHATJA.
FIGYELMEZTETÉS: A KONCENTRÁTORT OLYANHELYRE KELL TENNI, AHOL EL LEHET KERÜLNI A
FÜSTÖT, A SZENNYEZŐ ANYAGOKAT ÉS A PÁRÁT.
PN 15062781-C4 J | Felhasználói kézikönyv

261 - HUN

Companion 5
FIGYELMEZTETÉS: A NEM EHHEZ AZOXIGÉNKONCENTRÁTORHOZ VALÓ EGYES OXIGÉNBEVITELITARTOZÉKOK HASZNÁLATA BEFOLYÁSOLHATJA A
TELJESÍTMÉNYT. A JAVASOLT TARTOZÉKOK LISTÁJAMEGTALÁLHATÓ EBBEN A KÉZIKÖNYVBEN.
FIGYELMEZTETÉS: HA AZ OXIGÉNKONCENTRÁTORT
LEEJTETTE, AZ MEGSÉRÜLT VAGY VÍZ ÉRTE, AKKOR KERESSE
MEG AZ OTTHONI EGÉSZSÉGÜGYI ELLÁTÓJÁT, HOGY AZ
ELLENŐRIZHESSE, ÉS ESETLEG MEGJAVÍTSA A KÉSZÜLÉKET.
NE HASZNÁLJA AZ OXIGÉNKONCENTRÁTORT, HA A TÁPKÁBEL
VAGY A CSATLAKOZÓJA MEGSÉRÜLT.
FIGYELMEZTETÉS: NE HAGYJA, HOGY AKÁR A LÉGBESZÍVÓ,AKÁR A LÉGKIMENETI NYÍLÁSOK TAKARVA LEGYENEK.
NE EJTSEN VAGY HELYEZZEN BE SEMMILYEN TÁRGYATA KÉSZÜLÉK SEMELYIK NYÍLÁSÁBA. EZ UGYANISAZ OXIGÉNKONCENTRÁTOR TÚLHEVÜLÉSÉT ÉS ATELJESÍTMÉNYÉNEK ROMLÁSÁT OKOZHATJA.
FIGYELMEZTETÉS: NE TÖLTSE TÚL AZ OPCIONÁLIS
PÁRÁSÍTÓPALACKOT. AZ OPCIONÁLIS PÁRÁSÍTÓPALACKOTKIZÁRÓLAG VÍZZEL SZABAD FELTÖLTENI A PÁRÁSÍTÓ
GYÁRTÓJA ÁLTAL MEGJELÖLT SZINTIG.
FIGYELMEZTETÉS: NE HASZNÁLJA A BERENDEZÉSTADDIG, AMÍG EL NEM OLVASTA ÉS MEG NEM ÉRTETTE EZT
A KÉZIKÖNYVET. HA NEM ÉRTHETŐK AZ ÖN SZÁMÁRA A
FIGYELMEZTETÉSEK ÉS AZ UTASÍTÁSOK, FORDULJON A
BERENDEZÉS FORGALMAZÓJÁHOZ, MIELŐTT MEGKÍSÉRELNÉHASZNÁLNI. ELLENKEZŐ ESETBEN EGÉSZSÉGKÁROSODÁS
VAGY A BERENDEZÉS KÁROSODÁSA KÖVETKEZHET BE.
FIGYELMEZTETÉS: HA ROSSZUL ÉRZI MAGÁT VAGYEGÉSZSÉGÜGYI VÉSZHELYZETBE KERÜL, AZONNAL HÍVJONORVOSI SEGÍTSÉGET.
FIGYELMEZTETÉS: HA A MŰSZAKI JELLEMZŐKNÉLMEGADOTT ÜZEMELTETÉSI HŐMÉRSÉKLETEN KÍVÜL
MŰKÖDTETI AZ OXIGÉNKONCENTRÁTORT, AKKOR A
KONCENTRÁTOR CSAK KORLÁTOZOTTAN TUD MEGFELELNI AZOXIGÉNKONCENTRÁCIÓS JELLEMZŐKNEK. A HŐMÉRSÉKLETIHATÁRÉRTÉKEKET MEGTALÁLJA A KÉZIKÖNYV MŰSZAKIJELLEMZŐKRŐL SZÓLÓ RÉSZÉBEN.
FIGYELMEZTETÉS: NE HASZNÁLJON OLAJAT, ZSÍRT,PETRÓLEUMALAPÚ VAGY MÁS ÉGHETŐ TERMÉKETAZ OXIGÉNT SZÁLLÍTÓ TARTOZÉKOKHOZ VAGY AZOXIGÉNKONCENTRÁTORHOZ. CSAK OXIGÉNNELÖSSZEFÉRHETŐ, VÍZALAPÚ KRÉMEKET ÉS KENŐCSÖKETSZABAD HASZNÁLNI. AZ OXIGÉN FELGYORSÍTJA AZ ÉGHETŐANYAGOK ÉGÉSÉT.
FIGYELMEZTETÉS: AZ OXIGÉNKONCENTRÁTOR NEM
HASZNÁLHATÓ MÁS BERENDEZÉS MELLETT, ALATT VAGYFÖLÖTT ELHELYEZVE. HA NEM LEHET ELKERÜLNI, HOGYA KÉSZÜLÉK MÁS BERENDEZÉS MELLETT, ALATT VAGYFELETT LEGYEN ELHELYEZVE, ELLENŐRIZNI KELL, HOGY
MEGFELELŐEN MŰKÖDIK-E.

