Sprint
Stroller
Hi Flow Stroller

User Manual (US)Manuel d’utilisation (FR)Bedienungsanleitung (DE)Manual del usuario (ES)Manuale dell’utente (IT)Manual do utilizador (PT)
Gebruikershandleiding (NL)
Brugervejledning (DK)Bruksanvisning (NO)Bruksanvisning (SE)Käyttöopas (FI)Εγχειρίδιο χρήστη (GR)
Kullanıcı El Kitabı (TR)Uživatelská příručka (CZ)
Instrukcja obsługi (PL)Felhasználói kézikönyv (HU)

Sprint / Stroller / Hi Flow Stroller
Symbols Glossary
ISO 7000

Do not smoke near unit or while
operating unit. Reg. # P002

Keep away from rain, keep dry. Reg.
# 0626

Type BF applied part (degree ofprotection against electric shock).
Reg. # 5333

Stacking limit by number. Reg. #2403
Name and address of manufacturer.
Reg. # 3082
The country and date of manufacture.
The “CC” identifies the two lettercountry code of the country of manufacture. The date of manufacture is in
YYYY-MM-DD format. Reg. # 6049
Caution, consult accompanying documents. Reg. # 0434A
Catalog Number. Reg. # 2493
Serial Number. Reg. # 2498
Storage or operating temperature
range. Reg. # 0632

Warning. Reg. # W001
Council Directive 93/42/EEC
Authorized representative in theEuropean Community
If the product unique device identifier(UDI) label has the CE#### symbolon it, the device complies with therequirements of Directive 93/42/EEC
concerning medical devices. The
CE#### symbol indicates notified
body number.
This device complies with the requirements of Directive 2010/35/EU
concerning medical devices. It bears
the pi marking as shown.
ADR

Storage humidity range Reg. # 2620

Non-toxic gas.

Atmospheric pressure limitation. Reg.
# 2621

Do not cover unit or carry portable
unit under your clothing. These units
normally vent oxygen. No. 5641

This way up. Reg. # 0623
Fragile, handle with care. Reg. # 0621

Refrigerated Liquid, USP; Produced
by Air Liquefaction
Additional Symbols

Contains hazardous substances. Reg.
# 3723

ISO 7010

Keep unit well ventilated at all times

Importer. Reg. # 3725

Keep away from flammable materials,
oil and grease.

Frostbite may occur on contact withcold liquid or gaseous oxygen, or
frosted parts. Warning low temperature. To warn of low temperature or
freezing conditions. Reg. # W010

Authorized representative in
Switzerland.

The instruction manual must be read.
Reg. # M002
Keep away from open flame, fire,
sparks. Open ignition source andsmoking prohibited. Reg. # P003
2 - US PN MN236-C4 E | User Manual

If the device bears the UKCA markas shown with UKCA#### indicating
the notified body number, this device
complies with UKCA regulations.
IEC 60417

Do not cover unit or carry portable
unit under your clothing. These units
normally vent oxygen. No. 5641

Sprint / Stroller / Hi Flow Stroller
21 CFR 801.15
Federal law restricts this device tosale by or on the order of a physician.
IEC 60601-1

ISO 15223-1
Medical Device. Reg. # 5.7.7
Unique device identifier Reg. # 5.7.10

Drip Proof
Council Directive 2012/19/EU
WEEE
This symbol is to remind the equipment owners to return it to a recycling
facility at the end of its life, per Waste
Electrical and Electronic Equipment(WEEE) Directive.
Our products will comply with therestriction of Hazardous Substances(RoHS) directive. They will not contain more than trace amounts of leador other hazardous material content.

This product may be covered by one or more patents, US and international. Please visit our website
below for the listing of applicable patents. Pat.:www.caireinc.com/corporate/patents/ .

Specifications• Mode of Operation: Continuous Flow
• Type of Protection Against Electrical Shock: Internally Powered Equipment• Degree of Protection Against Electrical Shock: Type BF Applied Part• Classification According to the Degree of Protection Against Ingress of Water: IP22 – Ordinary
Equipment• Equipment not suitable for use in the presence of flammable mixtures

Product Specifications
Sprint
1.5 lb (0,68 kg)
513 L

LOX Capacity
Gaseous Equivalent
Capacity
Weight, Empty
Weight, Filled
HeightDimensions

Stroller
3.0 lb (1,36 kg)
1026 LHi Flow Stroller
3.0 lb (1,36 kg)
1025 L

Typical use time at 2 LPM
Operating Pressure
Normal Evaporation Rate

4.5 lb (2,04 kg)
5.0 lb (2,27 kg)
5.5 lb (2,49 kg)
6.0 lb (2,72 kg)
8.0 lb (3,63 kg)
8.5 lb (3,86 kg)
11.75 in. (298 mm)
13.5 in. (343 mm)
13.5 in. (343 mm)
4.875” (123.83 mm) D x 7.5" 5.875" (149.23 mm) D x 7.5" 5.875" (149 mm) D x 7.5"
(191 mm) W(191 mm) W(191 mm) W
4.3 hours
8.5 hours
8.5 hours
20 psi (137 kPa)
20 psi (137 kPa)
20 psi (137 kPa)
1.3 lb/ day (0,57 kg/ day)
1.3 lb/ day (0,57 kg/ day)
1.3 lb/ day (0,57 kg/ day)Standard Flow Control
Rangeoff, .25, .5, .75, 1, 1.5, 2,
2.5, 3, 4, 5, 6 LPMoff, .25, .5, .75, 1, 1.5, 2,
2.5, 3, 4, 5, 6 LPM

Off, .5, 1, 2, 2.5, 3, 4, 6, 8,
10, 12, 15 LPM

Flow Rate Accuracy*+/- 10%+/- 10%+/- 10%* This accuracy is only at 70F and 14.7 psig and with a calibrated accurate mass flow meter.

PN MN236-C4 E | User Manual US - 3

Sprint / Stroller / Hi Flow Stroller
Warning Information
Important: Read this manual thoroughly before
operating the Sprint/Stroller/Hi Flow Stroller.
RX Only.
WARNING: THIS DEVICE IS NOT INTENDED
FOR LIFE SUSTAINING USE.
WARNING: IF YOU FEEL THE EQUIPMENT IS NOTOPERATING PROPERLY, CALL YOUR HEALTH
CARE PROVIDER. DO NOT ATTEMPT TO REPAIR OR
ADJUST THE UNIT YOURSELF.
WARNING: DO NOT MODIFY THIS EQUIPMENTWITHOUT AUTHORIZATION FROM THE MANUFACTURER.
WARNING: IF CONTINUITY OF OXYGEN SUPPLY IS
REQUIRED, ENSURE THAT AN ADEQUATE SUPPLYOF OXYGEN AND/OR A SECONDARY OXYGENSUPPLY IS AVAILABLE AT ALL TIMES DURING
THERAPY.
WARNING: DO NOT ALLOW SMOKING, CANDLES,OR OPEN FLAMES WITHIN 10 FEET (3 METERS) OF
THE DEVICE, OR CLOSER THAN 8 INCHES (20 CM)
FROM ANY SOURCE OF IGNITION.
WARNING: KEEP YOUR UNIT IN A WELL-VENTILATED AREA.
WARNING: DO NOT STORE LIQUID OXYGENEQUIPMENT IN A CLOSET, CAR TRUNK, OR OTHERCONFINED AREA. DO NOT PLACE BLANKETS,
DRAPERIES, OR OTHER FABRICS OVER EQUIPMENT.
WARNING: THIS PRODUCT CAN EXPOSE YOU
TO CHEMICALS INCLUDING NICKEL, WHICH IS
KNOWN TO THE STATE OF CALIFORNIA TO CAUSECANCER. FOR MORE INFORMATION, GO TO WWW.
P65WARNINGS.CA.GOV.
WARNING: IN THE EVENT THERE IS A SERIOUS
INCIDENT OCCURRING WITH THIS DEVICE, THE
USER SHOULD IMMEDIATELY REPORT THE INCIDENT TO THE PROVIDER AND/OR THE MANUFACTURER. A SERIOUS INCIDENT IS DEFINED AS AN
INJURY, DEATH, OR POTENTIAL TO CAUSE INJURY/DEATH SHOULD THERE BE A REOCCURRENCE OF
THE INCIDENT. THE USER CAN ALSO REPORT THE
INCIDENT TO THE COMPETENT AUTHORITY IN THE
COUNTRY WHERE THE INCIDENT OCCURRED.
Caution: Use the Sprint/Stroller/Hi Flow
Stroller only as directed by your doctor.
Caution: Federal law restricts this device to sale byor on the order of a physician.
Caution: The unit contains liquid oxygen which isextremely cold, almost -300°F (-184°C). Exposure tosuch a low temperature can cause severe frostbite.
4 - US PN MN236-C4 E | User Manual

Caution: Liquid and gaseous oxygen, though nonflammable, cause other materials to burn faster thannormal. This hazard, along with the low temperatureof liquid oxygen, warrants certain safety precautions.
Caution: Keep flammable materials away from thisequipment. Aerosol sprays, oils and grease, including facial creams and petroleum jelly, ignite easily
and may burn rapidly in the presence of oxygen.
Caution: Smoking while wearing an oxygen cannula
can cause facial burns and possibly result in death.
Removing the cannula and placing it on clothing,bedding, sofas, or other cushion material will causea flash fire when exposed to a cigarette, heat source,
spark or flame.
If you smoke please: (1) turn off the portable, (2) takeoff the cannula, and (3) leave the room where thedevice is located.
Caution: In the event of an accidental tip-over, immediately but cautiously return the unit into an uprightposition if possible. If any liquid oxygen is escaping,leave the area immediately and call your healthcare
provider. Do not attempt to move the unit or stop theliquid oxygen from escaping.
Note: Do not touch frosted parts of any unit.
Note: Do not store or operate the portable coupled tothe stationary unit.
Note: Do not allow untrained personnel to handle oroperate this device.
Note: Use of this device is prohibited on commercial
passenger and cargo air flights by the Federal Aviation
Administration.

Intended Use

The CAIRE Sprint/Stroller/Hi Flow Stroller oxygen
unit is intended for the administration of supplemental oxygen. The device is not intended for life
support nor does it provide any patient monitoring
capabilities.

Introduction

Sprint / Stroller / Hi Flow Stroller
Controls
1. Electronic Liquid Level Gauge
2. Electronic Liquid Level Switch
3. Mechanical Liquid Level Gauge
4. Flow Control Knob
5. Oxygen Tube Connector
6. Vent Valve
7. Fill Connections (see next page for QDVIdentification)
8. Carrying Strap with Shoulder Pad
9. Condensation Pad
10. Condensation Cup

The Sprint, Stroller, and Hi Flow Stroller are
intended for the administration of supplemental
oxygen to the patient in the end user’s home andcan also be used in institutions such as nursing
homes or sub-acute care facilities. The Sprint/
Stroller/Hi Flow Stroller provide an ambulatory
source of oxygen for an extended period of time.
The device is used by COPD patients or thosewith diminished breathing capacity. The device
is prescribed to the patient. The device is sold toa provider that is trained to operate and service
the Sprint/Stroller/Hi Flow Stroller. The provider
trains the user.

1

The device is not intended for life support nor doesit provide any patient monitoring capabilities. It
is recommended to have an alternate source of
supplemental oxygen in the event of mechanicalfailure. These portable devices also have a quick
disconnect valve that can be coupled with areservoir lox device for transfilling the portabledevice. The reservoir unit is filled by your healthcare provider.

2
4
5
8

This user manual contains the instructions for using
the Sprint/Stroller/Hi Flow Stroller. Refer to theuser manual supplied with the stationary unit for its
operation.
Note: The service provider will assist with the initial
setup and instruct proper handling and usage of the
unit.
5
7

6
Sprint, Stroller and Hi Flow Stroller units

10
9
PN MN236-C4 E | User Manual US - 5

Sprint / Stroller / Hi Flow Stroller
Quick Disconnect Valve (QDV)
Identification
Portable
Connector

Stationary
Connector

QDV Style

CAIRE
Side Fill

SF
Locking
4. a. If your connector is a push-on:
Lower the Sprint/Stroller/Hi Flow Stroller
unit on to the stationary until you feel theconnector engage.

LIFE-OX
TF
Locking
PENOX

TAEMA
TF
Push‑OnPB

b. If the connector is a Penox or Life-Ox:
Rotate Sprint/Stroller/Hi Flow Stroller until
you feel the connector engage. Then, carefully
and firmly, rotate the Sprint/Stroller/Hi Flow
Stroller clockwise () while pressing
downward.

Operating Instructions
To Fill the Top Fill Sprint/Stroller/Hi Flow
Stroller

Note: Continued on step 6 in To Fill the Side Fill Sprint/
Stroller/Hi Flow Stroller.

1. Clean the fill connectors on both the stationary
and the Sprint/Stroller/Hi Flow Stroller with aclean, dry, lint-free cloth.

To Fill the Side Fill Sprint/Stroller/Hi Flow
Stroller

Caution: The fill connectors must be dry,because moisture can cause your Sprint/
Stroller/Hi Flow Stroller and stationary unit
to freeze together.
2. Make sure the flow control knobs on both the
Sprint/Stroller/Hi Flow Stroller and reservoir arein the off (0) position.
3. Position the fill connector in the upright position
over the stationary unit’s connector.

6 - US PN MN236-C4 E | User Manual

1. Clean the fill connectors on both the stationary
and the Sprint/Stroller/Hi Flow Stroller with aclean, dry, lint-free cloth.
WARNING: THE FILL CONNECTORSMUST BE DRY, BECAUSE MOISTURE CAN
CAUSE YOUR SPRINT/STROLLER/HI FLOWSTROLLER AND STATIONARY UNIT TO
FREEZE TOGETHER.
2. Make sure the flow control knobs on both the
Sprint/Stroller/Hi Flow Stroller and reservoir
are in the off (0) position

Sprint / Stroller / Hi Flow Stroller
3. Make the Sprint/Stroller/Hi Flow Stroller
fill connector is in the upright position to thestationary connector.

Note: Liquid level gauge readings are most accurate
30 seconds or more after the Sprint/Stroller/Hi Flow
Stroller is filled.

4. Rotate the Sprint/Stroller/Hi Flow Stroller
counterclockwise () until you feel thepin and slot engage.
7. You may also see vapor around the connection.
This is normal and is due to the great differencein temperature between the liquid oxygen and thewarm air surrounding the units.
8. When the unit is full, close the vent valve. You
can tell that the unit is full when the hissing soundchanges in tone, and some liquid comes out of thevent line.
5. Carefully and firmly rotate the Sprint/Stroller/Hi
Flow Stroller back to the upright position. Now
the units are locked together.

Note: Closing the valve after a nearly continuous
stream of liquid is discharging fills the Sprint/Stroller/Hi
Flow Stroller most completely, but wastes some liquid
oxygen.
Note: Closing the valve at the first sign of liquid
discharge does not fill the Sprint/Stroller/Hi Flow
Stroller as completely, but saves liquid oxygen.
9. For the push-on style connectors, push therelease button down until the units separate.
For all other connectors rotate the unit until it
separates from the stationary unit.

Note: Continued on step 6.
Note: For push-on style connector, hold the unit inplace throughout the fill.
6. Open the vent valve to begin filling your Sprint/
Stroller/Hi Flow Stroller. You should hear someoxygen escaping; do not worry. The hissing soundyou hear is perfectly normal.

Portable Release Button (push-on style only)

PN MN236-C4 E | User Manual US - 7

Sprint / Stroller / Hi Flow Stroller
Caution: Check liquid level gauge only after
the valve is closed.

Operating Instructions

Caution: If the Sprint/Stroller/Hi Flow Stroller isbeing refilled immediately after a period of use, fillit, and then wait 10 minutes after filling with the FCV
set at zero for the pressure to stabilize. Then operate
as normal.
WARNING: DO NOT OPERATE SPRINT/
STROLLER/HI FLOW STROLLER WHILE
ATTACHED TO THE STATIONARY UNIT.
WARNING: IF THE SPRINT/STROLLER/HI FLOWSTROLLER DOES NOT SEPARATE EASILY, DO
NOT USE FORCE. THE UNITS MAY BE FROZENTOGETHER. LEAVE THE UNITS CONNECTED
AND WAIT UNTIL THE UNITS WARM UP — THEN
THEY WILL SEPARATE EASILY. DO NOT TOUCHANY FROSTED PARTS. WARNING: SHOULD THE
LEAKAGE BE EXCESSIVE TO THE POINT THATA STREAM OF LIQUID IS PRESENT, LEAVE THE
AREA AND CALL YOUR HEALTH CARE PROVIDERIMMEDIATELY.
WARNING: LIQUID OR COLD GASEOUS OXYGENCAN FROSTBITE THE SKIN. OXYGEN WILL RESULTIN INCREASED FIRE HAZARD.
Caution: Should there be any liquid leakage
from the stationary unit or Sprint/Stroller/
Hi Flow Stroller separating the units, set
the Sprint/Stroller/Hi Flow Stroller aside,
ensuring it remains vertical, leave theroom, and call your health care provider
immediately.
Note: To Equipment Provider: The following oxygen
administration accessories are recommended for use
with the Sprint / Stroller / Hi Flow Stroller:• Nasal Cannula: CAIRE Part Number 5408-SEQ• Firebreak: CAIRE Part Number 21126636
A firebreak is recommended for use with any cannula.
• CAIRE offers a firebreak intended to be used inconjunction with the oxygen portable. The firebreak is
a thermal fuse to stop the flow of gas in the event thatthe downstream cannula or oxygen tubing is ignited
and burns to the firebreak. It is placed in-line with thenasal cannula or oxygen tubing between the patient
and the oxygen outlet of the Sprint / Stroller / Hi Flow
Stroller. For proper use of the firebreak, always referto the manufacturer’s instructions (included with eachfirebreak kit).
• Additional recommended accessories information isavailable online at www.caireinc.com.

8 - US PN MN236-C4 E | User Manual(Cannula is not included.)

1. Push the cannula breathing tube firmly on to theoxygen tube connector.
2. Adjust your cannula breathing tube to the proper
position so that you will be able to breathe theoxygen comfortably.
3. Turn the flow control knob on top of the Sprint/
Stroller/Hi Flow Stroller clockwise until theprescribed flow rate (numeral) is visible in theknob “window” and a positive detent is felt.
Caution: Out-of-specification oxygen flow
will result if the flow control knob is set
between flow rates.
4. You should be receiving oxygen now. To stop
flow of oxygen, turn the flow control knob
counterclockwise to the off (0) position.
Caution: Always turn the flow control knob
off (0) when not in use.

Sprint / Stroller / Hi Flow Stroller
5. Use the following chart as a guideline todetermine the length of time the Sprint/Stroller/
Hi Flow Stroller will operate:
FCV Setting
Model

SprintStroller

Hi Flow
Stroller

NominalNominalFCVPos.

SprintStroller

Hi Flow
Stroller

Off

30:00

60:00

71:00

0.25

18:00

30:00

NA

0.5

12:12

23:00

32:00

0.75

9:18

17:24

NA

1

7:30

14:06

16:00

1.5

5:30

10:12

NA

2

4:18

8:00

8:00

2.5

3:36

6:36

6:20

3

3:06

5:36

5:20

4

2:24

4:24

4:00

5

2:00

3:36

NA

6

1:42

3:00

2:40

8

N/A

N/A

2:00

10

N/A

N/A

1:40

12

N/A

N/A

1:20

15

N/A

N/A

1:00

• Gen 4 Gauge: Press the green operate buttonon the face of the gauge. Read the arc of LEDs,
which indicates content level.

Caution: The Sprint/Stroller/Hi Flow Stroller
is empty if only the last segment of the lightbar is lit.
If the Low Battery indicator lights up when
the button is depressed, notify your health careprovider.
Note: The Sprint/Stroller/Hi Flow Stroller will continue
to supply oxygen even if the Low Battery Indicatorlights up, as long as there is liquid oxygen in the unit.

Note: Times are in hours and minutes(format 00:00).

6b. To verify the level of liquid oxygen in unit withthe mechanical liquid level gauge:

Note: The “Nominal” times are for ideal conditions,i.e. maximum fill, exact flow rates, good loss
rate, unit not being moved, etc.

• Support the unit, by the fabric strap, directly
above the black contents indicator spring scalechamber.

These times are the maximum expected.

• Allow the unit to stabilize (i.e. not bounce up anddown), then read the exposed colored scale forapproximate liquid oxygen contents.

Note: Your individual results will vary.
6a.To verify the level of liquid oxygen in the unit
with the electronic liquid level gauge:• Gen 3 Gauge: Depress the push button (liquid
level switch) on top of the unit for two seconds
minimum. Read across the top of the light bar toindicate contents level.

PN MN236-C4 E | User Manual US - 9

Sprint / Stroller / Hi Flow Stroller
Note: If the unit is empty, only the red colored band of
the scale will be exposed. If this is the case, the unit
must be filled prior to use.
7. Under certain environmental conditions, with
continuous use, and especially with the portable
not being moved, it may develop excessive frost
around the vaporizer coil and on the case. You
may reduce this frost by tapping the unit and/or
wiping any accumulated frost off the case.

Condensate Pad Maintenance
After the unit is empty and has warmed up to
room temperature, remove any moisture from thecondensation cup:
For hard case
• Tilt the unit to one side. Grasp the knob at thecenter of the condensation cup (at the bottomof the unit) and pull straight out. This will
release the cup.
Note: If heavily soiled, the pad can be machine washed(with bleach to disinfect).
• To replace the cup, center it over the roundopening at the bottom of the unit and press
the knob in one click to engage the cup andthen a second click to lock the cup in place.

For soft case
• Unzip the bottom zipper and pull back thebottom cover.
• Remove the pad from the cup and wring out
any absorbed moisture. Allow the pad to drycompletely before replacing it in the cup.
Note: If heavily soiled, the pad can be machine washed(with bleach to disinfect).
• To replace, simply insert pad into bottom
cover and rezip the bottom zipper.
Note: If the knob will not push in and engage the cup,
hold the cup and pull the knob out until it clicks into
the open position. Now replace the cup as previously
described.

Frost Reduction
Under certain environmental conditions, with
continuous use, and especially with the portable
not being moved, it may develop excessive frost
around the vaporizer coil and on the case. You may
reduce this frost by tapping the unit and/or wiping
any accumulated frost off the case.
WARNING: DUE TO THE POSSIBILITY OF
EXCESSIVE MOISTURE / ICE BUILD UP IT
IS RECOMMENDED THAT THE HI FLOWSTROLLER REMAIN IDLE FOR 1 HOUR
AFTER USE TO DECREASE MOISTURE
BUILDUP.

LockedEngaged
Open

Knob Cup

Note: If the knob will not push in and engage the cup,
hold the cup and pull the knob out until it clicks into
the open position. Now replace the cup as previously
described.

10 - US PN MN236-C4 E | User ManualDue to the higher flow rates associated with the Hi
Flow Stroller, condensation and some frosting will
occur on the unit, particularly on flow settings of
10 LPM and higher. It is recommended that the unit
be allowed at least 1 hour of inactivity after use to
decrease the possibility of excessive build up due tocontinuous usage.

Troubleshooting

Sprint / Stroller / Hi Flow Stroller

The following information is intended to help you troubleshoot and solve simple operational problems thatyou may experience when using your Sprint/Stroller/Hi Flow Stroller.
Issue
The Portable makes ahissing sound.

The Portable flow stops
during use.

Solution• Hissing can occur to maintain the correct operating pressure within the Portable.
It is most likely to hiss after filling or when the position of the Portable is changed.
Hissing can last for approximately 10 minutes after filling. Additionally, it can occur
when the flow control valve is at a low setting.
• If the portable has been laid in an improper position return the portable to an uprightposition and allow several minutes for the unit to stabilize• Ensure that the cannula is firmly attached to the Oxygen Outlet.
• Ensure that the cannula is not kinked.
• Check the contents indicator/level gauge and fill the portable if needed.

The Portable does not fill.

The Portable vent valve
does not close properlyat the end of the filling
process.

The Portable does notdisengage easily from theReservoir after filling.

• Ensure the flow control knob is not in the off (“0”) position.
• Verify that there is oxygen in the Reservoir.
• Ensure that the Portable and Reservoir fill connectors are fully engaged throughout
the filling process.
• If the vent valve fails to close and the hissing sound and oxygen vapor cloudcontinue, carefully remove the Portable by depressing the release button on the
Reservoir. Venting from the bottom of the Portable will stop in a few minutes. Allow
the unit to warm until you can close the vent valve. The Portable may require as
much as 60 minutes to restore adequate pressure for accurate oxygen flow. If
needed, use an alternate source of oxygen such as a flow control valve attached to
the Reservoir.
• The Portable and Reservoir fill connectors may have become frozen.
• DO NOT USE FORCE. Allow a few minutes for the frozen parts to warm, thendisengage the Portable when the ice has melted. To prevent the units from freezing
together, always wipe the male fill connector on the Reservoir and the female fillconnector on the Portable with a clean, dry cloth before filling.

PN MN236-C4 E | User Manual US - 11

Sprint / Stroller / Hi Flow Stroller
Cleaning Standard
WARNING: CLEAN ONLY AFTER THE UNIT
IS EMPTY.
• Clean using a solution of mild dish washingdetergent and water.
• Apply cleaning solution directly to a lint-free
cloth. Approved cleaners include HydroPure
and HydroKlean. Do not spray cleaners
directly on the Sprint/Stroller/Hi Flow
Stroller.
• Wipe the outside surface with the lint-free
cloth until the outside surface is clean.
Caution: Do not use high temperature andhigh pressure washing equipment to clean
these units.
• Do not get cleaner on any internal
components or valves.
• Allow the unit to dry thoroughly before using.
Note: Note to health care provider – for reprocessing
procedures, see applicable service manual.

MaintenanceYour service provider is responsible for any
maintenance that my be required per the technicalmanual of this device. Call your service provider forany maintenance requirements.
The condensate pad is the only user-serviceable
part. No other parts are user-serviceable.
The expected service life is a minimum of five years.
Clean the fill connectors on both the stationary
and portable units with a clean, dry, lint-free cloth
between each fill to prevent freezing and possible
equipment failure.
Note: Any additional maintenance needed must be
done by a qualified service technician or service
provider.

Accessories

WEEE and RoHS

This symbol is to remind the equipment
owners to return it to a recycling facility atthe end of its life, per Waste Electrical andElectronic Equipment (WEEE) Directive.
Our products will comply with the restriction ofHazardous Substances (RoHS) directive. They will
not contain more than trace amounts of lead orother hazardous material content.

DisposalAlways return Sprint/Stroller/Hi Flow Stroller,
including all components, to your homecare
provider for proper disposal. You can also contact
your local city or town offices for instructions on
proper disposal of the battery.

Transport and Storage
The device should be stored in the upright position,and be well ventilated. Do not allow the device tolie on its side. Humidity up to 95% noncondensing.
Temperatures range from -40°F to 158°F (-40°C
to 70°C).
Operating temperature ranges from 14°F to 104°F(-10°C to 40°C). Relative humidity range from
30% to 75% noncondensing.
Note: The atmospheric pressure range is 700 hPa to1060 hPa (elevation of 10,000 Ft. to -1,000 Ft.).
12 - US PN MN236-C4 E | User Manual

Carrying Cart PN 20765565

Sprint / Stroller / Hi Flow Stroller

SafetyWARNING: PORTABLE RF COMMUNICATIONS EQUIPMENT (INCLUDING PERIPHERALS SUCH AS
ANTENNA CABLES AND EXTERNAL ANTENNAS) SHOULD BE USED NO CLOSER THAN 30 CM (12INCHES) TO ANY PART OF THE SPRINT/STROLLER/HI FLOW STROLLER, INCLUDING CABLESSPECIFIED BY THE MANUFACTURER. OTHERWISE, DEGRADATION OF THE PERFORMANCE OF
THIS EQUIPMENT COULD RESULT.
WARNING: USE OF ACCESSORIES, TRANSDUCERS AND CABLES OTHER THAN THOSE SPECIFIEDOR PROVIDED BY THE MANUFACTURER OF THIS EQUIPMENT COULD RESULT IN INCREASEDELECTROMAGNETIC EMISSIONS OR DECREASED ELECTROMAGNETIC IMMUNITY OF THIS EQUIPMENT
AND RESULT IN IMPROPER OPERATION.
WARNING: USE OF THIS EQUIPMENT ADJACENT TO OR STACKED WITH OTHER EQUIPMENT SHOULD
BE AVOIDED BECAUSE IT COULD RESULT IN IMPROPER OPERATION. IF SUCH USE IS NECESSARY,
THIS EQUIPMENT AND THE OTHER EQUIPMENT SHOULD BE OBSERVED TO VERIFY THAT THEY AREOPERATING NORMALLY.
Caution: Medical Electrical Equipment needs special precautions regarding Electromagnetic
compatibility (EMC) and needs to be installed and put into service according to the EMC information
provided in this manual.
Caution: Portable and mobile radio frequency (RF) communications equipment can affect Medical
Electrical Equipment.

Table 1

Guidance and Manufacturer’s declaration—electromagnetic emissions

The Sprint/Stroller/Hi Flow Stroller is intended for use in the electromagnetic environment specified
below. The customer or the user of the Sprint/Stroller/Hi Flow Stroller should assure that it is used insuch an environment.
Emissions test

ComplianceElectromagnetic environment—guidance

RF emission CISPR 11

Group 1The Sprint/Stroller/Hi Flow Stroller uses RF energy only forinternal function.

Therefore, its RF emissions are very low and are not likely tocause any interference in nearby electronic equipment.
RF emissions CISPR 11
Harmonic emissions
IEC 61000-3-2
Voltage fluctuations/
flicker emissions
IEC 61000-3-3Class BNot applicableNot applicable

The Sprint/Stroller/Hi Flow Stroller is suitable for use in all
establishments, including domestic establishments and those
directly connected to the public low-voltage power supply
network that supplies buildings used for domestic purposes.

PN MN236-C4 E | User Manual US - 13

Sprint / Stroller / Hi Flow Stroller
Table 2*: Recommended separation distances between portable and mobile RF communications equipment
and the Sprint/Stroller/Hi Flow Stroller
The Sprint/Stroller/Hi Flow Stroller is intended for use in an electromagnetic environment in which radiated RF disturbances arecontrolled. The customer or the user of the Sprint/Stroller/Hi Flow Stroller can help prevent electromagnetic interference by maintaining a minimum distance between portable and mobile RF communications equipment (transmitters) and the Sprint/Stroller/Hi Flow
Stroller as recommended below, according to the maximum output power of the communications equipment.
Rated maximum
Separation distance according to frequency of transmitter
output power ofmtransmitterfrom 150 kHz to 80 MHz

from 80 MHz to 800 MHz

from 800 MHz to 2,5 GHz

d=1.2 √P

d=1.2 √P

d=2.3 √PW
0,01
0.12 m0.12 m
0.23 m0,10.38 m0.38 m0.73 m
1
1.2 m
1.2 m
2.3 m
10
3.8 m
3.8 m
7.3 m
100
12 m
12 m
23 m
For transmitters rated at a maximum output power not listed above, the recommended separation distance (d) in meters (m) canbe estimated using the equation applicable to the frequency of the transmitter, where P is the maximum output power rating of thetransmitter in watts (W) according to the transmitter manufacturer.
NOTE 1 at 80 MHz and 800 MHz, the separation distance for the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection fromstructures, objects and people.

* This table is included as a standard requirement for equipment which has been tested to specific testlevels and over specific frequency ranges and been found compliant with regulations.
Table 3: Guidance and Manufacturer’s Declaration—Electromagnetic Immunity
The Sprint/Stroller/Hi Flow Stroller is intended for use in the electromagnetic environment specified below. The customer or the userof the Sprint/Stroller/Hi Flow Stroller should assure that it is used in such an environment
Immunity test
IEC 60601 test level
Compliance level
Electromagnetic environment – guidance±8 kV Contact
Electrostatic discharge ±8 kV Contact
Floors should be wood, concrete or ceramic(ESD)
tile. If floors are synthetic, the relative humidity±2 kV, ±4 kV,±2 kV, ±4 kV,should be at least 30 %.**IEC 61000-4-2±8 kV, ±15kV air±8 kV, ±15kV air
Not applicable
Electrical fast transient/±2 kV for power supply lines DC powered device
burst
Not applicable±1 kV for input/output lines Not applicable
IEC 61000-4-4
No data input/output lines
Surge±1 kV line(s) to line(s)Not applicableNot applicable±2 kV line(s) to earth
DC powered device
IEC 61000-4-5<5 % UT (>95 % dip in UT )for 0,5 cycle
Voltage dips, short in- 40 % UT (60 % dip in UT )terruptions and voltage for 5 cycles
variations on power
supply input lines
70 % UT (30 % dip in UT )IEC 61000-4-11
25 cycles

Not applicable
DC powered device

Not applicable

3A/m
50/60 HzPower frequency magnetic fields should be thatof a typical commercial or hospital environment.

<5 % UT (>95 % dip in UT )for 5 sec
Power frequency(50/60 Hz) magnetic
field
IEC 61000-4-8

3A/m
50/60 HzNOTE UT is the A.C. mains voltage prior to application of the test level.

** This statement indicates that the required testing was performed in a controlled environment and theSprint/Stroller/Hi Flow Stroller are found to be compliant with regulations.
14 - US PN MN236-C4 E | User ManualSprint / Stroller / Hi Flow Stroller
Table 4
Guidance and Manufacturer’s Declaration—Immunity ME Equipment and ME Systems
Guidance and Manufacturer’s Declaration—Immunity
The Sprint/Stroller/Hi Flow Stroller is intended for use in the electromagnetic environment specified
below. The customer or user of the Sprint/Stroller/Hi Flow Stroller should ensure that it is used in suchan environment.
Immunity Test

IEC 60601 Test Level

Compliance Level

Electromagnetic Environment—Guidance

Conducted RF IEC
61000-4-6

3 Vrms6 Vrms (In ISM Bands)
150 kHz to 80 MHz

Not applicable

Portable and mobile RF communications
equipment should be used no closer to any partof the Sprint/Stroller/Hi Flow Stroller, including
cables, than the recommended separation
distance calculated from the equation applicableto the frequency of the transmitter.

Battery powered
device, No SIP/SOP

Recommended separation distanced = 1,2 √P
Radiated RF IEC
61000-4-3

80 MHz to 2.7 GHz

10 V/m80 MHz—2,7 GHz
80 % AM at 1 kHz

d = 1,2 √P
d = 2,3 √P
where P is the maximum output power rating of thetransmitter in watts (W) according to the transmitter
manufacturer and d is the recommended
separation distance in meters (m).
Field strengths from fixed RF transmitters, as
determined by an electromagnetic site surveya,
should be less than the compliance level in eachfrequency rangeb.
Interference may occur in the vicinity ofequipment marked with the following symbol:

Test frequency(MHz)

Banda)(MHz)

Servicea)

Modulationb)

385

380–390TETRA 400450

430–470

GMRS 460, FRS 460

Pulse modulationb)
18 Hz
FMC)
±5 kHz deviation
1 kHz sine

Maximum
power (W)Distance(m)

Immunity Test
Level (V/m)

1.8

0.3

27

2

0.3

28

710
745
780
810
870
930
1720
1845
1970

704–787

LTE Band 13, 17

Pulse modulationb)
217 Hz

0.2

0.3

9

800–960

GSM 800/900, TETRA
800, IDEN 820, CDMA
850, LTE Band 5

Pulse modulationb)
18 Hz

2

0.3

28

1700–
1900

GSM 1800; CDMA 1900;
GSM 1900; DECT; LTEBand 1, 3, 4, 25; UMTS

Pulse modulationb)
217 Hz

2

0.3

28

2450

2400–
2570

Pulse modulationb)
217 Hz

2

0.3

28

5240
5500
5785

Bluetooth, WLAN, 802.11
b/g/n, RFID 2450, LTE
Band 7

5100–
5800

WLAN 802.11 a/n

Pulse modulationb)
217 Hz

0.2

0.3

9

Note: If necessary to achieve the IMMUNITY TEST LEVEL, the distance between the transmitting antenna and theSprint/Stroller/Hi Flow Stroller may be reduced to 1m. The 1m test distance is permitted by IEC 61000-4-3.
For some services, only the uplink frequencies are included.
The carrier shall be modulated using a 50% duty cycle square wave signal.
As an alternative to FM modulation, 50% pulse modulation at 18 Hz may be used because while it does not represent actual modulation, it
would be worst case.
abc

PN MN236-C4 E | User Manual US - 15

Sprint / Stroller / Hi Flow Stroller
Explication des symboles
ISO 7000

Ne pas fumer à proximité de l’appareilou pendant son fonctionnement.
Reg. # P002
Pièce appliquée de type BF (degré deprotection contre les chocs électriques).
Reg. # 5333

Maintenir à l’abri de l’humidité.
Reg. # 0626
Limite d’empilement par numéro. Reg.
# 2403
Nom et adresse du fabricant.
Reg. # 3082

Avertissement. Reg. # W001

Directive du Conseil européen 93/42/CEE
Représentant agréé au sein de l’Union
européenne
Si l’étiquette de l’identifiant uniquedu produit (UDI) du produit porte
le symbole CE ####, l’appareil est
conforme aux exigences de la directive
93/42 / CEE concernant les dispositifs
médicaux. Le symbole CE #### indiquele numéro de l’organisme notifié.

Le pays et la date de fabrication. Le «CC » identifie le code de pays à deux
lettres du pays de fabrication. La date defabrication est au format AAAA-MM-JJ.
Rég. # 6049
Attention, consulter la documentation
fournie. Reg. # 0434A
Référence catalogue. Reg. # 2493

Cet appareil est conforme aux exigencesde la Directive 2010/35/CEE relative
aux dispositifs médicaux. Il porte lemarquage pi, comme illustré.

Numéro de série. Reg. # 2498
Température de stockage ou defonctionnement. Nº enreg. 0632

ADR
Gaz non-toxique.

Plage d’humidité de stockage. Nº enreg.
2620

Substances oxydantes dangereuses:
risque d'intensification d'incendie. Lesagents oxydants font brûler les incendies
plus vigoureusement.
Liquide réfrigéré, USP ; produit par laliquéfaction de l’air.

Limite de température atmosphérique.
Nº enreg. 2621
Haut. Reg. # 0623
Fragile, manipuler avec soin.
Reg. # 0621

Symboles supplémentaires

Assurer une ventilation correcte del’unité en permanence

Contient des substances dangereuses.
Rég. # 3723

Tenir éloigné des matériaux, huiles etgraisses inflammables.

Importateur. Rég. # 3725

ISO 7010

Des gelures peuvent survenir au contact
d’un liquide froid, de l’oxygène gazeux
ou d’éléments givrés. Avertissement
basse température. Pour signaler lestempératures basses ou les conditions
de gel. Reg. # W010
Le manuel d’instruction doit être lu.
Reg. # M002
Tenir à l’écart des flammes nues, du feu,
des étincelles. Source d’incendie ouverteinterdite et interdiction de fumer.
Reg. # P003

16 - FR PN MN236-C4 E | Manuel d’utilisation

Représentant autorisé en Suisse.

CEI 60417

Si l'appareil porte la marque UKCA
comme indiqué avec UKCA####indiquant le numéro de l'organisme
notifié, cet appareil est conforme auxréglementations UKCA.
Ne couvrez pas l'appareil et ne letransportez pas sous vos vêtements. Ces
unités évacuent normalement l'oxygène.
N° 5641

Sprint / Stroller / Hi Flow Stroller
21 CFR 801.15

CEI 60601-1

La loi fédérale restreint cet appareil àla vente par, ou sur ordonnance d’unmédecin.

ISO 15223-1
Dispositif médical. Rég. # 5.7.7
Identifiant unique de l'appareil # 5.7.10

Preuve d'égouttement

Directive 2012/19/UE du Conseil européen

DEEE
Ce symbole rappelle aux propriétairesd'équipements qu'ils doivent le retournerà une installation de recyclage à la fin
de sa durée de vie, conformément à ladirective relative aux déchets d'équipements électriques et électroniques(DEEE).
Nos produits seront conformes à la restriction de la directive sur les substances
dangereuses (RoHS). Ils ne contiendront
pas plus que des traces de plomb ou
d'autres matières dangereuses.

Ce produit peut être protégé par un ou plusieurs
brevets américains ou internationaux. Consultez notresite Web ci-dessous pour voir la liste des brevets.
Brevet : www.caireinc.com/corporate/patents/.

Caractéristiques techniques• Mode de fonctionnement : Débit constant• Type de protection contre les chocs électriques : Équipement alimenté en interne• Degré de protection contre les chocs électriques : Pièce appliquée de Type BF• Classification par rapport au degré de protection contre la pénétration de l’eau : IP22 – Équipement ordinaire• Équipement inadapté pour une utilisation en présence de produits inflammables

Caractéristiques techniques

SprintStroller

Hi Flow Stroller

Capacité en oxygène liquide

0,68 kg (1,5 lb)

1,36 kg (3 lb)

1,36 kg (3 lb)

Capacité équivalente en gaz
Poids, à vide513 l
2,04 kg (4,5 lb)

1 026 l2,27 kg (5 lb)

1 025 l2,49 kg (5,5 lb)

Poids, plein

2,72 kg (6 lb)

3,63 kg (8 lb)

3,86 kg (8,5 lb)

Hauteur

298 mm (11,75 po)

343 mm (13,5 po)

343 mm (13,5 po)

Dimensions
Autonomie type à 2 l/min

123,83 mm (4,875 po) P x
191 mm (7,5 po) L4,3 h149,23 mm (5,875 po) P x
191 mm (7,5 po) L
8,5 h

149 mm (5,875 po) P x
191 mm (7,5 po) L
8,5 h

Pression de fonctionnement

137 kPa (20 psi)

137 kPa (20 psi)

137 kPa (20 psi)

Taux d’évaporation normal

0,57 kg/jour (1,3 lb/jour)

0,57 kg/jour (1,3 lb/jour)

0,57 kg/jour (1,3 lb/jour)Plage de réglage du débit
standard
Exactitude du débit*Éteint, 0,25; 0,5; 0,75; 1; 1,5; Éteint, 0,25; 0,5; 0,75; 1; 1,5; Éteint, 0,5, 1, 2, 2,5, 3, 4, 6, 8,
10, 12, 15 l/min
2; 2,5; 3; 4; 5; 6 l/min
2; 2,5; 3; 4; 5; 6 l/min+/- 10 %+/- 10 %+/- 10 %* Ce degré de précision s’applique uniquement à 21 °C (70 °F) et 101,4 kPa (14,7 psi) et avec un débitmètre massique calibré avec précision.

PN MN236-C4 E | Manuel d’utilisation FR - 17

Sprint / Stroller / Hi Flow Stroller
Message d’avertissement :Important : Lisez attentivement ce manuel avantd’utiliser l’équipement Sprint/Stroller/Hi Flow Stroller.
Sur ordonnance uniquement.
AVERTISSEMENT : CET APPAREIL N’EST PAS
CONÇU POUR LE MAINTIEN ARTIFICIEL DES
FONCTIONS VITALES.
AVERTISSEMENT : SI VOUS PENSEZ QUE L’APPAREIL
NE FONCTIONNE PAS CORRECTEMENT, PRENEZCONTACT AVEC VOTRE PRESTATAIRE DE SOINS.
N’ESSAYEZ PAS DE RÉPARER OU DE RÉGLER
L’APPAREIL VOUS-MÊME.
AVERTISSEMENT : NE MODIFIEZ PAS CETÉQUIPEMENT SANS L’AUTORISATION DU FABRICANT.
AVERTISSEMENT : SI UN APPROVISIONNEMENTCONTINU EN OXYGÈNE EST NÉCESSAIRE, VEILLEZ ÀCE QUE L’APPORT EN OXYGÈNE SOIT APPROPRIÉ ET/
OU À CE QU’UN APPROVISIONNEMENT EN OXYGÈNE
SECONDAIRE SOIT EN PERMANENCE DISPONIBLE AU
COURS DU TRAITEMENT.
AVERTISSEMENT : NE LAISSEZ PERSONNE FUMERET N’UTILISEZ PAS DE BOUGIES OU DE FLAMMESNUES À MOINS DE 3 M (10 PI) DE L’APPAREIL, OU
À MOINS DE 20 CM (8 PO) DE TOUTE SOURCE
D’INFLAMMATION.
AVERTISSEMENT : INSTALLEZ VOTRE UNITÉ DANSUNE PIÈCE BIEN AÉRÉE.
AVERTISSEMENT : NE STOCKEZ PAS
D’ÉQUIPEMENT D’OXYGÈNE LIQUIDEDANS UN PLACARD, DANS LE COFFRE
D’UNE VOITURE OU DANS UN AUTREENDROIT CONFINÉ. NE PLACEZ PAS DE
COUVERTURES, TENTURES OU D’AUTRESTISSUS SUR L’ÉQUIPEMENT.
AVERTISSEMENT : CE PRODUIT PEUT VOUSEXPOSER À DES PRODUITS CHIMIQUES, Y COMPRISLE NICKEL, RECONNU PAR L’ÉTAT DE CALIFORNIECOMME ÉTANT CANCÉRIGÈNE. POUR PLUS
D’INFORMATIONS, VEUILLEZ CONSULTER LE SITE
WWW.P65WARNINGS.CA.GOV.
AVERTISSEMENT: EN CAS D'INCIDENT GRAVESURVENANT AVEC CET APPAREIL, L'UTILISATEURDOIT IMMÉDIATEMENT SIGNALER L'INCIDENT AU
FOURNISSEUR ET / OU AU FABRICANT. UN INCIDENTGRAVE EST DÉFINI COMME UNE BLESSURE, LA MORTOU LA POSSIBILITÉ DE CAUSER DES BLESSURES OU
LA MORT EN CAS DE RÉAPPARITION DE L'INCIDENT.
L'UTILISATEUR PEUT ÉGALEMENT SIGNALER
L'INCIDENT À L'AUTORITÉ COMPÉTENTE DU PAYS OÙL'INCIDENT S'EST PRODUIT.
Mise en garde : Utilisez l’appareil Sprint/
Stroller/Hi Flow Stroller uniquement selon les
instructions du médecin.
Mise en garde : La loi fédérale restreint cet appareil à lavente par, ou sur ordonnance d’un médecin.

18 - FR PN MN236-C4 E | Manuel d’utilisation

Mise en garde : L’unité contient de l’oxygène liquideextrêmement froid, à presque -184 °C (-300 °F).
L’exposition à une température aussi basse peut provoquer de sérieuses brûlures.
Mise en garde : Bien qu’ils soient ininflammables,
l’oxygène liquide et l’oxygène gazeux sont susceptibles
d’accélérer la combustion d’autres substances. Ce risque ainsi que la basse température de l’oxygène liquide
exigent de prendre certaines précautions de sécurité.
Mise en garde : Éloignez les matériaux inflammables del’équipement. Les aérosols, les huiles et les graisses, ycompris les crèmes pour le visage et la gelée de pétrole,s’enflamment facilement et peuvent brûler rapidement
en présence d’oxygène.
Mise en garde : Fumer lors du port d’une canule à
oxygène peut provoquer des brûlures au visage oumême la mort.
Le fait de retirer la canule et de la placer sur unvêtement, un drap, un sofa ou toute autre matière textile
peut provoquer un embrasement instantané en cas
d’exposition à une cigarette, à une source de chaleur, à
une étincelle ou à une flamme.
Si vous fumez, veuillez : (1) arrêter l’unité portable,(2) retirer la canule et (3) quitter la pièce dans laquelle setrouve cet appareil.
Mise en garde : Si l’unité bascule accidentellement, larelever immédiatement, mais prudemment, en position
verticale, si possible. Si de l’oxygène liquide s’échappe,
quittez la pièce immédiatement et appelez votre prestataire de soins. N’essayez pas de déplacer l’unité ou
d’arrêter la fuite d’oxygène liquide.
Remarque : Ne touchez pas les éléments givrés.
Remarque : Ne rangez pas ou n’utilisez pas l’unité portable en la laissant connectée à l’unité fixe.
Remarque : Ne laissez pas des personnes non forméesmanipuler ou utiliser cet appareil.
Remarque : L’administration fédérale de l’aviation interdit
l’usage de cet appareil sur les vols commerciaux et dansles avions cargo.

Utilisation conforme

Le concentrateur d'oxygène CAIRE Sprint/Stroller/Hi
Flow Stroller supplémentaire. L'appareil n'est pas conçu
pour le maintien en vie et ne fournit aucune capacité desurveillance des patients.

Sprint / Stroller / Hi Flow Stroller
Introduction

Réglages

L’appareil Sprint, Stroller et Hi Flow Stroller viseà fournir un apport d’oxygène supplémentaire aupatient à domicile et peut être utilisé également dansdes établissements comme les maisons de soins
infirmiers ou les centres de soins continus. L’appareil
Sprint/Stroller/Hi Flow Stroller constitue une sourced’oxygène mobile sur une période prolongée.

1. Indicateur de niveau de liquide électronique
2. Commutateur de niveau de liquide électronique
3. Indicateur de niveau de liquide mécanique
4. Bouton de réglage du débit
5. Connecteur du tube à oxygène
6. Vanne de ventilation
7. Connexions de remplissage (voir page suivantepour l’identification du QDV)
8. Sangle de transport avec bandoulière matelassée
9. Coussinet anti-condensation
10. Coupelle de condensation

L’appareil est utilisé par les patients atteints de BPCO
(Bronchopneumopathie chronique obstructive) ou
dont la capacité respiratoire est diminuée. L’appareil
est prescrit au patient. L’appareil est vendu à unfournisseur formé à l’utilisation de l’appareil Sprint/
Stroller/Hi Flow Stroller. Le fournisseur forme
l’utilisateur.

1
2

L’appareil n’est pas conçu pour le maintien desfonctions vitales et ne permet pas non plus desurveiller les patients. Il est recommandé d’avoir une
autre source d’oxygène d’appoint en cas de pannemécanique. Ces appareils portables sont égalementéquipés d’une vanne à déconnexion rapide qu’il est
possible de coupler à un réservoir pour transvaser
l’appareil portable. Le réservoir est rempli par votre
prestataire de soins.

4

5
8

Ce manuel d’utilisation contient les instructions
d’utilisation de l’équipement Sprint/Stroller/Hi Flow
Stroller. Consultez le manuel d’utilisation fourni avec
l’unité fixe pour des informations concernant son
fonctionnement.
Remarque : le prestataire de services apportera son aidelors du paramétrage initial et fournira une formation sur la
manipulation et l’utilisation appropriées de l’unité.
5

7

6
Appareils Sprint, Stroller et Hi Flow Stroller

10
9

PN MN236-C4 E | Manuel d’utilisation FR - 19

Sprint / Stroller / Hi Flow Stroller
Identification de la vanne àdéconnexion rapide
Connecteur
portable

Connecteur
fixeverticale sur le connecteur de l’unité fixe.

Type de vanne àdéconnexion rapide
CAIRE Remplissage
latéralVerrouillage
SF
4. a. Si votre connecteur est de type poussée :abaissez l’unité Sprint/Stroller/Hi Flow Stroller
dans l’unité fixe jusqu’à l’enclenchement duconnecteur.

LIFE-OX
Verrouillage
TF
PENOX

TAEMA
Poussée TF
PB

b. Si le connecteur est de type Penox ou Life-Ox :faites pivoter l’unité Sprint/Stroller/Hi Flow
Stroller jusqu’à l’enclenchement du connecteur.
Faites ensuite pivoter soigneusement etfermement les appareils Sprint/Stroller/Hi Flow
Stroller dans le sens des aiguilles d’une montre() tout en appuyant vers le bas.

Instructions d’utilisation
Pour un remplissage par le haut du Sprint/
Stroller/Hi Flow Stroller

Remarque : suite à l’étape 6 dans Pour un remplissage
latéral du Sprint/Stroller/Hi Flow Stroller.

1. Nettoyez les connecteurs de remplissage sur l’unité
fixe et les appareils Sprint/Stroller/Hi Flow Stroller
avec un chiffon propre, sec et non pelucheux.

Pour un remplissage latéral du Sprint/
Stroller/Hi Flow Stroller

Mise en garde : Les connecteurs deremplissage doivent être secs, car l’humidité
peut entraîner le gel des appareils Sprint/
Stroller/Hi Flow Stroller et de l’unité fixe.
2. Assurez-vous que les boutons de réglage du débit
des appareils Sprint/Stroller/Hi Flow Stroller et duréservoir sont en positions d’arrêt (0).
3. Maintenez le connecteur de remplissage en position

20 - FR PN MN236-C4 E | Manuel d’utilisation

1. Nettoyez les connecteurs de remplissage sur l’unité
fixe et les appareils Sprint/Stroller/Hi Flow Stroller
avec un chiffon propre, sec et non pelucheux.
AVERTISSEMENT : LES CONNECTEURS DE
REMPLISSAGE DOIVENT ÊTRE SECS, CAR
L’HUMIDITÉ PEUT ENTRAÎNER LE GEL DESAPPAREILS SPRINT/STROLLER/HI FLOWSTROLLER ET DE L’UNITÉ FIXE.
2. Assurez-vous que les boutons de réglage du débit
des appareils Sprint/Stroller/Hi Flow Stroller et duréservoir sont en positions d’arrêt (0).

Sprint / Stroller / Hi Flow Stroller
3. Assurez-vous que le connecteur de remplissage
du Sprint/Stroller/Hi Flow Stroller est en position
verticale vers le connecteur fixe.

Remarque : les relevés de l’indicateur de niveau deliquide sont les plus précis 30 secondes minimum après leremplissage du Sprint/Stroller/Hi Flow Stroller.

4. Faites pivoter l’unité Sprint/Stroller/Hi Flow Stroller
dans le sens contraire des aiguilles d’une montre() jusqu’à l’enclenchement de la broche et dela fente.
7. Il est possible que vous voyiez de la vapeur autourde la connexion. Ceci est normal étant donné lagrande différence de température entre l’oxygène
liquide et l’air chaud ambiant.
8. Lorsque l’unité est pleine, fermez la vanne deventilation. Vous pouvez dire que l’unité est pleinequand le sifflement change de tonalité et que duliquide sort de la conduite d’évent.

5. Faites à nouveau pivoter soigneusement et fermement
l’unité Sprint/Stroller/Hi Flow Stroller en position
verticale. Les unités sont désormais verrouillées
ensemble.

Remarque : fermer la vanne après le déchargement d’un
jet de liquide presque continu remplit presque entièrement
l’unité Sprint/Stroller/Hi Flow Stroller, mais entraîne uneperte d’oxygène liquide.
Remarque : fermer la vanne au premier signe de rejet
liquide ne remplit pas autant le Sprint/Stroller/Hi Flow
Stroller, mais permet d’économiser de l’oxygène liquide.
9. Pour les connecteurs de type poussée, appuyez
sur le bouton de déblocage jusqu’à la séparation
des unités. Pour tous les autres connecteurs, faites
pivoter l’unité jusqu’à ce qu’elle se sépare de l’unité
fixe.

Remarque : suite à l’étape 6.
Remarque : pour le connecteur de type poussée,maintenez l’appareil en place pendant le remplissage.
6. Ouvrez la vanne de ventilation pour commencer
le remplissage de votre Sprint/Stroller/Hi Flow
Stroller. Vous entendrez l’oxygène s’échapper. Ne
vous inquiétez pas. Le sifflement émis est tout à faitnormal.

Bouton de déblocage portable(type poussée uniquement)

PN MN236-C4 E | Manuel d’utilisation FR - 21

Sprint / Stroller / Hi Flow Stroller
Mise en garde : vérifiez l’indicateur de niveau
de liquide uniquement après avoir fermé lavanne.
Mise en garde : si le remplissage du Sprint/Stroller/
Hi Flow Stroller est effectué tout de suite après une
période d’utilisation, remplissez-le, puis attendez
10 minutes avec la VCD réglée sur « 0 » pour que lapression se stabilise. Utilisez-le ensuite normalement.
AVERTISSEMENT : N’UTILISEZ PAS LE
SPRINT/STROLLER/HI FLOW STROLLERLORSQU’IL EST FIXÉ À L’UNITÉ FIXE.
AVERTISSEMENT : SI LE SPRINT/STROLLER/HI FLOWSTROLLER NE SE DISSOCIE PAS FACILEMENT,NE FORCEZ PAS. LES UNITÉS PEUVENT ÊTRESOUDÉES ENSEMBLE PAR LE GEL. LAISSEZ LESUNITÉS BRANCHÉES ET ATTENDEZ QU’ELLES SERÉCHAUFFENT. ELLES SE DISSOCIERONT ALORSFACILEMENT. NE TOUCHEZ PAS LES ÉLÉMENTS
GIVRÉS. AVERTISSEMENT : EN CAS DE FUITEIMPORTANTE AVEC UN ÉCOULEMENT DE LIQUIDE,QUITTEZ LA PIÈCE ET APPELEZ IMMÉDIATEMENT LE
PRESTATAIRE DE SOINS.
AVERTISSEMENT : DES GELURES PEUVENTSURVENIR AU CONTACT D’UN LIQUIDE OU
D’OXYGÈNE GAZEUX FROID. L’OXYGÈNE ENTRAÎNE
UNE AUGMENTATION DES RISQUES D’INCENDIE.
Mise en garde : en cas d’écoulement de liquide
de l’unité fixe ou du Sprint/Stroller/Hi Flow
Stroller entraînant une dissociation des unités,éloignez l’unité Sprint/Stroller/Hi Flow Stroller,en vous assurant qu’elle reste à la verticale,quittez la pièce et appelez immédiatement leprestataire de soins.
Remarque: Pour le fournisseur d'équipement: Lesaccessoires d'administration d'oxygène suivants sontrecommandés pour une utilisation avec le Sprint / Stroller/ Hi Flow Stroller:• Canule nasale avec 7 pieds (2,1 m) de tube (LPM maximum: 6 LPM): référence CAIRE 5408-SEQ• Coupe-feu: référence CAIRE 21126636
Un coupe-feu est recommandé pour une utilisation avec
n'importe quelle canule.
• CAIRE propose un coupe-feu destiné à être utilisé avec
le portable d'oxygène. Le coupe-feu est un fusible thermique pour arrêter le flux de gaz dans le cas où la canuleen aval ou le tube d'oxygène est allumé et brûle jusqu'aucoupe-feu. Il est placé en ligne avec la canule nasale ou latubulure d'oxygène entre le patient et la sortie d'oxygène
du Sprint / Stroller / Hi Flow Stroller. Pour une utilisation
correcte du coupe-feu, reportez-vous toujours aux instructions du fabricant (incluses avec chaque kit coupe-feu).
• Des informations supplémentaires sur les accessoires
recommandés sont disponibles en ligne sur www.caireinc.
com.

22 - FR PN MN236-C4 E | Manuel d’utilisation

Instructions d’utilisation(La canule n’est pas fournie.)

1. Poussez fermement le tube respiratoire de la canule
sur le connecteur du tube à oxygène.
2. Réglez le tube respiratoire de la canule sur labonne position de manière à pouvoir respirer
confortablement l’oxygène.
3. Tournez le bouton de réglage du débit situé en
haut du Sprint/Stroller/Hi Flow Stroller dans lesens des aiguilles d’une montre jusqu’à ce que ledébit prescrit (numérique) soit visible sur le bouton« fenêtre » et jusqu’à ressentir une détente positive.
Mise en garde : le débit d’oxygène sera hors
des spécifications si le bouton de réglage dudébit est réglé entre les débits.
4. Vous devriez désormais recevoir l’oxygène. Pourarrêter le débit d’oxygène, tournez le bouton deréglage du débit dans le sens contraire des aiguilles
d’une montre jusqu’à la position arrêt (0).
Mise en garde : mettez toujours le boutonde réglage du débit sur arrêt (0) quand vousn’utilisez pas l’équipement.

Sprint / Stroller / Hi Flow Stroller
5. Référez-vous au tableau suivant pour déterminer ladurée de fonctionnement du Sprint/Stroller/Hi Flow
Stroller :Réglage de la VCD
ModèleSprintStroller

Hi Flow
Stroller

NominalNominalPos.
VCD

SprintStroller

Hi Flow
StrollerÉteint

30:00

60:00

71:00

0.25

18:00

30:00

S/O

0.5

12:12

23:00

32:00

0.75

9:18

17:24

S/O

1

7:30

14:06

16:00

1.5

5:30

10:12

S/O

2

4:18

8:00

8:00

2.5

3:36

6:36

6:20

3

3:06

5:36

5:20

4

2:24

4:24

4:00

5

2:00

3:36

S/O

6

1:42

3:00

2:40

8

S/O

S/O

2:00

10

S/O

S/O

1:40

12

S/O

S/O

1:20

15

S/O

S/O

1:00

Remarque : les durées sont données en heures et en
minutes (format 00:00).
Remarque : les durées « nominales » sont prévues pourdes conditions idéales, c’est-à-dire le remplissage
maximal, les débits exacts, le bon coefficient deperte, le non-retrait de l’unité, etc.
Ces durées correspondent au maximum prévu.
Remarque : vos résultats individuels varieront.
6A. Pour vérifier le niveau de l’oxygène liquide dansl’unité avec l’indicateur de niveau du liquideélectronique :• 	 Indicateur Gen 4 : Appuyez sur le bouton vert del’indicateur. Lisez l’arc des témoins qui indique leniveau du contenu.

Mise en garde : le Sprint/Stroller/Hi Flow
Stroller est vide si seulement le derniersegment de la barre lumineuse est allumé.
Si l’indicateur Batterie faible s’allume lorsque l’on
appuie sur le bouton, contactez votre prestataire desoins.
Remarque : le Sprint/Stroller/Hi Flow Stroller continue àfournir de l’oxygène même si l’indicateur Batterie faibles’allume tant qu’il y a de l’oxygène liquide dans l’unité.
6b. Pour vérifier le niveau de l’oxygène liquide dansl’unité avec l’indicateur de niveau du liquide
mécanique :• Soulevez l’unité par la sangle en tissu directement
au-dessus de la chambre noire à ressort del’indicateur de contenu.
• Stabilisez l’unité (ne la faites pas rebondir), puis lisezl’échelle colorée affichée pour connaître le contenu
approximatif en oxygène liquide.

• Indicateur Gen 3 : Appuyez sur le bouton poussoir(commutateur de niveau de liquide) sur le dessus del’unité pendant deux secondes minimum. Reportezvous à la barre lumineuse pour indiquer le niveau ducontenu.

PN MN236-C4 E | Manuel d’utilisation FR - 23

Sprint / Stroller / Hi Flow Stroller
Remarque : si l’unité est vide, seule la bande rouge del’échelle va s’afficher. Si c’est le cas, l’unité doit êtreremplie avant d’être utilisée.
7. Dans certaines conditions environnementales, enparticulier lors d’une utilisation continue quand
l’unité portable n’est pas déplacée, du givre peut se
former en quantité excessive autour de la spirale duvaporisateur et sur le boîtier. Vous pouvez diminuer
son apparition en tapotant l’unité et/ou en essuyant
le boîtier.

Maintenance du coussinet anticondensation
Une fois l’unité vide et à température ambiante,évacuez l’humidité de la coupelle de condensation :Pour le boîtier rigide• Inclinez l’unité sur un côté. Saisissez le bouton
situé au centre de la coupelle de condensation(en bas de l’unité) et tirez-le vers l’extérieur.
Cette action permet de dégager la coupelle.
Remarque : si le coussinet est très sale, vous pouvez lelaver en machine (et le désinfecter à l’eau de Javel).
• Pour repositionner la coupelle, centrez-la
sur l’ouverture circulaire en bas de l’unitéet appuyez sur le bouton pour enclencher lacoupelle (un clic est émis). Un second clic estémis lors du verrouillage de la coupelle dans sa
position.

Pour le boîtier souple• Ouvrez la fermeture du bas et tirez le couvercle
inférieur.
• Retirez le coussinet de la coupelle et essorez
l’humidité absorbée. Faites sécher entièrement
le coussinet avant de le repositionner dans lacoupelle.
Remarque : si le coussinet est très sale, vous pouvez lelaver en machine (et le désinfecter à l’eau de Javel).
• Pour le repositionner, insérez le coussinet dansle couvercle inférieur et refermez la fermetureinférieure.
Remarque : si le bouton ne s’enfonce pas et n’enclenche
pas la coupelle, maintenez la coupelle et tirez sur lebouton jusqu’à ce qu’un déclic soit émis et que la coupellesoit en position ouverte. Repositionnez à présent lacoupelle tel que décrit précédemment.

Réduction du givre
Dans certaines conditions environnementales, enparticulier lors d’une utilisation continue quand l’unité
portable n’est pas déplacée, du givre peut se former enquantité excessive autour de la spirale du vaporisateur
et sur le boîtier. Vous pouvez diminuer son apparitionen tapotant l’unité et/ou en essuyant le boîtier.
AVERTISSEMENT : EN RAISON DE
LA POSSIBILITÉ D’UNE FORMATIONEXCESSIVE D’HUMIDITÉ/DE GLACE, IL ESTRECOMMANDÉ DE LAISSER LE HI FLOWSTROLLER INACTIF PENDANT 1 HEURE
APRÈS UTILISATION POUR DIMINUER
L’ACCUMULATION D’HUMIDITÉ.

Verrouillée
Enclenchée
OuverteBouton Coupelle

Remarque : si le bouton ne s’enfonce pas et n’enclenche
pas la coupelle, maintenez la coupelle et tirez sur lebouton jusqu’à ce qu’un déclic soit émis et que la coupellesoit en position ouverte. Repositionnez à présent lacoupelle tel que décrit précédemment.

24 - FR PN MN236-C4 E | Manuel d’utilisation

En raison des débits plus élevés associés au Hi Flow
Stroller, de la condensation et des traces de givre
peuvent se former sur l’unité, en particulier avec
des réglages de débit de 10 l/min minimum. Il est
recommandé de laisser l’unité inactive pendant au
moins 1 heure après utilisation pour diminuer cetteaccumulation excessive due à une utilisation continue.

Sprint / Stroller / Hi Flow Stroller
Dépannage
Les informations suivantes permettent de dépanner et de résoudre de simples problèmes de fonctionnement
rencontrés lors de l’utilisation de votre Sprint/Stroller/Hi Flow Stroller.
Problème
L’unité portable émet un
sifflement.

Le débit de l’unité portables’arrête pendant lefonctionnement.

Solution• Un sifflement peut apparaître pour maintenir la pression de fonctionnement appropriéedans l’unité portable. Il est émis généralement après le remplissage ou lorsque l’unité
portable a changé de position. Il peut durer pendant environ 10 minutes après leremplissage. De plus, il peut survenir quand la vanne de réglage du débit est en positionbasse.
• Si l’unité portable a été posée dans une mauvaise position, mettez-la en position verticale
et laissez-la se stabiliser pendant plusieurs minutes.
• Vérifiez que la canule est solidement fixée à l’embout de sortie d’oxygène.
• Assurez-vous que la canule n’est pas pliée.
• Vérifiez l’indicateur de contenu/le niveau de liquide et remplissez l’unité portable sinécessaire.

• Assurez-vous que le bouton de la vanne de contrôle du débit n’est pas en position d’arrêt
(« 0 »).
L’unité portable ne se remplit • Assurez-vous que le réservoir contient de l’oxygène.
pas.
• Assurez-vous que les connecteurs de remplissage de l’unité portable et du réservoir sontbien raccordés pendant la procédure de remplissage.
• Si la vanne de ventilation ne ferme pas et que le sifflement et un nuage de vapeur
La vanne de ventilation ded’oxygène persistent, retirer délicatement l’unité portable en appuyant sur le bouton
l’unité portable ne fermede déblocage présent sur le réservoir. La ventilation provenant du dessous de l’unité
pas correctement à la fin duportable s’arrêtera en quelques minutes. Laissez l’unité se réchauffer jusqu’à ce que vous
remplissage.
puissiez fermer la vanne de ventilation. Il ne faudra pas moins de 60 minutes à l’unité
portable pour rétablir une pression suffisante et assurer un débit d’oxygène approprié.
Si nécessaire, utilisez une autre source d’oxygène comme la vanne de réglage du débit
fixée au réservoir.
L’unité portable ne se libère • Les connecteurs de l’unité portable et du réservoir sont peut-être givrés.
pas facilement du réservoir• NE PAS FORCER. Laissez quelques minutes pour que les parties se réchauffent, puis
après le remplissage.
libérez l’unité portable quand la glace a fondu. Pour éviter que les unités ne gèlent
ensemble, avant de procéder au remplissage, essuyez toujours, à l’aide d’un tissu propre
et sec, le connecteur de remplissage mâle et le connecteur femelle de l’unité portable.

PN MN236-C4 E | Manuel d’utilisation FR - 25

Sprint / Stroller / Hi Flow Stroller
Nettoyage
AVERTISSEMENT : NETTOYEZ UNIQUEMENTAPRÈS QUE L’UNITÉ AIT ÉTÉ VIDÉE.
• Nettoyez en utilisant un mélange de liquide
vaisselle doux et d’eau.
• Appliquez directement la solution nettoyante
sur un tissu non pelucheux. Les nettoyants
approuvés incluent HydroPure et HydroKlean.
Ne vaporisez pas directement de produits
nettoyants sur le Sprint/Stroller/Hi Flow Stroller.
• Essuyez la surface externe avec un tissu non
pelucheux jusqu’à ce qu’elle soit propre.
Mise en garde : n’utilisez pas d’équipement
de lavage à haute température et à pressionélevée pour nettoyer ces unités.
• N’utilisez pas de nettoyant sur les composantsinternes ou les vannes.
• Laissez sécher complètement l’unité avant
l’utilisation.
Remarque : remarque à l’intention du prestataire de soins :pour connaître les procédures de retraitement, consultez lemanuel technique applicable.

DEEE et RoHS

Remarque : la plage de la pression atmosphérique va
de 700 hPa à 1 060 hPa (altitude de 3 048 m à-305 m[10 000 pi à -1 000 pi]).

Entretien
Le prestataire de services est responsable de toute
maintenance qui pourrait être nécessaire selon lemanuel technique de cet appareil. Appelez le prestatairede services en ce qui concerne toute exigence demaintenance.
Le coussinet anti-condensation est l’unique pièce quinécessite une intervention de l’utilisateur. Aucune autre
pièce ne nécessite une intervention de l’utilisateur.
La durée de vie estimée est d’au moins cinq ans.
Entre chaque remplissage, nettoyez les connecteurs deremplissage sur les unités fixe et portable, à l’aide d’untissu non pelucheux, propre et sec, afin d’éviter le gel etles défaillances éventuelles de l’équipement.
Remarque : tout entretien supplémentaire doit être réalisépar un technicien ou un prestataire de services agréé.

Accessoires

Ce symbole a pour objet d’inciter lepropriétaire de l’équipement à le retourner à
un centre de recyclage à la fin de son cycle de vie,conformément à la directive relative aux déchetsd’équipements électriques et électroniques (DEEE).
Nos produits sont conformes à la Directive sur lalimitation des substances dangereuses (RoHS). Ils ne
contiendront pas plus qu’une quantité infime de plombou d’autres substances dangereuses.

Élimination
Retournez toujours l’équipement Sprint/Stroller/
Hi Flow Stroller, y compris tous les composants,
à votre organisme de soins de santé pour qu’il soit
correctement éliminé. Vous pouvez également
contacter vos services municipaux locaux pour obtenir
des instructions concernant l’élimination appropriée dela batterie.

Transport et stockage
L’appareil doit être stocké en position verticale etbien aéré. Ne laissez pas l’appareil posé sur le côté.
Humidité atteignant jusqu’à 95 % sans condensation.
Plage de température : de -40 °C à 70 °C (-40 °F à
158 °F).
La plage de température de fonctionnement va de-10 °C à 40 °C (14 °F à 104 °F). La plage d’humidité
relative va de 30 % à 75 % sans condensation.

26 - FR PN MN236-C4 E | Manuel d’utilisation

Chariot de transport PN 20765565

Sprint / Stroller / Hi Flow Stroller
Sécurité
AVERTISSEMENT : L’ÉQUIPEMENT DE COMMUNICATION RF PORTABLE (Y COMPRIS LESPÉRIPHÉRIQUES TELS QUE LES CÂBLES D’ANTENNE ET LES ANTENNES EXTERNES) NE DOIT PAS
ÊTRE UTILISÉ À MOINS DE 30 CM (12 POUCES) DE TOUTE PARTIE DE L’APPAREIL SPRINT/STROLLER/HIFLOW STROLLER, Y COMPRIS LES CÂBLES SPÉCIFIÉS PAR LE FABRICANT. SINON, UNE DÉGRADATIONDES PERFORMANCES DE CET ÉQUIPEMENT POURRAIT SE PRODUIRE.
AVERTISSEMENT : L’UTILISATION D’ACCESSOIRES, DE TRANSDUCTEURS ET DE CÂBLES AUTRES QUE CEUXSPÉCIFIÉS OU FOURNIS PAR LE FABRICANT DE L’ÉQUIPEMENT PEUT ENTRAÎNER UNE AUGMENTATION DESÉMISSIONS ÉLECTROMAGNÉTIQUES OU UNE DIMINUTION DE L’IMMUNITÉ ÉLECTROMAGNÉTIQUE DE CETÉQUIPEMENT AINSI QU’UN FONCTIONNEMENT INADÉQUAT.
AVERTISSEMENT : L’UTILISATION DE CET ÉQUIPEMENT ADJACENT OU EMPILÉ AVEC D’AUTRES ÉQUIPEMENTS
DOIT ÊTRE ÉVITÉE CAR CELA POURRAIT ENTRAÎNER UN MAUVAIS FONCTIONNEMENT. SI UNE TELLEUTILISATION EST NÉCESSAIRE, CET ÉQUIPEMENT ET LES AUTRES ÉQUIPEMENTS DOIVENT ÊTRE OBSERVÉSPOUR VÉRIFIER QU’ILS FONCTIONNENT NORMALEMENT.
Mise en garde : l’appareil électro-médical doit faire l’objet de précautions particulières concernant lacompatibilité électromagnétique (CEM) et doit être installé et mis en service conformément aux informations
CEM fournies dans ce manuel.
Mise en garde : les équipements portables et mobiles de communication par radiofréquences (RF) peuvent
affecter le fonctionnement des équipements médicaux électriques.

Tableau 1

Directives et déclaration du fabricant - émissions électromagnétiques
Le système Sprint/Stroller/Hi Flow Stroller est prévu pour une utilisation dans les environnementsélectromagnétiques spécifiés ci-dessous. Le client ou l’utilisateur du système Sprint/Stroller/Hi Flow Stroller
doit s’assurer que celui-ci est utilisé dans un environnement de ce type.
Test d’émissions

Conformité

Environnement électromagnétique - Directives

Émissions RF

Groupe 1

Le système Sprint/Stroller/Hi Flow Stroller utilise del’énergie RF uniquement pour son fonctionnement interne.

CISPR 11Par conséquent, ses émissions RF sont très faibles et ne sontpas susceptibles de provoquer des interférences avec deséquipements électroniques situés à proximité.

Émissions RF CISPR 11

Classe B

Émissions de courant
harmonique
CEI 61000-3-2

Non applicable

Norme CEI 61000-3-3
Fluctuations de tension/
papillotement

Non applicable

Le Sprint/Stroller/Hi Flow Stroller convient à une utilisation
dans tous les établissements, y compris les habitationset les établissements directement connectés au réseau
d’alimentation public basse tension qui alimente les bâtiments
à usage d’habitation.

PN MN236-C4 E | Manuel d’utilisation FR - 27

Sprint / Stroller / Hi Flow Stroller
Tableau 2*: Distances de séparation recommandées entre les équipements de communication
RF mobiles et portables et l’appareil Sprint/Stroller/Hi Flow Stroller
Le Sprint/Stroller/Hi Flow Stroller est prévu pour une utilisation dans un environnement dans lequel les perturbations dues aux
radiofréquences rayonnées sont contrôlées. Le client ou l’utilisateur du système Sprint/Stroller/Hi Flow Stroller peut prévenir lesperturbations électromagnétiques en respectant les distances minimum recommandées ci-dessous pour la séparation entre leséquipements de communication RF portables et mobiles (émetteurs) et l’appareil Sprint/Stroller/Hi Flow Stroller, en fonction de lapuissance maximum de sortie de l’équipement de communication.
Puissance nominale
Puissance nominale de sortie maximum de l’émetteurde sortie maximum deml’émetteur
80 MHz et 800 MHz
800 MHz à 2,5 GHz
150 kHz à 80 MHz

d = 1,2√P

d = 1,2√P

d = 2,3√PW
0,010,12 m0,12 m0,23 m0,10,38 m0,38 m0,73 m
1
1,2 m
1,2 m
2,3 m
10
3,8 m
3,8 m
7,3 m
100
12 m
12 m
23 m
Pour les émetteurs avec une puissance nominale de sortie maximum ne figurant pas dans le tableau ci-dessus, la distance deséparation recommandée (d) en mètres (m) peut être déterminée à l’aide de l’équation applicable à la fréquence de l’émetteur, oùP correspond à la puissance nominale de sortie maximum de l’émetteur en watts W), indiquée par le fabricant de l’émetteur.
REMARQUE 1 : à 80 MHz et 800 MHz, la distance de séparation correspondant à la plage de fréquences supérieure s’applique.
REMARQUE 2 : il est possible que ces directives ne s’appliquent pas à toutes les situations. La propagation électromagnétiquedépend des phénomènes d’absorption et de réflexion induits par les structures, les objets et les personnes.

* Ce tableau est inclus pour servir d’exigence standard pour l’équipement qui a été testé à des niveaux de test
spécifiques et sur des plages de fréquence spécifiques et déclaré conforme aux réglementations.

Tableau 3**: Directives et déclaration du fabricant - Immunité électromagnétique
Le système Sprint/Stroller/Hi Flow Stroller est prévu pour une utilisation dans les environnements électromagnétiques spécifiés
ci-dessous. Le client ou l’utilisateur du système Sprint/Stroller/Hi Flow Stroller doit s’assurer que celui-ci est utilisé dans un environnement de ce type.
Environnement électromagnétique - DirecTest d’immunité
Niveau de test CEI 60601
Niveau de conformité
tives
Les sols doivent être en bois, en ciment ou+- 8 kV Contact+- 8 kV Contact
Décharge électrostacarrelés. Si les sols sont revêtus d’un matériau
tique (DES)+- 2 kV, +- 4 kV+- 2 kV, +- 4 kV
synthétique, le taux d’humidité relative doit êtreCEI 61000-4-2+- 8 kV, +- 15 kV air+- 8 kV, +- 15 kV aird’au moins 30 %.**
Non applicable
Lignes d’alimentation±2 kV pour les lignes
Appareil à alimentatransitoires électriques d’alimentation
tion CC
Non applicable
rapides/Salves±1 kV pour les lignes
Non applicabled’entrée/sortie
CEI 610004-4
Pas de données pour leslignes d’entrée/sortie
Non applicable
Surtension±1 kV entre fils de ligne
Non applicableà alimenta±2 kV entre un fil et la terre Appareil
CEI 61000-4-5
tion CCUT <5 % (baisse >95 %Creux de tension, cou- en UT) pendant 0,5 cycle
pures brèves et varia- UT de 40 % (baisse de 60 %Non applicabletions de tension sur les en UT) pendant 5 cycles
Non applicable
circuits d’alimentation UT de 70 % (baisse de 30 % Appareil à alimentation CCélectriqueen UT) pendant 25 cycles<5 % UT (baisse >95 %IEC 61000-4-11
en UT) pendant 5 sec
Champ magnétique àLes champs magnétiques à la fréquence dula fréquence du réseau 30 A/m
3A/m
réseau doivent correspondre à ceux d’un envi(50/60 Hz)
50/60 Hz50/60 Hzronnement commercial ou médical standard.
IEC 61000-4-8
Remarque : UT correspond à la tension secteur CA avant l’application du niveau de test.

** Cette déclaration indique que le test requis a été effectué dans un environnement contrôlé et que l’équipement
Sprint/Stroller/Hi Flow Stroller a été déclaré conforme aux réglementations.
28 - FR PN MN236-C4 E | Manuel d’utilisation

Sprint / Stroller / Hi Flow Stroller
Tableau 4

Directives et déclaration du fabricant - Immunité - Équipement ME et systèmes ME

Directives et déclaration du fabricant - Immunité
Le système Sprint/Stroller/Hi Flow Stroller est prévu pour une utilisation dans les environnementsélectromagnétiques spécifiés ci-dessous. Le client ou l’utilisateur du système Sprint/Stroller/Hi Flow Stroller
doit s’assurer que celui-ci est utilisé dans un environnement de ce type.
Test d’immunité
Niveau de test CEI 60601
Niveau de conformité
Environnement électromagnétique - Directives
RF conduites
CEI 61000-4-6

3 Vrms6 Vrms (dans les bandes ISM)
150 kHz à 80 MHz

Non applicable

Appareil alimenté à l’aide
d’une batterie, Pas de
SIP/SOPLes équipements de communication RF portables etmobiles ne doivent pas être utilisés à proximité descomposants de l’équipement Sprint/Stroller/Hi Flow
Stroller, y compris les câbles, à une distance inférieureà la distance de séparation recommandée calculée
sur la base de l’équation applicable à la fréquence del’émetteur.
Distance de séparation recommandéed = 1,2 √P

RF rayonnées
CEI 61000-4-3

80 MHz à 2,7 GHz

10 V/m80 MHz - 2,7 GHz
80 % AM à 1 kHz

d = 1,2 √P
d = 2,3 √Poù P correspond à la puissance nominale de sortie
maximum de l’émetteur en watts (W), indiquée par lefabricant de l’émetteur et d correspond à la distance deséparation recommandée en mètres (m).
L’intensité des champs émis par des émetteurs RF fixes,
déterminée par l’étude électromagnétique d’un site a, doitêtre inférieure au niveau de conformité pour chaque plagede fréquences b.
Des interférences peuvent se produire à proximité deséquipements portant le symbole suivant :Fréquence detest (MHz)Bandea)(MHz)

Servicea)

385

380-390TETRA 400450

430-470

GMRS 460, FRS 460

704-787

Bande LTE 13, 17

710
745
780
810
870
930
1720
1845
1970
2450
5240
5500
5785

GSM 800/900,
TETRA 800, IDEN 820,CDMA 850, Bande LTE 5
GSM 1800 ;CDMA 1900 ;
1700-1900
GSM 1900 ; DECT ;Bande LTE 1, 3, 4, 25 ;UMTS
Bluetooth, WLAN,
2400-2570 802.11 b/g/n, RFID 2450,
Bande LTE 7
800-960

5100-5800

WLAN 802.11 a/n

Puissance
maximale (W)Distance (m)Niveau de testd’immunité (V/m)

1,8

0,3

27

2

0,3

28

0,2

0,3

9

Modulation par
impulsionsb)
18 Hz

2

0,3

28

Modulation par
impulsionsb)
217 Hz

2

0,3

28

2

0,3

28

0,2

0,3

9

Modulationb)
Modulation par
impulsionsb)
18 Hz
FMC)
Écart ±5 kHz
Sinus 1 kHz
Modulation par
impulsionsb)
217 HzModulation par
impulsionsb)
217 HzModulation par
impulsionsb)
217 HzRemarque : le cas échéant, pour atteindre le NIVEAU DE TEST D’IMMMUNITÉ, la distance entre l’antenne émettrice et leSprint/Stroller/Hi Flow Stroller peut être réduite à 1 m. La distance de test de 1 m est autorisée par la norme CEI 61000-4-3.
Pour certains services, seules les fréquences de liaisons montantes sont incluses.
Le transporteur doit être modulé à l’aide d’un signal rectangulaire à rapport cyclique de 50 %.
Comme alternative à la modulation FM, une modulation par impulsions de 50 % à 18 Hz peut être utilisée, car si elle ne représente pas une
modulation réelle, il semble que ce soit le pire des scénarios.
abc

CAIRE et CAIRE Inc. sont des marques déposées de CAIRE Inc. Veuillez visiter notre site Web ci-dessous pour obtenir une liste complète des marques de commerce. Marques: www.
careinc.com/corporate/trademarks.
Copyright © 2023 CAIRE Inc. CAIRE Inc. se réserve le droit d’interrompre la commercialisation de ses produits ou de modifier les tarifs, les matériaux, les équipements, les niveaux dequalité, les descriptions, les caractéristiques et/ou les processus de ses produits, et ce à tout moment, sans avertissement préalable et sans aucune autre obligation ni conséquence. Nous
nous réservons tous les droits non expressément stipulés dans le présent document, selon les conditions applicables.

PN MN236-C4 E | Manuel d’utilisation FR - 29

Sprint / Stroller / Hi Flow Stroller
Definitionen der Symbole
ISO 7000

Nicht in der Nähe des Geräts oderwährend dessen Betrieb rauchen.
Reg.-Nr. P002

Vor Regen schützen, trocken
aufbewahren. Reg.-Nr. 0626

Anwendungsteil des Typs BF(Schutzgrad gegen Stromschlag).
Reg.-Nr. 5333

Stapelbegrenzung nach Anzahl. Reg.
# 2403Name und Adresse des Herstellers.
Reg.-Nr. 3082

Warnung. Reg.-Nr. W001
Richtlinie 93/42/EWG des Rates
Autorisierter Vertreter in derEuropäischen Gemeinschaft

Herstellungsland und -datum. Das„CC“ kennzeichnet den aus zwei
Buchstaben bestehenden Ländercodedes Herstellungslandes. Das
Herstellungsdatum hat das Format JJJJMM-TT. Reg.-Nr. # 6049
Vorsicht, Begleitdokumente beachten.
Reg. # 0434A

Wenn auf dem UDI-Etikett (ProductUnique Device Identifier) das CE ####
-Symbol angebracht ist, entspricht das
Gerät den Anforderungen der Richtlinie
93/42 / EWG für Medizinprodukte. Das
CE #### -Symbol zeigt die Nummer desangemeldeten Körpers an.
Dieses Gerät entspricht den Vorgaben
der Richtlinie 2010/35/EU für
Medizinprodukte. Es trägt die
Pi‑Kennzeichnung wie dargestellt.

Katalognummer. Reg.-Nr. 2493
Seriennummer. Reg.-Nr. 2498
Lager- oder Betriebstemperaturbereich.
Reg.-Nr. 0632

ADR
Ungiftiges Gas.

Lagerung luftfeuchtigkeit bereich. Reg.Nr. 2620
Luftdruckbegrenzung. Reg.-Nr. 2621
Diese Seite nach oben. Reg.-Nr. 0623

Brandfördernde Stoffe: Brandfördernde
Gefahr. Oxidationsmittel bewirken, dass
Brände stärker brennen.
Tiefkalte Flüssigkeit, USP; Hergestellt
durch Luftverflüssigung
Zusätzliche Symbole
Darauf achten, dass das Gerät immer gut
belüftet ist

Zerbrechlich, Vorsicht bei derHandhabung. Reg.-Nr. 0621
Enthält gefährliche Stoffe. Reg.-Nr. #3723

Von entzündlichen Materialien, Öl und
Fett fernhalten.

Importeur. Reg.-Nr. # 3725

Bevollmächtigter Vertreter in der
Schweiz.

ISO 7010
Bei Berührung mit kaltem flüssigem
oder gasförmigem Sauerstoff bzw.
vereisten Teilen kann es zu Erfrierungenkommen. Warnung: Temperatur niedrig.
Warnt vor niedriger Temperatur oder
Frost. Reg.-Nr. W010
Die Betriebsanleitung ist aufmerksam
durchzulesen. Reg.-Nr. M002
Von Flammen, offenem Feuer undFunken fernhalten. Offene Zündquellen
und Rauchen verboten. Reg.-Nr. P003
30 - DE PN MN236-C4 E | Bedienungsanleitung

IEC 60417

Wenn das Gerät das UKCA-Zeichen
trägt, wie gezeigt, wobei UKCA####
die Nummer der benannten Stelle
angibt, entspricht dieses Gerät den
UKCA-Bestimmungen.
Decken Sie das Gerät nicht ab undtragen Sie das tragbare Gerät nicht
unter Ihrer Kleidung. Diese Einheiten
entlüften normalerweise Sauerstoff. Nr.
5641

Sprint / Stroller / Hi Flow Stroller
21 CFR 801.15
Nach den in den USA geltenden
Gesetzen darf dieses Gerät nur voneinem Arzt oder im Auftrag eines Arztes
verkauft werden.
EN 60601-1
Tropfsicher
Richtlinie 2012/19/EU des Rates
WEEE
Dieses Symbol soll die Gerätebesitzer
daran erinnern, das Gerät am Endeseiner Lebensdauer gemäß der Richtlinieüber Elektro- und Elektronik-Altgeräte(WEEE) einer Recyclingeinrichtung
zuzuführen.
Unsere Produkte erfüllen die Richtlinie
zur Beschränkung gefährlicher Stoffe(RoHS). Sie dürfen nicht mehr als
Spuren von Blei oder anderen gefährlichen Stoffen enthalten.

ISO 15223-1
Medizinisches Gerät. Reg.-Nr. # 5.7.7
Eindeutige Gerätekennung # 5.7.10
Dieses Produkt kann von einem oder mehreren
USamerikanischen oder internationalen Patenten
abgedeckt sein. Auf unserer Webseite finden Sie eine
Auflistung der geltenden Patente. Pat.: www.caireinc.
com/corporate/patents/ .

Technische Daten• Betriebsmodus: Kontinuierlicher Durchfluss• Art des Schutzes gegen elektrischen Schlag: Gerät mit interner Stromversorgung• Grad des Schutzes gegen elektrischen Schlag: Anwendungsteil vom Typ BF• Klassifizierung gemäß dem Schutzgrad gegen das Eindringen von Wasser: IP22 – Gewöhnliches Gerät• Gerät nicht geeignet für die Verwendung in der Nähe von brennbaren Gemischen

Produktdaten

SprintStroller

Hi Flow Stroller

LOX-Kapazität

0,68 kg (1,5 lb)

1,36 kg (3,0 lb)

1,36 kg (3,0 lb)

Gasförmige gleichwertige
Kapazität
Leergewicht

513 l1026 l1025 l

2,04 kg (4,5 lb)

2,27 kg (5,0 lb)

2,49 kg (5,5 lb)

Gewicht in gefülltem Zustand 2,72 kg (6,0 lb)

3,63 kg (8,0 lb)

3,86 kg (8,5 lb)

Höhe

298 mm (11,75 in.)

343 mm (13,5 in.)

343 mm (13,5 in.)Abmessungen123,83 mm (4,875") D x
191 mm (7,5") W
4,3 Stunden

149,23 mm (5,875") D x
191 mm (7,5") W
8,5 Stunden

149 mm (5,875") D x
191 mm (7,5") W
8,5 Stunden

20 psi (137 kPa)

20 psi (137 kPa)

20 psi (137 kPa)

Normale Verdampfungsrate

0,57 kg/Tag (1.3 lb/Tag)

0,57 kg/Tag (1.3 lb/Tag)

0,57 kg/Tag (1.3 lb/Tag)

Standard-Messbereich fürden Durchfluss
Genauigkeit derDurchflussrate*

Aus, 0,25; 0,5; 0,75; 1; 1,5; 2; Aus, 0,25; 0,5; 0,75; 1; 1,5; 2; Aus, .5, 1, 2, 2.5, 3, 4, 6, 8,
10, 12, 15 LPM
2,5; 3; 4; 5; 6 LPM
2,5; 3; 4; 5; 6 LPM+/- 10 %+/- 10 %+/- 10 %Typische Nutzungsdauer
bei 2 L/MIN
Betriebsdruck* Diese Genauigkeit wird nur bei 70 F/21,1 °C und 14,7 psig/1,013 bar und mit einem kalibrierten präzisen Massendurchflussmesser erreicht.

PN MN236-C4 E | Bedienungsanleitung DE - 31

Sprint / Stroller / Hi Flow Stroller
Warnhinweise
Wichtig: Lesen Sie dieses Handbuch vor dem Betrieb
des Sprint/Stroller/Hi Flow Stroller aufmerksam durch.
Verschreibungspflichtig.
WARNUNG: DIESES GERÄT IST NICHT ZUR
VERWENDUNG FÜR LEBENSERHALTENDEMASSNAHMEN VORGESEHEN.
WARNUNG: WENN SIE VERMUTEN, DASS DASGERÄT NICHT ORDNUNGSGEMÄSS FUNKTIONIERT,WENDEN SIE SICH AN IHREN LEISTUNGSERBRINGER.
VERSUCHEN SIE NICHT, DAS GERÄT EIGENSTÄNDIGZU REPARIEREN ODER EINZUSTELLEN.
WARNUNG: NEHMEN SIE OHNE DIE ZUSTIMMUNGDES HERSTELLERS KEINERLEI VERÄNDERUNGEN AM
GERÄT VOR.
WARNUNG: WENN EINE KONTINUIERLICHESAUERSTOFFVERSORGUNG ERFORDERLICH
IST, STELLEN SIE SICHER, DASS WÄHREND DER
THERAPIE JEDERZEIT EINE AUSREICHENDESAUERSTOFFVERSORGUNG UND/ODER EINESEKUNDÄRE SAUERSTOFFVERSORGUNG ZUR
VERFÜGUNG STEHT.
WARNUNG: IN EINEM UMKREIS VON 3 M (10 FUSS)ZUM GERÄT SIND RAUCHEN, KERZEN ODER OFFENESFEUER VERBOTEN ODER EIN ABSTAND NÄHER ALS
20 CM (8 ZOLL) VON EINER ZÜNDQUELLE.
WARNUNG: BEWAHREN SIE DAS GERÄT IN EINEM
GUT DURCHLÜFTETEN BEREICH AUF.
WARNUNG: LAGERN SIE GERÄTE
MIT FLÜSSIGSAUERSTOFF NICHT IM
KOFFERRAUM EINES FAHRZEUGS,
SCHRÄNKEN ODER ANDEREN ENGEN
RÄUMEN. LEGEN SIE KEINE DECKEN,VORHÄNGE ODER ANDERE STOFFE ÜBERDAS GERÄT.
WARNUNG: DURCH DIESES PRODUKT KÖNNEN
SIE MIT CHEMIKALIEN EINSCHLIESSLICH NICKELIN KONTAKT KOMMEN, WELCHES IM STAATKALIFORNIEN ALS KREBSVERURSACHEND GILT.
WEITERE INFORMATIONEN ERHALTEN SIE AUF
WWW.P65WARNINGS.CA.GOV.
WARNUNG: WENN BEI DIESEM GERÄT EIN SCHWERWIEGENDER VORFALL AUFTRITT, SOLLTE DER
BENUTZER DEN VORFALL UNVERZÜGLICH DEM
ANBIETER UND / ODER DEM HERSTELLER MELDEN.
EIN SCHWERWIEGENDER VORFALL IST DEFINIERT ALS EINE VERLETZUNG, DER TOD ODER DIE
MÖGLICHKEIT, EINE VERLETZUNG / DEN TOD ZUVERURSACHEN, FALLS DER VORFALL ERNEUTAUFTRITT. DER BENUTZER KANN DEN VORFALL AUCH
DER ZUSTÄNDIGEN BEHÖRDE DES LANDES MELDEN,
IN DEM DER VORFALL AUFGETRETEN IST.
Vorsicht: Verwenden Sie den Sprint/Stroller/
Hi Flow Stroller nur gemäß den Anweisungen
Ihres Arztes.
Vorsicht: Nach den in den USA geltenden Gesetzen darf
dieses Gerät nur von einem Arzt oder im Auftrag eines
Arztes verkauft werden.
32 - DE PN MN236-C4 E | Bedienungsanleitung

Vorsicht: Das Gerät enthält extrem kalten Flüssigsauerstoff mit einer Temperatur von fast -184 °C. Der Kontakt
mit derart niedrigen Temperaturen kann zu schweren
Erfrierungen führen.
Vorsicht: Flüssiger und gasförmiger Sauerstoff sindselbst zwar nicht brennbar, können aber dazu führen,
dass andere Materialien schneller brennen als normal.
Aufgrund dieser Gefahr und wegen der niedrigen Temperatur von Flüssigsauerstoff sind gewisse Sicherheitsvorkehrungen zu beachten.
Vorsicht: Brennbare Materialien von diesem Gerät
fernhalten. Leicht entflammbare Stoffe, wie Sprays, Öle
und Fette, einschließlich Gesichtscremes und Vaselinekönnen sich in Gegenwart von Sauerstoff schnell
entzünden und brennen.
Vorsicht: Rauchen während des Tragens einer Sauerstoffkanüle kann zu Gesichtsverbrennungen und zum
Tode führen.
Ein Abnehmen der Kanüle und eine Lagerung auf Kleidung, Bettwäsche, Sofas oder anderem Polstermaterial
führt bei Vorhandensein einer Zigarette, Wärmequelleoder Flamme zu einer Verpuffung.
Wenn Sie rauchen sollten Sie folgendes beachten: (1)schalten Sie das Gerät aus, (2) nehmen Sie die Kanüle
ab und (3) verlassen Sie den Raum, in dem sich das
Gerät befindet.
Vorsicht: Sollte das Gerät versehentlich umfallen, bringen Sie (falls möglich) das Gerät sofort, aber vorsichtig,
wieder in eine aufrechte Position. Wenn Flüssigsauerstoff austritt, verlassen Sie den Bereich unverzüglich undwenden Sie sich an Ihren Leistungserbringer. Versuchen
Sie nicht, das Gerät zu bewegen oder das Austreten von
Flüssigsauerstoff zu verhindern.
Hinweis: Berühren Sie keine vereisten Teile an den Geräten.
Hinweis: Lagern oder betreiben Sie die tragbare Einheit
nicht, wenn sie an die stationäre Einheit angeschlossen ist.
Hinweis: Gestatten Sie keinem ungeschulten Personal die
Handhabung oder Bedienung dieses Geräts.
Hinweis: Die Verwendung dieses Geräts auf gewerb­lichen
Passagier- und Transportflügen ist durch die Federal Aviation
Administration (US-Bundesluftfahrtbehörde) eingeschränkt.