262 - HUNPN 15062781-C4 J | Felhasználói kézikönyv

FIGYELMEZTETÉS: A GYÁRTÓ AZT JAVASOLJA,HOGY TARTSON MAGÁNÁL KIEGÉSZÍTŐOXIGÉNFORRÁST ARRA AZ ESETRE, HAÁRAMSZÜNET LENNE, ILLETVE RIASZTÁSI
KÖRÜLMÉNY VAGY MŰSZAKI HIBA LÉPNE
FEL. ÉRDEKLŐDJÖN ORVOSÁNÁL VAGY A
BERENDEZÉS FORGALMAZÓJÁNÁL, HOGYMILYEN TÍPUSÚ TARTALÉKRENDSZERRE VAN
SZÜKSÉGE.
FIGYELMEZTETÉS: EZ A KÉSZÜLÉK NAGY KONCENTRÁCIÓJÚOXIGÉNT AD LE, AMI ELŐSEGÍTI A GYORS ÉGÉST. NE ENGEDJE
MEG A DOHÁNYZÁST VAGY NYÍLT LÁNG HASZNÁLATÁT
ABBAN A HELYISÉGBEN, AHOL (1) EZ A KÉSZÜLÉK VAGY (2)BÁRMILYEN OXIGÉNT SZÁLLÍTÓ TARTOZÉK VAN. ENNEK A
FIGYELMEZTETÉSNEK A FIGYELMEN KÍVÜL HAGYÁSA SÚLYOS
TÜZET, ANYAGI KÁROKAT ÉS/VAGY FIZIKAI SÉRÜLÉST VAGYHALÁLT OKOZHAT.
FIGYELMEZTETÉS: A NEM EHHEZ AZOXIGÉNKONCENTRÁTORHOZ VALÓ EGYES OXIGÉNBEVITELITARTOZÉKOK HASZNÁLATA BEFOLYÁSOLHATJA A
TELJESÍTMÉNYT. A JAVASOLT TARTOZÉKOK LISTÁJAMEGTALÁLHATÓ EBBEN A KÉZIKÖNYVBEN.
FIGYELMEZTETÉS: HA AZ OXIGÉNKONCENTRÁTORT
LEEJTETTE, AZ MEGSÉRÜLT VAGY VÍZ ÉRTE, AKKOR KERESSE
MEG AZ OTTHONI EGÉSZSÉGÜGYI ELLÁTÓJÁT, HOGY AZ
ELLENŐRIZHESSE, ÉS ESETLEG MEGJAVÍTSA A KÉSZÜLÉKET.
NE HASZNÁLJA AZ OXIGÉNKONCENTRÁTORT, HA A TÁPKÁBEL
VAGY A CSATLAKOZÓJA MEGSÉRÜLT.
FIGYELMEZTETÉS: KIZÁRÓLAG A GYÁRTÓ ÁLTAL JAVASOLTCSEREALKATRÉSZEKET HASZNÁLJON A MEGFELELŐ
MŰKÖDÉS BIZTOSÍTÁSA, VALAMINT A TŰZ ÉS ÉGÉSISÉRÜLÉSEK KOCKÁZATÁNAK ELKERÜLÉSE ÉRDEKÉBEN.
FIGYELMEZTETÉS: A TŰZ ÉS AZ ÉGÉSI SÉRÜLÉSEKELKERÜLÉSE ÉRDEKÉBEN NE HASZNÁLJON KENŐANYAGOKATAZ OXIGÉNKONCENTRÁTOR ILLESZTÉKEIHEZ,CSATLAKOZÓIHOZ ÉS MÁS TARTOZÉKAIHOZ.
FIGYELMEZTETÉS: AZ OXIGÉN HASZNÁLATA KÖZBENIDOHÁNYZÁS A TŰZZEL KAPCSOLATOS SÉRÜLÉSEKNEK ÉSHALÁLESETEKNEK AZ ELSŐ SZÁMÚ KIVÁLTÓ OKA. AZ ALÁBBI
BIZTONSÁGI FIGYELMEZTETÉSEKET SZIGORÚAN BE KELLTARTANI:FIGYELMEZTETÉS: NE ENGEDJE MEG A DOHÁNYZÁST,GYERTYA VAGY NYÍLT LÁNG HASZNÁLATÁT ABBAN AHELYISÉGBEN, AHOL A KÉSZÜLÉKET VAGY OXIGÉNT SZÁLLÍTÓTARTOZÉKAIT HASZNÁLJA.
FIGYELMEZTETÉS: A DOHÁNYZÁS AZ OXIGÉNKANÜL VISELÉSEKÖZBEN AZ ARC MEGÉGÉSÉT OKOZHATJA ÉS HALÁLHOZVEZETHET.

Companion 5
FIGYELMEZTETÉS: HA DOHÁNYZIK, AZ ALÁBBI
3 LÉPÉS ÉLETMENTŐ LEHET: KAPCSOLJA KI AZOXIGÉNKONCENTRÁTORT, VEGYE LE A KANÜLT, ÉS HAGYJA EL
A HELYISÉGET, AHOL A KÉSZÜLÉK TALÁLHATÓ.

FIGYELMEZTETÉS: MINDEN MINDEN HELYEZZE AZOXIGÉNELLÁTÓ CSÖVET ÉS AZ ÁRAMELLÁTÓ KÁBELEKET OLYANFÜLBE, AMELY MEGAKADÁLYOZZA AZ UTAZÁSI VESZÉLYT VAGYA VÉLETLENSZERŰ VÉLETLENSZERŰ MŰKÖDÉST.

FIGYELMEZTETÉS: „TILOS A DOHÁNYZÁS –OXIGÉNHASZNÁLAT” FELIRATÚ TÁBLÁKAT KELL JÓL LÁTHATÓ
MÓDON ELHELYEZNI AZ OTTHONÁBAN, VAGY AHOL AZOXIGÉNT HASZNÁLJA. A BETEGEKET ÉS GONDOZÓIKAT
TÁJÉKOZTATNI KELL, HOGY MILYEN VESZÉLYEKKEL JÁR AZ
ORVOSI OXIGÉN JELENLÉTÉBEN VAGY ANNAK HASZNÁLATAKÖZBEN TÖRTÉNŐ DOHÁNYZÁS.

FIGYELMEZTETÉS: EZ A TERMÉK VEGYÜLETEK, TÖBBEK KÖZÖTT
NIKKEL HATÁSÁNAK TEHETI KI ÖNT, AMELY KALIFORNIAÁLLAMBAN RÁKKELTŐKÉNT VAN NYILVÁNTARTVA. TOVÁBBI
INFORMÁCIÓKÉRT KERESSE FEL A WWW.P65WARNINGS.CA.GOVWEBOLDALT.

FIGYELMEZTETÉS: NE HASZNÁLJA AZOXIGÉNKONCENTRÁTORT GYÚLÉKONY GÁZOKJELENLÉTÉBEN. ELLENKEZŐ ESETBEN GYORS ÉGÉSALAKULHAT KI, AMI ANYAGI KÁROKAT, SZEMÉLYI SÉRÜLÉST
VAGY HALÁLT IS OKOZHAT.
FIGYELMEZTETÉS: A KANÜL LEVÉTELE ÉS RUHÁRA,ÁGYNEMŰRE, KANAPÉRA VAGY MÁS PÁRNAANYAGRAHELYEZÉSE FELLOBBANÓ TÜZET OKOZ, HA CIGARETTÁVAL,HŐFORRÁSSAL VAGY LÁNGGAL KERÜL ÉRINTKEZÉSBE.