Verwendungszweck

Der CAIRE Sprint/Stroller/Hi Flow Stroller ist für die
Verabreichung von zusätzlichem Sauerstoff vorgesehen.
Das Gerät ist weder für die Lebenserhaltung
vorgesehen, noch bietet es Funktionen zur
Patientenüberwachung.

Sprint / Stroller / Hi Flow Stroller
Einführung

Bedienelemente

Sprint, Stroller und Hi Flow Stroller sind für die
Versorgung des Patienten mit Zusatzsauerstoff inseinem Privathaus bestimmt und können auch in
Pflegeheimen oder Einrichtungen für subakute Pflege
verwendet werden. Der Sprint/Stroller/Hi Flow Stroller
stellt über einen längeren Zeitraum eine ambulante
Sauerstoffversorgung bereit.

1. Elektronische Füllstandanzeige
2. Knopf zum Abfragen des Füllstandes
3. Mechanische Füllstandanzeige (Federwaage)
4. Durchflussregler
5. O² Ausgangsstutzen
6. Entlüftungsventil
7. Füllstutzen (es gibt verschiedene Füllstutzen die
auf der nächsten Seite beschrieben sind)
8. Tragegurt mit Schulterpolster
9. Kondenswasserschwamm
10. Kondenswasserschale

Das Gerät wird von COPD-Patienten oder Patienten
mit verminderter Atemkapazität verwendet. Das Gerät
wird dem Patienten ärztlich verschrieben. Das Gerät
wird an einen Leistungserbringer verkauft, der inBedienung und Wartung des Sprint/Stroller/Hi Flow
Stroller geschult ist. Dieser Leistungserbringer schultden Benutzer in der Anwendung.

1

Das Gerät ist nicht für lebenserhaltende Maßnahmen
ausgelegt und bietet keine Patientenüberwachung.
Für den Fall eines mechanischen Versagens empfiehlt
sich eine alternative Quelle der Sauerstoffzufuhr.
Diese tragbaren Geräte verfügen zudem über ein
Schnellabsperrventil, das an einen Lox-Behälter
angeschlossen werden kann, um das tragbare
Gerät zu befüllen. Der Behälter wird von Ihrem
Leistungserbringer befüllt.

2
4
5
8

Diese Bedienungsanleitung enthält die Anweisungenfür die Verwendung des Sprint/Stroller/Hi Flow
Stroller. Einzelheiten zum Betrieb der stationären
Einheit entnehmen Sie bitte der Bedienungsanleitung.
Hinweis: Der Dienstleister unterstützt Sie bei der Inbetriebnahme und weist Sie in die richtige Handhabung undBedienung des Gerätes ein.

57Sprint, Stroller und Hi Flow Stroller

610
9
PN MN236-C4 E | Bedienungsanleitung DE - 33

Sprint / Stroller / Hi Flow Stroller
Zuordnung derSchnellabsperrventile
Anschluss
für tragbare
Einheit

3. 	Bringen Sie den Auffüllanschluss in die aufrechte
Position über dem Anschluss der stationären
Einheit.
Typ des
Schnellabsperrventils

Anschluss für
stationäres
Gerät

CAIRE
Seitenbefüllung

Absperren der
Seitenbefüllung

LIFE-OX
Aufsteckmodell

4. a. Wenn es sich um einen Aufsteckanschluss handelt:
Senken Sie den Sprint/Stroller/Hi Flow Stroller
auf der stationären Einheit ab, bis der Anschluss
spürbar einrastet.

PENOX

TAEMA
Obenbefüllung
PB

b. Wenn der Anschluss vom Typ Penox oder LifeOx ist:
Drehen Sie den Sprint/Stroller/Hi Flow Stroller,
bis der Anschluss spürbar einrastet. Drehen Sie
den Sprint/Stroller/Hi Flow Stroller anschließend
kräftig aber vorsichtig im Uhrzeigersinn ()
und drücken Sie ihn dabei nach unten.

GebrauchsanweisungenBefüllen des Sprint/Stroller/Hi Flow Stroller
mit Obenbefüllung
1. Reinigen Sie die Auffüllanschlüsse sowohl an derstationären Einheit als auch am Sprint/Stroller/Hi
Flow Stroller mit einem sauberen, trockenen und
flusenfreien Tuch.
Vorsicht: Die Auffüllanschlüsse müssen trocken sein, da der Sprint/Stroller/Hi Flow Stroller
und die stationäre Einheit durch Feuchtigkeit
aneinander festfrieren können.
2. Vergewissern Sie sich, dass die Durchflussschalter
sowohl am Sprint/Stroller/Hi Flow Stroller als aucham Behälter auf AUS (0) stehen.

34 - DE PN MN236-C4 E | Bedienungsanleitung

Hinweis: Weiter bei Schritt 6 in „Befüllung von Sprint/
Stroller/Hi Flow Stroller mit Seitenbefüllung“.

Befüllung von Sprint/Stroller/Hi Flow Stroller
mit Seitenbefüllung
1. Reinigen Sie die Auffüllanschlüsse sowohl an derstationären Einheit als auch am Sprint/Stroller/Hi
Flow Stroller mit einem sauberen, trockenen und
flusenfreien Tuch.
WARNUNG: DIE AUFFÜLLANSCHLÜSSE
MÜSSEN TROCKEN SEIN, DA DER SPRINT/
STROLLER/HI FLOW STROLLER UND DIE
STATIONÄRE EINHEIT DURCH FEUCHTIGKEITANEINANDER FESTFRIEREN KÖNNEN.

Sprint / Stroller / Hi Flow Stroller
2. Vergewissern Sie sich, dass die
Durchflussmengenregler sowohl am Sprint/Stroller/
Hi Flow Stroller als auch am Behälter auf AUS (0)stehen.

Hinweis: Die Messwerte der Füllstandanzeige sindmindestens 30 Sekunden nach der Befüllung des Sprint/
Stroller/Hi Flow am genauesten.

3. Führen Sie den Auffüllanschluss vom Sprint/Stroller/
Hi Flow Stroller in der aufrechten Position zum
Anschluss der stationären Einheit.

4. Drehen Sie den Sprint/Stroller/Hi Flow Stroller
gegen den Uhrzeigersinn (), bis der Stift spürbar
in der Nut einrastet.

5. Drehen Sie den Sprint/Stroller/Hi Flow Stroller
kräftig aber vorsichtig zurück in die aufrechte
Position. Die Einheiten sind jetzt fest aneinanderangeschlossen.

7. Möglicherweise ist am Anschluss außerdem
Dampf zu sehen. Dies ist aufgrund desgroßen Temperaturunterschieds zwischen dem
Flüssigsauerstoff und der warmen Umgebungsluft
um die Einheiten herum normal.
8. Schließen Sie das Entlüftungsventil, sobald die
Einheit gefüllt ist. Eine gefüllte Einheit können
Sie daran erkennen, dass das Zischen seinen
Klang verändert und etwas Flüssigkeit aus derEntlüftungsleitung austritt.
Hinweis: Wenn das Ventil geschlossen wird, nachdem
Flüssigkeit in einem nahezu gleichbleibenden Stromausgetreten ist, wird der Sprint/Stroller/Hi Flow Stroller fast
vollständig gefüllt, wobei allerdings etwas Flüssigsauerstoff
verloren geht.
Hinweis: Wenn Sie das Ventil beim ersten Anzeichen
austretender Flüssigkeit schließen, wird der Sprint/Stroller/
Hi Flow Stroller nicht vollständig gefüllt, aber Sie sparen
Flüssigsauerstoff.
9. Drücken Sie bei Modellen mit Aufsteckanschluss
den Entriegelungsknopf, bis sich die Einheiten
voneinander trennen. Bei allen anderen Anschlüssen
drehen Sie die Einheit, bis sie sich von derstationären Einheit löst.

Hinweis: Weiter mit Schritt 6.
Hinweis: Halten Sie die Einheit beim Modell mit
Aufsteckanschluss während der Befüllung gut fest.
6. Öffnen Sie das Entlüftungsventil, um mit dem
Befüllen Ihres Sprint/Stroller/Hi Flow Stroller zu
beginnen. Dabei muss entweichender Sauerstoff zu
hören sein, was aber kein Anlass zur Sorge ist. Daszischende Geräusch ist völlig normal..

Entriegelungsknopf für tragbare Einheit(nur Aufsteckmodell)PN MN236-C4 E | Bedienungsanleitung DE - 35

Sprint / Stroller / Hi Flow Stroller
Vorsicht: Prüfen Sie die Füllstandanzeige nur
bei geschlossenem Entlüftungsventil.

GebrauchsanweisungenVorsicht: Wenn Sie den Sprint/Stroller/Hi Flow
Stroller sofort nach Gebrauch wieder auffüllen,
warten Sie anschließend 10 Minuten, wobei für die
Durchflusseinstellung Null eingestellt ist, damit sichder Druck stabilisieren kann. Verwenden Sie das Gerät
anschließend wie gewohnt.
WARNUNG: NEHMEN SIE DEN SPRINT/
STROLLER/HI FLOW NICHT IN BETRIEB,SOLANGE ER AN DIE STATIONÄRE EINHEITANGESCHLOSSEN IST.
WARNUNG: WENN SICH DER SPRINT/STROLLER/HIFLOW STROLLER NICHT LEICHT TRENNEN LÄSST,WENDEN SIE KEINE GEWALT AN. DIE GERÄTE SINDMÖGLICHERWEISE ZUSAMMENGEFROREN. WARTEN
SIE MIT DEM TRENNEN, BIS SICH DIE EINHEITEN
ERWÄRMT HABEN. ANSCHLIESSEND KÖNNEN
SIE PROBLEMLOS VONEINANDER GETRENNTWERDEN. BERÜHREN SIE KEINE VEREISTEN TEILE.
WARNUNG: WENN DER FLÜSSIGKEITSAUSTRITTSO STARK IST, DASS EIN FLÜSSIGKEITSSTROMVORHANDEN IST, VERLASSEN SIE DEN BEREICH
UND WENDEN SIE SICH UNVERZÜGLICH AN IHRENLEISTUNGSERBRINGER.
WARNUNG: FLÜSSIGER ODER KALTERGASFÖRMIGER SAUERSTOFF KANN ZU
ERFRIERUNGEN DER HAUT FÜHREN. SAUERSTOFF
BEDEUTET EIN ERHÖHTES BRANDRISIKO.
Vorsicht: Sollte beim Trennen aus derstationären Einheit oder dem Sprint/Stroller/
Hi Flow Stroller Flüssigkeit austreten, stellen
Sie den Sprint/Stroller/Hi Flow Stroller
beiseite. Achten Sie darauf, dass er senkrecht
stehen bleibt, verlassen Sie den Raum undkontaktieren Sie sofort Ihren medizinischen
Leistungserbringer.
Hinweis: An den Geräteanbieter: Das folgende Zubehör für
die Sauerstoffverabreichung wird zur Verwendung mit dem
Sprint / Stroller / Hi Flow Stroller empfohlen:• Nasenkanüle mit 7 Fuß (2,1 m) Schlauch (maximale
LPM: 6 LPM): CAIRE-Teilenummer 5408-SEQ• Brandschutz: CAIRE-Teilenummer 21126636
Für die Verwendung mit einer Kanüle wird eine Feuerpause empfohlen.
• CAIRE bietet einen Brandschutz an, der in Verbindung
mit dem tragbaren Sauerstoff verwendet werden soll. Die
Brandschutzklemme ist eine thermische Zündschnur,um den Gasfluss zu stoppen, falls die nachgeschaltete
Kanüle oder der Sauerstoffschlauch entzündet werden
und zur Brandschutzstelle verbrennen. Es wird in Linie mit
der Nasenkanüle oder dem Sauerstoffschlauch zwischen
dem Patienten und dem Sauerstoffauslass des Sprint /Stroller / Hi Flow Stroller platziert. Beziehen Sie sich für die
ordnungsgemäße Verwendung des Brandschutzes immer
auf die Anweisungen des Herstellers (im Lieferumfang
jedes Brandschutz-Kits enthalten).
• Weitere Informationen zu empfohlenem Zubehör finden
Sie online unter www.caireinc.com.
36 - DE PN MN236-C4 E | Bedienungsanleitung(Kanüle ist nicht im Lieferumfang enthalten.)

1. Schieben Sie den Schlauch der Atemkanüle fest auf
den Sauerstoffschlauchanschluss.
2. Stellen Sie den Schlauch Ihrer Atemkanüle
entsprechend ein, sodass Sie bequem den Sauerstoff
einatmen können.
3. Drehen Sie den Durchflussmengenregler oben am
Sprint/Stroller/Hi Flow Stroller im Uhrzeigersinn,
bis die verordnete Durchflussrate (als Zahlenwert)
im Schalterfenster angezeigt wird und ein positives
Einrasten zu spüren ist.
Vorsicht: Ein Sauerstoffdurchfluss
außerhalb der Spezifikation ergibt sich,wenn der Durchflussregler zwischen den
Durchflussraten eingestellt wird.
4. Ihnen sollte jetzt Sauerstoff zugeführt werden.
Stellen Sie den Durchflussmengenregler
gegen den Uhrzeigersinn auf AUS (0), um den
Sauerstoffdurchfluss zu stoppen.
Vorsicht: Stellen Sie den
Durchflussmengenregler stets auf die AUSPosition (0), wenn das Gerät nicht verwendet
wird.

Sprint / Stroller / Hi Flow Stroller
5. Halten Sie sich beim Ermitteln der Betriebszeit desSprint/Stroller/Hi Flow Stroller an die Werte derfolgenden Tabelle:
Einstellung des Drosselventils
ModellSprintStroller

Hi Flow
Stroller

NominalNominalDrosselventil Pos.

SprintStroller

Hi Flow
Stroller

Aus

30:00

60:00

71:00

0,25

18:00

30:00

entfällt

0,5

12:12

23:00

32:00

0,75

09:18

17:24

entfällt

1

07:30

14:06

16:00

1,5

05:30

10:12

entfällt

2

04:18

08:00

08:00

2,5

03:36

06:36

06:20

3

03:06

05:36

05:20

4

02:24

04:24

04:00

5

02:00

03:36

entfällt

6

01:42

03:00

02:40

8

N/A

N/A

02:00

10

N/A

N/A

01:40

12

N/A

N/A

1:20

15

N/A

N/A

01:00

Hinweis: Die Zeiten sind in Stunden und Minuten
angegeben (im Format 00-00).
Hinweis: Die „Nennzeiten“ beziehen sich auf ideale
Bedingungen, d. h. maximale Befüllung, präzise
Durchflussraten, günstige Verlustrate, Einheit wird
nicht bewegt usw.
Diese Zeiten sind erwartete Höchstzeiten,
Hinweis: Ihre individuellen Ergebnisse können
variieren.
6a. Führen Sie zum Überprüfen des FlüssigsauerstoffFüllstands im Gerät mit der elektronischen
Füllstandanzeige folgende Schritte aus:

• Anzeige Typ 4: Drücken Sie die grüne Betriebstaste
an der Vorderseite des Messgeräts. Lesen Sie den
LED-Bogen ab, der den Füllstand anzeigt.

Vorsicht: Der Sprint/Stroller/Hi Flow Stroller
ist leer, wenn nur das letzte Segment desLichtbalkens beleuchtet ist.
Wenn die Anzeige für niedrigen Akkustand beim
Drücken des Knopfes aufleuchtet, benachrichtigen
Sie Ihren medizinischen Leistungserbringer.
Hinweis: Die Sauerstoffversorgung durch den Sprint/
Stroller/Hi Flow Stroller wird bei aufleuchtender
Akkustandanzeige fortgesetzt, solange sich im Gerät noch
Flüssigsauerstoff befindet.
6b. Zum Überprüfen des Flüssigsauerstoff-Füllstands
im Gerät mit der mechanischen Füllstandanzeige
folgende Schritte ausführen:• Setzen Sie das Gerät am Gewebeband direkt auf derschwarzen Messkammer mit der Federanzeige ab.
• Sorgen Sie dafür, dass die Einheit stabil bleibt (d. h.
nicht auf und ab schwingt), und lesen Sie dann dieangezeigte Farbskala ab, um den ungefähren Inhalt
des Flüssigsauerstoffs zu ermitteln.

• Anzeige Typ 3: Drücken Sie mindestens zwei
Sekunden den Druckknopf (Füllstandschalter) an der
Geräteoberseite. Lesen Sie oben am Lichtbalken den
Füllstand ab.

PN MN236-C4 E | Bedienungsanleitung DE - 37

Sprint / Stroller / Hi Flow Stroller
Hinweis: Wenn die Einheit leer ist, wird nur der rote
Streifen der Skala angezeigt. In diesem Fall müssen Sie
die Einheit vor Gebrauch auffüllen.
7. Unter bestimmten Umgebungsbedingungen, bei
fortlaufender Verwendung und besonders wenn
die tragbare Einheit nicht bewegt wird, kann es zuübermäßigem Frost um die Spirale des Zerstäubers
und am Gehäuse kommen. Sie können diese
Frostbildung durch Beklopfen der Einheit und/oderdurch Abwischen des angesammelten Frosts vom
Gehäuse verringern.

Wartung desKondenswasserschwamms
Wenn die Einheit leer ist und sich auf
Zimmertemperatur erwärmt hat, beseitigen Sie jegliche
Feuchtigkeit aus der Kondenswasserschale:
Für harte Taschen• Kippen Sie die Einheit auf eine Seite.
Halten Sie den Knopf in der Mitte derKondenswasserschale fest (Unterseite derEinheit) und ziehen Sie diesen gerade heraus.
Dadurch wird die Schale freigegeben.
Hinweis: Wenn der Schwamm stark verschmutzt ist, kanner in der Waschmaschine gewaschen werden (mit entsprechendem Desinfektionsmittel).
• Um die Schale zu ersetzen, zentrieren Sie sieüber der runden Öffnung unten am Gerät unddrücken Sie den Knopf mit einem Klick, sodass
die Schale einrastet. Arretieren Sie die Schale
mit einem zweiten Klick in ihrer Position.

Für weiche Taschen• Öffnen Sie den unteren Reißverschluss undziehen Sie die untere Abdeckung zurück.
• Entnehmen Sie den Schwamm aus der Schale
und wringen Sie die aufgesogene Feuchtigkeit
aus. Lassen Sie den Schwamm vollständig
trocknen, bevor Sie ihn wieder in die Schaleeinsetzen.
Hinweis: Wenn der Schwamm stark verschmutzt ist,kann er in der Waschmaschine gewaschen werden(mit entsprechendem Desinfektionsmittel).
• Zum Austauschen den Schwamm einfach nur indie untere Abdeckung einführen und den unteren
Reißverschluss wieder schließen.
Hinweis: Wenn sich der Knopf nicht eindrücken lässt und
die Schale nicht richtig einrastet, halten Sie diese fest undziehen am Knopf, bis sich die Schale mit einem Klickenöffnet. Setzen Sie die Schale jetzt gemäß vorheriger
Beschreibung ein.

Frostvermeidung
Unter bestimmten Umgebungsbedingungen, bei
fortlaufender Verwendung und besonders wenn
die tragbare Einheit nicht bewegt wird, kann es zuübermäßigem Frost um die Spirale des Zerstäubers undam Gehäuse kommen. Sie können diese Frostbildung
durch Beklopfen der Einheit und/oder durch
Abwischen des angesammelten Frosts vom Gehäuse
verringern.
WARNUNG: WEGEN DER MÖGLICHKEITEINER ÜBERMÄSSIGEN FEUCHTIGKEITS-/EISBILDUNG WIRD EMPFOHLEN, DEN
HI FLOW STROLLER FÜR EINE STUNDE
NACH DER VERWENDUNG NICHT MEHR ZU
BENUTZEN, UM FEUCHTIGKEITSBILDUNGZU VERMEIDEN.

Verriegelt
Eingerastet
Offen

Knopf Schale

Hinweis: Wenn sich der Knopf nicht eindrücken lässt und
die Schale nicht richtig einrastet, halten Sie diese fest undziehen am Knopf, bis sich die Schale mit einem Klickenöffnet. Setzen Sie die Schale jetzt gemäß vorheriger
Beschreibung ein.

38 - DE PN MN236-C4 E | Bedienungsanleitung

Wegen der höheren Durchflussraten beim Hi Flow
Stroller können am Gerät Kondensation und Eisbildung
auftreten, insbesondere bei Durchflusseinstellungenvon 10 LPM und mehr. Es wird empfohlen, das Gerätnach seiner Verwendung mindestens eine Stunde nicht
mehr zu benutzen, um die Möglichkeit übermäßige
Feuchtigkeits- oder Eisbildung zu verringern.

Sprint / Stroller / Hi Flow Stroller
Fehlerbehebung
Die folgenden Informationen sollen Ihnen bei der Behebung von Fehlern und der Lösung von einfachen
Problemen beim Betrieb helfen, die bei der Verwendung Ihres Sprint/Stroller/Hi Flow Stroller auftreten können.
Problem
LösungDas tragbare Gerät erzeugt • Das Zischen kann auftreten, um im tragbaren Gerät den richtigen Betriebsdruck
ein Zischgeräusch.
aufrechtzuerhalten. Das Zischen tritt am wahrscheinlichsten nach dem Befüllen auf,oder wenn die Position des tragbaren Geräts geändert wird. Das Zischen kann nachdem Befüllen ca. 10 Minuten lang andauern. Außerdem kann es auftreten, wenn das
Durchflussregelventil niedrig eingestellt ist.

Der Durchfluss der
tragbaren Einheit stoppt
während der Verwendung.

• Wenn die tragbare Einheit nicht in die richtige Position gelegt wurde, stellen Sie die
Einheit wieder aufrecht hin und warten Sie einige Minuten, bis sie sich stabilisiert hat.
• Stellen Sie sicher, dass die Kanüle sicher am Anschluss des Sauerstoffaustritts befestigt
ist.
• Achten Sie darauf, dass die Kanüle nicht geknickt ist.
• Inhaltsanzeige/Füllstand prüfen und ggf. nachfüllen.

Das tragbare Gerät lässt
sich nicht befüllen.

Das Entlüftungsventil
des tragbaren Geräts
schließt am Ende desBefüllungsvorgangs nicht
richtig.

Das tragbare Gerät lässt
sich nach dem Befüllen
nicht problemlos vom
Vorratsbehälter trennen.

• Sicherstellen, dass sich der Durchflussmengenregler nicht in der Stellung AUS(0) befindet.
• Überprüfen Sie, ob sich Sauerstoff im Vorratsbehälter befindet.
• Stellen Sie sicher, dass die Füllanschlüsse des tragbaren Geräts und desVorratsbehälters während des gesamten Befüllungsvorgangs vollständig miteinander
eingerastet sind.
• Wenn das Entlüftungsventil nicht schließt und weiterhin ein Zischen auftritt sowie eine
Sauerstoffdampfwolke austritt, entfernen Sie das tragbare Gerät vorsichtig, indem Sie
den Entriegelungsknopf am Vorratsbehälter herunterdrücken. Die Entlüftung von derUnterseite des tragbaren Geräts stoppt nach einigen Minuten. Das Gerät muss sichaufwärmen, bevor Sie das Entlüftungsventil schließen können. Es kann bis zu 60 Minuten
dauern, bis der korrekte Druck im tragbaren Gerät für einen präzisen Sauerstofffluss
wiederhergestellt ist. Verwenden Sie, falls notwendig, eine alternative Sauerstoffquelle,
wie z. B. ein am Vorratsbehälter angebrachtes Durchflussregelventil.
• Die Füllanschlüsse des tragbaren Geräts und des Vorratsbehälters sind möglicherweise
eingefroren.
• WENDEN SIE KEINE GEWALT AN. Lassen Sie die eingefrorenen Teile einige Minuten
lang aufwärmen, und trennen Sie das tragbare Gerät, sobald das Eis geschmolzen ist.
Um ein Zusammenfrieren der Einheiten zu vermeiden, wischen Sie den Füllstutzen am
Vorratsbehälter und den Füllanschluss am tragbaren Gerät vor dem Befüllen stets mit
einem sauberen, trockenen Tuch ab.

PN MN236-C4 E | Bedienungsanleitung DE - 39

Sprint / Stroller / Hi Flow Stroller
Reinigungsstandard
WARNUNG: REINIGEN SIE DAS GERÄT NUR
NACH DER ENTLEERUNG.
• Reinigung mit einer Lösung aus Waschmittel undWasser.
• Reinigungslösung direkt auf ein flusenfreies
Tuch auftragen. Zulässige Reinigungsmittel
sind u. a. HydroPure und HydroKlean. Keine
Reinigungsmittel direkt auf den Sprint/Stroller/Hi
Flow Stroller sprühen.
• Außenflächen mit dem flusenfreien Tuchabwischen, bis die Außenflächen sauber sind.
Vorsicht: Verwenden Sie
keine Hochtemperatur- und
Hochdruckreinigungsgeräte zur Reinigung
dieser Einheiten.
• Reinigungsmittel nicht mit im Inneren
befindlichen Bauteilen oder Ventilen in Berührung
bringen.
• Gerät vor der Verwendung gründlich trocknen
lassen.
Hinweis: Hinweis für Leistungserbringer – Informationen
zur Wiederaufbereitung sind dem entsprechenden
Wartungshandbuch zu entnehmen.

Wartung
Ihr Dienstanbieter ist für alle Wartungsarbeiten
verantwortlich, die gemäß dem technischen Handbuch
dieses Geräts erforderlich sind. Wenden Sie sich an Ihren
Dienstanbieter, wenn Sie Wartungsanforderungen haben.
Der Kondenswasserschwamm ist das einzige vom
Benutzer zu wartende Teil. Alle anderen Teile müssen
nicht vom Benutzer gewartet werden.
Die voraussichtliche Nutzungsdauer beträgt mindestensfünf Jahre.
Reinigen Sie die Füllanschlüsse sowohl an der stationären
als auch an der tragbaren Einheit jeweils zwischenzwei Befüllungen mit einem sauberen, trockenen undfusselfreien Tuch, um ein Einfrieren und ein mögliches
Geräteversagen zu vermeiden.
Hinweis: Sollten weitere Wartungsmaßnahmen
erforderlich werden, sind diese durch einen qualifizierten
Servicetechniker oder Dienstleister vorzunehmen.

ZubehörWEEE und RoHS

Das Symbol soll den Besitzer des Geräts darauf
hinweisen, dass das Gerät gemäß der Richtlinieüber Elektro- und Elektronik-Altgeräte am Endeseiner Lebensdauer zu einer Recyclingstelle gebracht
werden muss.
Unsere Produkte entsprechen der Richtlinie zur
Beschränkung der Verwendung bestimmter gefährlicher
Stoffe (RoHS). Sie enthalten Blei oder andere gefährliche
Materialien höchstens in Spuren.

Entsorgung
Geben Sie den Sprint/Stroller/Hi Flow Stroller,
einschließlich aller Komponenten, für eine
ordnungsgemäße Entsorgung stets an Ihren
Leistungserbringer zurück. Für Anweisungen zur
ordnungsgemäßen Entsorgung der Batterie können Sie
sich zudem an die zuständigen Stellen Ihres Wohnortes
wenden.

Transport und Aufbewahrung
Das Gerät sollte in aufrechter Position und unter guter
Belüftung aufbewahrt werden. Lassen Sie das Gerät
nicht auf der Seite liegen. Luftfeuchtigkeit bis zu 95 %nicht kondensierend. Die Temperaturen reichen von-40 °C bis 70 °C (-40 °F bis 158 °F).
Die Betriebstemperaturen reichen von 10 °C bis 40 °C(14 °F bis 104 °F). Relative Luftfeuchtigkeit von 30 %bis 75 % nicht kondensierend.
Hinweis: Der atmosphärische Druckbereich beträgt 700 hPa
bis 1060 hPa (Höhe von 10.000 Ft bis -1.000 Ft).
40 - DE PN MN236-C4 E | BedienungsanleitungTransportwagen TN 20765565

Sprint / Stroller / Hi Flow Stroller
Sicherheit
WARNUNG: TRAGBARE HF-KOMMUNIKATIONSGERÄTE (EINSCHLIESSLICH PERIPHERIEGERÄTE WIE
ANTENNENKABEL UND EXTERNE ANTENNEN) SOLLTEN NICHT IN EINEM ABSTAND VON WENIGER
ALS 30 CM ZU EINEM TEIL DES SPRINT/STROLLER/HI FLOW STROLLER VERWENDET WERDEN,
EINSCHLIESSLICH DER VOM HERSTELLER ANGEGEBENEN KABEL. ANDERNFALLS KANN DIE LEISTUNGDIESES GERÄTS BEEINTRÄCHTIGT WERDEN.
WARNUNG: DIE VERWENDUNG VON ANDEREN ZUBEHÖRTEILEN, WANDLERN UND KABELN ALS DEN VOMHERSTELLER DIESES GERÄTS ANGEGEBENEN KANN ZU ERHÖHTEN MAGNETISCHEN AUSSENDUNGEN ODEREINER VERRINGERTEN ELEKTROMAGNETISCHEN STÖRFESTIGKEIT DIESES GERÄTS UND SOMIT ZU EINEMUNSACHGEMÄSSEN BETRIEB FÜHREN.
WARNUNG: DIESES GERÄT SOLLTE NICHT NEBEN, AUF ODER UNTER ANDEREN GERÄTEN VERWENDETWERDEN. ANDERNFALLS KANN DER KORREKTE BETRIEB BEEINTRÄCHTIGT SEIN. SOLLTE DIES DENNOCH
ERFORDERLICH SEIN, SIND DIE BETREFFENDEN GERÄTE AUF KORREKTEN BETRIEB ZU PRÜFEN.
Vorsicht: Medizinische elektrische Geräte unterliegen hinsichtlich der elektromagnetischen Verträglichkeit(EMV) besonderen Vorsichtsmaßnahmen und müssen gemäß den in diesem Handbuch bereitgestellten
Informationen zur EMV installiert und in Betrieb genommen werden.
Vorsicht: Tragbare und mobile (HF) Kommunikationsausrüstung kann medizinische elektrische Geräte
beeinträchtigen.
Tabelle 1

Leitlinien und Herstellererklärung – elektromagnetische Aussendungen
Der Sprint/Stroller/Hi Flow Stroller ist für den Einsatz in der unten beschriebenen elektromagnetischen
Umgebung vorgesehen. Der Kunde oder Benutzer des Sprint/Stroller/Hi Flow Stroller muss sicherstellen, dasser in einer solchen Umgebung benutzt wird.
Störaussendungsmessungen
HF-AussendungÜbereinstimmung
Gruppe 1

CISPR 11

Elektromagnetische Umgebung – Leitlinien
Der Sprint/Stroller/Hi Flow Stroller verwendet
HF‑Energie ausschließlich für interne Funktionen.
Daher ist die HF-Aussendung sehr gering, und es ist
unwahrscheinlich, dass benachbarte elektronische
Geräte gestört werden.

HF-Emissionen CISPR 11

Klasse B

Aussendungen vonOberschwingungen nachIEC 61000-3-2

Nicht zutreffend

Aussendungen von
Spannungsschwankungen/
Flicker nach IEC 61000-3-3

Nicht zutreffend

Der Sprint/Stroller/Hi Flow Stroller ist für den
Gebrauch in allen Einrichtungen einschließlich
denen im Wohnbereich und solchen geeignet, dieunmittelbar an ein öffentliches Niederspannungsnetz
angeschlossen sind, das auch Gebäude versorgt, diezu Wohnzwecken benutzt werden.

PN MN236-C4 E | Bedienungsanleitung DE - 41

Sprint / Stroller / Hi Flow Stroller
Tabelle 2*: Empfohlene Mindestabstände zwischen tragbaren und mobilen
HF-Kommunikationsgeräten und dem Sprint/Stroller/Hi Flow Stroller
Der Sprint/Stroller/Hi Flow Stroller ist für den Betrieb in einer elektromagnetischen Umgebung vorgesehen, in der Störungendurch abgestrahlte Hochfrequenz kontrolliert sind. Der Kunde oder Benutzer des Sprint/Stroller/Hi Flow Stroller kanndazu beitragen, elektromagnetische Störungen zu vermeiden, indem er entsprechend der maximalen Ausgangsleistung
der Kommunikationsausrüstung den nachstehend empfohlenen Mindestabstand zwischen tragbaren und mobilen HFTelekommunikationsgeräten (Sendern) und dem Sprint/Stroller/Hi Flow Stroller einhält.
Nennleistung desTrennungsabstand je nach Frequenz des Senders
Sendersm
80 MHz und 800 MHz
800 MHz bis 2,5 GHz
150 kHz bis 80 MHz
d = 1,2√PW
d = 1,2√P
d = 2,3√P
0,01
0,12 m0,12 m0,23 m0,10,38 m0,38 m0,73 m
1
1,2 m
1,2 m
2,3 m
10
3,8 m
3,8 m
7,3 m
100
12 m
12 m
23 m
Für Sender, deren maximale Nennleistung in obiger Tabelle nicht angegeben ist, kann der empfohlene Schutzabstand (d) in Metern(m) unter Verwendung der Gleichung ermittelt werden, die für die jeweilige Sendefrequenz gilt. Dabei ist P die maximale Nennleistung des Senders in Watt (W) gemäß Angabe des Senderherstellers.
ANMERKUNG 1 Bei 80 MHz und 800 MHz gilt der Schutzabstand für den höheren Frequenzbereich.
ANMERKUNG 2 Diese Leitlinien sind möglicherweise nicht in allen Fällen anwendbar. Die Ausbreitung elektromagnetischer Wellen
wird durch Absorptionen und Reflexionen von Gebäuden, Gegenständen und Menschen beeinflusst.

* Diese Tabelle ist als Standardvoraussetzung für Geräte enthalten, die mit bestimmten Messpegeln und überbestimmte Frequenzbereiche getestet und als vorschriftskonform befunden wurden.

Tabelle 3: Leitlinien und Herstellererklärung – elektromagnetische Störfestigkeit
Der Sprint/Stroller/Hi Flow Stroller ist für den Einsatz in der unten beschriebenen elektromagnetischen Umgebung vorgesehen. Der
Kunde oder Benutzer des Sprint/Stroller/Hi Flow Stroller muss sicherstellen, dass er in einer solchen Umgebung benutzt wird.
StörfestigkeitsprüIEC 60601 PrüfpegelÜbereinstimmungElektromagnetische Umgebung – Leitlinien
fung
spegelFußböden sollten aus Holz oder Beton
Elektrostatische± 8 kV Kontakt± 8 kV Kontakt
bestehen oder mit Keramikfliesen versehen
Entladung (ESD)± 2 kV, ± 4 kV, ± 8 kV, ±± 2 kV, ± 4 kV, ± 8 kV, ± sein. Wenn der Fußboden mit synthetischem
IEC 61000-4-2
15 kV Luft
15 kV LuftMaterial versehen ist, muss die relative
Luftfeuchte mindestens 30 % betragen.**
Nicht zutreffend
Schnelle transiente±2 kV für Netzleitungenelektrische Störgrößen

Gerät mit Gleichstromversorgung

Transienten/Burst
IEC 610004-4±1 kV für Eingangs /AusgangsleitungenNicht zutreffend

Stoßspannungen
IEC 61000-4-5±1 kV Gegentaktspannung±2 kV Gleichtaktspannung

Spannungseinbrüche,Kurzzeitunterbrechungen und Schwankungen der Versorgungsspannung gemäß
IEC 61000-4-11
Magnetfeld der Netzfrequenz (50/60 Hz)
gemäß
EC 61000-4-8

Nicht zutreffend

Keine Dateneingabe/ausgabeleitungenNicht zutreffend
Gerät mit Gleichstromver- Nicht zutreffend.
sorgung<5% UT (>95% Einbruch der
UT ) für 0,5 Perioden
40% UT (60% Einbruch
Nicht zutreffend
der UT ) für 5 Perioden
Gerät mit Gleichstromver- Nicht zutreffend.
70% UT (30% Einbruchsorgung
der UT ) für 25 Perioden<5% UT (>95% Einbruch der
UT ) für 5 Sekunden
3A/m
50/60 Hz

3A/m
50/60 HzDie Magnetfelder bei Netzfrequenz sollten den
typischen Werten einer Geschäfts- oder Krankenhausumgebung entsprechen.

Hinweis: UT ist die Netzwechselspannung vor der Anwendung der Prüfpegel.
** Diese Aussage zeigt an, dass die erforderlichen Prüfungen in einer kontrollierten Umgebung durchgeführt wurden undder Sprint/Stroller/Hi Flow Stroller den Vorschriften entspricht.
42 - DE PN MN236-C4 E | BedienungsanleitungSprint / Stroller / Hi Flow Stroller
Tabelle 4
Richtlinie und Herstellererklärung – Störfestigkeit medizinischer elektrischer Geräte und Systeme
Richtlinie und Herstellererklärung – Störfestigkeit
Der Sprint/Stroller/Hi Flow Stroller ist für den Einsatz in der unten beschriebenen elektromagnetischen
Umgebung vorgesehen. Der Kunde oder Benutzer des Sprint/Stroller/Hi Flow Stroller muss sicherstellen, dasser in einer solchen Umgebung benutzt wird.
Störfestigkeitsprüfung IEC 60601 Prüfpegel Übereinstimmungspegel
Elektromagnetische Umgebung – Leitlinien
Leitungsgebundene RF IEC 3 Veff
61000-4-6
6 Veff (In ISM Bands)
150 kHz bis 80 MHz

Tragbare und mobile Funkgeräte sollten nicht ineinem geringeren Abstand zum Sprint/Stroller/Hi
Batteriebetriebenes Gerät, kein Flow Stroller verwendet werden, als im empfohlenen
SIP/SOP
Schutzabstand, der nach der für die Sendefrequenz
zutreffenden Gleichung berechnet wird. Dieser
Schutzabstand gilt auch für die Geräteleitungen.
Nicht zutreffend

Empfohlener Schutzabstandd = 1,2 √P

Abstrahlung RF IEC
61000-4-3

80 MHz bis 2,7 GHz

10 V/m80 MHz – 2,7 GHz
80 % AM bei 1 kHz

d = 1,2 √P
d = 2,3 √P
Mit P als der Nennleistung des Senders in Watt (W)gemäß Angaben des Senderherstellers und d als
empfohlenem Schutzabstand in Metern (m).
Die Feldstärken stationärer HF-Sendersollten gemäß einer Untersuchung vor Ortain jedem Frequenzbereichb innerhalb desÜbereinstimmungspegels liegen.
In der Umgebung von Geräten, die das folgende
Bildzeichen tragen, sind Störungen möglich:Prüffrequenz(MHz)

Banda)(MHz)

Diensta)Modulationb)

Höchstleistung (W)Abstand(m)385

380–390TETRA 400

1,8

0,3

27

450

430-470

GMRS 460, FRS 460

Pulsmodulationb)
18 Hz
FMC)
±5 kHz Abweichung1 kHz Sinus

Störfestigkeitsprüfpegel
(V/m)

2

0,3

28

704-787

LTE-Band 13, 17

Pulsmodulationb)
217 Hz

0,2

0,3

9

800-960

GSM 800/900, TETRA 800,
IDEN 820, CDMA 850,
LTE-Band 5

Pulsmodulationb)
18 Hz

2

0,3

28

GSM 1800; CDMA 1900;
GSM 1900; DECT; LTEBand 1, 3, 4, 25; UMTS
Bluetooth, WLAN,
802.11 b/g/n, RFID 2450,
LTE-Band 7

Pulsmodulationb)
217 Hz2

0,3

28

Pulsmodulationb)
217 Hz2

0,3

28

Pulsmodulationb)
217 Hz

0,2

0,3

9

710
745
780
810
870
930
1720
1845
1970

1700-1900

2450

2400-2570

5240
5500
5785

5100-5800

WLAN 802.11 a/n

Hinweis: Wenn dies zum Erreichen des STÖRFESTIGKEITSPRÜFPEGELS erforderlich ist, kann der Abstand zwischen der
Sendeantenne und dem Sprint/Stroller/Hi Flow Stroller auf 1 m verringert werden. Der Prüfabstand von 1 m ist gemäß IEC
61000-4-3 zulässig.
Bei einigen Diensten sind nur die Uplink-Frequenzen enthalten.
Der Träger muss anhand des Rechteckwellensignals eines halben Betriebszyklus moduliert werden.
Als Alternative zur FM-Modulation kann eine 50-prozentige Pulsmodulation bei 18 Hz verwendet werden, auch wenn es sich nicht um einetatsächliche Modulation handelt, wäre dies der ungünstigste Fall.
abc

CAIRE und CAIRE Inc. sind eingetragene Marken von CAIRE Inc. Eine vollständige Liste der Marken finden Sie auf unserer Website. Warenzeichen: www.careinc.com/corporate/
trademarks.
Copyright © 2023 CAIRE Inc. CAIRE Inc. behält sich das Recht vor, die Vermarktung seiner Produkte einzustellen bzw. Preise, Werkstoffe, Ausrüstungsteile, Qualität, Beschreibungen,Spezifikationen und/oder Prozesse ohne Vorankündigung zu einem beliebigen Zeitpunkt zu ändern, ohne dass hieraus irgendwelche Verpflichtungen oder Rechtsfolgen entstehen. Alle hiernicht ausdrücklich genannten Rechte bleiben im gesetzlich zulässigen Rahmen CAIRE Inc. vorbehalten.

PN MN236-C4 E | Bedienungsanleitung DE - 43

Sprint / Stroller / Hi Flow Stroller
Glosario de símbolos
Mantener alejado de las llamas, delfuego y de las chispas. No acercar afuentes de ignición abiertas. Se prohíbe
fumar. Reg. n.º P003

Norma ISO 7000
Mantenerlo alejado de la lluvia.
Mantenerlo seco. Reg. n.º 0626

No fumar cerca de la unidad ni mientras
la esté utilizando. Reg. n.º P002

Límite de apilamiento por número. Reg.
# 2403

Parte aplicada tipo BF (grado deprotección frente a descargas eléctricas).
Reg. n.º 5333

Nombre y domicilio del fabricante.
Reg. n.º 3082
El país y la fecha de fabricación. El"CC" identifica el código de país de dos
letras del país de fabricación. La fecha
de fabricación está en formato AAAAMM-DD. registro # 6049
Precaución: consulte los documentos
que acompañan el producto.
Reg. n.º 0434A

Advertencia. Reg. n.º W001
Directiva del Consejo 93/42/CEE
Representante autorizado en la
Comunidad Europea
Si la etiqueta del identificador único dedispositivo (UDI) del producto tiene elsímbolo CE ####, el dispositivo cumple
con los requisitos de la Directiva 93/42/ CEE sobre dispositivos médicos. El
símbolo CE #### indica el número delcuerpo notificado.

Número de catálogo. Reg. n.º 2493
Número de serie. Reg. n.º 2498

Este dispositivo cumple con losrequerimientos de la Directiva
2010/35/UE para dispositivos médicos.
Lleva la marca "pi", tal como se muestra.

Rango de temperatura dealmacenamiento u operación. Reg. n.º
0632
Rango de humedad de almacenamiento.
Reg. n.º 2620
Limitación de presión atmosférica. Reg.
n.º 2621
Este lado hacia arriba. Reg. n.º 0623
Frágil. Manipular con cuidado.
Reg. n.º 0621
Contiene sustancias peligrosas. registro#3723

ISO 7010

ADR
Gas no tóxico.
Peligro Sustancias comburentes:
riesgo de intensificación de incendios.
Los agentes oxidantes hacen que losincendios ardan con más fuerza.
Líquido refrigerado, USP; producido
mediante licuefacción del aire.
Símbolos adicionales

Importador. registro #3725

Mantener la unidad bien ventilada en
todo momento.

Puede ocurrir congelamiento ante
el contacto con oxígeno frío enestado líquido o gaseoso o con laspartes congeladas. Advertencia: baja
temperatura. Para advertir sobre
baja temperatura o condiciones decongelamiento. Reg. n.º W010

Mantener la unidad lejos de materiales
inflamables, aceite y grasa.

Debe leer el manual de instrucciones.
Reg. n.º M002

44 - ES PN MN236-C4 E | Manual del usuario

Representante autorizado en Suiza.
Si el dispositivo lleva la marca UKCA
como se muestra con UKCA#### que
indica el número del organismo notificado, este dispositivo cumple con lasregulaciones de UKCA.

Sprint / Stroller / Hi Flow Stroller
IEC 60417
No cubra la unidad ni lleve la unidad
portátil debajo de la ropa. Estas unidades
normalmente ventilan oxígeno. Nº 5641
21 CFR 801.15
La legislación federal limita la venta deeste dispositivo a médicos o por orden
médica.
IEC 60601-1
A prueba de goteos

ISO 15223-1
Dispositivo médico. registro # 5.7.7
Identificador único de dispositivo #
5.7.10Este producto puede estar cubierto por una o varias
patentes, de EE. UU. o internacionales. Visite nuestro
sitio web para obtener el listado de patentes aplicables.
Pat.: www.caireinc.com/corporate/patents/.

Directiva del Consejo 2012/19/UE
RAEE
Este símbolo es para recordar a los propietarios del equipo que lo devuelvan a
una instalación de reciclaje al final de su
vida útil, según la Directiva sobre residuos de equipos eléctricos y electrónicos(WEEE).
Nuestros productos cumplirán con la
directiva de restricción de sustancias
peligrosas (RoHS). No contendrán más
que trazas de contenido de plomo u otro
material peligroso.

Especificaciones• Modo de funcionamiento: flujo continuo• Tipo de protección frente a choques eléctricos: equipo con alimentación interna.
• Grado de protección frente a choques eléctricos: partes aplicadas de tipo BF.
• Clasificación en función del grado de protección frente a la entrada de agua: IP22 – equipo ordinario.
• Equipo no apto para uso en presencia de mezclas inflamables.
Especificaciones del producto
SprintStroller

Hi Flow Stroller

Capacidad de oxígeno
líquidoCapacidad equivalente en
estado gaseoso
Peso (vacío)

0,68 kg (1,5 libras)

1,36 kg (3,0 libras)

1,36 kg (3,0 libras)

513 L

1,026 L

1,025 L2,04 kg (4,5 libras)

2,27 kg (5,0 libras)

2,49 kg (5,5 libras)

Peso (lleno)

2,72 kg (6,0 libras)

3,63 kg (8,0 libras)

3,86 kg (8,5 libras)

Altura11,75 in (298 mm)

13,5 in (343 mm)

13,5 in (343 mm)

Dimensiones

123.83 mm (4.875 pulgadas) 149.23 mm (5.875 pulgadas) 149 mm (5,875 pulgadas)de profundidad x 191 mm (7,5 de profundidad x 191 mm (7,5 de profundidad x 191 mm (7,5pulgadas) de ancho
pulgadas) de ancho
pulgadas) de ancho
4,3 horas
8,5 horas
8,5 horas

Tiempo de uso típico a 2
LPM
Presión operativa
Rango de evaporación
normalRango de control del flujo
estándarExactitud de la velocidad
de flujo*

137 kPa (20 psi)

137 kPa (20 psi)

137 kPa (20 psi)

0,57 kg/día (1,3 libras/día)

0,57 kg/día (1,3 libras/día)

0,57 kg/día (1,3 libras/día)

Apagado, 0,25; 0,5; 0,75; 1;
1,5; 2; 2,5; 3; 4; 5; 6 LPM+/- 10%

Apagado, 0,25; 0,5; 0,75; 1;
1,5; 2; 2,5; 3; 4; 5; 6 LPM+/- 10%

Apagado, 0,5; 1; 2; 2,5; 3; 4;
6; 8; 10; 12; 15 LPM+/- 10%* Esta precisión se produce únicamente a 70 F y 14,7 psig y con un medidor de flujo másico calibrado.

PN MN236-C4 E | Manual del usuario ES - 45

Sprint / Stroller / Hi Flow Stroller
Advertencias
Importante: lea detenidamente este manual antes
de utilizar el Sprint/Stroller/Hi Flow Stroller.
Venta exclusiva bajo receta.
ADVERTENCIA: ESTE DISPOSITIVO NO ESTÁDESTINADO A DESEMPEÑARSE COMOSOPORTE VITAL.
ADVERTENCIA: SI CONSIDERA QUE EL EQUIPO NOFUNCIONA CORRECTAMENTE, CONSULTE A SUMÉDICO. NO INTENTE REPARAR NI AJUSTAR LA
UNIDAD USTED SOLO.
ADVERTENCIA: NO MODIFIQUE ESTE EQUIPO SINAUTORIZACIÓN DEL FABRICANTE.
ADVERTENCIA: SI ES NECESARIO UN CONTINUOSUMINISTRO DEL OXÍGENO, ASEGÚRESE DEDISPONER DE UN CORRECTO SUMINISTRO DEOXÍGENO Y/O UN SUMINISTRO DE OXÍGENOSECUNDARIO EN TODO MOMENTO DURANTE LATERAPIA.
ADVERTENCIA: MANTÉNGALO ALEJADO DEL HUMO,DE LAS VELAS O DE LAS CHISPAS UNOS 3 METROS(10 PIES) O A MÁS DE 20 CM (8 PULGADAS) DE
CUALQUIER FUENTE DE IGNICIÓN.
ADVERTENCIA: MANTENGA LA UNIDAD EN UN ÁREABIEN VENTILADA.
ADVERTENCIA: NO GUARDE EQUIPOSDE OXÍGENO LÍQUIDO EN ARMARIOS,
MALETEROS NI OTROS LUGARESCERRADOS. NO COLOQUE MANTAS,
CORTINAS NI OTRO TIPO DE TELASOBRE EL EQUIPO.
ADVERTENCIA: ESTE PRODUCTO PUEDE EXPONERLOA SUSTANCIAS QUÍMICAS COMO EL NÍQUEL,
QUE, SEGÚN EL ESTADO DE CALIFORNIA, CAUSACÁNCER. PARA OBTENER MÁS INFORMACIÓN, VISITE
WWW.P65WARNINGS.CA.GOV.
ADVERTENCIA: EN EL CASO DE QUE OCURRA UN
INCIDENTE GRAVE CON ESTE DISPOSITIVO, EL
USUARIO DEBE INFORMAR INMEDIATAMENTE EL
INCIDENTE AL PROVEEDOR Y / O AL FABRICANTE. UN
INCIDENTE GRAVE SE DEFINE COMO UNA LESIÓN,MUERTE O EL POTENCIAL DE CAUSAR LESIONES /MUERTE EN CASO DE QUE VUELVA A OCURRIR EL
INCIDENTE. EL USUARIO TAMBIÉN PUEDE INFORMAR
EL INCIDENTE A LA AUTORIDAD COMPETENTE EN EL
PAÍS DONDE OCURRIÓ EL INCIDENTE.
Precaución: utilice el Sprint/Stroller/Hi Flow
Stroller siguiendo las instrucciones de su
médico.
Precaución: la legislación federal limita la venta de este
dispositivo a médicos o por orden médica.
Precaución: La unidad contiene oxígeno líquido, queestá extremadamente frío, a casi -184 ºC (-300 ºF). La
exposición a una temperatura tan baja puede causar
congelamiento grave.
46 - ES PN MN236-C4 E | Manual del usuario

Precaución: El oxígeno líquido y gaseoso, si bien no es
inflamable, produce que otros materiales se incineren auna velocidad mayor que la normal. Este riesgo, junto
con la baja temperatura del oxígeno líquido, requiere
tomar determinadas precauciones de seguridad.
Precaución: Mantenga los materiales inflamables
alejados del equipo. Los espráis en aerosol, aceites ylubricantes, como cremas faciales y vaselina, prenden
fácilmente y pueden arder rápidamente en presencia deoxígeno.
Precaución: Fumar mientras se usa una cánula deoxígeno puede causar quemaduras faciales y posiblemente la muerte.
Retirar la cánula y colocarla sobre la ropa, la cama, elsofá u otro material acolchado causará una llamaradasi se la expone a un cigarrillo, una fuente de calor o una
llama.
Si fuma: (1) apague el dispositivo portátil, (2) saque lacánula, y (3) salga de la habitación en la que se encuentre el dispositivo.
Precaución: En caso de volcado accidental, vuelva a
colocar la unidad en posición vertical, si es posible,de forma inmediata, pero con cuidado. Si se escapa
oxígeno líquido, abandone el área inmediatamente yllame a su médico. No intente mover la unidad ni evitar
que se escape oxígeno líquido.

Aclaración: No toque ninguna de las partes congeladasde la unidad.
Aclaración: No guarde ni opere el dispositivo portátil
acoplado a la unidad fija.
Aclaración: No permita que personal no capacitado
manipule ni opere este dispositivo.
Aclaración: El uso de este dispositivo está restringido
en vuelos comerciales y de carga por la Administración
Federal de Aviación.

Uso previsto

El CAIRE Sprint/Stroller/Hi Flow Stroller está diseñado para la administración de oxígeno suplementario.
El dispositivo no está destinado a soporte vital ni proporciona ninguna capacidad de monitoreo del paciente.

Sprint / Stroller / Hi Flow Stroller
Introducción

Controles

Las unidades Sprint, Stroller y Hi Flow Stroller
están destinadas a la administración de oxígeno
suplementario a pacientes en el hogar del usuario
final y también se pueden utilizar en instituciones
como residencias de ancianos o centros de cuidados
subagudos. El dispositivo Sprint/Stroller/Hi Flow
Stroller ofrece una fuente de oxígeno ambulatoria
para un periodo de tiempo prolongado.

1. Medidor electrónico del nivel de líquido
2. Interruptor electrónico del nivel de líquido
3. Medidor mecánico del nivel de líquido
4. Perilla de control de flujo
5. Conector del tubo de oxígeno
6. Válvula de ventilación
7. Conectores de llenado (consulte la página
siguiente para identificar la VDR)
8. Correa de transporte con almohadilla para
hombro9. Almohadilla de condensación
10. Recipiente de condensación

El dispositivo se emplea con pacientes con EPOC ocon aquellos que sufren dificultades respiratorias. El
dispositivo se prescribe al paciente. El dispositivo sevende a un proveedor que está capacitado para operar
y dar servicio al Sprint/Stroller/Hi Flow Stroller. El
proveedor ofrece formación al usuario.

1

El dispositivo no está destinado para su uso comosoporte vital ni proporciona capacidad de supervisióndel paciente. Se recomienda contar con una fuente
alternativa de oxígeno suplementario por si se produce
un fallo mecánico. Estos dispositivos portátiles
también tienen una válvula de desconexión rápida quepuede acoplarse a un depósito de oxígeno líquido para
trasvasar el contenido al dispositivo portátil. El médico
es el encargado del llenado del depósito.

2
4
5
8

Este manual del usuario contiene las instrucciones de
uso del Sprint/Stroller/Hi Flow Stroller. Consulte el
manual del usuario que viene con la unidad fija paraobtener más información acerca de su funcionamiento.
Aclaración: el proveedor del servicio le ayudará con laconfiguración inicial y le indicará las instrucciones enreferencia a la manipulación y al uso de la unidad.
57

6Unidades Sprint, Stroller y Hi Flow Stroller

10
9
PN MN236-C4 E | Manual del usuario ES - 47

Sprint / Stroller / Hi Flow Stroller
Identificación de la válvula dedesconexión rápida (VDR)
Conectorportátil

Conector fijo

Tipo de VDR
Llenadolateral
CAIRE

LIFE-OX

PENOX

TAEMA

PB

Cierre delllenadopor laparte
lateralCierre delllenadopor laparte
superior

Empujedelllenadopor laparte
superior

4. a. Si su conector es de empuje:
baje la unidad Sprint/Stroller/Hi Flow Stroller
hasta colocarla encima de la unidad fija y notar
que se engancha al conector.

b. Si el conector es Penox o Life-Ox:
gire la unidad Sprint/Stroller/Hi Flow Stroller
hasta notar que se engancha al conector. A
continuación, gire el Sprint/Stroller/Hi Flow
Stroller con cuidado y firmeza en el sentido) mientras presiona
de las agujas del reloj (hacia abajo.

Instrucciones de funcionamiento
Llenado del Sprint/Stroller/Hi Flow Stroller
por la parte superior
1. Limpie los conectores de llenado de la unidad fija
y del Stroller/Sprint/Hi Flow Stroller con un paño
limpio, seco y sin pelusas.
Precaución: los conectores de llenado deben
estar secos, ya que la humedad puede
provocar que el Sprint/Stroller/Hi Flow Stroller
y la unidad fija se congelen y se quedenenganchados.
2. Asegúrese de que las perillas de control de flujo delSprint/Stroller/Hi Flow Stroller y del depósito estén
en la posición de apagado (0).
3. Coloque el conector de llenado en posición verticalsobre el conector de la unidad fija.

Aclaración: continúe en el paso 6 para llenar el Sprint/
Stroller/Hi Flow Stroller por la parte lateral.

Llenado del Sprint/Stroller/Hi Flow Stroller
por la parte lateral
1. Limpie los conectores de llenado de la unidad fija
y del Sprint/Stroller/Hi Flow Stroller con un paño
limpio, seco y sin pelusas.
ADVERTENCIA: LOS CONECTORES DE
LLENADO DEBEN ESTAR SECOS, YA QUE
LA HUMEDAD PUEDE PROVOCAR QUE ELSPRINT/STROLLER/HI FLOW STROLLER Y LAUNIDAD FIJA SE CONGELEN Y SE QUEDENENGANCHADOS.
2. Asegúrese de que las perillas de control de flujo delSprint/Stroller/Hi Flow Stroller y del depósito estén
en la posición de apagado (0).

48 - ES PN MN236-C4 E | Manual del usuario

Sprint / Stroller / Hi Flow Stroller
3. Coloque el conector de llenado del Sprint/Stroller/
Hi Flow Stroller en posición vertical y conéctelo
con el conector fijo.

4. Gire la unidad Sprint/Stroller/Hi Flow Stroller en el) hastasentido contrario de las agujas del reloj (notar que se engancha al pasador y la ranura.

7. También es probable que detecte vapor alrededordel conector. Esto es normal debido a la grandiferencia de temperatura entre el oxígeno líquido
y el aire cálido que rodea las unidades.
8. Cuando la unidad esté llena, cierre la válvula deventilación. Sabrá que la unidad está llena cuandocambie el tono del sonido sibilante y salga algo delíquido del conducto de ventilación.
Aclaración: si se cierra la válvula después de una
descarga de flujo casi continuo de líquido, el Sprint/
Stroller/Hi Flow Stroller se llenará más, pero sedesperdiciará algo de oxígeno líquido.

5. Gire el Sprint/Stroller/Hi Flow Stroller con cuidado
y firmeza hasta colocarlo de nuevo en posición
vertical. Las unidades ya estarían encajadas.

Aclaración: continúa en el paso 6.
Aclaración: para los conectores de empuje, sujete la
unidad y manténgala en su sitio durante el llenado.

Aclaración: si se cierra la válvula a la primera señal dedescarga de líquido, el Sprint/Stroller/Hi Flow Stroller sellenará menos, pero se ahorrará oxígeno líquido.
9. Para los conectores de empuje, pulse el botón deextracción hasta que las unidades se separen. Paraotro tipo de conectores, gire la unidad hasta que sesepare de la unidad fija.

Botón de apertura de la unidad portátil(solo para conectores de empuje)

6. Abra la válvula de ventilación para empezar a llenar
su unidad Sprint/Stroller/Hi Flow Stroller. Debería
oír como se escapa el oxígeno. No se preocupe. Ese
sonido sibilante es totalmente normal.
Aclaración: las lecturas del medidor del nivel de líquido
son más precisas pasados 30 segundos o más desde elllenado del Sprint/Stroller/Hi Flow Stroller.
PN MN236-C4 E | Manual del usuario ES - 49

Sprint / Stroller / Hi Flow Stroller
Precaución: verifique el medidor del nivel
de líquido solo después de haber cerrado laválvula.
Precaución: si la unidad Sprint/Stroller/Hi Flow Stroller
se rellena de inmediato después de un periodo de uso,
llénela y después espere 10 minutos con la VCF en
0 para que se estabilice la presión. Luego, úsela con
normalidad.
ADVERTENCIA: NO OPERE EL SPRINT/
STROLLER/HI FLOW STROLLER MIENTRASESTÁ ACOPLADO A LA UNIDAD FIJA.
ADVERTENCIA: SI LA UNIDAD SPRINT/STROLLER/HIFLOW STROLLER NO SE DESPRENDE FÁCILMENTE,
NO LA FUERCE. LAS UNIDADES PODRÍAN ESTARCONGELADAS Y ENGANCHADAS. DEJE LASUNIDADES CONECTADAS Y ESPERE HASTA QUESE CALIENTEN. LUEGO, SE DESPRENDERÁN
FÁCILMENTE. NO TOQUE LAS PARTES CONGELADAS.
ADVERTENCIA: SI LA FILTRACIÓN ES EXCESIVA
HASTA EL PUNTO DE FORMARSE UN CHORRO
DE LÍQUIDO, ABANDONE LA ZONA Y LLAME A
SU PROVEEDOR DE ATENCIÓN SANITARIA DE
INMEDIATO.
ADVERTENCIA: EL OXÍGENO LÍQUIDO O GASEOSO Y
FRÍO PUEDE CAUSAR CONGELAMIENTO EN LA PIEL.
EL OXÍGENO AUMENTA EL RIESGO DE INCENDIO.
Precaución: en caso de fuga de líquido dela unidad fija o del Sprint/Stroller/Hi Flow
Stroller después de separar las unidades,aparte la unidad Sprint/Stroller/Hi Flow
Stroller, asegúrese de que permanece en
posición vertical, salga de la habitación yllame a su proveedor de atención sanitaria
inmediatamente.
Nota: Para el proveedor del equipo: Se recomiendan lossiguientes accesorios de administración de oxígeno para
usar con el Sprint / Stroller / Hi Flow Stroller:• Cánula nasal con 7 pies (2.1 m) de tubo (LPM máximo: 6
LPM): Número de parte CAIRE 5408-SEQ

Instrucciones de funcionamiento(La cánula no está incluida).

1. Introduzca la cánula de respiración con firmeza
dentro del conector del tubo de oxígeno.
2. Ajuste la cánula de respiración en la posición
adecuada para poder respirar el oxígeno
cómodamente.
3. Gire la perilla de control de flujo de la partesuperior del Sprint/Stroller/Hi Flow Stroller ensentido de las agujas del reloj hasta que aparezcala velocidad de flujo indicada (en cifras) en la«ventana» de la perilla y note un punto de bloqueo.
Precaución: si la perilla de control de flujo seajusta entre velocidades de flujo se obtendrá
un flujo de oxígeno diferente al indicado.
4. Ya debería estar recibiendo oxígeno. Para detener elflujo de oxígeno, gire la perilla de control de flujo
en sentido opuesto a las agujas del reloj hasta la
posición de apagado (0).
Precaución: gire siempre la perilla de control
de flujo hasta la posición de apagado (0)cuando no esté usando la unidad.

• Cortafuegos: Número de parte CAIRE 21126636
Se recomienda un cortafuegos para usar con cualquiercánula.
• CAIRE ofrece un cortafuegos destinado a ser utilizado
junto con el oxígeno portátil. El cortafuegos es un fusible
térmico para detener el flujo de gas en caso de que se
encienda la cánula o el tubo de oxígeno aguas abajo yse queme hasta el cortafuegos. Se coloca en línea conla cánula nasal o el tubo de oxígeno entre el paciente yla salida de oxígeno del Sprint / Stroller / Hi Flow Stroller.
Para un uso adecuado del cortafuegos, consulte siempre
las instrucciones del fabricante (incluidas con cada kit decortafuegos).
• Información adicional sobre accesorios recomendados
está disponible en línea en www.caireinc.com.

50 - ES PN MN236-C4 E | Manual del usuario

5. Utilice el siguiente cuadro como guía para
determinar la cantidad de tiempo que estará
operativo el Sprint/Stroller/Hi Flow Stroller.

Sprint / Stroller / Hi Flow Stroller
Ajuste de la VCF
ModeloSprintStroller

Hi Flow
Stroller

NominalNominal

Posiciónde la VCF

SprintStroller

Hi Flow
Stroller

Apagado

30:00

60:00

71:00

0,25

18:00

30:00

NA

0,5

12:12

23:00

32:00

0,75

9:18

17:24

NA

1

7:30

14:06

16:00

1,5

5:30

10:12

NA

2

4:18

8:00

8:00

2,5

3:36

6:36

6:20

3

3:06

5:36

5:20

4

2:24

4:24

4:00

5

2:00

3:36

NA

6

1:42

3:00

2:40

8

N/A

N/A

2:00

10

N/A

N/A

1:40

12

N/A

N/A

1:20

15

N/A

N/A

1:00

Aclaración: los tiempos se indican en horas y minutos(formato 00:00).
Aclaración: las horas «nominales» son paracondiciones ideales, por ejemplo, llenado máximo,
velocidad de flujo exacta, tasa de pérdida, unidad
sin moverse, etc.
Esos tiempos son los máximos posibles.
Aclaración: sus resultados individuales pueden variar.
6a. Para verificar el nivel de oxígeno líquido en la
unidad con el medidor electrónico del nivel delíquido:• Medidor Gen 3: pulse el botón (interruptor del nivel
de líquido) situado en la parte superior de la unidad
durante al menos dos segundos. Consulte la partesuperior de la barra de luz para comprobar el nivel
de contenido.

• Medidor Gen 4: pulse el botón verde situado en laparte frontal del medidor. Consulte los LED para
comprobar el nivel de contenido.

Precaución: el Sprint/Stroller/Hi Flow Stroller
está vacío únicamente si el último segmento dela barra de luz está encendido.
Si se enciende el Indicador de batería baja cuando
se pulsa el botón, informe a su proveedor deatención sanitaria.
Aclaración: el Sprint/Stroller/Hi Flow Stroller seguirásuministrando oxígeno aunque se encienda el Indicadorde batería baja, siempre y cuando haya oxígeno en la
unidad.
6b. Para verificar el nivel de oxígeno líquido en la
unidad con el medidor mecánico del nivel delíquido:• Cuelgue la unidad directamente en la balanza demuelle del indicador de contenido negro mediantela correa.
• Deje que se estabilice (por ejemplo, que pare de subir
y bajar) y después consulte la escala de color parasaber el contenido de líquido aproximado.

PN MN236-C4 E | Manual del usuario ES - 51

Sprint / Stroller / Hi Flow Stroller
Aclaración: si la unidad está vacía, solo se verá la franja
de color rojo de la balanza. En este caso, deberá llenar la
unidad antes de utilizarla.
7. En ciertas condiciones ambientales, con uso
continuado y, sobre todo, si no se mueve la unidad
portátil, se puede crear una cantidad excesiva deescarcha alrededor de la bobina del vaporizador
y sobre la cubierta. Puede reducir dicha escarcha
dando pequeños golpes sobre la unidad o limpiando
la escarcha acumulada sobre la cubierta.

Mantenimiento de la almohadilla decondensación
Una vez la unidad está vacía y se haya calentado
a temperatura ambiente, elimine la humedad delrecipiente de condensación:
Para cubiertas duras• Incline la unidad hacia un lado. Agarre laperilla situada en el centro del recipiente decondensación (en la parte inferior de la unidad)
y tire hacia afuera. El recipiente se desprenderá.
Aclaración: si la almohadilla está muy sucia, puede lavarseen la lavadora (con lejía, para desinfectar).
• Para cambiar el recipiente, céntrelo en la abertura
circular situada en la parte inferior de la unidad
y presione la perilla. Cuando oiga el primer clic,
el recipiente estará enganchado y, cuando oiga el
segundo clic, estará ya encajado en su sitio.

• Retire la almohadilla del recipiente y escurra la
humedad que haya podido absorber. Deje quela almohadilla se seque por completo antes devolver a colocarla en la unidad.
Aclaración: si la almohadilla está muy sucia, puede lavarseen la lavadora (con lejía, para desinfectar).
• Para cambiarla, solo tiene que insertar laalmohadilla en la cubierta inferior y volver acerrar la cremallera inferior.
Aclaración: si no logra presionar la perilla y enganchar elrecipiente, sostenga el recipiente y tire de la perilla hasta
que haga clic y se quede abierta. A continuación, cambie
el recipiente siguiendo los pasos anteriores.

Reducción de la escarcha
En ciertas condiciones ambientales, con uso
continuado y, sobre todo, si no se mueve la unidad
portátil, se puede crear una cantidad excesiva deescarcha alrededor de la bobina del vaporizador ysobre la cubierta. Puede reducir dicha escarcha dando
pequeños golpes sobre la unidad o limpiando laescarcha acumulada sobre la cubierta.
ADVERTENCIA: DEBIDO A LA POSIBILIDAD
DE HUMEDAD EXCESIVA/FORMACIÓN DE
HIELO, SE RECOMIENDA DEJAR REPOSAR
EL HI FLOW STROLLER DURANTE 1 HORA
DESPUÉS DE SU USO PARA REDUCIR LA
FORMACIÓN DE HUMEDAD.

Encajado
Enganchado
AbiertoRecipiente
con mecanismo
para encajarlo

Aclaración: si no logra presionar la perilla y enganchar elrecipiente, sostenga el recipiente y tire de la perilla hasta
que haga clic y se quede abierta. A continuación, cambie elrecipiente siguiendo los pasos anteriores.
Para cubiertas blandas• Abra la cremallera de la parte inferior y retire lacubierta inferior.
52 - ES PN MN236-C4 E | Manual del usuario

Debido a las elevadas velocidades de flujo asociadas
al Hi Flow Stroller, se producirá condensación y algode escarcha en la unidad, sobre todo con ajustes deflujo de 10 LPM y superiores. Se recomienda dejar la
unidad inactiva durante al menos 1 hora después de su
uso para reducir la posibilidad de formación excesiva
de humedad o escarcha debido al uso continuado.

Sprint / Stroller / Hi Flow Stroller
Resolución de problemas
La siguiente información pretende ayudarlo a resolver problemas sencillos de funcionamiento a los que podría
enfrentarse al utilizar la unidad Sprint/Stroller/Hi Flow Stroller.
Problema
Solución
La unidad portátil emite un• El silbido puede producirse para mantener la presión operativa correcta dentro de la unidad
silbido.
portátil. Es más probable escuchar el silbido después de llenarla o al cambiar la posición
de la unidad portátil. El silbido puede durar aproximadamente 10 minutos después dellenar la unidad. Asimismo, puede producirse cuando la válvula de control de flujo estáen un valor bajo.

El flujo de la unidad portátil
se detiene durante el uso.

• Si la unidad portátil se ha dejado mal colocada, vuelva a ponerla en posición vertical y dejeque se estabilice durante unos minutos.
• Asegúrese de que la cánula esté bien conectada a la salida de oxígeno.
• Asegúrese de que la cánula no esté retorcida.
• Compruebe el indicador/nivel del contenido y llene la unidad portátil si fuera necesario.

La unidad portátil no sellena.
La válvula de ventilación dela unidad portátil no cierra
correctamente al final delproceso de llenado.

La unidad portátil no se
desprende fácilmentedel depósito después dellenarla.

• Asegúrese de que la perilla de control del flujo no esté en la posición de apagado («0»).
• Verifique que haya oxígeno en el depósito.
• Asegúrese de que los conectores de llenado de la unidad portátil y del depósito estén bien
enganchados durante todo el proceso de llenado.
• Si la válvula de ventilación no cierra y continúa el silbido y la nube de vapor de oxígeno,
retire con cuidado la unidad portátil presionando el botón de extracción del depósito. La
ventilación de la parte inferior de la unidad portátil se detendrá en pocos minutos. Dejeque la unidad se entibie hasta que pueda cerrar la válvula de ventilación. La unidad
portátil puede tardar hasta 60 segundos en restituir la presión adecuada para un flujo
preciso de oxígeno. Si es necesario, utilice una fuente alternativa de oxígeno, como una
válvula de control de flujo, conectada al depósito.
• Los conectores de llenado de la unidad portátil y del depósito podrían haberse congelado.
• NO LA FUERCE. Espere unos minutos hasta que las partes congeladas se entibien y,
luego, desenganche la unidad portátil cuando el hielo se haya derretido. Para evitar
que las unidades se congelen y se queden enganchadas, limpie siempre el conector dellenado macho del depósito y el conector de llenado hembra de la unidad portátil con un
paño limpio y seco antes del proceso de llenado.

PN MN236-C4 E | Manual del usuario ES - 53

Sprint / Stroller / Hi Flow Stroller
Norma de limpieza
ADVERTENCIA: LÍMPIELA SOLO SI LAUNIDAD ESTÁ VACÍA.
• Límpiela con una solución de agua tibia y jabón
suave.
• Aplique la solución limpiadora directamente
en un paño sin pelusas. Entre los limpiadores
aprobados figuran HydroPure y HydroKlean.
No rocíe limpiadores directamente sobre el
Sprint/Stroller/Hi Flow Stroller.
• Limpie la superficie externa con un paño sin
pelusas hasta que quede limpia.
Precaución: no utilice equipos de lavado aalta temperatura y con una presión elevada
para limpiar estas unidades.
• No permita el ingreso de limpiador en loscomponentes internos o en las válvulas.
• Deje que la unidad se seque bien antes de usarla.
Aclaración: nota para el médico. Consulte el manualde mantenimiento correspondiente para obtener más
información sobre los procedimientos a seguir para unnuevo uso.

Mantenimiento
Su proveedor del servicio es responsable de cualquiertipo de mantenimiento que la guía de mantenimiento
considere necesario. Llame a su proveedor del servicio
si necesita algún tipo de mantenimiento.
La almohadilla de condensación es la única pieza que elusuario puede reparar. El resto de las piezas no pueden
ser reparadas por el usuario.
La vida útil esperada es de un mínimo de cinco años.
Limpie los conectores de llenado del depósito y de launidad portátil con un paño limpio, seco y sin pelusas
entre cada llenado para evitar el congelamiento yposibles fallas del equipo.
Aclaración: las tareas de mantenimiento adicionales debe
realizarlas un técnico de mantenimiento o proveedor deservicio técnico habilitado.

Accesorios

RAEE y RoHS

Este símbolo pretende recordar a lospropietarios de los equipos que deben llevarlos
a un centro de reciclado al final de su vida útil,de conformidad con la Directiva sobre residuos deaparatos eléctricos y electrónicos (RAEE).
Nuestros productos cumplirán con la Restricción sobre
Sustancias Peligrosas (RoHS). No contendrán más que
trazas de plomo o de otros materiales peligrosos.

Eliminación
Devuelva siempre a su proveedor de cuidados en elhogar el Sprint/Stroller/Hi Flow Stroller junto con
todos sus componentes para su correcta eliminación.
También puede ponerse en contacto con las oficinas
pertinentes de su ciudad para obtener instrucciones
sobre la correcta eliminación de la batería.

Transporte y almacenamiento
El dispositivo debe guardarse en posición verticalen un lugar bien ventilado. No ponga la unidad en
posición horizontal. Humedad de hasta el 95% sincondensación. Rango de temperaturas desde -40 °C
hasta 70 ºC (-104 °F hasta 158 °F).
Rango de temperaturas de funcionamiento
desde -10 °C hasta 40 °C (14 °F hasta 104 °F). Rango
de humedad relativa del 30 al 75 % sin condensación.
Aclaración: el rango de presión atmosférica es de 700 hPa
hasta 1060 hPa (elevación de 3.000 m hasta -304 m[10.000 pies hasta -1.000 pies]).
54 - ES PN MN236-C4 E | Manual del usuario

Carro de transporte PN 20765565

Sprint / Stroller / Hi Flow Stroller
Seguridad
ADVERTENCIA: LOS EQUIPOS PORTÁTILES DE COMUNICACIONES POR RADIOFRECUENCIA (ENTRELOS QUE SE INCLUYEN LOS EQUIPOS PERIFÉRICOS COMO LOS CABLES DE ANTENA Y LAS ANTENAS
EXTERNAS) DEBEN UTILIZARSE A UNA DISTANCIA DE 30 CM (12 PULGADAS) O MÁS DE CUALQUIERPARTE DEL SPRINT/STROLLER/HI FLOW STROLLER, INCLUIDOS LOS CABLES ESPECIFICADOS POR EL
FABRICANTE. DE LO CONTRARIO, PODRÍA PRODUCIRSE UNA DEGRADACIÓN DEL RENDIMIENTO DE
ESTE EQUIPO.
ADVERTENCIA: EL USO DE ACCESORIOS, TRANSDUCTORES Y CABLES QUE NO SEAN LOS ESPECIFICADOSO PROVISTOS POR EL FABRICANTE DE ESTE EQUIPO PODRÍA PROVOCAR UN AUMENTO DE LAS EMISIONESELECTROMAGNÉTICAS O UNA DISMINUCIÓN DE LA INMUNIDAD ELECTROMAGNÉTICA DE ESTE EQUIPO YDESEMBOCAR EN UN FUNCIONAMIENTO INCORRECTO.
ADVERTENCIA: SE DEBE EVITAR EL USO DE ESTE EQUIPO ADYACENTE O APILADO CON OTROS EQUIPOS,YA QUE PODRÍA DAR LUGAR A UN FUNCIONAMIENTO INCORRECTO. SI ES NECESARIO, ESTE EQUIPO Y LOSDEMÁS DEBEN VIGILARSE PARA VERIFICAR QUE FUNCIONAN CON NORMALIDAD.
Precaución: los equipos médicos eléctricos requieren precauciones especiales con respecto a lacompatibilidad electromagnética (EMC) y deben instalarse y ponerse en servicio según la información
sobre compatibilidad electromagnética (EMC) provista en este manual.
Precaución: los equipos de comunicaciones portátiles y móviles de radiofrecuencia (RF) pueden afectar alos equipos eléctricos médicos.
Tabla 1

Orientación y declaración del fabricante: emisiones electromagnéticas
El Sprint/Stroller/Hi Flow Stroller está destinado para su uso en el ambiente electromagnético que se especifica
a continuación. El cliente o el usuario del Sprint/Stroller/Hi Flow Stroller deben asegurar que se use en elambiente que se indica.

Prueba de emisiones
Emisiones deradiofrecuencia

Cumplimiento
Grupo 1CISPR 11

Emisiones de
RF CISPR 11

Ambiente electromagnético: orientación
El Sprint/Stroller/Hi Flow Stroller usa energía deradiofrecuencia solo para su función interna.
En consecuencia, sus emisiones de radiofrecuencia son
muy bajas y no es probable que causen interferencias en
equipos electrónicos cercanos.

Clase B

Emisiones armónicas
IEC 61000-3-2

No aplicable

Fluctuaciones devoltaje/emisiones deflicker IEC 61000-3-3

No aplicable

El Sprint/Stroller/Hi Flow Stroller es adecuado para
usar en todo tipo de establecimiento, incluidos losestablecimientos domésticos y aquellos conectados
directamente a la red pública de energía eléctrica debajo voltaje que alimenta edificios usados para fines
domésticos.

PN MN236-C4 E | Manual del usuario ES - 55

Sprint / Stroller / Hi Flow Stroller
Tabla 2*: Distancias recomendadas de separación entre equipos portátiles y móviles de comunicaciones deradiofrecuencia y el sistema Sprint/Stroller/Hi Flow Stroller
El Sprint/Stroller/Hi Flow Stroller está diseñado para ser utilizado en un ambiente electromagnético en el que se controlan lasinterferencias de radiofrecuencia irradiada. El cliente o el usuario del sistema Sprint/Stroller/Hi Flow Stroller pueden ayudar aprevenir la interferencia electromagnética al mantener una distancia mínima entre los equipos portátiles y móviles de comunicación
por radiofrecuencia (transmisores) y el Sprint/Stroller/Hi Flow Stroller, tal como se recomienda a continuación, de acuerdo con lapotencia máxima de salida de los equipos de comunicaciones.
Potencia de salida
Distancia de separación en función de la frecuencia del transmisor
nominal máxima delmtransmisor
80 MHz y 800 MHz
de 800 MHz a 2,5 GHz
de 150 kHz a 80 MHz

d = 1,2√P

d = 1,2√P

d = 2,3√PW
0,010,12 m0,12 m0,23 m0,10,38 m0,38 m0,73 m
1
1,2 m
1,2 m
2,3 m
10
3,8 m
3,8 m
7,3 m
100
12 m
12 m
23 m
Para transmisores con una potencia máxima de salida no mencionada anteriormente, la distancia (d) de separación recomendada
en metros (m) puede calcularse usando la ecuación aplicable a la frecuencia del transmisor, donde P es la potencia nominal
máxima de salida del transmisor en vatios (W) de acuerdo con el fabricante del transmisor.
NOTA 1: a 80 MHz y 800 MHz, se aplica la distancia de separación para el rango de mayor frecuencia.
NOTA 2: estas pautas pueden no ser de aplicación en todas las situaciones. La propagación electromagnética se ve afectada por laabsorción y la reflexión de estructuras, objetos y personas.

* Esta tabla se incluye como un requisito estándar para equipos que ya han sido sometidos a varios niveles depruebas y rangos de frecuencia específicos y han resultado conformes con la normativa.

Tabla 3: Orientación y declaración del fabricante: inmunidad electromagnéticaEl Sprint/Stroller/Hi Flow Stroller está destinado para su uso en el ambiente electromagnético que se especifica a continuación. El
cliente o el usuario del Sprint/Stroller/Hi Flow Stroller deben asegurar que se use en el ambiente que se indica.
Prueba de inmunidad Nivel de prueba IEC 60601 Nivel de cumplimiento Ambiente electromagnético: orientación
Los suelos deben ser de madera, hormigón
Contacto de ± 8 kV
Contacto de ± 8 kV
Descarga electroso cerámica. Si los suelos están cubiertos con
tática (ESD)± 2kV, ± 4 Kv
± 2kV, ± 4 Kv
material sintético, la humedad relativa debe ser
IEC 61000-4-2
Aire de ± 8 kV, ± 15 kV
Aire de ± 8 kV, ± 15 kV
de al menos un 30 %.**
No aplicable± 2 kV para las líneas deDispositivo de corriTransitorios eléctricos alimentación
ente continua
No aplicablerápidos/en ráfagas IEC
No aplicable± 1 kV para líneas de610004-4
entrada/salida
Sin líneas de datos
de entrada/salida
No aplicable
Sobrecarga± 1 kV línea(s) a línea(s)No aplicable
Dispositivo de corri± 2 kV línea(s) a tierra
IEC 61000-4-5
ente continua< 5 % UT (caída de tensión
Caídas de voltaje,
de >95 %) durante 0,5 ciclointerrupciones breves 40 % UT (caída de tensión
y variaciones dede 60 %) durante 5 ciclos
No aplicable
voltaje en las líneas de 70 % UT (caída de tensión
No aplicable
Dispositivo de corrientrada de la fuente de de 30 %) durante 25 ciclos
ente continua
alimentación< 5 % UT (caída de tensiónde >95 %) durante 5 seIEC 61000-4-11
gundosCampo magnético a
Los campos magnéticos a frecuencia industrial
frecuencia industrial
3A/m
3A/m
deben ser los característicos de un entorno(50/60 Hz)
50/60 Hz50/60 Hzcomercial u hospitalario típico.
IEC 61000-4-8
Aclaración: UT es el voltaje de CA antes de la aplicación del nivel de prueba.

** Esta afirmación indica que la prueba requerida fue realizada en un ambiente controlado y que se estimó que elLiberator cumplía con la normativa.
56 - ES PN MN236-C4 E | Manual del usuario

Sprint / Stroller / Hi Flow Stroller
Tabla 4

Orientación y declaración del fabricante: inmunidad de los equipos EM y de los sistemas EM

Orientación y declaración del fabricante: inmunidad
El Sprint/Stroller/Hi Flow Stroller está destinado para su uso en el ambiente electromagnético que se especifica
a continuación. El cliente o el usuario del Sprint/Stroller/Hi Flow Stroller deben garantizar que se use en elambiente que se indica.
Prueba de inmunidad Nivel de prueba
Nivel deAmbiente electromagnético: orientación
IEC 60601
cumplimiento
Perturbaciones
conducidas, inducidas
por radiofrecuencia
IEC 61000-4-6

3 Vrms6 Vrms (en franjas de equipos
industriales, científicos ymédicos [ISM])
de 150 kHz a 80 MHz

Los equipos portátiles y móviles de comunicaciones
por radiofrecuencia no deben usarse más cerca de laspiezas del Sprint/Stroller/Hi Flow Stroller, incluidos los
cables, que la distancia de separación recomendadacalculada a partir de la ecuación aplicable a la frecuenciadel transmisor.

No aplicable

Dispositivo con batería,
sin terminales deentrada/salida de señal

Distancia recomendada de separaciónd = 1,2 √P

Radiofrecuencia radiada
IEC 61000-4-3

de 80 MHz a 2,7 GHz

d = 1,2 √P
d = 2,3 √P
donde P es la clasificación máxima de potencia desalida del transmisor en vatios (W) de acuerdo conel fabricante del transmisor y d es la distancia deseparación recomendada en metros (m).

10 V/m80 MHz—2,7 GHz
80 % AM a 1 kHz

Las concentraciones de campo de transmisores fijosde radiofrecuencia, determinadas por un estudioelectromagnético del sitioa, deberían ser inferiores al
nivel de cumplimiento en cada rango de frecuenciab.
Puede producirse interferencia en la cercanía de losequipos marcados con el símbolo siguiente:

Frecuencia de laprueba (MHz)Franjaa)(MHz)

Servicioa)

Modulaciónb)

385

380-390TETRA 400450

430-470

GMRS 460, FRS 460

Modulación del pulso
18 Hz
FMC)
desviación de ±5 kHz
onda sinusoidal de
1 kHz

704-787

banda LTE 13, 17

710
745
780
810
870
930
1720
1845
1970

800-960
1700-1900

2450

2400-2570

5240
5500
5785

5100-5800

b)

MáximaDistancia Nivel de prueba depotencia (W)(m)
inmunidad (V/m)
1,8

0,3

27

2

0,3

28

Modulación del pulsob)
217 Hz

0,2

0,3

9

GSM 800/900, TETRA 800, Modulación del pulsob)
IDEN 820, CDMA 850,
18 Hz
banda LTE 5

2

0,3

28

GSM 1800; CDMA 1900;
GSM 1900; DECT; banda
LTE 1, 3, 4, 25; UMTS
Bluetooth, WLAN,
802.11 b/g/n, RFID 2450,
banda LTE 7

Modulación del pulsob)
217 Hz20,328Modulación del pulsob)
217 Hz20,328

WLAN 802.11 a/n

Modulación del pulsob)
217 Hz

0,2

0,3

9

Aclaración: si es necesario alcanzar el NIVEL DE PRUEBA DE INMUNIDAD, la distancia entre la antena transmisora y el
Sprint/Stroller/Hi Flow Stroller debe reducirse a 1 m. La distancia de prueba de 1 m está permitida por la IEC 61000-4-3.
Para algunos servicios, solo se incluyen las frecuencias de subida.
La portadora deberá modularse mediante una señal de onda cuadrada de ciclo de trabajo del 50 %.
Como alternativa a la modulación FM, se puede usar una modulación del pulso del 50 % a 18 Hz, ya que, aunque no representa una
modulación real, sería el peor de los casos.
abc

CAIRE y CAIRE Inc. son marcas registradas de CAIRE Inc. Visite nuestro sitio web a continuación para obtener una lista completa de marcas registradas. Marcas registradas: www.
careinc.com/corporate/trademarks.
Copyright © 2023 CAIRE Inc. CAIRE Inc. se reserva el derecho de suspender sus productos o modificar los precios, los materiales, el equipo, la calidad, las descripciones, lasespecificaciones y/o los procesos de sus productos en cualquier momento, sin previo aviso y sin ningún compromiso ni ninguna consecuencia. Queda reservado cualquier otro derecho
que no se encuentre indicado explícitamente en este documento, según corresponda.

PN MN236-C4 E | Manual del usuario ES - 57

Sprint / Stroller / Hi Flow Stroller
Glossario dei simboli
ISO 7000

Non fumare nei pressi dell'unità odurante il suo utilizzo. N. reg. P002

Tenere lontano dalla pioggia e
conservare in luogo asciutto.
N. reg. 0626

Parte applicata di tipo BF (gradodi protezione da scossa elettrica).
N. reg. 5333

Limite di impilamento per numero. Reg.
# 2403
Nome e indirizzo del produttore.
N. reg. 3082

Avvertenza. N. reg. W001
Direttiva del Consiglio 93/42/CEE
Rappresentante autorizzato per la
Comunità Europea
Se sull’etichetta dell’identificatore
univoco del prodotto (UDI) del prodottoè riportato il simbolo CE ####, ildispositivo è conforme ai requisitidella Direttiva 93/42 / CEE relativa
ai dispositivi medici. Il simbolo CE#### indica il numero dell’organismo
notificato.

Il paese e la data di produzione. Il "CC"identifica il codice paese di due letteredel paese di produzione. La data di produzione è nel formato AAAA-MM-GG.
reg. # 6049
Attenzione, consultare i documenti diaccompagnamento. N. reg. 0434A
Numero di catalogo. N. reg. 2493

Il presente dispositivo è conforme ai
requisiti della Direttiva 2010/35/UE
relativa ai dispositivi medici e reca ilmarchio pi come illustrato.

Numero di serie. N. reg. 2498
Intervallo di temperatura diconservazione o di esercizio. N. reg.
0632

ADR

Intervallo di umidità di stoccaggio. N.
reg. 2620
Limite della pressione atmosferica. N.
reg. 2621
Lato alto. N. reg. 0623
Fragile, maneggiare con cura.
N. reg. 0621

Gas non tossico.
Pericolo Sostanze comburenti: rischio
di intensificazione dell'incendio. Gli
agenti ossidanti fanno sì che gli incendi
brucino più vigorosamente.
Liquido refrigerato, USP; prodotto
tramite liquefazione dell’aria
Símbolos adicionales

Contiene sostanze pericolose. reg. #3723

Tenere sempre l’unità ben ventilata

Importatore. reg. # 3725

Tenere lontano da materiali
infiammabili, olio e grasso.

Il contatto con ossigeno freddo allo
stato liquido o gassoso oppure con particongelate può causare congelamento.
Avviso bassa temperatura. Per segnalarecondizioni di bassa temperatura ocongelamento. N. reg. W010

Rappresentante autorizzato in Svizzera.

ISO 7010

Leggere il manuale di istruzioni.
N. reg. M002
Conservare lontano da fiamme libere,
fuoco e scintille. È vietato fumare eutilizzare fonti di accensione aperte.
N. reg. P003
58 - IT PN MN236-C4 E | Manuale dell’utente

IEC 60417

Se il dispositivo reca il marchio UKCA
come indicato con UKCA#### che
indica il numero dell'organismo notificato, questo dispositivo è conforme alle
normative UKCA.
Non coprire l'unità o trasportare l'unità
portatile sotto i vestiti. Queste unità
normalmente scaricano ossigeno. N.
5641

Sprint / Stroller / Hi Flow Stroller
21 CFR 801.15
La legge federale degli Stati Uniti limita
la vendita del presente dispositivo ai soli
medici o dietro prescrizione medica.
IEC 60601-1
A prova di gocciolamento
Direttiva del Consiglio 2012/19/UE
RAEE
Questo simbolo ricorda ai proprietaridelle apparecchiature di restituirle a un
impianto di riciclaggio al termine delloro ciclo di vita, in base alla Direttiva
sui rifiuti di apparecchiature elettriche
ed elettroniche (RAEE).
I nostri prodotti saranno conformi alla
direttiva sulla restrizione delle sostanze
pericolose (RoHS). Non conterranno piùdi tracce di piombo o altro contenuto dimateriale pericoloso.

ISO 15223-1
Dispositivo medico. reg. # 5.7.7
Identificatore univoco del dispositivo
n. 5.7.10
Questo prodotto può essere coperto da uno o più
brevetti, validi negli Stati Uniti e all’estero. Visitare il
nostro sito Web, sotto indicato, per l’elenco dei brevetti
applicabili. Brev.: www.caireinc.com/corporate/
patents/.

Specifiche• Modalità di funzionamento: flusso continuo• Tipo di protezione da elettrocuzione: apparecchiatura alimentata internamente• Grado di protezione da elettrocuzione: parte applicata di tipo BF• Classificazione a seconda del grado di protezione dall’infiltrazione di acqua: IP22 – Apparecchiatura
comune• L’apparecchio non va utilizzato in presenza di miscele infiammabili
Specifiche del prodotto

SprintStroller

Hi Flow Stroller

Capacità ossigeno liquido(LOX)
Capacità gassosa
equivalentePeso, vuoto

0,68 kg (1,5 lb)

1,36 kg (3,0 lb)

1,36 kg (3,0 lb)

513 l

1026 l1025 l

2,04 kg (4,5 lb)

2,27 kg (5,0 lb)

2,49 kg (5,5 lb)

Peso, pieno

2,72 kg (6,0 lb)

3,63 kg (8,0 lb)

3,86 kg (8,5 lb)

Altezza

298 mm (11,75 poll.)

343 mm (13,5 poll.)

343 mm (13,5 poll.)

Dimensioni

P 123,83 mm (4,875 poll.) x L P 149,23 mm (5,875 poll.) x L P 149 mm (5,875 poll.) x L 191
191 mm (7,5 poll)
191 mm (7,5 poll)
mm (7,5 poll)
4,3 ore8,5 ore8,5 oreTempo di utilizzo tipico a
2 l/minPressione di esercizio
Tasso normale dievaporazione
Intervallo di regolazione delflusso standard
Precisione della portata*

137 kPa (20 psi)

137 kPa (20 psi)

137 kPa (20 psi)

0,57 kg/giorno (1,3 lb/giorno) 0,57 kg/giorno (1,3 lb/giorno) 0,57 kg/giorno (1,3 lb/giorno)
Off, 0,25; 0,5; 0,75; 1; 1,5; 2;
2,5; 3; 4; 5; 6 l/min+/- 10%Off, 0,25; 0,5; 0,75; 1; 1,5; 2;
2,5; 3; 4; 5; 6 l/min+/- 10%Off, 0,5, 1, 2, 2,5, 3, 4, 6, 8,
10, 12, 15 l/min+/- 10%* Questa precisione vale solo a 70 °F e 14,7 psig e con un flussometro di massa accurato calibrato.