FIGYELMEZTETÉS: HA KOMOLY ESEMÉNY TÖRTÉNIK EZZELA KÉSZÜLÉKKEL, A FELHASZNÁLÓNAK AZONNAL ÉRTESÍTENIE KELL AZ ESEMÉNYT A SZOLGÁLTATÓVAL ÉS / VAGY A
GYÁRTÓVAL. A SÚLYOS ESEMÉNYT SÉRÜLÉS, HALÁL VAGYSÉRÜLÉS / HALÁL OKAKÉNT KELL MEGHATÁROZNI, HA AZ ESEMÉNY MEGISMÉTLŐDIK. A FELHASZNÁLÓ JELENTHETI AZ ESEMÉNYT ANNAK AZ ORSZÁGNAK AZ ILLETÉKES HATÓSÁGÁHOZ
IS, AHOL AZ ESEMÉNY BEKÖVETKEZETT.

FIGYELMEZTETÉS: NE HAGYJA AZ ORRKANÜLT A RUHÁZATON,ÁGYTAKARÓN VAGY SZÉKHUZATON, ILLETVE AZOK ALATT. HA
AZ EGYSÉGET BEKAPCSOLJA, DE NEM HASZNÁLJA, AKKORAZ OXIGÉN EZEKET AZ ANYAGOKAT GYÚLÉKONNYÁ TESZI. HA
NEM HASZNÁLJA AZ OXIGÉNKONCENTRÁTORT, ÁLLÍTSA A
FŐKAPCSOLÓ GOMBOT (I/0) A 0 (KI) ÁLLÁSBA.
FIGYELMEZTETÉS: NE KÍSÉRELJEN MEG
ELVÉGEZNI SEMMILYEN KARBANTARTÁSI
MUNKÁLATOT A JELEN KÉZIKÖNYVBENFELSOROLTAKON KÍVÜL. NE TÁVOLÍTSA EL
A BURKOLATOKAT; CSAK A BERENDEZÉSFORGALMAZÓJA VAGY EGY SZAKKÉPZETTSZERVIZTECHNIKUS TÁVOLÍTHATJA EL A
BURKOLATOKAT, VAGY SZERVIZELHETI A
KÉSZÜLÉKET.
FIGYELMEZTETÉS: CSAK A GYÁRTÓ ÁLTAL AJÁNLOTTTARTOZÉKOKAT HASZNÁLJA. AZ EGYÉB TARTOZÉKOKHASZNÁLATA VESZÉLYES LEHET, SÚLYOSAN KÁROSODHAT AZOXIGÉNKONCENTRÁTOR, ÉS A GARANCIA ÉRVÉNYÉT VESZÍTI.
VIGYÁZAT: Először mindig a fali váltóáramú
csatlakozóról válassza le a tápegységet, mielőtt
az oxigénkoncentrátorról választaná le azt.
VIGYÁZAT: Az oxigénszállító csöveket és a tápkábelt mindig
úgy helyezze el, hogy ne lehessen bennük megbotlani.
VIGYÁZAT: A garancia érvényességénekmegőrzése érdekében kövesse a gyártó minden
utasítását.
Megjegyzés: A hordozható és mobil RF távközlési berendezések
befolyásolhatják a gyógyászati villamos készülékek működését.

PN 15062781-C4 J | Felhasználói kézikönyv

263 - HUN

Companion 5Műszaki jellemzők
A következő táblázat fontos információkkal szolgál akészülék megfelelő használatához a javasolt működési
környezet és üzemi feltételek tekintetében.

Műszaki jellemzők
Áramlási
sebességek
O2-koncentráció
Méretek

0,5–5,0 LPM

90% (+5,5% / –3%)
54,6 × 31,8 × 34,2 cm(21,5 × 12,5 × 13,5 hüvelyk)
Tömeg
16,3 kg (36 font)
Hangnyomásszint 50 dB(A) 2,0 LPM áramlási sebességnél
Áramfelvétel
285 W 2 LPM áramlási sebességnél,
350 W maximumÜzemeltetési
5 °C és 40 °C (41 °F és 104 °F) között
hőmérséklet
Üzemeltetési
15% - 95% 28 ° C (82,4 ° F) harmatpontnál
páratartalom
Tárolási
–25 °C és 70 °C (–13 °F és 158 °F) között
hőmérséklet
Tárolási
0% - 95%, nem kondenzáló
páratartalom
Elektromos
Ne használjon hosszabítókábelt. Ne használjon
kapcsolóval vezérelt elektromos aljzatokat.
Tengerszint feletti –382 és 3011 méter (–1253 és 9879 láb)üzemi magasság között
Üzemelési idő
Akár napi 24 óra
A berendezés várható élettartama legalább öt év.
FIGYELMEZTETÉS: A KÉSZÜLÉK HASZNÁLATA
A MEGHATÁROZOTT MŰKÖDÉSI FELTÉTELEKNÉLKÜL VÉGREHAJTAN, HOGYAN KÖVETKEZTENMEG AZ OXIGÉN ÁRAMLÁSI SEBESSÉGÉT ÉSSZÁZALÉKOS ARÁNYÁT, ÉS KÖVETKEZÉSKÉPPEN
A TERÁPIA MINŐSÉGÉT.
Megjegyzés: Ha az oxigénkoncentrátort hosszabb ideig a normálüzemi hőmérséklet-tartományon kívül tárolták, bekapcsolás előtt
hagyni kell, hogy az egység hőmérséklete visszaálljon a normálüzemi értékre (lásd a jelen kézikönyv „Műszaki jellemzők” címűfejezetét).