PN MN236-C4 E | Manuale dell’utente IT - 59

Sprint / Stroller / Hi Flow Stroller
Informazioni di avvertenza
Importante: leggere attentamente il presente manuale
prima di utilizzare Sprint/Stroller/Hi Flow Stroller.
Solo su prescrizione medica.
AVVERTENZA: QUESTO DISPOSITIVONON È INTESO COME DISPOSITIVO DISOSTENTAMENTO VITALE.
AVVERTENZA: SE SI RITIENE CHE
L’APPARECCHIATURA NON STIA FUNZIONANDOCORRETTAMENTE, CONTATTARE L’OPERATORESANITARIO. NON TENTARE DI RIPARARE O
REGOLARE L’UNITÀ AUTONOMAMENTE.
AVVERTENZA: NON MODIFICARE QUESTAAPPARECCHIATURA SENZA L’AUTORIZZAZIONE DEL
PRODUTTORE.
AVVERTENZA: SE È RICHIESTA UN’EROGAZIONECONTINUA DELL’OSSIGENO, ASSICURARSI CHE
DURANTE LA TERAPIA SIA SEMPRE DISPONIBILEUNA FORNITURA ADEGUATA DI OSSIGENO E/O UNAFORNITURA DI OSSIGENO SECONDARIA.
AVVERTENZA: NON CONSENTIRE FUMO, CANDELEO FIAMME APERTE ENTRO 3 METRI (10 PIEDI)DAL DISPOSITIVO O ENTRO 20 CM (8 POLLICI) DAQUALSIASI FONTE DI IGNIZIONE.
AVVERTENZA: CONSERVARE L’UNITÀ IN UN’AREA
BEN VENTILATA.
AVVERTENZA: NON CONSERVARE
L’APPARECCHIATURA A OSSIGENO LIQUIDOIN UN RIPOSTIGLIO, NEL BAGAGLIAIO DI UN
AUTO O IN ALTRE AREE CONFINATE. NON
POSIZIONARE SOPRA L’APPARECCHIATURACOPERTE, TENDAGGI O ALTRI TESSUTI.
AVVERTENZA: IL PRESENTE PRODOTTO PUÒESPORRE L'UTENTE A SOSTANZE CHIMICHE,COMPRENDENTI IL NICHEL, CHE, SECONDO LO STATODELLA CALIFORNIA, PUÒ CAUSARE IL CANCRO.
PER MAGGIORI INFORMAZIONI, VISITARE WWW.
P65WARNINGS.CA.GOV.
AVVERTENZA: NEL CASO IN CUI SI VERIFICHI UN
GRAVE INCIDENTE CON QUESTO DISPOSITIVO,
L'UTENTE DEVE SEGNALARE IMMEDIATAMENTEL'INCIDENTE AL FORNITORE E / O AL PRODUTTORE.
UN INCIDENTE GRAVE È DEFINITO COME UNALESIONE, MORTE O POSSIBILITÀ DI CAUSARELESIONI / MORTE IN CASO DI RIPETIZIONE
DELL'INCIDENTE. L'UTENTE PUÒ ANCHE SEGNALAREL'INCIDENTE ALL'AUTORITÀ COMPETENTE NEL
PAESE IN CUI SI È VERIFICATO L'INCIDENTE.
Attenzione: utilizzare Sprint/Stroller/Hi Flow
Stroller esclusivamente come da indicazionedel medico.
Attenzione: la legge federale degli Stati Uniti limita lavendita del presente dispositivo ai soli medici o dietro
prescrizione medica.

60 - IT PN MN236-C4 E | Manuale dell’utente

Attenzione: L’unità contiene ossigeno liquido estremamente freddo la cui temperatura è prossima a -184 °C(-300 °F). L’esposizione a una temperatura così bassa
può provocare un grave congelamento.
Attenzione: L’ossigeno allo stato liquido e gassoso, perquanto non infiammabile, può aumentare la velocità di
combustione di altri materiali. Questo pericolo, assiemealla bassa temperatura dell’ossigeno liquido, giustifica
alcune precauzioni di sicurezza.
Attenzione: Tenere i materiali infiammabili lontano daquesta apparecchiatura. Vaporizzatori spray, oli e grassi,
incluse creme per il viso e vaselina, si incendiano facilmente e potrebbero bruciare rapidamente in presenza
di ossigeno.
Attenzione: Non fumare mentre si indossa una cannula
per l’ossigeno per evitare il rischio di ustioni al viso e
possibili conseguenze letali.
Rimuovendo la cannula e posizionandola su indumenti,
biancheria, divani o altri materiali imbottiti si possono
verificare lampi di fuoco in caso di esposizione a sigarette, fonti di calore o fiamme.
Se si fuma: (1) spegnere l’unità portatile, (2) rimuovere
la cannula e (3) lasciare la stanza in cui si trova ildispositivo.
Attenzione: Nell’eventualità di un ribaltamento accidentale, riportare l’unità in posizione verticale immediatamente ma prestando molta attenzione, se possibile. In
caso di fuoriuscita di ossigeno liquido, abbandonare
immediatamente l’area e contattare l’operatore sanitario.
Non tentare di spostare l’unità o interrompere la fuoriuscita di ossigeno liquido.
Nota: Non toccare le parti congelate di qualsiasi unità.
Nota: Non conservare né utilizzare l’elemento portatilecollegato all’unità stazionaria.
Nota: Non consentire l’accesso o l’utilizzo del presente
dispositivo da parte di personale non addestrato.
Nota: L’utilizzo del presente dispositivo è proibito sugliaerei commerciali per passeggeri e cargo dalla Federal
Aviation Administration (Amministrazione Aviazione
Federale).

Destinazione d'uso

Il Sprint/Stroller/Hi Flow Stroller CAIRE è destinato
alla somministrazione di ossigeno supplementare. Il
dispositivo non è destinato al supporto vitale né offre
funzionalità di monitoraggio del paziente.

Sprint / Stroller / Hi Flow Stroller
Introduzione

Controlli

Sprint, Stroller e Hi Flow Stroller sono progettati
per la somministrazione a domicilio di ossigeno
supplementare al paziente e possono essere utilizzati
anche in istituti di assistenza sanitaria quali case dicura o strutture per degenza subacuta. Sprint/Stroller/
Hi Flow Stroller fornisce una fonte di ossigeno
ambulatoriale per un periodo di tempo prolungato.

1. Indicatore elettronico di livello del liquido
2. Interruttore elettronico del livello del liquido
3. Indicatore meccanico di livello del liquido
4. Manopola di regolazione del flusso
5. Connettore del tubo dell’ossigeno
6. Valvola di aerazione
7. Collegamenti per riempimento (vedere nella
pagina seguente la descrizione delle QDV)
8. Cinghia per il trasporto con imbottitura incorrispondenza della spalla
9. Assorbitore di condensa
10. Recipiente per condensa

Il dispositivo viene usato da pazienti BPCO (portatoridi broncopneumopatia cronica ostruttiva) o da pazienti
con capacità respiratoria ridotta. Il dispositivo viene
prescritto al paziente. Il dispositivo Sprint/Stroller/
Hi Flow Stroller è venduto a fornitori formati per
il suo utilizzo e la sua manutenzione. Il fornitore è
responsabile della formazione dell’utente.

1

Il dispositivo non è progettato per il supporto vitale
artificiale e non offre funzionalità di monitoraggio delpaziente. Si consiglia di tenere a disposizione un’altra
sorgente di ossigeno supplementare in caso di guasti
meccanici. Questi apparecchi portatili sono dotati
anche di una valvola di scollegamento rapido, che può
essere abbinata a un dispositivo con serbatoio di LOX
che ne consente il rabbocco. L’unità serbatoio viene
riempita dall’operatore sanitario.

2
4
5
8

Il presente manuale dell’utente contiene le istruzioni
per l’utilizzo di Sprint/Stroller/Hi Flow Stroller. Fareriferimento al manuale dell’utente fornito con l’unità
stazionaria per il relativo funzionamento.
Nota: l’operatore sanitario aiuterà a eseguire l’installazione
iniziale e fornirà le istruzioni per la gestione e l’utilizzo
corretti dell’unità.
5

7

6
Unità Sprint, Stroller e Hi Flow Stroller

10
9
PN MN236-C4 E | Manuale dell’utente IT - 61

Sprint / Stroller / Hi Flow Stroller
Descrizione delle valvole discollegamento rapido (QDV)
Connettore
elemento
portatile

Connettore
unità
stazionaria

Modello di QDVCAIRE
con riempimento
laterale

ChiusuraSF

LIFE-OX

4. a. In caso di connettore a pressione:
premere l’unità Sprint/Stroller/Hi Flow Stroller
sull’unità stazionaria finché il connettore non si
incastra.

Chiusura
TF
PENOX

TAEMA
TF a
pressione
PB

b. In caso di connettore Penox o Life-Ox:
ruotare Sprint/Stroller/Hi Flow Stroller finché ilconnettore non si incastra. Successivamente, con
un movimento preciso e deciso, ruotare Sprint/)
Stroller/Hi Flow Stroller in senso orario (premendo al contempo verso il basso.

Istruzioni per l’uso
Riempimento del modello di Sprint/Stroller/
Hi Flow Stroller con riempimento dall’alto
1. Pulire i connettori di riempimento sull’unità
stazionaria e su Sprint/Stroller/Hi Flow Stroller con
un panno pulito, asciutto e privo di sfilacciature.
Attenzione: i connettori di riempimento devono essere asciutti in quanto l’umidità può far
sì che Sprint/Stroller/Hi Flow Stroller e l’unità
stazionaria congelino aderendo l’uno all’altra.
2. Verificare che le manopole di regolazione del flusso
di Sprint/Stroller/Hi Flow Stroller e del serbatoio
siano in posizione off (0).
3. Mettere il connettore di riempimento in posizione
verticale al di sopra del connettore dell’unità
stazionaria.

62 - IT PN MN236-C4 E | Manuale dell’utente

Nota: segue al passaggio 6 in Riempimento del modello di
Sprint/Stroller/Hi Flow Stroller con riempimento laterale.

Riempimento del modello di Sprint/Stroller/
Hi Flow Stroller con riempimento laterale
1. Pulire i connettori di riempimento sull’unità
stazionaria e su Sprint/Stroller/Hi Flow Stroller con
un panno pulito, asciutto e privo di sfilacciature.
AVVERTENZA: I CONNETTORI DI RIEMPIMENTO DEVONO ESSERE ASCIUTTI IN
QUANTO L’UMIDITÀ PUÒ FAR SÌ CHE SPRINT/
STROLLER/HI FLOW STROLLER E L’UNITÀSTAZIONARIA CONGELINO ADERENDOL’UNO ALL’ALTRA.
2. Verificare che le manopole di regolazione del flusso
di Sprint/Stroller/Hi Flow Stroller e del serbatoio
siano in posizione off (0)Sprint / Stroller / Hi Flow Stroller
3. Tenere il connettore di riempimento di Sprint/
Stroller/Hi Flow Stroller in posizione verticale
e unirlo al connettore dell’unità stazionaria.

Nota: i valori del livello del liquido diventano più precisi apartire da 30 secondi dopo il riempimento di Sprint/Stroller/
Hi Flow Stroller.

4. Ruotare Sprint/Stroller/Hi Flow Stroller in senso) finché il perno non si incastra
antiorario (nella fessura.

7. È possibile che venga rilevato del vapore attorno alconnettore. Questo fenomeno è normale e dipende
dalla grande differenza di temperatura tra l’ossigeno
liquido e l’aria calda circostante.

5. Con un movimento preciso e deciso, ruotare Sprint/
Stroller/Hi Flow Stroller riportandolo in posizione
verticale. A questo punto le unità sono collegate.

8. Quando l’unità è piena, chiudere la valvola diaerazione. L’unità è piena quando la tonalità delsibilo varia e dal canale di aerazione fuoriesce delliquido.
Nota: la chiusura della valvola dopo una fuoriuscita
pressoché continua di liquido consente di riempire al
massimo Sprint/Stroller/Hi Flow Stroller ma comporta lo
spreco di una certa quantità di ossigeno liquido.
Nota: la chiusura della valvola al primo segnale difuoriuscita di liquido non consente di riempire altrettanto
Sprint/Stroller/Hi Flow Stroller, ma comporta un risparmio
di ossigeno liquido.
9. In caso di connettori a pressione, premere ilpulsante di rilascio finché le unità non si separano.
Per tutti gli altri tipi di connettori, ruotare l’unità
portatile finché non si separa da quella stazionaria.

Nota: segue al passaggio 6.
Nota: in caso di connettore a pressione, mantenere l’unità
in posizione per tutta la durata del riempimento.
6. Aprire la valvola di aerazione per cominciarea riempire Sprint/Stroller/Hi Flow Stroller. Si
dovrebbe avvertire la fuoriuscita di una certa
quantità di ossigeno. Si percepirà un sibilo, che
non deve suscitare preoccupazione in quantoassolutamente normale.

Pulsante di rilascio dell’elemento portatile(solo per modelli a pressione)

PN MN236-C4 E | Manuale dell’utente IT - 63

Sprint / Stroller / Hi Flow Stroller
Attenzione: controllare l’indicatore del livellodi liquido solamente dopo aver chiuso lavalvola.
Attenzione: se Sprint/Stroller/Hi Flow Stroller viene
riempito immediatamente dopo un periodo di utilizzo,
riempirlo, quindi attendere 10 minuti dal riempimento
con l’FCV impostata su zero al fine di consentire lastabilizzazione della pressione. In seguito, procedere
con il normale utilizzo.
AVVERTENZA: NON UTILIZZARE SPRINT/
STROLLER/HI FLOW STROLLER MENTRE È
COLLEGATO ALL’UNITÀ STAZIONARIA.
AVVERTENZA: SE SPRINT/STROLLER/HI FLOWSTROLLER NON SI STACCA FACILMENTE, NON
ESERCITARE FORZA. LE UNITÀ POTREBBERO
ESSERSI CONGELATE ADERENDO L’UNA ALL’ALTRA.
LASCIARE LE UNITÀ COLLEGATE E ATTENDERECHE SI RISCALDINO; DOPODICHÉ, SARÀ POSSIBILESEPARARLE CON FACILITÀ. NON TOCCARE
ALCUNA PARTE CONGELATA. AVVERTENZA:NEL CASO IN CUI LE PERDITE SIANO TALMENTEECCESSIVE DA PROVOCARE UN FLUSSO DI
LIQUIDO, ABBANDONARE L’AREA E CONTATTAREIMMEDIATAMENTE L’OPERATORE SANITARIO.
AVVERTENZA: L’OSSIGENO FREDDO ALLO STATO
LIQUIDO O GASSOSO PUÒ CAUSARE USTIONI DA
CONGELAMENTO. LA PRESENZA DI OSSIGENO
AUMENTA IL RISCHIO DI INCENDI.
Attenzione: in caso di perdita di liquido
dall’unità stazionaria o da Sprint/Stroller/Hi
Flow Stroller durante la separazione, mettereda parte Sprint/Stroller/Hi Flow Stroller,
assicurandosi che rimanga in posizione
verticale, lasciare la stanza e contattare
immediatamente l’operatore sanitario.
Nota: al fornitore dell'apparecchiatura: si consiglia diutilizzare i seguenti accessori per la somministrazione diossigeno con il Sprint / Stroller / Hi Flow Stroller:• Cannula nasale con tubo da 2,1 m (7 piedi) (LPM massimo: 6 LPM): numero parte CAIRE 5408-SEQ• Firebreak: numero parte CAIRE 21126636
Un incendio è raccomandato per l'uso con qualsiasi
cannula.
• CAIRE offre una protezione antincendio destinata all'uso
in combinazione con l'ossigeno portatile. L'incendio è unfusibile termico per arrestare il flusso di gas nel caso in cuila cannula a valle o il tubo dell'ossigeno vengano accesi ebrucino all'incendio. È posizionato in linea con la cannula
nasale o il tubo dell'ossigeno tra il paziente e l'uscita
dell'ossigeno di Sprint / Stroller / Hi Flow Stroller. Per un
uso corretto del focolaio, consultare sempre le istruzionidel produttore (incluse in ciascun kit di focolaio).
• Ulteriori informazioni sugli accessori consigliati sono
disponibili online all'indirizzo www.caireinc.com.

64 - IT PN MN236-C4 E | Manuale dell’utente

Istruzioni per l’uso(la cannula non è inclusa)

1. Premere con decisione il tubo di respirazione dellacannula sul connettore del tubo dell’ossigeno.
2. Regolare la posizione del tubo di respirazione dellacannula in modo da poter respirare comodamente.
3. Ruotare in senso orario la manopola di regolazionedel flusso sulla sommità di Sprint/Stroller/Hi Flow
Stroller finché nella "finestra" della manopola nonviene visualizzata la portata prescritta (numero) e
non si avverte un arresto positivo.
Attenzione: impostando la manopola a unvalore tra le portate, ne conseguirà un flusso
di ossigeno fuori dalle specifiche.
4. A questo punto, si dovrebbe ricevere ossigeno.
Per interrompere il flusso di ossigeno, ruotare la
manopola di regolazione del flusso in posizione
off (0).
Attenzione: ruotare sempre la manopola diregolazione del flusso in posizione off (0)quando l’unità non viene utilizzata.

Sprint / Stroller / Hi Flow Stroller
5. Usare la seguente tabella come linea guida utile adeterminare la durata operativa di Sprint/Stroller/Hi
Flow Stroller:
Impostazioni FCVModello

SprintStroller

Hi Flow
Stroller

NominaleNominale

Pos.
FCV

SprintStroller

Hi Flow
Stroller

Spento

30:00

60:00

71:00

0,25

18:00

30:00

ND

0,5

12:12

23:00

32:00

0,75

09:18

17:24

ND

1

07:30

14:06

16:00

1,5

05:30

10:12

ND

2

04:18

08:00

08:00

2,5

03:36

06:36

06:20

3

03:06

05:36

05:20

4

02:24

04:24

04:00

5

02:00

03:36

ND

6

01:42

03:00

02:40

8

N/D

N/D

02:00

10

N/D

N/D

01:40

12

N/D

N/D

1:20

15

N/D

N/D

01:00

Nota: la durata è espressa in ore e minuti(formato 00:00).
Nota: la durata “Nominale” fa riferimento a condizioni
ideali, ossia riempimento massimo, portate esatte,buon tasso di perdita, unità non spostata, ecc.
Queste sono le durate massime previste.
Nota: i risultati individuali varieranno.
6a. Per verificare il livello di ossigeno liquido
nell’unità con l’indicatore elettronico di livello delliquido:• Indicatore Gen 4: premere il pulsante di funzionamento
verde sul quadrante dell’indicatore. Controllare l’arco dispie LED per verificare il livello di contenuto.

Attenzione: Sprint/Stroller/Hi Flow Stroller è
vuoto se risulta acceso solo l’ultimo segmentodella barra luminosa.
Se la spia della batteria scarica si accende quando
viene premuto il pulsante, informare l’operatore
sanitario.
Nota: finché nell’unità è presente ossigeno liquido, Sprint/
Stroller/Hi Flow Stroller continua a erogare ossigeno anche incaso di accensione della spia della batteria scarica.
6b. Per verificare il livello di ossigeno liquido
nell’unità con l’indicatore meccanico di livello delliquido:• Afferrando la cinghia di stoffa, tenere l’unità
direttamente al di sopra della camera della bilancia
a molla con indicatore del contenuto nera.
• Far stabilizzare l’unità (non deve andare su e giù),quindi leggere il valore sulla scala colorata indicanteil contenuto approssimativo di ossigeno liquido.

• Indicatore Gen 3: premere il pulsante (interruttore dellivello del liquido) sulla parte superiore dell’unità peralmeno due secondi. Leggere sulla parte superioredella barra luminosa il livello di contenuto.

PN MN236-C4 E | Manuale dell’utente IT - 65

Sprint / Stroller / Hi Flow Stroller
Nota: se l’unità è vuota, la bilancia mostra soltanto la
banda rossa. In questo caso, l’unità deve essere riempita
prima dell’uso.
7. In determinate condizioni ambientali, con unutilizzo continuo e soprattutto in assenza dimovimento, l’unità portatile può sviluppare
un accumulo di brina attorno alla bobina delvaporizzatore e sul telaio. Tale accumulo può essere
ridotto dando dei colpetti sull’unità e/o rimuovendo
la brina dal telaio.

Manutenzione dell’assorbitore dicondensa
Quando l’unità è vuota e ha raggiunto la temperatura
ambiente, rimuovere l’eventuale condensa dalrecipiente per condensa:
Involucro duro
• Inclinare da un lato l’unità. Afferrare la
manopola al centro del recipiente per condensa(alla base dell’unità) e tirarla in linea retta. Il
recipiente si stacca.
Nota: se è molto sporco, l’assorbitore può essere lavatoin lavatrice (con l’aggiunta di candeggina per un’azione
disinfettante).
• Per riposizionare il recipiente, centrarlo
sull’apertura alla base dell’unità e inserirlo
premendo la manopola in modo che faccia
un solo scatto; premere ancora la manopola
facendole fare un secondo scatto per bloccare il
recipiente in posizione.

• Aprire la cerniera inferiore e il coperchio
inferiore.
• Rimuovere l’assorbitore dal recipiente e
strizzarlo per eliminare l’acqua assorbita. Farasciugare completamente l’assorbitore prima direinserirlo nel recipiente.
Nota: se è molto sporco, l’assorbitore può essere lavatoin lavatrice (con l’aggiunta di candeggina per un’azione
disinfettante).
• Per riposizionarlo, inserirlo nel coperchio
inferiore e richiudere la cerniera inferiore.
Nota: se non è possibile premere la manopola per
inserire il recipiente, fare presa su quest’ultimo e tirare
la manopola finché non scatta in posizione aperta. A
questo punto riposizionare il recipiente come descritto inprecedenza.

Riduzione della brina
In determinate condizioni ambientali, con un utilizzo
continuo e soprattutto in assenza di movimento,
l’unità portatile può sviluppare un accumulo di brina
attorno alla bobina del vaporizzatore e sul telaio.
Tale accumulo può essere ridotto dando dei colpetti
sull’unità e/o rimuovendo la brina dal telaio.
AVVERTENZA: DATA L’EVENTUALITÀ DI
UN ACCUMULO ECCESSIVO DI BRINA/
GHIACCIO, SI CONSIGLIA DI FAR RIPOSAREHI FLOW STROLLER PER 1 ORA DOPOL’UTILIZZO PER RIDURRE LA FORMAZIONE
DI UMIDITÀ.

Bloccata
Inserita
ApertaRecipiente a manopola

Nota: se non è possibile premere la manopola per inserire
il recipiente, fare presa su quest’ultimo e tirare la manopola finché non scatta in posizione aperta. A questo punto
riposizionare il recipiente come descritto in precedenza.
Involucro morbido
66 - IT PN MN236-C4 E | Manuale dell’utente

Le maggiori portate che caratterizzano Hi Flow Stroller
fanno sì che sull’unità si accumulino condensa e brina,
soprattutto con impostazioni del flusso da 10 l/min insu. Si consiglia di lasciare a riposo l’unità per almeno
1 ora dopo l’utilizzo per ridurre la possibilità diaccumulo eccessivo risultante da un uso continuo.

Sprint / Stroller / Hi Flow Stroller
Risoluzione dei problemi
Le seguenti informazioni sono utili per risolvere eventuali problemi di funzionamento dell’apparecchiatura
riscontrabili durante l’utilizzo di Sprint/Stroller/Hi Flow Stroller.
Problema
Soluzione
L’elemento portatile emette • Il sibilo può verificarsi quando l’elemento portatile mantiene la corretta pressione diun sibilo.
esercizio. Il sibilo è più probabile dopo il riempimento o quando la posizione dell’elemento
portatile cambia. Il sibilo può durare circa 10 minuti dopo il riempimento. Inoltre, puòverificarsi quando la valvola di regolazione del flusso è impostata su un valore basso.

Il flusso dell’elemento
portatile si arresta durante
l’uso.

L’elemento portatile non si
riempie.
La valvola di sfiatodell’elemento portatile nonsi chiude correttamente
alla fine del processo diriempimento.

L’elemento portatile nonsi scollega facilmente
dal serbatoio dopo ilriempimento.

• Se l’elemento portatile è stato tenuto in posizione errata, rimetterlo in verticale e attenderequalche minuto affinché si stabilizzi• Accertarsi che la cannula sia ben collegata all’uscita dell’ossigeno.
• Accertarsi che la cannula non sia attorcigliata.
• Controllare l’indicatore del contenuto/livello e riempire l’unità portatile se necessario.
• Accertarsi che la manopola di regolazione del flusso non sia in posizione off ("0").
• Verificare che sia presente ossigeno all’interno del serbatoio.
• Verificare che i connettori di riempimento sull’elemento portatile e sul serbatoio siano fissati
saldamente nel corso dell’intero processo di riempimento.
• Se la valvola di sfiato non si chiude e il sibilo e la nuvola di vapore di ossigeno persistono,
rimuovere con cautela l’elemento portatile premendo il pulsante di rilascio sul serbatoio.
La fuoriuscita dal fondo dell’elemento portatile si arresterà dopo qualche minuto. Lasciare
che l'unità si scaldi fino a quando non si riesce a chiudere la valvola di sfiato. Potrebbero
essere necessari ben 60 minuti prima che all’interno dell’elemento portatile si ripristiniuna pressione adeguata a garantire un flusso di ossigeno accurato. Se necessario,
utilizzare una fonte di ossigeno alternativa, come ad esempio una valvola di regolazionedel flusso collegata al serbatoio.
• I connettori di riempimento dell’elemento portatile e del serbatoio potrebbero essersi
congelati.
• NON ESERCITARE FORZA. Attendere qualche minuto prima che le parti congelate si
riscaldino, quindi staccare l’elemento portatile quando il ghiaccio si sarà sciolto. Per
evitare che le unità si congelino e aderiscano l’una all’altra, pulire sempre il connettore diriempimento maschio sul serbatoio e il connettore di riempimento femmina sull’elemento
portatile con un panno pulito e asciutto prima di effettuare il riempimento.

PN MN236-C4 E | Manuale dell’utente IT - 67

Sprint / Stroller / Hi Flow StrollerStandard di pulizia
AVVERTENZA: PULIRE L’UNITÀ SOLO DOPOAVERLA SVUOTATA.
• Pulire utilizzando un detersivo per stoviglie
neutro e acqua.
• Applicare la soluzione detergente direttamente
su un panno privo di sfilacciature. I detergenti
approvati includono HydroPure e HydroKlean.
Non spruzzare detergenti direttamente su Sprint/
Stroller/Hi Flow Stroller.
• Strofinare la superficie esterna con il panno privo disfilacciature fino a quando essa non risulta pulita.
Attenzione: non utilizzare apparecchiature dipulizia ad alta temperatura e alta pressione
per pulire queste unità.
• Non applicare il detergente sui componenti
interni o sulle valvole.
• Lasciare asciugare completamente l’unità primadell’utilizzo.
Nota: per l’operatore sanitario: per le procedure di
rigenerazione, vedere il manuale di assistenza applicabile.

RAEE e RoHS

Questo simbolo ha lo scopo di ricordare ai
proprietari dell’apparecchio di consegnarlopresso un centro per il riciclaggio al terminedella durata utile, in conformità alla Direttiva sui rifiuti
di apparecchiature elettriche ed elettroniche (RAEE).
I nostri prodotti sono conformi alle restrizioni dettate
dalla Direttiva sulle sostanze pericolose (RoHS). È
possibile che essi contengano tracce, ma non quantità
rilevanti, di piombo o altri materiali pericolosi.

Manutenzione
Il fornitore di servizi è responsabile di qualsiasiintervento di manutenzione richiesto secondo ilmanuale tecnico di questo dispositivo. Contattare
il fornitore di servizi per qualsiasi richiesta dimanutenzione.
L’assorbitore di condensa è l’unico componente della
cui manutenzione può occuparsi l’utente. L’utente nonpuò occuparsi della manutenzione di altri componenti.
La durata prevista è di almeno cinque anni.
Pulire i connettori di riempimento tra un riempimento
e l’altro con un panno pulito, asciutto e privo disfilacciature sia sulle unità portatili sia su quellestazionarie, al fine di evitare il congelamento ed
eventuali guasti dell’apparecchiatura.
Nota: eventuali interventi di manutenzione supplementari
dovranno essere eseguiti da un tecnico dell’assistenzaqualificato o da un fornitore di assistenza.

Accessori

Smaltimento
Restituire sempre Sprint/Stroller/Hi Flow Stroller,
compresi tutti i suoi componenti, al fornitore di
assistenza domiciliare per il corretto smaltimento.
È inoltre possibile rivolgersi agli uffici locali per
istruzioni sul corretto smaltimento della batteria.

Trasporto e immagazzinaggio
Il dispositivo dovrà essere conservato in posizione
verticale e in un’area ben ventilata. Non appoggiare
il dispositivo sul lato. Umidità fino al 95% senzacondensa. Intervallo di temperatura da -40 °C a 70 °C(da -40 °F a 158 °F).
Intervalli di temperatura di esercizio da -10 °C a 40 °C(da 14 °F a 104 °F). Intervallo di umidità relativa dal
30% al 75% senza condensa.
Nota: l’intervallo di pressione atmosferica è compreso tra
700 hPa e 1060 hPa (altitudine da 3.000 m a -300 m).

68 - IT PN MN236-C4 E | Manuale dell’utente

Carrello per il trasporto PN 20765565

Sprint / Stroller / Hi Flow Stroller
Sicurezza
AVVERTENZA: GLI APPARECCHI DI COMUNICAZIONE A RF PORTATILI (COMPRESE LE PERIFERICHE,COME CAVI PER ANTENNE E ANTENNE ESTERNE) VANNO UTILIZZATI A NON MENO DI 30 CM(12 POLLICI) DA QUALSIASI PARTE DI SPRINT/STROLLER/HI FLOW STROLLER, INCLUSI I CAVISPECIFICATI DAL COSTRUTTORE. DIVERSAMENTE, L’APPARECCHIO POTREBBE SUBIRE UNADIMINUZIONE DELLE PRESTAZIONI.
AVVERTENZA: L’UTILIZZO DI ACCESSORI, TRASDUTTORI E CAVI DIVERSI DA QUELLI SPECIFICATI O VENDUTIDAL PRODUTTORE DI QUESTA APPARECCHIATURA PUÒ PROVOCARE UN AUMENTO DELLE EMISSIONIELETTROMAGNETICHE O UNA RIDUZIONE DELL’IMMUNITÀ ELETTROMAGNETICA DI QUESTA APPARECCHIATURA
E, CONSEGUENTEMENTE, UN FUNZIONAMENTO NON CORRETTO.
AVVERTENZA: EVITARE L'USO DI QUESTA APPARECCHIATURA VICINO O SOPRA ALTRE APPARECCHIATUREPERCHÉ POTREBBE COMPORTARE UN FUNZIONAMENTO NON CORRETTO. SE TALE USO È NECESSARIO,QUESTA APPARECCHIATURA E LE ALTRE DEVONO ESSERE CONTROLLATE PER VERIFICARE CHE STIANOFUNZIONANDO NORMALMENTE.
Attenzione: i dispositivi elettromedicali richiedono speciali precauzioni rispetto alla compatibilità
elettromagnetica (CEM) e vanno installati e messi in servizio secondo le informazioni sulla CEM fornite inquesto manuale.
Attenzione: le apparecchiature di comunicazione a radiofrequenza (RF) portatili e mobili possono avere
effetti sui dispositivi elettromedicali.
Tabella 1

Linee guida e dichiarazione del produttore – Emissioni elettromagnetiche
Sprint/Stroller/Hi Flow Stroller è destinato all’uso nell’ambiente elettromagnetico descritto di seguito. Il cliente
o l’utilizzatore di Sprint/Stroller/Hi Flow Stroller deve garantirne l’uso in un ambiente di questo tipo.

Test delle emissioni

Conformità

Ambiente elettromagnetico - Linee guida

Emissioni RF

Gruppo 1

Sprint/Stroller/Hi Flow Stroller usa energia RF solo per
le funzioni interne.

CISPR 11Pertanto, le emissioni RF sono molto basse ed è
improbabile che possano causare interferenze negli
apparecchi elettronici nelle vicinanze.

Emissioni RF CISPR 11

Classe B

Emissioni armoniche
IEC 61000-3-2

Nonapplicabile

Fluttuazioni di tensione/
emissioni di sfarfallii
IEC 61000-3-3

Nonapplicabile

Sprint/Stroller/Hi Flow Stroller è idoneo all’uso intutti gli edifici, compresi quelli a uso abitativo e quelli
collegati direttamente alla rete pubblica di alimentazione
a bassa tensione utilizzata per rifornire gli edifici a uso
abitativo.

PN MN236-C4 E | Manuale dell’utente IT - 69

Sprint / Stroller / Hi Flow Stroller
Tabella 2*: Distanze di separazione consigliate tra apparecchiature
di comunicazione RF portatili e mobili e Sprint/Stroller/Hi Flow Stroller
Sprint/Stroller/Hi Flow Stroller è destinato all’uso in un ambiente elettromagnetico in cui i disturbi da RF irradiata sono tenuti sotto
controllo. Il cliente o l’utilizzatore di Sprint/Stroller/Hi Flow Stroller può prevenire le interferenze elettromagnetiche mantenendo una
distanza minima tra le apparecchiature di comunicazione RF portatili e mobili (trasmettitori) e Sprint/Stroller/Hi Flow Stroller comeindicato di seguito, calcolando tale distanza in base alla potenza massima di uscita delle apparecchiature di comunicazione.
Potenza di uscita
Distanza di separazione in base alla frequenza del trasmettitore
nominale massima delm
trasmettitore
Da 80 MHz a 800 MHzd=1,2
Da 800 MHz a 2,5 GHz
Da 150 kHz a 80 MHz

d = 1,2√P

d = 1,2√P

d = 2,3√PW
0,010,12 m0,12 m0,23 m0,10,38 m0,38 m0,73 m
1
1,2 m
1,2 m
2,3 m
10
3,8 m
3,8 m
7,3 m
100
12 m
12 m
23 m
Per i trasmettitori con una potenza di uscita massima non elencata sopra, la distanza di separazione consigliata (d) in metri (m)può essere stimata utilizzando l’equazione applicabile alla frequenza del trasmettitore, dove P è la potenza di uscita massima deltrasmettitore in watt (W) ottenuta in base alle informazioni fornite dal produttore.
NOTA 1: a 80 MHz e 800 MHz, si applica la distanza di separazione per l’intervallo di frequenza superiore.
NOTA 2: queste linee guida potrebbero non essere applicabili in tutte le situazioni. La propagazione elettromagnetica è influenzata
dall’assorbimento e dalla riflessione di strutture, oggetti e persone.

* Questa tabella è inclusa come requisito standard per le apparecchiature che sono state testate secondo livelli ditest specifici e su intervalli di frequenza specifici e sono state ritenute conformi ai regolamenti.

Tabella 3: Linee guida e dichiarazione del produttore - Immunità elettromagnetica
Sprint/Stroller/Hi Flow Stroller è destinato all’uso nell’ambiente elettromagnetico descritto di seguito. Il cliente o l’utilizzatore di
Sprint/Stroller/Hi Flow Stroller deve garantirne l’uso in un ambiente di questo tipo.
Test dell’immunità
Livello di test IEC 60601
Livello di conformità
Ambiente elettromagnetico - Linee guida
Il pavimento deve essere di legno, cemento
Scarica elettrostatica±8 kV a contatto±8 kV a contattoo piastrelle in ceramica. Se i pavimenti sono(ESD)
± 2 kV, ± 4 kV
± 2 kV, ± 4 kV
ricoperti di materiale sintetico, l’umidità relativa
±8
kV,
±
15
nell’aria
±8
kV,
±
15
nell’aria
IEC 61000-4-2
deve essere almeno del 30%.**Non applicabile±2 kV per linee di alimentazione

Dispositivo alimentato
con CC±1 kV per le linee di ingresso/uscita

Non applicabile±1 kV da linea a linea±2 kV da linea a terra

Non applicabile
Dispositivo alimentato
con CCNon applicabile

Non applicabile Dispositivo alimentato con CC

Non applicabile

IEC 61000-4-11<5% UT (>95% di calo inUT) per 0,5 cicli
40% UT (60% di calo in UT)
per 5 cicli
70% UT (30% di calo in UT)
per 25 cicli<5% UT (>95% di calo inUT) per 5 s

Campo magneticoa frequenza di rete
(50/60 Hz)

3A/m
50/60 Hz

3A/m
50/60 HzI campi magnetici alla frequenza di alimentazione devono essere ai livelli tipici degli ambienti
commerciali o ospedalieri.

Transitori/Raffiche di
impulsi elettrici veloci
IEC 610004-4
Sovratensionetransitoria
IEC 61000-4-5
Cali di tensione, brevi
interruzioni e variazioni della tensione
di alimentazione sulle
linee di ingresso
dell’alimentazione

IEC 61000-4-8

Non applicabile

Nessuna linea di ingresso/uscita dati

Nota: UT indica la tensione di rete c.a. prima dell’applicazione del livello di test.

** Questa dichiarazione indica che il test richiesto è stato eseguito in un ambiente controllato e che Sprint/Stroller/Hi Flow Stroller è stato ritenuto conforme ai regolamenti.
70 - IT PN MN236-C4 E | Manuale dell’utenteSprint / Stroller / Hi Flow Stroller
Tabella 4Linee guida e dichiarazione del produttore – Immunità di sistemi e apparecchiature EM

Linee guida e dichiarazione del produttore – Immunità
Sprint/Stroller/Hi Flow Stroller è destinato all’uso nell’ambiente elettromagnetico descritto di seguito. Il cliente
o l’utilizzatore di Sprint/Stroller/Hi Flow Stroller deve assicurarne l’uso in un ambiente di questo tipo.
Test dell’immunità
Livello di test IEC 60601
Livello di conformità
Ambiente elettromagnetico – Linee guida
RF condotta IEC
61000-4-6

3 Veff
6 Veff (nelle bande ISM)
Da 150 kHz a 80 MHz

Non applicabile
Dispositivo a batteria,senza SIP/SOP

Le apparecchiature di comunicazione a RF mobili
e portatili devono essere utilizzate a una distanzada qualsiasi componente di Sprint/Stroller/Hi Flow
Stroller, compresi i cavi, superiore a quella calcolata
in base all’equazione applicabile alla frequenza deltrasmettitore.
Distanza di separazione consigliata
d = 1,2 √P

RF irradiata IEC
61000-4-3

Da 80 MHz a 2,7 GHz

10 V/m80 MHz–2,7 GHz
80% AM a 1 kHz

d = 1,2 √P
d = 2,3 √Pdove P è la potenza di uscita nominale massima deltrasmettitore in watt (W) secondo le indicazioni delproduttore e d è la distanza di separazione consigliata
in metri (m).
Le intensità di campo dai trasmettitori RF fissi,
determinate mediante un’indagine elettromagnetica
nella sedea, devono essere inferiori al livello diconformità in ogni intervallo di frequenzab.
Le interferenze possono verificarsi nei pressi delleapparecchiature contrassegnate da questo simbolo:

Potenza
massima(W)

Distanza(m)Livello del test
dell’immunità
(V/m)

1,8

0,3

27

2

0,3

28

Modulazione degli
impulsib)
217 Hz

0,2

0,3

9

GSM 800/900, TETRA 800,
IDEN 820, CDMA 850,
banda LTE 5

Modulazione degli
impulsib)
18 Hz

2

0,3

28

1700-1900

GSM 1800; CDMA 1900;
GSM 1900; DECT; banda
LTE 1, 3, 4, 25; UMTS

Modulazione degli
impulsib)
217 Hz

2

0,3

28

2450

2400-2570

Modulazione degli
impulsib)
217 Hz

2

0,3

28

5240
5500
5785

Bluetooth, WLAN, 802.11
b/g/n, RFID 2450, banda
LTE 75100-5800

WLAN 802.11 a/n

Modulazione degli
impulsib)
217 Hz

0,2

0,3

9

Frequenza deltest (MHz)Bandaa)(MHz)

Servizioa)

385

380-390TETRA 400450

430-470

GMRS 460, FRS 460

704-787

Banda LTE 13, 17

800-960

710
745
780
810
870
930
1720
1845
1970

Modulazioneb)
Modulazione degli
impulsib)
18 Hz
FMC)
±5 kHz deviazione
1 kHz sinusoidale

Nota: se necessario per raggiungere il LIVELLO DEL TEST DELL’IMMUNITÀ, la distanza fra l’antenna trasmittente e
Sprint/Stroller/Hi Flow Stroller può essere ridotta a 1 m. La distanza di test di 1 m è consentita da IEC 61000-4-3.
Per alcuni servizi, sono incluse solo le frequenze di uplink.
La portante deve essere modulata usando un segnale d’onda quadra con duty cycle del 50%.
In alternativa alla modulazione FM, può essere usata la modulazione degli impulsi del 50% a 18 Hz in quanto, sebbene non rappresenti lamodulazione effettiva, sarebbe il caso peggiore.
abc

CAIRE e CAIRE Inc. sono marchi registrati di CAIRE Inc. Visitare il nostro sito Web di seguito per un elenco completo dei marchi. Marchi: www.careinc.com/corporate/trademarks.
Copyright © 2023 CAIRE Inc. CAIRE Inc. si riserva il diritto di interrompere la produzione o di modificare prezzi, materiali, apparecchiature, qualità, descrizioni, specifiche e/o processi dei
propri prodotti in qualsiasi momento, senza preavviso e senza ulteriori obblighi o conseguenze. Tutti i diritti non espressamente indicati nel presente documento sono riservati, laddove
applicabili.

PN MN236-C4 E | Manuale dell’utente IT - 71

Sprint / Stroller / Hi Flow Stroller
Glossário de símbolos
ISO 7000Não fume perto da unidade ou quando aestiver a utilizar. Reg. # P002

Manter afastado da chuva, manter seco.
Reg. # 0626

Equipamento aplicado tipo BF (graude proteção contra choques elétricos).
Reg. # 5333

Limite de empilhamento por número.
Reg. # 2403
Nome e morada do fabricante.
Reg. # 3082

Aviso. Reg. # W001
Diretiva do Conselho 93/42/EEC

O país e a data de fabricação. O “CC”
identifica o código de país de duas letras
do país de fabricação. A data de fabricação está no formato AAAA-MM-DD.
Reg. # 6049
Atenção, consulte os documentos
incluídos Reg. # 0434A
Número de catálogo. Reg. # 2493
Número de série. Reg. # 2498
Faixa de temperatura de armazenamento
ou operação. Reg. # 0632

ADR

Gás não tóxico.

Faixa de umidade de armazenamento.
Reg. # 2620
Limites de pressão atmosférica. Reg.
# 2621
Este lado para cima. Reg. # 0623
Frágil, manusear com cuidado.
Reg. # 0621

Perigo Substâncias oxidantes: risco deintensificação de incêndio. Os agentes
oxidantes fazem com que os incêndios
queimem com mais vigor.
Líquido Refrigerado, USP; Produzido
por liquefação de ar
Símbolos Adicionais
Mantenha sempre a unidade bem
ventilada

Contém substâncias perigosas. Reg. #3723

Manter afastado de materiais
inflamáveis, óleo e gorduras.

Importador. Reg. # 3725
ISO 7010

Representante autorizado na Suíça.

O contacto com líquidos frios, oxigénio
gasoso, ou peças congeladas podeprovocar queimaduras de frio. Aviso
de temperatura baixa. Visa alertar paracondições de temperatura baixa ou decongelamento. Reg. # W010
O manual de instruções tem de ser lido.
Reg. # M002
Manter afastado de chamas e faíscas. É
proibido fumar e são proibidas fontes deignição abertas. Reg. # P003

72 - PT PN MN236-C4 E | Manual do utilizador

Representante autorizado naUnião Europeia
Se a etiqueta UDI (identificador exclusivo
de dispositivo) do produto contiver osímbolo CE ####, o dispositivo cumpriráos requisitos da Diretiva 93/42 / CEE
relativa a dispositivos médicos. O símbolo
CE #### indica o número do organismo
notificado.
Este dispositivo está em conformidade
com os requisitos da Diretiva
2010/35/EU referentes a dispositivos
médicos. O dispositivo possui a
marcação pi, tal como é mostrado.

IEC 60417

Se o dispositivo tiver a marca UKCA
conforme mostrado com UKCA####
indicando o número do organismo
notificado, este dispositivo está em conformidade com os regulamentos UKCA.
Não cubra a unidade nem carregue
a unidade portátil sob a roupa. Essas
unidades normalmente liberam oxigênio.
nº 5641

Sprint / Stroller / Hi Flow Stroller
21 CFR 801.15
A lei federal restringe a venda deste
dispositivo a ou por ordem de um
médico.
IEC 60601-1
À prova de gotejamento
Diretiva do Conselho 2012/19/EU
WEEE
Este símbolo é para lembrar os proprietários do equipamento de devolvê-lo a
uma instalação de reciclagem no final desua vida útil, de acordo com a Diretiva
de Resíduos de Equipamentos Elétricose Eletrônicos (WEEE).
Nossos produtos estarão em conformidade com a restrição da diretiva desubstâncias perigosas (RoHS). Elesnão conterão mais do que vestígios dechumbo ou outro conteúdo de material
perigoso.

ISO 15223-1
Aparelho médico. Reg. # 5.7.7
Identificador de dispositivo exclusivonº 5.7.10
Este produto pode estar abrangido por uma ou mais
patentes dos EUA e internacionais. Visite o nosso
website abaixo para obter uma lista das patentes
aplicáveis. Pat.: www.caireinc.com/corporate/patents/.

Especificações
• Modo de funcionamento: Fluxo Contínuo• Tipo de proteção contra choques elétricos: Equipamento alimentado internamente• Grau de proteção contra choques elétricos: Peça aplicada tipo BF• Classificação de acordo com o grau de proteção contra a entrada de água: IP22 – Equipamento comum• Equipamento não adequado para utilização na presença de misturas inflamáveis
Especificações do produto

SprintStroller

Hi Flow Stroller

Capacidade de LOX

0,68 kg (1,5 lb)

1,36 kg (3,0 lb)

1,36 kg (3,0 lb)

Capacidade gasosa
equivalentePeso, Vazio

513 L

1,026 L

1,025 L2,04 kg (4,5 lb)

2,27 kg (5,0 lb)

2,49 kg (5,5 lb)

Peso, Cheio

2,72 kg (6,0 lb)

3,63 kg (8,0 lb)

3,86 kg (8,5 lb)

Altura298 mm (11,75 pol.)343 mm (13,5 pol.)343 mm (13,5 pol.)Dimensões

123,83 mm (4,875”) P x
191 mm (7,5”) L
4,3 horas

149,23 mm (5,875”) P x
191 mm (7,5”) L
8,5 horas

149 mm (5,875") P x
191 mm (7,5”) L
8,5 horas

Tempo de utilização típico
a 2 LPM
Pressão de funcionamento

20 psi (137 kPa)

20 psi (137 kPa)

20 psi (137 kPa)

Taxa de evaporação normal

0,57 kg/dia (1,3 lb/dia)

0,57 kg/dia (1,3 lb/dia)

0,57 kg/dia (1,3 lb/dia)Intervalos de controlo defluxo padrão
Precisão dos fluxómetros*

Desligado, 0,25; 0,5; 0,75; 1; Desligado, 0,25; 0,5; 0,75; 1; Desligado, 0,5, 1, 2, 2,5, 3, 4,
1,5; 2; 2,5; 3; 4; 5; 6 LPM
1,5; 2; 2,5; 3; 4; 5; 6 LPM
6, 8, 10, 12, 15 LPM+/- 10%+/- 10%+/- 10%* Esta precisão só se verifica a 20 °C e 14,7 psig, e com um medidor de fluxo de massa preciso e calibrado.

PN MN236-C4 E | Manual do utilizador PT - 73

Sprint / Stroller / Hi Flow Stroller
Informações de alerta
Importante: Leia atentamente este manual antes deutilizar o Sprint/Stroller/Hi Flow Stroller.
Apenas com receita médica.
AVISO: ESTE DISPOSITIVO NÃO SE DESTINAA SER UTILIZADO COMO MEIO DE SUPORTE
DE VIDA.
AVISO: SE LHE PARECER QUE O EQUIPAMENTO NÃOESTÁ A FUNCIONAR CORRETAMENTE, LIGUE AO SEUPRESTADOR DE CUIDADOS DE SAÚDE. NÃO TENTEREPARAR OU AJUSTAR A UNIDADE PESSOALMENTE.
AVISO: NÃO MODIFIQUE ESTE EQUIPAMENTO SEM
AUTORIZAÇÃO DO FABRICANTE.
AVISO: SE SE EXIGIR UMA CONTINUIDADE NO
FORNECIMENTO DE OXIGÉNIO, CERTIFIQUE-SE DE
QUE EXISTE UMA FONTE DE OXIGÉNIO ADEQUADAE/OU UMA FONTE DE OXIGÉNIO SECUNDÁRIADISPONÍVEL A TODO O INSTANTE DURANTE ATERAPIA.
AVISO: NÃO PERMITA A PRESENÇA DE VELAS OU
CHAMAS NEM PERMITA QUE SE FUME A MENOS DE 3
METROS DO DISPOSITIVO, OU A MENOS DE 20 CM DEQUALQUER FONTE DE IGNIÇÃO.
AVISO: MANTENHA A UNIDADE NUMA ÁREA BEM
VENTILADA.
AVISO: NÃO GUARDE EQUIPAMENTOSCOM OXIGÉNIO LÍQUIDO NUM ARMÁRIO,NO PORTA-BAGAGENS DO CARRO, OU
NOUTRA ÁREA CONFINADA. NÃO COLOQUE
COBERTORES, CORTINADOS, OU OUTROSTECIDOS SOBRE O EQUIPAMENTO.
AVISO: ESTE PRODUTO PODE EXPÔ-LO A
SUBSTÂNCIAS QUÍMICAS COMO O NÍQUEL, QUE ÉCONSIDERADO UM AGENTE CANCERÍGENO PELOESTADO DA CALIFÓRNIA. PARA MAIS INFORMAÇÕES,VISITE WWW.P65WARNINGS.CA.GOV.
AVISO: NO CASO DE OCORRER UM INCIDENTE SÉRIOCOM ESTE DISPOSITIVO, O USUÁRIO DEVE RELATARIMEDIATAMENTE O INCIDENTE AO FORNECEDOR
E / OU AO FABRICANTE. UM INCIDENTE GRAVE É
DEFINIDO COMO UMA LESÃO, MORTE OU POTENCIAL
DE CAUSAR LESÃO / MORTE, CASO OCORRA UMA
REINCIDÊNCIA. O USUÁRIO TAMBÉM PODE RELATARO INCIDENTE À AUTORIDADE COMPETENTE DO PAÍSEM QUE O INCIDENTE OCORREU.
Atenção: Use o Sprint/Stroller/Hi Flow Stroller
apenas conforme indicado pelo seu médico.
Atenção: A lei federal restringe a venda deste dispositivo a ou por ordem de um médico.
Atenção: A unidade contém oxigénio líquido extremamente frio, a quase -184 °C (-300 ºF). A exposição atemperaturas tão baixas pode provocar leões graves
causadas pelo frio.

74 - PT PN MN236-C4 E | Manual do utilizador

Atenção: Embora sejam não inflamáveis, o oxigénio
líquido e gasoso aceleram a combustão de outros
materiais. Este perigo, juntamente com as baixas
temperaturas do oxigénio líquido, justificam certas
precauções de segurança.
Atenção: Mantenha o equipamento afastado demateriais inflamáveis. Os sprays de aerossol, óleos egorduras, incluindo cremes faciais e vaselina inflamamse facilmente e podem arder rapidamente na presençade oxigénio.
Atenção: Fumar durante a utilização de uma cânula deoxigénio pode provocar queimaduras faciais e possivelmente a morte.
Remover a cânula e colocá-la no vestuário, roupa decama, sofás ou outro material almofadado irá deflagrar
um incêndio repentino quando exposta a um cigarro,
fonte de calor, faísca ou chama.
Se for fumador(a): (1) desligue o dispositivo portátil, (2)remova a cânula e (3) saia da divisão onde o dispositivo
está localizado.
Atenção: Em caso de queda acidental, se possível, voltea colocar a unidade na vertical de imediato, mas com
cuidado. Se observar alguma fuga de oxigénio líquido,
abandone imediatamente a área e contacte o seu prestador de cuidados de saúde. Não tente mover a unidade
ou interromper a fuga de oxigénio líquido.
Nota: Não toque em peças congeladas da unidade.
Nota: Não armazene nem utilize o dispositivo portátil
acoplado à unidade fixa.
Nota: Não permita o manuseamento ou utilização dodispositivo por pessoal não qualificado.
Nota: A utilização deste dispositivo é proibida emvoos comerciais de passageiros e cargas pela Federal
Aviation Administration.

Uso pretendido

O CAIRE Sprint/Stroller/Hi Flow Stroller destina-se àadministração de oxigênio suplementar. O dispositivo
não se destina ao suporte de vida nem fornece recursosde monitoramento do paciente.

Sprint / Stroller / Hi Flow Stroller
Introdução

Controlos

O Sprint, Stroller e o Hi Flow Stroller destinam-se àadministração de oxigénio suplementar para o paciente
no domicílio do consumidor final e também podem
ser utilizados em instituições como lares de idosos
ou instalações de cuidados sub-intensivos. O Sprint/
Stroller/Hi Flow Stroller proporciona uma fonte deoxigénio móvel por um período de tempo alargado.

1. Medidor do nível de líquido eletrónico
2. Interruptor eletrónico do nível de líquido
3. Medidor do nível de líquido mecânico
4. Botão de controlo do fluxo
5. Conetor do tubo de oxigénio
6. Válvula de ventilação
7. Ligações para enchimento (consultar a página
seguinte para identificar a QDV)
8. Alça de transporte almofadada
9. Compressa de condensação
10. Depósito de condensação

O dispositivo é usado por pacientes com DPOC ou
com capacidade respiratória diminuída. O dispositivo éprescrito para o paciente. O dispositivo é vendido a um
fornecedor com a devida formação para a utilização
e manutenção do Sprint/Stroller/Hi Flow Stroller. O
fornecedor encarrega-se da formação do utilizador.

1

O dispositivo não foi concebido como aparelho desuporte de vida, nem possui quaisquer funcionalidadesde monitorização do doente. É recomendável que se
possua uma fonte alternativa de oxigénio suplementar
na eventualidade de uma falha mecânica. Estes
dispositivos portáteis possuem também uma válvula
de desconexão rápida que pode ser acoplada a um
reservatório de oxigénio líquido para a trasfega dodispositivo portátil. O reservatório é abastecido pelo
seu prestador de cuidados de saúde.

2
4
5
8

Este manual do utilizador contém as instruções deutilização do Sprint/Stroller/Hi Flow Stroller. Consulteo manual do utilizador fornecido com a unidade fixa
para obter informações sobre o funcionamento da
mesma.
Nota: O prestador de serviços irá ajudá-lo na configuração
inicial e informá-lo acerca da utilização e manuseamento
adequados da unidade.
57

6
Unidades Sprint, Stroller e Hi Flow Stroller

10
9
PN MN236-C4 E | Manual do utilizador PT - 75

Sprint / Stroller / Hi Flow Stroller
Identificação da válvula dedesconexão rápida (QDV)
Conetor da
unidadeportátil

Conetor da
unidade fixa

Modelo de QDVCAIRE
Bloqueio
EnchimentoSFlateral

4. a. Se o seu conetor for de pressão (push-on):Coloque a unidade Sprint/Stroller/Hi Flow Stroller
sobre a unidade fixa até sentir o conetor encaixar.

LIFE-OX
Bloqueio
TF
PENOX

TAEMA
TF
Push-On(pressão)PB

Instruções de utilização
Para encher o Sprint/Stroller/Hi Flow Stroller
de enchimento superior
1. Limpe os conetores de enchimento da unidade fixa
e do Stroller/Sprint/Hi Flow Stroller com um pano
limpo, seco e sem fiapos.
Atenção: Os conetores de enchimento têm deestar secos, uma vez que a humidade podefazer com que o Sprint/Stroller/Hi Flow e a
unidade fixa congelem em conjunto.
2. Certifique-se de que os botões de controlo do fluxo
do Sprint/Stroller/Hi Flow Stroller e do reservatório
se encontram na posição de desligado (0).
3. Coloque o conetor de enchimento na vertical sobre oconetor da unidade fixa.

76 - PT PN MN236-C4 E | Manual do utilizadorb. Se o tipo de conetor for Penox ou Life-Ox:
Rode a unidade Sprint/Stroller/Hi Flow Stroller
até sentir o conetor encaixar. Em seguida,rode o Sprint/Stroller/Hi Flow Stroller firme)
e cuidadosamente no sentido horário (pressionando para baixo.

Nota: Continuação no passo 6 Para encher o Sprint/Stroller/Hi Flow Stroller de enchimento lateral.

Para encher o Sprint/Stroller/Hi Flow Stroller
de enchimento lateral
1. Limpe os conetores de enchimento da unidade fixa
e do Sprint/Stroller/Hi Flow Stroller com um pano
limpo, seco e sem fiapos.
AVISO: OS CONETORES DE ENCHIMENTOTÊM DE ESTAR SECOS, UMA VEZ QUEA HUMIDADE PODE FAZER COM QUE OSPRINT/STROLLER/HI FLOW E A UNIDADEFIXA CONGELEM EM CONJUNTO.
2. Certifique-se de que os botões de controlo do fluxo
do Sprint/Stroller/Hi Flow Stroller e do reservatório
se encontram na posição de desligado (0).

Sprint / Stroller / Hi Flow Stroller
3. Certifique-se de que os conetores de enchimento do
Sprint/Stroller/Hi Flow Stroller se encontram na
vertical em relação ao conetor da unidade fixa.

4. Gire o Sprint/Stroller/Hi Flow Stroller no sentido) até sentir o pino encaixar na
anti-horário (ranhura.

7. Poderá ver também vapor em torno da ligação. Issoé normal devido à grande diferença de temperatura
entre o oxigénio líquido e o ar quente em torno das
unidades.
8. Quando a unidade estiver cheia, feche a válvula deventilação. A unidade está cheia quando o silvo mudade tom e sai algum líquido pela linha de ventilação.
Nota: Fechar a válvula após a descarga de um fluxo delíquido praticamente contínuo enche o Sprint/Stroller/Hi
Flow Stroller quase por completo, mas desperdiça algum
oxigénio líquido.

5. Volte a rodar cuidadosa e firmemente o Sprint/
Stroller/Hi Flow Stroller até ficar na vertical. As
unidades estão agora bloqueadas em conjunto.

Nota: Continuação no passo 6.
Nota: Em equipamentos com conetor de pressão (pushon), mantenha a unidade no devido lugar durante oenchimento.

Nota: Fechar a válvula ao primeiro sinal de descarga delíquido não enche o Sprint/Stroller/Hi Flow Stroller tão
completamente, mas poupa oxigénio líquido.
9. Com conetores do tipo “push-on” (pressão), primao botão de libertação até as unidades se separarem.
Com todos os outros conetores, rode a unidade até
esta se separar da unidade fixa.

Botão de libertação da unidade portátil(apenas no modelo “push-on”)

6. Abra a válvula de ventilação para começar a encher oSprint/Stroller/Hi Flow Stroller. Deverá ouvir algum
oxigénio a sair; não se preocupe. O som sibilante éperfeitamente normal.
Nota: As leituras do medidor do nível de líquido são mais
precisas 30 segundos após o abastecimento do Sprint/
Stroller/Hi Flow Stroller.
PN MN236-C4 E | Manual do utilizador PT - 77

Sprint / Stroller / Hi Flow Stroller
Atenção: Verifique o medidor do nível delíquido só depois de fechada a válvula deventilação.
Atenção: Se o Sprint/Stroller/Hi Flow Stroller forreabastecido imediatamente após um período deutilização, encha-o e aguarde 10 minutos após oenchimento com a FCV na posição zero para estabilizar
a pressão. Em seguida, utilize normalmente.
AVISO: NÃO UTILIZE O SPRINT/STROLLER/
HI FLOW STROLLER ACOPLADO ÀUNIDADE FIXA.
AVISO: SE O SPRINT/STROLLER/HI FLOW STROLLER
NÃO SE SEPARAR FACILMENTE, NÃO FORCE.
AS UNIDADES PODEM ESTAR CONGELADAS EMCONJUNTO. DEIXE AS UNIDADES LIGADAS EESPERE QUE ESTAS AQUEÇAM — SEPARAR-SE-ÃO
FACILMENTE. NÃO TOQUE EM PEÇAS CONGELADAS.
AVISO: SE A FUGA FOR SUFICIENTEMENTEACENTUADA PARA PROVOCAR UMA CORRENTE
DE LÍQUIDO, ABANDONE A ÁREA E TELEFONEIMEDIATAMENTE PARA O SEU PRESTADOR DE
CUIDADOS DE SAÚDE.
AVISO: O OXIGÉNIO LÍQUIDO OU O OXIGÉNIOGASOSO FRIO PODEM PROVOCAR QUEIMADURAS
DE FRIO NA PELE. O OXIGÉNIO AUMENTA O RISCO DEINCÊNDIO.
Atenção: Em caso de derramamento delíquidos da unidade fixa ou do Sprint/Stroller/
Hi Flow Stroller após a separação das
unidades, afaste o Sprint/Stroller/Hi Flow
Stroller garantindo que este fica na vertical,
abandone a divisão e contacte o seu prestador
de cuidados de saúde de imediato.
Nota: Para o fornecedor do equipamento: Os seguintes
acessórios de administração de oxigênio são recomendados para uso com o Sprint / Stroller / Hi Flow Stroller:• Cânula nasal com 7 pés (2,1 m) de tubulação (LPM
máximo: 6 LPM): Número de peça CAIRE 5408-SEQ• Firebreak: Número de peça CAIRE 21126636
Um quebra de fogo é recomendado para uso com
qualquer cânula.
• O CAIRE oferece um quebra-fogo destinado a ser usado
em conjunto com o oxigênio portátil. O quebra-fogo é um
fusível térmico para interromper o fluxo de gás no casode a cânula a jusante ou a tubulação de oxigênio ser
inflamada e queimar no fogo. É colocado em linha com
a cânula nasal ou tubo de oxigênio entre o paciente e asaída de oxigênio do Sprint / Stroller / Hi Flow Stroller.
Para o uso adequado do combate a incêndio, sempre
consulte as instruções do fabricante (incluídas em cada kit
de combate a incêndio).
• Informações adicionais sobre acessórios recomendadosestão disponíveis online em www.caireinc.com.

78 - PT PN MN236-C4 E | Manual do utilizador

Instruções de utilização

(Cânula não incluída.)1. Encaixe o tubo de respiração da cânula firmemente
no conetor do tubo de oxigénio.
2. Ajuste o tubo de respiração da cânula para aposição adequada de modo a conseguir inalar
confortavelmente o oxigénio.
3. Rode o botão de controlo do fluxo na parte superior
do Sprint/Stroller/Hi Flow Stroller no sentido
horário até a taxa de fluxo prescrita (numérica) ficar
visível na “janela” do botão e sentir uma retenção
positiva.
Atenção: Se o botão de controlo do fluxo
for posicionado entre duas taxas de fluxo,
o fluxo de oxigénio não corresponderá àsespecificações.
4. Neste momento, deverá começar a receber oxigénio.
Para interromper o fluxo de oxigénio, gire o botão
de controlo do fluxo no sentido anti-horário,colocando-o na posição de desligado(0).
Atenção: Quando não estiver a ser utilizado,
rode sempre o botão de controlo do fluxo paraa posição de desligado (0).

Sprint / Stroller / Hi Flow Stroller
5. Use o quadro seguinte como referência para
determinar o período de tempo de funcionamento
do Sprint/Stroller/Hi Flow Stroller:
Configuração da FCVModeloSprintStroller

Hi Flow
Stroller

NominalNominalPos. daFCV

SprintStroller

Hi Flow
Stroller

Desligado

30:00

60:00

71:00

0,25

18:00

30:00

NA

0,5

12:12

23:00

32:00

0,75

09:18

17:24

NA

1

07:30

14:06

16:00

1,5

05:30

10:12

NA

2

04:18

08:00

08:00

2,5

03:36

06:36

06:20

3

03:06

05:36

05:20

4

02:24

04:24

04:00

5

02:00

03:36

NA

6

01:42

03:00

02:40

8

N/A

N/A

02:00

10

N/A

N/A

01:40

12

N/A

N/A

1:20

15

N/A

N/A

01:00

Nota: Os tempos estão indicados em horas e minutos(formato 00:00).
Nota: Os tempos “nominais” referem-se a condições
ideais, ou seja, enchimento máximo, taxas de fluxo
precisas, boa taxa de perda, não deslocação da
unidade, etc.
Estes tempos são os máximos esperados.
Nota: Os seus resultados individuais podem variar.
6a. Para verificar o nível de oxigénio líquido na
unidade com o medidor do nível de líquido
eletrónico:• Medidor Gen 4: Prima o botão verde junto ao
medidor. Leia o arco de LEDs que indica o nível doconteúdo.

Atenção: O Sprint/Stroller/Hi Flow Stroller
está vazio quando apenas se acende o último
segmento da barra luminosa.
Se o indicador de bateria fraca acender quando
o botão é premido, informe o seu prestador decuidados de saúde.
Nota: Se o indicador de bateria fraca se acender, o Sprint/
Stroller/Hi Flow Stroller continuará a fornecer oxigénio
desde que exista oxigénio líquido na unidade.
6b. Para verificar o nível de oxigénio líquido na
unidade com o medidor do nível de líquido
mecânico:• Segure a unidade diretamente acima da câmara dabalança de mola do indicador de conteúdo preto.
• Permita que a unidade estabilize (isto é, evite que se
mova para cima e para baixo), e, em seguida, leia oconteúdo aproximado de oxigénio líquido na zonacolorida exposta da balança.

• Medidor Gen 3: Pressione o botão (interruptor
do nível de líquido) na parte superior da unidade
durante um mínimo de dois segundos. Leia o nível
de conteúdo indicado na parte superior da barra
luminosa.

PN MN236-C4 E | Manual do utilizador PT - 79

Sprint / Stroller / Hi Flow Stroller
Nota: Se a unidade estiver vazia, apenas a faixa vermelha
da balança ficará exposta. Se for esse o caso, a unidade
tem de ser abastecida antes da utilização.
7. Sob determinadas condições ambientais, utilização
contínua e especialmente se a unidade portátil não
for movida, poderá produzir-se geada em excesso em
torno da bobina do vaporizador e do invólucro. Podereduzir esta geada batendo levemente na unidade e/ou limpando a geada acumulada do invólucro.

Manutenção da compressa decondensação
Após o esvaziamento e aquecimento da unidade até àtemperatura ambiente, remova a humidade do depósito
de condensação.
Para invólucro rígido
• Incline a unidade para um dos lados. Agarre o
puxador no centro do depósito de condensação
(na parte inferior da unidade) e puxe para fora.
Isto libertará o depósito.
Nota: Se estiver muito suja, a compressa pode ser lavada
à máquina (com lixívia para desinfetar).
• Para voltar a colocar o depósito, centre-o sobre
a abertura redonda na parte inferior da unidadee pressione o puxador até ouvir um clique paraencaixar o depósito e, em seguida, um segundoclique para bloquear o depósito no devido lugar.

• Remova a compressa do depósito e torça-apara remover a humidade absorvida. Deixe acompressa secar completamente antes de voltar
a colocá-la no depósito.
Nota: Se estiver muito suja, a compressa pode ser lavada
à máquina (com lixívia para desinfetar).
• Para voltar a colocar a compressa, basta inseri-lana cobertura inferior e voltar a fechar o fechode correr.
Nota: Se não for possível pressionar o puxador parao interior de modo a encaixar o depósito, segure odepósito e puxe o puxador para fora até ouvir um clique,colocando-o na posição aberta. Agora, volte a colocar odepósito conforme descrito anteriormente.

Redução do gelo
Sob determinadas condições ambientais, com uma
utilização contínua e especialmente se a unidade portátil
não for movida, poderá produzir-se geada excessiva em
torno da bobina do vaporizador e no invólucro. Podereduzir esta geada batendo levemente na unidade e/ou
limpando a geada acumulada do invólucro.
AVISO: DEVIDO À POSSIBILIDADE
DE PRODUÇÃO DE HUMIDADEEXCESSIVA / ACUMULAÇÃO DE GELO,É RECOMENDÁVEL QUE O HI FLOWSTROLLER PERMANEÇA INATIVO DURANTE
1 HORA APÓS A UTILIZAÇÃO PARA REDUZIRA ACUMULAÇÃO DE HUMIDADE.

Bloqueado
Encaixado
AbertoPuxador Depósito

Nota: Se não for possível pressionar o puxador para ointerior de modo a encaixar o depósito, segure o depósito
e puxe o puxador para fora até ouvir um clique, colocando-o na posição aberta. Agora, volte a colocar o depósito
conforme descrito anteriormente.
Para invólucro flexível• Abra o fecho de correr inferior e puxe a coberturainferior.
80 - PT PN MN236-C4 E | Manual do utilizador

Devido às taxas de fluxo mais elevadas do Hi Flow
Stroller, pode produzir-se alguma condensação e geada
na unidade, particularmente nas configurações defluxo de 10 LPM ou superiores. É recomendável que a
unidade permaneça inativa durante pelo menos 1 hora
após a utilização, para reduzir a possibilidade de uma
acumulação excessiva como resultado da utilização
contínua.

Sprint / Stroller / Hi Flow Stroller
Resolução de Problemas
As informações seguintes destinam-se a ajudá-lo a resolver problemas operacionais simples que podem surgir
durante a utilização do Sprint/Stroller/Hi Flow Stroller.
Problema
Solução
A unidade portátil emite um • O som sibilante pode ocorrer para manter a pressão operacional adequada na unidadesom sibilante.
portátil. A emissão do silvo é mais provável após o enchimento ou quando a posição da
unidade portátil é alterada. O silvo pode durar cerca de 10 minutos após o enchimento.
Poderá ocorrer também quando a válvula de controlo do fluxo se encontrar numa
configuração baixa.

O fluxo da unidade portátil
é interrompido durante a
utilização.

A unidade Portátil
não enche.
A válvula de ventilação da
unidade portátil não fecha
corretamente no final doprocesso de enchimento.

A unidade portátil nãose liberta facilmente do
Reservatório após oenchimento.

• Se a unidade portátil for deitada numa posição incorreta, volte a colocá-la na verticale aguarde alguns minutos para permitir que a unidade estabilize• Certifique-se de que a cânula está firmemente ligada à Saída de oxigénio.
• Certifique-se de que a cânula não está dobrada.
• Verifique o indicador de conteúdo/medidor de nível e, se necessário, encha a unidade
portátil.
• Certifique-se de que o botão de controlo do fluxo não se encontra na posição de desligado(“0”).
• Verifique se existe oxigénio no Reservatório.
• Certifique-se de que os conetores de enchimento da unidade Portátil e do Reservatório se
encontram totalmente encaixados durante o processo de enchimento.
• Se a válvula de ventilação não fechar e se o som sibilante e a nuvem de vapor deoxigénio permanecerem, remova cuidadosamente a unidade Portátil premindo o botão
de libertação do Reservatório. A ventilação da parte inferior da unidade Portátil seráinterrompida no espaço de alguns minutos. Deixe a unidade aquecer até poder fechar aválvula de ventilação. A unidade Portátil pode requerer até 60 minutos para restabelecera pressão adequada a um fluxo de oxigénio preciso. Se necessário, use uma fonte
alternativa de oxigénio como uma válvula de controlo de fluxo ligada ao Reservatório.
• Os conetores de enchimento da unidade portátil e do reservatório podem ter congelado.
• NÃO FORCE. Permita que as peças congeladas aqueçam durante alguns minutos e, em
seguida, desengate a unidade portátil depois de o gelo ter derretido. Para evitar que
as unidades congelem em conjunto, limpe sempre o conetor de enchimento macho do
Reservatório e o conetor de enchimento fêmea da unidade Portátil com um pano limpo eseco antes do enchimento.

PN MN236-C4 E | Manual do utilizador PT - 81

Sprint / Stroller / Hi Flow Stroller
Limpeza padrãoAVISO: LIMPAR APENAS QUANDO A
UNIDADE ESTIVER VAZIA.
• Limpar usando uma solução de água e detergente
para a loiça moderado.
• Aplique a solução de limpeza diretamente
num pano sem fiapos. Os agentes de limpezaaprovados incluem o HydroPure e o HydroKlean.
Não pulverize agentes de limpeza diretamente
sobre o Sprint/Stroller/Hi Flow Stroller.
• Limpe a superfície exterior com o pano sem
fiapos até esta ficar limpa.
Atenção: Não use equipamentos de lavagem
a altas temperaturas ou de alta pressão para
limpar estas unidades.
• Não permita o contacto entre o agente de limpezae os componentes internos ou válvulas.
• Deixe a unidade secar por completo antes de autilizar.
Nota: Nota para o prestador de cuidados de saúde – para
procedimentos de reprocessamento, consulte o manual demanutenção aplicável.

Manutenção
O seu prestador de serviços é responsável pelas ações
de manutenção requeridas, de acordo com o manual
técnico do dispositivo. Para os eventuais requisitos demanutenção, contacte o seu prestador de serviços.
A compressa de condensação é a única peça reparável
pelo utilizador. Nenhuma outra peça é reparável pelo
utilizador.
A vida útil esperada do dispositivo é de, no mínimo,
cinco anos.
Limpe os conetores de enchimento das unidades fixa
e portátil com um pano limpo, seco e sem fiapos entrecada abastecimento para prevenir o congelamento e apossível falha do equipamento.
Nota: Qualquer manutenção adicional necessária deve serrealizada por um técnico de manutenção ou prestador deserviços qualificado.

AcessóriosWEEE e RoHS

Este símbolo destina-se a recordar os
proprietários do equipamento de que devem
entregá-lo a uma instalação de reciclagem
no final da sua vida útil, de acordo com a Diretiva
relativa aos Resíduos de Equipamentos Elétricos e
Eletrónicos (REEE).
Os nossos produtos estão em conformidade com adiretiva de Restrição de Substâncias Perigosas (RoHS).
Não contêm mais do que quantidades residuais dechumbo ou de outros materiais perigosos.

Eliminação
Devolva sempre o Sprint/Stroller/Hi Flow Stroller,
incluindo todos os componentes, ao seu prestador decuidados domiciliários para garantir uma eliminação
adequada. Pode também entrar em contracto com as
entidades locais para obter instruções acerca da correta
eliminação da bateria.

Transporte e armazenamento
O dispositivo deve ser armazenado na vertical enuma área bem ventilada. Não pouse o dispositivo
horizontalmente, sobre a sua parte lateral. Humidade
até 95% sem condensação. As temperaturas variam
entre os -40 °C e os 70 °C (-40 °F a 158 °F).
A temperatura de funcionamento varia entre os -10 °C
e os 40 °C (14 °F a 104 °F). A humidade relativa é de
30% a 75% sem condensação.
Nota: Os limites de pressão atmosférica são de 700 hPa a
1060 hPa (elevação de 3.000 m a -305 m).
82 - PT PN MN236-C4 E | Manual do utilizador

Carrinho de transporte PN 20765565

Sprint / Stroller / Hi Flow Stroller
Segurança
AVISO: OS EQUIPAMENTOS DE COMUNICAÇÃO POR RF PORTÁTEIS (INCLUINDO OS PERIFÉRICOSCOMO CABOS DE ANTENA E ANTENAS EXTERNAS) DEVEM SER USADOS A PELO MENOS 30 CM (12POLEGADAS) DE QUALQUER PEÇA DO SPRINT/STROLLER/HI FLOW STROLLER, INCLUINDO OS CABOSESPECIFICADOS PELO FABRICANTE. CASO CONTRÁRIO, O DESEMPENHO DESTE EQUIPAMENTOPODERÁ DEGRADAR-SE.
AVISO: A UTILIZAÇÃO DE ACESSÓRIOS, TRANSDUTORES E CABOS PARA ALÉM DOS ESPECIFICADOS OU
FORNECIDOS PELO FABRICANTE DESTE EQUIPAMENTO PODE RESULTAR EM EMISSÕES ELETROMAGNÉTICASACRESCIDAS OU NUMA REDUÇÃO DA IMUNIDADE ELETROMAGNÉTICA DO EQUIPAMENTO E PROVOCAR UM
FUNCIONAMENTO INDEVIDO.
AVISO: DEVE EVITAR-SE A UTILIZAÇÃO DESTE EQUIPAMENTO DE FORMA ADJACENTE OU EMPILHADO COM
OUTROS DISPOSITIVOS POIS TAL PODE RESULTAR NUM FUNCIONAMENTO INADEQUADO. SE TAL UTILIZAÇÃO
FOR NECESSÁRIA, ESTE E OS OUTROS EQUIPAMENTOS DEVEM SER VERIFICADOS A FIM DE GARANTIR O SEUNORMAL FUNCIONAMENTO.
Atenção: Os equipamentos médicos elétricos requerem precauções especiais no que respeita à
Compatibilidade eletromagnética (EMC) e têm de ser instalados e colocados em serviço de acordo com as
informações de EMC fornecidas neste manual.
Atenção: Os equipamentos de comunicação por radiofrequência (RF) portáteis ou móveis podem afetar os
equipamentos elétricos médicos.
Quadro 1

Orientações e Declaração do Fabricante­—emissões eletromagnéticas
O Sprint/Stroller/Hi Flow Stroller destina-se a ser usado no ambiente eletromagnético especificado abaixo. O
cliente ou utilizador do Sprint/Stroller/Hi Flow Stroller deve certificar-se de que este é usado no ambiente descrito.

Teste de Emissões
Emissões RF

Conformidade
Grupo 1CISPR 11

Emissões de
RF CISPR 11

Ambiente eletromagnético—orientaçõesO Sprint/Stroller/Hi Flow Stroller utiliza energia de RF
apenas para o seu funcionamento interno.
Por conseguinte, as suas emissões de RF são muito
baixas e não é provável que causem interferências nos
equipamentos eletrónicos próximos.

Classe B

Emissões harmónicas
IEC 61000-3-2

Não aplicável

Flutuações detensão/emissões
inconstantes
IEC 61000-3-3

Não aplicável

A utilização do Sprint/Stroller/Hi Flow Stroller éadequada em todos os estabelecimentos, incluindo os
estabelecimentos domésticos e os que se encontram
ligados à rede pública de fornecimento de energia debaixa voltagem que abastece os edifícios usados para finsdomésticos.

PN MN236-C4 E | Manual do utilizador PT - 83

Sprint / Stroller / Hi Flow Stroller
Quadro 2*: Distâncias de separação recomendadas entre os equipamentos de comunicação de RF portáteis emóveis e o Sprint/Stroller/Hi Flow Stroller
O Sprint/Stroller/Hi Flow Stroller destina-se a ser utilizado num ambiente eletromagnético no qual as perturbações de RF irradiadas
sejam controladas. O cliente ou o utilizador do Sprint/Stroller/Hi Flow Stroller pode ajudar a evitar interferências eletromagnéticas
mantendo uma distância mínima entre os equipamentos de comunicação por RF portáteis e móveis (transmissores) e o Sprint/
Stroller/Hi Flow Stroller, tal como se recomenda em baixo, de acordo com a potência máxima de saída do equipamento decomunicações.
Potência nominal
Distância de separação de acordo com a frequência do transmissor
máxima de saída domtransmissor
80 MHz e 800 MHz
800 MHz a 2,5 GHz
150 kHz a 80 MHzW

d = 1,2√P

d = 1,2√P

d = 2,3√P

0,01
0,12 m0,12 m0,23 m0,10,38 m0,38 m0,73 m
1
1,2 m
1,2 m
2,3 m
10
3,8 m
3,8 m
7,3 m
100
12 m
12 m
23 m
Para transmissores com uma potência nominal de saída máxima não indicada acima, a distância de separação recomendada(d) em metros (m) pode ser estimada utilizando a equação aplicável à frequência do transmissor, em que P é a potência nominal
máxima de saída do transmissor em Watts (W) segundo o fabricante do transmissor.
NOTA 1 A 80 MHz e 800 MHz, aplica-se a distância de separação para o intervalo de frequência superior.
NOTA 2 Estas diretrizes podem não se aplicar em todas as situações. A propagação eletromagnética é afetada pela absorção ereflexão de estruturas, objetos e pessoas.

* Este quadro é incluído como requisito padrão para equipamentos testados segundo níveis de teste e amplitudes
de frequência específicos e considerados em conformidade com os regulamentos.

Quadro 3: Orientações e Declaração dos Fabricantes—imunidade eletromagnéticaO Sprint/Stroller/Hi Flow Stroller destina-se a ser usado no ambiente eletromagnético especificado abaixo. O cliente ou utilizador do
Sprint/Stroller/Hi Flow Stroller deve certificar-se de que este é usado no ambiente descrito.
Teste de imunidade
Nível de teste IEC 60601
Nível de conformidade Ambiente eletromagnético—orientações
O piso deve ser de madeira, cimento ou
Contacto ± 8 kV
Descarga eletrostática Contacto ± 8 kVazulejo. Se o piso estiver coberto de material(ESD)± 2 kV, ± 4 kV
± 2 kV, ± 4 kV
sintético, a humidade relativa deve ser de 30%,
IEC 61000-4-2
Ar ± 8 kV, ± 15 kV
Ar ± 8 kV, ± 15 kV
no mínimo.**Não aplicável±2 kV para fornecimento
Dispositivo alimentado
Elétrico rápido
energético linhas de forpor CC
necimento
Não aplicáveltransiente
Não aplicável±1 kV para linhas de
IEC 610004-4
Semlinhasdeentrada/entrada/saída
saída de dados
Sobretensão
±1 kV linha(s) para linha(s) Não aplicável
Não aplicável
dispositivo alimentado±2 kV linha(s) para terra
IEC 61000-4-5
por CC<5% UT (>95% de queda
Quedas de tensão,em UT) para 0,5 ciclos
interrupções curtas e
40% UT (60% de queda em
variações de tensão
Não aplicável
nas linhas de entrada UT) para 5 ciclos
Não aplicável
alimentado
70% UT (30% de queda em dispositivo
de fornecimento de
por CC
UT)
para
25
ciclosenergia<5% UT (>95% de queda
IEC 61000-4-11
em UT) para 5 segundos
Campo magnético deOs campos magnéticos de frequência defrequência de energia 3 A / m
3A/menergia devem estar a um nível característico(50/60 Hz)
de um local típico, num ambiente comercial ou
50/60 Hz50/60 Hzhospitalar típico.
IEC 61000-4-8
Nota: UT é a tensão de CA antes da aplicação do nível de teste.

** Esta declaração indica que os testes requeridos foram realizados num ambiente controlado e que o Sprint/
Stroller/Hi Flow Stroller foi considerado em conformidade com os regulamentos.

84 - PT PN MN236-C4 E | Manual do utilizador

Sprint / Stroller / Hi Flow Stroller
Quadro 4

Orientações e Declaração do Fabricante—Imunidade Equipamentos ME e Sistemas ME

Orientações e Declaração do Fabricante—Imunidade
O Sprint/Stroller/Hi Flow Stroller destina-se a ser usado no ambiente eletromagnético especificado abaixo. O cliente
ou utilizador do Sprint/Stroller/Hi Flow Stroller deve certificar-se de que este é usado no ambiente descrito.
Teste de imunidade Nível de teste IEC 60601
Nível de conformidade
Ambiente eletromagnético—orientações
RF conduzidas
IEC 61000-4-6

3 Vrms6 Vrms (Em bandas ISM)
150 kHz a 80 MHz

Não aplicável

Dispositivo alimentado porbateria, No SIP/SOP

Os equipamentos portáteis e móveis decomunicações por RF não devem ser utilizados
mais perto de qualquer peça do Sprint/Stroller/Hi
Flow Stroller, incluindo cabos, do que a distância
de separação recomendada calculada a partir daequação aplicável à frequência do transmissor.
Distância de separação recomendada
d = 1,2 √P

RF radiadas IEC 610004-3

80 MHz a 2,7 GHz

d = 1,2 √P
d = 2,3 √P
onde P é a potência nominal máxima de saída
do transmissor em Watts (W), de acordo com ofabricante do transmissor, e d é a distância deseparação recomendada em metros (m).

10 V/m80 MHz—2,7 GHz
80% AM a 1 kHz

As forças de campo provenientes detransmissores de RF fixos, conforme determinado
por um levantamento eletromagnético do locala,devem ser inferiores ao nível de conformidade em
cada intervalo de frequênciab.
Poderão ocorrer interferências nas imediações
de equipamentos assinalados com o seguinte
símbolo:Frequência deteste (MHz)Bandaa)(MHz)

Serviçoa)Modulaçãob)385

380-390TETRA 400450

430-470

GMRS 460, FRS 460

Modulação do pulso
18 Hz
FMC)
desvio ±5 kHz
seno 1 kHz

704-787

Banda LTE 13, 17

800-960

GSM 800/900, TETRA 800,
IDEN 820, CDMA 850, LTE
Band 5

710
745
780
810
870
930
1720
1845
1970
2450
5240
5500
5785

Potênciamáxima (W)Distância(m)

Nível do teste deimunidade (V/m)

1,8

0,3

27

2

0,3

28

Modulação do pulsob)
217 Hz

0,2

0,3

9

Modulação do pulsob)
18 Hz

2

0,3

28

2

0,3

28

2

0,3

28

0,2

0,3

9

b)

GSM 1800; CDMA 1900;b)
1700-1900 GSM 1900; DECT; LTE Band Modulação do pulso
217 Hz1, 3, 4, 25; UMTS
Bluetooth, WLAN, 802.11
Modulação do pulsob)
2400-2570
b/g/n, RFID 2450, LTE
217 HzBand 75100-5800

WLAN 802.11 a/n

Modulação do pulsob)
217 Hz

Nota: Se tal for necessário para atingir o NÍVEL DE TESTE DE IMUNIDADE, a distância entre a antena transmissora e o Sprint/
Stroller/Hi Flow Stroller pode ser reduzida a 1 m. A distância de teste de 1 m é permitida pela norma IEC 61000-4-3.
Para alguns serviços, estão incluídas apenas as frequências de uplink.
O portador deve ser modulado usando um sinal de onda quadrada de ciclo de trabalho de 50%.
Como alternativa à modulação FM, pode ser usada uma modulação de pulso de 50% a 18 Hz uma vez que, embora não represente a
modulação real, esse seria o pior dos cenários.
abc

CAIRE e CAIRE Inc. são marcas registradas da CAIRE Inc. Visite nosso site abaixo para obter uma lista completa de marcas comerciais.Marcas comerciais: www.careinc.com/corporate/
trademarks.
Copyright © 2023 CAIRE Inc. A CAIRE Inc. reserva-se o direito de descontinuar os seus produtos ou alterar os preços, materiais, equipamento, qualidade, descrições, especificações e/ou
processos dos seus produtos a qualquer momento e sem aviso prévio, sem quaisquer obrigações ou consequências. Reservamos todos os direitos não expressamente aqui declarados,
conforme aplicável.

PN MN236-C4 E | Manual do utilizador PT - 85

Sprint / Stroller / Hi Flow Stroller
Overzicht van symbolen
ISO 7000

Rook niet in de buurt van het apparaatof tijdens de bediening ervan.
Reg.-nr. P002

Droog en uit de buurt van regenbewaren. Reg.-nr. 0626

Toegepast onderdeel van type BF (matevan bescherming tegen elektrischeschokken). Reg.-nr. 5333

Stapellimiet op nummer. Reg. # 2403
Naam en adres van de fabrikant.
Reg.-nr. 3082

Waarschuwing. Reg.-nr. W001
Richtlijn 93/42/EEG van de Raad
Gevolmachtigde vertegenwoordiger inde Europese Gemeenschap

Het land en de fabricagedatum. De"CC" identificeert de tweeletterige
landcode van het land van fabricage. De
fabricagedatum is in het formaat JJJJMM-DD. Reg. # 6049
Opgelet: raadpleeg de bijgeleverde
documenten. Reg.-nr. 0434A

Als op het etiket van het product uniquedevice identifier (UDI) het CE ####
symbool staat, voldoet het apparaat aande vereisten van Richtlijn 93/42 / EEG
betreffende medische hulpmiddelen. Het
CE #### symbool geeft het nummer van
de aangemelde instantie aan.

Catalogusnummer. Reg.-nr. 2493

Dit apparaat voldoet aan de eisen vanrichtlijn 2010/35/EU inzake medische
instrumenten. Het is voorzien van het
getoonde pi-keurmerk.

Serienummer. Reg.-nr. 2498
Opslag- of bedrijfstemperatuurbereik.
Reg.-nr. 0632
Luchtvochtigheid bij opslag. Reg.-nr.
2620
Luchtdrukbereik. Reg.-nr. 2621
Deze kant omhoog. Reg.-nr. 0623

ADR
Niet-giftig gas.
Gevaar Oxiderende stoffen:
brandverhogend risico.
Oxidatiemiddelen zorgen ervoor dat
branden heftiger branden.
Gekoelde vloeistof, USP; geproduceerd
door lucht vloeibaar te maken
Extra symbolen

Breekbaar, voorzichtig. Reg.-nr. 0621

Zorg dat het apparaat altijd goed isgeventileerd

Bevat gevaarlijke stoffen. Reg. # 3723
Uit de buurt bewaren van ontvlambare
materialen, olie en vet.

Importeur. Reg. # 3725

Geautoriseerde vertegenwoordiger inZwitserland.

ISO 7010
Bij contact met koude vloeistoffen,
gasvormig zuurstof of bevroren
onderdelen kan bevriezing optreden.
Waarschuwing voor lage temperatuur.
Ter waarschuwing voor lage
temperaturen of vrieskou. Reg.-nr. W010
Het lezen van de instructiehandleiding
is verplicht. Reg.-nr. M002
Uit de buurt van open vuur,
vlammen en vonken bewaren. Open
ontstekingsbronnen en roken verboden.
Reg.-nr. P003
86 - NL PN MN236-C4 E | Gebruikershandleiding

IEC 60417

Als het apparaat het UKCA-merkteken draagt, zoals weergegeven met
UKCA#### met vermelding van het
nummer van de aangemelde instantie,
voldoet dit apparaat aan de UKCAregelgeving.
Dek het apparaat niet af en draag het
draagbare apparaat niet onder uw
kleding. Deze units geven normaal
gesproken zuurstof af. Nr. 5641

Sprint / Stroller / Hi Flow Stroller
21 CFR 801.15
Dit product mag op grond van defederale wetgeving uitsluitend worden
verkocht door of namens een arts.
IEC 60601-1
Drupt niet
Richtlijn 2012/19/EU van de Raad
WEEE
Dit symbool herinnert de eigenaars vande apparatuur eraan om deze aan het
einde van zijn levensduur in te leveren
bij een recyclingbedrijf, volgens de richtlijn betreffende afgedankte elektrische
en elektronische apparatuur (AEEA).
Onze producten zullen voldoen aan debeperking van de richtlijn gevaarlijke
stoffen (RoHS). Ze bevatten niet meer
dan sporen van lood of andere gevaarlijke stoffen.

ISO 15223-1
Medisch apparaat. Reg. # 5.7.7
Unieke apparaat-ID # 5.7.10
Op dit product kunnen een of meer Amerikaanse eninternationale octrooien van toepassing zijn. Bezoek
onze website voor een lijst met toepasselijke octrooien.
Pat.: www.caireinc.com/corporate/patents/.

Specificaties• Bedrijfsmodus: continu debiet• Type bescherming tegen elektrische schok: apparatuur met interne voeding• Mate van bescherming tegen elektrische schok: toegepast onderdeel van het type BF• Classificatie overeenkomstig de mate van bescherming tegen binnendringend water:
IP22 - Standaardapparatuur• Het apparaat is niet geschikt voor gebruik in de nabijheid van ontvlambare mengsels
Productspecificaties

SprintStroller

Hi Flow Stroller

Capaciteit voor vloeibare
zuurstof
Capaciteit voor gasvormig
equivalent
Gewicht, leeg

0,68 kg (1,5 lb)

1,36 kg (3,0 lb)

1,36 kg (3,0 lb)

513 l

1026 l1025 l

2,04 kg (4,5 lb)

2,27 kg (5,0 lb)

2,49 kg (5,5 lb)

Gewicht, gevuld

2,72 kg (6,0 lb)

3,63 kg (8,0 lb)

3,86 kg (8,5 lb)

Hoogte298 mm (11,75 inch)

343 mm (13,5 inch)

343 mm (13,5 inch)

Afmetingen

123,83 mm (4,875") D x
191 mm (7,5") B
4,3 uur

149,23 mm (5,875") D x
191 mm (7,5") B
8,5 uur

149 mm (5,875") D x
191 mm (7,5") B
8,5 uur

137 kPa (20 psi)

137 kPa (20 psi)

137 kPa (20 psi)

0,57 kg/dag (1,3 lb/dag)

0,57 kg/dag (1,3 lb/dag)

0,57 kg/dag (1,3 lb/dag)

Uit; 0,25; 0,5; 0,75; 1; 1,5; 2;
2,5; 3; 4; 5; 6 L/MIN.
+/- 10%

Uit; 0,5; 1; 1,5; 2; 2,5; 3; 4; 5; Uit; 0,5; 1; 2; 2,5; 3; 4; 6; 8;
6; 8; 10; 15 L/MIN.
10; 12; 15 L/MIN.
+/- 10%+/- 10%Gangbare gebruikstijd bij
2 liter per minuut
Operationele druk
Normaleverdampingssnelheid
Standaardbereik voor
debietregeling
Nauwkeurigheid van het
debiet** Deze nauwkeurigheid is alleen van toepassing bij 21,1 °C (70 °F) en 101,35 kPa (14,7 psig) en bij een correct gekalibreerde debietmeter.

PN MN236-C4 E | Gebruikershandleiding NL - 87

Sprint / Stroller / Hi Flow Stroller
Waarschuwingsinformatie
Belangrijk: lees deze handleiding grondig voordat u
de Sprint/Stroller/Hi Flow Stroller gebruikt.
Uitsluitend op recept.
WAARSCHUWING: DIT INSTRUMENTHEEFT GEEN LEVENSONDERHOUDENDETOEPASSING.
WAARSCHUWING: NEEM CONTACT OP MET UWZORGVERLENER WANNEER HET APPARAAT VOLGENSU NIET GOED WERKT. PROBEER HET APPARAAT NIETZELF TE REPAREREN OF AAN TE PASSEN.
WAARSCHUWING: WIJZIG DIT APPARAAT NIETZONDER TOESTEMMING VAN DE FABRIKANT.
WAARSCHUWING: ALS EEN CONSTANTEZUURSTOFTOEVOER VEREIST IS, ZORG DAN VOOREEN GESCHIKTE ZUURSTOFVOORZIENING EN/OFZORG ERVOOR DAT TIJDENS DE THERAPIE TE ALLEN
TIJDE EEN SECUNDAIRE ZUURSTOFVOORZIENINGBESCHIKBAAR IS.
WAARSCHUWING: ROKEN, KAARSEN OF OPEN VUURZIJN NIET TOEGESTAAN OP MINDER DAN 3 METER(10 FT.) AFSTAND VAN HET APPARAAT OF OP MINDERDAN 20 CM (8 INCH) VAN EEN ONTSTEKINGSBRON.
WAARSCHUWING: GEBRUIK UW APPARAAT IN EEN
GOED GEVENTILEERDE RUIMTE.
WAARSCHUWING: BEWAAR APPARATUUR
MET VLOEIBARE ZUURSTOF NIET IN EEN
KAST, DE KOFFERBAK VAN EEN AUTOOF EEN ANDERE AFGESLOTEN RUIMTE.
LEG GEEN DEKENS, KLEDEN OF ANDERESTOFFEN OP HET APPARAAT.
WAARSCHUWING: DIT PRODUCT KAN UBLOOTSTELLEN AAN CHEMISCHE STOFFEN ZOALSNIKKEL, WAARVAN DE STAAT CALIFORNIË WEETDAT HET KANKERVERWEKKEND IS. GA VOOR MEER
INFORMATIE NAAR WWW.P65WARNINGS.CA.GOV.
WAARSCHUWING: IN HET GEVAL ER EEN ERNSTIGINCIDENT MET DIT APPARAAT OPTREEDT, MOET DEGEBRUIKER HET INCIDENT ONMIDDELLIJK MELDENAAN DE AANBIEDER EN / OF DE FABRIKANT. EEN
ERNSTIG INCIDENT WORDT GEDEFINIEERD ALS
EEN LETSEL, OVERLIJDEN OF MOGELIJK LETSEL/ OVERLIJDEN VEROORZAKEN ALS HET INCIDENTZICH OPNIEUW VOORDOET. DE GEBRUIKER KANHET INCIDENT OOK MELDEN AAN DE BEVOEGDEAUTORITEIT IN HET LAND WAAR HET INCIDENT
PLAATSVOND.
Let op: gebruik de Sprint/Stroller/Hi Flow
Stroller uitsluitend volgens de voor­schriften
van uw arts.
Let op: dit product mag op grond van de federale
wetgeving uitsluitend worden verkocht door of namens
een arts.
Let op: Het apparaat bevat vloeibare zuurstof. Dezezuurstof is extreem koud, bijna -184 °C (300 °F).
Blootstelling aan een dergelijke lage temperatuur kan
ernstige bevriezing veroorzaken.

88 - NL PN MN236-C4 E | Gebruikershandleiding

Let op: Vloeibare en gasvormige zuurstof zijn vanzichzelf niet ontvlambaar, maar kunnen er wel toe leiden
dat andere materialen sneller ontbranden dan anders.
Dit gevaar vereist, samen met de lage temperatuur vanvloeibare zuurstof, bepaalde voorzorgsmaatregelen.
Let op: Houd ontvlambare materialen uit de buurt vandeze apparatuur. Spuitbussen, olie en vet, waaronder
gezichtscrèmes en vaseline, vatten gemakkelijk vlam en
kunnen snel verbranden in aanwezigheid van zuurstof.
Let op: Roken terwijl u een zuurstofcanule draagt, kan
leiden tot brandwonden in het gezicht of zelfs tot dedood.
Als u de canule afneemt en op kleding, bedlinnen, een
sofa of kussens legt, ontstaat er een steekvlam bij blootstelling aan een sigaret, warmtebron, vonk of vlam.
Houd u aan de volgende regels als u rookt: (1) zet het
draagbare apparaat uit, (2) neem de canule af en (3)verlaat de kamer waar het apparaat zich bevindt.
Plaats het apparaat indien mogelijk direct maarvoorzichtig overeind wanneer het per ongeluk omvalt.
Verlaat de ruimte direct en bel uw zorgverlener wanneer
er vloeibare zuurstof ontsnapt. Probeer het apparaat
niet te verplaatsen of te voorkomen dat de vloeibare
zuurstof ontsnapt.
Opmerking: Raak geen bevroren onderdelen van een
apparaat aan.
Opmerking: Bewaar of bedien het draagbare apparaat niet
terwijl deze aan het stationaire apparaat is gekoppeld.
Opmerking: Voorkom dat dit apparaat wordt gebruikt ofbediend door onbevoegde medewerkers.
Opmerking: De Amerikaanse Federal Aviation Administration verbiedt het gebruik van dit apparaat tijdens commerciële passagiersvluchten en vrachtvluchten.

Beoogd gebruik

De CAIRE Sprint/Stroller/Hi Flow Stroller is bedoeldvoor het toedienen van extra zuurstof. Het apparaat isniet bedoeld voor levensondersteuning en biedt ook
geen mogelijkheden voor patiëntbewaking.

Sprint / Stroller / Hi Flow Stroller
InleidingBedieningselementen

De Sprint, Stroller en Hi Flow Stroller zijnbedoeld voor de toediening van aanvullende
zuurstof aan de patiënt in de woning van deeindgebruiker. Ze kunnen tevens worden gebruikt ininstellingen, zoals verzorgingstehuizen of subacuteverzorgingsinstellingen. De Sprint/Stroller/Hi Flow
Stroller is een ambulante zuurstofbron die u gedurende
een langere periode kunt gebruiken.

1. Elektronische vloeistofniveaumeter
2. Elektronische vloeistofniveauschakelaar
3. Mechanische vloeistofniveaumeter
4. Debietregelknop
5. Zuurstofslangaansluiting
6. Ontluchtingsklep
7. Vulaansluitingen (zie volgende pagina voor
identificatie snelloskleppen)
8. Draagriem met schouderkussen
9. Condenskussen
10. Condensbakje

Het apparaat wordt gebruikt bij COPD-patiënten ofpatiënten met een verminderde ademhalingsfunctie.
Het apparaat wordt aan de patiënt voorgeschreven.
Het apparaat wordt verkocht aan een zorgverlener die
is opgeleid om de Sprint/Stroller/Hi Flow Stroller tegebruiken en te onderhouden. De zorgverlener traint
de gebruiker.

1
2

Het apparaat is niet bedoeld ter ondersteuning of
instandhouding van de levensfuncties en biedt ook
geen voorzieningen voor het bewaken van patiënten.
Het is aangewezen over een alternatieve bron van
aanvullende zuurstof te beschikken in geval van
een mechanische storing. Deze draagbare apparaten
beschikken tevens over een snellosklep die aan een
apparaat met een vloeibaar zuurstofreservoir gekoppeld
kan worden om het draagbare systeem te vullen.
De reservoirunit wordt gevuld door uw zorgverlener.

4
58Deze gebruikershandleiding bevat de instructies voor
het gebruik van de Sprint/Stroller/Hi Flow Stroller.
Raadpleeg de gebruikershandleiding die bij het
stationaire apparaat is geleverd voor informatie overde werking ervan.
Opmerking: de dienstverlener zal u helpen met het instellen van het apparaat en zal u instructies geven voor een
juiste omgang met en een juist gebruik van het apparaat.

5

7

6

Sprint-, Stroller- en Hi Flow Stroller-apparaten

10
9

PN MN236-C4 E | Gebruikershandleiding NL - 89

Sprint / Stroller / Hi Flow Stroller
Identificatie snelloskleppen
Aansluiting
draagbaar
systeemAansluiting
stationair
systeem

3. Plaats de vulaansluiting rechtop over de aansluiting
op het stationaire apparaat.

Type snellosklep

CAIREZijvulverzijvullgrendelinging

LIFEOX
Bovenvulvergrendeling

4. a. Als uw aansluiting met drukknop is:
Laat de Sprint/Stroller/Hi Flow Stroller op het
stationaire apparaat zakken, totdat u voelt dat deaansluiting vergrendelt.

PENOX

TAEMA
Bovenvullingmet drukknop
PB

b. Als de aansluiting van Penox of Life-Ox is:
Draai de Sprint/Stroller/Hi Flow Stroller totdat
u voelt dat de aansluiting vergrendelt. Draaide Sprint/Stroller/Hi Flow Stroller vervolgens
voorzichtig maar krachtig rechtsom (), terwijl
u die naar beneden drukt.

Gebruiksaanwijzing
Sprint/Stroller/Hi Flow Stroller met
bovenvulling vullen

Opmerking: Vervolg in stap 6 van 'Sprint/Stroller/Hi Flow
Stroller met zijvulling vullen'.

1. Maak de vulaansluitingen van het stationaire
apparaat en de Sprint/Stroller/Hi Flow Strollerschoon met een schone, droge en pluisvrije doek.
Let op: de vulaansluitingen moeten droog
zijn, want uw Sprint/Stroller/Hi Flow Stroller
kan door vocht vastvriezen aan het stationaire
apparaat.

Sprint/Stroller/Hi Flow Stroller met zijvullingvullen2. Zorg dat de debietregelknoppen van de Sprint/
Stroller/Hi Flow Stroller en het reservoir in de uitstand (0) staan.

90 - NL PN MN236-C4 E | Gebruikershandleiding

1. Maak de vulaansluitingen van het stationaire
apparaat en de Sprint/Stroller/Hi Flow Strollerschoon met een schone, droge en pluisvrije doek.
WAARSCHUWING: DE VULAANSLUITINGEN
MOETEN DROOG ZIJN, WANT UW SPRINT/
STROLLER/HI FLOW STROLLER KAN
DOOR VOCHT VASTVRIEZEN AAN HETSTATIONAIRE APPARAAT.

Sprint / Stroller / Hi Flow Stroller
2. Zorg dat de debietregelknoppen van de Sprint/
Stroller/Hi Flow Stroller en het reservoir in de uitstand (0) staan.
3. Plaats de vulaansluiting van de Sprint/Stroller/Hi
Flow Stroller rechtop over de aansluiting van het
stationaire apparaat.

4. Draai de Sprint/Stroller/Hi Flow Stroller linksom(), totdat u voelt dat de pin zich vastklikt in degleuf.

Opmerking: de vloeistofniveaumeter geeft het meest
accurate resultaat 30 seconden of meer nadat de Sprint/
Stroller/Hi Flow Stroller werd gevuld.

7. U ziet mogelijk ook wat damp rond deaansluiting. Dit is normaal en komt door het grote
temperatuurverschil tussen de vloeibare zuurstof en
de warme lucht rondom de apparaten.
8. Als het apparaat vol is, sluit u de ontluchtingsklep.
U merkt dat het apparaat vol is wanneer het sissende
geluid van toon verandert en er wat vloeistof uit deontluchtingsleiding komt.
Opmerking: als u de klep sluit nadat er een vrijwel continue
stroom vloeistof wordt afgevoerd, dan wordt de Sprint/
Stroller/Hi Flow Stroller tot het maximale gevuld, maar zultu wat vloeibare zuurstof verspillen.

5. Draai de Sprint/Stroller/Hi Flow Stroller voorzichtig
maar stevig terug naar een rechtstaande positie. Nuzijn de apparaten met elkaar vergrendeld.

Opmerking: als u de klep sluit bij de eerste tekenen vanvloeistofafvoer, wordt de Sprint/Stroller/Hi Flow Stroller niet
helemaal gevuld, maar bespaart u wat vloeibare zuurstof.
9. Voor de aansluitingen met drukknop, duwt u deontgrendelingsknop in totdat de apparaten vanelkaar loskomen. Voor alle overige aansluitingendraait u het apparaat totdat dit loskomt van het
stationaire apparaat.

Opmerking: vervolg in stap 6.
Opmerking: zorg dat u bij de aansluiting met drukknop het
apparaat tijdens het vullen op zijn plek houdt.

Ontgrendelingsknop draagbare systeem(alleen voor aansluitingen met drukknop)

6. Open de ontluchtingsklep en begin met het vullen
van uw Sprint/Stroller/Hi Flow Stroller. U zult
wat zuurstof horen ontsnappen; dat is niet erg. Het
sissende geluid dat u hoort is volkomen normaal.

PN MN236-C4 E | Gebruikershandleiding NL - 91

Sprint / Stroller / Hi Flow Stroller
Let op: controleer de vloeistofniveaumeter pas
nadat de klep gesloten is.

Gebruiksaanwijzing

Let op: als de Sprint/Stroller/Hi Flow Stroller
onmiddellijk opnieuw wordt gevuld na een periode
van gebruik, moet u na het vullen ervan 10 minuten
wachten, met de debietregelaar op nul, totdat de drukgestabiliseerd is. Gebruik het systeem daarna weer
zoals gewoonlijk.
WAARSCHUWING: GEBRUIK DE SPRINT/
STROLLER/HI FLOW STROLLER NIETTERWIJL DIE AAN HET STATIONAIRE
APPARAAT GEKOPPELD IS.
WAARSCHUWING: ALS DE SPRINT/STROLLER/HIFLOW STROLLER NIET GEMAKKELIJK LOSRAAKT,GEBRUIK DAN GEEN GEWELD. DE APPARATENZIJN WELLICHT AAN ELKAAR GEVROREN. LAAT DEAPPARATEN AAN ELKAAR ZITTEN EN WACHT TOT ZEZIJN OPGEWARMD: DAN KUNNEN ZE GEMAKKELIJKWORDEN GESCHEIDEN. RAAK GEEN BEVRORENONDERDELEN AAN. WAARSCHUWING: VERLAATDE RUIMTE EN BEL DIRECT UW ZORGVERLENERWANNEER ER SPRAKE IS VAN EEN ZODANIG LEK DATER VLOEISTOF UIT HET APPARAAT STROOMT.
WAARSCHUWING: VLOEIBARE OF KOUDEGASVORMIGE ZUURSTOF KUNNEN DE HUID
BEVRIEZEN. ZUURSTOF VERHOOGT HET
BRANDGEVAAR.
Let op: lekt er na het scheiden van deapparaten vloeistof uit het stationaire apparaat
of de Sprint/Stroller/Hi Flow Stroller, zet de
Sprint/Stroller/Hi Flow Stroller dan in verticale
positie aan de kant, verlaat de ruimte en bel
onmiddellijk uw zorgverlener.
Opmerking: aan leverancier van apparatuur: De volgende
accessoires voor zuurstoftoediening worden aanbevolen
voor gebruik met de Sprint / Stroller / Hi Flow Stroller:(Canule is niet inbegrepen.)

1. Duw de ademhalingscanule stevig op de aansluitingvoor de zuurstofslang.
2. Breng de ademhalingscanule in de juiste positie om
de zuurstof comfortabel te kunnen inademen.
3. Draai de debietregelknop aan de bovenzijde van de
Sprint/Stroller/Hi Flow Stroller rechtsom totdat het
voorgeschreven debiet (numeriek) zichtbaar is in
het 'venster' van de knop en u een klik voelt.
Let op: als de debietregelknop tussen debieten
is ingesteld, zal het zuurstofdebiet buiten despecificaties vallen.
4. Als het goed is, ontvangt u nu zuurstof. Om
de zuurstoftoediening te stoppen, draait u dedebietregelknop linksom naar de uit-stand (0).
Let op: draai de debietregelknop altijd naar '0'
wanneer u het systeem niet gebruikt.

• Neuscanule met 2,1 m (7 voet) slang (maximale LPM: 6
LPM): CAIRE-onderdeelnummer 5408-SEQ• Firebreak: CAIRE-onderdeelnummer 21126636
Een brandgang wordt aanbevolen voor gebruik met elke
canule.
• CAIRE biedt een brandgang aan die bedoeld is om teworden gebruikt in combinatie met de draagbare zuurstof.
De brandgang is een thermische zekering om de gasstroom te stoppen in het geval dat de stroomafwaartse
canule of zuurstofslang wordt ontstoken en brandt aan
de brandgang. Het wordt in lijn met de neuscanule ofzuurstofslang tussen de patiënt en de zuurstofuitlaat vande Sprint / Stroller / Hi Flow Stroller geplaatst. Raadpleeg
voor een juist gebruik van de brandgang altijd de instructies van de fabrikant (meegeleverd bij elke brandgangset).
• Aanvullende informatie over aanbevolen accessoires isonline beschikbaar op www.caireinc.com.

92 - NL PN MN236-C4 E | Gebruikershandleiding

5. Gebruik de onderstaande tabel als een richtlijn omvast te stellen hoelang de Sprint/Stroller/Hi Flow
Stroller zal blijven werken:

Sprint / Stroller / Hi Flow Stroller
Debietregelaarinstelling
Model

SprintStroller

Hi Flow
Stroller

Nominaal

Nominaal

Pos.
debietregelaar

SprintStroller

Hi Flow
Stroller

Uit

30:00

60:00

71:00

0,25

18:00

30:00

n.v.t.
32:00

0,5

12:12

23:00

0,75

9:18

17:24

n.v.t.

1

7:30

14:06

16:00

1,5

5:30

10:12

n.v.t.

2

4:18

8:00

8:00

2,5

3:36

6:36

6:20

3

3:06

5:36

5:20

4

2:24

4:24

4:00

5

2:00

3:36

n.v.t.

6

1:42

3:00

2:40

8

N.v.t.

N.v.t.

2:00

10

N.v.t.

N.v.t.

1:40

12

N.v.t.

N.v.t.

1:20

15

N.v.t.

N.v.t.

1:00

Opmerking: tijden worden in uren en minuten
weergegeven (notatie 00:00).
Opmerking: de 'nominale' tijden gelden in ideale
omstandigheden, d.w.z. maximale vulling, exacte
debieten, een goed verliespercentage, geen
verplaatsing van het apparaat, enz.
Deze tijden zijn de maximaal verwachte tijden.
Opmerking: de individuele resultaten zullen variëren.
6a. Het vloeibaar zuurstofpeil in het apparaat
controleren met behulp van de elektronischevloeistofniveaumeter:• Gen 3-meter: houd de drukknop (vloeistofniveauschakelaar) bovenaan het apparaat ten minste
twee seconden lang ingedrukt. Lees de lichtbalk
bovenaan af om het vloeistofpeil te bepalen.

• Gen 4-meter: druk de groene bedieningsknop op demeter in. Lees de ledboog af, die het vloeistofpeil
aangeeft.

Let op: de Sprint/Stroller/Hi Flow Stroller isleeg als alleen het laatste deel van de lichtbalk
brandt.
Als de indicator voor een lege batterij brandt
wanneer de knop wordt ingedrukt, meld dit dan aan
uw zorgverlener.
Opmerking: zelfs als de indicator voor een lege batterij
brandt, blijft de Sprint/Stroller/Hi Flow Stroller zuurstof
leveren zolang het apparaat vloeibare zuurstof bevat.
6b. Het vloeibare zuurstofpeil in het apparaat
controleren met behulp van de mechanischevloeistofpeilmeter:• Houd het apparaat vast aan de stoffen band, direct
boven de kamer met de zwarte weeghaak met het
vloeistofpeil.
• Laat het apparaat stabiliseren (dus niet op en neer
stuiteren) en lees af hoeveel vloeibare zuurstof er
volgens de gekleurde schaalverdeling ongeveer nog
in zit.

PN MN236-C4 E | Gebruikershandleiding NL - 93

Sprint / Stroller / Hi Flow Stroller
Opmerking: als het apparaat leeg is, is alleen de rood
gekleurde band van de schaal zichtbaar. In dat geval moet
het apparaat voor gebruik eerst gevuld worden.
7. Bij bepaalde omgevingsomstandigheden, vooral
bij een continu gebruik en weinig beweging van
het draagbaar apparaat, kan deze overtollig ijs
rond de spoel van de verstuiver en op de behuizing
opbouwen. U kunt deze ijsvorming verminderen
door tegen het apparaat te tikken en/of deijsvorming van de behuizing te vegen.

Onderhoud van condenskussen
Als het apparaat leeg is en tot kamertemperatuur is
opgewarmd, kunt u het vocht uit het condensbakje
verwijderen:
Bij een harde behuizing• Kantel het apparaat naar een kant. Pak de knop
in het midden van het condensbakje (onderaan
het apparaat) vast en trek hem recht omhoog. Zo
komt het bakje los.
Opmerking: als het kussen zeer vuil is, kan het in dewasmachine gewassen worden (met bleekmiddel om teontsmetten).
• Om het bakje terug te plaatsen, plaatst u het in
het midden over de ronde opening onderaan het
apparaat en drukt u eenmaal op de knop om het
bakje vast te klikken en dan nogmaals op deknop om het bakje te vergrendelen.

Bij een zachte behuizing• Maak de rits aan de onderzijde open en trek deklep onderaan achteruit.
• Haal het kussen uit het bakje en wring alle
opgenomen vocht eruit. Laat het kussen volledig
drogen, voordat u het terugplaatst in het bakje.
Opmerking: als het kussen zeer vuil is, kan het in dewasmachine gewassen worden (met bleekmiddel om teontsmetten).
• Om het kussen terug te plaatsen, steekt u het
gewoon in de klep onderaan en doet u de rits
onderaan terug dicht.
Opmerking: als u de knop niet kunt indrukken en het bakje
niet wil vastklikken, houdt u het bakje vast en trekt u aan
de knop totdat deze in de open stand klikt. Plaats het bakje
nu terug volgens de bovenstaande instructies.

Beperking van ijs
Bij bepaalde omgevingsomstandigheden, vooral bij een
continu gebruik en weinig beweging van het draagbaar
apparaat, kan deze overtollig ijs rond de spoel van deverstuiver en op de behuizing opbouwen. U kunt deze
ijsvorming verminderen door tegen het apparaat tetikken en/of de ijsvorming van de behuizing te vegen.
WAARSCHUWING: AANGEZIEN HET RISICOBESTAAT OP OVERTOLLIGE VOCHT-/IJSOPHOPING, IS HET AANBEVOLEN
DE HI FLOW STROLLER NA GEBRUIK 1
UUR LANG TE LATEN RUSTEN OM DE
VOCHTOPHOPING TE LATEN AFNEMEN.

Vergrendeld
Vastgeklikt
Open

Knop bakje

Opmerking: als u de knop niet kunt indrukken en het bakje
niet wil vastklikken, houdt u het bakje vast en trekt u aan
de knop totdat deze in de open stand klikt. Plaats het bakje
nu terug volgens de bovenstaande instructies.

94 - NL PN MN236-C4 E | Gebruikershandleiding

Gezien de hogere debieten van de Hi Flow Stroller,
treden er condens en ijsvorming op het apparaat op,
in het bijzonder bij debietinstellingen van 10 l/min.
en meer. Het is aanbevolen om het apparaat minstens
1 uur te laten rusten na gebruik om het risico op
overtollige ophoping wegens een continu gebruik tebeperken.

Sprint / Stroller / Hi Flow Stroller
Probleemoplossing
De volgende informatie is bedoeld om u te helpen bij het oplossen van eenvoudige problemen die u kunt ervaren
wanneer u uw Sprint/Stroller/Hi Flow Stroller gebruikt.
Probleem
OplossingHet draagbare systeem• Het sissen kan ontstaan om de juiste operationele druk binnen het draagbare systeem
maakt een sissend geluid.
te handhaven. De kans op sissen is het grootst na het vullen of wanneer de positie van
het draagbare systeem wordt gewijzigd. Het sissen kan na het vullen nog ongeveer 10
minuten duren. Ook kan het optreden wanneer de debietklep op een lage instelling staat.

De toevoer van het
draagbare systeem valt stiltijdens gebruik.

Het draagbare apparaat
wordt niet gevuld.

• Als het draagbare systeem niet in de juiste positie is neergelegd, zet het dan weer
rechtop en wacht een aantal minuten totdat het apparaat gestabiliseerd is.
• Controleer of de canule stevig aan de zuurstofuitlaat is bevestigd.
• Controleer of de canule niet is geknikt.
• Controleer het vloeistofpeil/de niveaumeter en vul het draagbare systeem indien nodig bij.
• Controleer of de debietregelknop niet in de uit-stand ('0') staat.
• Controleer of er zuurstof in het reservoir zit.

• Controleer of de aansluitingen van het draagbare apparaat en het reservoir tijdens het
vulproces volledig vergrendeld zijn.
De ontluchtingsklep van het • Wanneer de ontluchtingsklep niet goed sluit en het sissende geluid en de wolk met
zuurstofdamp aanhoudt, dient u het draagbare apparaat voorzichtig te verwijderen door
draagbare apparaat sluit niet
de ontgrendelingsknop op het reservoir in te drukken. De ontluchting aan de onderkant
goed aan het einde van het
van het draagbare apparaat stopt binnen een paar minuten. Laat het apparaat opwarmen
vulproces.
tot u de ontluchtingsklep kunt sluiten. Het kan tot wel 60 minuten duren voordat in het
draagbare systeem de juiste druk is hersteld voor de zuurstofstroom. U kunt desgewenst
een andere zuurstofbron gebruiken, zoals een debietklep die aan het reservoir isbevestigd.
• Wellicht zijn de vulaansluitingen van het draagbare apparaat en reservoir bevroren.
Het draagbare apparaat
kan na het vullen niet
• GEBRUIK GEEN GEWELD. Laat de bevroren onderdelen een paar minuten opwarmen
gemakkelijk worden
en maak het draagbare apparaat los wanneer het ijs is gesmolten. Maak de mannelijke
losgemaakt van het
vulaansluiting op het reservoir en de vrouwelijke vulaansluiting op het draagbare apparaat
reservoir.
altijd schoon met een schone, droge doek voordat u het apparaat vult: zo voorkomt u dat
de apparaten aan elkaar vastvriezen.

PN MN236-C4 E | Gebruikershandleiding NL - 95

Sprint / Stroller / Hi Flow Stroller
Schoonmaken
WAARSCHUWING: REINIG HET APPARAATALLEEN WANNEER HET LEEG IS.
• Reinig het met mild afwasmiddel opgelost inwater.
• Breng het schoonmaakmiddel direct aan
op een pluisvrije doek. Goedgekeurdeschoonmaakmiddelen zijn HydroPure
en HydroKlean. Spuit nooit rechtstreeks
schoonmaakmiddelen op de Sprint/Stroller/Hi
Flow Stroller.
• Maak de buitenkant schoon met een pluisvrije
doek tot er geen vuil meer zichtbaar is.
Let op: gebruik geen hogedrukreinigersof reinigers op hoge temperatuur om dezeapparaten te reinigen.
• Voorkom dat er schoonmaakmiddelen op deinterne componenten of kleppen terecht komen.
• Laat het apparaat goed drogen voordat u het
opnieuw gebruikt.
Opmerking: opmerking voor de zorgverlener raadpleeg de toepasselijke servicehandleiding voor
herverwerkingsprocedures.

OnderhoudUw zorgverlener is verantwoordelijk voor benodigd
onderhoud zoals beschreven in de technischehandleiding van dit apparaat. Bel uw zorgverlener als
onderhoud nodig is.
Het condenskussen is het enige onderdeel dat een
gebruiker kan onderhouden. Er zijn geen andere
onderdelen die door de gebruiker kunnen worden
onderhouden.
De verwachte levensduur van dit apparaat bedraagt
minimaal vijf jaar.
Maak de vulaansluitingen van de stationaire en
draagbare apparaten schoon met een schone, droge enpluisvrije doek voordat u ze vult om zo bevriezing en
mogelijke defecte apparatuur te voorkomen.
Opmerking: elk extra onderhoud moet worden uitgevoerd
door een bevoegde onderhoudsmonteur of dienstverlener.

Accessoires

AEEA en RoHS

Dit symbool herinnert gebruikers eraan om deapparatuur aan het einde van de levensduur
naar een recyclingbedrijf te brengen, conformde AEEA-richtlijn (afgedankte elektrische enelektronische apparatuur).
Onze producten voldoen aan de beperkingen vande richtlijn inzake gevaarlijke stoffen (RoHS). Ze
bevatten geen sporen van lood of andere gevaarlijke
stoffen.

AfvoerLever de Sprint/Stroller/Hi Flow Stroller, inclusief alleonderdelen, altijd in bij uw thuiszorgaanbieder voor
een juiste verwijdering. U kunt ook contact opnemenmet de lokale autoriteiten voor inlichtingen over decorrecte afvoer van de batterij.

Transport en opslag
Het apparaat dient rechtop en goed geventileerd
te worden bewaard. Leg het apparaat niet op zijn
kant. Luchtvochtigheid tot 95% niet-condenserend.
Temperatuurbereik -40 °C tot 70 °C (-40 °F tot 158 °F).
Bedrijfstemperatuurbereik -10 °C tot 40 °C (14 °F tot
104 °F). Relatieve vochtigheid van 30% tot 75% nietcondenserend.
Opmerking: het atmosferische-drukbereik is 700 hPa tot
1060 hPa (hoogte 3048 m tot -304 m).

96 - NL PN MN236-C4 E | Gebruikershandleiding

Transportwagen onderdeelnr. 20765565

Sprint / Stroller / Hi Flow Stroller
Veiligheid
WAARSCHUWING: DRAAGBARE RF-COMMUNICATIEAPPARATUUR (WAARONDER RANDAPPARATUURZOALS ANTENNEKABELS EN EXTERNE ANTENNES) MOETEN STEEDS OP MEER DAN 30 CM (12 INCH)
VAN ALLE ONDERDELEN VAN DE SPRINT/STROLLER/HI FLOW STROLLER WORDEN GEBRUIKT, MET
INBEGRIP VAN DOOR DE FABRIKANT GESPECIFIEERDE KABELS. ANDERS KUNNEN DE PRESTATIESVAN DEZE APPARATUUR ERONDER LIJDEN.
WAARSCHUWING: HET GEBRUIK VAN ANDERE TOEBEHOREN, TRANSDUCERS EN KABELS DANOPGEGEVEN OF VERSTREKT DOOR DE FABRIKANT VAN DEZE APPARATUUR KAN LEIDEN TOT VERHOOGDEELEKTROMAGNETISCHE EMISSIES OF VERMINDERDE ELEKTROMAGNETISCHE IMMUNITEIT VAN DEZEAPPARATUUR EN LEIDEN TOT EEN ONGEWENSTE WERKING.
WAARSCHUWING: HET GEBRUIK VAN DEZE APPARATUUR NAAST OF GESTAPELD OP ANDERE APPARATUUR
MOET WORDEN VERMEDEN, WANT DIT KAN DE CORRECTE WERKING ERVAN VERSTOREN. ALS AANPALEND OFGESTAPELD GEBRUIK ONVERMIJDELIJK IS, MOET WORDEN GECONTROLEERD OF DEZE APPARATUUR EN DE
ANDERE APPARATUUR NORMAAL WERKEN.
Let op: voor medische elektrische apparatuur zijn speciale voorzorgsmaatregelen nodig met betrekking
tot Elektromagnetische Compatibiliteit (EMC) en deze moet worden geïnstalleerd en in bedrijf genomen inovereenstemming met de EMC-informatie in deze handleiding.
Let op: draagbare en mobiele RF-communicatie-apparatuur kan medische elektrische apparatuurbeïnvloeden.
Tabel 1

Richtlijnen en verklaring van de fabrikant - elektromagnetische emissies
De Sprint/Stroller/Hi Flow Stroller is bestemd voor gebruik in de hieronder gespecificeerde elektromagnetischeomgeving. De klant of de gebruiker van de Sprint/Stroller/Hi Flow Stroller dient te verzekeren dat het apparaat
in een dergelijke omgeving wordt gebruikt.
Emissietest

Naleving

Elektromagnetische omgeving - richtlijnen

RF-emissies

Groep 1De Sprint/Stroller/Hi Flow Stroller gebruikt alleen
RF‑energie voor zijn interne werking.

CISPR 11De RF-emissies zijn daarom erg laag en het is
onwaarschijnlijk dat deze enige storing zouden veroorzaken
in elektronische apparatuur die zich dichtbij bevindt.

RF-emissies CISPR 11

Klasse B

Harmonische emissies
IEC 61000-3-2

Niet vantoepassing

Spanningsfluctuaties/
flikkeremissies
IEC 61000-3-3

Niet vantoepassing

De Sprint/Stroller/Hi Flow Stroller is geschikt voor
gebruik in alle instellingen, waaronder instellingen met een
woonfunctie en instellingen die rechtstreeks zijn aangesloten
op een laagspanningsnetwerk dat gebouwen met een
woonfunctie van stroom voorziet.

PN MN236-C4 E | Gebruikershandleiding NL - 97

Sprint / Stroller / Hi Flow Stroller
Tabel 2*: Aanbevolen scheidingsafstanden tussen draagbare en mobiele RF-communicatieapparatuur en de Sprint/
Stroller/Hi Flow Stroller
De Sprint/Stroller/Hi Flow Stroller is bestemd voor gebruik in een elektromagnetische omgeving waarin uitgestraalde RFstoringen worden beheerd. De klant of de gebruiker van de Sprint/Stroller/Hi Flow Stroller kan elektromagnetische storing helpenvoorkomen door een minimale scheidingsafstand tussen de draagbare en mobiele RF-communicatieapparatuur (zenders) ende Sprint/Stroller/Hi Flow Stroller in acht te nemen zoals hieronder aanbevolen, overeenkomstig het maximumvermogen van decommunicatieapparatuur.
Nominaal maximuScheidingsafstand volgens frequentie van de zendermvermogen van demzender80 MHz tot 800 MHz
800 MHz tot 2,5 GHz
150 kHz tot 80 MHzW

d = 1,2√P

d = 1,2√P

d = 2,3√P

0,01
0,12 m0,12 m0,23 m0,10,38 m0,38 m0,73 m
1
1,2 m
1,2 m
2,3 m
10
3,8 m
3,8 m
7,3 m
100
12 m
12 m
23 m
Voor zenders met een ander nominaal maximum uitgangsvermogen dan hierboven vermeld, kan de aanbevolen scheidingsafstand(d) in meter (m) worden geschat met behulp van de vergelijking die van toepassing is op de frequentie van de zender, waarbij P
staat voor het nominale maximale uitgangsvermogen van de zender in watt (W) volgens de fabrikant van de zender.
OPMERKING 1 Bij 80 MHz en 800 MHz geldt de scheidingsafstand voor het hogere frequentiebereik.
OPMERKING 2 Het kan zijn dat deze richtlijnen niet op alle situaties van toepassing zijn. Elektromagnetische propagatie wordtbeïnvloed door absorptie en reflectie van gebouwen, voorwerpen en mensen.

* Deze tabel is opgenomen als standaardvereiste voor apparatuur die is getest volgens specifieke testniveaus en
specifieke frequentiebereiken en waarvan is vastgesteld dat deze voldoet aan de regelgeving.g.

Tabel 3: Richtlijnen en verklaring van de fabrikant - elektromagnetische immuniteit
De Sprint/Stroller/Hi Flow Stroller is bestemd voor gebruik in de hieronder gespecificeerde elektromagnetische omgeving. De klant
of de gebruiker van de Sprint/Stroller/Hi Flow Stroller dient te verzekeren dat het apparaat in een dergelijke omgeving wordt gebruikt
Immuniteitstest
IEC 60601-testniveau
Nalevingsniveau
Elektromagnetische omgeving - richtlijnen
Houten, betonnen of keramische tegelvloer
Elektrostatische± 8 kV contact± 8 kV contact
vereist. Indien de vloerbedekking bestaat
ontlading (ESD)± 2kV, ± 4 kV ±8 kV, ±± 2kV, ± 4 kV ±8 kV, ±uit synthetisch materiaal, dient de relatieve
15kV lucht
15kV lucht
IEC 61000-4-2
luchtvochtigheid minimaal 30% te zijn.**Niet van toepassing
Snelle elektrischetransiënten/burst
IEC 610004-4

Stroomstoten
IEC 61000-4-5
Spanningsdalingen,korte onderbrekingenen spanningsfluctuaties bij ingangslijnen
van de stroomvoorziening
IEC 61000-4-11±2 kV voor voedingskabels

Op gelijkstroom werkend
apparaat

Niet van toepassing±1 kV voor ingangs-/uitgangNiet van toepassing
sleidingen
Geen gegevens ingangs-/
uitgangsleidingen
Niet van toepassing± 1 kV lijn(en) naar lijn(en)Op gelijkstroom werkend Niet van toepassing±2 kV lijn(en) naar aarde
apparaat< 5% UT (> 95% daling inUT) gedurende 0,5 cyclus
40% UT (60% daling in UT)
gedurende 5 cycli
70% UT (30% daling in UT)
gedurende 25 cycli< 5% UT (> 95% daling inUT) gedurende 5 sec.

Netfrequentie (50/60 Hz) 3 A / mmagnetisch veld
50/60 HzIEC 61000-4-8

Niet van toepassing
Op gelijkstroom werkend
apparaat

Niet van toepassing

3A/m
50/60 HzDe magnetische velden van de netfrequentie
dienen die van een typische commerciële ofziekenhuisomgeving te zijn.

Opmerking: UT is de spanning van de wisselstroomnetvoeding voorafgaand aan de toepassing van het testniveau.

** Deze zin geeft aan dat de verplichte tests zijn uitgevoerd in een gecontroleerde omgeving en dat is gebleken
dat de Sprint/Stroller/Hi Flow Stroller voldoen aan de regelgeving.

98 - NL PN MN236-C4 E | Gebruikershandleiding

Sprint / Stroller / Hi Flow Stroller
Tabel 4Richtlijnen en verklaring van de fabrikant - Immuniteit voor elektromagnetische apparatuur enelektromagnetische systemen
Richtlijnen en verklaring van de fabrikant - Immuniteit
De Sprint/Stroller/Hi Flow Stroller is bestemd voor gebruik in de hieronder gespecificeerde elektromagnetischeomgeving. De klant of de gebruiker van de Sprint/Stroller/Hi Flow Stroller dient te verzekeren dat het apparaat
in een dergelijke omgeving wordt gebruikt.
Immuniteitstest
IEC 60601-testniveau
Nalevingsniveau
Elektromagnetische omgeving - Richtlijnen
Geleide RF
IEC 61000-4-6

3 Vrms6 Vrms (in ISMfrequentiebanden)
150 kHz tot 80 MHz

Niet van toepassing
Door batterijen
aangedreven apparaat,
geen SIP/SOPDraagbare en mobiele RF-communicatieapparatuur
mag niet dichter bij onderdelen van de Sprint/Stroller/
Hi Flow Stroller, inclusief de kabels, worden gebruikt
dan de aanbevolen scheidingsafstand die werd
berekend a.d.h.v. de vergelijking die toepasselijk is op
de frequentie van de zender.
Aanbevolen scheidingsafstandd = 1,2 √P

Uitgestraalde RF
IEC 61000-4-3

80 MHz tot 2,7 GHz

10 V/m80 MHz tot 2,7 GHz
80% AM bij 1 kHz

d = 1,2 √P
d = 2,3 √P
waarbij P staat voor het nominale maximale
uitgangsvermogen van de zender in watt (W) volgens
de fabrikant van de zender en d voor de aanbevolen
scheidingsafstand in meter (m).
Veldsterktes van stationaire RF-zenders,zoals bepaald door een elektromagnetischterreinonderzoeka, dienen lager te zijn dan het
nalevingsniveau in elk frequentiebereikb.
Interferentie kan voorkomen in de nabijheid van
apparatuur waarop het volgende symbool staat:

Maximaal
vermogen(W)

Afstand(m)

Immuniteits­testniveau
(V/m)

1,8

0,3

27

2

0,3

28

Pulseermodulatieb)
217 Hz

0,2

0,3

9

Pulseermodulatieb)
18 Hz

2

0,3

28

Pulseermodulatieb)
217 Hz20,328

Pulseermodulatieb)
217 Hz20,328

Pulseermodulatieb)
217 Hz

0,2

0,3

9

Testfrequentie(MHz)

Frequentiebanda)(MHz)

Servicea)

Modulatieb)

385

380–390TETRA 400450

430–470

GMRS 460, FRS 460

Pulseermodulatieb)
18 Hz
FMC)
±5 kHz afwijking
1 kHz sinus

704–787

LTE-frequentieband
13, 17

710
745
780
810
870
930
1720
1845
1970

800–960

1700–1900

2450

2400–2570

5240
5500
5785

5100–5800

GSM 800/900, TETRA
800, IDEN 820,CDMA 850, LTEfrequentieband 5
GSM 1800;
CDMA 1900;
GSM 1900; DECT;
LTE-frequentieband 1,
3, 4, 25; UMTS
Bluetooth, WLAN,
802.11 b/g/n,
RFID 2450, LTEfrequentieband 7
WLAN 802.11 a/n

Opmerking: als het noodzakelijk is om het IMMUNITEITSTESTNIVEAU te bereiken, dan kan de afstand tussen deantenne die uitzendt en de Sprint/Stroller/Hi Flow Stroller verkleind worden naar 1 meter. De testafstand van 1 meter wordt
toegestaan door IEC 61000-4-3.
Voor sommige services worden alleen de transmissiefrequenties opgenomen.
Het draagsignaal zal worden aangepast door middel van een blokgolfsignaal met een arbeidscyclus van 50%.
Als alternatief voor een FM-modulatie, kan een pulseermodulatie van 50% bij 18 Hz worden gebruikt. Ook al vertegenwoordigt dat niet dedaadwerkelijke modulatie, betreft het wel het ergste scenario.
abc

CAIRE en CAIRE Inc. zijn gedeponeerde handelsmerken van CAIRE Inc. Bezoek onze onderstaande website voor een volledige lijst van handelsmerken. Handelsmerken: www.careinc.
com/corporate/trademarks.
Copyright © 2023 CAIRE Inc. CAIRE Inc. behoudt zich het recht voor om te allen tijde zonder voorafgaande kennisgeving en zonder verdere verplichtingen of gevolgen producten uit het
assortiment te halen en om prijzen, materialen, uitrusting, kwaliteit, beschrijvingen, specificaties en/of processen van producten te wijzigen. Alle hierin niet uitdrukkelijk genoemde rechtenzijn, waar van toepassing, voorbehouden.

PN MN236-C4 E | Gebruikershandleiding NL - 99

Sprint / Stroller / Hi Flow Stroller
Symbolforklaring
ISO 7000

Der må ikke ryges i nærheden af
enheden eller under drift. Reg.nr. P002

Må ikke udsættes for regn. Opbevares
tørt. Reg.nr. 0626

Type BF-del (beskyttelsesgrad modelektrisk stød). Reg.nr. 5333

Stabelgrænse efter antal. Reg. # 2403
Producentens navn og adresse.
Reg.nr. 3082

Advarsel: Reg.nr. W001
Rådets direktiv 93/42/EØF
Autoriseret repræsentant i EU

Landet og fremstillingsdatoen.
"CC" identificerer landekoden på
to bogstaver for fremstillingslandet.
Fremstillingsdatoen er i formatetÅÅÅÅ-MM-DD. Reg. #6049
Forsigtig! Se medfølgende
dokumentation. Reg.nr. 0434A

Hvis den produkt-unikke
enhedsidentifikator (UDI) -mærkning
har CE #### -symbolet på, overholder
enheden kravene i direktiv 93/42 / EØFvedrørende medicinsk udstyr. CE ####
symbolet angiver det bemyndigedeorgan nummer.

Katalognummer. Reg.nr. 2493

Enheden overholder kravene i direktiv
2010/35/EU om transportabelt
trykbærende udstyr. Den bærer
pi‑mærkningen som vist.

Serienummer. Reg.nr. 2498
Opbevaring eller driftstemperaturområde.
Reg.nr. 0632
Opbevaring fugtighedsområde. Reg.
nr. 2620
Atmosfærisk trykbegrænsning. Reg.
nr. 2621
Denne side op. Reg.nr. 0623
Skrøbelig – skal håndteres forsigtigt.
Reg.nr. 0621

ADR
Ikke-giftig gas.
Fare Oxiderende stoffer:
brandforstærkende risiko.
Oxidationsmidler får ild til at brænde
kraftigere.
Nedkølet flydende gas, USP, fremstillet
ved likvefaktion af luft
Yderligere symboler
Opbevares altid et sted med god
ventilation

Indeholder farlige stoffer. Reg. #3723

Må ikke anbringes i nærheden afbrændbare materialer, olie og fedt.

Importør. Reg. #3725
ISO 7010

Autoriseret repræsentant i Schweiz.

Der kan forekomme forfrysninger ved
kontakt med kold væske, luftformig
ilt eller dele med is på. Advarselom lav temperatur. Advarer om lavetemperaturer eller frost. Reg.nr. W010
Brugervejledningen skal læses.
Reg.nr. M002
Må ikke udsættes for åben ild, brandeller gnister. Åben antændelseskilde og
rygning er ikke tilladt. Reg.nr. P003

100 - DK PN MN236-C4 E | Brugervejledning

Hvis enheden bærer UKCA-mærket som
vist med UKCA####, der angiver nummeret på det bemyndigede organ, overholder denne enhed UKCA-reglerne.
IEC 60417
Undlad at tildække enheden eller bæreden bærbare enhed under dit tøj. Disse
enheder udlufter normalt ilt. nr. 5641

Sprint / Stroller / Hi Flow Stroller
21 CFR 801.15
Ifølge føderal amerikansk lovgivningmå dette apparat kun sælges af en lægeeller efter ordination af en læge.
IEC 60601-1
Drypsikker
Rådets direktiv 2012/19/EU
WEEE
Dette symbol skal minde udstyrsejerne
om at returnere det til et genbrugsanlæg,når dets levetid er udløbet, i henhold til
direktivet om affald af elektrisk og elektronisk udstyr (WEEE).
Vores produkter vil overholde begrænsningen i direktivet om farlige stoffer(RoHS). De vil ikke indeholde mere end
spormængder af bly eller andet indhold
af farligt materiale.

ISO 15223-1
Medicinsk udstyr. Reg. # 5.7.7
Unik enhedsidentifikator # 5.7.10
Dette produkt er muligvis omfattet af ét eller flereamerikanske eller internationale patenter. Gå til vores
websted nedenfor for at se en liste over gældende
patenter. Pat.: www.caireinc.com/corporate/patents/.

Specifikationer• Funktionsmetode: Uafbrudt flow
• Beskyttelsestype mod elektrisk stød: Internt strømforsynet udstyr• Beskyttelsesgrad mod elektrisk stød: Type BF-del• Klassifikation i henhold til grad af beskyttelse mod indtrængen af vand: JP22 – Almindeligt udstyr• Udstyret må ikke anvendes i nærheden af brændbare blandinger
Produktspecifikationer

SprintStroller

Hi Flow Stroller

LOX-kapacitet

0,68 kg (1,5 lb)

1,36 kg (3,0 lb)

1,36 kg (3,0 lb)

Luftformig ækvivalent, kapacitet
Vægt, tom

513 l
2,04 kg (4,5 lb)

1026 L
2,27 kg (5,0 lb)

1025 L
2,49 kg (5,5 lb)Vægt, fyldt

2,72 kg (6,0 lb)

3,63 kg (8,0 lb)

3,86 kg (8,5 lb)

Højde

298 mm (11,75 in.)

343 mm (13,5 in.)

343 mm (13,5 in.)Mål

D 123,83 mm (4,875 in.) x
B 191 mm (7,5 in.)
4,3 timer

D 149,23 mm (5,875 in.) x
B 191 mm (7,5 in.)
8,5 timer

D 149 mm (5,87 in.) x
B 191 mm (7,5 in.)
8,5 timer

Typisk brugstid ved 2 LPM
Driftstryk

137 kPa (20 psi)

137 kPa (20 psi)

137 kPa (20 psi)

Normal fordampningshastighed

0,57 kg/dag (1,3 lb/dag)

0,57 kg/dag (1,3 lb/dag)

0,57 kg/dag (1,3 lb/dag)StandardflowreguleringsintervalFra, 0,25; 0,5; 0,75; 1; 1,5; 2; Fra, 0,25; 0,5; 0,75; 1; 1,5; 2; Fra, 0,5, 1, 2, 2,5, 3, 4, 6, 8,
2,5; 3; 4; 5; 6 LPM
2,5; 3; 4; 5; 6 LPM
10, 12, 15 LPM+/- 10 %+/- 10 %+/- 10 %Flowmålernøjagtighed** Denne nøjagtighed gælder kun ved 70F og 14,7 psig og med en kalibreret nøjagtig masseflowmåler.

PN MN236-C4 E | Brugervejledning DK - 101

Sprint / Stroller / Hi Flow Stroller
Advarsler
Vigtigt! Læs denne vejledning grundigt, før Sprint/
Stroller/Hi Flow Stroller tages i brug.
Kun på recept.
ADVARSEL: DETTE APPARAT ERIKKE EGNET TIL LIVSFORLÆNGENDEANVENDELSE.
ADVARSEL: HVIS DU MENER, AT UDSTYRET IKKEFUNGERER KORREKT, BEDES DU KONTAKTE
LEVERANDØREN. DU MÅ IKKE FORSØGE ATREPARERE ELLER JUSTERE ENHEDEN SELV.
ADVARSEL: DU MÅ IKKE ÆNDRE UDSTYRET UDEN
GODKENDELSE FRA PRODUCENTEN.
ADVARSEL: HVIS DER KRÆVES KONTINUITET I
ILTFORSYNINGEN, SKAL DET SIKRES, AT DER TIL
ENHVER TID ER TILSTRÆKKELIG FORSYNING AF
ILT, OG/ELLER AT EN SEKUNDÆR ILTFORSYNING ER
TILGÆNGELIG UNDER BEHANDLINGEN.
ADVARSEL: UNDGÅ RYGNING, LEVENDE LYS
ELLER ÅBEN ILD INDEN FOR 3 METER (10 FT) AF
ENHEDEN ELLER UNDER 20 CM (8 IN.) FRA EN
ANTÆNDELSESKILDE.
ADVARSEL: ENHEDEN SKAL BEFINDE SIG ET STEDMED GOD VENTILATION.
ADVARSEL: UDSTYR MED FLYDENDE
ILT MÅ IKKE OPBEVARES I ET SKAB,
I BAGAGERUMMET PÅ EN BIL ELLERNOGEN ANDRE INDELUKKEDE OMRÅDER.
UDSTYRET MÅ IKKE TILDÆKKES MED
TÆPPER, GARDINER ELLER ANDRETEKSTILER.
ADVARSEL: DETTE PRODUKT KAN UDSÆTTE
DIG FOR KEMIKALIER, HERUNDER NIKKEL,
SOM I STATEN CALIFORNIEN ER KENDT FOR AT
FORÅRSAGE KRÆFT. FÅ FLERE OPLYSNINGER PÅ
WWW.P65WARNINGS.CA.GOV.
ADVARSEL: HVIS DER OPSTÅR EN ALVORLIGHÆNDELSE MED DENNE ENHED, SKAL BRUGERENSTRAKS RAPPORTERE HÆNDELSEN TIL UDBYDEREN OG / ELLER PRODUCENTEN. EN ALVORLIGHÆNDELSE DEFINERES SOM EN SKADE, DØD ELLERPOTENTIALE TIL AT FORÅRSAGE SKADE / DØD, HVISDER ER EN GENTAGELSE AF HÆNDELSEN. BRUGEREN KAN OGSÅ RAPPORTERE HÆNDELSEN TIL
DEN KOMPETENTE MYNDIGHED I DET LAND, HVOR
HÆNDELSEN FANDT STED.
Forsigtig: Anvend kun Sprint/Stroller/Hi Flow
Stroller efter lægens anvisninger.
Forsigtig: Ifølge føderal amerikansk lovgivning må detteapparat kun sælges af en læge eller efter ordination afen læge.
Forsigtig: Enheden indeholder flydende ilt, som er ekstremt kold, næsten -184 °C (-300 °F). Eksponering for så
lav en temperatur kan give alvorlige forfrysninger.

102 - DK PN MN236-C4 E | Brugervejledning

Forsigtig: Flydende og luftformig ilt er ikke brændbar,
men kan få andre materialer til at brænde hurtigere end
normalt. Denne fare samt den flydende ilts lave temperatur nødvendiggør visse sikkerhedsforanstaltninger.
Forsigtig: Hold brændbare materialer væk fra dette udstyr. Aerosolspray, olie og fedt, herunder ansigtscremer
og vaseline, er let antændelige og kan brænde hurtigt inærheden af ilt.
Forsigtig: Rygning under brug af iltbrille kan udløse
forbrænding i ansigtet og risiko for dødsfald.
Hvis du tager iltbrillen af og anbringer den på tøj, sengetøj, møbler eller andet polstret materiale, udløses en
eksplosionsagtig brand, hvis den kommer i forbindelse
med cigaretter, varmekilder, gnister eller flammer.
Hvis du ryger, skal du gøre følgende: (1) Sluk den bærbare enhed, (2) tag iltbrillen af, og (3) forlad det lokale,
hvor udstyret er placeret.
Forsigtig: Hvis enheden ved et uheld vælter, skal dustraks, men forsigtigt, rejse den op, hvis det er muligt.
Hvis der slipper flydende ilt ud, skal du straks forladeområdet og kontakte leverandøren. Du må ikke forsøge
at flytte enheden eller forhindre den flydende ilt i atslippe ud.
Bemærk: Du må ikke røre ved dele af enheden med is på.
Bemærk: Du må ikke opbevare eller bruge den bærbareenhed, når den er koblet til den stationære enhed.
Bemærk: Personer, som ikke er oplært i systemet, må ikkehåndtere eller betjene udstyret.
Bemærk: De føderale amerikanske flymyndigheder har
forbudt anvendelse af dette udstyr på almindelige ruteflyog fragtfly.

Anvendelsesformål

CAIRE Sprint/Stroller/Hi Flow Stroller er beregnet
til administration af supplerende ilt. Enheden er ikke
beregnet til livsstøtte og giver heller ikke nogen patientovervågningsfunktioner.

Sprint / Stroller / Hi Flow Stroller
Indledning

Betjeningselementer

Sprint, Stroller og Hi Flow Stroller er beregnet tiltilførsel af supplerende ilt til patienten i slutbrugerens
hjem og kan endvidere anvedes i institutioner såsom
plejehjem eller subakutte plejefaciliteter. Sprint/
Stroller/Hi Flow Stroller er en ambulalant iltforsyningtil længere tids brug.

1. Elektronisk måleinstrument for væskeniveau
2. Elektronisk omskifter for væskeniveau
3. Mekanisk måleinstrument for væskeniveau
4. Flowreguleringsknap
5. Konnektor til iltslange
6. Udluftningsventil
7. Fyldningsåbninger (identificering af
lynudkoblingsventiler vises på næste side)
8. Bærerem med skulderpolster
9. Kondensationspude
10. Kondensationskop

Enheden anvendes af KOL-patienter og patienter med
nedsat lungefunktion. Enheden ordineres til patienten.
Enheden sælges til en leverandør, som er uddannet i at
betjene og servicere Sprint/Stroller/Hi Flow Stroller.
Leverandører oplærer brugeren.
Enheden er ikke beregnet som respirator, og den har
ingen yderligere patientovervågningsfunktioner. Det
anbefales at have en alternativ kilde til supplerende
ilt i tilfælde af mekanisk fejl. Disse bærbare enheder
har også en lynudkoblingsventil, som kan kobles til
en beholder med flydende ilt til omfyldning af denbærbare enhed. Beholderen opfyldes af leverandøren.

1
2
4

5

Denne brugervejledning indeholder instruktioner
i brugen af Sprint/Stroller/Hi Flow Stroller. Du kanlæse om betjening af den stationære enhed i den dertil
hørende brugervejledning.

8Bemærk: Leverandøren hjælper med den indledendeopsætning og underviser i korrekt håndtering og betjening
af enheden.

5

7
Sprint, Stroller og Hi Flow Stroller

6

10
9

PN MN236-C4 E | Brugervejledning DK - 103

Sprint / Stroller / Hi Flow Stroller
Identificering af lynudkoblingsventil
Konnektor tilbærbar enhed

Konnektor
til stationær
enhed

Lynudkoblingsventiltype
CAIREsidefyldningSFlåsning
4. a. Hvis konnektoren er en trykkonnektor:Sænk din Sprint/Stroller/Hi Flow Stroller ned
på den stationære enhed, indtil du kan mærke, atkonnektoren er i indgreb.

LIFE-OX
TF-låsning
PENOX

TAEMA
TF-trykkonnektorPB

b. H
 vis konnektoren er af typen Penox eller Life-Ox:
Drej din Sprint/Stroller/Hi Flow Stroller, indtil
du kan mærke, at konnektoren er i indgreb. Drej
derefter din Sprint/Stroller/Hi Flow Stroller –forsigtigt og fast med uret (), mens du trykker
nedad.

BrugsanvisningSådan fyldes Sprint/Stroller/Hi Flow Stroller
ved topfyldning.
1. Rengør fyldekonnektorerne både på den stationære
enhed og på din Sprint/Stroller/Hi Flow Stroller
med en ren, tør, fnugfri klud.
Forsigtig: Fyldekonnektorerne skal væretørre, fordi fugt kan få din Sprint/Stroller/Hi
Flow Stroller og den stationære enhed til atfryse fast til hinanden.
2. Sørg for, at flowreguleringsknapperne er i stilling
0 (Fra) – både på din Sprint/Stroller/Hi Flow
Stroller og på beholderen.
3. Placér fyldekonnektoren i den lodrette position overden stationære enheds konnektor.

104 - DK PN MN236-C4 E | BrugervejledningBemærk: Fortsættes i trin 6 i Sådan fyldes din Sprint/Stroller/Hi Flow Stroller ved sidefyldning.

Sådan fyldes din Sprint/Stroller/Hi Flow
Stroller ved sidefyldning.
1. Rengør konnektorerne både på den stationære enhedog på din Sprint/Stroller/Hi Flow Stroller med enren, tør, fnugfri klud.
ADVARSEL: FYLDEKONNEKTORERNE SKAL
VÆRE TØRRE, FORDI FUGT KAN FÅ DIN
SPRINT/STROLLER/HI FLOW STROLLER OG
DEN STATIONÆRE ENHED TIL AT FRYSEFAST TIL HINANDEN.

Sprint / Stroller / Hi Flow Stroller
2. Sørg for, at flowreguleringsknapperne er i stilling
0 – både på din Sprint/Stroller/Hi Flow Stroller og
på beholderen.

Bemærk: Visningen af væskeniveauet på instrumentet er
mest præcis 30 sekunder eller mere efter påfyldningen af
din Sprint/Stroller/Hi Flow Stroller.

3. Sørg for, at fyldekonnektoren til din Sprint/Stroller/
Hi Flow Stroller er i lodret position i forhold til
konnektoren til den stationære enhed.

4. Drej din Sprint/Stroller/Hi Flow Stroller mod uret(), indtil du kan mærke, at pinden og åbningen
er i indgreb.

7. Du kan muligvis også se damp omkringforbindelsen. Det er helt normalt og skyldes denstore temperaturforskel mellem den flydende ilt og
den varme, omgivende luft.
8. Luk udluftningsventilen, når enheden er fuld. Nårhvislelyden ændres, og der kommer væske ud af
udluftningsslangen, er enheden fuld.
Bemærk: Hvis ventilen lukkes efter efter en næsten
sammenhængende udstrømning af væske, bliver din
Sprint/Stroller/Hi Flow Stroller fyldt næsten helt op, men envis mængde ilt vil gå tabt.

5. Drej din Sprint/Stroller/Hi Flow Stroller forsigtigt
og fast tilbage til lodret position. Nu er enhederne låstsammen.

Bemærk: Fortsættes i trin 6.
Bemærk: Hvis konnektoren er en trykkonnektor, skalenheden holdes på plads under påfyldningen.

Bemærk: Hvis ventilen lukkes allerede ved det første tegnpå udstrømning af væske, fyldes din Sprint/Stroller/Hi Flow
Stroller ikke helt, men det sparer på den flydende ilt.
9. Hvis konnektorerne er trykkonnektorer, skal du
trykke på udløserknappen, indtil enheder bliver
adskilt fra hinanden. Hvis konnektorerne derimod
ikke er trykkonnektorer, skal du dreje enheden,
indtil den bliver adskilt fra den stationære enhed.

Udløserknap på den bærbare enhed(kun varianter med trykkonnektor)

6. Åbn udluftningsventilen for at påbegyndepåfyldningen af din Sprint/Stroller/Hi Flow Stroller.
Du vil kunne høre noget ilt slippe ud, men det gør
ikke noget. Den hvislende lyd er helt normal.
PN MN236-C4 E | Brugervejledning DK - 105

Sprint / Stroller / Hi Flow Stroller
Forsigtig: Kontrollér førstvæskeniveauinstrumentet, efter ventilen er
lukket.
Forsigtig: Hvis din Sprint/Stroller/Hi Flow Stroller
bliver fyldt op umiddelbart efter brug, skal
flowreguleringsventilen indstilles til nul i 10 minutter
efter påfyldningen, så trykket kan stabilisere sig. Brugderefter enheden som normalt.
ADVARSEL: DU MÅ IKKE BETJENE DINSPRINT/STROLLER/HI FLOW STROLLER,
MENS DEN ER SLUTTET TIL EN STATIONÆR
ENHED.
ADVARSEL: BRUG IKKE UNØDIG KRAFT, HVIS DINSPRINT/STROLLER/HI FLOW STROLLER ER SVÆRAT KOBLE FRA. ENHEDERNE KAN VÆRE FROSSET
SAMMEN. LAD ENHEDERNE SIDDE SAMMEN, OG
VENT, INDTIL ENHEDERNE VARMER OP – SÅ KAN DE
NEMT KOBLES FRA HINANDEN. DU MÅ IKKE RØREVED DELE MED IS PÅ. ADVARSEL: HVIS DER LØBER
SÅ MEGET VÆSKE UD, AT DER ER EN STRØM AFVÆSKE, SKAL DU FORLADE OMRÅDET, OG STRAKSKONTAKTE LEVERANDØREN.
ADVARSEL: FLYDENDE ILT ELLER KOLD,LUFTFORMIG ILT KAN FORSÅRSAGEFORFRYSNINGER PÅ HUDEN. ILT MEDFØRER
FORHØJET BRANDFARE.
Forsigtig: Hvis der opstår lækage, når den
stationære enhed og din Sprint/Stroller/Hi
Flow Stroller kobles fra hinanden, skal din
Sprint/Stroller/Hi Flow Stroller stilles af vejen –men den skal forblive lodret. Forlad lokalet, og
kontakt omgående leverandøren.

Bemærk: Til udstyrsudbyder: Følgende tilbehør til
iltadministration anbefales til brug sammen med Sprint/ Stroller / Hi Flow Stroller:

Brugsanvisning(Iltbrille medfølger ikke).

1. Tryk iltbrillens vejtrækningsslange fast på
konnektoren til iltslangen.
2. Juster iltbrillen til den rigtige position, så du let kanindånde ilten.
3. Drej flowreguleringsknappen på toppen af din
Sprint/Stroller/Hi Flow Stroller med uret, indtil det
fastsatte flow vises (med tal) i knappens "vindue",og der kan mærkes en positiv tilbageholdelse.
Forsigtig: Hvis flowreguleringsknappen
indstilles mellem flowhastigheder, opstår der
iltflow uden for specifikationen.
4. Du skulle nu få ilt. For at stoppe
iltgennemstrømningen skal flowreguleringknappen
drejes mod uret til stilling 0.
Forsigtig: Drej altid flowreguleringsknappen til(0), når enheden ikke er i brug.

• Nasal kanyle med 7 fod (2,1 m) af rør (Maksimal
LPM: 6 LPM): CAIRE Delnummer 5408-SEQ• Firebreak: CAIRE-delenummer 21126636
En brandbryd anbefales til brug med enhver kanyle.
• CAIRE tilbyder en brandbryd beregnet til at blive
brugt i forbindelse med den bærbare ilt. Firebreaker en termisk sikring for at stoppe strømmen af gas itilfælde af, at nedstrøms kanyle eller iltrør antændes
og brænder til ildstedet. Det anbringes på linje mednæsekanylen eller iltrøret mellem patienten og iltudløbet fra Sprint / Stroller / Hi Flow Stroller. For korrekt
brug af firebreak skal du altid henvise til producentens
instruktioner (inkluderet i hvert firebreak kit).
• Yderligere oplysninger om anbefalet tilbehør ertilgængelige online på www.caireinc.com.

106 - DK PN MN236-C4 E | Brugervejledning

5. Brug følgende skema som retningslinje for atfastsætte drifttiden for din Sprint/Stroller/Hi Flow
Stroller:

Sprint / Stroller / Hi Flow Stroller
Flowreguleringsventilens indstilling
Model

SprintStroller

Hi Flow
Stroller

Nominelt

Nominelt

Flowreguleringsventilens
position

SprintStroller

Hi Flow
Stroller

Fra

30.00

60.00

71.00

0,25

18.00

30.00

I/R

0,5

12.12

23.00

32.00

0,75

9.18

17.24

I/R

1

7.30

14.06

16.00

1,5

5.30

10.12

I/R

2

4.18

8.00

8.00

2,5

3.36

6.36

6.20

3

3.06

5.36

5.20
4.00

4

2.24

4.24

5

2.00

3.36

I/R

6

1.42

3.00

2.40

8

-

-

2.00

10

-

-

1.40

12

-

-

1.20

15

-

-

1.00

Bemærk: Tiderne er angivet i timer og minutter(på formatet tt.mm).
Bemærk: De "nominelle" tider opnås under ideellebetingelser, dvs. maksimal påfyldning, nøjagtige
flowhastigheder, god tabshastighed, ingen flytning
af enhed osv.
Disse tider er forventede maksimumstider.
Bemærk: Dine individuelle resultater kommer til atvariere.
6a. For at kontrollere mængden af flydende ilt
i enheden med den elektroniske måler forvæskeniveau:• Gen. 4-måleinstrument: Tryk på den grønne
betjeningsknap på instrumentets forside. Aflæs
væskestanden på den runde LED-skala.

Forsigtig: Hvis kun det sidste diodelys ertændt, er din Sprint/Stroller/Hi Flow Stroller
tom.
Hvis indikatorlyset for lavt batteriniveau tænder,
når knappen holdes nede, skal du kontakte
leverandøren.
Bemærk: Din Sprint/Stroller/HI Flow Stroller fortsætter med
at afgive ilt, så længe der er flydende ilt i enheden, ogsånår indikatorlyset for lavt batteriniveau er tændt.
6b. For at kontrollere mængden af flydende ilt
i enheden med den mekaniske måler forvæskeniveau:• Støt enheden ved at holde på stroppen lige oven overdet sorte målekammer på instrumentets indikator.
• Lad enheden stabilere sig (dvs. ikke hoppe op
og ned), og aflæs derefter den omtrentlige mængde
flydende ilt på den viste farveskala.

• Gen. 3-måleinstrument: Hold trykknappen(omskifteren for væskeniveau) nedtrykt i mindst to
sekunder. Aflæs væskestanden øverst i lysbjælken.

PN MN236-C4 E | Brugervejledning DK - 107

Sprint / Stroller / Hi Flow StrollerBemærk: Hvis enheden er tom, vises kun det røde båndpå skalaen. I så fald skal enheden fyldes op inden brug.
7. Under visse forhold – i forbindelse med
vedvarende brug og især, hvis den bærbare enhedikke bliver flyttet – kan der dannes is omkring
fordampningsspolen og på beholderen. Denne
isdannelse kan mindskes ved at banke på enheden
og/eller aftørre ophobet is af beholderen.

Vedligeholdelse af
kondensationspude
Fjern al fugt fra kondensationskoppen, når enheden er
tom og opvarmet til stuetemperatur:
Hardcase-beholder:• Tip enheden om på den ene side. Tag fat iknappen midt i kondensationskoppen (påbunden af enheden), og træk den lige ud. Herved
frigøres koppen.
Bemærk: Hvis puden er meget snavset, kan den vaskes ivaskemaskine (brug blegemiddel for at desinficere den).
• Koppen udskiftes ved at placere den lige
over midten af den runde åbning i bunden af
enheden, hvorefter knappen trykkes ind med
et klik, så koppen indkobles – og derefter med
endnu et klik for at låse koppen fast.

• Fjern puden fra koppen, og vrid den absorberedefugt. Lad puden blive helt tør, inden denplaceres i koppen igen.
Bemærk: Hvis puden er meget snavset, kan den vaskes
i vaskemaskine (brug blegemiddel for at desinficere den).
• Puden genindsættes ved at anbringe den
i bundbetrækket og derefter lukke lynlåsen
i bunden igen.
Bemærk: Hvis knappen ikke kan trykkes ind for at indkoble
koppen, så hold fast i koppen, og træk knappen ud, indtil
den klikker fast i den åbne position. Nu kan du så udskifte
koppen som beskrevet ovenfor.

Frostreduktion
Under visse forhold – i forbindelse med vedvarendebrug og især, hvis den bærbare enhed ikke bliver flyttet– kan der dannes is omkring fordampningsspolen og
på beholderen. Denne isdannelse kan mindskes ved
at banke på enheden og/eller aftørre ophobet is af
beholderen.
ADVARSEL: PÅ GRUND AF MULIGHEDENFOR STÆRK FUGTDANNELSE/ISDANNELSEANBEFALES DET, AT DIN HI FLOW STROLLERHOLDES I STILSTAND 1 TIME EFTER BRUG
FOR AT REDUCERE FUGTDANNELSE.

Låst
I indgrebÅben

Knap KopBemærk: Hvis knappen ikke kan trykkes ind for at indkoble
koppen, så hold fast i koppen, og træk knappen ud, indtil
den klikker fast i den åbne position. Nu kan du så udskifte
koppen som beskrevet ovenfor.
Softcase-beholder:• Åbn den bageste lynlås, og træk bundbetrækket
tilbage.

108 - DK PN MN236-C4 E | BrugervejledningPå grund af den højere grad af fugtdannelse ved din
Hi Flow Stroller vil der opstå kondensation og en delisdannelse på enheden, især ved flowindstillinger på
10 LPM og derover. Det anbefales, at enheden holdes
inaktiv i mindst 1 time efter brugen for at reducere
muligheden for kraftig fugtdannelse på grund af den
vedvarende brug.

Sprint / Stroller / Hi Flow Stroller
Fejlfinding
Følgende oplysninger kan hjælpe dig med at afhjælpe fejl og løse enkle funktionsproblemer, du kan komme ud
for, når du bruger din Sprint/Stroller/Hi Flow Stroller.
ProblemLøsning
Den bærbare enhed• Der kan forekomme hvislen for at opretholde det korrekte tryk i den bærbare enhed. Den
udsender en hvislende lyd.
vil oftest hvisle efter påfyldning, eller når dens position ændres. Denne hvislen kan vare ica. 10 minutter efter påfyldning. Det kan også forekomme, når flowreguleringsventilen er
på en lav indstilling.

Flowet på den bærbareenhed stopper under
brugen.

Den bærbare enhed bliver
ikke fyldt op.

• Hvis den bærbare enhed har ligget i en ukorrekt position, skal den vendes tilbage til lodret
position. Vent derefter nogle minutter, så enheden kan stabilisere sig.
• Kontrollér, at iltbrillen er koblet korrekt til iltudtaget.
• Kontrollér, at der ikke er knæk på iltbrillen.
• Kontrollér indholdsmåleren, og opfyld den bærbare enhed, hvis det er nødvendigt.
• Kontrollér, at flowreguleringsknappen ikke står i stilling 0.
• Kontrollér, at der er ilt i den stationære beholder.

• Kontrollér, at fyldekonnektorerne på den bærbare enhed og på beholderen er koblet helt
sammen under påfyldningen.
Den bærbare enheds• Hvis udluftningsventilen ikke lukker, og den hvislende lyd og iltdampskyen ikke forsvinder,
udluftningsventil lukker
skal du forsigtigt fjerne den bærbare enhed ved at trykke udløserknappen på den
ikke korrekt efter
stationære beholder ned. Udblæsningen fra bunden af den bærbare enhed stopper ipåfyldningsprocessen.
løbet af nogle minutter. Lad enheden varme op, indtil du kan lukke udluftningsventilen.
Den bærbare enhed kan bruge op til 60 minutter på at genoprette tilstrækkeligt tryk til
nøjagtigt iltflow. Hvis der er behov for det, kan du bruge en alternativ iltforsyning, f.eks. en
flowreguleringsventil monteret på den stationære beholder.
Den bærbare enhed er svær • Fyldkonnektorerne på den bærbare enhed og på beholderen kan være frosset.
at koble fra beholderen efter• DU MÅ IKKE TVINGE DEM FRA HINANDEN. Vent nogle minutter, indtil de frosne delepåfyldning.
er varmet op, og kobl den bærbare enhed fra, når isen er smeltet. Du kan forhindre,
at enhederne fryser sammen ved altid at tørre hankonnektoren på beholderen og
hunkonnektoren på den bærbare enhed med en ren, tør klud før påfyldning.

PN MN236-C4 E | Brugervejledning DK - 109

Sprint / Stroller / Hi Flow Stroller
Rengøringsstandard
ADVARSEL: ENHEDEN MÅ KUN RENGØRES,NÅR DEN ER TOM.
• Rengøres med en opløsning af mildt
opvaskemiddel og vand.
• Påfør rengøringsopløsningen på en fnugfri
klud. Godkender rengøringsmidler omfatter
HydroPure og HydroKlean. Sprøjt ikke
rengøringsmidler direkte på din Sprint/Stroller/
Hi Flow Stroller.
• Tør den udvendige flade af med en fnugfri klud,
indtil den er ren.
Forsigtig: Brug ikke vaskeudstyr med høje
temperaturer eller højt tryk til rengøring af
disse enheder.
• Der må ikke komme rengøringsmiddel på nogenaf de indvendige dele eller ventilerne.
• Lad enheden tørre helt, før den tages i brug igen.
Bemærk: Note til leverandøren – se den relevante
servicevejledning vedrørende oparbejdningsprocedurer.

WEEE og RoHS

Vedligeholdelse
Leverandøren er ansvarlig for al vedligeholdelse,der måtte blive nødvendig i henhold til den tekniske
vejledning til dette udstyr. Ring til leverandøren
vedrørende eventuelle vedligeholdelseskrav.
Kondensationspuden er den eneste løsdel, sombrugeren kan vedligeholde. Ingen andre løsdele kanvedligeholdes af brugeren.
Den forventede levetid er minimum fem år.
Rengør fyldekonnektorerne på både den stationære og
den bærbare enhed med en ren, tør, fnugfri klud mellem
hver påfyldning for at forhindre, at der dannes is, og at
udstyret svigter.
Bemærk: Yderligere nødvendig vedligeholdelseskal udføres af en kvalificeret servicetekniker eller
serviceleverandør.

Tilbehør

Dette symbol har til formål at minde
ejere af udstyr om at returnere det til etgenvindingsanlæg, når det er udtjent, i henhold
til direktivet om affald af elektrisk og elektronisk
udstyr (WEEE).
Vores produkter overholder begrænsningerne i RoHSdirektivet. De indeholder ikke mere end spor af bly
eller andre farlige stoffer.

Bortskaffelse
Returner altid din Sprint/Stroller/Hi Flow Stroller
sammen med alt tilbehørm så den kan bortskaffes
korrekt. Du kan også kontakte den lokale myndighedangående instruktioner om korrekt bortskaffelse afbatteriet.

Transport og opbevaring
Udstyret skal opbevares opretståendde et sted med
god ventilation. Udstyret må ikke ligge på siden.
Luftfugtighed op til 95 % ikke-kondenserende.
Temperaturområdet er -40 °C til 70 °C(-40 °F til 158 °F).
Driftstemperaturområdet er -10°C til 40 °C (14 °F til
104 °F). Området for relativ luftfugtighed er 30 % til
75 % ikke-kondenserende.
Bemærk: Området for atmosfærisk tryk er 700 hPa til 1060hPa (højde: 10.000 ft til -1.000 ft).

110 - DK PN MN236-C4 E | Brugervejledning

Vogn delnr. 20765565

Sprint / Stroller / Hi Flow Stroller
Sikkerhed
ADVARSEL: BÆRBART RF-KOMMUNIKATIONSUDSTYR (HERUNDER EKSTERNE ENHEDER SÅSOM
ANTENNEKABLER OG EKSTERNE ANTENNER) MÅ IKKE BRUGES INDEN FOR 30 CM (12 IN.) AF
NOGEN DEL AF DIN SPRINT/STROLLER/HI FLOW STROLLER, HERUNDER KABLER SOM ANGIVET AF
PRODUCENTEN. ELLERS KAN UDSTYRETS YDEEVNE BLIVE FORRINGET.
ADVARSEL: BRUG AF TILBEHØR, TRANSDUCERE ELLER KABLER, UD OVER DEM, DER ER ANGIVET ELLER
LEVERET AF PRODUCENTEN AF DETTE UDSTYR, KAN MEDFØRE FORØGET ELEKTROMAGNETISK EMISSION
ELLER REDUCERET ELEKTROMAGNETISK IMMUNITET FOR DETTE UDSTYR OG RESULTERE I UKORREKT DRIFT.
ADVARSEL: BRUG AF DETTE UDSTYR OP AD ELLER STABLET MED ANDET UDSTYR BØR UNDGÅS, DA DET KAN
MEDFØRE UKORREKT DRIFT. HVIS DET ER NØDVENDIGT AT BRUGE UDSTYRET PÅ DEN MÅDE, SKAL BÅDEDETTE OG DET ANDET UDSTYR HOLDES UNDER OPSYN FOR AT SIKRE, AT DET FUNGERER NORMALT.
Forsigtig: Elektrisk medicinsk udstyr kræver særlige foranstaltninger med hensyn til elektromagnetisk
kompatibilitet (EMC) og skal installeres og anvendes i overensstemmelse med EMC-oplysningerne i denne
vejledning.
Forsigtig: Bærbart og mobilt RF-kommunikationsudstyr (radiofrekvens) kan påvirke elektrisk medicinsk
udstyr.
Tabel 1

Vejledning og producentens erklæring – elektromagnetiske emissioner
Sprint/Stroller/Hi Flow Stroller er beregnet til brug i elektromagnetiske miljøer som angivet nedenfor. Kundeneller brugeren af Sprint/Stroller/Hi Flow Stroller bør sikre sig, at den bruges i et sådant miljø.
Emissionstest

Overensstemmelse

RF-emissioner

Gruppe 1

CISPR 11

Elektromagnetisk miljø – vejledning
Sprint/Stroller/Hi Flow Stroller bruger RF-energi
udelukkende til interne funktioner.
Derfor er RF-emissionerne meget lave og vil
sandsynligvis ikke forårsage interferens i elektronisk
udstyr i nærheden.

RF-emissioner CISPR 11

Klasse B

Harmoniske emissioner
IEC 61000-3-2

Ikke relevantSpændingsudsving/flickeremissioner
IEC 61000-3-3

Ikke relevant

Sprint/Stroller/Hi Flow Stroller er egnet til brug i alle
bygninger, såvel privatboliger som bygninger, der er
direkte forbundet til det offentlige lavspændingsnet, som
forsyner beboelsesejendomme.

PN MN236-C4 E | Brugervejledning DK - 111

Sprint / Stroller / Hi Flow Stroller
Tabel 2*: Anbefalede afstande mellem bærbart og mobilt RF-kommunikationsudstyr og Sprint/Stroller/Hi Flow
Stroller
Sprint/Stroller/Hi Flow Stroller er beregnet til brug i et elektromagnetisk miljø, hvor forstyrrelser af RF‑stråling er kontrolleret. Kundeneller brugeren af Sprint/Stroller/Hi Flow Stroller kan hjælpe til med at forhindre elektromagnetisk interferens ved at bevare en
minimumsafstand mellem bærbart og mobilt RF‑kommunikationsudstyr (sendere) og Sprint/Stroller/Hi Flow Stroller, som anbefalet
nedenfor, i henhold til kommunikationsudstyrets maksimale udgangsefffekt.
Senderens maksimale
Sikkerhedsafstand i henhold til senderens frekvens
udgangseffektm
80 MHz og 800 MHz
800 MHz til 2,5 GHz
150 kHz til 80 MHz
d = 1,2√PW
d = 1,2√P
d = 2,3√P
0,01
0,12 m0,12 m0,23 m0,10,38 m0,38 m0,73 m
1
1,2 m
1,2 m
2,3 m
10
3,8 m
3,8 m
7,3 m
100
12 m
12 m
23 m
For sendere med en maksimal udgangseffekt, der ikke er angivet ovenfor, kan den anbefalede sikkerhedsafstand (d) i meter (m)beregnes ved hjælp af den ligning, der gælder for den pågældende senders frekvens, hvor P er den maksimale udgangseffekt for
senderen i watt (W) i henhold til producenten af senderen.
NOTE 1 Ved 80 MHz og 800 MHz gælder sikkerhedsafstanden for det højere frekvensområde.
NOTE 2 Disse retningslinjer gælder muligvis ikke i alle situationer. Elektromagnetisk interferens påvirkes af absorption og refleksion
fra bygninger, genstande og mennesker.

* Denne tabel er medtaget som standardkrav for udstyr, der er testet til specifikke testniveauer og over specifikke
frekvensområder og er konstateret i overensstemmelse med bestemmelserne.

Tabel 3: Vejledning og producentens erklæring – elektromagnetisk immunitet
Sprint/Stroller/Hi Flow Stroller er beregnet til brug i elektromagnetiske miljøer som angivet nedenfor. Kunden eller brugeren af Sprint/
Stroller/Hi Flow Stroller bør sikre sig, at den bruges i et sådant miljø.
Immunitetstest
IEC 60601-testniveau
Overensstemmelsesn- Elektromagnetisk miljø – vejledningiveau
Gulvene bør være af træ, beton eller keramiske
Elektrostatisk udlad± 8 kV kontakt± 8 kV kontakt
fliser. Hvis gulvene er beklædt med syntetiskning (ESD)± 2 kV, ± 4 kV, ± 8 kV, ±± 2 kV, ± 4 kV, ± 8 kV,materiale, skal den relative fugtighed være15 luft± 15 luft
IEC 61000-4-2
mindst 30 %.**
Ikke relevant±2 kV for strøm
Udstyr, der drives af
strømledninger
Elektrisk hurtigjævnstrømtransient/burst
Ikke relevant
Ikke relevant
IEC 610004-4±1 kV for indgangs-/
udgangsledninger
Ingen dataindgangs-/
udgangsledninger
Ikke relevant
Strømstød±1 kV linje til linje
Ikke relevant
Udstyr, der drives af±2 kV linje(r) til jordIEC 61000-4-5jævnstrøm< 5 % UT (> 95 % fald i UT)Spændingsfald,
i 0,5 cyklus
korte afbrydelser og< 40 % UT (> 60 % fald i UT)
Ikke relevantspændingsvariationer i 5 cyklusser
Ikke relevantpå strømforsyningens < 70 % UT (> 30 % fald i UT) Udstyr, der drives af
jævnstrøm
indgangsledninger
i 25 cyklusser< 5 % UT (> 95 % fald i UT)
IEC 61000-4-11
i 5 sek.
Netfrekvensmagnetfelterne skal være på et
Driftsfrekvens (50/60
3A/m
3A/m
niveau svarende til niveauet på en typisk
Hz) magnetfelt
placering i et typisk kommercielt miljø eller et50/60 Hz50/60 HzIEC 61000-4-8
hospitalsmiljø.
Bemærk: UT er netspændingen før anvendelse af testniveauet.

** Denne erklæring angiver, at de påkrævede test er udført i et kontrolleret miljø, og at Sprint/Stroller/Hi Flow
Stroller er konstateret i overensstemmelse med bestemmelserne.
112 - DK PN MN236-C4 E | BrugervejledningSprint / Stroller / Hi Flow Stroller
Tabel 4

Vejledning og producentens erklæring – immunitet for ME-udstyr og ME-systemer

Vejledning og producentens erklæring – immunitet
Sprint/Stroller/Hi Flow Stroller er beregnet til brug i elektromagnetiske miljøer som angivet nedenfor. Kundeneller brugeren af Sprint/Stroller/Hi Flow Stroller bør sikre sig, at den bruges i et sådant miljø.
Immunitetstest

IEC 60601-testniveau

Overensstemmelsesniveau

Elektromagnetisk miljø – vejledning

Ledet RFIEC 61000-4-6

3 Vrms6 Vrms (i ISM-båndene)
150 kHz til 80 MHz

Ikke relevantBærbart og mobilt RF-kommunikationsudstyr måikke anvendes tættere på nogen del af Sprint/
Stroller/Hi Flow Stroller, herunder kabler, end den
anbefalede afstand beregnet ud fra den ligning, somgælder for senderens frekvens.

Batteridrevet udstyr – ingen
SIP/SOP

Anbefalet sikkerhedsafstandd = 1,2 √P
Udstrålet RF
IEC 61000-4-3

80 MHz til 2,7 GHz

10 V/m80 MHz – 2,7 GHz
80 % AM ved 1 kHz

d = 1,2 √P
d = 2,3 √Phvor P er senderens maksimale udgangseffekt i watt(W) i henhold til producenten af senderen, og d er
den anbefalede afstand i meter (m).
Feltstyrker fra faste RF-sendere som fastsat af enelektromagnetisk undersøgelsea skal være mindre
end compliance-niveauet i hvert frekvensområdeb.
Interferens kan forekomme i nærheden af​udstyrmærket med følgende symbol:

Testfrekvens(MHz)Bånda)(MHz)

Tjenestea)

Modulationb)

385

380–390TETRA 400450

430-470

GMRS 460, FRS 460

704-787

800-960

710
745
780
810
870
930
1720
1845
1970

1700-1900

2450

2400-2570

5240
5500
5785

5100-5800

Maksimaleffekt (W)Afstand(m)

Immunitetstestniveau (V/m)

Pulsmodulationb)
18 Hz
FMC)
±5 kHz afvigelse
1 kHz sinus

1,8

0,3

27

2

0,3

28

LTE-bånd 13, 17

Pulsmodulationb)
217 Hz

0,2

0,3

9

GSM 800/900, TETRA 800,
IDEN 820, CDMA 850,
LTE-bånd 5

Pulsmodulationb)
18 Hz

2

0,3

28

Pulsmodulationb)
217 Hz2

0,3

28

Pulsmodulationb)
217 Hz2

0,3

28

Pulsmodulationb)
217 Hz

0,2

0,3

9

GSM 1800, CDMA 1900,
GSM 1900, DECT,
LTE‑bånd 1, 3, 4 og 25,
UMTS
Bluetooth, WLAN,
802.11 b/g/n, RFID 2450,
LTE-bånd 7
WLAN 802.11 a/nBemærk: Hvis det er nødvendigt for at opnå IMMUNITETSTESTNIVEAUET, kan afstanden mellem senderantennen og
Sprint/Stroller/Hi Flow Stroller nedsættes til 1 m. Testafstanden på 1 m er tilladt, jævnfør IEC 61000-4-3.
abc

For visse tjenester er kun uplink-frekvenserne medtaget.
Bærebølgen skal moduleres med et firkantsignal med 50 % pulsvarighed.
Som alternativ til frekvensmodulation kan der bruges 50 % pulsmodulation på 18 Hz, fordi selvom denne ikke repræsenterer en aktuel
modulation, vil det være det værste tilfælde.

CAIRE og CAIRE Inc. er registrerede varemærker tilhørende CAIRE Inc. Besøg vores websted herunder for en fuld liste over varemærker. Varemærker: www.careinc.com/corporate/
trademarks.
Copyright © 2023 CAIRE Inc. CAIRE Inc. forbeholder sig retten til at lade deres produkter udgå eller ændre priserne, materialerne, udstyret, kvaliteten, beskrivelserne, specifikationerne
og/eller processerne for deres produkter til enhver tid uden forudgående varsel og uden yderligere forpligtelser eller konsekvens. Alle rettigheder, der ikke udtrykkeligt er angivet heri, er
forbeholdt os, hvor det er relevant.

PN MN236-C4 E | Brugervejledning DK - 113

Sprint / Stroller / Hi Flow Stroller
Symbolbibliotek
ISO 7000

Røyking forbudt nær enheten eller mens
enheten brukes. Reg.nr. P002

Må holdes unna regn – oppbevares tørt.
Reg. nr. 0626

Anvendt del type BF (grad av
beskyttelse mot elektrisk støt).
Reg. nr. 5333

Stabelgrense etter antall. Reg. # 2403
Produsentens navn og adresse.
Reg. nr. 3082

Advarsel. Reg.nr. W001
Rådsdirektiv 93/42/EØS
Autorisert representant i EU

Landet og produksjonsdatoen.
"CC" identifiserer landskoden på
to bokstaver for produksjonslandet.
Produksjonsdatoen er i formatet ÅÅÅÅMM-DD. Reg. #6049
Forsiktig – se dokumentene som
medfølger. Reg. nr. 0434A

Hvis produktets unikeenhetsidentifikator (UDI) -etikett har CE#### -symbolet på, oppfyller enheten
kravene i direktiv 93/42 / EØF om
medisinsk utstyr. CE #### symboletindikerer varslet organ nummer.
Denne enheten overholder kravene
i direktiv 2010/35/EF vedrørende
medisinsk utstyr. Pi-merket som vist.

Katalognummer. Reg. nr. 2493
Serienummer. Reg. nr. 2498
Lagring eller driftstemperaturområde.
Reg. nr. 0632

ADR
Ikke giftig gass.

Oppbevaring fuktighetsområde. Reg.
nr. 2620

Fare Oksiderende stoffer:brannforsterkende risiko.
Oksidasjonsmidler får branner til åbrenne kraftigere.
Nedkjølt væske, USP; produsert ved
luftkondensasjon

Atmosfærisk trykkbegrensning. Reg.
nr. 2621
Denne siden opp. Reg. nr. 0623

Ytterligere symboler

Glass – håndteres med forsiktighet. Reg.
nr. 0621

Hold enheten godt ventilert til enhver tid

Inneholder farlige stoffer. Reg. #3723

Må holdes unna brennbare materialer,olje og smørefett.

Importør. Reg. #3725

Autorisert representant i Sveits.

ISO 7010
Frostskader kan oppstå ved kontakt med
kalde væsker eller oksygengass, eller
frosne deler. Advarsel lav temperatur /frostfare. Varsler om lav temperatur ellerfrysing. Reg.nr. W010
Instruksjonsmanualen må leses.
Reg. nr. M002
Holdes unna åpen flamme, brann,
gnister. Åpne tennkilder og røyking
forbudt. Reg. nr. P003

114 - NO PN MN236-C4 E | Bruksanvisning

IEC 60417

Hvis enheten bærer UKCA-merket som
vist med UKCA#### som indikerer det
varslede organets nummer, er denne
enheten i samsvar med UKCA-forskriftene.
Ikke dekk til enheten eller bær denbærbare enheten under klærne. Disse
enhetene lufter normalt ut oksygen. nr.
5641

Sprint / Stroller / Hi Flow Stroller
21 CFR 801.15
Amerikansk føderal lovgivningbegrenser denne enheten til salg fra enlege eller foreskrevet av en lege.
IEC 60601-1
Dryppsikker
Rådsdirektiv 2012/19/EF
WEEE
Dette symbolet er for å minne eierne av
utstyret om å returnere det til et gjenvinningsanlegg ved slutten av levetiden, ihenhold til direktivet om avfall av elektrisk og elektronisk utstyr (WEEE).
Produktene våre vil være i samsvar med
direktivet om restriksjoner av farlige
stoffer (RoHS). De vil ikke inneholde
mer enn spormengder av bly eller annet
innhold av farlig materiale.

ISO 15223-1
Medisinsk enhet. Reg. # 5.7.7
Unik enhetsidentifikator # 5.7.10
Dette produktet kan være dekket av ett eller flereamerikanske eller internasjonale patenter. Gå til vår
hjemmeside nedenfor for en liste med gjeldende
patenter. Pat.: www.caireinc.com/corporate/patents/.

Spesifikasjoner• Driftsmodus: Kontinuerlig oksygentilførsel
• Type beskyttelse mot elektrisk sjokk: Selvdrevet utstyr• Grad av beskyttelse mot elektrisk sjokk: Type BF anvendt del• Klassifisering i henhold til grad av beskyttelse mot skadelig vanninntrengning: IP22 – ordinært utstyr• Utstyret er ikke passende til bruk sammen med brennbare anestesimidler
Produktspesifikasjoner

SprintStroller

Hi Flow Stroller

LOX-kapasitet

0,68 kg (1,5 pund)

1,36 kg (3,0 pund)

1,36 kg (3,0 pund)Gassmessig tilsvarendekapasitet
Vekt, tom

513 L

1026 L1025 L

2,04 kg (4,5 pund)2,27 kg (5,0 pund)2,49 kg (5,5 pund)

Vekt, fylt

2,72 kg (6,0 pund)

3,63 kg (8,0 pund)

3,86 kg (8,5 pund)Høyde

298 mm (11,75 tommer)

343 mm (13,5 tommer)

343 mm (13,5 tommer)

Dimensjoner

123,83 mm D x 191 mm B

149,23 mm D x 191 mm B

149 mm D x 191 mm B

Typisk brukstid ved 2 LPM

4,3 timer

8,5 timer

8,5 timer

Driftstrykk

20 kPa (137 psi)

20 kPa (137 psi)

20 kPa (137 psi)

Normal fordampningsrate

0,57 kg/dag (1,3 pund/dag)

0,57 kg/dag (1,3 pund/dag)

0,57 kg/dag (1,3 pund/dag)Standard flytreguleringsområde
Nøyaktighet av
strømningshastighet*

Av, 0,25; 0,5; 0,75; 1; 1,5; 2;
2,5; 3; 4; 5; 6 LPM+/–10 %

Av, 0,25; 0,5; 0,75; 1; 1,5; 2;
2,5; 3; 4; 5; 6 LPM+/–10 %

Av, 0,5, 1, 2, 2,5, 3, 4, 6, 8,
10, 12, 15 LPM+/–10 %* Denne nøyaktigheten forekommer bare ved 70 F (21,1 °C) og 14,7 psig (1,01 bar) og med en kalibrert nøyaktig massestrømmåler.

PN MN236-C4 E | Bruksanvisning NO - 115

Sprint / Stroller / Hi Flow Stroller
Varselinformasjon
Viktig: Les denne bruksanvisningen nøye før dubruker Sprint / Stroller / Hi Flow Stroller.
Fås kun på resept.
ADVARSEL: ENHETEN ER IKKE BEREGNETFOR LIVSBEVARENDE BRUK.
ADVARSEL: HVIS DU MENER AT UTSTYRETIKKE FUNGERER SOM DET SKAL, KONTAKTHELSEPERSONELL. IKKE PRØV Å REPARERE ELLERJUSTERE ENHETEN SELV.
ADVARSEL: DETTE UTSTYRET MÅ IKKE ENDRESUTEN AUTORISASJON FRA PRODUSENTEN.
ADVARSEL: HVIS KONTINUERT OKSYGENFORSYNINGER NØDVENDIG, SØRG FOR AT TILSTREKKELIG
TILFØRSEL AV OKSYGEN OG/ELLER SEKUNDÆROKSYGENFORSYNING ALLTID ER TILGJENGELIGUNDER BEHANDLINGEN.
ADVARSEL: IKKE TILLAT RØYKING, STEARINLYS
ELLER ÅPEN ILD INNEN 3 METER (10 FOT) FRAENHETEN, ELLER NÆRMERE ENN 20 CM (8 TOMMER)FRA NOEN ANTENNELSESKILDE.
ADVARSEL: HOLD ENHETEN I ET GODT VENTILERTOMRÅDE.
ADVARSEL: IKKE OPPBEVAR UTSTYR MED
FLYTENDE OKSYGEN I SKAP, BAGASJEROM
ELLER ANDRE BEGRENSEDE OMRÅDER.
IKKE LEGG TEPPER, DRAPERIER ELLER
ANDRE STOFFER OVER UTSTYRET.
ADVARSEL: DETTE PRODUKTET KAN EKSPONEREDEG FOR KJEMIKALIER, INKLUDERT NIKKEL, SOMIFØLGE STATEN CALIFORNIA KAN FORÅRSAKEKREFT. FOR MER INFORMASJON, GÅ TIL
WWW.P65WARNINGS.CA.GOV.
ADVARSEL: I TILFELLE DET OPPSTÅR EN ALVORLIGHENDELSE MED DENNE ENHETEN, SKAL BRUKEREN UMIDDELBART RAPPORTERE HENDELSEN
TIL LEVERANDØREN OG / ELLER PRODUSENTEN.
EN ALVORLIG HENDELSE ER DEFINERT SOM ENSKADE, DØD ELLER POTENSIELL FOR Å FORÅRSAKESKADE / DØD DERSOM DET SKULLE OPPSTÅ EN NY
HENDELSE. BRUKEREN KAN OGSÅ RAPPORTERE
HENDELSEN TIL DEN KOMPETENTE MYNDIGHETEN I
LANDET DER HENDELSEN SKJEDDE.
Forsiktig: Sprint / Stroller / Hi Flow Stroller
skal bare brukes som anvist av legen.
Forsiktig: Amerikansk føderal lovgivning begrenser
denne enheten til salg fra en lege eller foreskrevet av
en lege.
Forsiktig: Enheten inneholder flytende oksygen som
er ekstremt kaldt, nesten –184 °C (–300 °F). Hvis duutsettes for slike lave temperaturer, kan det forårsake
alvorlige frostskader.

116 - NO PN MN236-C4 E | Bruksanvisning

Forsiktig: Oksygen, enten flytende eller som gass, gjør
at andre materialer brenner raskere, selv om oksygenet
selv ikke er brennbart. Denne faren, sammen med denlave temperaturen på flytende oksygen, gjør det nødvendig med visse sikkerhetsforanstaltninger.
Forsiktig: Hold brennbare materialer vekk fra dette
utstyret. Aerosolsprayer, oljer og fett, inkludert
ansiktskremer og vaselin, antennes lett og kan brenneraskt når det er oksygen til stede.
Forsiktig: Dersom man røyker samtidig som man bruker
en oksygenkanyle, kan det føre til brannskade i ansiktetog mulig død.
Hvis kanylen blir tatt ut og lagt på klær, sofaer eller
andre myke materialer når den tas ut, vil det kunneforårsake øyeblikkelig brann hvis det utsettes for ensigarett, varmekilde, gnist åpen ild.
Hvis du røyker, bør du: (1) slå av den bærbare enheten,(2) ta av kanylen og (3) forlate rommet enheten befinner
seg i.
Forsiktig: Hvis den tippes over ved et uhell, sett denomgående, men forsiktig, tilbake til stående stilling hvis
mulig. Hvis noe flytende oksygen slippes ut, forlate området umiddelbart og kontakt helsepersonell. Ikke forsøk
å flytte enheten eller hindre det flytende oksygenet i åslippe ut.
Merk: Ikke berør noen deler med rim på, uansett
enhet.
Merk: Ikke oppbevar eller betjen den bærbare enheten
som er koblet til den stasjonære enheten.
Merk: Ikke la personer uten opplæring håndtere eller
bruke denne enheten.
Merk: De føderale luftfartsmyndighetene forbyr brukav denne enheten i frakt- og passasjerfly.

Tiltenkt bruk

CAIRE Sprint/Stroller/Hi Flow Stroller er beregnet
på administrering av tilleggsstoff. Enheten er ikke
beregnet på livsstøtte, og gir heller ikke noen evne til å
overvåke pasienten.

Sprint / Stroller / Hi Flow Stroller
Innledning

Kontroller

De bærbare Sprint-, Stroller- og Hi Flow Strollerenhetene er beregnet på å administrere supplerende
oksygen til pasienten i sluttbrukerens hjem. De kanogså brukes i institusjoner slik som pleiehjem eller ikkekritiske helsefasiliteter. Sprint / Stroller / Hi Flow Stroller
sørger for en flyttbar oksygenkilde over lengre tid.

1. Elektronisk væskenivåmåler
2. Elektronisk væskenivåbryter
3. Mekanisk væskenivåmåler
4. Strømmingsregulering
5. Kontakt for oksygenslange
6. Lufteventil
7. Fyllekontakter (se neste side for
QDV‑identifikasjon)
8. Bærestropp med skulderpute
9. Kondenseringspute
10. Kondenseringskopp

Enheten brukes av KOLS-pasienter eller de med
redusert pustekapasitet. Enheten foreskrives til
pasienten. Enheten selges til en leverandør som harfått opplæring i å betjene og utføre service på debærbare Sprint- / Stroller- / Hi Flow Stroller-enhetene.
Leverandøren lærer opp brukeren.
Enheten er ikke beregnet for å værelivsopprettholdende, og har heller ikke mulighet til å
overvåke en pasient. Det anbefales å ha en alternativ
oksygenkilde i tilfelle mekanisk feil. Disse bærbare
enhetene har også en hurtigfrakoblingsventil som
kan kobles til en LOX-reservoarenhet for transfylling
av den bærbare enheten. Reservoarenheten fylles av
helsepersonell.