264 - HUNPN 15062781-C4 J | Felhasználói kézikönyv

A Companion 5 készülék megfelelő elhelyezése

Válasszon olyan helyet a készüléknek, ahol elkerülheti,
hogy füstöt, párát vagy szennyezőanyagokat
szívjon be. A készülék helyes elhelyezése lehetővé
teszi a levegő beszívását a készülékszekrény felső
hátsó részén, illetve alatta elhelyezkedő három
levegőbemeneti nyíláson át, továbbá az elhasznált
levegő szabad távozását a készülék bal alsó részén
található szellőzőnyíláson keresztül.
A levegőbemeneti nyílások a koncentrátor felső
hátsó részén, illetve az alján találhatók. Helyezze el
az egységet úgy, hogy legalább 30 cm (12 hüvelyk.)
távolság legyen a koncentrátor és a fal, bútorok,
függönyök, illetve egyéb akadályt jelentő tárgyak
között.
Helyezze el úgy a készüléket, hogy a riasztások
hallhatók legyenek.
Helyezze el úgy az oxigénellátást biztosító csövet,hogy az ne törjön meg és ne záródjon el.
FIGYELMEZTETÉS: NE HASZNÁLJA AZOXIGÉNKONCENTRÁTORT GYÚLÉKONYGÁZOK JELENLÉTÉBEN. ELLENKEZŐ ESETBENGYORS ÉGÉS ALAKULHAT KI, AMI ANYAGI
KÁROKAT, SZEMÉLYI SÉRÜLÉST VAGY HALÁLTIS OKOZHAT. NE HASZNÁLJON OLAJAT, ZSÍRT,PETRÓLEUMALAPÚ VAGY MÁS ÉGHETŐ TERMÉKETAZ OXIGÉNT SZÁLLÍTÓ TARTOZÉKOKHOZ VAGY AZOXIGÉNKONCENTRÁTORHOZ. CSAK OXIGÉNNELÖSSZEFÉRHETŐ, VÍZALAPÚ KRÉMEKET ÉSKENŐCSÖKET SZABAD HASZNÁLNI. AZ OXIGÉNFELGYORSÍTJA AZ ÉGHETŐ ANYAGOK ÉGÉSÉT.
FIGYELMEZTETÉS: EZ A KÉSZÜLÉK NAGY KONCENTRÁCIÓJÚOXIGÉNT AD LE, AMI ELŐSEGÍTI A GYORS ÉGÉST. NE ENGEDJE
MEG A DOHÁNYZÁST VAGY NYÍLT LÁNG HASZNÁLATÁT
ABBAN A HELYISÉGBEN, AHOL (1) EZ A KÉSZÜLÉK VAGY (2)BÁRMILYEN OXIGÉNT SZÁLLÍTÓ TARTOZÉK VAN. ENNEK A
FIGYELMEZTETÉSNEK A FIGYELMEN KÍVÜL HAGYÁSA SÚLYOS
TÜZET, ANYAGI KÁROKAT ÉS/VAGY FIZIKAI SÉRÜLÉST VAGYHALÁLT OKOZHAT.

Companion 5
Kezelési utasítások
Használat előtti teendők

Ez a felhasználói kézikönyv referenciaként szolgál akészülék üzemeltetéséhez és karbantartásához. Ha akészülékkel kapcsolatban kérdése vagy aggálya van,forduljon otthoni egészségügyi ellátójához.
Fontos! NE próbálja üzemeltetni a Companion 5
készüléket, mielőtt elolvasná a kézikönyv „Biztonsági
irányelvek” című részét. Kérjük, kövesse az összes
kezelési utasítást. Kérjük, tartsa szem előtt akészüléken és a felhasználói kézikönyvben ismertetett
valamennyi figyelmeztetést. A tűz, a személyi sérülés
és a Companion 5 súlyos károsodása kockázatának
csökkentése érdekében, kérjük, tartsa be az összes
biztonsági óvintézkedést.
A CAIRE Companion 5 nem hoz létre saját maga
oxigént. A környező szoba levegőjéből kivont, erősen
koncentrált oxigént adja le Önnek. A szoba levegője
természetes formájában csak körülbelül 21% oxigént
tartalmaz térfogatszázalékban számolva.
A szoba levegőjét egy kompresszor szívja be akoncentrátorba. A levegő ezután számos filteren és
egy molekulaszűrő ágyon halad át, amely kémiailag
elnyeli a nitrogénmolekulákat. Ennek következtében
a koncentrátorból kilépő oxigéndús levegő 87–95%
oxigént tartalmaz térfogatszázalékban számolva.

1. lépés: A Companion 5 készülék elhelyezése a használathoz
Helyezze a Companion 5 készüléket egy jól szellőző,jól megvilágított területre. Ügyeljen arra, hogy alevegőbemeneti és a szellőzőnyílások ne legyenek
eltakarva. Helyezze el a Companion 5 készüléket
úgy, hogy minden hallható és látható jelzést könnyen
meglássanak és meghalljanak.
Csatlakoztassa a készüléket a váltóáramú elektromos
aljzathoz.
2. lépés: Ha a párásítópalackot NEM FOGJÁK használni

LevegőbemenetHűtőlevegőbemenet
Szellőzőnyílás
Levegőbemenet

a. Csatlakoztassa az orrkanült vagy az oxigéncsövet az
oxigénkimeneti csőcsatlakozóhoz.

FIGYELMEZTETÉS: A KÉSZÜLÉKSZEKRÉNY, A
VEZÉRLŐPULT ÉS A TÁPKÁBEL TISZTÍTÁSÁHOZ
KIZÁRÓLAG KÍMÉLŐ HÁZTARTÁSI TISZTÍTÓSZERTSZABAD HASZNÁLNI, AMELYET NEDVES (DENEM VIZES) KENDŐVEL VAGY SZIVACCSAL
KELL FELVINNI, MAJD AZ ÖSSZES FELÜLETETSZÁRAZRA KELL TÖRÖLNI. ÜGYELJEN ARRA,NEHOGY FOLYADÉK KERÜLJÖN A KÉSZÜLÉKBE.
FORDÍTSON KÜLÖNÖS FIGYELMET A
KANÜLCSATLAKOZÁS OXIGÉNKIMENETÉRE,ÜGYELVE ARRA, HOGY POR-, VÍZ- ÉSRÉSZECSKEMENTES MARADJON.

b. Ugorjon a 4. lépésre.

VIGYÁZAT: Ne hagyja, hogy akár a légbeszívó,
akár a légkimeneti nyílások takarva legyenek.
Ez ugyanis az oxigénkoncentrátor túlhevülését
és a teljesítményének romlását okozhatja. Neműködtesse az oxigénkoncentrátort behelyezett
légbeszívó szűrő nélkül! Ha van másik szűrője
is, akkor a szennyezett szűrő megtisztítása
előtt helyezze be a pótszűrőt. Meleg, szappanosvízzel tisztítsa meg a szennyezett szűrőt, majdhasználat előtt teljesen szárítsa meg.