1

Denne brukerhåndboken inneholder instruksjonene
for bruk av Sprint / Stroller / Hi Flow Stroller. Se
brukerhåndboken som fulgte med den stasjonære
enheten, for informasjon om drift.

8

2
4
5

Merk: Tjenesteleverandøren vil bistå med det første oppsettet og gi instrukser om håndtering og bruk av enheten.

57
Sprint-, Stroller, og Hi Flow Stroller-enheter

6

10
9

PN MN236-C4 E | Bruksanvisning NO - 117

Sprint / Stroller / Hi Flow Stroller
QDV-identifikasjon(hurtigfrakoblingsventil)
Kontakt forbærbar enhet

Stasjonær
kontakt

3. Plasser fyllekontakten i opprettstående posisjon
over den stasjonære enhetens kontakt.
QDV-type

CAIREsidefyllingSFlåsende

LIFE-OX
TFlåsende

4. a. Hvis kontakten er av skyv-på-typen:
Senk Sprint- / Stroller- / Hi Flow Stroller-enheten
på den stasjonære enheten til du føler at kontakten
går i inngrep.

PENOX

TAEMA
TF
Push‑OnPB

b. Hvis kontakten er av typen Penox eller Life-Ox:
Roter Sprint- / Stroller- / Hi Flow Strollerenheten til du føler at kontakten går i inngrep.
Roter deretter forsiktig og bestemt Sprint /Stroller / Hi Flow Stroller med klokken ()
mens du presser nedover.

Instruksjoner for brukSlik fyller du Sprint / Stroller / Hi Flow
Stroller med toppåfylling
1. Rengjør fyllekontaktene på både den stasjonære
enheten og Stroller / Spirit / Hi Flow Stroller med
en ren, tørr lofri klut.
Forsiktig: Fyllekontaktene må være tørre,fordi fuktighet kan føre til at Sprint / Stroller/ Hi Flow Stroller og den stasjonære enhet
fryser sammen.
2. Sørg for at strømningskontrollknottene på bådeSprint / Stroller / Hi Flow Stroller og reservoaret er
i av-posisjonen (0).

118 - NO PN MN236-C4 E | Bruksanvisning

Merk: Fortsetter i trinn 6 under Slik fyller du Sprint / Stroller/ Hi Flow Stroller med sidepåfylling.

Slik fyller du Sprint / Stroller / Hi Flow
Stroller med sidepåfylling.
1. Rengjør fyllekontaktene på både den stasjonære
enheten og Sprint / Stroller / Hi Flow Stroller med
en ren, tørr lofri klut.
ADVARSEL: FYLLEKONTAKTENE MÅ VÆRETØRRE, FORDI FUKTIGHET KAN FØRE TIL ATSPRINT / STROLLER / HI FLOW STROLLER
OG DEN STASJONÆRE ENHET FRYSERSAMMEN.

Sprint / Stroller / Hi Flow Stroller
2. Sørg for at strømningskontrollknottene på bådeSprint / Stroller / Hi Flow Stroller og reservoaret er i
av-posisjonen (0).

Merk: Avlesninger på væskenivåmåleren er mest
nøyaktige 30 sekunder eller mer etter at Sprint / Stroller /Hi Flow Stroller har blitt fylt.

3. Sørg for at Sprint- / Stroller- / Hi Flow Strollerfyllekontakten er i opprettstående posisjon i forhold
til den stasjonære kontakten.

4. Roter Sprint / Stroller / Hi Flow Stroller mot klokken() til du føler at pinnen og sporet går i inngrep.
7. Du kan også se damp rundt forbindelsen. Detteer normalt og skyldes den store forskjellen itemperatur mellom det flytende oksygenet og den
varme luften som omgir enhetene.
8. Når enheten er full, lukker du lufteventilen. Du
merker at enheten er full, når hveselydens tone endres
og noe væske kommer ut av ventilasjonsslangen.
Merk: Ved lukking av ventilen når en nesten kontinuerligvæskestrøm slipper ut, fylles Sprint / Stroller / Hi Flow
Stroller mest fullstendig, men noe flytende oksygen går
til spille.
5. Roter Sprint / Stroller / Hi Flow Stroller forsiktig
og bestemt tilbake til opprettstående posisjon. Nå er
enhetene låst sammen.

Merk: Fortsetter i trinn 6.
Merk: For kontakt av skyv-på-typen holder du enheten påplass gjennom hele fyllingen.

Merk: Ved lukking av ventilen ved første tegn påvæskeutslipp, fylles ikke Sprint / Stroller / Hi Flow Stroller
like fullstendig, men det sparer flytende oksygen.
9. For kontaktene av skyv-på-typen skyver duutløserknappen ned til enhetene skilles. For alle
andre kontakter roterer du enheten til den skilles fra
den stasjonære enheten.

Utløserknapp for bærbar enhet (bare skyv-på-typen)

6. Åpne lufteventilen for å begynne påfylling av
Sprint / Stroller / Hi Flow Stroller. Du skal høre
at noe oksygen slipper ut, så ikke bekymre deg på
grunn av det. Hveselyden du hører, er helt normal.

PN MN236-C4 E | Bruksanvisning NO - 119

Sprint / Stroller / Hi Flow Stroller
Forsiktig: Kontroller væskenivåmåleren først
etter at ventilen har blitt lukket.

Instruksjoner for brukForsiktig: Hvis Sprint / Stroller / Hi Flow Stroller fylles
på umiddelbart etter en bruksperiode, fyller den opp
og venter deretter i 10 minutter etter påfylling med
strømningsstyringsventilen (FCV) satt til null for at
trykket skal stabilisere seg. Bruk deretter som normalt.
ADVARSEL: IKKE BRUK SPRINT / STROLLER/ HI FLOW STROLLER NÅR DEN ER KOBLETTIL DEN STASJONÆRE ENHETEN.
ADVARSEL: HVIS SPRINT / STROLLER / HI FLOWSTROLLER IKKE LØSNER LETT, MÅ DU IKKE BRUKEKRAFT. ENHETENE KAN VÆRE FROSSET SAMMEN. LA
ENHETENE FORBLI SAMMENKOBLET OG VENT TIL DE
VARMES OPP – DA ER DET ENKLERE Å TA DEM FRA
HVERANDRE. IKKE BERØR DELER MED FROST PÅ.
ADVARSEL: HVIS LEKKASJEN BLIR FOR STOR, SLIKAT DET FAKTISK STRØMMER UT VÆSKE, FORLATERDU OMRÅDET OG KONTAKTER HELSEPERSONELLUMIDDELBART.
ADVARSEL: FLYTENDE OKSYGEN ELLER KALDTOKSYGEN I GASSFORM KAN FORFRYSE HUDEN.
OKSYGEN FØRER TIL ØKT BRANNFARE.
Forsiktig: Skulle det oppstå væskelekkasje fra
den stasjonære enheten eller Sprint / Stroller/ Hi Flow Stroller som skiller enhetene, setter
du Sprint / Stroller / Hi Flow Stroller til sideog sikrer at den forblir i vertikal posisjon.
Forlat så rommet og kontakt helsepersonellet
umiddelbart.

Merk: Til utstyrsleverandør: Følgende tilbehør foroksygenadministrasjon anbefales for bruk sammen
med Sprint / Stroller / Hi Flow Stroller:• Nasal kanyle med 2,1 m (2 fot) rør (Maksimal LPM: 6
LPM): CAIRE-delenummer 5408-SEQ• Brannbane: CAIRE-delenummer 21126636
Det anbefales brannbrytning for bruk med eventuell
kanyle.
• CAIRE tilbyr en brannbrytning beregnet på å brukes
i forbindelse med den bærbare oksygen. Brannbrynet
er en termisk sikring for å stoppe strømmen av gass itilfelle at nedstrøms kanyle eller oksygenrør tennes og
brenner for brannknappen. Den plasseres på linje med
nesekanylen eller oksygenrøret mellom pasienten og
oksygenutløpet til Sprint / Stroller / Hi Flow Stroller. For
riktig bruk av brannkjøringen, se alltid produsentens
instruksjoner (følger med hvert brannkjøresett).
• Ytterligere anbefalt tilbehørsinformasjon er tilgjengelig
online på www.caireinc.com.

120 - NO PN MN236-C4 E | Bruksanvisning(Kanyle er ikke inkludert.)

1. Skyv kanylepusteslangen bestemt påoksygenslangekontakten.
2. Juster kanylepusteslangen til riktig posisjon slik at
du kan puste oksygenet komfortabelt.
3. Drei strømreguleringsknappen på toppen av Sprint /Stroller / Hi Flow Stroller med urviseren til den
foreskrevne strømningshastigheten (numerisk tegn)
er synlig i knappen «window» (vindu) og du føler etpositivt hakk.
Forsiktig: Oksygenstrøm utenfor
spesifikasjonene vil oppstå hvis
strømningsreguleringsknappen er innstilt
mellom strømningshastigheter.
4. Du bør nå motta oksygen. Hvis du vil stoppe
oksygenstrømmen, dreier du strømstyringsknotten
mot urviseren til av-posisjonen (0).
Forsiktig: Slå alltid
strømmingsreguleringsknotten av (0) når den
ikke er i bruk.

Sprint / Stroller / Hi Flow Stroller
5. Bruk følgende diagram som veiledning for å
bestemme hvor lenge Sprint / Stroller / Hi Flow
Stroller vil fungere:
FCV-innstilling (strømningsstyringsventil)
ModellSprintStroller

Hi Flow
Stroller

NominellNominell

FCVpos.

SprintStroller

Hi Flow
Stroller

Av

30:00

60:00

71:00

0,25

18:00

30:00

NA

0,5

12:12

23:00

32:00

0,75

9:18

17:24

NA

1

7:30

14:06

16:00

1,5

5:30

10:12

NA

2

4:18

8:00

8:00

2,5

3:36

6:36

6:20

3

3:06

5:36

5:20

4

2:24

4:24

4:00

5

2:00

3:36

NA

6

1:42

3:00

2:40

8

N/A

N/A

2:00
1:40

10

N/A

N/A

12

N/A

N/A

1:20

15

N/A

N/A

1:00

Merk: Tidene oppgis i timer og minutter(i formatet 00:00).
Merk: «Nominelle» tider er for ideelle forhold.
Det vil si maksimal påfylling, eksakte
strømningshastigheter, god tapshastighet, en enhet
som ikke flyttes, osv.
Disse tidene er de maksimale forventede verdiene.
Merk: Individuelle resultater vil variere.

• Gen 4-måler: Trykk på den grønne
betjeningsknappen på forsiden av måleren. Les av
LED-lampebuen, som indikerer innholdsnivå.

Forsiktig: Sprint / Stroller / Hi Flow Stroller er
tom hvis bare det siste segmentet av lysfeltet
lyser.
Hvis indikatoren for lavt batterinivå lysernår knappen er trykket inn, må du varsle
helsepersonellet.
Merk: Sprint / Stroller / Hi Flow Stroller vil fortsette å levere
oksygen selv om indikatoren for lavt batterinivå tennes, sålenge det er flytende oksygen i enheten.
6b. For å kontrollere nivået av flytende oksygen ienheten med den mekaniske væskenivåmåleren:• Støtt enheten med fabrikkstroppen direkte over det
svarte kammerets indikatorfjærskala for innhold.
• La enheten stabilisere seg (dvs. ikke bevege seg
opp og ned), og les deretter den eksponerte fargede
skalaen for omtrentlig væskeinnhold.

6a. Slik verifiserer du nivået av flytende oksygen ienheten med den elektroniske væskenivåmåleren:• Gen 3-måler: Trykk på trykknappen(væskenivåbryteren) på toppen av enheten i minst to
sekunder. Les over toppen av lysfeltet for å indikere
innholdsnivå.

PN MN236-C4 E | Bruksanvisning NO - 121

Sprint / Stroller / Hi Flow Stroller
Merk: Hvis enheten er tom, vil bare skalaens rødfargedebånd bli eksponert. Hvis dette er tilfelle, må enheten fylles
før bruk.
7. Under visse miljømessige betingelser med
kontinuerlig bruk, og spesielt hvis den bærbare
enheten ikke flyttes, kan den utvikle overdreven
frost rundt fordamperspolen og på huset. Du kan
redusere denne frosten ved å banke lett på enheten
og/eller tørke eventuell akkumulert frost av huset.

Vedlikehold av kondensatputeNår enheten har blitt tom og er varmet opp til
romtemperatur, fjerner du eventuell fuktighet fra
kondenseringskoppen:
For hardt hus• Vipp enheten til én side. Ta tak i knotten i midten
av kondenseringskoppen (nederst på enheten) og
trekk den rett ut. Dette frigjør koppen.
Merk: Hvis puten er svært tilsmusset, kan den vaskes(med blekemiddel for å desinfisere).
• Hvis du vil bytte ut koppen, sentrerer du den
over den runde åpningen nederst på enheten
og trykker inn knotten i ett klikk for å aktivere
koppen. Deretter trykker du med et klikk til forå låse koppen på plass.

• Fjern puten fra koppen og vri ut eventuell
absorbert fuktighet. La puten tørke helt før dusetter den tilbake i koppen.
Merk: Hvis puten er svært tilsmusset, kan den vaskes(med blekemiddel for å desinfisere).
• Når du skal få på plass igjen puten, setter du den
ganske enkelt inn i bunndekselet og lukker igjen
bunnglidelåsen.
Merk: Hvis knotten ikke lar seg skyves inn og ikke aktiverer
koppen, holder du koppen og trekker ut knotten til den
klikker inn i åpen posisjon. Bytt nå ut koppen som tidligere
beskrevet.

Frostreduksjon
Under visse miljømessige betingelser med kontinuerlig
bruk, og spesielt hvis den bærbare enheten ikke
flyttes, kan den utvikle overdreven frost rundt
fordamperspolen og på huset. Du kan redusere denne
frosten ved å banke lett på enheten og/eller tørke
eventuell akkumulert frost av huset.
ADVARSEL: PÅ GRUNN AV MULIGHETENFOR OVERDREVEN FUKTIGHETS-/ISOPPSAMLING ANBEFALES DET AT HI-FLOW
STROLLER FORBLIR I TOMGANGSMODUS
I 1 TIME ETTER BRUK FOR Å REDUSEREFUKTOPPSAMLING.

Låst
I inngrep
ÅpenKnottkopp

Merk: Hvis knotten ikke lar seg skyves inn og ikke aktiverer
koppen, holder du koppen og trekker ut knotten til den
klikker inn i åpen posisjon. Bytt nå ut koppen som tidligere
beskrevet.
For mykt hus• Åpne den nederste glidelåsen og trekk tilbake
bunndekselet.

122 - NO PN MN236-C4 E | BruksanvisningPå grunn av de høyere strømningshastighetene som
er forbundet med Hi Flow Stroller, vil det oppstå
kondens og noe frost på enheten, spesielt ved
strømningsinnstillinger på 10 LPM og høyere. Det
anbefales at enheten etterlates i minst 1 time med
inaktivitet etter bruk for å redusere muligheten for
overdreven oppsamling på grunn av kontinuerlig bruk.

Sprint / Stroller / Hi Flow Stroller
Feilsøking
Følgende informasjon er ment å hjelpe deg med å feilsøke og løse enkle driftsproblemer som du kan oppleve ved
bruk av Sprint / Stroller / Hi Flow Stroller.
ProblemLøsning
Portable gir fra seg en• Hvesingen kan skje for å opprettholde riktig arbeidstrykk i enheten. Det er mest
hvesende lyd.
sannsynlig at den hveser etter fylling eller når den bærbare enheten endrer stilling.
Hvesingen kan vare i cirka 10 minutter etter påfylling. Dessuten kan den oppstå når
strømningsreguleringsventilen står på en lav innstilling.

Den bærbare enhetens
strømning stanser underbruk.

Den bærbare enheten kan
ikke fylles.
Den bærbare enhetenstømmeventil ble ikke lukket
ordentlig på slutten av
fyllingen.

• Hvis den bærbare enheten har blitt lagt i feil posisjon, setter du den bærbare enheten
tilbake i opprettstående posisjon og lar det gå flere minutter til enheten stabiliseres• Kontroller at kanylen er godt festet til oksygenutløpet.
• Kontroller at kanylen ikke har en knekk.
• Kontroller innholdsindikatoren/nivåmåleren og fyll om nødvendig den bærbare enheten.
• Kontroller at strømstyringsknotten ikke er i av-posisjonen («0»).
• Kontroller at det er oksygen i tanken.
• Kontroller at den bærbare enhetens og tankens påfyllingskontakter er helt sammenkoblet
gjennom fylleprosessen.
• Hvis tømmeventilen ikke kan lukkes og hveselyden og oksygenrøyken fortsetter, fjern
den bærbare enheten forsiktig ved å trykke på utløserknappen på tanken. Utluftingen fra
bunnen av den bærbare enheten vil stoppe etter et par minutter. La enheten bli varm før
du kan lukke tømmeventilen. Den bærbare enheten kan kreve så mye som 60 minutterfor å gjenopprette tilstrekkelig trykk for nøyaktig oksygenstrømming. Om nødvendig, bruken alternativ kilde til oksygen slik som en strømningsreguleringsventil festet til tanken.
• Den bærbare enhetens og tankens påfyllingskontakter kan ha frosset.

Den bærbare enheten kan
ikke enkelt kobles fra tanken• IKKE BRUK KRAFT. La det gå noen minutter slik at frosne deler blir varme, løsne deretter
etter fylling.
den bærbare enheten når isen har smeltet. For å hindre at enhetene fryser sammen,
tørk alltid hannkontakten for påfylling på tanken og hunnkontakten for påfylling på denbærbare enheten med en ren, tørr klut før du fyller.

PN MN236-C4 E | Bruksanvisning NO - 123

Sprint / Stroller / Hi Flow Stroller
Rengjøringsstandard
ADVARSEL: RENGJØR BARE ETTER ATENHETEN TOM.
• Rengjør med en løsning av mildt oppvaskmiddel
og vann.
• Bruk rengjøringsmiddelet direkte på en lofri
klut. Godkjente rengjøringsmidler inkluderer
HydroPure og HydroKlean. Ikke spray
rengjøringsmidler direkte på Sprint / Stroller /Hi Flow Stroller.
• Tørk overflaten med den lofrie kluten til den
ytre overflaten er ren.
Forsiktig: Ikke bruk høytemperatur- og
høytrykksvaskutstyr for å rengjøre disse
enhetene.
• Ikke få rengjøringsmiddelet på eventuelle interne
komponenter eller ventiler.
• La enheten tørke grundig før bruk.
Merk: Merknad til helsepersonell – forreprosesseringsprosedyrer, se gjeldende servicehåndbok.

WEEE og RoHS

Vedlikehold
Tjenesteleverandøren er ansvarlig for alt vedlikehold
som kreves i henhold til den tekniske håndboken
for denne enheten. Ring tjenesteleverandøren foreventuelle vedlikeholdskrav.
Kondensatputen er den eneste delen som brukeren kan
utføre service på. Brukeren skal ikke utføre service pånoen andre deler.
Forventet levetid er minst fem år.
Rengjør påfyllingskontaktene på både stasjonære ogbærbare enheter med en ren, tørr, lofri klut mellom
hver påfylling for å forhindre frysing og mulige feil på
utstyret.
Merk: Eventuelt ekstra vedlikehold som er nødvendig,
må utføres av en kvalifisert servicetekniker eller
tjenesteleverandøren.

Tilbehør

Dette symbolet skal påminne eierne av
utstyret om at utstyret skal leveres til etgjenvinningsanlegg når brukstiden er omme,
i henhold til direktiv om avfall fra elektrisk og
elektronisk utstyr (WEEE-direktivet).
Produktene våre skal være i samsvar med begrensningav farlige stoffer (RoHS-direktivet). De vil ikke
inneholde mer enn spormengder av bly eller andre
farlige materialer.

Avhending
Returner alltid Sprint / Stroller / Hi Flow
Stroller, inkludert alle komponenter, tilhjemmepleieleverandøren for riktig kassering. Du kanogså kontakte lokale myndigheter for anvisning om
korrekt deponering av batteriet.

Transport og lagring
Enheten skal oppbevares i stående stilling og
godt ventilert. Ikke la enheten ligge på siden.
Ikke-kondenserende fuktighet opp til 95 %.
Temperaturområde fra -40 °C til 70 °C (-40 °F til
158 °F).
Driftstemperaturen ligger i området -10 til 40 °C (14 til
104 °F). Ikke-kondenserende luftfuktighet fra 30 %til 75 %.
Merk: Det atmosfæriske trykket spenner fra 700 hPa til
1060 hPa (høyde fra 10 000 fot til -1000 fot).

124 - NO PN MN236-C4 E | BruksanvisningBærevogn, delenr. 20765565

Sprint / Stroller / Hi Flow Stroller
Sikkerhet
ADVARSEL: BÆRBART RADIOKOMMUNIKASJONSUTSTYR (INKLUDERT EKSTERNE ENHETER SOM FOR
EKSEMPEL ANTENNEKABLER OG EKSTERNE ANTENNER) BØR IKKE BRUKES NÆRMERE ENN 30 CM(12 TOMMER) TIL ALLE DELER AV SPRINT / STROLLER / HI FLOW STROLLER, INKLUDERT KABLER
SPESIFISERT AV PRODUSENTEN. ELLERS KAN UTSTYRETS YTELSE BLI REDUSERT.
ADVARSEL: BRUK AV TILBEHØR, OMFORMERE OG KABLER SOM IKKE ER SPESIFISERT ELLER LEVERT AV
PRODUSENTEN AV DETTE UTSTYRET, KAN FØRE TIL ØKTE ELEKTROMAGNETISKE UTSLIPP ELLER REDUSERTELEKTROMAGNETISK IMMUNITET FOR DETTE UTSTYRET OG FØRE TIL FEIL DRIFT.
ADVARSEL: BRUK AV DETTE UTSTYRET VED SIDEN AV ELLER OPPÅ ANNET UTSTYR BØR UNNGÅS FORDI
DET KAN FØRE TIL FEILFUNKSJON. HVIS SLIK BRUK ER NØDVENDIG, BØR UTSTYRET OG ANNET UTSTYROBSERVERES FOR Å KONTROLLERE AT DE FUNGERER SOM DE SKAL.
Forsiktig: Medisinsk elektrisk utstyr krever spesielle forholdsregler med hensyn til EMC (elektromagnetisk
samsvar) og må installeres og tas i drift i henhold til EMC-informasjonen som gis i denne håndboken.
Forsiktig: Bærbart og mobilt RF-kommunikasjonsutstyr (radiofrekvens) kan påvirke medisinsk elektrisk
utstyr.
Tabell 1

Veiledning og produsenterklæring – elektromagnetisk stråling
Sprint / Stroller / Hi Flow Stroller er beregnet på bruk i det elektromagnetiske miljøet som er spesifisert nedenfor.
Kunden eller brukeren av Sprint / Stroller / Hi Flow Stroller må sikre at enheten brukes i et slikt miljø.
Emisjonstest

SamsvarElektromagnetisk miljø – veiledning

RF-utslipp

Gruppe 1

Sprint / Stroller / Hi Flow Stroller bruker bare RF-energi forintern funksjonalitet.

CISPR 11

Derfor er RF-utslippet meget lavt, og det er ikke sannsynlig
at det fører til forstyrrelser for elektronisk utstyr i nærheten.

RF-emisjoner CISPR 11

Klasse B

Harmoniske emisjoner
IEC 61000-3-2

Ikke relevant

Spenningssvingninger/
flimmerutstråling
IEC 61000-3-3

Ikke relevant

Sprint / Stroller / Hi Flow Stroller er egnet for bruk i alle
typer bygninger, inklusive bosteder og de som er direktekoblet til offentlig lavspent forsyningsnettverk, som gir strøm
til bygninger som brukes for boformål.

PN MN236-C4 E | Bruksanvisning NO - 125

Sprint / Stroller / Hi Flow Stroller
Tabell 2*: Anbefalt separasjonsavstand mellom bærbart og mobilt RF-kommunikasjonsutstyr og Sprint / Stroller /Hi Flow Stroller
Sprint / Stroller / Hi Flow Stroller er beregnet på bruk i et elektromagnetisk miljø der utstrålte RF‑forstyrrelser kontrolleres. Kundeneller brukeren av Sprint / Stroller / Hi Flow Stroller kan bidra til å forhindre elektromagnetisk forstyrrelse ved å opprettholde en
minimumsavstand mellom bærbart og mobilt RF‑kommunikasjonsutstyr (sendere) og Sprint / Stroller / Hi Flow Stroller, som anbefalt
nedenfor, i henhold til maksimal utgangseffekt for kommunikasjonsutstyr.
Maksimal nominell
Avstand i henhold til senderens frekvensr
utgangseffekt formsenderen
80 MHz og 800 MHz
800 MHz til 2,5 GHz
150 kHz til 80 MHz

d = 1,2√P

d = 1,2√P

d = 2,3√PW
0,010,12 m0,12 m0,23 m0,10,38 m0,38 m0,73 m
1
1,2 m
1,2 m
2,3 m
10
3,8 m
3,8 m
7,3 m
100
12 m
12 m
23 m
For sendere med en maksimal utgangseffekt som ikke oppført ovenfor, kan den anbefalte avstanden (d) i meter (m) beregnes ved
hjelp av ligningen som gjelder for senderfrekvensen, der P er maksimal utgangseffekt for senderen i watt (W) i henhold til produsenten av senderen.
MERKNAD 1 Ved 80 MHz og 800 MHz gjelder avstanden for høyere frekvensområder.
MERKNAD 2 Retningslinjene gjelder kanskje ikke i alle situasjoner. Elektromagnetisk spredning påvirkes av absorpsjon og refleksjon fra strukturer, gjenstander og mennesker.

* Denne tabellen er inkludert som et standardkrav for utstyr som er testet til bestemte testnivåer og over bestemte
frekvensområder, og funnet å være i samsvar med forskrifter.

Tabell 3: Veiledning og produsentens erklæring – elektromagnetisk immunitet
Sprint / Stroller / Hi Flow Stroller er beregnet på bruk i det elektromagnetiske miljøet som er spesifisert nedenfor. Kunden eller
brukeren av Sprint / Stroller / Hi Flow Stroller må sikre at enheten brukes i et slikt miljø.
Immunitetstest
IEC 60601 testnivå
Samsvarsnivå
Elektromagnetisk miljø – veiledning
Gulvet bør være av tre, betong eller keramiske± 8 kV kontakt± 8 kV kontakt
Utlading av statisk
fliser. Hvis gulvet er dekket med syntetiskeelektrisitet (ESD)± 2 kV, ± 4 kV
± 2 kV, ± 4 kVmaterialer, skal den relative fuktigheten væreIEC 61000-4-2± 8 kV, ± 15 luft
± 8 kV, ± 15 luft
minst 30 %.**
Ikke relevant±2 kV for strøm forsyningElektrisk rask
Likestrømdrevet enhet
slinjer
forbigående/utbrudd
Ikke relevant
Ikke relevant±1 kV for inndata/utdataIEC 61000-4-4Ingen datainngangs/linjer
utgangslinjer
Overspenning±1 kV linje(r) til linje(r)Ikke relevant
Ikke relevant±2 kV linje(r) til jordlikestrømdrevet enhet
IEC 61000-4-5
Spenningsfall, korte
avbrudd og spenningssvingninger istrømforsyningens
inngangslinjer
IEC 61000-4-11
Strømfrekvensens(50/60 Hz) magnetfelt
IEC 61000-4-8<5 % UT (>95 % fall på UT) i
0,5 periode
40 % UT (60 % fall på UT) i
5 perioder
Ikke relevant
70 % UT (30 % fall på UT) i likestrømdrevet enhet
25 perioder<5 % UT (>95 % fall på UT) i
5 sekunder

Ikke relevant

3A/m
50/60 Hz

Strømfrekvensmagnetfeltene bør være på nivåmed en typisk plassering i et typisk kommersielt
eller sykehusmiljø.

3A/m
50/60 Hz

Merk: UT er vekselstrømmen i strømnettet før testnivået ble brukt.

** Denne erklæringen indikerer at den nødvendige testingen ble utført i et kontrollert miljø, og Sprint / Stroller /Hi Flow Stroller ble funnet å være i samsvar med forskriftene.

126 - NO PN MN236-C4 E | BruksanvisningSprint / Stroller / Hi Flow Stroller
Tabell 4

Veiledning og produsenterklæring – Immunitet for ME-utstyr og ME-systemer

Veiledning og produsentens erklæring – Immunitet
Sprint / Stroller / Hi Flow Stroller er beregnet på bruk i det elektromagnetiske miljøet som er spesifisert
nedenfor. Kunden eller brukeren av Sprint / Stroller / Hi Flow Stroller må sikre at enheten brukes i et slikt
miljø.
Immunitetstest

IEC 60601 testnivå

Samsvarsnivå

Elektromagnetisk miljø – Veiledning

Ledet RFIEC 61000-4-6

3 Vrms6 Vrms (i ISM-bånd)150 kHz til 80 MHz

Ikke relevantBærbart og mobilt RF-kommunikasjonsutstyr skal
ikke brukes nærmere noen del av Sprint / Stroller /Hi Flow Stroller, inkludert kabler, enn den anbefalte
avstanden beregnet ut fra ligningen som gjelder for
senderfrekvensen.

Batteridrevet enhet, nr.
SIP/SOP

Anbefalt avstandd = 1,2 √P
Utstrålt RF
IEC 61000-4-3

80 MHz til 2,7 GHz

10 V/m80 MHz – 2,7 GHz
80 % AM ved 1 KHz

d = 1,2 √P
d = 2,3 √P
der P er maksimal utgangseffekt for senderen i watt(W) i henhold til produsenten av senderen, og d er
den anbefalte avstanden i meter (m).
Feltstyrker fra faste RF-sendere, som målt ved enelektromagnetisk områdeundersøkelsea, skal væremindre enn samsvarsnivået i hvert frekvensområdeb.
Det kan forekomme forstyrrelser i nærheten av utstyr
merket med følgende symbol:

Testfrekvens(MHz)Bånda)(MHz)

Tjenestea)

Modulasjonb)

Maksimaleffekt (W)Avstand(m)

Immunitetstestnivå
(V/m)

385

380–390TETRA 400

Pulsmodulasjonb)
18 Hz
FMC)
±5 kHz avvik
1 kHz sinus

1,8

0,3

27

450

430–470

GMRS 460, FRS 460

2

0,3

28

704–787

LTE-bånd 13, 17

Pulsmodulasjonb)
217 Hz

0,2

0,3

9

800–960

GSM 800/900, TETRA 800,
IDEN 820, CDMA 850,
LTE-bånd 5

Pulsmodulasjonb)
18 Hz

2

0,3

28

1700–1900

GSM 1800; CDMA 1900;
GSM 1900; DECT;
LTE‑bånd 1, 3, 4, 25; UMTS

Pulsmodulasjonb)
217 Hz

2

0,3

28

2450

2400–2570

Pulsmodulasjonb)
217 Hz

2

0,3

28

5240
5500
5785

Bluetooth, WLAN,
802,11 b/g/n, RFID 2450,
LTE-bånd 7

5100–5800

WLAN 802.11 a/n

Pulsmodulasjonb)
217 Hz

0,2

0,3

9

710
745
780
810
870
930
1720
1845
1970

Merk: Hvis det er nødvendig for å oppnå IMMUNITETSTESTNIVÅET, kan avstanden mellom senderantennen og Sprint /Stroller / Hi Flow Stroller reduseres til 1 m. Testavstanden på 1 m er tillatt i henhold til IEC 61000-4-3.
For noen tjenester er bare uplink-frekvensene inkludert.
Operatøren skal moduleres ved bruk av en arbeidssyklus på 50 % kvadratbølgesignal.
Som et alternativ til FM-modulasjon kan pulsmodulasjon på 50 % ved 18 Hz brukes fordi det ikke representerer faktisk modulasjon. Det
hadde vært det verste tilfellet.
abc

CAIRE og CAIRE Inc. er registrerte varemerker for CAIRE Inc. Vennligst besøk vår hjemmeside nedenfor for en fullstendig oversikt over varemerker. Varemerker: www.careinc.com/
corporate/trademarks.
Copyright © 2023 CAIRE Inc. CAIRE Inc. forbeholder seg retten til å stanse sine produkter, eller endre priser, materialer, utstyr, kvalitet, spesifikasjoner og/eller prosesser for sine produkternår som helst, uten forvarsel og uten videre forpliktelser eller konsekvenser. Alle rettigheter som ikke er uttrykkelig nevnt her, er forbeholdt oss, som anvendelig.

PN MN236-C4 E | Bruksanvisning NO - 127

Sprint / Stroller / Hi Flow Stroller
Symbolbeskrivning
ISO 7000

Rök inte nära enheten eller när duanvänder den. Reg.nr P002

Skydda den mot regn, håll den torr.
Reg.nr 0626

Patientansluten del av typ BF (grad av
skydd mot elektrisk stöt). Reg.nr 5333

Staplingsgräns efter antal. Reg. # 2403
Tillverkarens namn och adress.
Reg.nr 3082

Varning. Reg.nr W001
Rådets direktiv 93/42/EEG
Auktoriserad representant i den
Europeiska gemenskapen
Om den produkt unika
enhetsidentifieraren (UDI) -etiketten
har CE-nummer #### på, uppfyller
enheten kraven i direktiv 93/42 / EEG
om medicinsk utrustning. CE ####
symbolen indikerar anmält organ
nummer.

Land och tillverkningsdatum. "CC"identifierar den två bokstäverna
landskoden för tillverkningslandet.
Tillverkningsdatumet är i formatetÅÅÅÅ-MM-DD. Reg. #6049
Var försiktig, se bifogade dokument.
Reg.nr 0434A
Katalognummer. Reg.nr 2493

Denna enhet uppfyller kraven i direktiv
2010/35/EU om medicintekniska
produkter. Den har pi-märkning såsom
visas.

Serienummer. Reg.nr 2498
Lagring eller driftstemperaturområde.
Reg.nr 0632

ADR
Giftfri gas.

Lagringsfuktighetsområde. Reg.nr 2620

Fara Oxiderande ämnen:brandförstärkande risk. Oxidationsmedel
gör att bränder brinner kraftigare.

Begränsning av atmosfärtryck. Reg.
nr 2621Denna sida upp. Reg.nr 0623

Kyld vätska, USP; producerad av Air
Liquefaction
Ytterligare symbolerÖmtåligt, hanteras varsamt.
Reg.nr 0621

Enheten ska alltid vara väl ventilerad

Innehåller farliga ämnen. Reg. #3723
Håll på avstånd från lättantändliga
material, olja och fett.

Importör. Reg. #3725
ISO 7010

Auktoriserad representant i Schweiz.
Köldskador kan uppkomma vid kontakt
med kallt syre i flytande form eller
gasform, eller delar med frostbildning.
Varning låg temperatur. Varnar för
låg temperatur eller frysskador.
Reg.nr W010
Bruksanvisningen måste läsas.
Reg.nr M002Håll på avstånd från öppna lågor, eld
och gnistor. Öppna antändningskällor
och rökning är förbjudet. Reg.nr P003

128 - SE PN MN236-C4 E | Bruksanvisning

Om enheten bär UKCA-märket som
visas med UKCA#### som anger det
anmälda organets nummer, uppfyller
denna enhet UKCA-reglerna.
IEC 60417

Täck inte över enheten eller bär den
bärbara enheten under dina kläder.
Dessa enheter ventilerar normalt syre.
nr 5641

Sprint / Stroller / Hi Flow Stroller
21 CFR 801.15
Enligt amerikansk federal lag får denna
anordning endast säljas av läkare ellerpå läkares anmodan.
IEC 60601-1
DroppsäkerRådets direktiv 2012/19/EU
WEEE
Denna symbol är för att påminna utrustningens ägare om att lämna tillbaka
den till en återvinningsanläggning vid
slutet av dess livslängd, enligt direktivet
om avfall från elektrisk och elektronisk
utrustning (WEEE).
Våra produkter kommer att följa direktivet om begränsning av farliga ämnen
(RoHS). De kommer inte att innehålla
mer än spårmängder av bly eller annat
innehåll av farligt material.

ISO 15223-1
Medicinsk utrustning. Reg. # 5.7.7
Unik enhetsidentifierare # 5.7.10
Den här produkten kan omfattas av ett eller flera
amerikanska eller internationella patent. Besök vår
webbplats nedan för att se en lista över tillämpliga
patent. Pat.: www.caireinc.com/corporate/patents/.

Specifikationer• Driftsätt: Kontinuerligt flöde• Typ av skydd mot elstöt: Utrustning med intern drift• Grad av skydd mot elektrisk stöt: Patientansluten del av typ BF• Klassificering enligt graden av skydd mot inträngande vatten: IP22 - vanlig utrustning• Utrustningen är inte lämplig att använda i närvaro av lättantändliga blandningar
Produktspecifikationer

SprintStroller

Hi Flow Stroller

LOX-kapacitet

0,68 kg (1,5 lb)

1,36 kg (3,0 lb)

1,36 kg (3,0 lb)

Kapacitet motsvarande
gasmängd
Vikt, tom

513 l

1 026 l

1 025 l2,04 kg (4,5 lb)

2,27 kg (5,0 lb)

2,49 kg (5,5 lb)

Vikt, fylld

2,72 kg (6,0 lb)

3,63 kg (8,0 lb)

3,86 kg (8,5 lb)

Höjd

298 mm (11,75 tum)

343 mm (13,5 tum)

343 mm (13,5 tum)D 123,83 mm (4,875 tum)
B 191 mm (7,5 tum)
Normal användningstid vid 2 4,3 timmar
LPM
Drifttryck
20 psi (137 kPa)

D 149,23 mm (5,875 tum)
B 191 mm (7,5 tum)
8,5 timmar

D 149 mm (5,875 tum)
B 191 mm (7,5 tum)
8,5 timmar

20 psi (137 kPa)

20 psi (137 kPa)

Normalavdunstningshastighet
Standardintervall för
flödeskontroll
Flödeshastighetens
noggrannhet*

0,57 kg/dag (1,3 lb/dag)

0,57 kg/dag (1,3 lb/dag)

0,57 kg/dag (1,3 lb/dag)

Av, 0,25; 0,5; 0,75; 1; 1,5; 2;
2,5; 3; 4; 5; 6 LPM+/- 10 %Av, 0,25; 0,5; 0,75; 1; 1,5; 2;
2,5; 3; 4; 5; 6 LPM+/- 10 %Av, 0,5, 1, 2, 2,5, 3, 4, 6, 8,
10, 12, 15 LPM+/- 10 %Mått* Denna noggrannhet gäller bara vid 70 °F och 14,7 psig samt med en kalibrerad exakt massflödesmätare.

PN MN236-C4 E | Bruksanvisning SE - 129

Sprint / Stroller / Hi Flow Stroller
Varningsinformation
Viktigt! Läs handboken noggrant innan du använder
Sprint/Stroller/Hi Flow Stroller.
Endast på läkarordination.
VARNING: ENHETEN ÄR INTE AVSEDD FÖRLIVSUPPEHÅLLANDE ANVÄNDNING.
VARNING: OM DU KÄNNER ATT UTRUSTNINGEN INTE
FUNGERAR SOM DEN SKA, RING DIN VÅRDGIVARE.
FÖRSÖK INTE REPARERA ELLER JUSTERA ENHETENSJÄLV.
VARNING: MODIFIERA INTE UTRUSTNINGEN UTANAUKTORISERING FRÅN TILLVERKAREN.
VARNING: OM DET KRÄVS KONTINUERLIGSYREFÖRSÖRJNING KONTROLLERAR DU ATTLÄMPLIG SYRGASFÖRSÖRJNING OCH/ELLERSEKUNDÄR SYRGASFÖRSÖRJNING ÄR TILLGÄNGLIGUNDER BEHANDLINGEN.
VARNING: TILLÅT INTE RÖKNING, LEVANDE LJUS
ELLER ÖPPNA LÅGOR INOM 3 METER FRÅN ENHETENELLER NÄRMARE AVSTÅND ÄN 20 CM FRÅN ENANTÄNDNINGSKÄLLA.
VARNING: ANVÄND OCH FÖRVARA DIN ENHET PÅ EN
VÄL VENTILERAD PLATS.
VARNING: FÖRVARA INTE UTRUSTNINGFÖR FLYTANDE SYRE I EN GARDEROB, I
BILENS BAGAGEUTRYMME ELLER I ANDRAAVGRÄNSADE OMRÅDEN. LÄGG INTEFILTAR, GARDINER ELLER ANNAT TYG ÖVERUTRUSTNINGEN.
VARNING: DENNA PRODUKT KAN UTSÄTTA DIG
FÖR KEMIKALIER, BLAND ANNAT NICKEL, SOM I
STATEN KALIFORNIEN I USA HAR KLASSATS SOM
CANCERFRAMKALLANDE. MER INFORMATION FINNSPÅ WWW.P65WARNINGS.CA.GOV.
VARNING: OM DET INTRÄFFAR EN ALLVARLIG
HÄNDELSE MED DEN HÄR ENHETEN BÖRANVÄNDAREN OMEDELBART RAPPORTERA
HÄNDELSEN TILL LEVERANTÖREN OCH / ELLERTILLVERKAREN. EN ALLVARLIG HÄNDELSE
DEFINIERAS SOM EN SKADA, DÖDSFALL ELLERPOTENTIAL ATT ORSAKA SKADA / DÖDSFALL OM
HÄNDELSEN SKULLE ÅTERKOMMA. ANVÄNDARENKAN OCKSÅ RAPPORTERA HÄNDELSEN TILL
DEN BEHÖRIGA MYNDIGHETEN I DET LAND DÄRHÄNDELSEN INTRÄFFADE.
Viktigt: Använd endast Sprint/Stroller/Hi Flow
Stroller enligt anvisningar från din läkare.
Viktigt: Enligt amerikansk federal lag får denna anordning endast säljas av läkare eller på läkares anmodan.
Viktigt: Enheten innehåller flytande syre som är extremt
kallt, nästan -184 °C (-300 °F). Att utsättas för sådana
temperaturer kan ge svåra köldskador.

130 - SE PN MN236-C4 E | Bruksanvisning

Viktigt: Syre i flytande form och gasform är i sig inte
lättantändligt men det kan göra att andra material
brinner fortare. Denna risk, tillsammans med den lågatemperaturen hos flytande syre, gör att vissa säkerhetsåtgärder behövs.
Viktigt: Håll brännbart material borta från enheten. Aerosolspray, olja och smörjmedel, inklusive ansiktskrämer
och vaselin antänds lätt och kan brinna snabbt inärheten av syre.
Viktigt: Rökning i samband med användning av näskanyl kan orsaka brännskador i ansiktet och eventuellt
dödsfall.
Om man tar av näskanylen och lägger den på kläder,sängkläder, soffor eller andra mjuka material uppstår enbrinnande låga om näskanylen utsätts för cigarrettglöd,
en värmekälla, en gnista eller en låga.
Om du ska röka: (1) stäng av den bärbara enheten, (2) ta
bort kanylen och (3) lämna det rum där enheten finns.
Viktigt: Om den råkar falla omkull, räta försiktigt men
genast upp den igen till stående position om det går.
Om syre läcker ut ska du genast gå bort från enheten
och ringa din vårdgivare. Försök inte flytta enheten eller
stoppa det flytande syret från att läcka ut.
Obs! Rör inte vid de delar av enheten som har frost på sig.
Obs! Förvara eller använd inte den bärbara enheten när
den är kopplad till den stationära enheten.
Obs! Låt inte outbildad personal hantera eller använda
enheten.
Obs! Användning av denna enhet ombord på kommersiellaflyg och transportflyg är förbjuden av Federal Aviation
Administration.

Avsedd användning

CAIRE Sprint/Stroller/Hi Flow Stroller är avsedd för
administrering av extra syre. Enheten är inte avsedd för
livsstöd och ger inte heller någon patientövervakningsfunktion.

Sprint / Stroller / Hi Flow Stroller
Inledning

Reglage

Sprint, Stroller och Hi Flow Stroller är avsedda för
tillförsel av syrgas till patienten i slutanvändarens
hem och kan även användas inom institutioner somvårdhem och inrättningar för subakut vård. Sprint/
Stroller/Hi Flow Stroller utgör en flyttbar syrgaskälla
under en längre tidsperiod.

1. Elektronisk vätskenivåmätare
2. Elektronisk vätskenivåkontakt
3. Mekanisk vätskenivåmätare
4. Vred för flödeskontroll
5. Anslutning för syrgasslang
6. Luftningsventil
7. Fyllningsanslutningar (se nästa sida för
QDV‑identifiering)
8. Bärrem med axeldyna
9. Kondensdyna
10. Kondenskopp

Enheten används av KOL-patienter eller personer
med nedsatt andningskapacitet. Enheten är ordinerad
till patienten. Enheten säljs till en leverantör som ärutbildad i att använda och utföra service på Sprint/
Stroller/Hi Flow Stroller. Leverantören utbildar
användaren.

1

Enheten är inte avsedd för livsuppehållandeverksamhet och den är inte ämnad för
patientövervakning. Vi rekommenderar att du har
tillgång till en annan källa med extra syrgas i händelse
av ett mekaniskt fel. De här bärbara enheterna haräven en snabbkopplingsventil som kan kopplas tillen behållarenhet för påfyllning av flytande syre iden bärbara enheten. Behållarenheten fylls på av dinvårdgivare.

2
4

5
8

Den här bruksanvisningen innehåller instruktioner
för användning av Sprint/Stroller/Hi Flow Stroller.
I bruksanvisningen som levereras med den stationära
enheten finns uppgifter om dess användning.
Obs! Serviceleverantören hjälper till med den inledande
inställningen och instruerar om korrekt hantering och
användning av enheten.

57

Sprint-, Stroller- och Hi Flow Stroller-enheter

6

10
9

PN MN236-C4 E | Bruksanvisning SE - 131

Sprint / Stroller / Hi Flow Stroller
Identifiering av snabbkopplingsventil(QDV)
Bärbar,anslutning

Stationär,
anslutning

3. Placera påfyllningsanslutningen i upprätt läge överden stationära enhetens anslutning.

QDV-typ

CAIRE,
sidofyllningSF,låsning

LIFE-OX
TF,
låsning

4. a. Om anslutningen är påskjutningsbar:
Sänk ned Sprint-/Stroller-/Hi Flow Strollerenheten på den stationära enheten tills du känner
att anslutningarna kopplas ihop.

PENOX

TAEMA
TF,
påskjutning
PB

b. Om anslutningen är en Penox eller Life-Ox:
Vrid Sprint/Stroller/Hi Flow Stroller tills dukänner att anslutningarna kopplas ihop. Sedan
vrider du försiktigt och bestämt Sprint/Stroller/
Hi Flow Stroller medurs ( ) samtidigt som dutrycker nedåt.

AnvändarinstruktionerÖvre påfyllning av Sprint/Stroller/
Hi Flow Stroller

Obs! Fortsättning finns i steg 6 i Sidopåfyllning av Sprint/
Stroller/Hi Flow Stroller.

1. Rengör påfyllningsanslutningarna på både denstationära enheten och Sprint/Stroller/Hi Flow
Stroller med en ren, torr, luddfri trasa.
Viktigt: Påfyllningsanslutningarna måste vara
torra eftersom fukt kan orsaka att Sprint/Stroller/Hi Flow Stroller och den stationära enheten
fryser ihop.

Sidopåfyllning av Sprint/Stroller/
Hi Flow Stroller

2. Se till att vreden för flödeskontroll på både Sprint/
Stroller/Hi Flow Stroller och behållaren är iavstängt läge (0).

132 - SE PN MN236-C4 E | Bruksanvisning

1. Rengör påfyllningsanslutningarna på både denstationära enheten och Sprint/Stroller/Hi Flow
Stroller med en ren, torr, luddfri trasa.
VARNING: PÅFYLLNINGSANSLUTNINGARNAMÅSTE VARA TORRA EFTERSOM FUKT KANORSAKA ATT SPRINT/STROLLER/HI FLOWSTROLLER OCH DEN STATIONÄRA ENHETENFRYSER IHOP.

Sprint / Stroller / Hi Flow Stroller
2. Se till att vreden för flödeskontroll på både Sprint/
Stroller/Hi Flow Stroller och behållaren är i avstängt
läge (0).
3. Se till att påfyllningsanslutningen på Sprint/Stroller/
Hi Flow Stroller är i upprätt läge mot den stationära
enhetens anslutning.

Obs! Vätskenivåmätarens avläsningar är mest exakta 30
sekunder eller mer efter att Sprint/Stroller/Hi Flow Stroller
har fyllts.

4. Vrid Sprint/Stroller/Hi Flow Stroller moturs ( )tills du känner att sprinten och spåret kopplas ihop.
7. Du kan också se ånga runt anslutningen.
Det är normalt och orsakas av den stora
temperaturskillnaden mellan det flytande syret och
den varma luften som omger enheterna.

5. Vrid försiktigt och bestämt Sprint/Stroller/Hi Flow
Stroller tillbaka till upprätt läge. Nu är enheterna
ihopkopplade.

8. När enheten är full stänger du ventilationsventilen.
Du vet att enheten är full när det väsande ljudets
ton förändras och lite vätska kommer ut ur
ventilationsledningen.
Obs! Vid stängning av ventilen efter att ett nästan
kontinuerligt vätskeflöde matas ut fylls Sprint/Stroller/Hi
Flow Stroller på det mest fullständiga sättet, men visst
flytande syre går förlorat.
Obs! Vid stängning av ventilen efter första tecknet på
vätskeutmatning fylls inte Sprint/Stroller/Hi Flow Stroller
helt, men detta sparar flytande syre.
9. För anslutningarna av påskjutningstyp trycker duned frigöringsknappen tills enheterna kopplas isär.
För alla andra anslutningar vrider du enheten tills
den frigörs från den stationära enheten.

Obs! Fortsättning följer i steg 6.
Obs! För anslutningen av påskjutningstyp håller duenheten på plats under hela påfyllningen.

Frigöringsknapp för bärbara enheter(endast påskjutningstyp)6. Öppna ventilationsventilen för att börja fylla din
Sprint/Stroller/Hi Flow Stroller. Du ska kunna höra
att viss syrgas läcker ut. Det är inget att oroa sig över.
Det väsande ljudet du hör är helt normalt.

PN MN236-C4 E | Bruksanvisning SE - 133

Sprint / Stroller / Hi Flow Stroller
Viktigt: Kontrollera mätaren för vätskenivå
endast efter att ventilen har stängts.

Användarinstruktioner

Viktigt: Om Sprint/Stroller/Hi Flow Stroller fylls på
direkt efter en tids användning fyller du på den
och väntar sedan i 10 minuter efter påfyllning med
flödeskontrollventilen (FCV) inställd på noll så att
trycket stabiliseras. Använd sedan som vanligt.
VARNING: ANVÄND INTE SPRINT/STROLLER/
HI FLOW STROLLER OM DEN ÄR ANSLUTEN
TILL DEN STATIONÄRA ENHETEN.
VARNING: OM SPRINT/STROLLER/HI FLOW STROLLER
INTE GÅR LÄTT ATT KOPPLA BORT SKA DU INTEANVÄNDA FÖR MYCKET KRAFT. ENHETERNA KANHA FRYST FAST I VARANDRA. LÅT ENHETERNA
VARA IHOPKOPPLADE TILLS DE HAR VÄRMTS UPP.
DÅ ÄR DET LÄTT ATT KOPPLA ISÄR DEM. RÖR INTE
VID DELAR SOM ÄR FROSTTÄCKTA. VARNING:
OM LÄCKAGET ÄR SÅ KRAFTIGT ATT VÄTSKASTRÖMMAR UT, SKA DU LÄMNA PLATSEN OCH RINGADIN VÅRDGIVARE OMEDELBART.
VARNING: SYRE I FLYTANDE FORM OCH KALLT SYREI GASFORM KAN MEDFÖRA FROSTSKADOR PÅ
HUDEN. SYRGAS LEDER TILL ÖKAD BRANDRISK.
Viktigt: Om vätskeläckage skulle inträffa frånden stationära enheten eller Sprint/Stroller/
Hi Flow Stroller vid frånkoppling av enheternaska Sprint/Stroller/Hi Flow Stroller ställas åt
sidan. Se till att den är i upprätt läge, lämna
rummet och ring din vårdgivare omedelbart.
Obs: Till utrustningsleverantören: Följande syretillbehör
rekommenderas för användning med Sprint / Stroller / Hi
Flow Stroller:• Nasal kanyl med 7 fot (2,1 m) slang (maximalt LPM: 6
LPM): CAIRE artikelnummer 5408-SEQ• Firebreak: CAIRE artikelnummer 21126636
En eldbrytning rekommenderas för användning med
kanyler.
• CAIRE erbjuder en eldbrytare som är avsedd att
användas i samband med det bärbara syret. Firebreak är
en termisk säkring för att stoppa gasflödet i händelse av att
nedströms kanylen eller syrgasröret antänds och bränns
till firebreak. Den placeras i linje med näsekanylen ellersyrgasröret mellan patienten och syreutloppet från Sprint/ Stroller / Hi Flow Stroller. För korrekt användning av
brandbrytaren, se alltid tillverkarens anvisningar (medföljer
varje brandbrytarkit).
• Ytterligare rekommenderad tillbehörsinformation finns
online på www.caireinc.com.

134 - SE PN MN236-C4 E | Bruksanvisning(Näskanyl medföljer inte.)

1. Skjut på näskanylens luftslang ordentligt påsyrgasslangens anslutning.
2. Placera näskanylens luftslang i rätt läge så att du
bekvämt kan andas in syrgasen.
3. Vrid vredet för flödeskontroll längst upp på Sprint/
Stroller/Hi Flow Stroller medurs tills den angivna
flödeshastigheten (siffra) syns i vredets ”fönster”
och du känner en ”klickning”.
Viktigt: Om vredet står mellan två hastigheter
uppstår ett ospecificerat syrgasflöde.
4. Du bör nu få syre. Stoppa syrgasflödet genom
att vrida vredet för flödeskontroll moturs till det
avstängda läget (0).
Viktigt: Vrid alltid vredet för flödeskontroll till
läget av (0) när enheten inte används.

Sprint / Stroller / Hi Flow Stroller
5. Använd följande tabell som riktlinje för att fastställa
tidsperioden då Sprint/Stroller/Hi Flow Stroller är idrift:
FCV-inställning
ModellSprintStroller

Hi Flow
Stroller

Nominellt

Nominellt

FCVpos.

SprintStroller

Hi Flow
Stroller

Av

30:00

60:00

71:00

0,25

18:00

30:00

Ej tillämpligt

0,5

12:12

23:00

32:00

0,75

9:18

17:24

Ej tillämpligt

1

7:30

14:06

16:00

1,5

5:30

10:12

Ej tillämpligt

2

4:18

8:00

8:00

2,5

3:36

6:36

6:20

3

3:06

5:36

5:20

4

2:24

4:24

4:00

5

2:00

3:36

Ej tillämpligt

6

1:42

3:00

2:40

8

Ej tillämpligt

Ej tillämpligt

2:00

10

Ej tillämpligt

Ej tillämpligt

1:40

12

Ej tillämpligt

Ej tillämpligt

1:20

15

Ej tillämpligt

Ej tillämpligt

1:00

Obs! Tiden visas i timmar och minuter (format 00:00).
Obs! ”Nominella” tider gäller vid idealiskaförhållanden, dvs. maximal påfyllning, exakta
flödeshastigheter, bra förlustfrekvens, ingen
förflyttning av enheten osv.
De här tidsangivelserna är de högsta förväntade.
Obs! Dina enskilda resultat varierar.

• Mätare, gen. 4: Tryck på den gröna driftsknappen på
mätarens framsida. Läs av lysdiodsbågen som visar
innehållsnivån.

Viktigt: Sprint/Stroller/Hi Flow Stroller är tom
om endast det sista segmentet på ljusremsan
lyser.
Om indikatorn för låg batterinivå tänds när knappen
trycks ned ska du meddela det till din vårdgivare.
Obs! Sprint/Stroller/Hi Flow Stroller fortsätter att matasyrgas även om indikatorn för låg batterinivå tänds, underförutsättning att det finns flytande syre i enheten.
6b. Kontrollera nivån av flytande syre i enheten med
den mekaniska vätskenivåmätaren:• Stötta enheten med tygremmen direkt ovanför densvarta innehållsindikatorns fjädervågskammare.
• Låt enheten stabiliseras (dvs. inte röra sig uppåt ochnedåt) och läs sedan av det ungefärliga flytande
syreinnehållet på den färgade skalan som visas.

6a. Kontrollera nivån av flytande syre i enheten med
den elektroniska vätskenivåmätaren:• Mätare, gen. 3: Tryck på knappen (brytare för
vätskenivå) på maskinen i minst två sekunder. Läs av
innehållsnivån längs ljusremsans övre del.

PN MN236-C4 E | Bruksanvisning SE - 135

Sprint / Stroller / Hi Flow Stroller
Obs! Om enheten är tom visas endast skalans rödfärgadedel. Om så är fallet måste enheten fyllas före användning.
7. Under vissa miljöförhållanden, vid kontinuerlig
användning och i synnerhet om den bärbara
enheten inte flyttas, kan den utveckla kraftig frost
runt förångarspolen och på höljet. Du kan minska
frostbildningen genom att knacka på enheten och/eller torka av eventuell ansamlad frost på höljet.

Obs! Om dynan är mycket smutsig kan den tvättas imaskin (med blekmedel för desinficering).
• Sätt tillbaka genom att sätta i dynan i det nedre
skyddet och stänga den nedre dragkedjan.
Obs! Om vredet inte går att trycka in och koppla i koppenhåller du i koppen och drar ut vredet tills det klickar påplats i det öppna läget. Sätt nu tillbaka koppen enligt
tidigare beskrivning.

Underhåll av kondensdynan

Minskning av frostbildningNär enheten är tom och har värmts upp till
rumstemperatur avlägsnar du eventuell vätska frånkondenskoppen:

Under vissa miljöförhållanden, vid kontinuerlig
användning och i synnerhet om den bärbara
enheten inte flyttas, kan den utveckla kraftig frost
runt förångarspolen och på höljet. Du kan minska
frostbildningen genom att knacka på enheten och/eller
torka av eventuell ansamlad frost på höljet.
VARNING: PÅ GRUND AV RISKEN FÖRÖVERDRIVEN FUKTIGHET/ANSAMLING
AV IS REKOMMENDERAR VI ATT HI
FLOW STROLLER ÄR INAKTIV I 1 TIMMEEFTER ANVÄNDNING FÖR ATT MINSKAUPPBYGGNAD AV FUKT.

För hårt hölje• Luta enheten åt ena sidan. Ta tag i vredet
i mitten av kondenskoppen (längst ned på
enheten) och dra det rakt ut. Då frigörs koppen.
Obs! Om dynan är mycket smutsig kan den tvättas imaskin (med blekmedel för desinficering).
• Sätt tillbaka koppen genom att centrera den överden runda öppningen i botten på enheten och
trycka på vredet med ett klick för att koppla in
koppen och ett andra klick för att låsa koppen påplats.

Låst
InkoppladÖppenVredkopp

Obs! Om vredet inte går att trycka in och koppla i koppenhåller du i koppen och drar ut vredet tills det klickar påplats i det öppna läget. Sätt nu tillbaka koppen enligt
tidigare beskrivning.
För mjukt hölje• Dra upp den nedre dragkedjan och dra tillbaka
det nedre skyddet.
• Ta bort dynan från koppen och vrid ur eventuellt
absorberad fukt. Låt dynan torka helt innan den
sätts tillbaka i koppen.

136 - SE PN MN236-C4 E | Bruksanvisning

Till följd av de högre flödeshastigheterna i Hi Flow
Stroller skapas kondens- och viss frostbildning på
enheten, i synnerhet vid flödesinställningar på 10 LPM
och högre. Vi rekommenderar att enheten är inaktiv iminst 1 timme efter användning för att minska risken
för överdriven ansamling till följd av kontinuerlig
användning.

Sprint / Stroller / Hi Flow Stroller
Felsökning
Följande information är avsedd att hjälpa till vid felsökning och lösa enklare driftsproblem som du kan stöta på
när du använder Sprint/Stroller/Hi Flow Stroller.
ProblemLösning
Ett väsande ljud hörs från• Väsljud kan uppstå då den bärbara enheten jämnar ut trycket. Det inträffar oftast precis
den bärbara enheten.
efter påfyllning eller när den bärbara enhetens position ändras. Väsljudet kan pågå iungefär 10 minuter efter påfyllningen. Det kan dessutom inträffa när flödeskontrollventilen
är lågt inställd.

Den bärbara enhetens flöde
upphör under användning.

• Om den bärbara enheten har lagts i ett felaktigt läge ska den ställas i upprätt läge och fåmöjlighet att stabiliseras under flera minuter.
• Kontrollera att kanylen sitter fast ordentligt vid syrgasutloppet.
• Kontrollera så att inte näskanylen är klämd.
• Kontrollera innehållsindikatorn/nivåmätaren och fyll vid behov på den bärbara enheten.

Den bärbara enheten fylls
inte på.

• Kontrollera att vredet för flödeskontroll inte är i avstängt läge (”0”).
• Kontrollera att det finns syre i behållaren.

• Kontrollera att kopplingen mellan den bärbara enheten och behållaren är korrekt genom
hela påfyllningsproceduren.
Den bärbara enhetens ventil • Om ventilen inte stängs och väsljudet och syreångan fortsätter, ta försiktigt bort denbärbara enheten genom att trycka ned bortkopplingsknappen på behållaren. Ventileringen
stängs inte ordentligt närfrån enhetens botten slutar efter några minuter. Låt enheten bli varm igen så att ventilenpåfyllningen avslutats.
kan stängas. Den bärbara enheten kan behöva upp till 60 minuter innan trycket återställs
så att syreflödet kan fungera korrekt. Om det behövs använder du en alternativ syrekälla,t.ex. en flödeskontrollventil som ansluts till behållaren.
• Anslutningen mellan den bärbara enheten och behållaren har fryst.
Den bärbara enheten
sitter fast i behållaren efter• ANVÄND INTE VÅLD. Låt de frysta delarna värmas upp i några minuter och ta sedanpåfyllningen.
loss den bärbara enheten när isen har smält. Förhindra att enheterna fryser fast genom
att alltid torka av behållarens hankoppling och den bärbara enhetens honkoppling med entorr, ren och luddfri duk innan påfyllning.

PN MN236-C4 E | Bruksanvisning SE - 137

Sprint / Stroller / Hi Flow Stroller
Rengöringsstandard
VARNING: RENGÖR BARA NÄR ENHETENÄR TOM.
• Rengör med en lösning av milt diskmedel och
vatten.
• Fukta en luddfri duk med rengöringsmedlet.
Godkända rengöringsmedel omfattar HydroPure
och HydroKlean. Spruta inte rengöringsmedel
direkt på Sprint/Stroller/Hi Flow Stroller.
• Gnugga enhetens utsida med den luddfria duken
tills den är ren.
Viktigt: Använd inte hög temperatur eller
högtryckstvätt för att rengöra de här delarna.
• Se till att det inte kommer rengöringsmedel på
inre delar eller ventiler.
• Låt enheten torka innan den används.
Obs! Till vårdgivare – Se tillämplig servicehandbok för
bearbetningsrutiner.

Underhåll
Din serviceleverantör ansvarar för alla underhållskrav
enligt den tekniska handboken för enheten. Ring din
serviceleverantör för alla underhållskrav.
Kondensdynan är den enda del som användaren kan
utföra service på. Användaren kan inte utföra service påandra delar.
Den förväntade livslängden är minst fem år.
Rengör påfyllningsanslutningarna på både stationära
och bärbara enheter med en ren, torr och luddfri duk
mellan varje påfyllning för att förhindra att de fryser,
vilket kan göra att utrustningen inte fungerar.
Obs! Allt annat underhåll måste utföras av en kvalificerad
servicetekniker eller serviceleverantör.

Tillbehör

WEEE och RoHS

Syftet med denna symbol är att påminnaägaren av utrustningen att lämna den till enåtervinningsanläggning när den ska kasseras,
enligt WEEE-direktivet om kassering av
elektriskt avfall (Waste Electrical and Electronic
Equipment).
Våra produkter uppfyller kraven i RoHS-direktivet(begränsning av användningen av vissa farliga ämnen
i elektriska och elektroniska produkter). De innehåller
inte mer än spårelement av bly och andra farliga
material.

Bortskaffande
Lämna alltid tillbaka Sprint/Stroller/Hi Flow Stroller,
inklusive alla komponenter till din hemvårdgivare för
korrekt kassering. Du kan även kontakta ditt lokala
förvaltnings- eller kommunkontor för instruktioner om
korrekt avfallshantering av batteriet.

Transport och förvaring
Enheten ska förvaras i upprätt läge på en väl ventileradplats. Låt inte enheten ligga på sidan. Luftfuktighet upp
till 95 % icke-kondenserande. Temperaturintervall från-40 °C till 70 °C (-40 °F till 158 °F).
Drifttemperaturintervall från -10 °C till 40 °C (14 °F
till 104 °F). Relativ luftfuktighet 30 % till 75 % ickekondenserande.
Obs! Atmosfärstrycksområdet är mellan 700 hPa och
1 060 hPa (stigning på 10 000 ft. till -1 000 ft.).

138 - SE PN MN236-C4 E | Bruksanvisning

Vagn, artikelnr 20765565

Sprint / Stroller / Hi Flow Stroller
Säkerhet
VARNING: BÄRBAR RF-KOMMUNIKATIONSUTRUSTNING (INKLUSIVE KRINGUTRUSTNING SOM
ANTENNKABLAR OCH EXTERNA ANTENNER) SKA INTE ANVÄNDAS NÄRMARE ÄN 30 CM (12 TUM)
FRÅN NÅGON DEL AV SPRINT/STROLLER/HI FLOW STROLLER, DÄRIBLAND KABLAR SOM ANGES AV
TILLVERKAREN. I ANNAT FALL KAN UTRUSTNINGENS PRESTANDA FÖRSÄMRAS.
VARNING: ANVÄNDNING AV ANDRA TILLBEHÖR, GIVARE OCH KABLAR UTÖVER DE SOM SPECIFICERATSELLER TILLHANDAHÅLLITS AV TILLVERKAREN AV DEN HÄR UTRUSTNINGEN KAN LEDA TILL ÖKADE
ELEKTROMAGNETISKA EMISSIONER ELLER MINSKAD ELEKTROMAGNETISK IMMUNITET FÖR DEN HÄR
UTRUSTNINGEN, OCH GÖRA ATT DEN FUNGERAR FELAKTIGT.
VARNING: ANVÄNDNING AV DENNA UTRUSTNING DIREKT INTILL ELLER STAPLAD PÅ ANNAN UTRUSTNING BÖRUNDVIKAS EFTERSOM DET KAN LEDA TILL FELAKTIG FUNKTION. OM SÅDAN ANVÄNDNING ÄR NÖDVÄNDIG
MÅSTE UTRUSTNINGEN OCH ANNAN UTRUSTNING ÖVERVAKAS FÖR ATT BEKRÄFTA ATT DEN FUNGERARNORMALT.
Viktigt: Elektrisk utrustning för medicinskt bruk kräver särskilda försiktighetsåtgärder med avseende på
elektromagnetisk kompatibilitet (EMC) och måste installeras och tas i bruk enligt EMC-informationen i den
här handboken.
Viktigt: Bärbar och mobil RF-kommunikationsutrustning kan påverka medicinsk elektronisk utrustning.
Tabell 1

Vägledning och tillverkarens deklaration – elektromagnetiska emissioner
Sprint/Stroller/Hi Flow Stroller är avsedda för användning i nedan angiven elektromagnetisk miljö. Kundeneller användaren av Sprint/Stroller/Hi Flow Stroller måste försäkra sig om att utrustningen används i en sådan
miljö.
EmissionstesterÖverensstämmelse

RF-emissioner

Grupp 1CISPR 11

RF-emissioner
CISPR 11

Elektromagnetisk miljö – vägledning
Sprint/Stroller/Hi Flow Stroller använder endast
RF‑energi för interna funktioner.
Därför är RF-emissionerna väldigt låga och det ärinte sannolikt att de inverkar störande på elektronisk
utrustning i närheten.

Klass B

Harmoniska emissioner
IEC 61000-3-2

Ej tillämpligt

Spänningsvariationer/
flimmeremissioner
IEC 61000-3-3

Ej tillämpligt

Sprint/Stroller/Hi Flow Stroller lämpar sig för användningi alla inrättningar, inklusive hemmiljöer och sådana somär direktanslutna till det allmänna lågspänningsnätet som
försörjer bostadsbyggnader.

PN MN236-C4 E | Bruksanvisning SE - 139

Sprint / Stroller / Hi Flow Stroller
Tabell 2*: Rekommenderat avstånd mellan bärbar och mobil RF-kommunikationsutrustning och Sprint/Stroller/Hi
Flow Stroller
Sprint/Stroller/Hi Flow Stroller är avsedd att användas i en elektromagnetisk miljö där utstrålade RF-störningar är kontrollerade.
Kunden eller användaren av Sprint/Stroller/Hi Flow Stroller kan hjälpa till att förhindra elektromagnetiska störningar genom att
upprätthålla ett minimiavstånd mellan bärbar och mobil RF-kommunikationsutrustning (sändare) och Sprint/Stroller/Hi Flow Stroller
enligt rekommendationerna nedan, i enlighet med kommunikationsutrustningens maximala uteffekt.
Sändarens uppmätta
Separationsavstånd enligt sändarens frekvens
maximala uteffektm
80 MHz till 800 MHz
800 MHz till 2,5 GHz
150 kHz till 80 MHzW
d = 1,2√P
d = 2,3√P
d = 1,2√P
0,01
0,12 m0,12 m0,23 m0,10,38 m0,38 m0,73 m
1
1,2 m
1,2 m
2,3 m
10
3,8 m
3,8 m
7,3 m
100
12 m
12 m
23 m
För sändare med en maximal uteffekt som inte listas ovan kan rekommenderat separationsavstånd (d) i meter (m) beräknas med
hjälp av den ekvationen som tillämpas för sändarens frekvens, där P är sändarens maximala uteffekt i watt (W) enligt sändartillverkaren.
OBS 1! Vid 80 MHz och 800 MHz är det separationsavståndet för det högre frekvensområdet som gäller.
OBS 2! Dessa riktlinjer kanske inte gäller i alla situationer. Elektromagnetisk vågutbredning påverkas av absorption och reflektionfrån byggnader, föremål och människor.

* Denna tabell ingår som ett standardkrav för utrustning som har testats för specifika testnivåer och över specifika
frekvensintervall och överensstämmer med bestämmelserna.

Tabell 3: Vägledning och tillverkarens deklaration – elektromagnetisk immunitet
Sprint/Stroller/Hi Flow Stroller är avsedda för användning i nedan angiven elektromagnetisk miljö. Kunden eller användaren av
Sprint/Stroller/Hi Flow Stroller måste försäkra sig om att utrustningen används i en sådan miljö.
Immunitetstest
IEC 60601 testnivå
Överensstämmelsenivå Elektromagnetisk miljö – vägledning± 8 kV kontakt± 8 kV kontakt
Elektrostatisk urGolven ska vara av trä, betong eller kakel. Om
laddning (ESD)± 2 kV, ± 4 kV
± 2 kV, ± 4 kV
golven är täckta med syntetmaterial ska den
relativa fuktigheten vara minst 30 %.**IEC 61000-4-2±8 kV, ± 15 kV luft
±8 kV, ± 15 kV luft
Snabb elektrisk
transient/skur
IEC 610004-4±2 kV för strömförsörjningmatningsledningar±1 kV för ingångs-/utgångsledningar±1 kV ledning(ar) till
ledning(ar)±2 kV ledning(ar) till jord
<5 % UT (>95 % fall i UT) i
Spänningsfall, korta
0,5 cykler
avbrott och spännings- 40 % UT (60 % fall i UT) i
variationer hos
5 cyklerströmförsörjningens
70 % UT (30 % fall i UT) iingångsledningar
25 cykler<5 % UT (>95 % fall i UT)
IEC 61000-4-11
i 5 sek
Spänningsfrekvensens 3 A / m(50/60 Hz) magnetfält
50/60 HzIEC 61000-4-8
Toppar

IEC 61000-4-5

Ej tillämpligt
DC-aktiverad enhet
Ej tillämpligt

Ej tillämpligt

Inga in- och utgångsledningar för data
Ej tillämpligt
DC-aktiverad enhet

Ej tillämpligt

Ej tillämpligt
DC-aktiverad enhet

Ej tillämpligt

3A/m
50/60 Hz

Magnetfält som uppstår på grund av spänningsfrekvens ska ligga på nivåer typiska för en
kommersiell miljö eller en sjukhusmiljö.

Obs! UT är nätspänningen innan testnivån tillämpades.

** Det här meddelandet anger att nödvändig testning har utförts i en kontrollerad miljö och att Sprint/Stroller/Hi
Flow Stroller överensstämmer med bestämmelserna.

140 - SE PN MN236-C4 E | BruksanvisningSprint / Stroller / Hi Flow Stroller
Tabell 4
Vägledning och tillverkarens deklaration – immunitet gällande magnetelektrisk utrustning och
magnetelektriska system
Vägledning och tillverkarens deklaration – immunitet
Sprint/Stroller/Hi Flow Stroller är avsedda för användning i nedan angiven elektromagnetisk miljö. Kunden eller
användaren av Sprint/Stroller/Hi Flow Stroller måste försäkra sig om att utrustningen används i en sådan miljö.
Immunitetstest

IEC 60601-testnivå

Överensstämmelsenivå

Elektromagnetisk miljö – vägledning

Ledningsburen RF IEC
61000-4-6

3 Vrms6 Vrms (i ISM-band)
150 kHz till 80 MHz

Ej tillämpligt

Bärbar och mobil RF-kommunikationsutrustning
ska inte användas närmare någon del av Sprint/
Stroller/Hi Flow Stroller, inklusive kablarna, än det
rekommenderade minsta avstånd som beräknats
med den ekvation som är tillämplig för sändarens
frekvens.

Batteridriven enhet, ingen
SIP/SOP

Rekommenderat separationsavstånd
d = 1,2 √P
Utstrålad RF IEC
61000-4-3

80 MHz till 2,7 GHz

10 V/m80 MHz till 2,7 GHz
80 % AM vid 1 kHz

d = 1,2 √P
d = 2,3 √ P
där P är sändarens maximala uteffekt i watt (W) enligt
sändartillverkaren och d är det rekommenderade
separationsavståndet i meter (m).
Fältstyrkor från fasta RF-sändare, somfastställts genom en elektromagnetisk
anläggningsundersökninga, ska vara mindre änöverensstämmelsenivån för varje frekvensområdeb.
Störningar kan inträffa i närheten av utrustning somär märkt med följande symbol:

Testfrekvens(MHz)

Banda)(MHz)

Servicea)

Moduleringb)

385

380–390TETRA 400450

430–470

GMRS 460, FRS 460

704–787

800–960

710
745
780
810
870
930
1720
1845
1970

1700–1900

2450

2400–2570

5240
5500
5785

5100–5800

Maximaleffekt (W)Avstånd(m)

Immunitetstestnivå
(V/m)

Pulsmoduleringb)
18 Hz
FMC)
±5 kHz avvikelse
1 kHz sinus

1,8

0,3

27

2

0,3

28

LTE-band 13, 17

Pulsmoduleringb)
217 Hz

0,2

0,3

9

GSM 800/900, TETRA 800,
IDEN 820, CDMA 850,
LTE-band 5

Pulsmoduleringb)
18 Hz

2

0,3

28

Pulsmoduleringb)
217 Hz2

0,3

28

Pulsmoduleringb)
217 Hz2

0,3

28

Pulsmoduleringb)
217 Hz

0,2

0,3

9

GSM 1800, CDMA 1900,
GSM 1900, DECT,
LTE‑band 1, 3, 4, 25,
UMTS
Bluetooth, WLAN,
802.11 b/g/n, RFID 2450,
LTE-band 7
WLAN 802.11 a/n

Obs! Om det är nödvändigt att uppnå IMMUNITETSTESTNIVÅN kan avståndet mellan sändarantennen och Sprint/Stroller/
Hi Flow Stroller minskas till 1 m. Testavståndet på 1 m är tillåtet enligt IEC 61000-4-3.
För vissa tjänster inkluderas endast upplänksfrekvenser.
Bärvågen ska vara modulerad med en 50-procentig effektperiod med fyrkantsvågsignal.
Som alternativ till FM-modulering kan 50 % pulsmodulering vid 18 Hz användas eftersom det, även om det inte representerar faktisk
modulering, skulle vara den värsta situationen.
abc

CAIRE och CAIRE Inc. är registrerade varumärken som tillhör CAIRE Inc. Besök vår webbplats nedan för en fullständig lista över varumärken. Varumärken: www.careinc.com/corporate/
trademarks.
Copyright © 2023 CAIRE Inc. CAIRE Inc. förbehåller sig rätten att upphöra med försäljning av sina produkter eller ändra priser, material, utrustning, kvalitet, beskrivningar, specifikationer
och/eller processer för sina produkter när som helst utan förvarning och utan vidare förpliktelser eller följder. Vi förbehåller oss alla rättigheter som inte uttryckligen anges häri om tillämpligt.

PN MN236-C4 E | Bruksanvisning SE - 141

Sprint / Stroller / Hi Flow Stroller
Symbolien selitteet
ISO 7000

Älä tupakoi yksikön lähellä tai kun
käytät sitä. Sääntö P002

Säilytettävä kuivassa ja
pidettävä kuivana. Sääntö 0626

Tyypin BF potilaskosketuksessa
oleva osa (sähköiskusuojauksen taso).
Sääntö 5333

Pinoamisraja lukumäärän mukaan. Reg.
# 2403
Valmistajan nimi ja osoite. Sääntö 3082

Varoitus. Sääntö W001
Neuvoston direktiivi 93/42/ETY
Valtuutettu edustaja Euroopan
yhteisössä
Jos tuotteen yksilöivässä
tunnistemerkinnässä (UDI) on CEmerkki, laite on direktiivin 93/42 /ETY lääkinnällisiä laitteita koskevien
vaatimusten mukainen. Symboli CE#### tarkoittaa ilmoitetun laitoksen
numeroa.

Valmistusmaa ja päivämäärä. "CC" tunnistaa valmistusmaan kaksikirjaimisen
maakoodin. Valmistuspäivämäärä on
VVVV-KK-PP muodossa. Reg. # 6049
Vaara, katso mukana tulleita ohjeita.
Tuotenumero 0434A
Luettelonumero. Sääntö 2493

Tämä laite noudattaa lääkinnällisiä
laitteita koskevaa 2010/35/EUdirektiiviä. Siinä on kuvan mukainen
pi-merkintä.

Sarjanumero. Sääntö 2498
Varastointi- tai käyttölämpötila-alue.
Sääntö 0632

ADR

Varastoinnin kosteusalue. Sääntö 2620

Myrkytön kaasu.

Ilmanpainerajoitus. Sääntö 2621
Tämä puoli ylös. Sääntö 0623
Särkyvää, käsittele varoen. Sääntö 0621

ISO 7010

Lisäsymbolit

Vaara Hapettavat aineet: tulipaloa
voimistava vaara. Hapettavat aineet
aiheuttavat tulipalojen voimakkaamman
palamisen.
Jäähdytetty neste, USP: valmistettu
ilmanesteytyksellä

Sisältää vaarallisia aineita. Reg. # 3723

Pidä yksikkö aina hyvin tuuletettuna

Maahantuoja. Reg. # 3725

Pidä kaukana syttyvistä materiaaleista,öljystä ja rasvasta.

Kylmän nestemäisen tai kaasumaisen
hapen tai jäätyneiden osien
koskettaminen voi aiheuttaa paleltumia.
Alhaisen lämpötilan varoitus.
Varoittaa alhaisesta lämpötilasta tai
jäätymisvaarasta. Sääntö W010

Valtuutettu edustaja Sveitsissä.

Käyttöohjeet on luettava. Sääntö M002
Pidettävä kaukana avotulesta, tulesta ja
kipinöistä. Avoimet syttymislähteet ja
tupakointi kielletty. Sääntö P003

142 - FI PN MN236-C4 E | Käyttöopas

Jos laitteessa on UKCA-merkki kuvan
mukaisesti ja UKCA####, joka osoittaa
ilmoitetun laitoksen numeron, tämä laite
on UKCA-määräysten mukainen.
IEC 60417
Älä peitä yksikköä tai kanna kannettavaa
laitetta vaatteiden alla. Nämä yksiköt
yleensä poistavat happea. nro 5641

Sprint / Stroller / Hi Flow Stroller
21 CFR 801.15
Yhdysvaltojen liittovaltion laki sallii
tämän laitteen myynnin vain lääkärilletai lääkärin määräyksestä.
IEC 60601-1
Tippumiskestävä
Neuvoston direktiivi 2012/19/EU
WEEE
Tämä symboli muistuttaa laitteen omistajia palauttamaan se kierrätyslaitokseen
sen käyttöiän päätyttyä sähkö- ja elektroniikkalaiteromudirektiivin (WEEE)
mukaisesti.
Tuotteemme ovat vaarallisten aineiden(RoHS) direktiivin rajoitusten mukaisia.
Ne eivät sisällä enempää kuin pieniä
määriä lyijyä tai muuta vaarallista materiaalia.

ISO 15223-1
Lääketieteellinen laite. Reg. # 5.7.7
Ainutlaatuinen laitetunniste # 5.7.10
Tämä tuote saattaa olla suojattu yhdellä tai useammalla
patentilla Yhdysvalloissa ja muissa maissa. Alla
olevalla verkkosivullamme on lueteltu kaikki
sovellettavat patentit. Pat.: www.caireinc.com/
corporate/patents/.

Tekniset tiedot• Toimintatila: Jatkuva virtaus• Suojaus sähköiskuilta (tyyppi): Sisäisesti virran saava laite• Suojaus sähköiskuilta (luokka): Tyypin BF sovellettu osa• Luokitus veden sisääntulon suojausasteen mukaan: IP22 – tavallinen laitteisto• Laite ei sovellu käytettäväksi syttyvien seosten lähellä
Tuotteen tekniset tiedot

SprintStroller

Hi Flow Stroller

0,68 kg

1,36 kg

1,36 kg

Kaasua vastaava kapasiteetti 513 l
Paino tyhjänä
2,04 kg

1026 l2,27 kg

1025 l2,49 kg

Paino täytettynä

3,63 kg

3,86 kg

LOX-kapasiteetti

2,72 kg

Korkeus298 mm

343 mm

343 mmMitat

123,83 mm D x 191 mm W

149,23 mm D x 191 mm W

149 mm D x 191 mm W

Tyypillinen käyttöaika per
2 l/minToimintapaine

4,3 tuntia

8,5 tuntia

8,5 tuntia

20 psi (137 kPa)

20 psi (137 kPa)

20 psi (137 kPa)

Normaali haihtumisnopeus

0,57 kg/päivä

0,57 kg/päivä

0,57 kg/päivä

Off, 0,25; 0,5; 0,75; 1; 1,5; 2;
2,5; 3; 4; 5; 6 l/min+/- 10 %Off, .5, 1, 2, 2.5, 3, 4, 6, 8, 10,
12, 15 l/min+/- 10 %Normaali virtauksensäätöalue Off, 0,25; 0,5; 0,75; 1; 1,5; 2;
2,5; 3; 4; 5; 6 l/min
Virtausmittarin tarkkuus*+/- 10 %* Tämä tarkkuus on voimassa vain arvoilla 70F ja 14,7 psig ja kalibroidulla, tarkalla massan virtausanturilla.

PN MN236-C4 E | Käyttöopas FI - 143

Sprint / Stroller / Hi Flow Stroller
Varoitustiedot
Tärkeää: Lue tämä käyttöohje perusteellisesti ennen
kuin otat käyttöön Sprint/Stroller/Hi Flow Stroller-laitteen.
Vain RX.
VAROITUS: LAITETTA EI OLE TARKOITETTU
ELINTOIMINTOJEN YLLÄPITÄMISEEN.
VAROITUS: JOS LAITE EI MIELESTÄSI TOIMI OIKEIN,OTA YHTEYS TERVEYSPALVELUJEN TARJOAJAAN.
ÄLÄ YRITÄ KORJATA TAI SÄÄTÄÄ LAITETTA ITSE.
VAROITUS: ÄLÄ MUUTA LAITETTA ILMAN
VALMISTAJAN LUPAA.
VAROITUS: JOS HAPENSYÖTÖN JATKUVUUTTATARVITAAN, VARMISTA, ETTÄ RIITTÄVÄ
HAPENSAANTI JA/TAI TOISSIJAISEN HAPEN SAANTION AINA SAATAVILLA HOIDON AIKANA.
VAROITUS: ÄLÄ SALLI TUPAKOINTIA, KYNTTILÖITÄTAI AVOTULTA 3 METRIN ETÄISYYDELLÄ
LAITTEESTA TAI 20 CM:N ETÄISYYDELLÄ MISTÄÄN
SYTTYMISLÄHTEISTÄ.
VAROITUS: LAITTEEN ON OLTAVA HYVIN
TUULETETUSSA TILASSA.
VAROITUS: ÄLÄ SÄILYTÄ
NESTEHAPPILAITTEITA KAAPISSA, AUTONTAVARATILASSA TAI MUUSSA SULJETUSSA
TILASSA. ÄLÄ LAITA HUOPIA, VERHOJA TAIMUITA KANKAITA LAITTEEN PÄÄLLE.
VAROITUS: TÄMÄ TUOTE VOI ALTISTAA SINUTKEMIKAALEILLE, MUKAAN LUKIEN NIKKELI, JOKA
KALIFORNIAN OSAVALTION TIETOJEN MUKAANAIHEUTTAA SYÖPÄÄ. KATSO LISÄTIETOJA SIVULTA
WWW.P65WARNINGS.CA.GOV.
VAROITUS: JOS LAITTEESSA TAPAHTUUVAKAVA VAARATILANNE, KÄYTTÄJÄN TULEEILMOITTAA TAPAUKSESTA VÄLITTÖMÄSTIPALVELUNTARJOAJALLE JA / TAI VALMISTAJALLE.
VAKAVANA TAPAHTUMANA MÄÄRITELLÄÄNLOUKKAANTUMINEN, KUOLEMA TAI MAHDOLLISUUSAIHEUTTAA VAMMOJA / KUOLEMIA, JOS TAPAHTUMATOISTUU. KÄYTTÄJÄ VOI MYÖS ILMOITTAATAPAHTUMASTA SEN MAAN TOIMIVALTAISELLEVIRANOMAISELLE, JOSSA TAPAHTUMA TAPAHTUI.
Huomio: Käytä Sprint/Stroller/Hi Flow Stroller-laitetta vain lääkärisi ohjeen mukaan.
Huomio: Yhdysvaltojen liittovaltion laki sallii
tämän laitteen myynnin vain lääkärille tai lääkärin
määräyksestä.
Huomio: Laite sisältää nestemäistä happea, joka on erittäin kylmää, lähes -184 °C. Näin alhaiselle lämpötilalle
altistuminen saattaa aiheuttaa vakavia paleltumia.

144 - FI PN MN236-C4 E | Käyttöopas

Huomio: Nestemäinen ja kaasumainen happi eivät syty,
mutta ne voivat kiihdyttää muiden materiaalien palamista. Tämän vaaran sekä nestemäisen hapen matalan
lämpötilan vuoksi on noudatettava erityisiä varotoimia.
Huomio: Säilytä helposti syttyviä materiaaleja erillään
laitteesta. Aerosolisuihkeet, öljyt ja rasva, mukaan
lukien kasvovoiteet ja vaseliini, syttyvät helposti ja
voivat palaa nopeasti hapen läsnä ollessa.
Huomio: Tupakointi happikanyylin käyttämisen aikana
voi aiheuttaa palovammoja kasvoille ja mahdollisesti
kuoleman.
Irrotetun kanyylin asettaminen vaatteiden, vuodevaatteiden, sohvan tai muun tekstiilimateriaalin päälle voi
aiheuttaa räjähtävän palon, jos kanyyli altistuu savukkeelle, lämmönlähteelle, kipinälle tai avotulelle.
Jos tupakoit: (1) sammuta siirrettävä yksikkö, (2) irrota
kanyyli ja (3) poistu huoneesta, jossa laite sijaitsee.
Huomio: Jos laite kallistuu vahingossa, palauta seheti varovasti pystysuoraan asentoon, jos se on
mahdollista. Jos laitteesta vuotaa nestemäistä happea,
poistu alueelta välittömästi ja soita terveydenhuollon
palveluntarjoajallesi. Älä yritä siirtää yksikköä tai estää
nestemäistä happea pakenemasta.
Huomautus: Älä koske yksikköjen jäätyneisiin osiin.
Huomautus: Älä varastoi tai käytä siirrettävää yksikköä
yhdessä kiinteän yksikön kanssa.
Huomautus: Älä anna kouluttamattomien henkilöiden
käsitellä tai käyttää laitetta.
Huomautus: Federal Aviation Administration on kieltänyt
laitteen käytön kaupallisilla matkustaja- ja rahtilennoilla.

Käyttötarkoitus

CAIRE Sprint/Stroller/Hi Flow Stroller on tarkoitettu
lisähapen annosteluun. Laitetta ei ole tarkoitettu elämän
tukemiseen, eikä se tarjoa potilaan tarkkailuvalmiuksia.

Sprint / Stroller / Hi Flow Stroller
Johdanto

Säätimet

Sprint, Stroller ja Hi Flow Stroller on tarkoitettu
lisähapen annostelemiseen potilaalle loppukäyttäjän
kodissa, ja sitä voidaan käyttää myös sairaskodeissa ja
muissa ei-akuuttia hoitoa antavissa laitoksissa. Sprint/
Stroller/Hi Flow Stroller -laite toimii liikkuvana,
pitkäaikaisena hapen lähteenä.

1. Elektroninen nestetason mittari
2. Elektroninen nestetason kytkin
3. Mekaaninen nestetason mittari
4. Virtauksen säätönuppi
5. Happiputken liitin
6. Ilmausventtiili
7. Täyttöliitännät (pikairrotusventtiilin
tunnistuksesta ks. seuraavaa sivua)
8. Kantohihna, jossa olkapehmike
9. Tiivistyspehmike
10. Tiivistyskuppi

Laitetta käyttävät COPD-potilaat tai potilaat, joiden
hengityskapasiteetti on heikentynyt. Laite on määrätty
potilaalle. Laite myydään hoidonantajalle, joka on
koulutettu käyttämään ja huoltamaan Sprint/Stroller/
Hi Flow Stroller -laitetta. Hoidonantaja kouluttaa
käyttäjän.
Tätä laitetta ei ole tarkoitettu elämää
ylläpitäviin sovelluksiin, eikä siinä ole potilaan
monitorointiominaisuutta. On suositeltavaa, että
käytettävissä on vaihtoehtoinen lisähapen lähde
mekaanisen vian varalta. Näissä siirrettävissä laitteissa
on myös pikairrotusventtiili, joka voidaan yhdistää
nestemäisen hapen säilytysvälineeseen siirrettävän
laitteen täyttämiseksi. Säilytysvälineen täyttämisestä
vastaa hoidonantaja.

1
2
4
5
8

Tämä käyttöopas sisältää ohjeet Sprint/Stroller/
Hi Flow Stroller -laitteen käyttämiseksi. Lue kiinteän
yksikön toiminnasta sitä koskevasta käyttöoppaasta.
Huomautus: Palveluntuottaja auttaa alkuasennuksessa ja
ohjeistaa laitteen asianmukaiseen käsittelyyn ja käyttöön.

5

7
Sprint-, Stroller- ja Hi Flow Stroller -yksiköt

610
9
PN MN236-C4 E | Käyttöopas FI - 145

Sprint / Stroller / Hi Flow Stroller
Pikairrotusventtiilin(QDV) tunnistaminen
Siirrettävän
laitteen liitin

Kiinteänlaitteen liitin

QDV-tyyli

CAIREsivutäyttö

SF-lukitus
4. a. Jos liittimesi on päällepainettavaa tyyppiä:
Laske Sprint/Stroller/Hi Flow Stroller -yksikkö
kiinteälle yksikölle, kunnes liitin kiinnittyy
havaittavasti.

LIFE-OX
TF-lukitus
PENOX

TAEMA
TFpäällepainaminen
PB

Käyttöohjeet
Ylhäältä täytettävän Sprint/Stroller/Hi Flow
Stroller -laitteen täyttäminen
1. Puhdista sekä kiinteän laitteen että Sprint/Stroller/
Hi Flow Strollerin täyttöliittimet puhtaalla, kuivalla
ja nukkautumattomalla liinalla.
Huomio: Täyttöliittimien on oltava kuivat,
koska muuten Sprint/Stroller/Hi Flow Stroller
ja kiinteä yksikkö voivat sulaa yhteen.
2. Valmista, että virtauksen säätönupit sekä
Sprint/Stroller/Hi Flow Stroller -laitteessa että
säiliöyksikössä ovat off-asennossa (0).
3. Aseta täyttöliitin pystysuoraan asentoon kiinteän
yksikön liittimen yläpuolelle.

146 - FI PN MN236-C4 E | Käyttöopas

b. Jos liitin on Penox tai Life-Ox:
Pyöritä Sprint/Stroller/Hi Flow Stroller -laitetta,
kunnes liitin kiinnittyy havaittavasti. Pyöritä
sitten Sprint/Stroller/Hi Flow Stroller -laitetta
myötäpäivään () samalla, kun painat sitä
alaspäin.

Huomautus: Jatkuu vaiheesta 6 kohdasta Sprint/Stroller/Hi
Flow Stroller -laitteen sivutäyttö.

Sprint/Stroller/Hi Flow Stroller -laitteen
sivutäyttö
1. Puhdista sekä kiinteän laitteen että Sprint/Stroller/
Hi Flow Strollerin täyttöliittimet puhtaalla, kuivalla
ja nukkautumattomalla liinalla.
VAROITUS: TÄYTTÖLIITTIMIEN ON OLTAVAKUIVAT, KOSKA MUUTEN SPRINT/STROLLER/
HI FLOW STROLLER JA KIINTEÄ YKSIKKÖVOIVAT SULAA YHTEEN.
2. Valmista, että virtauksen säätönupit sekä
Sprint/Stroller/Hi Flow Stroller -laitteessa että
säiliöyksikössä ovat off-asennossa (0).

Sprint / Stroller / Hi Flow Stroller
3. Varmista, että Sprint/Stroller/Hi Flow Stroller-laitteen täyttöliitin on pystysuorassa kiinteän
laitteen liittimeen nähden.

4. Pyöritä Sprint/Stroller/Hi Flow Stroller -laitetta), kunnes pinne ja ura
vastapäivään (kiinnittyvät havaittavasti.

7. Saatat nähdä höyryä liitännän ympärillä. Tämä on
normaali ilmiö ja johtuu suuresta lämpötilaerosta
nestemäisen hapen ja yksiköitä ympäröivän
lämpimän ilman välillä.
8. Kun yksikkö on täynnä, sulje ilmausventtiili. Voit
todeta yksikön olevan täynnä, kun sihinän sävy
muuttuu ja ilmauslinjalta tulee ulos hiukan nestettä.
Huomautus: Jos venttiili suljetaan sen jälkeen, kun lähes
jatkuva nestevirta tulee ulos, Sprint/Stroller/Hi Flow Stroller-laite täyttyy mahdollisimman täydellisesti, mutta hiukan
nestemäistä happea menee hukkaan.

5. Pyöritä varoen, mutta vakaasti Sprint/Stroller/
Hi Flow Stroller -laite takaisin pystysuoraan
asentoon. Nyt yksiköt on lukittu yhteen.

Huomautus: Jatkuu vaiheessa 6.
Huomautus: Päälle painettavaa liitintä käytettäessä pidä
yksikköä paikallaan koko täytön aikana.

Huomautus: Jos venttiili suljetaan, kun nesteen
purkautumisesta näkyy ensimmäinen merkki, Sprint/
Stroller/Hi Flow Stroller -laite ei täyty yhtä tarkkaan, mutta
nestemäistä happea säästyy.
9. Päälle painettavilla liittimillä pidä irrotuspainiketta
painettuna, kunnes yksiköt irtautuvat. Kaikilla
muilla liitintyypeillä pyöritä yksikköä, kunnes se
irtautuu kiinteästä yksiköstä.