3. lépés: Ha a párásítópalackot használni FOGJÁK
a. Csavarja le a párásítópalack tetejét.
b. Töltse fel a palackot desztillált vízzel. Ügyeljen arra,
hogy a víz szintje a gyártó által megadott, a palackon
feltüntetett maximális és minimális szint között legyen.
c. Rögzítse a párásítópalack fedelét. Ügyeljen arra,
hogy ne legyen szivárgás.
d. Csatlakoztassa az orrkanült vagy a szabványos
oxigéncsövet a párásítópalackon található kimeneticsőcsatlakozóhoz.
Megjegyzés: A nem ehhez a koncentrátorhoz való párásító
használata befolyásolhatja a teljesítményt.

PN 15062781-C4 J | Felhasználói kézikönyv

265 - HUN

Companion 5
g. Helyezze a palackot a helyére a párásítópalacktartóállványán.
h. Rögzítse a palackot az állványhoz a rugalmas
hevederrel.

e. Csatlakoztassa a párásítócső menetes végét apárásítópalack fedeléhez.

i. Ugorjon a 4. lépésre.
4. lépés: Bekapcsolás és bemelegedésa. Csatlakoztassa a CAIRE Companion 5 tápkábelét
egy váltóáramú aljzathoz.
FIGYELMEZTETÉS: HASZNÁLAT ELŐTT VIZSGÁLJAMEG A TÁPKÁBELT, NEM SÉRÜLT-E MEG.
HA A KÁBEL SÉRÜLT, NE CSATLAKOZTASSAELEKTROMOS ALJZATHOZ, ÉS NE PRÓBÁLJA MEG
ÜZEMELTETNI A KONCENTRÁTORT.

f. Csatlakoztassa a párásítócső másik végét a
CAIRE Companion 5 készülék kimeneticsőcsatlakozójához.
b. Állítsa a főkapcsolót bekapcsolt (I) állásba. Amikora Companion 5 készüléket megfelelően kapcsolják
be, a LED kijelzőn kigyullad a zöld jelzőfény.
Elindításkor az összes LED lámpa kigyullad. A
bemelegedési ciklus befejeződése után csak a zöld
lámpa világít.
Megjegyzés: Bekapcsolás után hagyja a készüléket legfeljebb
tíz (10) percig, hogy elérje a teljesítményjellemzőit.

266 - HUNPN 15062781-C4 J | Felhasználói kézikönyv

Companion 5
FIGYELMEZTETÉS: NAGYON FONTOS, HOGY CSAKA FELÍRT OXIGÉNSZINTET ÁLLÍTSA BE. CSAKAKKOR MÓDOSÍTSA A KIVÁLASZTOTT ÁRAMLÁSISEBESSÉGET, HA EGY ILLETÉKES ORVOSTÓL ERREKAP UTASÍTÁST. AZ OXIGÉNKONCENTRÁTORT
ALVÁS KÖZBEN IS SZABAD HASZNÁLNI, HA AZTSZAKORVOS JAVASOLJA.
6. lépés: Ellenőrizze az áramlást és lélegezzen normálisan.

5. lépés: Az áramlási sebesség beállítása
a. Forgassa el az áramlásszabályozó gombot az orvosaáltal előírt oxigénáramlási sebességre (LPM).

a. Ellenőrizze, hogy az oxigén tényleg az orrkanülön
keresztül áramlik-e, és nincs-e a cső megtörve,meghajolva vagy eltömődve.
Megjegyzés: Ellenőrizze, hogy a kanült teljesen behelyezteés megfelelően rögzül-e. Hallania vagy éreznie kell az oxigénáramlását az orrkanül ágainál.
b. Helyezze el megfelelően az orrkanüljét, éslélegezzen normálisan az orrkanülön keresztül.
7. lépés: Kikapcsolása. Kapcsolja a főkapcsolót kikapcsolt (O) állásba,ha már nem használja a CAIRE Companion 5
készüléket.
VIGYÁZAT: Csak a gyártó által ajánlotttartozékokat használja. Az egyéb tartozékok
használata veszélyes lehet, súlyosan károsodhat
az oxigénkoncentrátor, és a garancia érvényét
veszíti.

Az áramlási sebesség beállítása:
Az áramlás fokozásához forgassa el a gombot azóramutató járásával ellentétes irányba.
Az áramlás csökkentéséhez forgassa el a gombot azóramutató járásával megegyező irányba.
A gömb közepe jelzi az áramlási sebességet. Az alábbiábrán 4,0 LPM-es áramlási sebesség látható.

VIGYÁZAT: Ne használjon az egységhez hosszabbítót, és ne
csatlakoztasson túl sok dugót ugyanahhoz az elektromos
aljzathoz. A hosszabbító használata károsan befolyásolhatja
a készülék teljesítményét. Ha túl sok fogyasztó vancsatlakoztatva egy aljzathoz, az az elektromos panel
túlterhelését eredményezheti, amitől a megszakító/biztosíték
aktiválódhat, illetve tűz keletkezhet, ha a megszakító vagy abiztosíték meghibásodik.
VIGYÁZAT: A megadottaktól eltérő kábelek és adapterek
alkalmazása – a gyógyászati villamos készülék gyártója által
a belső alkatrészek cseréjére árusított kábelek és adapterek
kivételével – az oxigénkoncentrátor megnövekedettkibocsátását és csökkent zavartűrését eredményezheti.

Helyes

PN 15062781-C4 J | Felhasználói kézikönyv

267 - HUN

Companion 5
FIGYELMEZTETÉS: NE HAGYJA, HOGY AKÁRA LÉGBESZÍVÓ, AKÁR A LÉGKIMENETINYÍLÁSOK TAKARVA LEGYENEK. EZ UGYANIS AZOXIGÉNKONCENTRÁTOR TÚLHEVÜLÉSÉT ÉS ATELJESÍTMÉNYÉNEK ROMLÁSÁT OKOZHATJA.
Megjegyzés: A berendezés szolgáltatójának: A következő
oxigénbeviteli kiegészítők használata ajánlott az Companion 5:• Orrkanül: CAIRE cikkszám: CU002-1• Párásító adapter cső: CAIRE cikkszám: 20843882• Párásító palack: CAIRE cikkszám HU003-1• Tűzjelzés: CAIRE cikkszám: 20629671
Bármely kanül használatához tűzjelzés javasolt / szükséges.
• A CAIRE tűzjelzést kínál, amelyet az oxigénkoncentrátorral
együtt kell használni. A tűzjelzés hőbiztosíték, amely megállítja
a gázáramot abban az esetben, ha az alsó irányú kanül vagy az
oxigéncső meggyullad és a tűzjelző égésbe kerül. Az orrkanültel
vagy az oxigéncsővel összhangban helyezkedik el a beteg és
az Companion 5oxigénkivezetése között. A tűzjelzés megfelelő
használatához mindig olvassa el a gyártó utasításait (mindentűzjelző készlethez mellékelve).
• A további javasolt tartozékokkal kapcsolatban lásd a www.
caireinc.com weboldalon található Tartozékkatalógust (PN
MLLOX0010).

268 - HUNPN 15062781-C4 J | Felhasználói kézikönyv

Companion 5
Riasztási körülményekA CAIRE Companion 5 két vagy három (OCSI-val
ellátott típusok esetében) LED lámpa kombinációjávalés hallható riasztással figyelmeztet a koncentrátor
meghibásodására.
1) Zöld LED () – normál működést jelez.

2) Piros LED ( ) – a rendszer hibás működésére
utal. További részletekért lásd az alábbi „Riasztási
körülmények” című táblázatot.
3) Sárga LED () – alacsony oxigénkoncentrációt
jelez. További részletekért lásd az alábbi „Riasztási
körülmények” című táblázatot.
A következő oldalon található „Riasztási
körülmények” című táblázatban a koncentrátor által
kijelzett különböző riasztási körülményeket mutatjuk
be. A táblázat a lehetséges okokat és a riasztás esetén
szükségessé váló teendőket tartalmazza. Minden
riasztás esetén javasolt mihamarabb felvennie akapcsolatot az egészségügyi ellátójával.

Zöld

Piros

Sárga

Fent: Companion 5

FIGYELMEZTETÉS: A RIASZTÁSOKAT NE HAGYJAFIGYELMEN KÍVÜL!
FIGYELMEZTETÉSEK: A GYÁRTÓ AZT JAVASOLJA,HOGY TARTSON MAGÁNÁL KIEGÉSZÍTŐOXIGÉNFORRÁST ARRA AZ ESETRE, HAÁRAMSZÜNET LENNE, ILLETVE RIASZTÁSI
KÖRÜLMÉNY VAGY MŰSZAKI HIBA LÉPNE FEL.
ÉRDEKLŐDJÖN ORVOSÁNÁL VAGY A BERENDEZÉSFORGALMAZÓJÁNÁL, HOGY MILYEN TÍPUSÚTARTALÉKRENDSZERRE VAN SZÜKSÉGE.

PN 15062781-C4 J | Felhasználói kézikönyv

269 - HUN

Companion 5
Riasztási körülmények táblázata
Hangjelzések

SzínesLED

Kikapcsolva
zöldenIdőszakos
PIROSANIdőszakos

A hiba oka
A koncentrátor megfelelően
működik.

Ellenőrizze, hogy a koncentrátor elektromos kábele csatlakoztatva
van-e elektromos aljzathoz, és az aljzatban van-e feszültség.
Próbáljon ki egy másik aljzatot. Ha a probléma továbbra is fennáll,
csatlakozzon egy tartalék oxigénellátó eszközhöz, és forduljon az
egészségügyi ellátójához.

RendszerhibaÁllítsa azonnal kikapcsolt helyzetbe a főkapcsolót. Húzza ki atápkábelt a váltóáramú csatlakozóaljzatból. Várjon 5 percig.
Csatlakoztassa újra a tápkábelt a fali aljzathoz, majd kapcsolja beújra a koncentrátort. Ha a riasztás továbbra is fennáll, szervizelésre
van szükség. Csatlakozzon egy tartalék oxigénellátó eszközhöz, és
forduljon azonnal az egészségügyi ellátójához.

Túl magas vagy túl alacsonyáramlási sebesség

1. Ügyeljen arra, hogy a kanül ne törjön meg és ne tömődjön el.
Párásítópalack használata esetén ügyeljen arra, hogy a palack
megfelelően fel legyen töltve és ne okozzon elzáródást.
2. Biztosítsa a Companion 5 megfelelő szellőzését. A készülék
legalább 30 cm (12 hüvelyk) távolságra helyezkedjen el bármely
felülettől, hogy a szellőzőnyílásai ne legyenek eltakarva.
3. Ha a probléma továbbra is fennáll, váltson másik oxigénforrásra,és kérjen segítséget az egészségügyi ellátójától.

PIROSANA Companion 5 alacsony
oxigénszintet észlelt.

Időszakos
sárga

A Companion 5 alacsony
oxigénszintet észlelt.

Időszakossárga

270 - HUN

Nincs

A koncentrátor nem kapáramellátást, de a főkapcsoló
bekapcsolt állásban van.

PIROSANIdőszakos

Az Ön teendője

PN 15062781-C4 J | Felhasználói kézikönyv

1. Ellenőrizze, hogy a bemeneti szűrő és a kimeneti nyílások
nincsenek-e eltömődve vagy eltakarva. 2. Ügyeljen arra, hogy aCompanion 5 készülék egy jól szellőző területen helyezkedjen el.
Ügyeljen arra, hogy a Companion 5 hátlapja, oldalai és bármilyen
akadályt képző tárgy (bútor, függöny stb.) között legalább 30 cm(12 hüvelyk) távolság legyen.
3. Ha a probléma továbbra is fennáll, váltson másik oxigénforrásra,és azonnal lépjen kapcsolatba az egészségügyi ellátójával.

Companion 5
Hibaelhárítás
Ajánlott felvennie a kapcsolatot az egészségügyi
ellátójával, ha a koncentrátora valamilyen riasztást
jelenít meg, vagy ha Ön úgy érzi, hogy a koncentrátor
nem működik megfelelően. Ne próbálja meg saját
maga megjavítani a koncentrátort. A javítást kizárólag
szakképzett szerviztechnikus végezheti. Van azonban

néhány hibaelhárítási lépés, amelyeket Ön is megtehet,
ha a CAIRE Companion 5 készülékkel kapcsolatos
problémákat észlel. Ezeket az alábbi táblázatban
tüntettük fel. A táblázatot használhatja úgy, hogy az
ellenőrzést számsorrendben végzi.

Felhasználói hibaelhárítási táblázat
Probléma

A koncentrátor be
van kapcsolva, denem működik.

Az oxigénáramlás
megszűnt vagy
alacsony.

Lehetséges ok

Az Ön teendője

1. Az elektromos tápkábel nincs csatlakoztatva
az elektromos aljzathoz.

1. Csatlakoztassa a tápkábelt.

2. Az elektromos aljzat nem ad le áramot.

2. Ellenőrizze otthonában a biztosítékot és az áramkört. Próbáljon ki
egy másik aljzatot.

3. Belső hiba.

3. Csatlakozzon egy tartalék oxigénellátó eszközhöz. Azonnal vegye fel
a kapcsolatot az egészségügyi ellátójával.

1. Módosították az áramlásszabályozó
beállítását.

1. Ellenőrizze, hogy az áramlásszabályozó gomb az orvosa által előírt,megfelelő liter per perc (LPM) értékre van-e beállítva.