Siirrettävän laitteen irrotuspainike (vain päälle
painettavilla liittimillä)

6. Avaa ilmausventtiili kun alat täyttää Sprint/Stroller/
Hi Flow Stroller -laitetta. Tällöin saatat kuulla
pienen happimäärän karkaavan; älä huolestu.
Kuulemasi sihahtava ääni on täysin normaali.
Huomautus: Nestetasonmittarien lukemat ovat
tarkimmillaan 30 sekuntia tai enemmän sen jälkeen, kun
Sprint/Stroller/Hi Flow Stroller -laite on täytetty.
PN MN236-C4 E | Käyttöopas FI - 147

Sprint / Stroller / Hi Flow Stroller
Huomio: Tarkasta nestetason mittari vasta,
kun venttiili on kiinni.

Käyttöohjeet

Huomio: Jos Sprint/Stroller/Hi Flow Stroller -laitetta
ollaan täyttämässä välittömästi käyttöjakson jälkeen,
täytä se ja odota sitten 10 minuuttia täytön jälkeen
paineen vakiintumista niin, että FCV on asetettu nollalle.
Tämän jälkeen voit käyttää laitetta normaalisti.
VAROITUS: ÄLÄ KÄYTÄ SPRINT/STROLLER/
HI FLOW STROLLER -LAITETTA, KUN SE ON
KIINNITETTYNÄ KIINTEÄÄN YKSIKKÖÖN.
VAROITUS: JOS SPRINT/STROLLER/HI FLOWSTROLLER -LAITE EI IRTOA HELPOSTI, ÄLÄ KÄYTÄVOIMAA. YKSIKÖT VOIVAT OLLA JÄÄTYNEETTOISIINSA KIINNI. JÄTÄ YKSIKÖT KIINNI TOISIINSAJA ODOTA, ETTÄ NE LÄMPENEVÄT – SITTEN NE
IRTOAVAT HELPOSTI. ÄLÄ KOSKE JÄÄTYNEISIINOSIIN. VAROITUS: JOS VUOTO KIIHTYY
NESTEVIRRAKSI, POISTU ALUEELTA JA SOITA HETITERVEYDENHOIDON PALVELUNTARJOAJALLESI.
VAROITUS: NESTEMÄINEN HAPPI JA KYLMÄ
HAPPIKAASU VOIVAT AIHEUTTAA IHOLLAPALELTUMIA. HAPPI KASVATTAA TULIPALOVAARAA.
Huomio: Jos kiinteästä yksiköstä tai Sprint/
Stroller/Hi Flow Stroller -laitteesta vuotaa
nestettä yksiköitä toisistaan irrotettaessa,
aseta Sprint/Stroller/Hi Flow Stroller-laite erilleen, varmista, että se pysyy
pystysuorassa, poistu huoneesta ja ota
välittömästi yhteyttä hoidonantajaan.
Huomautus: Laitevalmistajalle: Seuraavia hapen annostelun lisävarusteita suositellaan käytettäväksi Sprint / Stroller/ Hi Flow Stroller kanssa:• nenänkanyyli, jossa on 7 jalkaa (2,1 m) letkua (suurin
LPM: 6 LPM): CAIRE-osanumero 5408-SEQ• Tulipalo: CAIRE-osanumero 21126636
Tulipalo suositellaan käytettäväksi minkä tahansa kanyylin
kanssa.
• CAIRE tarjoaa palomuurin, joka on tarkoitettu
käytettäväksi kannettavan hapen kanssa. Tulipalo on
lämpövaroke, joka estää kaasun virtauksen, jos alavirran
kanyyli tai happiputki sytytetään ja palaa paloputkeen. Se
sijoitetaan linjaan nenän kanyylin tai happiputken kanssa
potilaan ja Sprint / Stroller / Hi Flow Stroller happea
poistoaukon väliin. Tietoja palomuurin oikeasta käytöstä
on aina valmistajan ohjeissa (joka sisältyy jokaiseenpalopalorasiaan).
• Lisätietoja suositelluista lisävarusteista on saatavana
verkossa osoitteessa www.caireinc.com.

148 - FI PN MN236-C4 E | Käyttöopas(Kanyyli ei sisälly toimitukseen.)

1. Paina kanyylihengitysputki lujasti happiputken
liittimeen.
2. Säädä kanyylihengitysputkesi oikeaan asentoon
niin, että voit hengittää happea mukavasti.
3. Käännä virtauksen säätönuppia Sprint/Stroller/
Hi Flow Stroller -laitteen yläosassa myötäpäivään,
kunnes määräyksen mukainen virtausmäärä (luku)
näkyy nupin "ikkunassa" ja tuntuu selvä naksahdus.
Huomio: Epäyhtenäinen hapen virtaus
syntyy, jos virtauksen säätönuppia säädetään
virtausnopeuksien väliin.
4. Hapen pitäisi nyt virrata. Pysäytä hapen virtaus
kääntämällä virtauksen säätönuppi vastapäivään
off-asentoon (0).
Huomio: Käännä virtausnuppi aina off(0) -asentoon, kun laite ei ole käytössä.

Sprint / Stroller / Hi Flow Stroller
5. Käytä seuraavaa kaaviota ohjenuorana Sprint/
Stroller/Hi Flow Stroller -laitteen toiminta-ajan
pituuden määrittämiseksi:
FCV-asetus
MalliSprintStroller

Hi Flow
Stroller

Nimellisarvo

Nimellisarvo

FCVasento.

SprintStroller

Pois

30.00

60:00

71:00

0,25

18.00

30.00

Ei käytössä

0,5

12.12

23.00

32:00

0,75

9.18

17.24

Ei käytössä

1

7.30

14.06

16.00

1,5

5.30

10.12

Ei käytössä

2

4.18

8.00

8.00

2,5

3.36

6.36

6.20

3

3.06

5.36

5.20

4

2.24

4.24

4.00

5

2.00

3.36

Ei käytössä

Hi Flow
Stroller

6

1.42

3.00

2.40

8

Ei mitään

Ei mitään

2.00

10

Ei mitään

Ei mitään

1.40

12

Ei mitään

Ei mitään

1.20

15

Ei mitään

Ei mitään

1.00

Huomautus: Ajat ovat tunteina ja minuutteina(muodossa 00.00).
Huomautus: Nimellisarvoajat on tarkoitettu
ihanneolosuhteisiin, ts. maksimitäyttö, tarkat
virtausnopeudet, suosiollinen tappioaste,liikkumatta pidetty laite ym.
Nämä ajat ovat odotettuja maksimiarvoja.
Huomautus: Yksilölliset tulokset vaihtelevat.

• Gen 4 -mittari: Paina vihreää käyttöpainiketta
mittarin etupuolella. Lue sisällön taso
LED‑valojen kaarelta.

Huomio: Sprint/Stroller/Hi Flow Stroller laite on tyhjä, jos vain valotangon viimeinensegmentti on valaistu.
Jos heikon pariston merkkivalo syttyy, kun
painiketta painetaan, ilmoita hoidonantajallesi.
Huomautus: Sprint/Stroller/Hi Flow Stroller -laite jatkaa
hapen syöttämistä siinäkin tapauksessa, että heikonpariston merkkivalo palaa, niin kauan kuin yksikössä on
nestemäistä happea.
6b. Nestemäisen hapen täyttötason todentaminen
yksikössä mekaanisen nestetason mittarin avulla:• Pidä yksikköä tuettuna kantohihnalla aivan mustan
sisältöilmaisimen jousivaakakammion yläpuolella.
• Anna yksikön vakaantua (niin että se ei kimpoile ylös
ja alas), lue sitten näkyviin tulevalta väriasteikolta
nestemäisen hapen määrän likiarvo.

6A. Nestemäisen hapen tason todentaminen yksikössä
elektronisella nestetason mittarilla:• Gen 3 -mittari: Pidä painonappia (nestetason
kytkintä) yksikön yläosassa painettuna ainakinkahden sekunnin ajan. Lue sisällön taso valotangon
poikkisuunnassa.

Huomautus: Jos yksikkö on tyhjä, vain punaiseksi väritetty
asteikon nauha näkyy. Jos tilanne on tämä, yksikkö on
täytettävä ennen käyttöä.
PN MN236-C4 E | Käyttöopas FI - 149

Sprint / Stroller / Hi Flow Stroller
7. Tietynlaisissa ympäristöoloissa, jatkuvan käytön
myötä tai erityisesti jätettäessä siirrettävä laite
liikuttamatta höyrystinkäämin ja kotelon päälle
voi syntyä runsaasti huurretta. Huurre voidaan
poistaa naputtamalla laitetta ja/tai pyyhkimällä
huurrekeräymät pois kotelolta.

Tiivistyspehmikkeen huolto
Kun laite on tyhjä ja huoneenlämpöinen, poista kaikki
kosteus tiivistyskupista:
Kovalle kotelolle• Kallista laitetta toiselle puolelle. Tartu
tiivistyskupin keskellä olevaan nuppiin (yksikön
pohjassa) ja vedä se ulos. Tällöin kuppi irtoaa.
Huomautus: Jos pehmike likaantuu pahasti, se voidaan
pestä koneessa (ja desinfioida valkaisuaineella).
• Sijoita kuppi takaisin paikalleen keskittämällä
se yksikön pohjassa olevaan pyöreään aukkoon,
napsauta nuppia kerran kupin kiinnittämiseksi ja
toisen kerran sen lukitsemiseksi paikalleen.

Huomautus: Jos pehmike likaantuu pahasti, se voidaan
pestä koneessa (ja desinfioida valkaisuaineella).
• Palauta pehmike paikalleen asettamalla sepohjakanteen ja vetämällä pohjan vetoketju
taas kiinni.
Huomautus: Jos nuppi ei suostu kiinnittämään kuppia, pidä
kuppia kiinni ja vedä nuppi ulos, kunnes se napsahtaa
auki-asentoon. Sijoita nyt kuppi paikalleen edellä kuvatulla
tavalla.

Huurtumisen rajoittaminen
Tietynlaisissa ympäristöoloissa, jatkuvan käytön myötä
tai erityisesti jätettäessä siirrettävä laite liikuttamatta
höyrystinkäämin ja kotelon päälle voi syntyä runsaasti
huurretta. Huurre voidaan poistaa naputtamalla laitetta
ja/tai pyyhkimällä huurrekeräymät pois kotelolta.
VAROITUS: KOSTEUDEN/JÄÄN RUNSAANKERTYMISEN RISKIN VUOKSI ON
SUOSITELTAVAA, ETTÄ HI FLOW STROLLER-LAITE JÄTETÄÄN 1 TUNNIKSI KÄYTÖN
JÄLKEEN POIS PÄÄLTÄ KOSTEUDEN
KERÄÄNTYMISEN VÄHENTÄMISEKSI.

Lukossa
Kiinnittynyt
Auki

Nuppi ja kuppi

Huomautus: Jos nuppi ei suostu kiinnittämään kuppia, pidä
kuppia kiinni ja vedä nuppi ulos, kunnes se napsahtaa
auki-asentoon. Sijoita nyt kuppi paikalleen edellä kuvatulla
tavalla.
Pehmeälle kotelolle• Vedä auki pohjassa oleva vetoketju ja vedä
taaksepäin pohjakansi.
• Poista pehmike kupilta ja väännä se kuivaksi.
Anna pehmikkeen kuivua kokonaan ennen kuin
asetat sen ja kupin takaisin paikalleen.

150 - FI PN MN236-C4 E | KäyttöopasHi Flow Stroller -laitteen suurempien virtausarvojen
vuoksi laitteessa esiintyy veden tiivistymistä jahuurteenmuodostusta erityisesti, kun virran asetus
on 10 l/min tai enemmän. On suotavaa antaa laitteenolla ainakin 1 tunnin ajan poissa toiminnasta käytön
jälkeen, jotta jatkuva käyttö ei aiheuttaisi kosteuden
ylenmääräistä kertymistä.

Sprint / Stroller / Hi Flow Stroller
Vianmääritys
Seuraavat ohjeet auttavat Sprint/Stroller/Hi Flow Stroller -laitteen vianmäärityksessä ja yksinkertaisten
toimintahäiriöiden korjaamisessa.
OngelmaSiirrettävä yksikkö sihisee.

Siirrettävän yksikön virta
katkeaa käytön aikana.

Ratkaisu• Sihinä voi olla merkki siirrettävän yksikön käyttöpaineen säätymisestä. Sihinää esiintyy
yleensä täytön jälkeen tai silloin, kun siirrettävän yksikön asento on muuttunut. Sihinää
voi kestää noin 10 minuuttia täytön jälkeen. Sitä voi esiintyä myös, kun virtausventtiili on
pienellä asetuksella.
• Jos siirrettävä yksikkö on asetettu virheelliseen asentoon, palauta siirrettävä yksikkö
pystyasentoon ja anna sen vakautua joitakin minuutteja.
• Varmista, että kanyyli on liitetty tiukasti hapen lähtöliitäntään.
• Varmista, ettei kanyyli ole kiertynyt.
• Tarkasta sisältöilmaisin/tasomittari ja täytä siirrettävä yksikkö, jos tarpeen.

Siirrettävä yksikkö ei täyty.

Siirrettävän yksikön
tuuletusventtiili ei sulkeudu
kunnolla täytön lopuksi.

Siirrettävä yksikkö ei irtoa
helposti säiliöstä täytön
jälkeen.

• Varmista, että virtauksen säätönuppi ei ole off-asennossa (”0”).
• Varmista, että säiliössä on happea.
• Varmista, että siirrettävän yksikön ja säiliön täyttöliittimet ovat kunnolla paikoillaan koko
täytön ajan.
• Jos tuuletusventtiili ei sulkeudu ja jos sihinää ja happihöyryä ilmenee edelleen, irrota
siirrettävä yksikkö varovasti painamalla säiliön vapautuspainiketta. Tuuletus siirrettävän
yksikön pohjan kautta loppuu muutamassa minuutissa. Anna yksikön lämmetä, jotta
voit sulkea tuuletusventtiilin. Siirrettävän yksikön paineen palautuminen voi kestää jopa
60 minuuttia, kunnes happivirtauksen vaatima paine on saavutettu. Tarvittaessa käytä
vaihtoehtoista happilähdettä, kuten säiliöön kiinnitettyä virtausventtiiliä.
• Siirrettävän yksikön ja säiliön täyttöliittimet ovat ehkä jäätyneet.
• ÄLÄ KÄYTÄ VOIMAA. Anna jäätyneiden osien lämmetä muutaman minuutin ajan ja irrota
siirrettävä yksikkö, kun jää on sulanut. Jotta yksiköt eivät jäädy kiinni toisiinsa, pyyhi
säiliön urostäyttöliitin ja siirrettävän yksikön naarastäyttöliitin puhtaalla kuivalla liinalla
ennen täyttöä.

PN MN236-C4 E | Käyttöopas FI - 151

Sprint / Stroller / Hi Flow Stroller
Puhdistusohjeet
VAROITUS: PUHDISTA VASTA, KUN YKSIKKÖON TYHJÄ.
• Puhdista käyttämällä mietoa astianpesuainetta
ja vettä.
• Käytä puhdistusliuosta nukkaamattoman liinan
avulla. Hyväksyttyihin puhdistusaineisiin
kuuluvat HydroPure ja HydroKlean. Älä
ruiskuta puhdistusaineita suoraan Sprint/
Stroller/Hi Flow Stroller -laitteelle.
• Pyyhi ulkopinta puhtaaksi nukkaamattomalla
liinalla.
Huomio: Älä käytä korkeaa lämpötilaa tai
korkeapainepesuria näiden yksiköiden
puhdistamiseen.
• Puhdistusainetta ei saa mennä sisäosiin tai
venttiileihin.
• Anna yksikön kuivua kunnolla ennen käyttöä.
Huomautus: Huomautus terveydenhuollon
ammattihenkilölle - jälleenkäsittelymenetelmistä on annettu
ohjeet kyseisessä huoltokäsikirjassa.

HuoltoPalveluntarjoajasi vastaa kaikista tämän laitteen
teknisen käyttöoppaan edellyttämästä huollosta. Jos
laitetta on huollettava, ota yhteys huoltoon.
Tiivistyspehmike on ainoa käyttäjän huollettavissa
oleva osa. Mitkään muut osat eivät ole käyttäjän
huollettavissa.
Odotettavissa oleva käyttöikä on vähintään viisi vuotta.
Puhdista sekä paikallaan olevan että siirrettävän
yksikön liittimet puhtaalla kuivalla nukkaamattomalla
liinalla täyttöjen välillä jäätymisen ja laitevikojen
välttämiseksi.
Huomautus: Muut huoltotoimet on jätettävä
ammattitaitoisen huoltoasentajan tai palveluntarjoajan
tehtäväksi.

Tarvikkeet

WEEE ja RoHS

Tämä symboli muistuttaa laitteen
omistajaa siitä, että elinkaarensa
lopussa laite on kierrätettävä sähkö- ja
elektroniikkalaiteromudirektiivin (WEEE) mukaisesti.
Tuotteemme täyttävät vaarallisia aineita koskevan
RoHS-direktiivin vaatimukset. Tuotteissamme on vain
vähäisiä jäämiä lyijystä ja muista vaarallisista aineista.

Hävittäminen
Palauta aina Sprint/Stroller/Hi Flow Stroller -laite
kaikkine rakenneosineen hoidonantajalle asianmukaista
hävittämistä varten. Voit myös ottaa yhteyttä
paikalliseen kaupungin tai kunnan viranomaiseen ja
kysyä ohjeita akun asianmukaisesta hävittämisestä.

Kuljetus ja säilytys
Laite on säilytettävä pystyasennossa ja hyvintuulettuvassa tilassa. Älä aseta laitetta kyljelleen.
Kosteus maks. 95 % ei kondensoituva. Lämpötila-alue-40–70 °C.
Käyttölämpötila-alue -10–40 °C. Suhteellinen kosteus
alueella 30–75 % ei kondensoituva.
Huomautus: Ilmanpainealue on 700–1060 hPa (korkeus
300 m...-300 m).

152 - FI PN MN236-C4 E | Käyttöopas

Kuljetuskärry PN 20765565

Sprint / Stroller / Hi Flow Stroller
Turvallisuus
VAROITUS: KANNETTAVIA RADIOTAAJUUSLAITTEITA (MUKAAN LUKIEN OHEISLAITTEET, KUTEN
ANTENNIKAAPELIT JA ULKOISET ANTENNIT) EI SAA KÄYTTÄÄ ALLE 30 CM:N ETÄISYYDELLÄ MISTÄÄNSPRINT/STROLLER/HI FLOW STROLLER -LAITTEEN OSASTA, MUKAAN LUKIEN VALMISTAJAN
MÄÄRITTÄMÄT KAAPELIT. MUUTOIN SEURAUKSENA VOI OLLA TÄMÄN LAITTEEN SUORITUSKYVYNHEIKENTYMINEN.
VAROITUS: MUIDEN KUIN LAITTEEN VALMISTAJAN TOIMITTAMIEN LISÄVARUSTEIDEN, ANTUREIDEN JA
KAAPELEIDEN KÄYTTÖ VOI LISÄTÄ LAITTEEN SÄHKÖMAGNEETTISIA PÄÄSTÖJÄ TAI HEIKENTÄÄ SEN
SÄHKÖMAGNEETTISTA HÄIRIÖNSIETOKYKYÄ JA AIHEUTTAA VIRHEELLISTÄ TOIMINTAA.
VAROITUS: TÄMÄN LAITTEEN KÄYTTÖÄ MUIDEN LAITTEIDEN VIERESSÄ TAI PINOTTUNA MUIDEN LAITTEIDENKANSSA PITÄÄ VÄLTTÄÄ, KOSKA SEURAUKSENA VOI OLLA VIRHEELLINEN TOIMINTA. JOS TÄLLAINEN KÄYTTÖ
ON VÄLTTÄMÄTÖNTÄ, TÄTÄ LAITETTA JA MUITA LAITTEITA PITÄÄ TARKKAILLA, JOTTA VOIDAAN VARMISTAA,ETTÄ NE TOIMIVAT NORMAALISTI.
Huomio: Sähköiset lääkinnälliset laitteet tarvitsevat erityisiä varotoimia sähkömagneettisenyhteensopivuuden (EMC) suhteen, ja ne on asennettava ja otettava käyttöön tässä käsikirjassa annettuja
EMC-määräyksiä noudattaen.
Huomio: Kannettavat ja langattomat radiotaajuuslaitteet (RF) voivat vaikuttaa lääkinnällisiin sähkölaitteisiin.
Taulukko 1

Valmistajan vakuutus - sähkömagneettiset häiriöt
Sprint/Stroller/Hi Flow Stroller -laite on suunniteltu käytettäväksi alla selostetun mukaisessa
sähkömagneettisessa ympäristössä. Asiakkaan tai Sprint/Stroller/Hi Flow Stroller -laitteen käyttäjän tulisi
varmistaa, että sitä käytetään sellaisessa ympäristössä.

Häiriöpäästötesti

Vastaavuus

Sähkömagneettinen ympäristö – ohjeistus

Radiotaajuussäteily

Ryhmä 1

Sprint/Stroller/Hi Flow Stroller -laite käyttää
radiotaajuusenergiaa vain sisäiseen toimintaansa.

CISPR 11

Tämän vuoksi radiotaajuussäteily on hyvin matalaa eikä
se todennäköisesti häiritse lähellä olevia elektronisia
laitteita.

RadiotaajuuspäästötCISPR 11

Luokka B

Harmonisoidut päästöt
IEC 61000-3-2

Ei sovelleta

Jännitteenvaihtelut/kohinapäästöt
IEC 61000-3-3

Ei sovelleta

Sprint/Stroller/Hi Flow Stroller -laite soveltuu
käytettäväksi kaikissa tiloissa, kuten kotona ja tiloissa,
jotka on liitetty suoraan julkiseen pienjänniteverkkoon,
joka syöttää sähköä kotitalouksille tarkoitettuihin
rakennuksiin.

PN MN236-C4 E | Käyttöopas FI - 153

Sprint / Stroller / Hi Flow Stroller
Taulukko 2*: Suositellut erotusetäisyydet siirrettävän ja liikuteltavan radiotaajuisen viestintälaitteen ja Sprint/Stroller/Hi Flow Stroller -laitteen välillä
Sprint/Stroller/Hi Flow Stroller -laite on tarkoitettu käytettäväksi sellaisessa elektromagneettisessa ympäristössä, jossa radiotaajuisia
häiriöitä valvotaan. Asiakas eli Sprint/Stroller/Hi Flow Stroller -laitteen käyttäjä voi auttaa estämään sähkömagneettisia häiriöitä
säilyttämällä alla olevan suosituksen mukaisen vähimmäisvälimatkan kannettavan, siirtyvän radiotaajuuslaitteen (lähettimet) ja
Sprint/Stroller/Hi Flow Stroller -laitteen välillä viestintälaitteen enimmäistehon mukaan.
Lähettimen suurin
Etäisyys lähettimen taajuuden mukaisesti
nimellislähtöteho
m
80 MHz – 800 MHz
800 MHz – 2,5 GHz
150 kHz – 80 MHzW
d = 1,2√P
d = 2,3√P
d = 1,2√P
0,01
0,12 m0,12 m0,23 m0,10,38 m0,38 m0,73 m
1
1,2 m
1,2 m
2,3 m
10
3,8 m
3,8 m
7,3 m
100
12 m
12 m
23 m
Jos lähettimen suurinta lähtötehoa ei ole mainittu alla, suositeltu etäisyys d metreinä (m) voidaan arvioida yhtälöllä lähettimen taajuuden perusteella, kun P on lähettimen suurin lähtöteho watteina (W) lähettimen valmistajan ilmoituksen mukaisesti.
HUOMAA 1 Korkean taajuusalueen etäisyys pätee tehoilla 80 MHz ja 800 MHz.
HUOMAA 2 Tämä ohjeistus ei ehkä päde kaikissa tilanteissa. Sähkömagneettiseen johtumiseen vaikuttavat pintojen, esineiden ja
ihmisten aiheuttama imeytyminen ja heijastuminen.

* Tämä taulukko on standardin vaatimus laitteille, jotka on testattu tiettyihin testitasoihin ja tiettyihintaajuusalueisiin ja jotka on todettu sääntöjen mukaisiksi.

Taulukko 3: Valmistajan vakuutus - sähkömagneettiset häiriöt
Sprint/Stroller/Hi Flow Stroller -laite on suunniteltu käytettäväksi alla selostetun mukaisessa sähkömagneettisessa ympäristössä.
Asiakkaan tai Sprint/Stroller/Hi Flow Stroller -laitteen käyttäjän tulisi varmistaa, että sitä käytetään sellaisessa ympäristössä.
Immuniteettitestaus IEC 60601 - testaustaso
Vastaavuustaso
Sähkömagneettinen ympäristö – ohjeistus
Lattian tulee olla puuta, betonia tai keraamista
Sähköstaattinen±8 kV kontakti±8 kV kontakti
laattaa. Jos lattia on peitetty synteettisellä
purkaus (ESD)± 2 kV, ± 4 kV,± 2 kV, ± 4 kV,materiaalilla, suhteellisen kosteuden on oltava±8 kV, ± 15 ilma
±8 kV, ± 15 ilma
IEC 61000-4-2
vähintään 30 %.**
Ei sovelleta
Tasavirtakäyttöinen laite

Nopeat sähköiset
transientit/purskeet
IEC 610004-4±2 kV tulojohdoille

Syöksyaalto±1 kV johtojen välillä±2 kV johdoista maahan

Ei sovelleta
Tasavirtakäyttöinen laite

Ei sovelleta<5 % UT (>95 %:n pudotus
UT:ssä) 0,5 jakson ajan
40 % UT (60 %:n pudotus
UT:ssä) 5 jakson ajan
70 % UT (30 %:n pudotus
UT:ssä) 25 jakson ajan
<5 % UT (>95 %:n pudotus
UT:ssä) 5 sekunnin ajan

Ei sovelleta
Tasavirtakäyttöinen laite

Ei sovelleta

3A/m
50/60 Hz

3A/m
50/60 Hz

Tehotaajuusmagneettikenttien tulisi olla
tyypillisen kaupallisen tai sairaalaympäristön
mukaisia.

IEC 61000-4-5
Jännitekuopat, lyhyet
häiriöt ja jännitteenvaihtelut virranjakeluverkon tulojohdoissa
IEC 61000-4-11
Verkkotaajuuden(50/60 Hz) magneettikenttä
IEC 61000-4-8±1 kV for tulo-/lähtöjohdoille

Ei sovelleta

Ei sovelleta

Ei tietoa tulo-/lähtöjohdoista

Huomautus: UT tarkoittaa AC-verkkojännitettä ennen testaustason soveltamista.
** Tämä ilmoitus tarkoittaa, että vaadittu testi on suoritettu valvotussa ympäristössä ja että Sprint/Stroller/Hi Flow
Stroller -laitteen on todettu vastaavan määräyksiä.

154 - FI PN MN236-C4 E | Käyttöopas

Sprint / Stroller / Hi Flow Stroller
Taulukko 4

Ohjeistus ja valmistajan vakuutus – ME-laitteiston ja ME-järjestelmien immuniteetti

Ohjeistus ja valmistajan vakuutus – immuniteetti
Sprint/Stroller/Hi Flow Stroller -laite on suunniteltu käytettäväksi alla selostetun mukaisessa
sähkömagneettisessa ympäristössä. Asiakkaan tai Sprint/Stroller/Hi Flow Stroller -laitteen käyttäjän tulisi
varmistaa, että sitä käytetään sellaisessa ympäristössä.
Immuniteettitestaus
IEC 60601 -testaustaso Vastaavuustaso
Sähkömagneettinen ympäristö – ohjeistus
Johtava radiotaajuus RF
IEC 61000-4-6

3 Vrms6 Vrms (ISM-taajuusalueet)
150 kHz–80 MHz

Jos kannettavia ja mobiileja radiotaajuuslaitteita
käytetään Sprint/Stroller/Hi Flow Stroller -laitteen
jonkin osan lähellä, mukaan lukien kaapelit, etäisyyden
on oltava vähintään kyseisen lähettimen taajuuden
perusteella laskettu suositeltu etäisyys.

Ei sovelleta

Akkukäyttöinen laite,
ei SIP/SOP

Suositeltu etäisyys
d = 1,2 √P

Säteilevä radiotaajuus
IEC 61000-4-3

80 MHz–2,7 GHz

10 V/m80 MHz–2,7 GHz
80 % AM taajuudella 1 kHz

d = 1,2 √P
d = 2,3 √P
jossa P on lähettimen suurin lähtöteho watteina (W)lähettimen valmistajan ilmoituksen mukaisesti ja d on
suositeltu etäisyys metreinä (m).
Kiinteiden radiotaajuuslähettimien kenttien
voimakkuuksien (paikan sähkömagneettisen
mittauksena mukaisesti) on oltava alhaisempia kuin
kunkin taajuusalueenb säädöstenmukaisen tason.
Häiriöitä voi esiintyä, jos lähellä on seuraavalla
symbolilla merkittyjä laitteita:

Testaustaajuus(MHz)

Aluea)(MHz)

Huoltoa)

Modulointib)

385

380–390TETRA 400

1,8

0,3

27

450

430–470

GMRS 460, FRS 460

Pulssimodulointib)
18 Hz
FMC)
±5 kHz poikkeama
1 kHz sinus

2

0,3

28

704–787

LTE-alue 13, 17

Pulssimodulointib)
217 Hz

0,2

0,3

9

Pulssimodulointib)
18 Hz

2

0,3

28

Pulssimodulointib)
217 Hz20,328

Pulssimodulointib)
217 Hz20,328

Pulssimodulointib)
217 Hz

0,2

0,3

9

710
745
780
810
870
930
1720
1845
1970

GSM 800/900, TETRA
800–960 800, IDEN 820, CDMA
850, LTE-alue 5

2450

2400–
2570

5240
5500
5785

GSM 1800; CDMA
1900; GSM 1900;
DECT; LTE-alueet
1, 3, 4, 25; UMTS
Bluetooth, WLAN,
802.11 b/g/n, RFID
2450, LTE-alue 7

5100–
5800

WLAN 802,11 a/n

1700–
1900

Enimmäisteho Etäisyys(W)(m)

Immuniteettitestaustaso(V/m)

Huomautus: Jos tämä on IMMUNITEETIN TESTAUSTASON saavuttamiseksi välttämätöntä, etäisyys lähetysantennin ja
Sprint/Stroller/Hi Flow Stroller -laitteen välillä voidaan vähentää 1 m:iin. IEC 61000-4-3 sallii 1 m testausetäisyyden.
Eräiden palvelujen osalta vain nousevan siirtotien taajuudet on sisällytetty.
Kantoaaltoa tulee moduloida 50 % pulssisuhteen neliöaaltosignaalia.
Vaihtoehtona FM-moduloinnille voidaan käyttää 50 % pulssimodulaatiota 18 Hz:n taajuudella, koska vaikka se ei ole todellisuudessa
käytettävä modulointi, se toimii pahimman vaihtoehdon skenaariona.
abc

CAIRE ja CAIRE Inc. ovat CAIRE Inc: n rekisteröityjä tavaramerkkejä. Katso täydellinen luettelo tavaramerkeistä alla olevalta verkkosivustoltamme. Tavaramerkit: www.careinc.com/
corporate/trademarks.
Tekijänoikeus © 2022 CAIRE Inc. pidättää oikeuden lakkauttaa tuotteitaan tai muuttaa hintoja, materiaaleja, laitteita, laatua, nimityksiä, teknisiä tietoja ja/tai tuotteiden valmistustapoja milloin
tahansa ilman ennakkoilmoitusta ja ilman, että tästä aiheutuu sille lisävelvoitteita tai seuraamuksia. Kaikki oikeudet, joita tässä ei ilmaista suoraan, pidätetään soveltuvissa määrin.

PN MN236-C4 E | Käyttöopas FI - 155

Sprint / Stroller / Hi Flow StrollerΓλωσσάριο συμβόλων
ISO 7000Μην καπνίζετε κοντά στη μονάδα ήκατά τη διάρκεια της λειτουργίας τηςμονάδας. Reg. # P002
Εφαρμοζόμενο μέρος τύπουBF (βαθμός προστασίας απόηλεκτροπληξία). Reg. # 5333

Φυλάσσετε μακριά από τη βροχή,διατηρείτε στεγνό. Reg. # 0626
Όριο οριοθέτησης κατά αριθμό. Reg.
# 2403Όνομα και διεύθυνση τουκατασκευαστή. Reg. # 3082
Χώρα και ημερομηνία κατασκευής. Το
"CC" προσδιορίζει τον κωδικό χώρας
δύο γραμμάτων της χώρας κατασκευής.
Η ημερομηνία κατασκευής είναι σεμορφή ΕΕΕΕ-ΜΜ-ΗΗ. Καν. #6049
Προσοχή, συμβουλευτείτε τασυνοδευτικά έγγραφα. Reg. # 0434AΑριθμός καταλόγου. Reg. # 2493
Σειριακός αριθμός. Reg. # 2498
Περιοχή θερμοκρασίας αποθήκευσης ήλειτουργίας. Reg. # 0632

Προειδοποίηση. Reg. # W001
Οδηγία του Συμβουλίου 93/42/ΕΟΚΕξουσιοδοτημένος αντιπρόσωπος στηνΕυρωπαϊκή ΚοινότηταΕάν η ετικέτα μοναδικού
αναγνωριστικού συσκευής (UDI)φέρει το σύμβολο CE ####, η συσκευή
συμμορφώνεται με τις απαιτήσεις τηςοδηγίας 93/42 / ΕΟΚ σχετικά με ταιατροτεχνολογικά προϊόντα. Το σύμβολοCE #### υποδεικνύει τον κοινοποιημένο
αριθμό σώματος.
Αυτή η συσκευή συμμορφώνεται με τιςαπαιτήσεις της Οδηγίας 2010/35/ΕΕ για
τα ιατροτεχνολογικά προϊόντα. Φέρει τησήμανση Π όπως φαίνεται στην εικόνα.
ADRΕύρος υγρασίας αποθήκευσης. Reg. #2620Μη τοξικό αέριο.
Κίνδυνος Οξειδωτικές ουσίες: κίνδυνος
εντατικοποίησης της πυρκαγιάς. Οιοξειδωτικοί παράγοντες κάνουν τιςπυρκαγιές να καίγονται πιο έντονα.
Υγρό υπό ψύξη, USP· παράγεται μέσωυγροποίησης του αέραΠεριορισμός ατμοσφαιρικής πίεσης. Reg.
# 2621Με αυτή την όψη προς τα πάνω.
Reg. # 0623
Εύθραυστο, χειριστείτε με προσοχή.
Reg. # 0621

Πρόσθετα σύμβολαΔιασφαλίστε τον καλό αερισμό τηςμονάδας ανά πάσα στιγμή

Περιέχει επικίνδυνες ουσίες. Καν. #3723

Μακριά από εύφλεκτα υλικά, λάδια καιγράσο.

Εισαγωγέας. Καν. # 3725
ISO 7010

Εξουσιοδοτημένος αντιπρόσωπος στην
Ελβετία.

Σε περίπτωση επαφής με ψυχρό
υγρό ή αέριο οξυγόνο, ή παγωμένα
μέρη, ενδέχεται να προκληθούν
κρυοπαγήματα. Προειδοποίηση χαμηλής
θερμοκρασίας. Προειδοποιεί σχετικά
με συνθήκες χαμηλής θερμοκρασίας ήπαγώματος. Reg. # W010
Απαιτείται να διαβαστεί το εγχειρίδιο
οδηγιών. Reg. # M002
Μακριά από γυμνές φλόγες, φωτιά,σπίθες. Απαγορεύονται οι ανοιχτές πηγές
ανάφλεξης και το κάπνισμα. Reg. # P003

156 - GR PN MN236-C4 E | Εγχειρίδιο χρήστη

IEC 60417

Εάν η συσκευή φέρει το σήμα UKCA
όπως φαίνεται με το UKCA####που υποδεικνύει τον αριθμό τουειδοποιημένου οργανισμού, αυτή ησυσκευή συμμορφώνεται με τουςκανονισμούς UKCA.
Μην καλύπτετε τη μονάδα και μηνμεταφέρετε τη φορητή μονάδα κάτω απότα ρούχα σας. Αυτές οι μονάδες συνήθως
εξαερώνουν οξυγόνο. Νο. 5641

Sprint / Stroller / Hi Flow Stroller
21 CFR 801.15
Η ομοσπονδιακή νομοθεσία τωνΗΠΑ επιτρέπει την πώληση αυτής τηςσυσκευής μόνο από ιατρό ή κατόπιν
εντολής ιατρού.
IEC 60601-1
Απόδειξη σταγόνων
Οδηγία του Συμβουλίου 2012/19/ΕΕ
WEEEΑυτό το σύμβολο υπενθυμίζει στους ιδιοκτήτες του εξοπλισμού να τον επιστρέψουν σε εγκατάσταση ανακύκλωσης στοτέλος της ζωής του, σύμφωνα με τηνΟδηγία για τα Απόβλητα Ηλεκτρικού καιΗλεκτρονικού Εξοπλισμού (ΑΗΗΕ).
Τα προϊόντα μας θα συμμορφώνονται μετην οδηγία περιορισμού των επικίνδυνων ουσιών (RoHS). Δεν θα περιέχουν
περισσότερες από ελάχιστες ποσότητες
μολύβδου ή άλλου επικίνδυνου υλικού.

ISO 15223-1Ιατρική συσκευή. Καν. # 5.7.7Μοναδικό αναγνωριστικό συσκευής #5.7.10Αυτό το προϊόν μπορεί να καλύπτεται από έναή περισσότερα διπλώματα ευρεσιτεχνίας, τωνΗΠΑ και διεθνή. Παρακαλούμε επισκεφθείτε
τον δικτυακό τόπο της εταιρείας μας παρακάτω
για τον κατάλογο των εφαρμοστέων διπλωμάτων
ευρεσιτεχνίας. Διπλώματα ευρεσιτεχνίας: www.
caireinc.com/corporate/patents/.

Προδιαγραφές
• Τρόπος λειτουργίας: Συνεχούς ροής• Τύπος προστασίας από ηλεκτροπληξία: Εσωτερικά τροφοδοτούμενος εξοπλισμός• Βαθμός προστασίας από ηλεκτροπληξία: Εφαρμοζόμενο εξάρτημα τύπου BF• Ταξινόμηση σύμφωνα με τον βαθμό προστασίας από την είσοδο νερού: IP22 – Συνήθης εξοπλισμός• Εξοπλισμός ακατάλληλος για χρήση υπό την παρουσία εύφλεκτων μειγμάτων

Χαρακτηριστικά προϊόντος

SprintStroller

Hi Flow StrollerΧωρητικότητα LOX

0,68 kg (1,5 lb)

1,36 kg (3,0 lb)

1,36 kg (3,0 lb)Ισοδύναμη χωρητικότητα
σε αέριοΒάρος, άδειο

513 L

1026 L1025 L

2,04 kg (4,5 lb)

2,27 kg (5,0 lb)

2,49 kg (5,5 lb)Βάρος, γεμάτο

2,72 kg (6,0 lb)

3,63 kg (8,0 lb)

3,86 kg (8,5 lb)

Ύψος

298 mm (11,75 in.)

343 mm (13,5 in.)

343 mm (13,5 in.)

123,83 mm (4,875 in.) Β x
191 mm (7,5 in.) ΠΤυπικός χρόνος χρήσης στα 4,3 ώρες2 LPM
Πίεση λειτουργίας
137 kPa (20 psi)

149,23 mm (5,875 in.) Β x
191 mm (7,5 in.) Π
8,5 ώρες

149 mm (5,875 in.) Β x 191 mm(7,5 in.) Π
8,5 ώρες

137 kPa (20 psi)

137 kPa (20 psi)Κανονικός ρυθμός εξάτμισης 0,57 kg/ημέρα (1,3 lb/ημέρα)0,57 kg/ημέρα (1,3 lb/ημέρα)0,57 kg/ημέρα (1,3 lb/ημέρα)Εύρος ελέγχου τυπικής ροής Απενεργοποιημένο, 0,25;
0,5; 0,75; 1; 1,5; 2; 2,5; 3; 4;
5; 6 LPM
Ακρίβεια ρυθμού ροής*+/-10%

Απενεργοποιημένο, 0,25;
0,5; 0,75; 1; 1,5; 2; 2,5; 3; 4;
5; 6 LPM+/-10%

Απενεργοποιημένο, 0,5, 1, 2,
2,5, 3, 4, 6, 8, 10, 12, 15 LPM

Διαστάσεις

+/-10%* Αυτή η ακρίβεια είναι μόνο για 70 F και 14,7 psig και με βαθμονομημένο ακριβές ροόμετρο μάζας.

PN MN236-C4 E | Εγχειρίδιο χρήστη GR - 157

Sprint / Stroller / Hi Flow StrollerΠληροφορίες προειδοποιήσεων
Σημαντικό: Διαβάστε προσεκτικά αυτό το εγχειρίδιο
πριν τη λειτουργία του Sprint/Stroller/Hi Flow Stroller.
Μόνο με συνταγή γιατρού.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΥΤΉ Η ΣΥΣΚΕΥΉ ΔΕΝΠΡΟΟΡΊΖΕΤΑΙ ΓΙΑ ΖΩΤΙΚΉ ΥΠΟΣΤΉΡΙΞΗ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΆΝ ΝΟΜΊΖΕΤΕ ΌΤΙ Ο ΕΞΟΠΛΙΣΜΌΣ
ΔΕΝ ΛΕΙΤΟΥΡΓΕΊ ΚΑΝΟΝΙΚΆ, ΚΑΛΈΣΤΕ ΤΟΝΠΆΡΟΧΟ ΥΓΕΙΟΝΟΜΙΚΉΣ ΠΕΡΊΘΑΛΨΉΣ ΣΑΣ. ΜΗΝΕΠΙΧΕΙΡΉΣΕΤΕ ΝΑ ΕΠΙΣΚΕΥΆΣΕΤΕ Ή ΝΑ ΡΥΘΜΊΣΕΤΕΤΗ ΜΟΝΆΔΑ ΕΣΕΊΣ ΟΙ ΊΔΙΟΙ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΤΡΟΠΟΠΟΙΕΊΤΕ ΑΥΤΌΝ ΤΟΝΕΞΟΠΛΙΣΜΌ ΧΩΡΊΣ ΕΞΟΥΣΙΟΔΌΤΗΣΗ ΑΠΌ ΤΟΝΚΑΤΑΣΚΕΥΑΣΤΉ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΆΝ ΑΠΑΙΤΕΊΤΑΙ ΣΥΝΕΧΉΣ
ΠΑΡΟΧΉ ΟΞΥΓΌΝΟΥ, ΔΙΑΣΦΑΛΊΣΤΕ ΌΤΙ ΕΊΝΑΙ
ΔΙΑΘΈΣΙΜΗ ΕΠΑΡΚΉΣ ΠΑΡΟΧΉ ΟΞΥΓΌΝΟΥ Ή/ΚΑΙΜΙΑ ΔΕΥΤΕΡΕΎΟΥΣΑ ΠΑΡΟΧΉ ΟΞΥΓΌΝΟΥ ΑΝΆ ΠΆΣΑΣΤΙΓΜΉ ΚΑΤΆ ΤΗ ΔΙΆΡΚΕΙΑ ΤΗΣ ΘΕΡΑΠΕΊΑΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΕΠΙΤΡΈΠΕΤΕ ΤΟ ΚΆΠΝΙΣΜΑ,ΚΕΡΙΆ Ή ΓΥΜΝΈΣ ΦΛΌΓΕΣ ΕΝΤΌΣ 3 ΜΈΤΡΩΝ (10 FT)
ΑΠΌ ΤΗ ΣΥΣΚΕΥΉ, Ή ΣΕ ΑΠΌΣΤΑΣΗ ΜΙΚΡΌΤΕΡΗ ΑΠΌ
20 CM (8 IN.) ΑΠΌ ΟΠΟΙΑΔΉΠΟΤΕ ΠΗΓΉ ΑΝΆΦΛΕΞΗΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΔΙΑΤΗΡΕΊΤΕ ΤΟ ΠΡΟΪΌΝ ΣΕ ΚΑΛΆΑΕΡΙΖΌΜΕΝΟ ΧΏΡΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΑΠΟΘΗΚΕΎΕΤΕ
ΕΞΟΠΛΙΣΜΌ ΥΓΡΟΎ ΟΞΥΓΌΝΟΥ ΜΈΣΑ ΣΕΕΡΜΆΡΙΟ, ΠΟΡΤ-ΜΠΑΓΚΆΖ ΑΥΤΟΚΙΝΉΤΟΥΉ ΆΛΛΟ ΠΕΡΙΟΡΙΣΜΈΝΟ ΧΏΡΟ. ΜΗΝΤΟΠΟΘΕΤΕΊΤΕ ΚΟΥΒΈΡΤΕΣ, ΚΟΥΡΤΊΝΕΣΉ ΆΛΛΑ ΥΦΆΣΜΑΤΑ ΕΠΆΝΩ ΣΤΟΝΕΞΟΠΛΙΣΜΌ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΥΤΌ ΤΟ ΠΡΟΪΌΝ ΜΠΟΡΕΊΝΑ ΣΑΣ ΕΚΘΈΣΕΙ ΣΕ ΧΗΜΙΚΈΣ ΟΥΣΊΕΣ
ΣΥΜΠΕΡΙΛΑΜΒΑΝΟΜΈΝΟΥ ΤΟΥ ΝΙΚΕΛΊΟΥ, ΤΟ ΟΠΟΊΟ
ΕΊΝΑΙ ΓΝΩΣΤΌ ΣΤΗΝ ΠΟΛΙΤΕΊΑ ΤΗΣ ΚΑΛΙΦΌΡΝΙΑΌΤΙ ΠΡΟΚΑΛΕΊ ΚΑΡΚΊΝΟ. ΓΙΑ ΠΕΡΙΣΣΌΤΕΡΕΣ
ΠΛΗΡΟΦΟΡΊΕΣ, ΜΕΤΑΒΕΊΤΕ ΣΤΟ WWW.
P65WARNINGS.CA.GOV.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΣΕ ΠΕΡΊΠΤΩΣΗ ΠΟΥ ΣΥΜΒΕΊ
ΚΆΠΟΙΟ ΣΟΒΑΡΌ ΣΥΜΒΆΝ ΜΕ ΑΥΤΉΝ ΤΗ ΣΥΣΚΕΥΉ,Ο ΧΡΉΣΤΗΣ ΘΑ ΠΡΈΠΕΙ ΝΑ ΑΝΑΦΈΡΕΙ ΑΜΈΣΩΣΤΟ ΣΥΜΒΆΝ ΣΤΟΝ ΠΑΡΟΧΈΑ Ή / ΚΑΙ ΣΤΟΝΚΑΤΑΣΚΕΥΑΣΤΉ. ΈΝΑ ΣΟΒΑΡΌ ΣΥΜΒΆΝ ΟΡΊΖΕΤΑΙΩΣ ΤΡΑΥΜΑΤΙΣΜΌΣ, ΘΆΝΑΤΟΣ Ή ΠΙΘΑΝΌΤΗΤΑ
ΠΡΌΚΛΗΣΗΣ ΤΡΑΥΜΑΤΙΣΜΟΎ / ΘΑΝΆΤΟΥ ΣΕΠΕΡΊΠΤΩΣΗ ΕΠΑΝΆΛΗΨΗΣ ΤΟΥ ΣΥΜΒΆΝΤΟΣ.
Ο ΧΡΉΣΤΗΣ ΜΠΟΡΕΊ ΕΠΊΣΗΣ ΝΑ ΑΝΑΦΈΡΕΙ ΤΟΠΕΡΙΣΤΑΤΙΚΌ ΣΤΗΝ ΑΡΜΌΔΙΑ ΑΡΧΉ ΤΗΣ ΧΏΡΑΣ
ΌΠΟΥ ΣΥΝΈΒΗ ΤΟ ΠΕΡΙΣΤΑΤΙΚΌ.
Προσοχή: Χρησιμοποιείτε το Sprint/Stroller/
Hi Flow Stroller μόνο σύμφωνα με τις οδηγίες
του γιατρού σας.
Προσοχή: Η ομοσπονδιακή νομοθεσία των ΗΠΑ επιτρέπει την πώληση αυτής της συσκευής μόνο από ιατρό ήκατόπιν εντολής ιατρού.
158 - GR PN MN236-C4 E | Εγχειρίδιο χρήστη

Προσοχή: Η μονάδα περιέχει υγρό οξυγόνο το οποίο
είναι υπερβολικά ψυχρό, σχεδόν -184 °C (-300 °F).
Η έκθεση σε τόσο χαμηλή θερμοκρασία μπορεί ναπροκαλέσει σοβαρά κρυοπαγήματα.
Προσοχή: Το υγρό και το αέριο οξυγόνο, αν καιδεν είναι εύφλεκτα, μπορούν να προκαλέσουν τηνταχύτερη από το συνηθισμένο καύση άλλων υλικών.
Αυτός ο κίνδυνος, σε συνδυασμό με τη χαμηλή
θερμοκρασία του υγρού οξυγόνου, απαιτεί ορισμένες
προφυλάξεις ασφάλειας.
Προσοχή: Διατηρείτε τα εύφλεκτα υλικά μακριά απόαυτόν τον εξοπλισμό. Σπρέι αερολύματος, λάδια καιγράσο, συμπεριλαμβανομένων κρεμών προσώπου
και βαζελίνης, αναφλέγονται εύκολα και μπορούν νακαούν γρήγορα υπό την παρουσία οξυγόνου.
Προσοχή: Το κάπνισμα ενώ φοράτε κάνουλα
οξυγόνου μπορεί να προκαλέσει εγκαύματα τουπροσώπου και δυνητικά να έχει ως αποτέλεσμα τονθάνατο.
Η αφαίρεση της κάνουλας και η απόθεσή της επάνω
σε ενδύματα, κλινοσκεπάσματα, καναπέδες, μαξιλάρια
ή παρόμοια υλικά θα προκαλέσει κατάκαυση αερίου
νέφους όταν εκτεθεί σε τσιγάρο, πηγή θερμότητας,σπινθήρα ή φλόγα.
Εάν καπνίζετε, παρακαλείστε να: (1)απενεργοποιήσετε τη φορητή μονάδα, (2) αφαιρέσετε
την κάνουλα και (3) βγείτε από τον χώρο όπουβρίσκεται η συσκευή.
Προσοχή: Σε περίπτωση τυχαίας ανατροπής,επαναφέρετε αμέσως αλλά προσεκτικά, τη μονάδα
σε όρθια θέση αν είναι δυνατόν. Σε περίπτωσηδιαφυγής υγρού οξυγόνου, απομακρυνθείτε από τονχώρο αμέσως και καλέστε τον πάροχο υγειονομικής
περίθαλψής σας. Μην επιχειρήσετε να μετακινήσετε τημονάδα ή να σταματήσετε τη διαφυγή υγρού οξυγόνου.
Σημειωση: Μην αγγίζετε τα παγωμένα μέρη οποιασδήποτε μονάδας.
Σημειωση: Μην αποθηκεύετε ή λειτουργείτε τη φορητή μονάδα συνδεδεμένη με τη σταθερή μονάδα.
Σημειωση: Μην επιτρέπετε σε μη εκπαιδευμένο
προσωπικό να χειρίζεται ή να λειτουργεί αυτήν τησυσκευή.
Σημειωση: Η χρήση αυτής της συσκευής απαγορεύεται στις εμπορικές πτήσεις μεταφοράς επιβατώνή πτήσεις μεταφοράς εμπορευμάτων από τηνΟμοσπονδιακή Υπηρεσία Πολιτικής Αεροπορίας
των ΗΠΑ.

Προβλεπόμενη χρήσηΟ CAIRE Sprint/Stroller/Hi Flow Stroller προορίζεταιγια τη χορήγηση συμπληρωματικού οξυγόνου. Ησυσκευή δεν προορίζεται για υποστήριξη ζωής ούτεπαρέχει δυνατότητες παρακολούθησης ασθενούς.

Sprint / Stroller / Hi Flow StrollerΕισαγωγή

Στοιχεία χειρισμού

Τα Sprint, Stroller και Hi Flow Stroller προορίζονταιγια τη χορήγηση συμπληρωματικού οξυγόνου στον
ασθενή στο σπίτι του τελικού χρήστη και μπορούν,επίσης, να χρησιμοποιηθούν σε ιδρύματα, όπως οίκοι
ευγηρίας ή εγκαταστάσεις παροχής περίθαλψης γιαυποξείες παθήσεις. Τα Sprint/Stroller/Hi Flow Strollerπαρέχουν μια περιπατητική πηγή οξυγόνου γιαπαρατεταμένη χρονική περίοδο.

1. Ηλεκτρονικός μετρητής στάθμης υγρού
2. Ηλεκτρονικός διακόπτης στάθμης υγρού
3. Μηχανικός μετρητής στάθμης υγρού
4. Περιστροφικό κουμπί ελέγχου ροής5. Σύνδεσμος σωλήνα οξυγόνου
6. Βαλβίδα εξαερισμού
7. Συνδέσεις πλήρωσης (βλ. την επόμενη σελίδα
για την αναγνώριση της QDV)
8. Ιμάντας μεταφοράς με μαξιλαράκι ώμου9. Επίθεμα συμπυκνώματος
10. Κύπελλο συμπυκνώματος

Η συσκευή χρησιμοποιείται από ασθενείς με ΧΑΠή ασθενείς με μειωμένη αναπνευστική ικανότητα. Ησυσκευή συνταγογραφείται στον ασθενή. Η συσκευή
πωλείται σε έναν πάροχο που έχει εκπαιδευτεί στηλειτουργία και συντήρηση των Sprint/Stroller/
Hi Flow Stroller. Ο πάροχος εκπαιδεύει τον χρήστη.

1

Η συσκευή δεν προορίζεται για ζωτική υποστήριξη,ούτε παρέχει οποιεσδήποτε δυνατότητες
παρακολούθησης ασθενούς. Συνιστάται να υπάρχει
μια εναλλακτική πηγή συμπληρωματικού οξυγόνου
σε περίπτωση μηχανικής βλάβης. Αυτές οι φορητές
συσκευές διαθέτουν επίσης μια βαλβίδα γρήγορηςαποσύνδεσης, η οποία μπορεί να συνδεθεί με
μια συσκευή δεξαμενής LOX για τη μετακένωση
της φορητής συσκευής. Η πλήρωση της μονάδας
δεξαμενής πραγματοποιείται από τον πάροχο
υγειονομικής περίθαλψής σας.

2
4
5
8

Αυτό το εγχειρίδιο χρήστη περιέχει τις οδηγίες για τηχρήση των Sprint/Stroller/Hi Flow Stroller. Ανατρέξτεστο εγχειρίδιο χρήστη που παρέχεται με τη σταθερή
μονάδα για τη λειτουργία της.
Σημειωση: Ο πάροχος σέρβις θα βοηθήσει με την αρχική
διαμόρφωση και θα παρέχει καθοδήγηση σχετικά με τονσωστό χειρισμό και χρήση της μονάδας.

5

7

6
Μονάδες Sprint, Stroller και Hi Flow Stroller

10
9
PN MN236-C4 E | Εγχειρίδιο χρήστη GR - 159

Sprint / Stroller / Hi Flow StrollerΑναγνώριση βαλβίδας γρήγορηςαποσύνδεσης (QDV)Σύνδεσμοςφορητής
μονάδαςΣύνδεσμος
σταθερής
μονάδας3. Τοποθετήστε τον σύνδεσμο πλήρωσης σε όρθια
θέση επάνω από τον σύνδεσμο της σταθερής
μονάδας.

Τύπος QDVCAIREπλήρωσης SF
από το
κλειδώματοςπλάι

LIFE-OX
TF
κλειδώματος

4. α. Εάν ο σύνδεσμός σας είναι τύπου push-on:Χαμηλώστε τη μονάδα Sprint/Stroller/Hi Flow
Stroller επάνω στη σταθερή μονάδα μέχρι νααισθανθείτε τον σύνδεσμο να κουμπώνει.

PENOX

TAEMA
TF τύπου
push-onPB

β. Ε άν ο σύνδεσμος είναι Penox ή Life-Ox:Περιστρέψτε το Sprint/Stroller/Hi Flow Strollerμέχρι να αισθανθείτε τον σύνδεσμο να κουμπώνει.
Στη συνέχεια, προσεκτικά και σταθερά,περιστρέψτε το Sprint/Stroller/Hi Flow Stroller), ενώ πιέζετε προς τα κάτω.
δεξιόστροφα (Οδηγίες λειτουργίας
Για πλήρωση του Sprint/Stroller/Hi Flow
Stroller πλήρωσης από το επάνω μέρος
1. Καθαρίστε τους συνδέσμους πλήρωσης τόσοστη σταθερή μονάδα όσο και στο Sprint/Stroller/
Hi Flow Stroller με ένα καθαρό, στεγνό πανίχωρίς χνούδι.
Προσοχή: Οι σύνδεσμοι πλήρωσης πρέπει
να είναι στεγνοί, επειδή η υγρασία μπορεί ναπαγώσει και να ενώσει το Sprint/Stroller/Hi
Flow Stroller και τη σταθερή μονάδα.
2. Βεβαιωθείτε ότι τα περιστροφικά κουμπιά ελέγχου
ροής στο Sprint/Stroller/Hi Flow Stroller και στηδεξαμενή βρίσκονται στη θέση απενεργοποίησης (0).

160 - GR PN MN236-C4 E | Εγχειρίδιο χρήστη

Σημειωση: Συνέχεια στο βήμα 6 στην ενότητα «Γιαπλήρωση του Sprint/Stroller/Hi Flow Stroller πλήρωσηςαπό το πλάι».

Για πλήρωση του Sprint/Stroller/Hi Flow
Stroller πλήρωσης από το πλάι1. Καθαρίστε τους συνδέσμους πλήρωσης τόσοστη σταθερή μονάδα όσο και στο Sprint/Stroller/
Hi Flow Stroller με ένα καθαρό, στεγνό πανί χωρίς
χνούδι.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΟΙ ΣΎΝΔΕΣΜΟΙ ΠΛΉΡΩΣΗΣ
ΠΡΈΠΕΙ ΝΑ ΕΊΝΑΙ ΣΤΕΓΝΟΊ, ΕΠΕΙΔΉ ΗΥΓΡΑΣΊΑ ΜΠΟΡΕΊ ΝΑ ΠΑΓΏΣΕΙ ΚΑΙ ΝΑΕΝΏΣΕΙ ΤΟ SPRINT/STROLLER/HI FLOWSTROLLER ΚΑΙ ΤΗ ΣΤΑΘΕΡΉ ΜΟΝΆΔΑ.

Sprint / Stroller / Hi Flow Stroller
2. Βεβαιωθείτε ότι τα περιστροφικά κουμπιά ελέγχου
ροής στο Sprint/Stroller/Hi Flow Stroller και στηδεξαμενή βρίσκονται στη θέση απενεργοποίησης (0).

Σημειωση: Οι ενδείξεις του μετρητή στάθμης υγρού είναι
πιο αξιόπιστες 30 δευτερόλεπτα ή περισσότερο μετά τηνπλήρωση του Sprint/Stroller/Hi Flow Stroller.

3. Βεβαιωθείτε ότι ο σύνδεσμος πλήρωσης του Sprint/
Stroller/Hi Flow Stroller βρίσκεται στην όρθια θέσηως προς τον σύνδεσμο της σταθερής μονάδας.

4. Περιστρέψτε το Sprint/Stroller/Hi Flow Stroller) μέχρι να αισθανθείτε τοναριστερόστροφα (πείρο και την υποδοχή να κουμπώνουν μεταξύ τους.

7. Μπορεί επίσης να δείτε ατμό γύρω από τη σύνδεση.
Αυτό είναι φυσιολογικό και οφείλεται στη μεγάλη
διαφορά θερμοκρασίας μεταξύ του υγρού οξυγόνου
και του θερμού αέρα που περιβάλλει τις μονάδες.
8. Όταν η μονάδα γεμίσει, κλείστε τη βαλβίδα
εξαερισμού. Μπορείτε να καταλάβετε ότι η μονάδα
είναι γεμάτη όταν ο συριγμός αλλάξει τόνο καιεξέρχεται λίγο υγρό από τη γραμμή εξαερισμού.
Σημειωση: Το κλείσιμο της βαλβίδας αφού εξέρχεται μιασχεδόν συνεχής ροή υγρού γεμίζει το Sprint/Stroller/Hi
Flow Stroller σχεδόν τελείως, αλλά προκαλεί τη σπατάλη
κάποιας ποσότητας υγρού οξυγόνου.

5. Προσεκτικά και σταθερά, περιστρέψτε το Sprint/
Stroller/Hi Flow Stroller πίσω στην όρθια θέση.
Τώρα οι μονάδες έχουν κλειδώσει μεταξύ τους.

Σημειωση: Το κλείσιμο της βαλβίδας με την πρώτη ένδειξηεξόδου υγρού δεν γεμίζει το Sprint/Stroller/Hi Flow Strollerτελείως, αλλά εξοικονομεί υγρό οξυγόνο.
9. Για τους συνδέσμους τύπου push-on, πιέστε τοκουμπί απελευθέρωσης προς τα κάτω μέχρι οιμονάδες να διαχωριστούν η μία από την άλλη.
Για όλους τους άλλους συνδέσμους, περιστρέψτετη μονάδα μέχρι να διαχωριστεί από τη σταθερή
μονάδα.

Σημειωση: Συνεχίστε στο βήμα 6.
Σημειωση: Για τον σύνδεσμο τύπου push-on, κρατήστε τημονάδα στη θέση της καθ' όλη τη διάρκεια της πλήρωσης.
6. Ανοίξτε τη βαλβίδα εξαερισμού για να αρχίσετε τηνπλήρωση του Sprint/Stroller/Hi Flow Stroller. Θαπρέπει να ακούσετε λίγο οξυγόνο να διαφεύγει·μην ανησυχείτε. Ο ήχος συριγμού που ακούτε είναι
απόλυτα φυσιολογικός.

Κουμπί απελευθέρωσης φορητής μονάδας
(τύπος push-on μόνο)PN MN236-C4 E | Εγχειρίδιο χρήστη GR - 161

Sprint / Stroller / Hi Flow StrollerΠροσοχή: Ελέγξτε τον μετρητή στάθμης τουυγρού μόνο αφού κλείσει η βαλβίδα.

Οδηγίες λειτουργίας

Προσοχή: Εάν το Sprint/Stroller/Hi Flow Strollerγεμίζεται ξανά αμέσως μετά από μια περίοδο χρήσης,γεμίστε το και, στη συνέχεια, περιμένετε περίπου
10 λεπτά μετά την πλήρωση με τη βαλβίδα ελέγχου
ροής (FCV) στη θέση μηδέν για να σταθεροποιηθεί ηπίεση. Κατόπιν λειτουργήστε τη μονάδα ως συνήθως.
ΠΡΟΕΙΔΟΠΟΊΗΣΗ: ΜΗ ΛΕΙΤΟΥΡΓΕΊΤΕ ΤΟ
SPRINT/STROLLER/HI FLOW STROLLERΕΝΏ ΕΊΝΑΙ ΣΥΝΔΕΔΕΜΈΝΟ ΣΤΗ ΣΤΑΘΕΡΉ
ΜΟΝΆΔΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΆΝ ΤΟ SPRINT/STROLLER/
HI FLOW STROLLER ΔΕΝ ΔΙΑΧΩΡΊΖΕΤΑΙΕΎΚΟΛΑ, ΜΗΝ ΑΣΚΉΣΕΤΕ ΔΎΝΑΜΗ. ΟΙ ΜΟΝΆΔΕΣ
ΜΠΟΡΕΊ ΝΑ ΈΧΟΥΝ ΠΑΓΏΣΕΙ ΜΑΖΊ. ΑΦΉΣΤΕ ΤΙΣΜΟΝΆΔΕΣ ΣΥΝΔΕΔΕΜΈΝΕΣ ΚΑΙ ΠΕΡΙΜΈΝΕΤΕ
ΜΈΧΡΙ ΟΙ ΜΟΝΆΔΕΣ ΝΑ ΘΕΡΜΑΝΘΟΎΝ — ΤΌΤΕΘΑ ΔΙΑΧΩΡΙΣΤΟΎΝ ΕΎΚΟΛΑ. ΜΗΝ ΑΓΓΊΖΕΤΕ
ΟΠΟΙΑΔΉΠΟΤΕ ΠΑΓΩΜΈΝΑ ΜΈΡΗ. ΠΡΟΕΙΔΟΠΟΊΗΣΗ:ΣΕ ΠΕΡΊΠΤΩΣΗ ΠΟΥ Η ΔΙΑΡΡΟΉ ΕΊΝΑΙ ΥΠΕΡΒΟΛΙΚΉΣΕ ΒΑΘΜΌ ΠΟΥ ΕΊΝΑΙ ΠΑΡΟΎΣΑ ΡΟΉ ΥΓΡΟΎ,ΑΠΟΜΑΚΡΥΝΘΕΊΤΕ ΑΠΌ ΤΟΝ ΧΏΡΟ ΚΑΙ ΚΑΛΈΣΤΕ
ΤΟΝ ΠΆΡΟΧΟ ΥΓΕΙΟΝΟΜΙΚΉΣ ΠΕΡΊΘΑΛΨΉΣ ΣΑΣΑΜΈΣΩΣ.
ΠΡΟΕΙΔΟΠΟΊΗΣΗ: ΥΓΡΌ Ή ΨΥΧΡΌ ΑΈΡΙΟ ΟΞΥΓΌΝΟ
ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΕΙ ΚΡΥΟΠΑΓΉΜΑΤΑ ΣΤΟΔΈΡΜΑ. ΤΟ ΟΞΥΓΌΝΟ ΘΑ ΟΔΗΓΉΣΕΙ ΣΕ ΑΥΞΗΜΈΝΟ
ΚΊΝΔΥΝΟ ΠΥΡΚΑΓΙΆΣ.
Προσοχή: Σε περίπτωση που υπάρχει
οποιαδήποτε διαρροή υγρού από τη σταθερή
μονάδα ή το Sprint/Stroller/Hi Flow Strollerμετά τον διαχωρισμό των μονάδων, θέστε το
Sprint/Stroller/Hi Flow Stroller κατά μέρος,διασφαλίζοντας ότι παραμένει σε όρθια θέση,απομακρυνθείτε από τον χώρο και καλέστε
τον πάροχο υγειονομικής περίθαλψής σαςαμέσως.
Σημείωση: Για τον παροχέα εξοπλισμού: Τα παρακάτω
εξαρτήματα χορήγησης οξυγόνου συνιστώνται για χρήση
με το Sprint / Stroller / Hi Flow Stroller:• Ρινική κάνουλα με σωλήνα (μέγιστο LPM: 6 LPM):Αριθμός CAIRE 5408-SEQ• Firebreak: CAIRE Αριθμός Αρ. 21126636
Συνιστάται συνύπαρξη για χρήση με οποιονδήποτε
σωληνίσκο.
• Το CAIRE προσφέρει ένα πυροσβέστη που προορίζεταινα χρησιμοποιηθεί σε συνδυασμό με το φορητό οξυγόνο.
Η πυροσβεστική είναι μια θερμική ασφάλεια για να
σταματήσει η ροή του αερίου σε περίπτωση που οσωληνίσκος ή το σωληνάριο οξυγόνου που βρίσκεται
κάτω από τη ροή καύσης αναφλέγεται και καίγεται στηνπυρκαγιά. Τοποθετείται σε σειρά με τη ρινική κάνουλαή τη σωλήνωση οξυγόνου μεταξύ του ασθενούς και τηςεξόδου οξυγόνου του Sprint / Stroller / Hi Flow Stroller. Γιατη σωστή χρήση του πυροσβέστη, ανατρέξτε πάντα στιςοδηγίες του κατασκευαστή (που περιλαμβάνονται σε κάθεκιτ πυροπροστασίας).
• Πρόσθετες προτεινόμενες πληροφορίες αξεσουάρ είναι
διαθέσιμες στο Διαδίκτυο στη διεύθυνση www.caireinc.
com.
162 - GR PN MN236-C4 E | Εγχειρίδιο χρήστη(Η κάνουλα δεν περιλαμβάνεται.)1. Ωθήστε τον αναπνευστικό σωλήνα της κάνουλας
επάνω στον σύνδεσμο του σωλήνα οξυγόνου.
2. Προσαρμόστε τον αναπνευστικό σωλήνα τηςκάνουλάς σας στη σωστή θέση έτσι ώστε ναμπορείτε να αναπνέετε το οξυγόνο άνετα.
3. Στρίψτε το περιστροφικό κουμπί ελέγχου ροής πουβρίσκεται στο επάνω μέρος του Sprint/Stroller/Hi
Flow Stroller δεξιόστροφα μέχρι να είναι ορατός οσυνταγογραφημένος ρυθμός ροής (αριθμητικό) στο«παράθυρο» του περιστροφικού κουμπιού και γίνει
αισθητή μια θετική συγκράτηση.
Προσοχή: Εάν το περιστροφικό κουμπί
ελέγχου ροής τεθεί μεταξύ ρυθμών
ροής, θα προκύψει ροή οξυγόνου εκτός
προδιαγραφών.
4. Θα πρέπει τώρα να λαμβάνετε οξυγόνο. Γιανα διακόψετε τη ροή οξυγόνου, στρίψτε τοπεριστροφικό κουμπί ελέγχου ροής αριστερόστροφαστη θέση απενεργοποίησης (0).
Προσοχή: Στρίβετε πάντα το περιστροφικό
κουμπί ελέγχου ροής στη θέσηαπενεργοποίησης (0) όταν το σύστημα δενχρησιμοποιείται.

Sprint / Stroller / Hi Flow Stroller
5. Χρησιμοποιήστε το ακόλουθο διάγραμμα ωςκατευθυντήρια γραμμή για να καθορίσετε τοχρονικό διάστημα λειτουργίας του Sprint/Stroller/
Hi Flow Stroller:ΜοντέλοΡύθμιση FCV
SprintStrollerΘέση FCVΟνομαστικό Ονομαστικό
SprintStroller

Απενεργοποιημένο

30:00

60:00

0,25

18:00

30:00

0,5
0,75

12:12
9:18

23:00
17:24

1
1,5

7:30
5:30

14:06
10:12

2
2,5
3
4
5

4:18
3:36
3:06
2:24
2:00

8:00
6:36
5:36
4:24
3:36

6
8
10
12
15

1:42
Δεν ισχύει
Δεν ισχύει
Δεν ισχύει
Δεν ισχύει

3:00
Δεν ισχύει
Δεν ισχύει
Δεν ισχύει
Δεν ισχύειHi Flow
Stroller
Hi Flow
Stroller
71:00
Δενισχύει
32:00
Δενισχύει
16:00
Δενισχύει
8:00
6:20
5:20
4:00
Δεν
ισχύει
2:40
2:00
1:40
1:20
1:00

Σημείωση: Οι χρόνοι είναι σε ώρες και λεπτά
(μορφή 00:00).
Σημείωση: Οι «Ονομαστικοί» χρόνοι αναφέρονται σειδανικές συνθήκες, δηλ. μέγιστη πλήρωση, ακριβείς
ρυθμούς ροής, ικανοποιητικό ρυθμό απώλειας,μη μετακίνηση της μονάδας κ.λπ.
Αυτοί οι χρόνοι είναι οι μέγιστοι αναμενόμενοι.
Σημείωση: Τα ατομικά σας αποτελέσματα θαδιαφέρουν.
6α. Για να επαληθεύσετε τη στάθμη του υγρού
οξυγόνου στη μονάδα με τον ηλεκτρονικό μετρητή
στάθμης υγρού:• Μετρητής 3ης γενιάς: Πιέστε το πιεζόμενο κουμπί
(κουμπί στάθμης υγρού) στο επάνω μέρος τηςμονάδας για δύο δευτερόλεπτα τουλάχιστον.
Διαβάστε το επάνω μέρος της φωτεινής ράβδου για
να καθορίσετε τη στάθμη του περιεχομένου.

• Μετρητής 4ης γενιάς: Πατήστε το πράσινο κουμπί
λειτουργίας στην πρόσοψη του μετρητή. Διαβάστε τοτόξο των LED, το οποίο υποδεικνύει τη στάθμη τουπεριεχομένου.

Προσοχή: Το Sprint/Stroller/Hi Flow Strollerείναι κενό εάν είναι αναμμένο μόνο τοτελευταίο τμήμα της φωτεινής ράβδου.
Εάν ο δείκτης χαμηλής μπαταρίας ανάψει ότανπιεστεί το κουμπί, ενημερώστε τον πάροχο
υγειονομικής περίθαλψής σας.
Σημειωση: Το Sprint/Stroller/Hi Flow Stroller θα συνεχίσει
να παρέχει οξυγόνο ακόμα και αν ανάψει η ένδειξηχαμηλής μπαταρίας, εφόσον υπάρχει υγρό οξυγόνο μέσαστη μονάδα.
6β. Για να επαληθεύσετε τη στάθμη του υγρού
οξυγόνου στη μονάδα με τον μηχανικό μετρητή
στάθμης υγρού:• Υποστηρίξτε τη μονάδα, από τον υφασμάτινο
ιμάντα, ακριβώς επάνω από τον μαύρο θάλαμο
ελατηριωτού ζυγού ένδειξης περιεχομένου.
• Αφήστε τη μονάδα να σταθεροποιηθεί (δηλ. να μηναναπηδά πάνω-κάτω) και στη συνέχεια διαβάστε
τον εκτεθειμένο έγχρωμο ζυγό για το περιεχόμενο
κατά προσέγγιση σε υγρό οξυγόνο.

PN MN236-C4 E | Εγχειρίδιο χρήστη GR - 163

Sprint / Stroller / Hi Flow StrollerΣημειωση: Εάν η μονάδα είναι κενή, θα είναι εκτεθειμένη
μόνο η ταινία κόκκινου χρώματος του ζυγού. Στηνπερίπτωση αυτή, πρέπει να γίνει πλήρωση της μονάδας
πριν από τη χρήση.
7. Υπό ορισμένες περιβαλλοντικές συνθήκες, μεσυνεχή χρήση, και ιδίως εάν η φορητή μονάδα δενμετακινείται, μπορεί να αναπτύξει υπερβολικό πάγογύρω από το πηνίο εξατμιστήρα και επάνω στοπερίβλημα. Μπορείτε να μειώσετε αυτόν τον πάγοκτυπώντας ελαφρά τη μονάδα ή/και σκουπίζοντας
τυχόν συσσωρευμένο πάγο από το περίβλημα.

Συντήρηση επιθέματοςσυμπυκνώματοςΑφού η μονάδα αδειάσει και θερμανθεί σεθερμοκρασία δωματίου, αφαιρέστε τυχόν υγρασία απότο κύπελλο συμπυκνώματος:Για σκληρό περίβλημα• Γείρετε τη μονάδα προς μία πλευρά. Κρατήστε
σταθερά το κουμπί στο κέντρο του κυπέλλου
συμπυκνώματος (στο κάτω μέρος της μονάδας)και τραβήξτε το ευθεία προς τα έξω. Η ενέργεια
αυτή θα απελευθερώσει το κύπελλο.
Σημειωση: Εάν φέρει έντονους ρύπους, το επίθεμα μπορεί
να πλυθεί σε πλυντήριο (με χλωρίνη για απολύμανση).
• Για να επανατοποθετήσετε το κύπελλο, κεντράρετέ
το επάνω από το στρογγυλό άνοιγμα στο κάτωμέρος της μονάδας και πιέστε το κουμπί προςτα μέσα κατά ένα κλικ για να κουμπώσετε τοκύπελλο και, στη συνέχεια, κατά ένα δεύτερο κλικγια να κλειδώσετε το κύπελλο στη θέση του.

Για μαλακό περίβλημα• Ανοίξτε το κάτω φερμουάρ και τραβήξτε προς ταπίσω το κάτω κάλυμμα.
• Αφαιρέστε το επίθεμα από το κύπελλο καιστύψτε την υγρασία που έχει απορροφήσει.
Αφήστε το επίθεμα να στεγνώσει τελείως πριντο επανατοποθετήσετε μέσα στο κύπελλο.
Σημειωση: Εάν φέρει έντονους ρύπους, το επίθεμα μπορεί
να πλυθεί σε πλυντήριο (με χλωρίνη για απολύμανση).
• Για να το επανατοποθετήσετε, απλά εισαγάγετε
το επίθεμα μέσα στο κάτω κάλυμμα και κλείστετο κάτω φερμουάρ.
Σημειωση: Εάν το κουμπί δεν πιέζεται προς τα μέσακουμπώνοντας το κύπελλο, κρατήστε το κύπελλο καιτραβήξτε το κουμπί προς τα έξω μέχρι να τεθεί στηνανοικτή θέση με ένα κλικ. Τώρα επανατοποθετήστε τοκύπελλο όπως περιγράφηκε προηγουμένως.

Μείωση του πάγουΥπό ορισμένες περιβαλλοντικές συνθήκες, μεσυνεχή χρήση, και ιδίως εάν η φορητή μονάδα δενμετακινείται, μπορεί να αναπτύξει υπερβολικό πάγογύρω από το πηνίο εξατμιστήρα και επάνω στοπερίβλημα. Μπορείτε να μειώσετε αυτόν τον πάγοκτυπώντας ελαφρά τη μονάδα ή/και σκουπίζοντας
τυχόν συσσωρευμένο πάγο από το περίβλημα.
ΠΡΟΕΙΔΟΠΟΊΗΣΗ: ΛΌΓΩ ΤΗΣ ΠΙΘΑΝΌΤΗΤΑΣ
ΣΥΣΣΏΡΕΥΣΗΣ ΥΠΕΡΒΟΛΙΚΉΣ ΥΓΡΑΣΊΑΣ/ΠΆΓΟΥ, ΣΥΝΙΣΤΆΤΑΙ ΤΟ HI FLOW STROLLERΝΑ ΠΑΡΑΜΈΝΕΙ ΣΕ ΑΔΡΆΝΕΙΑ ΓΙΑ 1 ΏΡΑΜΕΤΆ ΤΗ ΧΡΉΣΗ, ΓΙΑ ΝΑ ΜΕΙΩΘΕΊ Η ΣΥΣΣΏΡΕΥΣΗ ΥΓΡΑΣΊΑΣ.

Κλειδωμένο
Κουμπωμένο
Ανοικτό

Κύπελλο με κουμπί

Σημειωση: Εάν το κουμπί δεν πιέζεται προς τα μέσακουμπώνοντας το κύπελλο, κρατήστε το κύπελλο καιτραβήξτε το κουμπί προς τα έξω μέχρι να τεθεί στηνανοικτή θέση με ένα κλικ. Τώρα επανατοποθετήστε τοκύπελλο όπως περιγράφηκε προηγουμένως.
164 - GR PN MN236-C4 E | Εγχειρίδιο χρήστη

Λόγω των υψηλών ρυθμών ροής που συσχετίζονταιμε το Hi Flow Stroller, συμπυκνωμένη υγρασία καικάποια ποσότητα πάγου μπορεί να παρατηρηθούν
επάνω στη μονάδα, ιδιαίτερα σε ρυθμίσεις ροής10 LPM και υψηλότερες. Συνιστάται η μονάδα ναπαραμένει ανενεργή για τουλάχιστον 1 ώρα μετά τηχρήση, για να μειωθεί η πιθανότητα υπερβολικής
συσσώρευσης λόγω συνεχούς χρήσης.

Sprint / Stroller / Hi Flow StrollerΑντιμετώπιση προβλημάτων
Οι ακόλουθες πληροφορίες σκοπό έχουν να σας βοηθήσουν να αναγνωρίσετε και να επιλύσετε απλά προβλήματα
λειτουργίας που μπορεί να αντιμετωπίσετε όταν χρησιμοποιείτε το Sprint/Stroller/Hi Flow Stroller.
ΠρόβλημαΗ φορητή μονάδα παράγειήχο συριγμού.

Λύση• Συριγμός μπορεί να παρουσιαστεί για τη διατήρηση της σωστής πίεσης λειτουργίας εντός
της φορητής μονάδας. Είναι πιθανότερο να παρουσιαστεί συριγμός μετά την πλήρωση ήόταν αλλάξει η θέση της φορητής μονάδας. Ο συριγμός μπορεί να διαρκέσει για περίπου
10 λεπτά μετά την πλήρωση. Επιπλέον, μπορεί να παρουσιαστεί όταν η βαλβίδα ελέγχου
ροής βρίσκεται σε χαμηλή ρύθμιση.

• Εάν η φορητή μονάδα έχει τοποθετηθεί σε ακατάλληλη θέση, επαναφέρετε τη φορητή
μονάδα σε όρθια θέση και αφήστε τη μονάδα να σταθεροποιηθεί για αρκετά λεπτάΗ ροή της φορητής μονάδας • Βεβαιωθείτε ότι η κάνουλα έχει συνδεθεί σταθερά στην έξοδο οξυγόνου.
σταματά κατά τη διάρκεια
• Βεβαιωθείτε ότι η κάνουλα δεν έχει συστραφεί.
της χρήσης.
• Ελέγξτε την ένδειξη περιεχομένου/τον μετρητή στάθμης και γεμίστε τη φορητή μονάδα εάνχρειάζεται.

Η φορητή μονάδα δενγεμίζει.
Η βαλβίδα εξαερισμού
της φορητής μονάδας δενκλείνει καλά στο τέλος τηςδιαδικασίας πλήρωσης.

Η φορητή μονάδα δεναποδεσμεύεται εύκολααπό τη δεξαμενή μετά τηνπλήρωση.

• Διασφαλίστε ότι το περιστροφικό κουμπί ελέγχου ροής δεν βρίσκεται στη θέσηαπενεργοποίησης («0»).
• Επαληθεύστε ότι υπάρχει οξυγόνο μέσα στη δεξαμενή.
• Βεβαιωθείτε ότι οι σύνδεσμοι πλήρωσης της φορητής μονάδας και της δεξαμενής είναι
πλήρως κουμπωμένοι καθ' όλη τη διάρκεια της διαδικασίας πλήρωσης.
• Εάν η βαλβίδα εξαερισμού δεν μπορεί να κλείσει και ο συριγμός και το νέφος ατμού
οξυγόνου συνεχίζονται, αφαιρέστε προσεκτικά τη φορητή μονάδα πατώντας το κουμπί
απελευθέρωσης στη δεξαμενή. Η εξαέρωση από το κάτω μέρος της φορητής μονάδας
θα σταματήσουν σε λίγα λεπτά. Αφήστε τη μονάδα να θερμανθεί για να μπορέσετε
να κλείσετε τη βαλβίδα εξαέρωσης. Η φορητή μονάδα μπορεί να χρειαστεί μέχρι και60 λεπτά για την αποκατάσταση επαρκούς πίεσης για ακριβή ροή οξυγόνου. Εάναπαιτείται, χρησιμοποιήστε μια εναλλακτική πηγή οξυγόνου όπως βαλβίδα ελέγχου ροήςπροσαρτημένη στη δεξαμενή.
• Οι σύνδεσμοι πλήρωσης της φορητής μονάδας και της δεξαμενής μπορεί να έχουν
παγώσει.
• ΜΗΝ ΑΣΚΗΣΕΤΕ ΔΥΝΑΜΗ. Αφήστε να περάσουν μερικά λεπτά για να θερμανθούν ταπαγωμένα μέρη και στη συνέχεια αποδεσμεύστε τη φορητή μονάδα όταν λιώσει ο πάγος.
Για να μην παγώσουν οι μονάδες μεταξύ τους, πάντοτε να σκουπίζετε τον αρσενικό
σύνδεσμο πλήρωσης στη δεξαμενή και τον θηλυκό σύνδεσμο πλήρωσης στη φορητή
μονάδα με ένα καθαρό, στεγνό πανί πριν την πλήρωση.

PN MN236-C4 E | Εγχειρίδιο χρήστη GR - 165

Sprint / Stroller / Hi Flow StrollerΠρότυπο καθαρισμού

ΣυντήρησηΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΚΑΘΑΡΙΣΜΌΣ ΠΡΈΠΕΙΝΑ ΠΡΑΓΜΑΤΟΠΟΙΕΊΤΑΙ ΜΌΝΟ ΑΦΟΎ ΗΜΟΝΆΔΑ ΈΧΕΙ ΑΔΕΙΆΣΕΙ.
• Καθαρίστε χρησιμοποιώντας ένα διάλυμα ήπιου
απορρυπαντικού πιάτων και νερό.
• Εφαρμόστε το διάλυμα καθαρισμού απευθείας
επάνω σε ένα πανί χωρίς χνούδι. Τα εγκεκριμένα
καθαριστικά συμπεριλαμβάνουν το HydroPureκαι το HydroKlean. Μην ψεκάζετε καθαριστικά
απευθείας στο Sprint/Stroller/Hi Flow Stroller.
• Σκουπίστε την εξωτερική επιφάνεια με το πανίχωρίς χνούδι μέχρι να καθαρίσει.

Ο πάροχος σέρβις σας είναι υπεύθυνος για οποιαδήποτε
συντήρηση μπορεί να απαιτείται σύμφωνα με τοτεχνικό εγχειρίδιο αυτής της συσκευής. Καλέστε
τον πάροχο σέρβις σας για οποιεσδήποτε απαιτήσεις
συντήρησης.

Προσοχή: Μη χρησιμοποιείτε εξοπλισμό
καθαρισμού υψηλής θερμοκρασίας καιυψηλής πίεσης για τον καθαρισμό αυτών τωνμονάδων.
• Μην εφαρμόζετε καθαριστικό σε οποιαδήποτε
εσωτερικά εξαρτήματα ή βαλβίδες.
• Αφήστε τη μονάδα να στεγνώσει τελείως πριντη χρήση.

Καθαρίζετε τους συνδέσμους πλήρωσης τόσο στησταθερή όσο και στη φορητή μονάδα με ένα καθαρό,στεγνό πανί χωρίς χνούδι μεταξύ των πληρώσεων,προκειμένου να αποφευχθεί το πάγωμα και πιθανή
βλάβη του εξοπλισμού.

Σημείωση: Σημείωση προς τον πάροχο υγειονομικής
περίθαλψης – για τις διαδικασίες επανεπεξεργασίας,ανατρέξτε στο αντίστοιχο εγχειρίδιο σέρβις.

ΑΗΗΕ και RoHSΤο επίθεμα συμπυκνώματος είναι το μόνο μέρος πουεπιδέχεται συντήρηση από τον χρήστη. Κανένα άλλομέρος δεν επιδέχεται συντήρηση από τον χρήστη.
Η ελάχιστη αναμενόμενη ωφέλιμη διάρκεια ζωής είναι
πέντε χρόνια.

Σημείωση: Οποιαδήποτε απαιτούμενη πρόσθετησυντήρηση πρέπει να πραγματοποιείται από ειδικευμένο
τεχνικό σέρβις ή πάροχο σέρβις.

Παρελκόμενα

Αυτό το σύμβολο υπενθυμίζει στους κατόχους
του εξοπλισμού να τον επιστρέψουν σεεγκαταστάσεις ανακύκλωσης στο τέλος τηςωφέλιμης ζωής του, σύμφωνα με την Οδηγία για ταΑπόβλητα Ηλεκτρικού και Ηλεκτρονικού Εξοπλισμού
(ΑΗΗΕ).
Τα προϊόντα μας συμμορφώνονται με την οδηγία γιατον περιορισμό των επικίνδυνων ουσιών (RoHS). Δενπεριέχουν περισσότερο από ιχνοποσότητες μολύβδουή άλλων επικίνδυνων υλικών.

ΑπόρριψηΠάντοτε να επιστρέφετε το Sprint/Stroller/Hi Flow
Stroller, συμπεριλαμβανομένων όλων των εξαρτημάτων,στον πάροχο κατ' οίκον φροντίδας σας για σωστή
απόρριψη. Μπορείτε επίσης να επικοινωνήσετε με τατοπικά γραφεία της πόλης σας για οδηγίες σχετικά με τησωστή απόρριψη της μπαταρίας.

Μεταφορά και αποθήκευσηΟι συσκευές πρέπει να αποθηκεύονται σε όρθια θέση καινα αερίζονται καλά. Μην αφήνετε τη συσκευή να είναι
ξαπλωμένη επάνω στην πλευρά της. Υγρασία έως 95%χωρίς συμπύκνωση υδρατμών. Εύρος θερμοκρασιών
από -40 °C έως 70 °C (-40 °F έως 158 °F).
Εύρος θερμοκρασιών λειτουργίας από -10 °C έως
40 °C (14 °F έως 104 °F). Εύρος σχετικής υγρασίας
από 30% έως 75% χωρίς συμπύκνωση υδρατμών.
Σημείωση: Το εύρος ατμοσφαιρικής πίεσης είναι 700 hPa
έως 1.060 hPa (υψόμετρο 3.048 m έως -304,8 m [10.000 ft
έως -1.000 ft]).
166 - GR PN MN236-C4 E | Εγχειρίδιο χρήστη

Τροχήλατο μεταφοράς Αρ. προϊόντος 20765565

Sprint / Stroller / Hi Flow StrollerΑσφάλειαΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΟΙ ΦΟΡΗΤΈΣ ΣΥΣΚΕΥΈΣ ΕΠΙΚΟΙΝΩΝΙΏΝ ΜΕ ΡΑΔΙΟΣΥΧΝΌΤΗΤΕΣ(ΣΥΜΠΕΡΙΛΑΜΒΑΝΟΜΈΝΩΝ ΤΩΝ ΠΕΡΙΦΕΡΕΙΑΚΏΝ ΌΠΩΣ ΚΑΛΏΔΙΑ ΚΕΡΑΊΑΣ ΚΑΙ ΕΞΩΤΕΡΙΚΈΣ
ΚΕΡΑΊΕΣ) ΔΕΝ ΠΡΈΠΕΙ ΝΑ ΧΡΗΣΙΜΟΠΟΙΟΎΝΤΑΙ ΣΕ ΑΠΌΣΤΑΣΗ ΜΙΚΡΌΤΕΡΗ ΑΠΌ 30 CM (12 IN.) ΑΠΌΟΠΟΙΟΔΉΠΟΤΕ ΜΈΡΟΣ ΤΟΥ SPRINT/STROLLER/HI FLOW STROLLER, ΣΥΜΠΕΡΙΛΑΜΒΑΝΟΜΈΝΩΝ ΤΩΝΚΑΛΩΔΊΩΝ ΠΟΥ ΚΑΘΟΡΊΖΟΝΤΑΙ ΑΠΌ ΤΟΝ ΚΑΤΑΣΚΕΥΑΣΤΉ. ΣΕ ΑΝΤΊΘΕΤΗ ΠΕΡΊΠΤΩΣΗ, ΘΑ ΜΠΟΡΟΎΣΕΝΑ ΠΡΟΚΛΗΘΕΊ ΥΠΟΒΆΘΜΙΣΗ ΤΗΣ ΑΠΌΔΟΣΗΣ ΑΥΤΟΎ ΤΟΥ ΕΞΟΠΛΙΣΜΟΎ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΧΡΉΣΗ ΠΑΡΕΛΚΟΜΈΝΩΝ, ΜΟΡΦΟΤΡΟΠΈΩΝ ΚΑΙ ΚΑΛΩΔΊΩΝ ΕΚΤΌΣ ΑΠΌ ΕΚΕΊΝΑ ΠΟΥΚΑΘΟΡΊΖΟΝΤΑΙ Ή ΠΑΡΈΧΟΝΤΑΙ ΑΠΌ ΤΟΝ ΚΑΤΑΣΚΕΥΑΣΤΉ ΑΥΤΟΎ ΤΟΥ ΕΞΟΠΛΙΣΜΟΎ ΘΑ ΜΠΟΡΟΎΣΕ ΝΑ ΈΧΕΙΩΣ ΑΠΟΤΈΛΕΣΜΑ ΑΥΞΗΜΈΝΕΣ ΗΛΕΚΤΡΟΜΑΓΝΗΤΙΚΈΣ ΕΚΠΟΜΠΈΣ Ή ΜΕΙΩΜΈΝΗ ΗΛΕΚΤΡΟΜΑΓΝΗΤΙΚΉ ΑΤΡΩΣΊΑ
ΑΥΤΟΎ ΤΟΥ ΕΞΟΠΛΙΣΜΟΎ ΚΑΙ ΝΑ ΟΔΗΓΉΣΕΙ ΣΕ ΑΚΑΤΆΛΛΗΛΗ ΛΕΙΤΟΥΡΓΊΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΧΡΉΣΗ ΑΥΤΟΎ ΤΟΥ ΕΞΟΠΛΙΣΜΟΎ ΠΑΡΑΚΕΊΜΕΝΑ Ή ΣΕ ΔΙΆΤΑΞΗ ΣΤΟΊΒΑΣ ΜΕ ΆΛΛΟΕΞΟΠΛΙΣΜΌ ΠΡΈΠΕΙ ΝΑ ΑΠΟΦΕΎΓΕΤΑΙ ΔΙΌΤΙ ΜΠΟΡΕΊ ΝΑ ΟΔΗΓΉΣΕΙ ΣΕ ΑΚΑΤΆΛΛΗΛΗ ΛΕΙΤΟΥΡΓΊΑ. ΣΕΠΕΡΊΠΤΩΣΗ ΠΟΥ ΤΈΤΟΙΑ ΧΡΉΣΗ ΕΊΝΑΙ ΑΠΑΡΑΊΤΗΤΗ, ΑΥΤΌΣ Ο ΕΞΟΠΛΙΣΜΌΣ ΚΑΙ Ο ΆΛΛΟΣ ΕΞΟΠΛΙΣΜΌΣ ΘΑΠΡΈΠΕΙ ΝΑ ΠΑΡΑΚΟΛΟΥΘΗΘΟΎΝ ΓΙΑ ΝΑ ΕΠΑΛΗΘΕΥΤΕΊ ΌΤΙ ΛΕΙΤΟΥΡΓΟΎΝ ΚΑΝΟΝΙΚΆ.
Προσοχή: Ο ιατρικός ηλεκτρικός εξοπλισμός απαιτεί ειδικές προφυλάξεις όσον αφορά τηνηλεκτρομαγνητική συμβατότητα (EMC) και πρέπει να εγκαθίσταται και να τίθεται σε λειτουργία σύμφωνα με
τις πληροφορίες περί EMC που παρέχονται στο παρόν εγχειρίδιο.
Προσοχή: Φορητός και κινητός εξοπλισμός επικοινωνιών ραδιοσυχνοτήτων (RF) μπορεί να επηρεάσει τονιατρικό ηλεκτρικό εξοπλισμό.
Πίνακας 1Οδηγία και δήλωση του κατασκευαστή — ηλεκτρομαγνητικές εκπομπές
Το Sprint/Stroller/Hi Flow Stroller προορίζεται για χρήση στο ηλεκτρομαγνητικό περιβάλλον πουπροσδιορίζεται παρακάτω. Ο αγοραστής ή ο χρήστης του Sprint/Stroller/Hi Flow Stroller θα πρέπει ναδιασφαλίσει ότι το σύστημα χρησιμοποιείται σε ανάλογο περιβάλλον.

Δοκιμή εκπομπών
Εκπομπές
ραδιοσυχνοτήτων (RF)Συμμόρφωση
Ομάδα 1CISPR 11Ηλεκτρομαγνητικό περιβάλλον — οδηγίες
Το Sprint/Stroller/Hi Flow Stroller χρησιμοποιεί ενέργεια
ραδιοσυχνοτήτων μόνο για την εσωτερική λειτουργία του.
Συνεπώς, οι εκπομπές ραδιοσυχνοτήτων της συσκευής
είναι πολύ χαμηλές και δεν είναι πιθανό να προκαλέσουν
παρεμβολή σε κοντινό ηλεκτρονικό εξοπλισμό.

Εκπομπές
ραδιοσυχνοτήτων(RF) CISPR 11Κατηγορία BΕκπομπές αρμονικών
IEC 61000-3-2ΔενεφαρμόζεταιΔιακυμάνσεις τάσης/εκπομπές αναλαμπής
IEC 61000-3-3ΔενεφαρμόζεταιΤο Sprint/Stroller/Hi Flow Stroller είναι κατάλληλο γιαχρήση σε όλους τους χώρους, συμπεριλαμβανομένων
των οικιακών εγκαταστάσεων και εκείνων πουείναι απευθείας συνδεδεμένοι στο δημόσιο δίκτυο
παροχής χαμηλής τάσης, που τροφοδοτεί κτίρια πουχρησιμοποιούνται για οικιακούς σκοπούς.