2. A cső levált.

2. Ellenőrizze, hogy a cső minden csatlakozóponton (kimeneticsőcsatlakozó, párásítópalack, vízcsapda stb.) szorosan csatlakozik-e.
Szükség esetén igazítsa szorosabbra a csatlakozásokat.

3. Az oxigéncső vagy a kanül megtört.

3. Ellenőrizze, hogy a cső nem tört-e meg vagy nem tömődött-e el.
Szükség esetén javítsa a problémát.

4. A kanül vagy a cső szivárog.

4. Ellenőrizze, hogy a cső vagy a kanül nem szivárog-e. Ezt megteheti
azáltal, hogy meghallgatja vagy megtapintja, hogy érez-e
levegőszivárgást. Cserélje ki a szivárgó csövet vagy kanült.

5. Gátolt a levegő beáramlása a koncentrátorba.

5. Ellenőrizze, hogy nem zárja-e el valami a koncentrátor bemeneti
nyílásait. Ügyeljen arra, hogy a készülék legalább 30 cm (12 hüvelyk)
távolságban legyen a bútoroktól, a faltól vagy más akadályt jelentő
tárgyaktól.

6. Belső hiba.

6. Csatlakozzon egy tartalék oxigénellátó eszközhöz. Azonnal vegye fel
a kapcsolatot az egészségügyi ellátójával.

Megjegyzés: Ha Companion 5 készüléke még a „Riasztási körülmények” és „Hibaelhárítás” című táblázatok használata után is riaszt, forduljon aműszaki ügyfélszolgálathoz az 1-800-482-2473 telefonszámon, az LCD képernyőn megjelenő riasztási kód bemondásával.

PN 15062781-C4 J | Felhasználói kézikönyv

271 - HUN

Companion 5
Tisztítás, ápolás és rendszeres karbantartás
Rendszeres karbantartás

A rutin karbantartás a fogantyú nagyszemcsés
részecskeszűrőjének (ha van ilyen) heti egyszeri
cseréjéből áll.

A készülék felhasználó általi ápolása és tisztítása

Kanülcsere
Cserélje a csövet és a kanült az otthoni egészségügyi
ellátója által ajánlott rendszerességgel. Orvosa vagy az
otthoni egészségügyi ellátója ellátja Önt a tisztításra,fertőtlenítésre és cserére vonatkozó megfelelő
információkkal.
Megjegyzés: Mindig kövesse a kanül gyártójának a helyes
használatra vonatkozó utasításait. Az eldobható kanül cseréjét
végezze a kanül gyártójának vagy a berendezés forgalmazójának
a javaslata szerint. A készülék forgalmazójától egyéb kellékek isbeszerezhetők.

Levegőbemeneti szűrő (kazetta)
A levegőbemeneti szűrőt kizárólag az egészségügyi
ellátó cserélheti ki.
Nagyszemcsés részecskeszűrő (opcionális):

A szemcsés szűrő a Companion 5 legelején található.

Megjegyzés: A gyártó nem javasolja a berendezés sterilizálását.
Megjegyzés: Ne működtesse az oxigénkoncentrátort behelyezett
légbeszívó szűrő nélkül!
Párásítópalack (ha van ilyen)
Naponta ellenőriznie kell, hogy a párásítópalackban
a vízszint a megadott minimális és maximális szint
között van-e. Töltse fel a palackot szükség szerint
desztillált vízzel. A gyártó utasításait követve tisztítsa
meg a párásítópalackot.
Tisztítási eljárás:Kapcsolja KI a Companion 5 készüléket és válasszale a váltóáramú tápellátásról, mielőtt bármilyen
tisztítási vagy fertőtlenítési műveletet végezne.
SEMMIT NE permetezzen közvetlenül a külső
burkolatra. Használjon nedves (nem átitatott) kendőt
vagy szivacsot. Permetezze be a kendőt vagy szivacsot
kímélő mosószeres oldattal a készülékszekrény
és a tápkábel letisztításához. A Companion 5fertőtlenítéséhez használjon Lysol® Brand II (vagyezzel egyenértékű) fertőtlenítőszert. Hígítsa fel atisztítószer gyártójának utasításai szerint, de nepermetezzen folyadékot közvetlenül a Companion 5
készülékre.
FIGYELMEZTETÉS: ÁRAMÜTÉSVESZÉLY.
KAPCSOLJA KI AZ EGYSÉGET ÉS HÚZZA KI A
TÁPKÁBELT AZ ELEKTROMOS ALJZATBÓL AZ
EGYSÉG TISZTÍTÁSA ELŐTT, HOGY ELKERÜLJEA VÉLETLEN ÁRAMÜTÉS ÉS ÉGÉS VESZÉLYÉT.
KIZÁRÓLAG SZAKKÉPZETT SZERVIZTECHNIKUS
TÁVOLÍTHATJA EL A KÉSZÜLÉK BURKOLATAIT ÉSVÉGEZHET SZERVIZELÉST.

Nagyszemcsés
részecskeszűrő

Hetente egyszer meg kell vizsgálnia és meg kelltisztítania. A szűrő tisztításához használja a következő eljárást:
1) Vegye ki a nagyszemcsés részecskeszűrőt.
2) Mossa ki a szűrőt meleg csapvízzel és kímélő
szappanos oldattal.
3) Öblítse ki alaposan a szűrőt tiszta csapvízzel, éscsavarja ki belőle a vizet.
4) Hagyja a szűrőt a levegőn megszáradni.
5) Helyezze vissza a szűrőt a készülékszekrénybe.
272 - HUNPN 15062781-C4 J | Felhasználói kézikönyv

FIGYELMEZTETÉS: ÁRAMÜTÉSVESZÉLY. KAPCSOLJA KI AZ
EGYSÉGET ÉS HÚZZA KI A TÁPKÁBELT AZ ELEKTROMOSALJZATBÓL AZ EGYSÉG TISZTÍTÁSA ELŐTT, HOGY ELKERÜLJEA VÉLETLEN ÁRAMÜTÉS ÉS ÉGÉS VESZÉLYÉT. KIZÁRÓLAGSZAKKÉPZETT SZERVIZTECHNIKUS TÁVOLÍTHATJA EL A
KÉSZÜLÉK BURKOLATAIT ÉS VÉGEZHET SZERVIZELÉST.