PN MN236-C4 E | Εγχειρίδιο χρήστη GR - 167

Sprint / Stroller / Hi Flow StrollerΠίνακας 2*: Συνιστώμενες αποστάσεις διαχωρισμού μεταξύ φορητού και κινητού εξοπλισμού επικοινωνίας
ραδιοσυχνοτήτων και του Sprint/Stroller/Hi Flow StrollerΤο Sprint/Stroller/Hi Flow Stroller προορίζεται για χρήση σε ηλεκτρομαγνητικό περιβάλλον με ελεγχόμενες διαταραχές
ακτινοβολούμενων ραδιοσυχνοτήτων. Ο πελάτης ή ο χρήστης του Sprint/Stroller/Hi Flow Stroller μπορεί να αποτρέψει τιςηλεκτρομαγνητικές παρεμβολές διατηρώντας μια ελάχιστη απόσταση μεταξύ του φορητού και κινητού εξοπλισμού επικοινωνιών
ραδιοσυχνοτήτων (πομποί) και του Sprint/Stroller/Hi Flow Stroller όπως συνιστάται παρακάτω, σύμφωνα με τη μέγιστη ισχύ εξόδου
του εξοπλισμού επικοινωνιών.
Απόσταση διαχωρισμού ανάλογα με τη συχνότητα του πομπού
Ονομαστική τιμήmμέγιστης ισχύος
εξόδου του πομπού
80 MHz έως 800 MHz
800 MHz έως 2,5 GHz
150 kHz έως 80 MHz

d = 1,2√P

d = 1,2√P

d = 2,3√PW
0,010,12 m0,12 m0,23 m0,10,38 m0,38 m0,73 m
1
1,2 m
1,2 m
2,3 m
10
3,8 m
3,8 m
7,3 m
100
12 m
12 m
23 m
Για πομπούς με ονομαστική μέγιστη ισχύ εξόδου που δεν παρατίθεται παραπάνω, η συνιστώμενη απόσταση διαχωρισμού (d) σεμέτρα (m) μπορεί να προσδιοριστεί με χρήση της εξίσωσης που εφαρμόζεται στη συχνότητα του πομπού, όπου P είναι η μέγιστητιμή ισχύος εξόδου του πομπού σε watt (W), σύμφωνα με τον κατασκευαστή του πομπού.
ΣΗΜΕΙΩΣΗ 1 Στα 80 MHz και 800 MHz, ισχύει η απόσταση διαχωρισμού για το υψηλότερο εύρος συχνοτήτων.
ΣΗΜΕΙΩΣΗ 2 Αυτές οι οδηγίες μπορεί να μην ισχύουν σε όλες τις καταστάσεις. Η διάδοση των ηλεκτρομαγνητικών κυμάτων
επηρεάζεται από την απορρόφηση και την αντανάκλαση από δομές, αντικείμενα και ανθρώπους.
* Αυτός ο πίνακας περιλαμβάνεται ως τυπική απαίτηση για εξοπλισμό που έχει ελεγχθεί σύμφωνα με συγκεκριμένα επίπεδα
δοκιμής και άλλα συγκεκριμένα εύρη συχνοτήτων και έχει βρεθεί συμμορφούμενος με τους κανονισμούς.

Πίνακας 3: Οδηγίες και δήλωση κατασκευαστή — ηλεκτρομαγνητική ατρωσία
Το Sprint/Stroller/Hi Flow Stroller προορίζεται για χρήση στο ηλεκτρομαγνητικό περιβάλλον που προσδιορίζεται παρακάτω. Οαγοραστής ή ο χρήστης του Sprint/Stroller/Hi Flow Stroller θα πρέπει να διασφαλίσει ότι το σύστημα χρησιμοποιείται σε ανάλογο
περιβάλλον.
Δοκιμή ατρωσίας
Επίπεδο δοκιμής IEC 60601 Επίπεδο συμμόρφωσης Ηλεκτρομαγνητικό περιβάλλον — οδηγίες
Τα δάπεδα θα πρέπει να είναι
± 8 kV επαφή
± 8 kV επαφή
Ηλεκτροστατική
κατασκευασμένα από ξύλο, τσιμέντο ήεκφόρτιση (ESD)±2 kV, ± 4 kV,±2 kV, ± 4 kV,
κεραμικά πλακίδια. Εάν τα δάπεδα είναι
επιστρωμένα με συνθετικό υλικό, η σχετική
IEC 61000-4-2±8 kV, ±15 kV αέρας
±8 kV, ±15 kV αέρας
υγρασία πρέπει να είναι τουλάχιστον 30%.**Δεν εφαρμόζεται±2 kV για γραμμές
Συσκευή τροφοδοτούμενη
Ταχεία ηλεκτρική
τροφοδοσίας ρεύματος
με συνεχές ρεύμα (DC)Δεν εφαρμόζεται
πάροδος/ριπή
±1 kV για γραμμές εισόδου/ Δεν εφαρμόζεταιIEC 610004-4
εξόδου
Χωρίς γραμμές εισόδου/εξόδου δεδομένων
±1 kV γραμμή(ές) προςΔεν εφαρμόζεται
Υπέρτασηγραμμή(ές)Συσκευή τροφοδοτούμενη Δεν εφαρμόζεται±2 kV γραμμή(ές) προς
IEC 61000-4-5
με συνεχές ρεύμα (DC)γείωση
<5% UT (>95% πτώση στηνΠτώσεις τάσης,UT) για 0,5 κύκλο
σύντομες διακοπές και 40% UT (60% πτώση στηνΔεν εφαρμόζεταιδιακυμάνσεις τάσης
UT) για 5 κύκλους
Συσκευή τροφοδοτούμενη Δεν εφαρμόζεταιστις γραμμές παροχής 70% UT (30% πτώση στηνμε συνεχές ρεύμα (DC)τροφοδοσίας εισόδου UT) για 25 κύκλους
<5% UT (>95% πτώση στηνIEC 61000-4-11UT) για 5 δευτερόλεπταΜαγνητικό πεδίο
Τα μαγνητικά πεδία συχνότητας ισχύος θασυχνότητας ισχύος
3A/m
3A/m
πρέπει να είναι αυτά ενός τυπικού εμπορικού
(50/60 Hz)
50/60 Hz50/60 Hzή νοσοκομειακού περιβάλλοντος.
IEC 61000-4-8Σημείωση: UT είναι η τάση κύριας παροχής τάσης δικτύου (AC) πριν την εφαρμογή του επιπέδου δοκιμής.

** Αυτή η δήλωση υποδεικνύει ότι έχουν πραγματοποιηθεί οι απαιτούμενες δοκιμές σε ελεγχόμενο περιβάλλον και τα Sprint/
Stroller/Hi Flow Stroller συμμορφώνονται αποδεδειγμένα με τους κανονισμούς.

168 - GR PN MN236-C4 E | Εγχειρίδιο χρήστη

Sprint / Stroller / Hi Flow StrollerΠίνακας 4Οδηγίες και δήλωση κατασκευαστή – Ατρωσία σε ιατρικό
ηλεκτρικό εξοπλισμό και ιατρικά ηλεκτρικά συστήματα

Οδηγίες και δήλωση κατασκευαστή – Ατρωσία
Το Sprint/Stroller/Hi Flow Stroller προορίζεται για χρήση στο ηλεκτρομαγνητικό περιβάλλον που προσδιορίζεται
παρακάτω. Ο αγοραστής ή ο χρήστης του Sprint/Stroller/Hi Flow Stroller θα πρέπει να διασφαλίσει ότι το σύστημαχρησιμοποιείται σε ανάλογο περιβάλλον.
Δοκιμή ατρωσίας
Επίπεδο δοκιμής IECΕπίπεδοΗλεκτρομαγνητικό περιβάλλον – Οδηγίες
60601
συμμόρφωσης
Αγώγιμεςραδιοσυχνότητες (RF)IEC 61000-4-6Ακτινοβολούμενες
ραδιοσυχνότητες (RF)IEC 61000-4-3

3 Vrms6 Vrms (στις ζώνες ISM)
150 kHz έως 80 MHz

80 MHz έως 2,7 GHz

Δεν εφαρμόζεταιΣυσκευή τροφοδοτούμενη
με μπαταρία, χωρίς SIP/
SOPΟ φορητός και κινητός εξοπλισμός επικοινωνίας
ραδιοσυχνοτήτων δεν θα πρέπει να χρησιμοποιείται
κοντά σε οποιοδήποτε τμήμα του Sprint/Stroller/Hi
Flow Stroller, συμπεριλαμβανομένων των καλωδίων
του, και συγκεκριμένα σε απόσταση μικρότερηαπό τη συνιστώμενη απόσταση διαχωρισμού πουυπολογίζεται από την εξίσωση η οποία ισχύει για τησυχνότητα του πομπού.
Συνιστώμενη απόσταση διαχωρισμού
d = 1,2 √P

10 V/m80 MHz–2,7 GHz
80% AM στο 1 kHz

d = 1,2 √P
d = 2,3 √Pόπου P είναι η μέγιστη τιμή ισχύος εξόδου τουπομπού σε watt (W) σύμφωνα με τον κατασκευαστή
του πομπού και d είναι η συνιστώμενη ενδιάμεση
απόσταση σε μέτρα (m).
Η ισχύς πεδίου από σταθερούς πομπούς
ραδιοσυχνοτήτων, όπως καθορίζεται από μιαηλεκτρομαγνητική μελέτη του χώρου,α πρέπει να είναι
χαμηλότερη από το επίπεδο συμμόρφωσης σε κάθεεύρος συχνοτήτων.βΕνδέχεται να εμφανιστεί παρεμβολή στο περιβάλλον
εξοπλισμού που επισημαίνεται με το ακόλουθοσύμβολο:Συχνότητα ελέγχου Ζώνηα)
(MHz)
(MHz)

Σέρβιςα)Διαμόρφωσηβ)Διαμόρφωση παλμού
18 Hz
FMγ)
±5 kHz απόκλιση1 kHz ημιτονοειδής

385

380–390TETRA 400450

430–470

GMRS 460, FRS 460

704–787

LTE Ζώνη 13, 17β)Μέγιστηισχύς (W)Απόσταση Επίπεδο δοκιμής(m)ατρωσίας (V/m)

1,8

0,3

27

2

0,3

28

Διαμόρφωση παλμούβ)217 Hz

0,2

0,3

9

710
745
780
810
870
930
1720
1845
1970

800–960

GSM 800/900, TETRA 800, Διαμόρφωση παλμούβ)IDEN 820, CDMA 850, LTE
18 HzΖώνη 5

2

0,3

28

1700–
1900

GSM 1800, CDMA 1900,
GSM 1900, DECT, LTEΖώνη 1, 3, 4, 25, UMTS

Διαμόρφωση παλμούβ)217 Hz

2

0,3

28

2450

2400–
2570

Διαμόρφωση παλμούβ)217 Hz

2

0,3

28

5240
5500
5785

Bluetooth, WLAN, 802.11
b/g/n, RFID 2450, LTEΖώνη 7

5100–
5800

WLAN 802.11 a/nΔιαμόρφωση παλμούβ)217 Hz

0,2

0,3

9

Σημειωση: Εάν είναι απαραίτητο να επιτευχθεί το ΕΠΙΠΕΔΟ ΔΟΚΙΜΗΣ ΑΤΡΩΣΙΑΣ, η απόσταση μεταξύ της κεραίας μετάδοσης καιτου Sprint/Stroller/Hi Flow Stroller μπορεί να μειωθεί στο 1 m. Η απόσταση δοκιμής 1 m επιτρέπεται από το IEC 61000-4-3.
Για ορισμένες υπηρεσίες, περιλαμβάνονται μόνο οι συχνότητες ανοδικής ζεύξης.
Το φέρον σήμα πρέπει να διαμορφωθεί με χρήση σήματος τετραγωνικού κύματος κύκλου λειτουργίας 50%.
Ως εναλλακτική της διαμόρφωσης FM, μπορεί να χρησιμοποιηθεί διαμόρφωση παλμού 50% στα 18 Hz διότι, παρότι δεν αντιπροσωπεύει
την πραγματική διαμόρφωση, θα ήταν η χειρότερη περίπτωση.
αβγΗ CAIRE και η CAIRE Inc. είναι σήματα κατατεθέντα της CAIRE Inc. Επισκεφθείτε την ιστοσελίδα μας παρακάτω για μια πλήρη λίστα εμπορικών σημάτων. Εμπορικά σήματα: www.
careinc.com/corporate/trademarks.
Copyright © 2023 CAIRE Inc. Η CAIRE Inc. διατηρεί το δικαίωμα να διακόψει τα προϊόντα της, ή να αλλάξει τις τιμές, τα υλικά, τον εξοπλισμό, την ποιότητα, τις περιγραφές, τιςπροδιαγραφές ή/και τις διαδικασίες για τα προϊόντα της ανά πάσα στιγμή χωρίς προηγούμενη ειδοποίηση και χωρίς περαιτέρω υποχρέωση ή συνέπεια. Όλα τα δικαιώματα που δεναναφέρονται ρητά στο παρόν διατηρούνται από την εταιρεία μας, ανάλογα με την περίπτωση.

PN MN236-C4 E | Εγχειρίδιο χρήστη GR - 169

Sprint / Stroller / Hi Flow Stroller
Sembol Sözlüğü
ISO 7000Ünitenin yakınındayken veya üniteyiçalıştırırken sigara içmeyin. P002 sayılı
yönetmelik
BF Tipi uygulamalı parça (elektrikçarpmasına karşı koruma derecesi).
5333 sayılı yönetmelikYağmurdan koruyun, kuru tutun.
0626 sayılı yönetmelik
Sayıya göre yığınlama sınırı. Reg. #2403Üreticinin adı ve adresi. 3082 sayılı
yönetmelik

Uyarı. W001 sayılı yönetmelik
Konsey Direktifi 93/42/EEC
Avrupa Topluluğundaki yetkili temsilciÜlke ve üretim tarihi. "CC", üretildiği
ülkenin iki harfli ülke kodunu
tanımlar. Üretim tarihi YYYY-AA-GG
formatındadır. Kayıt # 6049
Dikkatli olun, ürünle verilen belgelere
bakın. 0434A sayılı yönetmelik
Katalog Numarası. 2493 sayılı
yönetmelikÜrüne özgü cihaz tanımlayıcı (UDI)
etiketi üzerinde CE #### sembolü varsa,
cihaz tıbbi cihazlarla ilgili 93/42 / EEC
Direktifinin gerekliliklerine uygundur.
CE #### sembolü onaylanmış kuruluş
numarasını gösterir.
Bu cihaz, tıbbi cihazlara ilişkin
2010/35/EU Direktifinin
gereksinimlerini karşılamaktadır.
Gösterilen pi işaretini taşır.

Seri Numarası. 2498 sayılı yönetmelik
Depolama veya çalışma sıcaklığı
aralığı.. 0632 sayılı yönetmelik

ADR
Zehirli olmayan gaz.

Depolama nem aralığı. 2620 sayılı
yönetmelik

Tehlike Oksitleyici maddeler: yangınıyoğunlaştırma riski. Oksitleyici
maddeler yangınların daha şiddetli
yanmasına neden olur.
Dondurulmuş Sıvı, USP; Havanın
Sıvılaştırılmasıyla Elde Edilir

Atmosfer basıncı sınırlaması. 2621
sayılı yönetmelikÜst kısmıdır. 0623 sayılı yönetmelik
Ek Semboller
Dikkat, kırılır. 0621 sayılı yönetmelikÜnitenin her zaman yeterli şekilde
havalandırılmasını sağlayın

Tehlikeli maddeler içerir. Kayıt # 3723

Yanıcı malzemeler, yağ ve gresten uzak
tutun.

İthalatçı. Kayıt # 3725
ISO 7010Soğuk sıvıyla veya gazlı oksijenle yada donmuş parçalarla temas durumundasoğuktan yanma meydana gelebilir.
Düşük sıcaklık uyarısı. Düşük sıcaklığa
veya dondurucu koşullara karşı uyarmak
içindir. W010 sayılı yönetmelik
Kullanıcı el kitabı okunmalıdır.
M002 sayılı yönetmelik
Açık alev, yangın ve kıvılcımdan uzak
tutun. Yakınında açık alev kaynağı
bulundurmak ve sigara içmek yasaktır.
P003 sayılı yönetmelik
170 - TR PN MN236-C4 E | Kullanıcı El Kitabıİsviçre'deki yetkili temsilci.
Cihaz, onaylanmış kuruluş numarasını
gösteren UKCA#### ile gösterildiği
gibi UKCA işaretini taşıyorsa, bu cihaz
UKCA yönetmeliklerine uygundur.
IEC 60417

Ünitenin üzerini örtmeyin veya portatifüniteyi giysinizin altında taşımayın. Buüniteler normalde oksijeni tahliye eder.
5641 numara

Sprint / Stroller / Hi Flow Stroller
21 CFR 801.15
Federal yasalar uyarınca bu cihazyalnızca bir hekim tarafından veya
hekimin talimatıyla satılabilir.
IEC 60601-1
Su geçirmez
Konsey Direktifi 2012/19/EU
Weee
Bu sembol, ekipman sahiplerine atık
elektrik ve elektronik ekipman (WEEE)
yönergesine göre ömrünün sonunda bir
geri dönüşüm tesisine iade etmelerini
hatırlatmaktır.
Ürünlerimiz Tehlikeli Maddelerin(ROHS) yönergesinin kısıtlanmasına uygun olacaktır. Eser miktarda kurşun veya
diğer tehlikeli maddi içerik içermezler.

ISO 15223-1
Tıbbi cihaz. Kayıt # 5.7.7
Benzersiz cihaz tanımlayıcısı # 5.7.10
Bu ürün, ABD’de ve tüm dünyada geçerli bir veya
daha fazla patent kapsamında olabilir. Geçerli
patentlerin listesini görmek için lütfen aşağıdaki
web sitemizi ziyaret edin. Pat.: www.caireinc.com/
corporate/patents/.

Teknik Özellikler• Çalışma Modu: Sürekli Akış• Elektrik Çarpmasına Karşı Koruma Türü: Dahili Güç Beslemeli Ekipman• Elektrik Çarpmasına Karşı Koruma Derecesi: BF Tipi Uygulamalı Parça• Su Girişine Karşı Koruma Derecesine Göre Sınıflandırma: IP22 – Normal Ekipman• Ekipman, yanıcı karışımlar bulunan ortamlarda kullanım için uygun değildirÜrün Teknik Özellikleri

SprintStroller

Hi Flow Stroller

LOX Kapasitesi

0,68 kg (1,5 lb)

1,36 kg (3,0 lb)

1,36 kg (3,0 lb)Gazlı Eşdeğer KapasiteAğırlık, Boş

513 L
2,04 kg (4,5 lb)

1.026 L2,27 kg (5,0 lb)

1.025 L2,49 kg (5,5 lb)Ağırlık, Dolu

2,72 kg (6,0 lb)

3,63 kg (8,0 lb)

3,86 kg (8,5 lb)

Yükseklik

298 mm (11,75 inç)

343 mm (13,5 inç)

343 mm (13,5 inç)

123,83 mm (4,875 inç) D x
191 mm (7,5 inç) G
2 LPM’de tipik kullanım süresi 4,3 saat

149,23 mm (5,875 inç) D x
191 mm (7,5 inç) G
8,5 saat

149 mm (5,875 inç) D x
191 mm (7,5 inç) G
8,5 saatÇalışma basıncı

137 kPa (20 psi)

137 kPa (20 psi)

137 kPa (20 psi)

Normal Buharlaşma Oranı

0,57 kg/gün (1,3 lb/gün)

0,57 kg/gün (1,3 lb/gün)

0,57 kg/gün (1,3 lb/gün)

BoyutlarStandart Akış Kontrolü Aralığı Kapalı, 0,25; 0,5; 0,75; 1; 1,5; Kapalı, 0,25; 0,5; 0,75; 1; 1,5; Kapalı, 0,5; 1; 2; 2,5; 3; 4; 6;
2; 2,5; 3; 4; 5; 6 LPM
2; 2,5; 3; 4; 5; 6 LPM
8; 10; 12; 15 LPMAkış Hızı Doğruluğu*+/-%10+/-%10+/-%10* Bu doğruluk değeri yalnızca 70 F, 14,7 psig ve doğru kalibre edilmiş bir kütle akış ölçer için geçerlidir.

PN MN236-C4 E | Kullanıcı El Kitabı TR - 171

Sprint / Stroller / Hi Flow Stroller
Uyarı BilgileriÖnemli: Sprint/Stroller/Hi Flow Stroller’ıçalıştırmadan önce bu el kitabının tamamını okuyun.
Yalnızca reçete ile satılır.
UYARI: BU CIHAZ HAYATI IŞLEVLER IÇINKULLANILAMAZ.
UYARI: EKIPMANIN DÜZGÜN ÇALIŞMADIĞINI
DÜŞÜNÜYORSANIZ SAĞLIK HIZMETI SAĞLAYICINIZIARAYIN. ÜNITEYI KENDINIZ ONARMAYA VEYAAYARLAMAYA ÇALIŞMAYIN.
UYARI: BU EKIPMANI, ÜRETICININ IZNI OLMADAN
MODIFIYE ETMEYIN.
UYARI: SÜREKLI ŞEKILDE OKSIJEN SAĞLANMASIGEREKIYORSA TEDAVI SIRASINDA HER ZAMANYETERLI OKSIJEN VE/VEYA IKINCI BIR OKSIJEN
KAYNAĞINI HAZIR BULUNDURUN.
UYARI: CIHAZIN 3 METRE (10 FIT) YAKININDA SIGARAIÇILMESINE, MUM YAKILMASINA VEYA AÇIK ALEVKAYNAKLARININ BULUNMASINA IZIN VERMEYIN VE
CIHAZI HERHANGI BIR TUTUŞTURMA KAYNAĞINA 20
CM’DEN (8 INÇ) FAZLA YAKLAŞTIRMAYIN.
UYARI: ÜNITENIZI YETERLI DERECEDEHAVALANDIRILAN BIR ALANDA TUTUN.
UYARI: SIVI OKSIJEN EKIPMANLARINIBIR DOLAPTA, ARABA BAGAJINDA VEYABAŞKA BIR KAPALI ALANDA SAKLAMAYIN.
EKIPMANLARIN ÜSTÜNE BATTANIYE, BEZ
VEYA BAŞKA BIR TÜR KUMAŞ ÖRTMEYIN.
UYARI: BU ÜRÜN, CALIFORNIA EYALETI TARAFINDANKANSERE NEDEN OLDUĞU BILINEN NIKEL GIBIKIMYASAL MADDELERE MARUZ KALMANIZAYOL AÇABILIR. DAHA FAZLA BILGI IÇIN WWW.
P65WARNINGS.CA.GOV WEB SITESINI ZIYARET EDIN.
UYARI: BU CIHAZLA CIDDI BIR OLAY OLMASIDURUMUNDA, KULLANICI OLAYI DERHALSAĞLAYICIYA VE / VEYA ÜRETICIYE BILDIRMELIDIR.
CIDDI BIR OLAY, OLAYIN TEKRARLANMASIDURUMUNDA YARALANMA, ÖLÜM VEYA YARALANMA/ ÖLÜME NEDEN OLMA POTANSIYELI OLARAKTANIMLANIR. KULLANICI AYRICA OLAYI, OLAYIN
MEYDANA GELDIĞI ÜLKEDEKI YETKILI MAKAMARAPOR EDEBILIR.
Dikkat: Sprint/Stroller/Hi Flow Stroller’ı, yalnızca doktorunuz tarafından belirtilen biçimde
kullanın.
Dikkat: Federal yasalar uyarınca bu cihaz yalnızca bir
hekim tarafından veya hekimin talimatıyla satılabilir.

172 - TR PN MN236-C4 E | Kullanıcı El Kitabı

Dikkat: Ünite, yaklaşık -184°C (-300°F) olan çok soğuk
bir sıvı oksijen içerir. Böyle düşük bir sıcaklığa maruz
kalmak, ciddi soğuk yanmasına neden olabilir.
Dikkat: Sıvı ve gaz halindeki oksijen, yanıcı olmasa
da başka malzemelerin normalden daha hızlı şekilde
yanmasına neden olur. Bu tehlikeden ve sıvı oksijenindüşük sıcaklığından dolayı belirli güvenlik önlemleri
almak gerekir.
Dikkat: Yanıcı malzemeleri bu ekipmandan uzak tutun.
Yüz kremleri ve vazelin gibi aerosol spreylerle yağlar
kolayca alev alır ve ortamda oksijen bulunduğunda
bunlar hızla yanabilir.
Dikkat: Oksijen kanülü takılıyken sigara içmek, yüzde
yanmalara neden olabilir ve ölümle sonuçlanabilir.
Kanülün çıkarılıp giysi, yatak örtüsü, kanepe ya dabu türde diğer eşyaların üzerine bırakılması sigara,
ısı kaynağı, kıvılcım veya aleve maruz kalındığında
parlama sonucu alev almaya neden olacaktır.
Sigara içiyorsanız lütfen: (1) taşınabilir üniteyi kapatın,(2) kanülü çıkarın ve (3) cihazın bulunduğu odadançıkın.
Dikkat: Yanlışlıkla devrildiğinde mümkünse üniteyi
hemen dikkatli bir şekilde dik konuma getirin. Sıvı
oksijen sızıntısı varsa alanı derhal terk edin ve sağlık
hizmeti sağlayıcınızı arayın. Üniteyi hareket ettirmeyeveya sıvı oksijen kaçağını durdurmaya çalışmayın.
Not: Ünitenin donmuş parçalarına dokunmayın.
Not: Taşınabilir üniteyi sabit üniteye takılı halde saklamayın
veya çalıştırmayın.
Not: Eğitimsiz personelin bu cihazı taşımasına veya
çalıştırmasına izin vermeyin.
Not: Bu cihazın ticari yolcu ve hava nakliye uçuşlarında
kullanımı Federal Havacılık İdaresi tarafından yasaklanmıştır.

Kullanım amacı

CAIRE Sprint/Stroller/Hi Flow Stroller ilave oksijenin
uygulanması için tasarlanmıştır. Cihaz yaşam desteği
için tasarlanmamıştır ve herhangi bir hasta izleme
yeteneği sağlamamaktadır.

Sprint / Stroller / Hi Flow Stroller
Giriş

Kontroller

Sprint, Stroller ve Hi Flow Stroller, son kullanıcının
evinde hastaya ek oksijen uygulamaya yöneliktir.
Bakım evleri veya yarı sürekli bakım merkezlerindede kullanılabilir. Sprint/Stroller/Hi Flow Stroller, uzun
süreli ayakta tedavi için oksijen kaynağı sağlar.

1. Elektronik Sıvı Seviye Göstergesi
2. Elektronik Sıvı Seviye Anahtarı
3. Mekanik Sıvı Seviye Göstergesi
4. Akış Kontrolü Düğmesi
5. Oksijen Tüpü Konnektörü
6. Havalandırma Valfi
7. Dolum Bağlantıları (QDV Tanımı için sonraki
sayfaya bakın)
8. Omuz Tamponlu Taşıma Askısı
9. Yoğuşma Tamponu
10. Yoğuşma Kabı

Cihaz, KOAH hastaları veya solunum kapasitelerinde
azalma olan kişiler tarafından kullanılır. Cihaz hastaya
reçete edilir. Cihaz, Sprint/Stroller/Hi Flow Stroller’ınçalıştırılması ve servisinin yapılması konusunda eğitim
almış sağlayıcılara satılır. Sağlayıcı, kullanıcıyı eğitir.
Bu cihaz yaşam desteğine yönelik olmadığı gibi hasta
izleme özelliklerine de sahip değildir. Mekanik arıza
olasılığına karşı alternatif bir ek oksijen kaynağının
hazır bulundurulması önerilir. Bu taşınabilir cihazlarda,taşınabilir cihaza dolum yapmak için bir hazne sıvı
oksijen cihazına bağlanabilecek hızlı açılır bir valf
de bulunur. Hazne ünitesi, sağlık hizmeti sağlayıcınız
tarafından doldurulur.

1
2
4
5

Bu kullanıcı el kitabı, Sprint/Stroller/Hi Flow
Stroller’ın kullanımına ilişkin yönergeleri içerir.
Sabit ünitenin kullanımı için üniteyle birlikte verilen
kullanıcı el kitabını inceleyin.

8

Not: Hizmet sağlayıcı, ünitenin ilk kurulumunda yardımcı
olacak, doğru idare ve kullanımı konusunda bilgi verecektir.

5

7

Sprint, Stroller ve Hi Flow Stroller üniteleri

6

10
9

PN MN236-C4 E | Kullanıcı El Kitabı TR - 173

Sprint / Stroller / Hi Flow Stroller
Hızlı Açılır Valf (QDV) TanımıTaşınabilirÜniteKonnektörü

Sabit Ünite
Konnektörü

3. Dolum konnektörünü sabit ünitenin konnektörüüzerine dik konumda yerleştirin.

QDV Stili

CAIRE
SF
YandanKilitleme
Doldurmalı

LIFE-OX
TF
Kilitleme

4. a. Konnektörünüz bastırmalı ise:
Konnektörün yerine oturduğunu hissedene kadar
Sprint/Stroller/Hi Flow Stroller ünitesini sabitünitenin üzerine indirin.

PENOX

TAEMA
TF
Bastırma
PB

b. Konnektör Penox veya Life-Ox ise:
Konnektörün takıldığını görene dek Sprint/
Stroller/Hi Flow Stroller’ı döndürün. Ardından,
dikkatli ve kuvvetli bir şekilde Sprint/Stroller/
Hi Flow Stroller’ı aşağı doğru bastırırken saat
yönünde () döndürün.

Çalıştırma TalimatlarıÜstten Doldurmalı Sprint/Stroller/Hi Flow
Stroller’ı Doldurma
1. Hem sabit ünite hem de Sprint/Stroller/Hi Flow
Stroller üzerindeki dolum konnektörlerini temiz,
kuru ve tüy bırakmayan bir bezle temizleyin.
Dikkat: Nem, Sprint/Stroller/Hi Flow Stroller vesabit ünitenin donarak birbirine yapışmasınaneden olabileceğinden dolum konnektörleri
kuru olmalıdır.
2. Hem Sprint/Stroller/Hi Flow Stroller hem de hazneüzerindeki akış kontrolü düğmelerinin kapalı (0)konumda olduğundan emin olun.

174 - TR PN MN236-C4 E | Kullanıcı El Kitabı

Not: Yandan Doldurmalı Sprint/Stroller/Hi Flow Stroller’ı
Doldurma için 6. adımdan devam etmektedir.

Yandan Doldurmalı Sprint/Stroller/Hi Flow
Stroller’ı Doldurma
1. Hem sabit ünite hem de Sprint/Stroller/Hi Flow
Stroller üzerindeki dolum konnektörlerini temiz,
kuru ve tüy bırakmayan bir bezle temizleyin.
UYARI: NEM, SPRINT/STROLLER/HI FLOWSTROLLER VE SABIT ÜNITENIN DONARAKBIRBIRINE YAPIŞMASINA NEDEN OLABILECEĞINDEN DOLUM KONNEKTÖRLERI KURU
OLMALIDIR.

Sprint / Stroller / Hi Flow Stroller
2. Hem Sprint/Stroller/Hi Flow Stroller hem de hazneüzerindeki akış kontrolü düğmelerinin kapalı (0)konumda olduğundan emin olun

Not: Sprint/Stroller/Hi Flow Stroller doldurulduktan
30 saniye veya daha uzun süre sonra sıvı seviyesi
göstergesinden en doğru okumalar elde edilir.

3. Sprint/Stroller/Hi Flow Stroller dolum konnektörünü
dik konumda sabit ünite konnektörüne bağlayın.

4. İğne ile yuvanın birbirine geçtiğini hissedene kadar
Sprint/Stroller/Hi Flow Stroller’ı saat yönünün
tersine () döndürün.

7. Bağlantının çevresinde buhar da görebilirsiniz.
Bu durum normaldir ve sıvı oksijen ile ünitelerinçevresindeki sıcak hava arasındaki büyük sıcaklık
farkından kaynaklanır.
8. Ünite dolu olduğunda havalandırma valfini kapatın.
Tıslama sesinin tonu değiştiğinde ve tahliye
hattından biraz sıvı çıktığında ünitenin dolu olduğunu
anlayabilirsiniz.
Not: Neredeyse kesintisiz şekilde devam eden bir sıvı
tahliyesinden sonra valf kapatıldığında Sprint/Stroller/Hi
Flow Stroller tamamen dolar ancak bir miktar sıvı oksijenkaçağı olur.

5. Sprint/Stroller/Hi Flow Stroller’ı dikkatli ve kuvvetli
bir şekilde dik konumuna geri döndürün. Üniteler
birbirine kenetlenir.

Not: İlk sıvı tahliyesi işaretinden sonra valf kapatılırsa
Sprint/Stroller/Hi Flow Stroller tamamen dolmaz ancak sıvı
oksijen korunur.
9. Bastırmalı konnektörlerde üniteler birbirinden
ayrılana kadar serbest bırakma düğmesini basılı
tutun. Diğer tüm konnektörlerde üniteyi sabitüniteden ayrılana kadar döndürün.

Not: 6. adımdan devam etmektedir.
Not: Bastırmalı konnektörde dolum boyunca üniteyi
yerinde tutun.

Taşınabilir Üniteyi Serbest Bırakma Düğmesi
(yalnızca bastırmalı stil)6. Sprint/Stroller/Hi Flow Stroller’ı doldurmayabaşlamak için havalandırma valfini açın. Oksijençıkma sesini duyacaksınız; endişelenmeyin.
Duyduğunuz tıslama sesi tamamen normaldir.

PN MN236-C4 E | Kullanıcı El Kitabı TR - 175

Sprint / Stroller / Hi Flow Stroller
Dikkat: Sıvı seviyesi ölçeğini yalnızca valf
kapatıldıktan sonra kontrol edin.

Çalıştırma Talimatları

Dikkat: Sprint/Stroller/Hi Flow Stroller bir sürekullanıldıktan sonra hemen dolduruluyorsa üniteyi
doldurun ve sonra FCV sıfıra ayarlanmış durumdayken
basıncın kararlı hale gelmesi için dolumdan sonra 10
dakika bekleyin. Daha sonra üniteyi normal şekildeçalıştırın.
UYARI: SPRINT/STROLLER/HI FLOWSTROLLER’I SABIT ÜNITEYE BAĞLIYKEN
ÇALIŞTIRMAYIN.
UYARI: SPRINT/STROLLER/HI FLOW STROLLER
KOLAYCA AYRILMIYORSA KUVVET UYGULAMAYIN.
ÜNITELER DONARAK BIRBIRINE YAPIŞABILIR.
ÜNITELERI BIRBIRINE YAPIŞIK HALDE BIRAKIN VEÜNITELER ISINANA KADAR BEKLEYIN. DAHA SONRAKOLAYCA AYRILACAKLARDIR. DONMUŞ PARÇALARADOKUNMAYIN. UYARI: SIVI AKIŞI GÖRÜLECEK KADARÇOK SIZMA VARSA DERHAL ALANI TERK EDIN VESAĞLIK HIZMETI SAĞLAYICINIZI ARAYIN.
UYARI: SIVI VEYA SOĞUK GAZ HALINDEKI OKSIJEN,
CILTTE SOĞUK YANMASINA NEDEN OLABILIR.
OKSIJEN, YANGIN TEHLIKESINI ARTIRIR.
Dikkat: Üniteleri ayırdıktan sonra sabitüniteden veya Sprint/Stroller/Hi Flow
Stroller’dan sıvı sızıntısı olması durumunda
Sprint/Stroller/Hi Flow Stroller’ı dikey şekilde
bir kenara koyun, odadan çıkın ve derhalsağlık hizmeti sağlayıcınızı arayın.
Not: Ekipman Sağlayıcıya: Sprint / Stroller / Hi Flow
Stroller ile kullanım için aşağıdaki oksijen uygulama
aksesuarları önerilir:• 7 fit (2,1 m) tüplü Burun Kanülü (Maksimum LPM: 6
LPM): CAIRE Parça Numarası 5408-SEQ• Ateş Çukuru: CAIRE Parça Numarası 21126636
Herhangi bir kanül ile kullanım için bir ateş kesilmesi
önerilir.
• CAIRE, portatif oksijen ile birlikte kullanılması amaçlanan
bir ateş kesmesi sunar. Firebreak, aşağı akış kanülünün
veya oksijen tüpünün tutuşması ve ateş bariyerine
yanması durumunda gaz akışını durdurmak için termal
bir sigortadır. Hasta ve Sprint / Stroller / Hi Flow Stroller
oksijen çıkışı arasındaki burun kanülü veya oksijen tüpü
ile aynı hizadadır. Firebreak'in doğru kullanımı için daimaüreticinin talimatlarına bakın (her firebreak kitiyle birlikte
verilir).
• Ek önerilen aksesuar bilgileri çevrimiçi olarak www.
caireinc.com adresinde mevcuttur.

176 - TR PN MN236-C4 E | Kullanıcı El Kitabı(Kanül ürünle birlikte gönderilmez.)

1. Kanül nefes alma tüpünü, oksijen tüpü
konnektörüne sıkıca bastırın.
2. Kanül nefes alma tüpünüzü, oksijeni rahatça
soluyabileceğiniz doğru konuma ayarlayın.
3. Reçete edilmiş akış hızı (sayı olarak) çevirmelidüğmenin “penceresinde” görünene ve kilitlenmehissedilene kadar Sprint/Stroller/Hi Flow Stroller’ınüst kısmındaki akış kontrolü düğmesini saat
yönünde çevirin.
Dikkat: Akış kontrolü düğmesinin akış hızları
arasında bir değere ayarlanması durumunda,
belirtilen koşulların dışında oksijen akışı
gerçekleşecektir.
4. Artık oksijen alıyor olmanız gerekir. Oksijenakışını durdurmak için akış kontrolü düğmesini saat
yönünün tersine çevirerek kapalı (0) konumuna
getirin.
Dikkat: Cihazı kullanmadığınızda akış kontrolüdüğmesini her zaman kapalı (0) konumuna
getirin.

Sprint / Stroller / Hi Flow Stroller
5. Sprint/Stroller/Hi Flow Stroller’ın çalıştırılacağı
süreyi belirlemek için aşağıdaki tabloyu kılavuz
olarak kullanın:
FCV Ayarı
Model

SprintStroller

Hi Flow
Stroller

NominalNominal

FCV
Kon.

SprintStroller

Hi Flow
Stroller

Kapalı

30:00

60:00

71:00

0,25

18:00

30:00

Yok

0,5

12:12

23:00

32:00

0,75

9:18

17:24

Yok

1

7:30

14:06

16:00

1,5

5:30

10:12

Yok

2

4:18

8:00

8:00

2,5

3:36

6:36

6:20

3

3:06

5:36

5:20

4

2:24

4:24

4:00

5

2:00

3:36

Yok

6

1:42

3:00

2:40

8

Yok

Yok

2:00

10

Yok

Yok

1:40

12

Yok

Yok

1:20

15

Yok

Yok

1:00

Not: Süreler, saat ve dakika (00:00 formatında) olarak
gösterilmiştir.
Not: “Nominal” süreler, maksimum doldurma, tamakış hızları, kabul edilebilir kayıp oranı, ünitenin
yerinden oynatılmaması vs. gibi ideal koşullar içinverilmiştir.
Bu süreler, öngörülen maksimum sürelerdir.

• 4. Nesil Gösterge: Göstergenin dış tarafındakiyeşil çalıştırma düğmesine basın. LED’lerin içerik
seviyesini gösteren eğrilerini okuyun.

Dikkat: Işık çubuğunun yalnızca son segmenti
yanıyorsa Sprint/Stroller/Hi Flow Strollerboştur.
Düğmeye basıldığında sadece Düşük Pil Göstergesi
yanarsa sağlık hizmeti sağlayıcınızı bilgilendirin.
Not: Sprint/Stroller/Hi Flow Stroller, ünitede sıvı oksijenolduğu sürece Düşük Pil Göstergesi yansa bile oksijensağlamaya devam eder.
6b. Ünitedeki sıvı oksijen seviyesini mekanik sıvı
seviye göstergesiyle doğrulamak için:• Siyah içerik göstergesi yaylı ölçek bölmesinin
hemen üzerinden üniteyi kumaş askıyla destekleyin.
• Ünitenin stabilize olmasını (sıvının hareketsiz halegelmesini) bekleyin, ardından yaklaşık sıvı oksijen
içeriği için açığa çıkan renkli ölçeği okuyun.

Not: Sizin elde edeceğiniz gerçek sonuçlar farklı
olacaktır.
6a. Ünitedeki sıvı oksijen seviyesini elektronik sıvı
seviye göstergesiyle doğrulamak için:• 3. Nesil Gösterge: Ünitenin üstündeki düğmeyi
(sıvı seviye anahtarı) en az iki saniye basılı tutun.
İçerik seviyesini görmek için ışık çubuğunun üstünü
okuyun.

PN MN236-C4 E | Kullanıcı El Kitabı TR - 177

Sprint / Stroller / Hi Flow Stroller
Not: Ünite boş ise yalnızca ölçeğin kırmızı renkli bandı
gösterilir. Böyle bir durumda, ünite kullanılmadan öncedoldurulmalıdır.
7. Kullanımın sürekli olduğu ve özellikle taşınabilirünitenin hareket etmediği bazı ortam koşullarında,vaporizatör bobininin çevresinde ve kasa üzerindeaşırı buzlanma oluşabilir. Bu buzlanmayı, üniteye
hafifçe vurarak ve/veya biriken buzları kasadan
temizleyerek azaltabilirsiniz.

Yoğuşma Tamponu BakımıÜnite boşalıp oda sıcaklığına geldikten sonra yoğuşma
kabında kalan nemi temizleyin.
Sert çanta için• Üniteyi bir tarafa doğru yatırın. Yoğuşma
kabının ortasındaki düğmeyi (ünitenin altında)tutun ve çekin. Böylece kap serbest kalacaktır.
Not: Tampon çok kirlenmişse makinede yıkanabilir (dezenfeksiyon için çamaşır suyu kullanılabilir).
• Kabı ünitenin altındaki yuvarlak açıklığın
üzerine ortalayarak takın. Kabı takmak içindüğmeye tek bir tıklama sesi duyana dek
bastırın ve ardından kabı yerine kilitlemek için
ikinci bir tıklama sesi duyana dek düğmeye
tekrar bastırın.

Yumuşak çanta için• Üst fermuarı açın ve alt kapağı geri çekin.
• Tamponu kaptan çıkarın ve emilen nemi sıkarak
kurutun. Kaba tekrar koymadan önce tamponun
tamamen kurumasını bekleyin.
Not: Tampon çok kirlenmişse makinede yıkanabilir(dezenfeksiyon için çamaşır suyu kullanılabilir).
• Tamponu kapağa yerleştirip yerine takın ve alt
fermuarı tekrar kapatın.
Not: Düğme itilmez ve kabı yerine takmazsa kabı tutun vedüğme bir tık sesiyle açık konuma gelene dek düğmeyi
çekin. Şimdi daha önce açıklandığı gibi kabı serbest
bırakın.

Buzlanmayı Azaltma
Kullanımın sürekli olduğu ve özellikle taşınabilirünitenin hareket etmediği bazı ortam koşullarında,vaporizatör bobininin çevresinde ve kasa üzerinde aşırı
buzlanma oluşabilir. Bu buzlanmayı, üniteye hafifçevurarak ve/veya biriken buzları kasadan temizleyerek
azaltabilirsiniz.
UYARI: AŞIRI NEM/BUZ OLUŞMA OLASILIĞI
NEDENIYLE, NEM OLUŞUMUNU AZALTMAKIÇIN HI FLOW STROLLER’IN KULLANIMDANSONRA 1 SAAT BOYUNCA ÇALIŞTIRILMAMASI ÖNERILIR.

Kilitli
TakılıAçık

Kap Düğmesi

Not: Düğme itilmez ve kabı yerine takmazsa kabı tutun vedüğme bir tık sesiyle açık konuma gelene dek düğmeyi çekin. Şimdi daha önce açıklandığı gibi kabı serbest bırakın.

178 - TR PN MN236-C4 E | Kullanıcı El Kitabı

Hi Flow Stroller’da görülen yüksek akış hızları
nedeniyle ünitede, özellikle de 10 LPM ve dahayüksek akış ayarlarında yoğuşma ve biraz buzlanmaoluşacaktır. Sürekli kullanım nedeniyle aşırı yoğuşma
ve buzlaşma oluşumu olasılığını azaltmak içinünitenin kullanımdan sonra en az 1 saat boyuncaçalıştırılmaması önerilir.

Sprint / Stroller / Hi Flow Stroller
Sorun GidermeAşağıdaki bilgiler, Sprint/Stroller/Hi Flow Stroller’ı kullanırken karşılaşabileceğiniz basit çalışma sorunlarını
gidermenize yardımcı olmak için verilmiştir.
Sorun
Çözüm
Taşınabilir ünite tıslama sesi • Taşınabilir ünitede doğru çalışma basıncını korumak için tıslama sesi oluşabilir. Doldurmaçıkarıyor.
işleminden sonra veya Taşınabilir ünitenin konumu değiştirildiğinde tıslama sesinin duyulma
olasılığı daha yüksektir. Tıslama sesi, doldurma işleminden sonra yaklaşık 10 dakika
sürebilir. Akış kontrolü valfi düşük ayarda olduğunda da bu ses duyulabilir.

Taşınabilir akışı kullanım
sırasında duruyor.

• Taşınabilir ünite hatalı bir pozisyonda yerleştirilmişse taşınabilir üniteyi dik konuma getirinve ünitenin stabilize olması için birkaç dakika bekleyin.
• Kanülün, Oksijen Çıkışına sıkı bir şekilde takılı olduğundan emin olun.
• Kanülün bükülmediğinden emin olun.
• İçerik göstergesini/seviye ölçeri kontrol edin ve gerekiyorsa taşınabilir üniteyi doldurun.

Taşınabilir ünite dolmuyor.

• Akış kontrolü düğmesinin kapalı (“0”) konumda olmadığından emin olun.
• Haznede oksijen olduğunu doğrulayın.

• Doldurma işlemi boyunca Taşınabilir ünitenin ve Hazne doldurma konnektörlerinin birbirine
tamamen geçtiğinden emin olun.
Taşınabilir ünite• Havalandırma valfi kapanmazsa ve tıslama sesi ve oksijen buhar bulutu devam ediyorsa
havalandırma valfi, doldurma Haznedeki serbest bırakma düğmesine basarak Taşınabilir üniteyi dikkatlice çıkarın.
işleminin sonunda düzgün
Taşınabilir ünitenin altından yapılan havalandırma birkaç dakika içinde duracaktır.
şekilde kapanmıyor.
Havalandırma valfi kapatabileceğiniz sıcaklığa gelene dek ünitenin ısınmasını bekleyin.
Taşınabilir ünitede, doğru oksijen akışı için basıncın yeterli hale gelmesi 60 dakikayı
bulabilir. Gerekirse Hazneye takılı akış kontrolü valfi gibi alternatif bir oksijen kaynağı
kullanın.
Taşınabilir ünite, doldurma• Taşınabilir ünite ve Hazne doldurma konnektörleri donmuş olabilir.
işleminden sonra Hazneden• KUVVET UYGULAMAYIN. Donmuş parçaların ısınması için birkaç dakika bekleyip buz
kolayca çıkmıyor.
eridiğinde Taşınabilir Üniteyi çıkarın. Ünitelerin birlikte donmasını önlemek için Haznedeki
erkek doldurma konnektörünü ve Taşınabilir ünitedeki dişi doldurma konnektörünü
doldurma işleminden önce temiz ve kuru bir bezle silin.

PN MN236-C4 E | Kullanıcı El Kitabı TR - 179

Sprint / Stroller / Hi Flow Stroller
Temizlik Standardı
UYARI: ÜNITEYI YALNIZCA BOŞKENTEMIZLEYIN.
• Hafif bir bulaşık deterjanı ve su çözeltisi
kullanarak temizleyin.
• Temizlik çözeltisini doğrudan tüy bırakmayan
bir beze uygulayın. Onaylı temizleyicilerarasında HydroPure ve HydroKlean vardır.
Sprint/Stroller/Hi Flow Stroller’a temizleyicileridoğrudan püskürtmeyin.
• Dış yüzeyi, temizlenene dek tüy bırakmayan
bezle silin.
Dikkat: Bu üniteleri temizlemek için yüksek
sıcaklığa ve yüksek basınca sahip yıkama
ekipmanları kullanmayın.
• Temizlik malzemesinin dahili bileşenlere veya
valflere bulaşmamasına dikkat edin.
• Kullanmadan önce ünitenin tamamen
kurumasını bekleyin.
Not: Sağlık hizmeti sağlayıcısının dikkatine: Yenidenişleme prosedürleri için ilgili servis kılavuzunu inceleyin.

Bakım
Bu cihazın teknik kılavuza uygun olarak bakımının
yapılması, hizmet sağlayıcınızın sorumluluğudur. Her
türlü bakım talebi için hizmet sağlayıcınızla görüşün.
Yoğuşma tamponu, bakımı kullanıcı tarafından
yapılabilecek tek parçadır. Diğer parçaların bakımı
kullanıcı tarafından yapılamaz.
Öngörülen kullanım ömrü en az beş yıldır.
Buzlanma ve olası ekipman hasarlarını önlemek içinhem sabit hem de taşınabilir ünitelerdeki doldurma
konnektörlerini, doldurma işlemlerinin arasında temiz,
kuru ve tüy bırakmayan bezle silin.
Not: Gerekli tüm ek bakım işlemleri, yetkili bir servis
teknisyeni veya servis sağlayıcısı tarafından yapılmalıdır.

Aksesuarlar

WEEE ve RoHS

Bu sembol, ekipmanın sahiplerine ekipmanı
kullanım ömrü sonunda Atık Elektrikli ve
Elektronik Ekipmanlar (WEEE) Direktifine
uygun olarak bir geri dönüşüm tesisine iadeetmelerini hatırlatmaya yöneliktir.
Ürünlerimiz, Tehlikeli Maddelerin Kısıtlanması(RoHS) direktifine uyar. Eser miktardan fazla kurşun
veya başka tehlikeli malzeme içermezler.

Elden Çıkarma
Sprint/Stroller/Hi Flow Stroller ve tüm bileşenlerini,doğru şekilde elden çıkarmak için her zaman ev bakım
hizmeti sağlayıcınıza iade edin. Pilin doğru bir biçimde
elden çıkarılması için şehrinizdeki yerel idari birimlerlede iletişime geçebilirsiniz.

Taşıma ve Depolama
Cihaz dik konumda depolanmalı ve yeterince
havalandırılmalıdır. Cihazın yana yatmasına izinvermeyin. %95’e kadar yoğunlaşmasız nem. Sıcaklık
aralığı: -40°C ile 70°C (-40°F ile 158°F) arası.
Çalıştırma sıcaklığı -10°C ile 40°C (14°F ile 104°F)
arasındadır. Bağıl nem aralığı: yoğunlaşmasız %30 ile%75 arası.
Not: Atmosfer basıncı aralığı: 700 hPa - 1.060 hPa(3.048 m - 305 m rakım) arası.

180 - TR PN MN236-C4 E | Kullanıcı El KitabıTaşıma Arabası PN 20765565

Sprint / Stroller / Hi Flow Stroller
Güvenlik
UYARI: TAŞINABILIR RF ILETIŞIM EKIPMANI (ANTEN KABLOLARI VE HARICI ANTENLER GIBIAKSESUARLAR DAHIL), ÜRETICI TARAFINDAN BELIRTILEN KABLOLAR DAHIL OLMAK ÜZERE SPRINT/
STROLLER/HI FLOW STROLLER’IN HERHANGI BIR PARÇASINA 30 CM’DEN (12 INÇ) YAKIN OLMAYACAKŞEKILDE KULLANILMALIDIR. AKSI TAKDIRDE BU EKIPMANIN PERFORMANSINDA DÜŞÜŞ YAŞANABILIR.
UYARI: BU EKIPMANIN ÜRETICISI TARAFINDAN BELIRTILEN VEYA SAĞLANANLARIN DIŞINDAKI AKSESUARLARIN,
TRANSDÜSERLERIN VE KABLOLARIN KULLANILMASI, BU EKIPMAN IÇIN DAHA FAZLA ELEKTROMANYETIKEMISYONA VEYA DAHA AZ ELEKTROMANYETIK BAĞIŞIKLIĞA NEDEN OLABILIR VE EKIPMANIN DÜZGÜN ŞEKILDEÇALIŞMASINI ÖNLEYEBILIR.
UYARI: BU EKIPMANIN, BAŞKA EKIPMANLARLA BITIŞIK VEYA ÜST ÜSTE KONMUŞ OLARAK KULLANILMASINDANKAÇINILMALIDIR ÇÜNKÜ BU KULLANIM ŞEKLI DOĞRU BIR BIÇIMDE ÇALIŞMAMAYA YOL AÇABILIR. BU
KULLANIM ŞEKLI GEREKLIYSE BU EKIPMAN VE DIĞER EKIPMANLARIN NORMAL ŞEKILDE ÇALIŞIP ÇALIŞMADIĞIGÖZLEMLENMELIDIR.
Dikkat: Elektrikli Tıbbi Ekipmanlar, Elektromanyetik uyumluluk (EMC) açısından özel önlemler gerektirir vebu el kitabında sağlanan EMC bilgileri uyarınca kurulup kullanıma alınmalıdır.
Dikkat: Taşınabilir ve mobil radyo frekansı (RF) haberleşme ekipmanı, Tıbbi Elektrikli Ekipmanı etkileyebilir.
Tablo 1

Yönergeler ve Üreticinin beyanı - Elektromanyetik emisyonlar
Sprint/Stroller/Hi Flow Stroller aşağıda belirtilen elektromanyetik ortamda kullanıma yöneliktir. Sprint/Stroller/
Hi Flow Stroller müşterisi veya kullanıcısı, ünitenin bu tür bir ortamda kullanılmasını sağlamalıdır.
Emisyon testi

Uyumluluk

RF emisyonları

Grup 1CISPR 11

RF emisyonları
CISPR 11

Elektromanyetik ortam - yönergeler
Sprint/Stroller/Hi Flow Stroller, RF enerjisini yalnızca dahili
fonksiyonu için kullanır.
Bu nedenle, RF emisyonları oldukça düşüktür ve bu
emisyonların, yakındaki elektronik ekipmanla herhangi bir
parazite neden olması beklenmez.

Sınıf B

Harmonik emisyonlar
IEC 61000-3-2

Geçerli değil

Voltaj dalgalanmaları/
titreme emisyonları
IEC 61000-3-3

Geçerli değil

Sprint/Stroller/Hi Flow Stroller, konutlar ve konutsal
amaçlarla kullanılan binalara elektrik sağlayan kamusaldüşük voltajlı güç kaynağı şebekesine bağlı olan tesisler dedahil olmak üzere tüm tesislerde kullanım için uygundur.

PN MN236-C4 E | Kullanıcı El Kitabı TR - 181

Sprint / Stroller / Hi Flow Stroller
Tablo 2*: Taşınabilir ve hareketli RF haberleşme ekipmanları ile Sprint/Stroller/Hi Flow Stroller arasında bırakılması
önerilen ayrım mesafeleri
Sprint/Stroller/Hi Flow Stroller, ışınan RF bozulmalarının kontrol altında olduğu bir elektromanyetik alanda kullanıma yöneliktir.
Sprint/Stroller/Hi Flow Stroller müşterisi veya kullanıcısı, taşınabilir ve mobil RF iletişim ekipmanı (vericiler) ve Sprint/Stroller/Hi Flow
Stroller arasında, iletişim ekipmanının maksimum çıkış gücüne göre aşağıda önerildiği şekilde minimum bir mesafeyi koruyarak
elektromanyetik parazitin önlenmesine yardımcı olabilir.
Vericinin nominal
Vericinin frekansına göre ayrım mesafesi
maksimum çıkış gücüm
80 MHz ve 800 MHz
800 MHz - 2,5 GHz
150 kHz - 80 MHzW
d = 1,2√P
d = 2,3√P
d = 1,2√P
0,01
0,12 m0,12 m0,23 m0,10,38 m0,38 m0,73 m
1
1,2 m
1,2 m
2,3 m
10
3,8 m
3,8 m
7,3 m
100
12 m
12 m
23 m
Yukarıda listelenmeyen bir nominal maksimum çıkış gücüne sahip olan vericiler için metre (m) cinsinden önerilen ayrım mesafesi (d),vericinin frekansı için geçerli olan denklem kullanılarak tahmin edilebilir; burada P, verici üreticisine göre vat (W) cinsinden vericinin
maksimum çıkış gücü derecelendirmesidir.
NOT 1: 80 MHz ve 800 MHz’de, daha yüksek frekans aralığına yönelik ayrım mesafesi geçerlidir.
NOT 2: Bu yönergeler her durumda geçerli olmayabilir. Elektromanyetik yayılım; yapılar, nesneler ve insanlar tarafından
gerçekleştirilen absorpsiyondan ve yansımadan etkilenir.

* Bu tablo, belirli düzeylerde ve frekans aralıklarında test edilip yönetmeliklere uygun olduğu anlaşılmış
ekipmanlar için standart gereklilik olarak eklenmiştir.

Tablo 3: Yönergeler ve üreticinin beyanı - Elektromanyetik bağışıklık
Sprint/Stroller/Hi Flow Stroller aşağıda belirtilen elektromanyetik ortamda kullanıma yöneliktir. Sprint/Stroller/Hi Flow Strollermüşterisi veya kullanıcısı, ünitenin bu tür bir ortamda kullanılmasını sağlamalıdır.
Bağışıklık testi
IEC 60601 test düzeyi
Uyumluluk düzeyi
Elektromanyetik ortam - yönerge
Elektrostatik boşalma ± 8 kV kontakt± 8 kV kontakt
Zemin ahşap, beton veya seramik karo
(ESD)olmalıdır. Zemin sentetik materyalle±2 kV, ± 4 kV,±2 kV, ± 4 kV,
kaplanmışsa bağıl nem en az %30 olmalıdır.**±8 kV, ± 15 kV hava
±8 kV, ± 15 kV hava
IEC 61000-4-2
Geçerli değil
Elektriksel hızlı
geçici rejim/patlama
IEC 610004-4
Dalgalanma
IEC 61000-4-5
Güç besleme giriş
hatlarında gerilim düşmeleri, kısa
kesintiler ve gerilimdeğişimleri
IEC 61000-4-11
Güç frekansı(50/60 Hz) manyetikalanı
IEC 61000-4-8±2 kV güç besleme hatları
için

DC ile güç verilen cihaz

hattan hatta ±1 kV
hattan toprağa ±2 kVGeçerli değil
DC ile güç verilen cihazGeçerli değil

<%5 UT (UT’de >%95düşüş), 0,5 döngü için
%40 UT (UT’de %60 düşüş),5 döngü için
Geçerli Değil
%70 UT (UT’de %30 düşüş), DC ile güç verilen cihaz
25 döngü için
< %5 UT (UT’de>%95 düşüş), 5 saniye için

Geçerli değil

3A/m
50/60 HzGüç frekans manyetik alanları, tipik ticari
ortama veya hastane ortamına uygun olmalıdır.

Geçerli değil
±1 kV (giriş/çıkış hatları için) giriş/çıkış hatları içinveri yok

3A/m
50/60 Hz

Geçerli değil

Not: UT, test düzeyinin uygulanmasından önceki A.C. şebeke gerilimini ifade eder.

** Bu beyanda, gerekli test işleminin kontrollü bir ortamda yapıldığı ve Sprint/Stroller/Hi Flow Stroller cihazınınyönetmeliklere uygun olduğu ifade edilmektedir.

182 - TR PN MN236-C4 E | Kullanıcı El Kitabı

Sprint / Stroller / Hi Flow Stroller
Tablo 4

Yönergeler ve Üreticinin Beyanı—Bağışıklık ME Ekipmanları ve ME Sistemleri

Yönergeler ve Üreticinin Beyanı—Bağışıklık
Sprint/Stroller/Hi Flow Stroller aşağıda belirtilen elektromanyetik ortamda kullanıma yöneliktir. Sprint/
Stroller/Hi Flow Stroller müşterisi veya kullanıcısı, ünitenin bu tür bir ortamda kullanılmasını sağlamalıdır.
Bağışıklık Testi

IEC 60601 Test Düzeyi

Uyumluluk Düzeyi

Elektromanyetik Ortam - Yönergelerİletilen RF IEC
61000-4-6

3 Vrms6 Vrms (ISM Bantlarında)150 kHz - 80 MHz

Geçerli değil

Taşınabilir ve mobil RF iletişim ekipmanı, kablolar
da dahil olmak üzere Sprint/Stroller/Hi Flow Strollerünitesinin herhangi bir parçasına, vericinin frekansı
için geçerli olan denklem kullanılarak hesaplananönerilen ayrım mesafesinden yakın olmayacakşekilde kullanılmalıdır.

Pille çalışan cihaz, SIP/
SOP yokÖnerilen ayrım mesafesi
d = 1,2 √PIşınan RF IEC
61000-4-3

80 MHz - 2,7 GHz

10 V/m80 MHz—2,7 GHz
1 kHz’de %80 AM

d = 1,2 √P
d = 2,3 √P
Burada P, verici üreticisine göre vat (W) cinsinden
vericinin maksimum çıkış gücü derecesidir ve d,metre (m) cinsinden önerilen ayrım mesafesidir.
Bir elektromanyetik saha araştırmasındaa belirlendiği
üzere, Sabit RF vericilerinden gelen alan kuvvetleri,
her bir frekans aralığında uyumluluk düzeyindendüşük olmalıdırb.
Şu sembolle işaretlenmiş ekipmanın yakınında
parazit oluşabilir:

Test frekansı(MHz)

Banta)(MHz)

Servisa)Modülasyonb)

Maksimumgüç (W)

Mesafe(m)Bağışıklık Testi
Düzeyi (V/m)

385

380-390TETRA 400

Puls Modülasyonub)
18 Hz
FMC)
±5 kHz sapma
1 kHz sine

1,8

0,3

27

450

430-470

GMRS 460, FRS 460

2

0,3

28

710
745
780
810
870
930
1.720
1.845
1.970

704-787

LTE Band 13, 17

Puls Modülasyonub)
217 Hz

0,2

0,3

9

800-960

GSM 800/900, TETRA 800,
IDEN 820, CDMA 850, LTE
Band 5

Puls Modülasyonub)
18 Hz

2

0,3

28

1.7001.900

GSM 1800; CDMA 1900;
GSM 1900; DECT; LTEBand 1, 3, 4, 25; UMTS

Puls Modülasyonub)
217 Hz

2

0,3

28

2.450

2.4002.570

Puls Modülasyonub)
217 Hz

2

0,3

28

5.240
5.500
5.785

Bluetooth, WLAN, 802.11
b/g/n, RFID 2450, LTE
Band 75.1005.800

WLAN 802.11 a/n

Puls Modülasyonub)
217 Hz

0,2

0,3

9

Not: BAĞIŞIKLIK TEST DÜZEYİ’nin elde edilmesi için gerekliyse verici anten ile Sprint/Stroller/Hi Flow Stroller arasındaki
mesafe 1 m’ye indirilebilir. IEC 61000-4-3 tarafından 1 m test mesafesine izin verilir.
Bazı hizmetler için yalnızca yer-uydu bağı frekansları eklenmiştir.
Taşıyıcı, %50 görev döngüsü kare dalga sinyali kullanılarak modüle edilmelidir.
Gerçek modülasyonu temsil etmemesine karşın en kötü senaryo olacağından, FM modülasyonuna alternatif olarak 18 Hz’de %50 puls
modülasyonu kullanılabilir.
abc

CAIRE ve CAIRE Inc., CAIRE Inc.’in tescilli ticari markalarıdır. Ticari markaların tam listesi i.in lütfen web sitemizi ziyaret edin. Ticari markalar: www.careinc.com/corporate/trademarks.
Telif hakkı © 2023 CAIRE Inc. CAIRE Inc. herhangi bir zamanda, önceden haber vermeksizin veya ek bir yükümlülük ya da sonuç olmaksızın ürünlerinin üretimini durdurma veya ürünlerinin
fiyatları, materyalleri, ekipmanı, kaliteyi, tanımları, teknik özellikleri ve/veya süreçlerini değiştirme hakkını saklı tutar. Bu belgede açıkça ifade edilmeyen tüm haklar, geçerli olduğu üzere
tarafımızca saklıdır.

PN MN236-C4 E | Kullanıcı El Kitabı TR - 183

Sprint / Stroller / Hi Flow Stroller
Glosář symbolů
ISO 7000

V blízkosti jednotky ani při manipulacis ní nekuřte. Reg. č. P002Chraňte před deštěm, uchovávejte
v suchu. Reg. č. 0626

Příložná část typu BF (stupeň ochrany
proti úrazu elektrickým proudem).
Reg. č. 5333

Limit stohování podle počtu. Reg. #2403
Název a adresa výrobce. Reg. č. 3082

Varování. Reg. č. W001
Směrnice Rady 93/42/EHS
Zplnomocněný zástupce v Evropském
společenstvíZemě a datum výroby. „CC“ označuje
dvoupísmenný kód země výroby. Datumvýroby je ve formátu RRRR-MM-DD.
Reg. # 6049
Upozornění: Prostudujte si průvodní
dokumentaci. Reg. č. 0434A

Pokud je na štítku jedinečného
identifikátoru prostředku (UDI)
produktu uveden symbol CE####, zařízení splňuje požadavky
směrnice 93/42 / EHS týkající sezdravotnických prostředků. Symbol
CE #### označuje číslo oznámeného
subjektu.

Katalogové číslo. Reg. č. 2493
Sériové číslo. Reg. č. 2498

Toto zařízení splňuje požadavky
směrnice 2010/35/EU o zdravotnických
prostředcích. Nese zobrazené
označení pí.

Rozsah skladovacích nebo provozních
teplot. Reg. č. 0632
Rozsah vlhkosti při skladování. Reg.
č. 2620ADR
Netoxický plyn.

Omezení atmosférického tlaku. Reg.
č. 2621

Nebezpečí Oxidující látky: nebezpečí
zesilující požár. Oxidační činidlazpůsobují, že požáry hoří intenzivněji.

Touto stranou nahoru. Reg. č. 0623
Křehké, manipulujte opatrně.
Reg. č. 0621

Hluboce zchlazená kapalina, USP;
vyrobeno zkapalněním vzduchu
Další symboly

Obsahuje nebezpečné látky. Reg. # 3723

Vždy zajistěte dobré odvětrání jednotky.

Dovozce. Reg. # 3725
ISO 7010Chraňte před hořlavými materiály, olejia mazivy.

Při styku se studeným kapalným či
plynným kyslíkem nebo namrzlými
částmi může dojít k omrzlinám.
Varování při nízké teplotě. Upozornění
na mrazivé podmínky nebo nízkou
teplotu. Reg. č. W010
Je nezbytné přečíst příručku k obsluze.
Reg. č. M002Chraňte před otevřeným plamenem,ohněm, jiskrami. Zákaz kouření
a manipulace s plamenem. Reg. č. P003

184 - CZ PN MN236-C4 E | Uživatelská příručka

Autorizovaný zástupce ve Švýcarsku.

IEC 60417

Pokud je zařízení opatřeno značkou
UKCA, jak je znázorněno s kódem
UKCA#### s číslem notifikované
osoby, toto zařízení vyhovuje předpisům
UKCA.
Nezakrývejte jednotku a nenostepřenosnou jednotku pod oblečením.
Tyto jednotky normálně odvětrávají
kyslík. č. 5641

Sprint / Stroller / Hi Flow Stroller
21 CFR 801.15
Tato zařízení smí být podle federálních
zákonů prodávána pouze lékaři nebo na
jejich příkaz.
IEC 60601-1
Odolné proti kapáníSměrnice Rady 2012/19/EU
OEEZ
Tento symbol má připomínat vlastníkům
zařízení, aby je po skončení životnosti
vrátili do recyklačního zařízení v
souladu se směrnicí o odpadních elektrických a elektronických zařízeních(WEEE).
Naše produkty budou splňovat omezení směrnice o nebezpečných látkách(RoHS). Nebudou obsahovat více než
stopová množství olova nebo jiného obsahu nebezpečného materiálu.

ISO 15223-1
Zdravotnické zařízení. Reg. # 5.7.7
Jedinečný identifikátor zařízení # 5.7.10
Na tento výrobek se může v USA i mezinárodně
vztahovat jeden nebo více patentů. Seznam platných
patentů najdete na níže uvedené webové stránce. Pat.:www.caireinc.com/corporate/patents/.

Specifikace• Provozní režim: Kontinuální průtok
• Typ ochrany před úrazem elektrickým proudem: Zařízení s vnitřním napájením• Stupeň ochrany před úrazem elektrickým proudem: Příložná část typu BF• Klasifikace podle stupně ochrany proti vniknutí vody: IP22 – běžná zařízení
• Zařízení není vhodné pro použití v prostředí hořlavých směsí

Technické údaje výrobku

SprintStroller

Hi Flow Stroller

Kapacita kapalného kyslíku

0,68 kg (1,5 lb)

1,36 kg (3,0 lb)

1,36 kg (3,0 lb)

Kapacita plynného
ekvivalentu
Hmotnost, prázdný

513 l1026 l1025 l

2,04 kg (4,5 lb)

2,27 kg (5,0 lb)

2,49 kg (5,5 lb)

Hmotnost, naplněný

2,72 kg (6,0 lb)

3,63 kg (8,0 lb)

3,86 kg (8,5 lb)

Výška

298 mm (11,75")

343 mm (13,5")343 mm (13,5")Rozměry

123,83 mm (4,875") H x
191 mm (7,5") Š
4,3 hodiny

149,23 mm (5,875") H x
191 mm (7,5") Š
8,5 hodin

149 mm (5,875") H x 191 mm(7,5") Š
8,5 hodin

Typická doba použití připrůtoku 2 l/min
Provozní tlak

137 kPa (20 psi)

137 kPa (20 psi)

137 kPa (20 psi)Běžná rychlost odpařování

0,57 kg/den (1,3 lb/den)

0,57 kg/den (1,3 lb/den)

0,57 kg/den (1,3 lb/den)Standardní rozsah regulaceprůtokuPřesnost průtoku*

Vypnuto, 0,25; 0,5; 0,75; 1;
1,5; 2; 2,5; 3; 4; 5; 6 l/min±10 %Vypnuto, 0,25; 0,5; 0,75; 1;
1,5; 2; 2,5; 3; 4; 5; 6 l/min±10 %Vypnuté, 0,5; 1; 2; 2,5; 3; 4; 6;
8; 10; 12; 15 l/min±10 %* Tato přesnost platí pouze při 21 °C a 14,7 psig a s kalibrovaným přesným hmotnostním průtokoměrem.

PN MN236-C4 E | Uživatelská příručka CZ - 185

Sprint / Stroller / Hi Flow Stroller
Varovné informaceDůležité: Před použitím přístrojů Sprint / Stroller /Hi Flow Stroller si pečlivě přečtěte tuto příručku.
Pouze na předpis.
VAROVÁNÍ: TOTO ZAŘÍZENÍ NENÍ URČENOPRO UDRŽOVÁNÍ PACIENTA PŘI ŽIVOTĚ.
VAROVÁNÍ: MÁTE-LI PODEZŘENÍ, ŽE ZAŘÍZENÍNEPRACUJE SPRÁVNĚ, ZAVOLEJTE SVÉMUPOSKYTOVATELI ZDRAVOTNÍ PÉČE. NEPOKOUŠEJTESE JEDNOTKU SAMI OPRAVOVAT ANI SEŘIZOVAT.
VAROVÁNÍ: NEUPRAVUJTE TOTO ZAŘÍZENÍ BEZ
SCHVÁLENÍ VÝROBCEM.
VAROVÁNÍ: JE-LI NUTNÝ NEPŘETRŽITÝ PŘÍVOD
KYSLÍKU, ZAJISTĚTE, ABY BYLO BĚHEM LÉČBY
K DISPOZICI DOSTATEČNÉ MNOŽSTVÍ KYSLÍKU A/
NEBO SEKUNDÁRNÍ PŘÍVOD KYSLÍKU.
VAROVÁNÍ: VYVARUJTE SE KOUŘENÍ A POUŽITÍ
SVÍČEK ČI OTEVŘENÉHO OHNĚ VE VZDÁLENOSTI
DO 3 METRŮ (10 STOP) OD PŘÍSTROJE A ZAJISTĚTE
VZDÁLENOST NEJMÉNĚ 20 CM (8 PALCŮ) OD ZDROJEVZNÍCENÍ.
VAROVÁNÍ: UMÍSTĚTE JEDNOTKU DO DOBŘE
VĚTRANÝCH PROSTOR.
VAROVÁNÍ: ZAŘÍZENÍ OBSAHUJÍCÍ KAPALNÝ
KYSLÍK NESKLADUJTE VE SKŘÍNI, KUFRUAUTA ANI JINÉM UZAVŘENÉM PROSTORU.
NEPOKLÁDEJTE NA ZAŘÍZENÍ PŘIKRÝVKY,ZÁVĚSY ANI JINÉ TEXTILIE.
VAROVÁNÍ: TENTO VÝROBEK VÁS MŮŽE VYSTAVITCHEMICKÝM LÁTKÁM, VČETNĚ NIKLU, KTERÝ JE
VE STÁTĚ KALIFORNIE ZNÁM JAKO KARCINOGEN.
PODROBNĚJŠÍ INFORMACE NAJDETE NA WWW.
P65WARNINGS.CA.GOV.
VAROVÁNÍ: V PŘÍPADĚ VÁŽNÉHO INCIDENTU S TÍMTO
ZAŘÍZENÍM BY MĚL UŽIVATEL OKAMŽITĚ NAHLÁSIT
INCIDENT POSKYTOVATELI A / NEBO VÝROBCI. VÁŽNÝ
INCIDENT JE DEFINOVÁN JAKO ZRANĚNÍ, SMRT NEBO
POTENCIÁL ZPŮSOBIT ZRANĚNÍ / SMRT V PŘÍPADĚ
OPAKOVANÉHO VÝSKYTU INCIDENTU. UŽIVATELMŮŽE TAKÉ NAHLÁSIT INCIDENT PŘÍSLUŠNÉMU
ORGÁNU V ZEMI, KDE K INCIDENTU DOŠLO.
Upozornění: Přístroje Sprint / Stroller / Hi
Flow Stroller používejte pouze podle pokynů
svého lékaře.
Upozornění: Tato zařízení smí být podle federálních
zákonů prodávána pouze lékaři nebo na jejich příkaz.
Upozornění: Jednotka obsahuje kapalný kyslík,který má extrémně nízkou teplotu, téměř –184 °C(–300 °F). Vystavení takto nízké teplotě může způsobitzávažné omrzliny.
Upozornění: Ačkoli je kapalný i plynný kyslíknehořlavý, urychluje hoření jiných materiálů. Toto
riziko spolu s nízkou teplotou kapalného kyslíkuvyžadují jistá bezpečnostní opatření.
186 - CZ PN MN236-C4 E | Uživatelská příručka

Upozornění: Uchovávejte hořlavé materiály
v dostatečné vzdálenosti od tohoto zařízení.
Aerosolové spreje, oleje a maziva, včetně krémů na
obličej či vazelíny, se snadno vznítí a za přítomnosti
kyslíku mohou rychle hořet.
Upozornění: Kouření při nošení kyslíkové kanyly můžezpůsobit popáleniny obličeje a může mít za následek
i smrt.
Jestliže sejmete kanylu a položíte ji na oblečení,povlečení, pohovku nebo jiný výplňový materiál,
po kontaktu s cigaretou, zdrojem tepla, jiskrou čiotevřeným plamenem dojde ke vzplanutí ohně.
Pokud kouříte, je nutné dbát těchto pokynů: (1)vypněte přenosnou jednotku, (2) odpojte kanylu a (3)opusťte místnost, v níž se přístroj nachází.
Upozornění: Dojde-li k neúmyslnému převržení
jednotky, pokud je to možné, ihned ji opatrně znovu
postavte do svislé polohy. Jestliže uniká kapalný
kyslík, ihned opusťte dané prostory a zavolejte svému
poskytovateli zdravotní péče. Nesnažte se jednotkupřemístit ani zastavit únik kapalného kyslíku.
Poznámka: Nedotýkejte se namrzlých částí jakékoli
jednotky.
Poznámka: Přenosnou jednotku neskladujte ani neprovozujte, když je připojena ke stacionární jednotce.
Poznámka: Zajistěte, aby tento přístroj neobsluhoval ani
s ním nemanipuloval neškolený personál.
Poznámka: Používání tohoto přístroje v komerční osobníi nákladní letecké přepravě je zakázáno Federální leteckou
správou (Federal Aviation Administration).

Zamýšlené použití

CAIRE Sprint/Stroller/Hi Flow Stroller je určen kpodávání doplňkového kyslíku. Zařízení není určeno
pro podporu života a neposkytuje žádné funkce pro
sledování pacienta.

Sprint / Stroller / Hi Flow Stroller
Úvod

Ovládací prvkyPřístroje Sprint, Stroller a Hi Flow Stroller jsou
určeny k podávání doplňkového kyslíku pacientovi
v domácnosti koncového uživatele a lze je rovněž
použít v zařízeních, jako jsou pečovatelské ústavy čizařízení subakutní péče. Přístroje Sprint / Stroller /Hi Flow Stroller jsou určeny jako ambulantní zdroj
kyslíku na delší časové období.

1. Elektronické měřidlo hladiny kapaliny
2. Elektronický hladinový spínač
3. Mechanické měřidlo hladiny kapaliny
4. Knoflík regulace průtoku
5. Konektor kyslíkové hadičky
6. Odvzdušňovací ventil
7. Plnicí připojení (identifikaci rychloodpojovacích
ventilů najdete na další straně)8. Popruh pro přenášení s ramenním polštářkem
9. Podložka na kondenzát
10. Nádobka na kondenzátPřístroj používají pacienti s CHOPN nebo sníženou
schopností dýchání. Přístroj se předepisuje pacientovi.
Přístroj se prodává poskytovateli, který je vyškolen
k provozování přístrojů Sprint / Stroller / Hi Flow
Stroller a provádění jejich servisu. Poskytovatel
vyškolí uživatele.
Přístroj není určen k podpoře života ani neposkytuježádné možnosti sledování pacienta. Doporučujeme,
aby byl pro případ mechanické poruchy k dispozici
náhradní zdroj doplňkového kyslíku. Tyto přenosné
jednotky jsou rovněž opatřeny rychloodpojovacím
ventilem, který lze připojit k zásobníku kapalného
kyslíku, a přenosnou jednotku tak doplnit. Zásobník
plní váš poskytovatel zdravotní péče.

1

Tato uživatelská příručka obsahuje pokyny pro použití
přístrojů Sprint / Stroller / Hi Flow Stroller. Informacetýkající se provozu stacionární jednotky najdetev uživatelské příručce dodané s jednotkou.

8

2
4
5

Poznámka: Poskytovatel služeb vám pomůže s počátečníinstalací a poskytne vám pokyny ohledně správné manipulace s jednotkou a jejího správného používání.

5

7

Jednotky Sprint, Stroller a Hi Flow Stroller

610
9
PN MN236-C4 E | Uživatelská příručka CZ - 187

Sprint / Stroller / Hi Flow Stroller
Identifikace rychloodpojovacího
ventilu
Konektorpřenosnéjednotky

Konektorstacionární
jednotky

3. Shora přiložte plnicí konektor ve vzpřímené poloze
ke konektoru stacionární jednotky.

Typ rychloodpojovacího ventilu
Bočníplnění
CAIRE

LIFE-OX

PENOX

TAEMA

PB

Boční
plnění sezablokováním

Horníplnění sezablokováním

Horníplnění
s nástrčným
provedením

4. a. Pokud se jedná o nástrčný konektor:
Zasunujte konektor přístroje Sprint / Stroller /Hi Flow Stroller do stacionární jednotky, dokud
neucítíte, že konektor zacvakl.

b. Pokud jde o konektor Penox nebo Life-Ox:
Otáčejte přístrojem Sprint / Stroller / Hi Flow
Stroller, dokud neucítíte, že konektor zacvakl.
Pak opatrně, ale pevně otáčejte přístrojem Sprint /Stroller / Hi Flow Stroller ve směru hodinových
) a přitom tlačte směrem dolů.
ručiček (Provozní pokynyPlnění přístrojů Sprint / Stroller / Hi Flow
Stroller s horním plněním
1. Očistěte plnicí konektory na stacionární jednotce
i na přístroji Stroller / Spirit / Hi Flow Strollerčistým a suchým hadříkem, který nepouští vlákna.
Upozornění: Plnicí konektory musí být suché,
protože vlhkost může způsobit vzájemnépřimrznutí přístroje Sprint / Stroller / Hi Flow
Stroller a stacionární jednotky.
2. Ujistěte se, že knoflíky regulující průtok na přístroji
Sprint / Stroller / Hi Flow Stroller i na zásobníku
jsou v poloze vypnuto (0).

188 - CZ PN MN236-C4 E | Uživatelská příručka

Poznámka: Pokračujte krokem 6 v části Plnění přístrojů
Sprint / Stroller / Hi Flow Stroller s bočním plněním.

Plnění přístrojů Sprint / Stroller / Hi Flow
Stroller s bočním plněním
1. Očistěte plnicí konektory na stacionární jednotce
i na přístroji Sprint / Stroller / Hi Flow Strollerčistým a suchým hadříkem, který nepouští vlákna.
VAROVÁNÍ: PLNICÍ KONEKTORY MUSÍ BÝT
SUCHÉ, PROTOŽE VLHKOST MŮŽE ZPŮSOBIT VZÁJEMNÉ PŘIMRZNUTÍ PŘÍSTROJESPRINT / STROLLER / HI FLOW STROLLERA STACIONÁRNÍ JEDNOTKY.

Sprint / Stroller / Hi Flow Stroller
2. Ujistěte se, že knoflíky regulující průtok na přístroji
Sprint / Stroller / Hi Flow Stroller i na zásobníku
jsou v poloze vypnuto (0).

Poznámka: Odečty měřidla hladiny kapaliny jsounejpřesnější 30 nebo více sekund po naplnění přístroje
Sprint / Stroller / Hi Flow Stroller.

3. Přiložte plnicí konektor přístroje Sprint / Stroller /Hi Flow Stroller ve vzpřímené poloze ke konektoru
stacionární jednotky.

4. Otáčejte přístrojem Sprint / Stroller / Hi Flow),
Stroller proti směru hodinových ručiček (dokud neucítíte, že kolík zapadl do výřezu.

7. Kolem přípojky se také může tvořit pára. Jedná
se o normální jev způsobený velkým teplotním
rozdílem mezi kapalným kyslíkem a teplým
vzduchem obklopujícím jednotky.
8. Po naplnění jednotky zavřete odvzdušňovací ventil.
To, že je jednotka naplněná, poznáte podle toho,
že syčení náhle začne znít jinak a z odvzdušňovací
hadičky začne vycházet kapalina.
Poznámka: Nejúplnější naplnění přístroje Sprint / Stroller/ Hi Flow Stroller dosáhnete tehdy, když ventil zavřete, ažkdyž z jednotky vychází téměř nepřetržitý proud kapaliny.
Takovýto postup však vede k vyplýtvání určitého množství
kapalného kyslíku.

5. Opatrně, ale pevně otáčejte přístrojem Sprint / Stroller /Hi Flow Stroller tak, aby se opět ocitl ve vzpřímené
poloze. Nyní jsou jednotky vzájemně spojeny.

Poznámka: Zavřete-li ventil při první známce toho, žez jednotky vychází kapalina, nebude naplnění přístroje
Sprint / Stroller / Hi Flow Stroller úplné, ale vyhnete seplýtvání kapalným kyslíkem.
9. V případě nástrčných konektorů tiskněte tlačítko pro
uvolnění, dokud se jednotky neoddělí. V případě
všech ostatních konektorů otáčejte jednotkou,
dokud se neoddělí od stacionární jednotky.

Poznámka: Pokračuje krokem 6.
Poznámka: V případě konektoru s nástrčným provedenímpřidržujte během plnění jednotku na místě.
6. Otevřete odvzdušňovací ventil a začněte plnitpřístroj Sprint / Stroller / Hi Flow Stroller. Mělibyste slyšet zvuk unikajícího kyslíku, tím se ovšem
neznepokojujte. Syčení, které slyšíte, je zcela
normální.

Tlačítko pro uvolnění na přenosné jednotce(pouze nástrčné provedení)

PN MN236-C4 E | Uživatelská příručka CZ - 189

Sprint / Stroller / Hi Flow Stroller
Upozornění: Měřidlo hladiny kapalinykontrolujte až po uzavření ventilu.

Provozní pokyny

Upozornění: Pokud přístroj Sprint / Stroller / Hi
Flow Stroller doplňujete bezprostředně po určitédobě používání, doplňte ho a pak počkejte 10 minuts regulačním ventilem průtoku nastaveným na nulu, aby
se stabilizoval tlak. Poté můžete začít jednotku používat
běžným způsobem.
VAROVÁNÍ: NEPOUŽÍVEJTE PŘÍSTROJSPRINT / STROLLER / HI FLOW STROLLER,
KDYŽ JE PŘIPOJEN KE STACIONÁRNÍJEDNOTCE.
VAROVÁNÍ: JESTLIŽE PŘÍSTROJ SPRINT / STROLLER/ HI FLOW STROLLER NELZE SNADNO ODPOJIT,NEPOUŽÍVEJTE SÍLU. PRAVDĚPODOBNĚ DOŠLO
K VZÁJEMNÉMU PŘIMRZNUTÍ JEDNOTEK. PONECHTEJEDNOTKY SPOJENÉ A VYČKEJTE, DOKUD SE
NEOHŘEJÍ – POTÉ JE LZE SNADNO ROZPOJIT.
NEDOTÝKEJTE SE ŽÁDNÝCH NAMRZLÝCH ČÁSTÍ.
VAROVÁNÍ: DOJDE-LI K NATOLIK ZÁVAŽNÉMU ÚNIKU,KDY PROUDÍ KAPALINA, IHNED OPUSŤTE DANÉPROSTORY A ZAVOLEJTE SVÉMU POSKYTOVATELIZDRAVOTNÍ PÉČE.
VAROVÁNÍ: KAPALNÝ NEBO STUDENÝ PLYNNÝ
KYSLÍK MŮŽE ZPŮSOBIT OMRZLINY KŮŽE.
PŘÍTOMNOST KYSLÍKU ZVYŠUJE RIZIKO POŽÁRU.
Upozornění: Jestliže po rozpojení jednotek
uniká ze stacionární jednotky nebo přístroje
Sprint / Stroller / Hi Flow Stroller kapalina,
postavte přístroj Sprint / Stroller / Hi Flow
Stroller stranou a zajistěte, aby zůstal ve
svislé poloze, poté ihned opusťte místnosta zavolejte svému poskytovateli zdravotní
péče.
Poznámka: Poskytovateli zařízení: K použití s Sprint/ Stroller / Hi Flow Stroller se doporučuje následujícípříslušenství pro podávání kyslíku:• Nasální kanyla s hadicí o délce 2,1 m (maximální LPM: 6
LPM): CAIRE Číslo dílu 5408-SEQ• Firebreak: CAIRE Číslo dílu 21126636
Pro použití s jakoukoli kanylou se doporučuje požární
štěrk.
• CAIRE nabízí ohnivzdorný materiál určený k použití
ve spojení s přenosným kyslíkem. Firebreak je tepelná
pojistka, která zastaví tok plynu v případě, že se zapálí
kanyla nebo kyslíková trubka po proudu a spálí se do firebreak. Je umístěna in-line s nosní kanylou nebo kyslíkovou
hadičkou mezi pacientem a výstupem kyslíku z Sprint /Stroller / Hi Flow Stroller. Pro správné použití firebreaku sevždy řiďte pokyny výrobce (součástí každé sady firebreak).
• Další doporučené informace o příslušenství jsou kdispozici online na www.caireinc.com.

190 - CZ PN MN236-C4 E | Uživatelská příručka(Kanyla není zahrnuta.)

1. Pevně nasaďte dýchací hadičku kanyly na konektor
kyslíkové hadičky.
2. Nastavte dýchací hadičku kanyly do vhodné polohy,
abyste mohli pohodlně vdechovat kyslík.
3. Otáčejte knoflíkem regulace průtoku na horní straně
přístroje Sprint / Stroller / Hi Flow Stroller ve směru
hodinových ručiček, dokud se v „okénku“ knoflíku
nezobrazí předepsaný průtok (číselný údaj) a dokud
neucítíte aretaci.
Upozornění: Je-li knoflík regulace průtoku
nastaven mezi průtoky, dojde k průtoku
kyslíku, který nespadá do specifikací.
4. Nyní byste měli být zásobováni kyslíkem. Chcete-li
zastavit proudění kyslíku, otočte knoflíkem regulaceprůtoku proti směru hodinových ručiček do polohy
vypnuto (0).
Upozornění: Když jednotku nepoužíváte, vždyotočte knoflík pro volbu průtoku do polohy
vypnuto (0).

Sprint / Stroller / Hi Flow Stroller
5. Chcete-li zjistit, jak dlouho bude přístroj Sprint /Stroller / Hi Flow Stroller pracovat, použijte jako
vodítko následující tabulku:
Nastavení regulačního ventilu průtoku (FCV)
Model

SprintStroller

Hi Flow
StrollerJmenovité

Jmenovité

PolohaFCV

SprintStroller

Hi Flow
Stroller

Vypnuto

30:00

60:00

71:00

0,25

18:00

30:00

Není
k dispozici

0,5

12:12

23:00

32:00

0,75

9:18

17:24

Není
k dispozici

1

7:30

14:06

16:00

1,5

5:30

10:12

Není
k dispozici

2

4:18

8:00

8:00

2,5

3:36

6:36

6:20

3

3:06

5:36

5:20

4

2:24

4:24

4:00

5

2:00

3:36

Není
k dispozici• Měřidlo Gen 4: Stiskněte zelené ovládací tlačítko načelní ploše měřidla. Hodnotu odečtěte na oblouku
LED kontrolek, který ukazuje hladinu obsahu.

Upozornění: Přístroj Sprint / Stroller / Hi Flow
Stroller je prázdný, jestliže svítí pouze poslednísegment světelné lišty.
Pokud se při stisknutí tlačítka rozsvítí ukazatel
vybité baterie, informujte svého poskytovatele
zdravotní péče.

6

1:42

3:00

2:40

8

Není
k dispozici

Není
k dispozici

2:00

10

Není
k dispozici

Není
k dispozici

1:40

Poznámka: I když je ukazatel vybité baterie rozsvícený,přístroj Sprint / Stroller / Hi Flow Stroller bude nadálepřivádět kyslík, a to tak dlouho, dokud bude v jednotce
kapalný kyslík.

12

Není
k dispozici

Není
k dispozici

1:20

6b. Ověření hladiny kapalného kyslíku v jednotce
pomocí mechanického měřidla hladiny kapaliny:

15

Není
k dispozici

Není
k dispozici

1:00

Poznámka: Doby jsou uvedeny v hodinách a minutách(formát 00:00).
Poznámka: „Jmenovité“ doby platí za ideálních
podmínek, tj. maximální naplnění, přesné průtoky,příznivá míra ztrát, jednotka se nepřemísťuje atd.

• Uchopte jednotku za textilní popruh přímo nadčerným pouzdrem pružinové váhy s ukazatelemobsahu.
• Počkejte, až se jednotka stabilizuje (tj. nebude se
houpat nahoru a dolů), a pak na barevné stupnici
odečtěte přibližný obsah kapalného kyslíku.

Jde o maximální předpokládané doby.
Poznámka: Vaše individuální výsledky se budou různit.
6a. Ověření hladiny kapalného kyslíku v jednotce
pomocí elektronického měřidla hladiny kapaliny:• Měřidlo Gen 3: Nejméně dvě sekundy podržte
stisknuté tlačítko (hladinový spínač) na horní straně
jednotky. Odečtením v horní části světelné lišty
zjistěte hladinu obsahu.

PN MN236-C4 E | Uživatelská příručka CZ - 191

Sprint / Stroller / Hi Flow Stroller
Poznámka: Je-li jednotka prázdná, na stupnici bude vidět
pouze červený proužek. V takovém případě je třeba jednotkupřed použitím naplnit.
7. Za určitých podmínek prostředí při kontinuálním
používání a zejména když se přenosná jednotka
nepohybuje, může se kolem vaporizační cívky a na
pouzdře vytvořit nadměrná námraza. Tuto námrazumůžete odstranit tak, že na jednotku zaťukáte nebo
nashromážděnou námrazu z pouzdra setřete.

Údržba podložky na kondenzát
Jakmile bude přístroj prázdný a ohřeje se na pokojovou
teplotu, odstraňte z nádobky na kondenzát veškerou
vlhkost:V případě tvrdého pouzdra• Nakloňte přístroj k jedné straně. Uchopte knoflík
uprostřed nádobky na kondenzát (v dolní části
přístroje) a vytáhněte ho přímým pohybem. Tím
se nádobka uvolní.
Poznámka: Pokud je podložka silně znečištěná, je možné
ji vyprat v pračce (s bělicím prostředkem pro dezinfikování).
• Chcete-li nádobku vrátit na místo, umístěte ji nadstředem okrouhlého otvoru v dolní části přístroje
a stiskněte knoflík, jedním cvaknutím nádobka
zapadne a druhým cvaknutím se nádobka
zablokuje na svém místě.

V případě měkkého pouzdra• Rozepněte dolní zip a stáhněte dolní kryt.
• Vyjměte podložku z nádobky a vyždímejte
veškerou absorbovanou vlhkost. Než podložku
vrátíte do nádobky, nechte ji zcela vyschnout.
Poznámka: Pokud je podložka silně znečištěná, jemožné ji vyprat v pračce (s bělicím prostředkem pro
dezinfikování).
• Chcete-li ji vrátit na místo, jednoduše podložku
vložte do dolního krytu a znovu zapněte dolní zip.
Poznámka: Jestliže se knoflík nezasune, a nádobka tedy
nezapadne, držte nádobku a vytahujte knoflík, dokuds cvaknutím nepřejde do polohy otevření. Nyní vraťte
nádobku na místo, jak bylo popsáno dříve.

Omezení námrazyZa určitých podmínek prostředí při kontinuálním
používání a zejména když se přenosná jednotka
nepohybuje, může se kolem vaporizační cívky a na
pouzdře vytvořit nadměrná námraza. Tuto námrazumůžete odstranit tak, že na jednotku zaťukáte nebo
nashromážděnou námrazu z pouzdra setřete.
VAROVÁNÍ: VZHLEDEM K MOŽNOSTINADMĚRNÉ VLHKOSTI / TVORBY LEDUDOPORUČUJEME, ABY PŘÍSTROJ HI FLOWSTROLLER ZŮSTAL NEČINNÝ PO DOBU
1 HODINY PO POUŽÍVÁNÍ, ABY SE SNÍŽILO
HROMADĚNÍ VLHKOSTI.

Zablokováno
Na místě
Otevřeno

Nádobka s knoflíkem

Poznámka: Jestliže se knoflík nezasune, a nádobka tedy
nezapadne, držte nádobku a vytahujte knoflík, dokud s cvaknutím nepřejde do polohy otevření. Nyní vraťte nádobku na
místo, jak bylo popsáno dříve.

192 - CZ PN MN236-C4 E | Uživatelská příručka

V důsledku vyšších průtokových rychlostí spojenýchs přístrojem Hi Flow Stroller bude na jednotce
docházet ke kondenzaci a určité námraze, zvláště
při nastaveních 10 l/min a vyšších. Doporučujeme
po používání ponechat jednotku po dobu minimálně
1 hodiny v nečinnosti, aby se snížila možnost jejichnadměrné tvorby v důsledku kontinuálního používání.

Sprint / Stroller / Hi Flow Stroller
Odstraňování potíží
Následující informace jsou určeny k tomu, aby vám pomohly při odstraňování potíží a řešení jednoduchých
provozních problémů, se kterými se můžete setkat při používání přístroje Sprint / Stroller / Hi Flow Stroller.
PotížePřenosná jednotka vydává
syčivý zvuk.

Řešení
• Syčení se může ozývat z důvodu udržování správného provozního tlaku v přenosné
jednotce. K syčení nejpravděpodobněji dochází po naplnění nebo po změně polohypřenosné jednotky. Po naplnění může být syčení slyšet zhruba po dobu 10 minut. Kromě
toho může být slyšet tehdy, když je ventil regulující průtok nastaven na nízký průtok.