Companion 5
FIGYELMEZTETÉS: ÁRAMÜTÉSVESZÉLY. HÚZZA KI A TÁPKÁBELT
AZ ELEKTROMOS ALJZATBÓL AZ EGYSÉG TISZTÍTÁSA ELŐTT,HOGY ELKERÜLJE A VÉLETLEN ÁRAMÜTÉS ÉS ÉGÉS VESZÉLYÉT.
KIZÁRÓLAG SZAKKÉPZETT SZERVIZTECHNIKUS TÁVOLÍTHATJA
EL A KÉSZÜLÉK BURKOLATAIT ÉS VÉGEZHET SZERVIZELÉST.
FIGYELMEZTETÉS: AZ OXIGÉNKONCENTRÁTORT ÓVNI KELL
ATTÓL, HOGY NEDVESSÉG ÉRJE VAGY FOLYADÉK KERÜLJÖN
BELE. EZ UGYANIS MEGHIBÁSODÁST ÉS LEÁLLÁST, ILLETVE
FOKOZOTT ÁRAMÜTÉS- ÉS ÉGÉSVESZÉLYT OKOZHAT.
FIGYELMEZTETÉS: NEM FÜGGETLEN AZ 5 VÁLLALATBEKAPCSOLÁSA MÁS BERENDEZÉSEKKEL.

FIGYELMEZTETÉS: A KÉSZÜLÉKSZEKRÉNY, A VEZÉRLŐPULTÉS A TÁPKÁBEL TISZTÍTÁSÁHOZ KIZÁRÓLAG KÍMÉLŐ
HÁZTARTÁSI TISZTÍTÓSZERT SZABAD HASZNÁLNI, AMELYETNEDVES (DE NEM VIZES) KENDŐVEL VAGY SZIVACCSAL KELLFELVINNI, MAJD AZ ÖSSZES FELÜLETET SZÁRAZRA KELLTÖRÖLNI. ÜGYELJEN ARRA, NEHOGY FOLYADÉK KERÜLJÖNA KÉSZÜLÉKBE. FORDÍTSON KÜLÖNÖS FIGYELMET A
KANÜLCSATLAKOZÁS OXIGÉNKIMENETÉRE, ÜGYELVE ARRA,HOGY POR-, VÍZ- ÉS RÉSZECSKEMENTES MARADJON.
FIGYELMEZTETÉS: NE HASZNÁLJON SEMMILYEN FOLYADÉKOTKÖZVETLENÜL AZ EGYSÉGEN. A KÖVETKEZŐ LISTA ATELJESSÉG IGÉNYE NÉLKÜL TARTALMAZZA A KERÜLENDŐ
VEGYSZEREKET: ALKOHOL ÉS ALKOHOLALAPÚ TERMÉKEK,TÖMÉNY KLÓRBÁZISÚ TERMÉKEK (ETILÉN-KLORID) ÉS
OLAJALAPÚ TERMÉKEK (PINE-SOL®, LESTOIL®). EZEKET NEM
SZABAD HASZNÁLNI AZ OXIGÉNKONCENTRÁTOR MŰANYAGBURKOLATÁNAK TISZTÍTÁSÁRA, MIVEL KÁROSÍTHATJÁK AZ
EGYSÉG MŰANYAG BURKOLATÁT.

Disposal
Minden esetben juttassa el a Companion 5 készüléket
az összes alkatrészével együtt az otthonához tartozó
szolgáltatóhoz a megfelelő hulladékkezelés céljából.
Az akkumulátor megfelelő hulladékkezeléséhez a helyiönkormányzatokat is felkeresheti.

WEEE és RoHS

Termékeink megfelelnek a készülékekben található
Veszélyes anyagokra vonatkozó korlátozások (RoHS)
irányelvének. Az ólom- és egyéb veszélyesanyagtartalom nem haladja meg a nyomokban tartalmazott
mennyiséget.
VIGYÁZAT: A megfelelő hulladékkezeléshez forduljon
a berendezés szállítójához vagy az önkormányzatiügynökséghez a hulladékkezelési útmutatásokért.

Ez a szimbólum arra emlékezteti a készülék
tulajdonosait, hogy Az elektromos és elektronikus
berendezések hulladékainak legmegfelelőbb kezeléséről
szóló irányelv (WEEE) alapján a készüléket vigyék
el egy hulladék-újrahasznosító telepre az élettartamalejárta után.

CAIRE i CAIRE Inc. są zastrzeżonymi znakami towarowymi firmy
CAIRE Inc. Aby uzyskać pełną listę znaków towarowych, odwiedź
naszą stronę internetową poniżej.
Znaki towarowe: www.careinc.com/corporate/trademarks.
Copyright © 2020 CAIRE Inc. A CAIRE Inc. fenntartja a jogot,
hogy bármikor, előzetes értesítés nélkül és további kötelezettség
nélkül megszakítsa termékeit, vagy megváltoztassa termékeinekárait, anyagait, felszereléseit, minőségét, leírásait, specifikációit és /vagy folyamatait a termékekre. vagy következménye. Minden olyan
jogot, amelyet itt kifejezetten nem tüntetünk fel, adott esetben mi
fenntartjuk.
PN 15062781-C4 J | Felhasználói kézikönyv

273 - HUN

Companion 5
NOTES:

274 - HUNPN 15062781-C4 J | Felhasználói kézikönyv

Companion 5
NOTES:PN 15062781-C4 J | Felhasználói kézikönyv

275 - HUN

www.c a ire inc .c om

CAIRE Inc.
2200 Airport Industrial Dr., Ste. 500
Ball Ground, GA 30107   U.S.A.
Medical Product Service GmbH
Borngasse 20
35619 Braunfels, Germany
Tel: +49 (0) 6442-962073
E-mail: info@mps-gmbh.eu
CAIRE and CAIRE Inc. are registered trademarks of CAIRE Inc. Please visit our
website below for a full listing of trademarks. Trademarks: www.caireinc.com/
corporate/trademarks/.
Copyright © 2021 CAIRE Inc. CAIRE Inc. reserves the right to discontinue its products,
or change the prices, materials, equipment, quality, descriptions, specifications and/or, processes to its products at any time without prior notice and with no further
obligation or consequence. All rights not expressly stated herein are reserved by us,
as applicable.

|15062781-C4-JY~

1 Dec 2021

PN 15062781-C4 J