• Jestliže se přenosná jednotka nachází v nevhodné poloze, vraťte ji do vzpřímené polohya několik minut počkejte, aby se stabilizovala.
Průtok přenosné jednotky se • Ujistěte se, zda je kanyla pevně připevněna k výstupu kyslíku.
během používání zastaví.
• Ujistěte se, zda není kanyla přehnutá.
• Zkontrolujte ukazatel obsahu / měřidlo hladiny a podle potřeby přenosnou jednotku naplňte.
Přenosná jednotka se
neplní.
Odvzdušňovací ventilpřenosné jednotky se na
konci plnicího procesu
správně nezavírá.

Přenosnou jednotku nelze
po naplnění snadno odpojit
od zásobníku.

• Ujistěte se, že knoflík pro regulaci tlaku není v poloze vypnuto („0“).
• Ověřte, zda je v zásobníku kyslík.
• Ujistěte se, zda jsou plnicí konektory přenosné jednotky a zásobníku po celou dobu
plnicího procesu zcela připojeny.
• Pokud se odvzdušňovací ventil nezavře a přetrvává syčení a závoj kyslíkových par,opatrně odeberte přenosnou jednotku stisknutím uvolňovacího tlačítka na zásobníku.
Odvzdušnění ve spodní části přenosné jednotky během několika minut skončí. Nechtejednotku zahřát, dokud nebude možné zavřít odvzdušňovací ventil. Může trvat až
60 minut, než se v přenosné jednotce obnoví dostatečný tlak pro přesný průtok kyslíku.
V případě potřeby použijte náhradní zdroj kyslíku, například regulační ventil průtoku
připojený k zásobníku.
• Plnicí konektory přenosné jednotky a zásobníku pravděpodobně zamrzly.
• NEPOUŽÍVEJTE SÍLU. Ponechte zamrzlé části několik minut zahřát, a jakmile ledroztaje, odpojte přenosnou jednotku. Abyste předešli vzájemnému přimrznutí jednotek,otřete vždy před plněním zástrčku plnicího konektoru na zásobníku a zásuvku plnicího
konektoru na přenosné jednotce čistým, suchým hadříkem.

PN MN236-C4 E | Uživatelská příručka CZ - 193

Sprint / Stroller / Hi Flow StrollerStandardní čištění
VAROVÁNÍ: JEDNOTKU ČISTĚTE, POUZE
JE-LI PRÁZDNÁ.
• Čistěte roztokem slabého mycího prostředku nanádobí a vody.
• Čisticí roztok aplikujte přímo na hadřík,který nepouští vlákna. Mezi schválené čisticí
prostředky patří HydroPure a HydroKlean.
Čisticí prostředky nestříkejte přímo na přístroj
Sprint / Stroller / Hi Flow Stroller.
• Hadříkem, který nepouští vlákna, otřete do čista
vnější povrch.
Upozornění: K čištění těchto jednotek
nepoužívejte zařízení pracující při vysokém
tlaku nebo vysoké teplotě.
• Dbejte na to, aby se čisticí prostředek nedostal do
kontaktu s vnitřními součástmi ani ventily.
• Před použitím nechte jednotku dokonale
oschnout.
Poznámka: Poznámka pro poskytovatele zdravotní péče –postupy před opětovným uvedením do provozu naleznetev příslušné servisní příručce.

ÚdržbaZa veškerou údržbu, která je vyžadována dle technicképříručky k tomuto přístroji, nese odpovědnost
poskytovatel služeb. S veškerými požadavky naúdržbu se obracejte na svého poskytovatele služeb.
Uživatel smí provádět pouze údržbu podložky na
kondenzát. Žádné jiné části údržbu ze strany uživatele
neumožňují.
Předpokládaná životnost je nejméně pět let.
Plnicí konektory na stacionární i přenosné jednotce
mezi každým plněním očistěte suchým a čistým
hadříkem, který nepouští vlákna, abyste předešlizamrznutí a potenciální poruše zařízení.
Poznámka: Veškerou další potřebnou údržbu musí provést
kvalifikovaný servisní technik nebo poskytovatel služeb.

PříslušenstvíSměrnice OEEZ a RoHS

Tento symbol má majitelům zařízení
připomenout, že jej mají podle směrniceo odpadních elektrických a elektronických
zařízeních (OEEZ) na konci životnosti
odevzdat sběrnému recyklačnímu zařízení.
Naše výrobky odpovídají směrnici o omezenípoužívání některých nebezpečných látek v elektrických
a elektronických zařízeních (RoHS). Neobsahují vícenež stopové množství olova ani jiných nebezpečných
materiálů.

LikvidaceZa účelem řádné likvidace vždy vraťte přístroj Sprint /Stroller / Hi Flow Stroller, včetně všech součástí,poskytovateli domácí zdravotní péče. Můžete také
kontaktovat místní městský úřad a požádat o pokynyk řádné likvidaci baterie.

Přeprava a skladováníPřístroj je nutné uložit ve svislé poloze a řádně jejodvětrat. Dbejte na to, aby přístroj neležel na boku.
Vlhkost do 95 %, bez kondenzace. Rozsah teplot
od −40 do 70 °C (−40 až 158 °F).
Provozní teplota se pohybuje od −10 do 40 °C (14 až
104 °F). Relativní vlhkost se pohybuje od 30 do 75 %,bez kondenzace.
Poznámka: Rozsah atmosférického tlaku je 700 až
1060 hPa (výška 10 000 až −1000 stop).

194 - CZ PN MN236-C4 E | Uživatelská příručka

Vozík na převážení PN 20765565

Sprint / Stroller / Hi Flow Stroller
Bezpečnost
VAROVÁNÍ: PŘENOSNÁ VYSOKOFREKVENČNÍ KOMUNIKAČNÍ ZAŘÍZENÍ (VČETNĚ PERIFERIÍ, JAKO
JSOU ANTÉNNÍ KABELY A EXTERNÍ ANTÉNY) BY NEMĚLA BÝT POUŽÍVÁNA BLÍŽE NEŽ 30 CM (12 PALCŮ)OD JAKÉKOLI ČÁSTI PŘÍSTROJE SPRINT / STROLLER / HI FLOW STROLLER, VČETNĚ KABELŮ
SPECIFIKOVANÝCH VÝROBCEM. V OPAČNÉM PŘÍPADĚ MŮŽE DOJÍT K ZHORŠENÍ FUNKCE TOHOTO
ZAŘÍZENÍ.
VAROVÁNÍ: POUŽITÍ JINÉHO PŘÍSLUŠENSTVÍ, PŘEVODNÍKŮ ČI KABELŮ, NEŽ KTERÉ JSOU SPECIFIKOVÁNY NEBOPOSKYTNUTY VÝROBCEM TOHOTO ZAŘÍZENÍ, MŮŽE MÍT ZA NÁSLEDEK ZVÝŠENÍ ELEKTROMAGNETICKÝCH
EMISÍ NEBO SNÍŽENÍ ELEKTROMAGNETICKÉ ODOLNOSTI TOHOTO ZAŘÍZENÍ A ZPŮSOBIT NESPRÁVNOU FUNKCI.
VAROVÁNÍ: JE NUTNÉ DBÁT NA TO, ABY TOTO ZAŘÍZENÍ NEBYLO POUŽÍVÁNO VEDLE JINÉHO ZAŘÍZENÍ NEBO NANĚM ČI POD NÍM, PROTOŽE BY TÍM MOHLO DOJÍT K NESPRÁVNÉ FUNKCI. JE-LI TAKOVÉ POUŽITÍ NEZBYTNÉ, JE
NUTNÉ SLEDOVÁNÍM OVĚŘIT, ZDA TOTO I DRUHÉ ZAŘÍZENÍ FUNGUJÍ NORMÁLNĚ.
Upozornění: Zdravotnické elektrické zařízení vyžaduje speciální preventivní bezpečnostní opatření ohledně
elektromagnetické kompatibility (EMC) a je nutné, aby bylo instalováno a uvedeno do provozu v souladu
s informacemi o EMC uvedenými v této příručce.
Upozornění: Přenosná a mobilní vysokofrekvenční (RF) komunikační zařízení mohou ovlivnit
zdravotnické elektrické zařízení.
Tabulka 1

Pokyny a prohlášení výrobce – elektromagnetické emisePřístroj Sprint / Stroller / Hi Flow Stroller je určen k používání v elektromagnetickém prostředí, které jespecifikováno níže. Zákazník nebo uživatel přístroje Sprint / Stroller / Hi Flow Stroller by měl zajistit, aby bylpoužíván v takovém prostředí.

Zkouška emisí
RF emise

Kompatibilita
Skupina 1

CISPR 11

Elektromagnetické prostředí – pokynyPřístroj Sprint / Stroller / Hi Flow Stroller používá
vysokofrekvenční energii pouze pro vnitřní funkce.
Jeho vysokofrekvenční emise jsou tedy velmi nízké
a je nepravděpodobné, že by mohly způsobovat rušení
blízkých elektronických zařízení.

Vysokofrekvenčníemise, CISPR 11Třída B

Harmonické emise,
IEC 61000-3-2

Nehodí se

Kolísání napětí /generování kmitavého
signálu, IEC 61000-3-3

Nehodí sePřístroj Sprint / Stroller / Hi Flow Stroller je vhodný
k použití ve všech objektech, včetně domácnostía zařízení přímo napojených na veřejnou síť nízkéhonapětí, která napájí budovy používané pro obytné účely.

PN MN236-C4 E | Uživatelská příručka CZ - 195

Sprint / Stroller / Hi Flow Stroller
Tabulka 2*: Doporučené oddělovací vzdálenosti mezi přenosnými a mobilními vysokofrekvenčními komunikačními
zařízeními a přístrojem Sprint / Stroller / Hi Flow StrollerPřístroj Sprint / Stroller / Hi Flow Stroller je určen k používání v elektromagnetickém prostředí, v němž je vyzařované
vysokofrekvenční rušení regulováno. Zákazník nebo uživatel přístroje Sprint / Stroller / Hi Flow Stroller může pomoci předcházet
elektromagnetickému rušení tím, že bude dodržovat níže uvedenou minimální vzdálenost mezi přenosnými a mobilními
vysokofrekvenčními komunikačními zařízeními (vysílači) a přístrojem Sprint / Stroller / Hi Flow Stroller podle maximálního výstupníhovýkonu komunikačního zařízení.
Jmenovitý maximální
Vzdálenost podle frekvence vysílače
výstupní výkon
mvysílače
80 MHz až 800 MHz
800 MHz až 2,5 GHz
150 kHz až 80 MHz

d = 1,2√P

d = 1,2√P

d = 2,3√PW
0,010,12 m0,12 m0,23 m0,10,38 m0,38 m0,73 m
1
1,2 m
1,2 m
2,3 m
10
3,8 m
3,8 m
7,3 m
100
12 m
12 m
23 m
U vysílačů, jejichž jmenovitý maximální výstupní výkon není uveden výše, může být doporučená oddělovací vzdálenost (d) v metrech (m) odhadnuta pomocí rovnice platné pro daný kmitočet vysílače, kde P je jmenovitý maximální výstupní výkon vysílače ve
wattech (W) podle výrobce vysílače.
POZNÁMKA 1: U frekvencí 80 MHz a 800 MHz platí vzdálenost pro vyšší frekvenční rozsah.
POZNÁMKA 2: Tyto pokyny nemusí platit ve všech situacích. Na šíření elektromagnetických vln má vliv míra jejich pohlcováníbudovami, předměty a lidmi i míra jejich odrazu od nich.

* Tato tabulka je zahrnuta jako standardní požadavek na zařízení, které bylo ozkoušeno dle konkrétních
zkušebních úrovní a v konkrétních frekvenčních rozsazích a bylo zjištěno, že splňuje předpisy.

Tabulka 3: Pokyny a prohlášení výrobce – elektromagnetická odolnostPřístroj Sprint / Stroller / Hi Flow Stroller je určen k používání v elektromagnetickém prostředí, které je specifikováno níže. Zákazník
nebo uživatel přístroje Sprint / Stroller / Hi Flow Stroller by měl zajistit, aby byl používán v takovém prostředí.
Zkouška odolnosti
IEC 60601, zkušební úroveň Úroveň kompatibility
Elektromagnetické prostředí – pokyny
Podlahy by měly být dřevěné, betonové nebo
±8 kV, kontaktní výboj
Vybití elektrostatického ±8 kV, kontaktní výboj
z keramické dlažby. Jsou-li podlahy pokryty
náboje (ESD)± 2 kV, ± 4 kV, ±8 kV,± 2 kV, ± 4 kV, ±8 kV,
syntetickým materiálem, relativní vlhkost by
IEC 61000-4-2± 15 kV vzduchový výboj
± 15 kV vzduchový výboj měla dosahovat alespoň 30 %.**
Nehodí se
Rychlé elektricképřechodové jevy /skupiny impulsů
IEC 610004-4
Rázový impulz
IEC 61000-4-5
Poklesy napětí, krátkápřerušení a kolísání
napětí na přívodním
vedení napájecích
zdrojů
IEC 61000-4-11
Magnetické polesíťového kmitočtu(50/60 Hz)
IEC 61000-4-8±2 kV pro elektrické napájecí
vedení

Přístroj napájený
stejnosměrným proudem±1 kV pro vstupní/výstupnívedení

Nehodí se±1 kV mezi vedeními±2 kV vedení–zeměŽádný datový vstup/výstup
Nehodí sePřístroj napájený
stejnosměrným proudem

Nehodí se

Nehodí se<5 % UT (>95 % pokles UT)
po dobu 0,5 cyklu
40 % UT (60 % pokles UT)
po dobu 5 cyklů
70 % UT (30 % pokles UT)
po dobu 25 cyklů
<5 % UT (>95 % pokles UT)
po dobu 5 sekund

Nehodí sePřístroj napájený
stejnosměrným proudem

Nehodí se

3A/m
50/60 Hz

3A/m
50/60 Hz

Magnetické pole síťové frekvence musíodpovídat typickému komerčnímu nebo
nemocničnímu prostředí.

Poznámka: UT je napětí sítě střídavého proudu před uplatněním zkušební úrovně.

** Toto prohlášení uvádí, že požadované zkoušení bylo provedeno v kontrolovaném prostředí a bylo zjištěno, žepřístroj Sprint / Stroller / Hi Flow Stroller splňuje předpisy.
196 - CZ PN MN236-C4 E | Uživatelská příručka

Sprint / Stroller / Hi Flow Stroller
Tabulka 4

Pokyny a prohlášení výrobce – odolnost zdravotnických elektrických
přístrojů a zdravotnických elektrických systémů

Pokyny a prohlášení výrobce – odolnostPřístroj Sprint / Stroller / Hi Flow Stroller je určen k používání v elektromagnetickém prostředí, které jespecifikováno níže. Zákazník nebo uživatel přístroje Sprint / Stroller / Hi Flow Stroller by měl zajistit, aby bylpoužíván v takovém prostředí.
Zkouška odolnosti
IEC 60601, zkušební
Úroveň kompatibility
Elektromagnetické prostředí – pokynyúroveň
RF vedením, IEC 61000- 3 Vrms
4-6
6 Vrms (pásmo ISM)
150 kHz až 80 MHz

Nehodí seZařízení napájené
bateriemi, žádný vstup/výstup signáluPřenosná a mobilní vysokofrekvenční komunikačnízařízení by se neměla používat v menší vzdálenosti
od jakékoli části přístroje Sprint / Stroller / Hi Flow
Stroller, včetně kabelů, než je doporučená oddělovací
vzdálenost vypočítaná z rovnice platné pro frekvenci
vysílače.
Doporučená vzdálenostd = 1,2 √P

RF zářením, IEC 610004-3

80 MHz až 2,7 GHz

10 V/m80 MHz – 2,7 GHz
80 % AM při 1 kHz

d = 1,2 √P
d = 2,3 √P
kde P je maximální jmenovitý výstupní výkon vysílače
ve wattech (W) udávaný výrobcem vysílače a d jedoporučená oddělovací vzdálenost v metrech (m).
Intenzita pole pevných radiofrekvenčních vysílačů
zjištěná elektromagnetickým průzkumem lokalityaby měla být nižší než povolená úroveň v každém
z frekvenčních pásemb.
K rušení může docházet v blízkosti zařízení
označených následujícím symbolem:

Zkušební
frekvence (MHz)Pásmoa)(MHz)Službaa)

Modulaceb)385

380–390TETRA 400450

430–470

GMRS 460, FRS 460

Pulzní modulaceb)
18 Hz
FMC)
Odchylka ±5 kHz
Sinus 1 kHz

Maximální
výkon (W)

Vzdálenost(m)

Zkušebníúroveň
odolnosti (V/m)

1,8

0,3

27

2

0,3

28

710
745
780
810
870
930
1720
1845
1970

704–787

Pásmo LTE 13, 17Pulzní modulaceb)
217 Hz

0,2

0,3

9

800–960

GSM 800/900, TETRA 800,
IDEN 820, CDMA 850,
pásmo LTE 5

Pulzní modulaceb)
18 Hz

2

0,3

28

1700–
1900

GSM 1800; CDMA 1900;
GSM 1900; DECT; pásmo
LTE 1, 3, 4, 25; UMTS

Pulzní modulaceb)
217 Hz

2

0,3

28

2450

2400–
2570

Pulzní modulaceb)
217 Hz

2

0,3

28

5240
5500
5785

Bluetooth, WLAN, 802.11
b/g/n, RFID 2450, pásmo
LTE 7

5100–
5800

WLAN 802.11 a/n

Pulzní modulaceb)
217 Hz

0,2

0,3

9

Poznámka: Pokud je to nutné k dosažení ZKUŠEBNÍ ÚROVNĚ ODOLNOSTI, může se vzdálenost mezi vysílací anténoua přístrojem Sprint / Stroller / Hi Flow Stroller zkrátit na 1 m. Tuto 1 m zkušební vzdálenost dovoluje norma IEC 61000-4-3.
U některých služeb jsou zahrnuty pouze frekvence odchozího připojení.
Nosič musí být modulován s použitím signálu obdélníkové vlny s 50% činitelem využití.
Jako alternativu k FM modulaci lze použít 50% pulzní modulaci při 18 Hz, protože by se jednalo o nejhorší případ, přestože nepředstavuje
skutečnou modulaci.
abc

CAIRE a CAIRE Inc. jsou registrovan. ochrann. zn.mky společnosti CAIRE Inc. .plný seznam ochranných zn.mek naleznete na n.že uveden. webov. str.nce. Ochrann. zn.mky: www.careinc.
com/corporate/trademarks.
Copyright © 2023 CAIRE Inc. Společnost CAIRE Inc. si vyhrazuje právo kdykoli bez předchozího upozornění a bez dalších závazků či důsledků přestat vyrábět své výrobky nebo změnit
ceny, materiály, zařízení, kvalitu, popisy, specifikace svých výrobků nebo procesy související s těmito výrobky. Veškerá práva, která zde nejsou výslovně uvedena, si vyhrazujeme jako
uplatnitelná.

PN MN236-C4 E | Uživatelská příručka CZ - 197

Sprint / Stroller / Hi Flow Stroller
Wykaz symbolów
ISO 7000

Nie palić w pobliżu urządzenia ani
podczas jego obsługi. Reg. # P002

Chronić przed deszczem. Przechowywać
w suchym miejscu.
Reg. # 0626

Część klasy BF (stopień ochrony przedporażeniem elektrycznym). Reg. # 5333

Limit układania według liczb. Reg. #2403
Nazwa i adres producenta. Reg. # 3082

Ostrzeżenie. Reg. # W001
Dyrektywa Rady 93/42/EWG
Autoryzowany przedstawiciel w Europie

Kraj i data produkcji. „CC” oznaczadwuliterowy kod kraju produkcji. Data
produkcji ma format RRRR-MM-DD.
Rozp. # 6049
Przestroga: Zapoznać się z dołączoną
dokumentacją. Reg. # 0434A

Jeżeli na etykiecie niepowtarzalnego
identyfikatora urządzenia (UDI)
znajduje się symbol CE ####,
urządzenie to spełnia wymagania
dyrektywy 93/42 / EWG dotyczącej
wyrobów medycznych. Symbol
CE #### wskazuje numer jednostki
notyfikowanej.

Numer katalogowy. Reg. # 2493

To urządzenie jest zgodne z wymogami
Dyrektywy 2010/35/EWG dotyczącej
wyrobów medycznych. Posiada
zaprezentowane oznaczenie pi.

Numer seryjny. Reg. # 2498
Zakres temperatur przechowywania lub
pracy. Reg. # 0632

ADR

Zakres wilgotności podczas
przechowywania. Reg. # 2620

Limit ciśnienia atmosferycznego.
Reg. # 2621
Góra. Reg. # 0623
Ostrożnie. Reg. # 0621

Gaz nietoksyczny.
Zagrożenie Substancje utleniające:ryzyko intensyfikacji pożaru. Utleniacze
powodują intensywniejsze palenie sięognia.
Zmrożona ciecz, USP; powstała
w wyniku skraplania powietrza
Dodatkowe symbole

Zawiera niebezpieczne substancje.
Rozp. # 3723

ISO 7010

Dbać o ciągłą wentylację urządzenia

Importer. Rozp. # 3725

Przechowywać z dala od materiałówłatwopalnych, olejów i smarów

Kontakt z zimnym płynem, tlenem
w postaci gazowej lub zamarzniętymi
częściami może doprowadzić doodmrożeń. Ostrzeżenie przed niską
temperaturą. Ostrzega przed niską
temperaturą lub zamarzaniem.
Reg. # W010

Autoryzowany przedstawiciel w
Szwajcarii.

Przeczytać instrukcję obsługi.
Reg. # M002
Przechowywać z dala od nieosłoniętego
płomienia, ognia i źródeł iskrzenia.
Chronić przed źródłem zapłonu. Zakaz
palenia. Reg. # P003
198 - PL PN MN236-C4 E | Instrukcja obsługi

IEC 60417

Jeśli urządzenie nosi znak UKCA, jakpokazano z UKCA#### wskazującym
numer jednostki notyfikowanej, tourządzenie jest zgodne z przepisami
UKCA.
Nie przykrywaj urządzenia ani nie nośprzenośnego urządzenia pod ubraniem.
Jednostki te normalnie odprowadzają
tlen. nr 5641

Sprint / Stroller / Hi Flow Stroller
21 CFR 801.15
Prawo federalne dopuszcza sprzedaż
tego urządzenia wyłącznie przez lekarza
lub na jego zlecenie
IEC 60601-1
Odporny na kapanie
Dyrektywa Rady 2012/19/UE
ZSEE
Ten symbol ma przypominać właścicielom sprzętu o zwróceniu go do zakładu
recyklingu po zakończeniu jego eksploatacji, zgodnie z dyrektywą w sprawiezużytego sprzętu elektrycznego i elektronicznego (WEEE).
Nasze produkty będą zgodne z dyrektywą dotyczącą substancji niebezpiecznych (RoHS). Nie będą zawierać więcejniż śladowe ilości ołowiu lub innychmateriałów niebezpiecznych.

ISO 15223-1
Urządzenie medyczne. Rozp. # 5.7.7
Unikalny identyfikator urządzenia #5.7.10Ten produkt może być chroniony jednym lub wieloma
patentami w Stanach Zjednoczonych i innych krajach.
Odpowiednie patenty wyszczególnione są w naszej
witrynie pod adresem podanym poniżej. Pat.: www.
caireinc.com/corporate/patents/.

Dane techniczne• Tryb działania: ciągły przepływ• Typ zabezpieczenia przed porażeniem elektrycznym: zasilanie wewnętrzne• Stopień ochrony przed porażeniem elektrycznym: część klasy BF• Klasyfikacja na podstawie stopnia ochrony przed przedostawaniem się wody: IP22 – brak ochrony• Nie stosować w obecności łatwopalnych środków
Dane techniczne produktu

SprintStroller

Hi Flow Stroller

Pojemność zbiornika na
ciekły tlen (LOX)
Pojemność równoważna dla
substancji gazowej
Masa, zbiornik pusty

0,68 kg (1,5 lb)

1,36 kg (3,0 lb)

1,36 kg (3,0 lb)

513 l

1026 l1025 l

2,04 kg (4,5 lb)

2,27 kg (5,0 lb)

2,49 kg (5,5 lb)

Masa, zbiornik napełniony

2,72 kg (6,0 lb)

3,63 kg (8,0 lb)

3,86 kg (8,5 lb)

Wysokość

298 mm (11,75")

343 mm (13,5")343 mm (13,5")

Wymiary

123,83 mm (4,875") G x
191 mm (7,5") S
4,3 godziny

149,23 mm (5,875") G x
191 mm (7,5") S
8,5 godzin

149 mm (5,875") G x
191 mm (7,5") S
8,5 godzin

137 kPa (20 psi)

137 kPa (20 psi)

0,57 kg/dzień (1,3 lb/dzień)0,57 kg/dzień (1,3 lb/dzień)Standardowy czasużytkowania przy przepływie
2 L/MINCiśnienie robocze
137 kPa (20 psi)
Normalna szybkość
parowania
Standardowy zakres
ustawień przepływu
Dokładność ustawień
przepływu*0,57 kg/dzień (1,3 lb/dzień)Wył., 0,25; 0,5; 0,75; 1; 1,5; 2; Wył., 0,25; 0,5; 0,75; 1; 1,5; 2; Wył., 0,5; 1; 2; 2,5; 3; 4; 6; 8;
2,5; 3; 4; 5; 6 L/MIN
2,5; 3; 4; 5; 6 L/MIN
10; 12; 15 L/MIN
±10%
±10%
±10%

* Dokładność gwarantowana tylko przy temperaturze 21,11°C (70°F) i ciśnieniu 14,7 psig oraz pod warunkiem, że stosowany jest precyzyjny,
skalibrowany przepływomierz masowy.

PN MN236-C4 E | Instrukcja obsługi PL - 199

Sprint / Stroller / Hi Flow Stroller
Ostrzeżenia
Ważne: Przed rozpoczęciem użytkowania urządzenia
Sprint / Stroller / Hi Flow Stroller należy dokładnie
przeczytać tę instrukcję.
Tylko na receptę.
OSTRZEŻENIE: URZĄDZENIE NIE JEST
PRZEZNACZONE DO PODTRZYMYWANIA
ŻYCIA.
OSTRZEŻENIE: JEŚLI URZĄDZENIE NIE DZIAŁA
PRAWIDŁOWO, NALEŻY SKONTAKTOWAĆ
SIĘ Z PRACOWNIKIEM SŁUŻBY ZDROWIA. NIE
PODEJMOWAĆ SAMODZIELNYCH PRÓB NAPRAWYANI REGULACJI.
OSTRZEŻENIE: NIE MODYFIKOWAĆ URZĄDZENIA BEZUPOWAŻNIENIA OD PRODUCENTA.
OSTRZEŻENIE: JEŚLI UŻYTKOWNIK WYMAGACIĄGŁEGO PODAWANIA TLENU, NALEŻY UPEWNIĆ
SIĘ, ŻE PRZEZ CAŁY CZAS TRWANIA TERAPIIDOSTĘPNE JEST ODPOWIEDNIE ŹRÓDŁO TLENU I/LUB
ZAPASOWE ŹRÓDŁO TLENU.
OSTRZEŻENIE: NIE WOLNO PALIĆ, UŻYWAĆ
ŚWIECZEK ANI OTWARTEGO OGNIA W ODLEGŁOŚCI
MNIEJSZEJ NIŻ 3 M (10 STÓP) OD URZĄDZENIA ORAZ
MNIEJSZEJ NIŻ 20 CM (8 CALI) OD DOWOLNEGOŹRÓDŁA ZAPŁONU.
OSTRZEŻENIE: TRZYMAĆ URZĄDZENIE W DOBRZEWENTYLOWANYM MIEJSCU.
OSTRZEŻENIE: NIE PRZECHOWYWAĆ
SPRZĘTU ZAWIERAJĄCEGO CIEKŁY TLENW SZAFIE, BAGAŻNIKU SAMOCHODU ANIINNYCH ZAMKNIĘTYCH MIEJSCACH. NIE
UMIESZCZAĆ NA URZĄDZENIU KOCÓW,FIRAN ANI INNYCH MATERIAŁÓW.
OSTRZEŻENIE: STOSOWANIE TEGO PRODUKTU MOŻENARAŻAĆ NA KONTAKT Z RÓŻNYMI SUBSTANCJAMI,W TYM Z NIKLEM, KTÓRY W STANIE KALIFORNIAUZNAWANY JEST ZA SUBSTANCJĘ RAKOTWÓRCZĄ.
DODATKOWE INFORMACJE DOSTĘPNE SĄ POD
ADRESEM WWW.P65WARNINGS.CA.GOV.
OSTRZEŻENIE: W PRZYPADKU WYSTĄPIENIAPOWAŻNEGO INCYDENTU Z TYM URZĄDZENIEM,
UŻYTKOWNIK POWINIEN NATYCHMIAST ZGŁOSIĆ
INCYDENT DOSTAWCY I / LUB PRODUCENTOWI.
POWAŻNY INCYDENT JEST DEFINIOWANY JAKO
URAZ, ŚMIERĆ LUB MOŻLIWOŚĆ SPOWODOWANIA
OBRAŻEŃ / ŚMIERCI W PRZYPADKU PONOWNEGO
WYSTĄPIENIA ZDARZENIA. UŻYTKOWNIK MOŻERÓWNIEŻ ZGŁOSIĆ INCYDENT WŁAŚCIWEMU
ORGANOWI W KRAJU, W KTÓRYM ZDARZENIE MIAŁOMIEJSCE.
Przestroga: Urządzenia Sprint / Stroller /Hi Flow Stroller należy używać zgodnie z
zaleceniami lekarza.
Przestroga: Prawo federalne dopuszcza sprzedaż tegourządzenia wyłącznie przez lekarza lub na jego zlecenie
200 - PL PN MN236-C4 E | Instrukcja obsługi

Przestroga: Urządzenie zawiera ciekły tlen, który
jest bardzo zimny — ma prawie -184°C (-300°F).
Kontakt z substancją o tak niskiej temperaturze możedoprowadzić do poważnych odmrożeń.
Przestroga: Ani w stanie ciekłym, ani gazowym tlen
nie jest łatwopalny, ale przyśpiesza spalanie innychmateriałów. W związku z tym zagrożeniem orazniską temperaturą ciekłego tlenu należy stosować
odpowiednie środki ostrożności.
Przestroga: Trzymać materiały łatwopalne z dala odtego urządzenia. Aerozole, oleje i smary, w tym kremy
do twarzy i wazelina, łatwo ulegają zapłonowi i mogąspalać się dynamicznie w obecności tlenu.
Przestroga: Palenie podczas stosowania wąsów
tlenowych może doprowadzić do oparzeń twarzy, anawet śmierci.
Jeśli wąsy zostaną zdjęte i umieszczone na ubraniach,pościeli, sofie lub innym miękkim materiale, a w
pobliżu będzie znajdować się papieros, źródło ciepła
lub płomień albo wystąpi iskrzenie, dojdzie do nagłego
wybuchu pożaru.
Palący użytkownicy powinni: (1) wyłączyć modułprzenośny, (2) wyjąć wąsy i (3) opuścić pomieszczenie,
w którym znajduje się urządzenie.
Przestroga: Jeśli dojdzie do przypadkowego
wywrócenia urządzenia, o ile to możliwe, natychmiast
podnieść je do pozycji pionowej, zachowując
ostrożność. W razie wycieku ciekłego tlenu
natychmiast opuścić miejsce, w którym znajduje
się zbiornik, i skontaktować się z pracownikiemsłużby zdrowia. Nie podejmować prób przeniesieniaurządzenia ani zatrzymania wycieku.
Uwaga: Nie dotykać zamarzniętych części urządzenia.
Uwaga: Nie przechowywać ani nie obsługiwać modułu
przenośnego podłączonego do urządzenia stacjonarnego.
Uwaga: Nie pozwalać osobom bez przeszkolenia używać
ani obsługiwać urządzenia.
Uwaga: Korzystanie z tego urządzenia w samolotach pasażerskich i towarowych jest zabronione przez Federalną
Administrację Lotnictwa.

Przeznaczenie

Sprint/Stroller/Hi Flow Stroller CAIRE jest
przeznaczony do podawania dodatkowego tlenu.
Urządzenie nie jest przeznaczone do podtrzymywaniażycia ani nie zapewnia możliwości monitorowania
pacjenta.

Sprint / Stroller / Hi Flow Stroller
Wprowadzenie

Elementy sterowaniaUrządzenia Sprint / Stroller / Hi Flow Strollersą przeznaczone do podawania tlenu pacjentowi
w warunkach domowych. Mogą też być
wykorzystywane w instytucjach, takich jak domy
opieki czy placówki opieki nieintensywnej. Urządzenia
Sprint / Stroller / Hi Flow Stroller mogą być
ambulatoryjnym źródłem tlenu przed dłuższy okres.

1. Elektroniczny wskaźnik poziomu płynu
2. Elektroniczny przełącznik poziomu płynu
3. Mechaniczny wskaźnik poziomu płynu
4. Pokrętło regulacji przepływu
5. Złącze przewodu doprowadzania tlenu
6. Zawór spustowy
7. Złącza do napełniania (złącza QDV opisano nanastępnej stronie)
8. Pasek z poduszką na ramię
9. Poduszka kondensacyjna
10. Zbiornik kondensacyjnyUrządzenie używane jest przez pacjentów z POChP lubze zmniejszoną pojemnością oddechową. Stosowane
jest na zlecenie lekarza. Urządzenie Sprint / Stroller/ Hi Flow Stroller sprzedawane jest pracownikowi
służby zdrowia, który odbył szkolenie w zakresie
jego obsługi i serwisowania. Pracownik szkoliużytkownika.

124Urządzenie nie jest przeznaczone do podtrzymywaniażycia. Nie zapewnia też żadnych narzędzi domonitorowania pacjenta. Zaleca się, aby w razie awarii
mechanicznej dostępne było alternatywne źródło
tlenu. Urządzenia przenośne wyposażone są w zawór
szybkozłączny, który po podłączeniu do zbiornika
z ciekłym tlenem umożliwia napełnianie. Zbiornik
napełniany jest przez personel medyczny.

58Niniejsza instrukcja obsługi zawiera instrukcjeużytkowania urządzenia Sprint / Stroller / Hi Flow
Stroller. Informacje na temat działania modułu
stacjonarnego zawarte są w dostarczonej z nim
instrukcji obsługi.
Uwaga: Pomoc w przygotowaniu oraz instrukcje prawidłowego użytkowania urządzenia można uzyskać odserwisanta.

576Urządzenia Sprint, Stroller i Hi Flow Stroller

10
9

PN MN236-C4 E | Instrukcja obsługi PL - 201

Sprint / Stroller / Hi Flow Stroller
Identyfikacja zaworów
szybkozłącznych (QDV)Złącze modułu Złącze modułu
przenośnego stacjonarnego

3. Umieścić złącze napełniania pionowo nad złączem
modułu stacjonarnego.
Typ złącza QDVCAIRE –
Bocznenapełnianie
z blokadą
boczneLIFE-OX
Od góryz blokadą

4. a. Jeśli złącze ma mocowanie wciskane:Nałożyć urządzenie Sprint / Stroller / Hi Flow
Stroller na moduł stacjonarny i docisnąć, aż do
odczuwalnego połączenia złącza.

PENOX

TAEMA
Od góry
wciskanePB

b. Jeśli urządzenie ma złącze Penox lub Life-Ox:
Obracać urządzenie Sprint / Stroller / Hi Flow
Stroller aż do odczuwalnego połączenia złącza.
Następnie ostrożnie, ale zdecydowanie, obracaćurządzenie Sprint / Stroller / Hi Flow Stroller
zgodnie z ruchem wskazówek zegara (),
jednocześnie naciskając w dół.

Instrukcja obsługi
Aby napełnić urządzenie Sprint / Stroller / Hi
Flow Stroller ze złączem u góry
1. Wyczyścić złącza napełniania na modulestacjonarnym i urządzeniu Stroller / Spirit /Hi Flow Stroller za pomocą czystej, suchej
i niepozostawiającej włókien ściereczki.
Przestroga: Złącza muszą być suche,
ponieważ wilgoć może spowodować
przymarznięcie urządzenia Sprint / Stroller / Hi
Flow Stroller i modułu stacjonarnego.
2. Upewnić się, że pokrętła regulacji przepływuurządzenia Sprint / Stroller / Hi Flow Stroller
i zbiornika są w pozycji wyłączonej (0).

202 - PL PN MN236-C4 E | Instrukcja obsługi

Uwaga: Dalszy opis umieszczono od kroku 6 w sekcji „Aby
napełnić urządzenie Sprint / Stroller / Hi Flow Stroller zezłączem z boku”.

Aby napełnić urządzenie Sprint / Stroller / Hi
Flow Stroller ze złączem z boku
1. Wyczyścić złącza napełniania na modulestacjonarnym i urządzeniu Sprint / Stroller /Hi Flow Stroller za pomocą czystej, suchej
i niepozostawiającej włókien ściereczki.
OSTRZEŻENIE: ZŁĄCZA MUSZĄ BYĆ SUCHE,PONIEWAŻ WILGOĆ MOŻE SPOWODOWAĆ
PRZYMARZNIĘCIE URZĄDZENIA SPRINT /STROLLER / HI FLOW STROLLER I MODUŁUSTACJONARNEGO.

Sprint / Stroller / Hi Flow Stroller
2. Upewnić się, że pokrętła regulacji przepływuurządzenia Sprint / Stroller / Hi Flow Stroller
i zbiornika są w pozycji wyłączonej (0).

Uwaga: Dokładność odczytów wskaźnika poziomu płynu
zwiększa się 30 sekund po napełnieniu urządzenia Sprint /Stroller / Hi Flow Stroller.

3. Umieścić urządzenie Sprint / Stroller / Hi Flow
Stroller pionowo i zrównać jego złącze napełniania
ze złączem modułu stacjonarnego.

4. Obracać urządzenie Sprint / Stroller / Hi Flow
Stroller w lewo (), aż do odczuwalnego
zablokowania zawleczki i szczeliny.

7. Wokół złącza może też pojawić się para. To
normalne zjawisko spowodowane bardzo dużąróżnicą temperatur pomiędzy ciekłym tlenema ciepłym powietrzem otaczającym moduły.
8. Po napełnieniu urządzenia zamknąć zawór spustowy.
Moduł będzie pełny, gdy ton syczenia zmieni się,a w przewodzie spustowym pojawi się płyn.
Uwaga: Największy stopień napełnienia urządzenia Sprint/ Stroller / Hi Flow Stroller uzyskać można, zamykając
zawór dopiero wtedy, gdy płyn wypływa niemal ciągłym
strumieniem. Metoda ta powoduje jednak utratę pewnejilości tlenu.

5. Ostrożnie, ale zdecydowanie ponownie obrócićurządzenie Sprint / Stroller / Hi Flow Stroller dopozycji pionowej. Moduły są teraz połączone.

Uwaga: Aby zaoszczędzić tlen, należy zamknąć zawór
natychmiast po zaobserwowaniu wypływającego płynu.
Urządzenie Sprint / Stroller / Hi Flow Stroller nie zostanie
jednak w pełni napełnione.
9. W przypadku złącza z mocowaniem wciskanymnależy nacisnąć przycisk zwalniania i rozłączyć
moduły. W przypadku wszystkich pozostałychzłączy należy obracać urządzenie aż do odłączenia
od modułu stacjonarnego.

Uwaga: Czynności od kroku 6.
Uwaga: Moduły ze złączem wciskanym trzymać
nieruchomo podczas napełniania.
6. Otworzyć zawór spustowy, aby rozpocząć
napełnianie urządzenia Sprint / Stroller / Hi Flow
Stroller. Nie przejmować się odgłosem ulatującego
tlenu. Syczenie jest w tym przypadku normalne.

Przycisk zwalniania urządzenia przenośnego
(tylko w przypadku modułów z mocowaniem
wciskanym)

PN MN236-C4 E | Instrukcja obsługi PL - 203

Sprint / Stroller / Hi Flow Stroller
Przestroga: Sprawdzić wskaźnik poziomu
płynu dopiero po zamknięciu zaworu.

Instrukcja obsługi

Przestroga: Jeśli urządzenie Sprint / Stroller / Hi
Flow Stroller jest napełniane natychmiast po okresieużytkowania, napełnić je, ustawić zawór sterowania
przepływem w pozycji zero i odczekać 10 minut,
aby wyrównało się ciśnienie. Następnie użytkować
normalnie.
OSTRZEŻENIE: NIE UŻYWAĆ URZĄDZENIASPRINT / STROLLER / HI FLOW STROLLER,
GDY JEST PRZYMOCOWANE DO MODUŁUSTACJONARNEGO.
OSTRZEŻENIE: JEŚLI PODCZAS ODŁĄCZANIAURZĄDZENIA SPRINT / STROLLER / HI FLOWSTROLLER WYCZUWANY JEST OPÓR, NIE UŻYWAĆ
SIŁY. URZĄDZENIA MOGŁY DO SIEBIE PRZYMARZNĄĆ.
ABY JE ROZŁĄCZYĆ, TRZEBA POCZEKAĆ, AŻ SIĘ
OGRZEJĄ. NIE DOTYKAĆ ZAMARZNIĘTYCH CZĘŚCI.
OSTRZEŻENIE: JEŚLI ZAOBSERWOWANY ZOSTANIEPOWAŻNY WYCIEK W POSTACI STRUMIENIAPŁYNU, NALEŻY NATYCHMIAST OPUŚCIĆMIEJSCE, W KTÓRYM ZNAJDUJE SIĘ URZĄDZENIE,I SKONTAKTOWAĆ SIĘ Z PRACOWNIKIEM SŁUŻBYZDROWIA.
OSTRZEŻENIE: TLEN W POSTACI CIEKŁEJ LUB
ZIMNEGO GAZU MOŻE SPOWODOWAĆ ODMROŻENIE
SKÓRY. TLEN ZWIĘKSZA RYZYKO POŻARU.
Przestroga: Jeśli po rozłączeniu elementów
zaobserwowany zostanie wyciek z modułu
stacjonarnego lub urządzenia Sprint / Stroller /Hi Flow Stroller, należy natychmiast odstawić
produkt Sprint / Stroller / Hi Flow Stroller
tak, aby znajdował się w pozycji pionowej,opuścić pomieszczenie i skontaktować się z
pracownikiem służby zdrowia.
Uwaga: do dostawcy sprzętu: Do NAZWY URZĄDZENIA
zalecane są następujące akcesoria do podawania tlenu:• Kaniula nosowa z rurką o długości 7 stóp (2,1 m) (maks.
LPM: 6 LPM): Numer katalogowy CAIRE 5408-SEQ• Firebreak: Numer katalogowy 21126636
Firebreak jest zalecany do użycia z każdą kaniulą.
• CAIRE oferuje ochronę przeciwpożarową przeznaczoną
do użycia w połączeniu z przenośnym aparatem tlenowym.
Firebreak to bezpiecznik termiczny, który zatrzymuje
przepływ gazu w przypadku zapalenia następnej kaniuli
lub rurki tlenowej i zapalenia się do wybuchu pożaru. Jest
on umieszczony w linii z kaniulą nosową lub rurką tlenową
między pacjentem a wylotem tlenu w Sprint / Stroller/ Hi Flow Stroller. W celu prawidłowego użycia osłony
przeciwpożarowej należy zawsze postępować zgodnie z
instrukcjami producenta (dołączonymi do każdego zestawu
przeciwpożarowego).
• Dodatkowe informacje na temat zalecanych akcesoriów
są dostępne online na stronie www.caireinc.com.
204 - PL PN MN236-C4 E | Instrukcja obsługi(Wąsy nie są częścią zestawu)

1. Zdecydowanie nacisnąć rurkę wąsów tlenowych nazłącze przewodu doprowadzania tlenu.
2. Ustawić wąsy tlenowe tak, aby nie odczuwać
dyskomfortu podczas oddychania tlenem.
3. Obrócić w prawo pokrętło regulacji przepływu
na górze urządzenia Sprint / Stroller / Hi Flow
Stroller, tak aby w okienku widoczne było zalecone
ustawienie natężenia przepływu (wartość). Upewnićsię, że pokrętło przeskoczyło na miejsce.
Przestroga: Jeśli pokrętło będzie ustawionepomiędzy pozycjami natężenia, przepływbędzie niezgodny z oznaczeniami.
4. Powinno rozpocząć się podawanie tlenu. Aby
zatrzymać przepływ tlenu, obrócić pokrętło
regulacji przepływu w lewo do pozycjiwyłączonej (0).
Przestroga: Gdy urządzenie nie jest używane,
zawsze ustawiać pokrętło regulacji przepływu
w pozycji wyłączonej (0).

Sprint / Stroller / Hi Flow Stroller
5. Poniższa tabela pozwala ustalić czas pracyurządzenia Sprint / Stroller / Hi Flow Stroller:
Ustawienie zaworu sterowania przepływem
Model

SprintStroller

Hi Flow
Stroller

Nominalne

Nominalne

Ust.
zaw.

SprintStroller

Hi Flow
Stroller
71:00

Wył.

30:00

60:00

0,25

18:00

30:00

ND

0,5

12:12

23:00

32:00

0,75

09:18

17:24

ND

1

07:30

14:06

16:00

1,5

05:30

10:12

ND

2

04:18

08:00

08:00

2,5

03:36

06:36

06:20

3

03:06

05:36

05:20

4

02:24

04:24

04:00

5

02:00

03:36

ND

6

01:42

03:00

02:40

8

ND

ND

02:00

10

ND

ND

01:40

12

ND

ND

01:20

15

ND

ND

01:00

Uwaga: Czasy podano w godzinach i minutach(format 00:00).
Uwaga: Czasy „nominalne” dotyczą warunków
idealnych, tzn. przy maksymalnym napełnieniu,
dokładnym ustawieniu natężenia przepływu,niskim poziomie strat, gdy urządzenie nie jest
przesuwane itd.
Podano maksymalne oczekiwane czasy pracy.
Uwaga: Czasy uzyskiwane przez użytkownika mogąsię różnić od podanych.
6a. Aby sprawdzić poziom płynu w urządzeniu za
pomocą elektronicznego wskaźnika poziomu płynu:• Wskaźnik 4. generacji: Nacisnąć zielony przycisk
funkcyjny na wskaźniku. Poziom płynu zostanie
wskazany przez kontrolki LED ułożone w kształtłuku.

Przestroga: Jeśli świeci tylko ostatni segment
paska kontrolki, urządzenie Sprint / Stroller / Hi
Flow Stroller jest puste.
Jeśli po naciśnięciu przycisku zapali się wskaźnik
niskiego poziomu baterii, poinformować
pracownika służby zdrowia.
Uwaga: Urządzenie Sprint / Stroller / Hi Flow Stroller nie
przerwie podawania tlenu nawet po zaświeceniu się lampki
niskiego poziomu baterii, chyba że skończy się ciekły tlen.
6b. Aby sprawdzić poziom płynu w urządzeniu za
pomocą mechanicznego wskaźnika poziomu płynu:• Chwycić urządzenie za pasek materiałowy tuż
nad czarną obudową mechanizmu sprężynowegowskaźnika poziomu.
• Poczekać, aż urządzenie ustabilizuje się (przestaniesię bujać), a następnie odczytać wynik na widocznej
podziałce kolorowej, wskazującej przybliżony
poziom ciekłego tlenu.

• Wskaźnik 3. generacji: Nacisnąć przycisk(przełącznik poziomu płynu) na górze urządzenia
i przytrzymać przez przynajmniej dwie sekundy.
Poziom płynu zostanie wskazany w górnej części
paska kontrolki.

PN MN236-C4 E | Instrukcja obsługi PL - 205

Sprint / Stroller / Hi Flow Stroller
Uwaga: Jeśli urządzenie jest puste, widoczny będzie
tylko czerwony pasek podziałki. W takim wypadku przedużyciem należy napełnić moduł.
7. W określonych warunkach atmosferycznych, gdyurządzenie jest używane ciągle i, w szczególności,jeśli nie jest przenoszone, może dojść na nim do
znacznego oszronienia w okolicy wężownicy
parownika i na obudowie. Można ograniczyć tozjawisko, stukając w urządzenie i strącając szron
z obudowy.

W przypadku osłony miękkiej• Odsunąć zamek u dołu i zdjąć dolną osłonę.
• Wyjąć poduszkę ze zbiornika i wykręcić,aby pozbyć się wilgoci. Przed ponownymwłożeniem poduszki do zbiornika poczekać aż
całkowicie wyschnie.
Uwaga: Jeśli poduszka jest mocno zanieczyszczona,można ją wyprać w pralce (dezynfekcja za pomocą
wybielacza).
• Aby ponownie zamontować, umieścić poduszkę
w osłonie dolnej i zasunąć zamek.

Konserwacja poduszki
kondensacyjnej

Uwaga: Jeśli nie można nacisnąć gałki, aby zamocować
zbiornik, chwycić go i pociągnąć za gałkę. Kliknięcie
oznacza, że gałka jest w pozycji otwartej. Zamontowaćzbiornik w sposób opisany powyżej.

Gdy urządzenie Sprint / Stroller / Hi Flow Stroller jest
puste i ogrzane do temperatury pokojowej, osuszyć
zbiornik kondensacyjny.

Ograniczanie oszronienia

W przypadku osłony twardej• Przechylić urządzenie. Chwycić gałkę
w środkowej części zbiornika kondensacyjnego(w dolnej części urządzenia) i pociągnąć dosiebie. Spowoduje to zwolnienie zbiornika.

W określonych warunkach atmosferycznych, gdyurządzenie jest używane ciągle i, w szczególności, jeśli
nie jest przenoszone, może dojść na nim do znacznego
oszronienia w okolicy wężownicy parownika i na
obudowie. Można ograniczyć to zjawisko, stukając
w urządzenie i strącając szron z obudowy.

Uwaga: Jeśli poduszka jest mocno zanieczyszczona,można ją wyprać w pralce (dezynfekcja za pomocą
wybielacza).
• Aby ponownie zamontować zbiornik, ustawić
go nad okrągłym otworem w dolnej części
urządzenia i nacisnąć gałkę. Pierwsze kliknięcie
oznacza połączenie zbiornika, a drugie —zablokowanie.

OSTRZEŻENIE: ZE WZGLĘDU NAMOŻLIWOŚĆ GROMADZENIA SIĘ DUŻEJ
ILOŚCI WILGOCI / POWSTAWANIA LODU,
PO UŻYCIU ZALECA SIĘ PRZEZ GODZINĘ
NIE KORZYSTAĆ Z URZĄDZENIA HI FLOWSTROLLER. POZWOLI TO ZMNIEJSZYĆ
ILOŚĆ GROMADZĄCEJ SIĘ WILGOCI.

ZablokowanePołączone
OtwarteGałka zbiornika

Uwaga: Jeśli nie można nacisnąć gałki, aby zamocować
zbiornik, chwycić go i pociągnąć za gałkę. Kliknięcie
oznacza, że gałka jest w pozycji otwartej. Zamontowaćzbiornik w sposób opisany powyżej.

206 - PL PN MN236-C4 E | Instrukcja obsługi

Ze względu stosowane w urządzeniu Hi Flow Strollerwyższe natężenie przepływu, dochodzi na nim do
kondensacji i oszraniania, szczególnie w przypadku
stosowania ustawień przepływu 10 L/MIN i wyższych.
Zaleca się, aby po użyciu przynajmniej przez
godzinę nie korzystać z urządzenia. Pozwoli tozmniejszyć powstawanie lodu spowodowane ciągłym
użytkowaniem.

Sprint / Stroller / Hi Flow Stroller
Rozwiązywanie problemówPoniższe informacje ułatwiają rozwiązywanie prostych problemów związanych z obsługą, które mogą wystąpić
podczas użytkowania urządzenia Sprint / Stroller / Hi Flow Stroller.
Problem
Z modułu przenośnego
słychać syczenie.

Podawanie tlenu ustaje
podczas użytkowania.

Rozwiązanie• Syczenie może występować podczas regulacji ciśnienia roboczego modułu przenośnego.
Dzieje się tak najczęściej po napełnianiu lub zmianie pozycji modułu przenośnego. Po
napełnianiu syczenie może być słyszane przez około 10 minut. Może też występować, gdy
za pomocą zaworu regulacji przepływu wybrano niskie ustawienie.
• Jeśli moduł przenośny znajduje się w nieprawidłowej pozycji, ustawić go pionowo ipoczekać kilka minut, aż się ustabilizuje.
• Upewnić się, że wąsy tlenowe są prawidłowo podłączone do złącza wylotowego tlenu.
• Upewnić się, że wąsy tlenowe nie są zagięte.
• Sprawdzić wskaźnik napełnienia/poziomu i w razie potrzeby napełnić moduł przenośny.

Moduł przenośny nie jest
napełniany.

• Upewnić się, że pokrętło regulacji przepływu nie jest ustawione w pozycji wyłączonej („0”).
• Sprawdzić, czy w zbiorniku jest tlen.

• Upewnić się, że złącza napełniania modułu przenośnego i zbiornika są dobrze połączone
podczas całego procesu napełniania.
Zawór spustowy modułu• Jeśli zawór spustowy nie zamyka się, a syczenie i wydobywanie się tlenu nie ustaje,przenośnego nie zamyka się
ostrożnie zdjąć moduł przenośny, naciskając przycisk zwalniania na zbiorniku.
prawidłowo po zakończeniu
Upuszczanie gazu ze spodu modułu przenośnego ustanie po kilku minutach. Poczekać,napełniania.
aż moduł ogrzeje się, aby można było zamknąć zawór spustowy. Przywracanie
prawidłowego ciśnienia w module przenośnym, koniecznego do generowania
wymaganego natężenia przepływu, może potrwać do 60 minut. W razie potrzeby użyćalternatywnego źródła tlenu, takiego jak zawór sterowania przepływem podłączony dozbiornika.
Odłączanie modułu• Mogło dojść do zamarznięcia złączy napełniania w module przenośnym i zbiorniku.
przenośnego od zbiornika po• NIE UŻYWAĆ NADMIERNEJ SIŁY. Poczekać kilka minut, aż zamarznięte elementy
napełnianiu jest utrudnione.
ogrzeją się, a po stopieniu lodu odłączyć moduł przenośny. Aby zapobiec przymarzaniu
do siebie modułów, przed napełnianiem zawsze wycierać męskie złącze napełniania nazbiorniku i żeńskie złącze napełniania na module przenośnym za pomocą czystej, suchej
szmatki.

PN MN236-C4 E | Instrukcja obsługi PL - 207

Sprint / Stroller / Hi Flow Stroller
Czyszczenie
OSTRZEŻENIE: CZYŚCIĆ TYLKO, GDYURZĄDZENIE JEST PUSTE.
• Czyścić za pomocą roztworu łagodnego płynu
do zmywania naczyń w wodzie.
• Nałożyć środek czyszczący bezpośrednio
na niepozostawiającą włókien ściereczkę.
Zatwierdzone środki czyszczące to m.in.
HydroPure i HydroKlean. Nie nanosić środków
czyszczących bezpośrednio na urządzenie
Sprint / Stroller / Hi Flow Stroller.
• Wycierać powierzchnię zewnętrzną
niepozostawiającą włókien ściereczką aż dowyczyszczenia.
Przestroga: Do czyszczenia urządzenia
nie używać sprzętu generującego wysoką
temperaturę i ciśnienie.
• Uważać, aby nie nanieść środka czyszczącego
na komponenty wewnętrzne ani zawory.
• Przed użyciem poczekać, aż urządzenie
całkowicie wyschnie.
Uwaga: Dla pracownika służby zdrowia — informacje na
temat procedury przygotowania sprzętu do ponownegoużycia znajdują się w instrukcji technicznej.

Dyrektywy WEEE i RoHS

Ten symbol przypomina właścicielom sprzętu,że zgodnie z Dyrektywą w sprawie zużytegosprzętu elektrycznego i elektronicznego(WEEE) po zakończeniu eksploatacji należy go oddać
do zakładu utylizacji.
Nasze produkty są zgodne z Dyrektywą w sprawie
ograniczenia stosowania niektórych niebezpiecznych
substancji w sprzęcie elektrycznym i elektronicznym(RoHS). Zawierają jedynie śladowe ilości ołowiu
i innych niebezpiecznych materiałów.

Uwaga: Zakres dozwolonego ciśnienia atmosferycznego
wynosi od 700 hPa do 1060 hPa (wysokość od 3050 m(10 000 ft) do -300 m (-1000 ft)).

KonserwacjaZa wszelkie prace konserwacyjne wskazane
w instrukcji technicznej odpowiedzialny jest
serwisant. Jeśli konieczne jest przeprowadzenie
prac konserwacyjnych, należy skontaktować sięz serwisantem.
Jedyną częścią serwisowaną przez użytkownika jest
poduszka kondensacyjna. Pozostałe części nie są
serwisowane przez użytkownika.
Przewidywany okres eksploatacji urządzenia toprzynajmniej pięć lat.
Pomiędzy napełnieniami należy oczyszczać złącza
napełniania na module stacjonarnym i przenośnym za
pomocą czystej, suchej i niepozostawiającej włókien
ściereczki, aby zapobiec zamarzaniu i uszkodzeniu
sprzętu.
Uwaga: Wszelkie inne wymagane prace konserwacyjne
powinien wykonywać serwisant.

Akcesoria

Utylizacja
W celu prawidłowej utylizacji urządzenie Sprint /Stroller / Hi Flow Stroller wraz z wszystkimi
komponentami należy oddać do placówki opieki.
Można też skontaktować się z urzędem gminy
lub miasta, aby dowiedzieć się, jak prawidłowo
zutylizować akumulator.

Transport i przechowywanieUrządzenie należy przechowywać w pozycji pionowej
w dobrze wentylowanym miejscu. Nie należy kłaść
urządzenia na boku. Dozwolona wilgotność to 95%,
bez kondensacji. Zakres temperatur wnosi od -40°F do
158°F (od -40°C do 70°C).
Zakres temperatur pracy wynosi od 14°F do 104°F(od -10°C do 40°C). Zakres wilgotności względnej
wynosi od 30% do 75%, bez kondensacji.
208 - PL PN MN236-C4 E | Instrukcja obsługi

Wózek transportowy PN 20765565

Sprint / Stroller / Hi Flow Stroller
Bezpieczeństwo
OSTRZEŻENIE: PRZENOŚNY SPRZĘT RADIOKOMUNIKACYJNY PRACUJĄCY NA CZĘSTOTLIWOŚCIACHRADIOWYCH (NP. KABLE ANTENOWE I ANTENY ZEWNĘTRZNE) NIE POWINIEN BYĆ UŻYWANY BLIŻEJDOWOLNEJ CZĘŚCI URZĄDZENIA SPRINT / STROLLER / HI FLOW STROLLER NIŻ 30 CM (12 CALI), W
TYM PRZEWODÓW WSKAZANYCH PRZEZ PRODUCENTA. W PRZECIWNYM RAZIE MOŻE DOJŚĆ DO
POGORSZENIA WYDAJNOŚCI URZĄDZENIA.
OSTRZEŻENIE: STOSOWANIE AKCESORIÓW, PRZETWORNIKÓW I KABLI INNYCH, NIŻ WSKAZANE LUB
DOSTARCZONE PRZEZ PRODUCENTA TEGO URZĄDZENIA, MOŻE DOPROWADZIĆ DO ZWIĘKSZENIA EMISJI
PROMIENIOWANIA ELEKTROMAGNETYCZNEGO LUB ZMNIEJSZENIA ODPORNOŚCI ELEKTROMAGNETYCZNEJ
URZĄDZENIA ORAZ SPOWODOWAĆ NIEPRAWIDŁOWE DZIAŁANIE.
OSTRZEŻENIE: NIE NALEŻY USTAWIAĆ TEGO URZĄDZENIA W POBLIŻU INNEGO SPRZĘTU ANI NA NIM, PONIEWAŻ
MOŻE TO DOPROWADZIĆ DO NIEPRAWIDŁOWEGO DZIAŁANIA. JEŚLI NIE DA SIĘ TEGO UNIKNĄĆ, NALEŻY
MONITOROWAĆ URZĄDZENIE ORAZ INNY SPRZĘT, ABY MIEĆ PEWNOŚĆ, ŻE DZIAŁA PRAWIDŁOWO.
Przestroga: Elektryczny sprzęt medyczny wymaga zastosowania specjalnych środków ostrożności w
zakresie kompatybilności elektromagnetycznej. Należy go instalować i wdrażać zgodnie z informacjami EMC
zawartymi w tej instrukcji.
Przestroga: Przenośny sprzęt komunikacyjny wykorzystujący częstotliwości radiowe może oddziaływać na
elektryczny sprzęt medyczny.
Tabela 1

Wskazówki i deklaracja producenta — emisja promieniowania elektromagnetycznegoUrządzenie Sprint / Stroller / Hi Flow Stroller przeznaczone jest do użytku w środowisku elektromagnetycznym
opisanym poniżej. Klient lub użytkownik urządzenia Sprint / Stroller / Hi Flow Stroller powinien przestrzegać
tej zasady w danym środowisku.
Test emisji

Zgodność

Środowisko elektromagnetyczne — wskazówki

Emisje o częstotliwości
fal radiowych

Grupa 1Urządzenie Sprint / Stroller / Hi Flow Stroller wykorzystuje
energię o częstotliwości radiowej wyłącznie dowewnętrznego działania.

CISPR 11

Emisje o częstotliwości
fal radiowych CISPR 11

W związku z tym emisje o częstotliwości radiowej sąniewielkie i nie powinny powodować żadnych zakłóceń w
działaniu znajdującego się w pobliżu sprzętu elektronicznego.
Klasa B

Emisje harmoniczne
IEC 61000-3-2

Nie dotyczy

Emisje fluktuacji /migotania napięcia
IEC 61000-3-3

Nie dotyczyUrządzenie Sprint / Stroller / Hi Flow Stroller jest
dostosowane do działania w każdym środowisku, w tym
w warunkach domowych oraz miejscach bezpośredniopodłączanych do publicznej sieci zasilania o niskim napięciu,która zasila budynki mieszkalne.

PN MN236-C4 E | Instrukcja obsługi PL - 209

Sprint / Stroller / Hi Flow Stroller
Tabela 2*: Zalecane odległości między przenośnym sprzętem komunikacyjnym pracującym na częstotliwościachradiowych a urządzeniem Sprint / Stroller / Hi Flow StrollerUrządzenie Sprint / Stroller / Hi Flow Stroller powinno być używane w środowisku elektromagnetycznym, w którym zakłócenia
promieniowania o częstotliwościach radiowych są pod kontrolą. Klient lub użytkownik urządzenia Sprint / Stroller / Hi Flow Strollermoże zapobiec zakłóceniom elektromagnetycznym poprzez utrzymywanie minimalnej odległości pomiędzy przenośnym sprzętem
komunikacyjnym działającym na częstotliwościach radiowych (nadajnikami) a produktem Sprint / Stroller / Hi Flow Stroller w sposób
opisany poniżej, zgodnie z maksymalną mocą wyjściową sprzętu komunikacyjnego.
Maksymalna wartość
Odległość w zależności od częstotliwości nadajnika
mocy wyjściowejmnadajnika
80 MHz do 800 MHz
800 MHz do 2,5 GHz
150 kHz do 80 MHz

d = 1,2√P

d = 1,2√P

d = 2,3√PW
0,010,12 m0,12 m0,23 m0,10,38 m0,38 m0,73 m
1
1,2 m
1,2 m
2,3 m
10
3,8 m
3,8 m
7,3 m
100
12 m
12 m
23 m
W przypadku nadajników, których maksymalna moc wyjściowa nie jest wymieniona powyżej, zalecaną odległość (d) pomiędzy
urządzeniami w metrach (m) można oszacować, stosując odpowiednie równanie dla częstotliwości nadajnika, gdzie P jest
maksymalną wartością mocy wyjściowej nadajnika w watach (W), zgodnie z danymi producenta nadajnika.
UWAGA 1: Przy 80 MHz i 800 MHz stosowana jest odległość dla zakresu większych częstotliwości.
UWAGA 2: Te zalecenia mogą nie mieć zastosowania we wszystkich sytuacjach. Na propagację elektromagnetyczną wpływa
pochłanianie i odbijanie od struktur, przedmiotów i ludzi.

* Tabela ta jest zgodnie z wymaganiami dostarczana ze sprzętem przetestowanym na odpowiednim poziomie
i w określonym zakresie częstotliwości, w wyniku czego stwierdzono zgodność z przepisami.

Tabela 3: Wskazówki i deklaracja producenta — odporność elektromagnetycznaUrządzenie Sprint / Stroller / Hi Flow Stroller przeznaczone jest do użytku w środowisku elektromagnetycznym opisanym poniżej.
Klient lub użytkownik urządzenia Sprint / Stroller / Hi Flow Stroller powinien przestrzegać tej zasady w danym środowisku.
Test odporności
Poziom testu IEC 60601 Poziom zgodności
Środowisko elektromagnetyczne - wskazówki
Podłogi powinny być drewniane, betonowe lub± 8 kV przy dotyku
Wyładowanie elektro- ± 8 kV przy dotyku
wyłożone płytkami ceramicznymi. Jeśli podłogi
statyczne (ESD)±2 kV, ± 4 kV, ±8 kV,±2 kV, ± 4 kV, ±8 kV,
pokryte są materiałem syntetycznym, wilgotność
IEC 61000-4-2± 15 kV przez powietrze± 15 kV przez powietrze względna powinna wynosić przynajmniej 30%.**
Nie dotyczy
Nagły skok/impuls±2 kV dla linii zasilających Urządzenie zasilaneprądem DC
Nie dotyczyelektryczny±1 kV dla linii wejścia/Nie dotyczywyjścia
IEC 610004-4
Brak danych dla liniiwejścia/wyjścia
Nie
dotyczy
Skok napięcia
±1 kV tryb różnicowy
Nie dotyczyUrządzenie zasilane
±2
kV
tryb
wspólny
IEC 61000-4-5prądem DC<5% UT (>95% spadek w
UT) przez 0,5 cyklu
Spadki, krótkie przerwy
40% UT (60% spadek w
oraz zmiany napięcia
Nie dotyczyna liniach wejściowych UT) przez 5 cyklów
Nie dotyczyUrządzenie zasilane
70%
UT
(30%
spadekw
zasilaniaprądem DC
UT) przez 25 cyklów
EC 61000-4-11<5% UT (>95% spadek w
UT) przez 5 s
Pole magnetyczne oPole magnetyczne o częstotliwości źródłaczęstotliwości źródła
3A/m
3A/m
zasilania powinno być na poziomie typowym dlazasilania (50/60 Hz)
50/60 Hz50/60 Hzśrodowiska komercyjnego lub szpitalnego.
IEC 61000-4-8
Uwaga: UT to napięcie sieci AC przed zastosowaniem poziomu testowego.

** To oświadczenie wskazuje, że przeprowadzono wymagane testy w środowisku kontrolowanym i stwierdzono
zgodność urządzenia Sprint / Stroller / Hi Flow Stroller z przepisami.

210 - PL PN MN236-C4 E | Instrukcja obsługi

Sprint / Stroller / Hi Flow Stroller
Tabela 4
Wskazówki i deklaracja producenta — odporność elektrycznego sprzętu medycznego i elektrycznych
systemów medycznych
Wskazówki i deklaracja producenta — odporność
Urządzenie Sprint / Stroller / Hi Flow Stroller przeznaczone jest do użytku w środowisku
elektromagnetycznym opisanym poniżej. Klient lub użytkownik urządzenia Sprint / Stroller / Hi Flow Stroller
powinien przestrzegać tej zasady w danym środowisku.
Test odporności

Poziom testu IEC 60601

Poziom zgodności

Środowisko elektromagnetyczne —wskazówki

Przewodzonaczęstotliwość radiowa
IEC 61000-4-6

3 Vrms6 Vrms (pasma ISM)
od 150 kHz do 80 MHz

Nie dotyczyPrzenośny sprzęt radiokomunikacyjny nie powinien
być używany bliżej dowolnej części urządzenia Sprint/ Stroller / Hi Flow Stroller, włącznie z przewodami,
niż zalecana odległość wyliczona ze wzoru
odpowiedniego dla częstotliwości nadajnika.

Urządzenie zasilane
akumulatorem, brak
SIP/SOP

Zalecana odległość pomiędzy urządzeniami
d = 1,2 √P

Promieniowanieo częstotliwości
radiowej IEC 61000-4-3

od 80 MHz do 2,7 GHz

10 V/m80 MHz – 2,7 GHz
80% AM przy 1 kHz

d = 1,2 √P
d = 2,3 √P
Gdzie P jest maksymalną wartością mocy wyjściowej
nadajnika w watach (W) zgodnie z danymi
producenta nadajnika, a d jest zalecaną odległością
pomiędzy urządzeniami w metrach (m).
Natężenia pola pochodzącego z umieszczonych na
stałe nadajników pracujących na częstotliwościachradiowych, określone na podstawie pomiaru poziomu
zakłóceń elektromagnetycznych w miejscu montażua
powinny być mniejsze niż poziom zgodności wkażdym zakresie częstotliwościb.
Zakłócenia mogą występować w pobliżu urządzeń
oznaczonych następującymi symbolami:Częstotliwość
testowa (MHz)Pasmoa)(MHz)

Usługaa)

Modulacjab)

385

380–390TETRA 400450

430–470

GMRS 460, FRS 460

Modulacja impulsowab)
18 Hz
FMC)
odchylenie ±5 kHz
fala sinusoidalna 1 kHz

704–787

Pasmo LTE 13, 17

Moc
Odległość
maksymalna(m)(W)

Poziom testu
odporności
(V/m)

1,8

0,3

27

2

0,3

28

Modulacja impulsowab)
217 Hz

0,2

0,3

9

800–960

GSM 800/900, TETRA 800, Modulacja impulsowab)
IDEN 820, CDMA 850,
18 Hz
pasmo LTE 5

2

0,3

28

1700–1900

GSM 1800; CDMA 1900;
GSM 1900; DECT; pasmo
LTE 1, 3, 4, 25; UMTS

Modulacja impulsowab)
217 Hz

2

0,3

28

2450

2400–2570

Modulacja impulsowab)
217 Hz

2

0,3

28

5240
5500
5785

Bluetooth, WLAN, 802.11
b/g/n, RFID 2450, pasmo
LTE 75100–5800

WLAN 802.11 a/n

Modulacja impulsowab)
217 Hz

0,2

0,3

9

710
745
780
810
870
930
1720
1845
1970

Uwaga: Aby uzyskać POZIOM TESTU ODPORNOŚCI, można zmniejszyć odległość pomiędzy anteną nadawczą a urządzeniem Sprint / Stroller / Hi Flow Stroller do 1 m. Odległość testowa 1 m jest dozwolona w normie IEC 61000-4-3.
W przypadku niektórych usług podano tylko częstotliwości łączy w górę.
Nośna jest modulowana za pomocą 50% roboczego sygnału o przebiegu prostokątnym.
Jako alternatywy dla modulacji FM można użyć 50% modulacji impulsowej przy 18 Hz. Mimo że nie jest to modulacja rzeczywista, stwarza
trudniejsze warunki.
abc

CAIRE i CAIRE Inc. są zastrzeżonymi znakami towarowymi firmy CAIRE Inc. Aby uzyskać pełną listę znak.w towarowych, odwiedź naszą stronę internetową poniżej. Znaki towarowe: www.
careinc.com/corporate/trademarks.
Copyright © 2023 CAIRE Inc. Firma CAIRE Inc. zastrzega sobie prawo do wstrzymania sprzedaży produktów, zmiany cen, materiałów, sprzętu, jakości, opisów, danych technicznych lubprocesów związanych z produktami w dowolnym momencie, bez wcześniejszego powiadomienia oraz bez dodatkowych zobowiązań i konsekwencji. Firma zastrzega sobie wszystkie
prawa, które nie zostały wyraźnie określone w niniejszym dokumencie (jeśli ma to zastosowanie).

PN MN236-C4 E | Instrukcja obsługi PL - 211

Sprint / Stroller / Hi Flow Stroller
Szimbólumok jegyzéke
ISO 7000Az egység közelében vagy annaküzemeltetésekor tilos a dohányzás!
Reg. szám: P002Esőtől védje, tartsa szárazon.
Reg. sz.: 0626

BF típusú alkalmazott alkatrész(elektromos áramütés elleni védelem
fokozata). Reg. szám: 5333

Halmozási határ szám szerint. Reg. #2403
A gyártó neve és címe. Reg. sz.: 3082

Vigyázat! Reg. szám: W001
A 93/42/EGK tanácsi irányelv
Meghatalmazott képviselő az Európai
Közösségben

A gyártás országa és dátuma. A „CC”
a gyártás országának kétbetűs országkódját jelöli. A gyártás dátuma ÉÉÉÉHH-NN formátumban van megadva.
Reg. # 6049
Figyelem, nézze meg akísérődokumentációt! Reg. sz.: 0434A

Ha a termék egyedi eszköz
azonosítójának (UDI) címkéjén CE ####
szimbólum szerepel, akkor az eszköz
megfelel a 93/42 / EGK irányelvnek az
orvostechnikai eszközökre vonatkozó
követelményeinek. A CE ####
szimbólum jelzi a bejelentett szervezet
számát.

Katalógusszám. Reg. sz.: 2493

Ez az eszköz megfelel a 2010/35/
EU orvostechnikai eszközökről szóló
irányelv követelményeinek. Rendelkezik
a feltüntetett PI jelöléssel.

Sorozatszám. Reg. sz.: 2498
Tárolási vagy működési hőmérsékleti
tartomány. Reg. sz.: 0632

ADR
Nem mérgező gáz.

PTárolási páratartalom. Reg. sz.: 2620
Légköri nyomás szerinti korlátozás.
Reg. sz.: 2621

Veszély Oxidáló anyagok: tűzfokozó
veszély. Az oxidálószerek a tüzeterőteljesebben égetik.

Ezzel a felével felfelé. Reg. sz.: 0623

Hűtött folyadék, USP; a levegő
cseppfolyósításával előállítva.

Törékeny, óvatosan kezelendő! Reg. sz.:
0621

További szimbólumok
Az egységet mindig tartsa jól szellőző
helyen.

Veszélyes anyagokat tartalmaz. Reg.
# 3723

Tartsa távol a gyúlékony anyagoktól,
olajtól és zsírtól.

Importőr. Reg. # 3725
ISO 7010

Hideg folyadékkal, a gázhalmazállapotú oxigénnel vagya jeges részekkel való érintkezés
fagyást okozhat. Vigyázat: alacsonyhőmérséklet. Az alacsony hőmérsékletrevagy a fagyás állapotára vonatkozó
figyelmeztetés. Reg. szám: W010
Olvassa el a használati útmutatót.
Reg. szám: M002
Tartsa távol nyílt lángtól, tűztől,szikrától. A dohányzás és a nyílt lánghasználata tilos! Reg. szám: P003

212 - HU PN MN236-C4 E | Felhasználói kézikönyv

Meghatalmazott képviselő Svájcban.
Ha az eszközön az UKCA jelölés
látható, amint az UKCA#### jelzi a bejelentett szervezet számát, ez az eszköz
megfelel az UKCA előírásainak.
IEC 60417

Ne takarja le az egységet, és ne hordja
a hordozható egységet a ruhája alatt.
Ezek az egységek általában oxigént
bocsátanak ki. 5641 sz

Sprint / Stroller / Hi Flow Stroller
ISO 15223-1

21 CFR 801.15
Az Amerikai Egyesült Államok
szövetségi törvényei szerint a készülék
kizárólag orvos által vagy orvosi
megrendelésre értékesíthető.
IEC 60601-1
Cseppálló
A 2012/19/EU tanácsi irányelv
ZSEE
Ten symbol ma przypominać właścicielom sprzętu o zwróceniu go do
zakładu recyklingu po zakończeniu jegoeksploatacji, zgodnie z dyrektywą w
sprawie zużytego sprzętu elektrycznego
i elektronicznego (WEEE).
Nasze produkty będą zgodne z dyrektywą dotyczącą substancji niebezpiecznych (RoHS). Nie będą zawierać więcejniż śladowe ilości ołowiu lub innychmateriałów niebezpiecznych.

Orvosi eszköz. Reg. # 5.7.7
Egyedi eszközazonosító # 5.7.10
Ez a termék egy vagy több USA-beli és nemzetközi
szabadalom védelme alatt állhat. A vonatkozó
szabadalmak listájának megtekintéséhez, k.rjük,
látogasson el weboldalunkra az alábbi címen.
Szabadalmak: www.caireinc.com/corporate/patents/.

Műszaki jellemzők
• Üzemmód: folyamatos áramlás• Elektromos áramütés elleni védelem típusa: belső tápellátású berendezés• Elektromos áramütés elleni védelem fokozata: BF típusú alkalmazott alkatrész• Osztályozás a víz behatolása ellen nyújtott védelem fokozata szerint: IP22 – Általános berendezés• A berendezés nem használható tűzveszélyes keverékek jelenlétében

Termékjellemzők

SprintStroller

Hi Flow Stroller

Cseppfolyósoxigén-tárolási
kapacitás
Gázhalmazállapotra
vonatkozó
befogadóképességÜres tömeg

0,68 kg

1,36 kg

1,36 kg

513 l1026 l1025 l

2,04 kg

2,27 kg

2,49 kg

Feltöltött tömeg

2,72 kg

3,63 kg

3,86 kg

Magasság

298 mm

343 mm

343 mmMéretek

123,83 mm D × 191 mm W

149,23 mm D × 191 mm W

149 mm D × 191 mm W

Tipikus használati idő
2 LITER/PERC eseténÜzemi nyomás

4,3 óra8,5 óra8,5 óra20 psi (137 kPa)

20 psi (137 kPa)

20 psi (137 kPa)

Normál párolgási arány

0,57 kg/nap

0,57 kg/nap

0,57 kg/nap

Standardáramlásszabályozási
tartományAz áramlási sebesség
pontossága*

Ki; 0,25; 0,5; 0,75; 1; 1,5; 2;
2,5; 3; 4; 5; 6 l/perc

Ki; 0,25; 0,5; 0,75; 1; 1,5; 2;
2,5; 3; 4; 5; 6 l/perc

Ki; 0,5; 1; 2; 2,5; 3; 4; 6; 8; 10;12; 15 l/perc

±10%

±10%

±10%

* Ez a pontosság kizárólag 70 F-en és 14,7 négyzethüvelyken egy hitelesített, pontos tömegáramlás-mérővel mérve érvényes.

PN MN236-C4 E | Felhasználói kézikönyv HU - 213

Sprint / Stroller / Hi Flow Stroller
Figyelmeztető információ
Fontos: A Sprint/Stroller/Hi Flow Stroller készülékek
használata előtt figyelmesen olvassa el az útmutatót.
Kizárólag orvosi rendelvényre.
VIGYÁZAT: EZ AZ ESZKÖZ NEM ALKALMASÉLETMENTÉS CÉLJÁBÓL TÖRTÉNŐ
FELHASZNÁLÁSRA.
VIGYÁZAT: AMENNYIBEN ÚGY ÉRZI, HOGY A
BERENDEZÉS NEM MŰKÖDIK MEGFELELŐEN, VEGYEFEL A KAPCSOLATOT ORVOSÁVAL. NE PRÓBÁLJAMEG AZ EGYSÉGET MEGJAVÍTANI VAGY MÓDOSÍTANI.
VIGYÁZAT: A GYÁRTÓ JÓVÁHAGYÁSA NÉLKÜL A
BERENDEZÉS MÓDOSÍTÁSA TILOS.
VIGYÁZAT: HA FOLYAMATOS OXIGÉNELLÁTÁSRA
VAN SZÜKSÉG, GONDOSKODJON ARRÓL,HOGY ELEGENDŐ OXIGÉNELLÁTÁS ÉS/VAGY
EGY MÁSODLAGOS OXIGÉNELLÁTÁS MINDIG
RENDELKEZÉSRE ÁLLJON A TERÁPIA IDEJE ALATT.
VIGYÁZAT: NE HAGYJON FÜSTÖLŐT, GYERTYÁT
VAGY NYÍLT LÁNGOT AZ ESZKÖZTŐL 3 MÉTERESTÁVOLSÁGON BELÜL, ÉS SEMMILYEN
GYÚJTÓSZERKEZETET NE HAGYJON 20 CM-NÉL
KÖZELEBB AZ ESZKÖZHÖZ.
VIGYÁZAT: AZ EGYSÉGET JÓL SZELLŐZŐ HELYENTARTSA.
VIGYÁZAT: NE TÁROLJON FOLYÉKONYOXIGÉNT TARTALMAZÓ BERENDEZÉSTSZEKRÉNYBEN, AZ AUTÓCSOMAGTARTÓJÁBAN VAGY MÁS SZŰKHELYEN. NE TAKARJA LE A BERENDEZÉSTTAKARÓKKAL, LEPEDŐKKEL VAGY EGYÉBSZÖVETTEL.
VIGYÁZAT: EBBŐL A TERMÉKBŐL KÉMIAI ANYAGOK– TÖBBEK KÖZT NIKKEL – OLDÓDHATNAK KI, AMI
KALIFORNIA ÁLLAMBAN ISMERTEN RÁKKELTŐ.
TOVÁBBI INFORMÁCIÓÉRT LÁTOGASSON EL A WWW.
P65WARNINGS.CA.GOV OLDALRA.
FIGYELMEZTETÉS: HA KOMOLY ESEMÉNY TÖRTÉNIKEZZEL A KÉSZÜLÉKKEL, A FELHASZNÁLÓNAKAZONNAL ÉRTESÍTENIE KELL AZ ESEMÉNYT ASZOLGÁLTATÓVAL ÉS / VAGY A GYÁRTÓVAL. A
SÚLYOS ESEMÉNYT SÉRÜLÉS, HALÁL VAGY SÉRÜLÉS
/ HALÁL OKAKÉNT KELL MEGHATÁROZNI, HA AZESEMÉNY MEGISMÉTLŐDIK. A FELHASZNÁLÓJELENTHETI AZ ESEMÉNYT ANNAK AZ ORSZÁGNAK
AZ ILLETÉKES HATÓSÁGÁHOZ IS, AHOL AZ ESEMÉNYBEKÖVETKEZETT.
Figyelem! Csak az orvosa utasításainak
megfelelően használja a Sprint/Stroller/Hi Flow
Stroller készülékeket.
Figyelem! Az Amerikai Egyesült Államok szövetségi
törvényei szerint a készülék kizárólag orvos által vagy
orvosi megrendelésre értékesíthető.

214 - HU PN MN236-C4 E | Felhasználói kézikönyv

Figyelem! Az egység folyékony oxigént tartalmaz,
amely rendkívül hideg, közel –184 °C hőmérsékletű. Az
ilyen alacsony hőmérséklet súlyos fagyási sérüléseket
okozhat.
Figyelem! A cseppfolyós és a gáz-halmazállapotú
oxigén ugyan nem gyúlékony, azonban más anyagokéghetőségét felgyorsítja. Ennek kockázata és az
alacsony hőmérsékletű folyékony oxigén megkövetel
bizonyos biztonsági intézkedéseket.
Figyelem! A tűzveszélyes anyagokat tartsa távol aberendezéstől. Az aeroszolos spray-k, olajok és zsírok,beleértve az arckrémeket, és a vazelint, könnyen
meggyulladnak, és oxigén jelenlétében gyorsan
eléghetnek.
Figyelem! A dohányzás az oxigénkanül viselése mellett
az arc megégését okozhatja és halálhoz vezethet.
A kanül levétele és ruhára, ágyneműre, kanapéra vagy
más párnaanyagra helyezése tűzrobbanást okozhat, hacigarettával, hőforrással, szikrával vagy lánggal kerülérintkezésbe.
Ha dohányzik, kérjük: (1) kapcsolja ki a hordozható
eszközt, (2) vegye le a kanült és (3) hagyja el ahelyiséget, ahol az eszköz található.
Figyelem! Abban az esetben, ha az eszköz véletlenül
felborul, óvatosan, de azonnal állítsa vissza az
egységet álló helyzetbe. Ha a folyékony oxigén
szivárogna, azonnal hagyja el a helyiséget és vegye fel
a kapcsolatot orvosával. Ne próbálja meg elmozdítani
az egységet vagy megállítani a folyékony oxigén
szivárgását.
Megjegyzés: Ne érjen az egység jeges részeihez.
Megjegyzés: Ne tárolja vagy használja a hordozható
eszközt a rögzített egységhez csatlakoztatva.
Megjegyzés: Ne engedje, hogy képzetlen személyzet
kezelje vagy működtesse az eszközt.
Megjegyzés: Az eszköz használatát tiltja a Szövetségi
Légiközlekedési Felügyelet az utas- és áruszállítási légi
járatokon.

Rendeltetésszerű használat

A CAIRE Sprint/Stroller/Hi Flow Stroller
kiegészítő oxigén beadására szolgál. A készüléket
nem életmentésre szánják, és nem nyújt semmilyen
betegfigyelési képességet.

Sprint / Stroller / Hi Flow Stroller
Bevezető

VezérlésA Sprint, a Stroller és a Hi Flow Stroller készülékek
a páciens otthonában, kiegészítő oxigén adagolására
szolgálnak, és olyan intézményekben is alkalmazhatóak,
mint a szanatóriumok vagy a szociális ellátó
intézmények. A Sprint/Stroller/Hi Flow Stroller készülék
hordozható oxigénforrást biztosít hosszú időre.

1. Elektronikus folyadékszintmérő
2. Elektronikus folyadékszint-szabályozó
3. Mechanikus folyadékszintmérő
4. Áramlásszabályozó gomb
5. Az oxigénszállító cső csatlakozója
6. Ventilátorszelep
7. Töltési csatlakozók (A gyorscsatlakozású
szelepek [QDV] azonosítását lásd a következő
oldalon)
8. Vállpárnás hordozópánt
9. Kondenzációs betét
10. Kondenzációs kosár

Az eszközt COPD-s vagy a csökkent légzőképességű
betegek használják. Az eszközt orvos írja fel a
betegnek. Az eszköz olyan szolgáltató számára
értékesíthető, amely képzett a Sprint/Stroller/
Hi Flow Stroller készülékek működtetésében és
szervizelésében. Az eszköz használatát a betegnek aszolgáltató tanítja meg.

1

Az eszköz nem alkalmas életmentő kezelésre, ésbetegmonitorozó funkciókkal sem rendelkezik. Egy
esetleges mechanikai meghibásodás esetére ajánlott
egy tartalék oxigénellátó forrás biztosítása. Ezek ahordozható eszközök egy gyors leválasztó szeleppel
is rendelkeznek, amellyel a hordozható eszköz egy
folyékony oxigént tartalmazó tartállyal kapcsolhatóössze az áttöltéséhez. A tartályegységet az Ön
egészségügyi ellátója tölti meg.

2
4

5
8

Ez a használati útmutató a Sprint/Stroller/Hi
Flow Stroller készülékek használatára vonatkozó
utasításokat tartalmazza. A rögzített egységműködtetéséhez olvassa el a hozzá tartozó felhasználói
kézikönyvet.
Megjegyzés: Az egészségügyi szolgáltató segíteni fog akezdeti beállításnál, valamint ismerteti az eszköz megfelelő
kezelését és használat.

5

7

6

Sprint, Stroller és Hi Flow Stroller egységek

10
9
PN MN236-C4 E | Felhasználói kézikönyv HU - 215

Sprint / Stroller / Hi Flow Stroller
A gyorscsatlakozású szelepek (QDV)
azonosítása
Hordozható
csatlakozó

Rögzítettcsatlakozó

3. A töltőcsatlakozót állítsa függőleges helyzetben arögzített egység csatlakozója fölé.

QDV típus

CAIRE Oldaltöltős
oldaltöltő rögzítő

LIFEOX
Felültöltős
rögzítő

4. a. Ha a csatlakozó rányomásos:engedje rá a Sprint/Stroller/Hi Flow Stroller
egységet a rögzített egységre, hogy a csatlakozók
rögzüljenek.

PENOX

TAEMA
Felültöltős
rányomásos
rögzítőPB

b. Ha Penox vagy Life-Ox csatlakozót használ:
addig forgassa a Sprint/Stroller/Hi Flow Stroller
készüléket, amíg a csatlakozók nem rögzülnek.
Óvatosan és biztosan tartva forgassa vissza a
Sprint/Stroller/Hi Flow Stroller készüléket azóramutató járásával megegyezően (), mialattlefelé nyomja.

Kezelési utasítások
A felültöltős Sprint/Stroller/Hi Flow Stroller
készülékek feltöltése
1. A Sprint/Stroller/Hi Flow Stroller készülékek
rögzített és a hordozható egységeken találhatótöltőcsatlakozókat egyaránt tisztítsa meg tiszta,
száraz, szöszmentes törlőronggyal.
Figyelem! Figyeljen arra, hogy a töltőcsatlakozók legyenek szárazak: ellenkező esetbena Sprint/Stroller/Hi Flow Stroller egység anedvesség hatására összefagyhat a rögzített
egységgel.
2. Bizonyosodjon meg arról, hogy a Sprint/
Stroller/Hi Flow Stroller készülékek és a tartályáramlásszabályozó gombjai kikapcsolt (0)helyzetben vannak.

216 - HU PN MN236-C4 E | Felhasználói kézikönyv

Megjegyzés: Folytatás az oldaltöltős Sprint/Stroller/Hi Flow
Stroller készülékek feltöltésének 6. lépésével.

Az oldaltöltős Sprint/Stroller/Hi Flow Stroller
készülékek feltöltése
1. A rögzített és a Sprint/Stroller/Hi Flow Stroller
egységek töltőcsatlakozóit egyaránt tisztítsa meg
tiszta, száraz, szöszmentes törlőronggyal.
VIGYÁZAT: FIGYELJEN ARRA, HOGY A TÖLTŐCSATLAKOZÓK LEGYENEK SZÁRAZAK:ELLENKEZŐ ESETBEN A SPRINT/STROLLER/
HI FLOW STROLLER EGYSÉG A NEDVESSÉGHATÁSÁRA ÖSSZEFAGYHAT A RÖGZÍTETT
EGYSÉGGEL.

Sprint / Stroller / Hi Flow Stroller
2. Bizonyosodjon meg arról, hogy a Sprint/
Stroller/Hi Flow Stroller készülék és a tartályáramlásszabályozó gombjai kikapcsolt (0)helyzetben vannak.

Megjegyzés: A Sprint/Stroller/Hi Flow Stroller készülék
töltését követő 30 másodpercben lehet a legpontosabban
megmérni a készülék folyadékszint-értékeit.

3. Állítsa a Sprint/Stroller/Hi Flow Strollertöltőcsatlakozót álló helyzetben a rögzített
csatlakozóhoz.

4. Forgassa a Sprint/Stroller/Hi Flow Stroller
készüléket az óramutató járásával ellentétesen (amíg a tüske nem csatlakozik a résbe.

)7. Előfordulhat, hogy a csatlakozás körül páraészlelhető. Ez megszokott jelenség: a folyékony
oxigén és az egységet körülvevő meleg levegő közti
jelentős hőmérséklet-különbségnek tudható be.
8. Ha az egység tele van, zárja el a szelepet. Onnan
lehet tudni, hogy az egység tele van, hogy a sziszegő
hang magassága megváltozik, és a töltőcsőnél egy
kevés folyadék szivárog ki.
Megjegyzés: Úgy tölthető fel teljesen a Sprint/Stroller/Hi
Flow Stroller készülék, ha a szelepet majdnem folyamatos
folyadékáramlás mellett zárja el, azonban így elvész egy
kevés folyékony oxigén is.

5. Óvatosan, de határozottan forgassa vissza a Sprint/
Stroller/Hi Flow Stroller készüléket álló helyzetbe.
Az egységek most már csatlakoztatva vannak.

Megjegyzés: Amennyiben a folyadékszivárgás első jelénél
elzárja a szelepet, a Sprint/Stroller/Hi Flow Stroller egység
nem töltődik fel teljesen, azonban megtakarítható bizonyos
folyékonyoxigén-mennyiség.
9. A rányomásos csatlakozók esetében addig tartsa
benyomva a kioldógombot, amíg az egységek
szét nem válnak. Minden más csatlakozóesetében
addig forgassa az egységet, amíg az el nem válik arögzített egységtől.

Megjegyzés: Folytatás 6. lépésnél.
Megjegyzés: A rányomásos típusú csatlakozók esetében,a töltés ideje alatt tartsa mozdulatlanul az egységeket.

Hordozható kioldógomb (csak rányomásos típusnál)

6. Nyissa ki a légtelenítő szelepet a Sprint/Stroller/
Hi Flow Stroller készülék feltöltéséhez. Ilyenkor
oxigén szabadul fel; ne aggódjon. Az Ön által hallottsziszegő hang teljesen helyén való.
PN MN236-C4 E | Felhasználói kézikönyv HU - 217

Sprint / Stroller / Hi Flow Stroller
Figyelem! Csak a szelep elzárása után
ellenőrizze a folyadékszintmérőt.

Kezelési utasítások

Figyelem! Amennyiben használat után azonnalújratölti a Stroller/Sprint egységet, állítsa nullára azáramlásszabályozó szelepet és várjon 10 percet, hogya nyomás kiegyenlítődjön. Ezután üzemeltesse amegszokott módon.
FIGYELMEZTETÉS: NE HASZNÁLJA ASPRINT/STROLLER/HI FLOW STROLLERKÉSZÜLÉKET A RÖGZÍTETT EGYSÉGHEZCSATLAKOZTATVA.
VIGYÁZAT: HA A SPRINT/STROLLER/HI FLOWSTROLLER NEM VÁLIK SZÉT KÖNNYEN, NE
ERŐLTESSE. ELŐFORDULHAT, HOGY AZ EGYSÉGEKÖSSZEFAGYTAK. HAGYJA AZ EGYSÉGEKETÖSSZEKAPCSOLÓDVA, ÉS VÁRJON, AMÍG FEL NEM
MELEGEDNEK – EZUTÁN KÖNNYEN SZÉTVÁLNAK.
NE ÉRJEN HOZZÁ EGYIK FAGYOTT RÉSZHEZ SEM.
FIGYELMEZTETÉS: HA A SZIVÁRGÁS MÉRTÉKE MÁR
ELÉRI AZT A PONTOT, HOGY FOLYADÉK ÁRAMLIKKI AZ ESZKÖZBŐL, HAGYJA EL A HELYISÉGET,ÉS AZONNAL VEGYE FEL A KAPCSOLATOTEGÉSZSÉGÜGYI ELLÁTÓJÁVAL.
FIGYELMEZTETÉS: A CSEPPFOLYÓS VAGY HIDEG,
GÁZ HALMAZÁLLAPOTÚ OXIGÉN FAGYÁST OKOZHATA BŐRÖN. AZ OXIGÉN NÖVELI A TŰZVESZÉLYT.
Figyelem! Ha az egységek szétválasztása
során folyadék szivárogna a rögzített
egységből vagy a Sprint/Stroller/Hi Flow
Stroller készülékből, tegye félre a Sprint/
Stroller/Hi Flow Stroller készüléket,gondoskodjon arról, hogy függőlegesen
maradjon, majd hagyja el a szobát, ésazonnal vegye fel a kapcsolatot egészségügyi
ellátójával.
Megjegyzés: A berendezés szolgáltatójának: A következő
oxigénbeviteli kiegészítők használata ajánlott az Sprint /Stroller / Hi Flow Stroller:• Orrkanül 7,1 láb (2 m) csővel (maximális LPM: 6 LPM):
CAIRE cikkszám 5408-SEQ• Tűzjelzés: CAIRE cikkszám: 21126636
Bármely kanül használatához tűzjelzés javasolt.
• A CAIRE tűzjelzést kínál, amelyet a hordozható
oxigénnel együtt kell használni. A tűzjelzés hőbiztosíték,amely megállítja a gázáramot abban az esetben, ha az
alsó irányú kanül vagy az oxigéncső meggyullad és atűzjelző égésbe kerül. Az orrkanültel vagy az oxigéncsővel
összhangban helyezkedik el a beteg és az Sprint / Stroller/ Hi Flow Stroller oxigénkivezetése között. A tűzjelzés
megfelelő használatához mindig olvassa el a gyártó utasításait (minden tűzjelző készlethez mellékelve).
• Az ajánlott kiegészítőkről további információk elérhetők awww.caireinc.com weboldalon.
218 - HU PN MN236-C4 E | Felhasználói kézikönyv(Kanült nem tartalmaz a csomag.)

1. Nyomja a kanül légzőcsövét erősen az
oxigénszállító cső csatlakozójára.
2. Állítsa be megfelelő helyzetbe a légzőcső kanüljét
úgy, hogy az oxigén kényelmesen belélegezhető
legyen.
3. Addig forgassa a Sprint/Stroller/HI Flow Stroller
egység tetején található áramlásszabályozó gombotaz óramutató járásával megegyező irányban, amíga gomb ellenőrzőablakán láthatóvá nem válik azelőírt áramlási sebesség (számjegy) és a gomb egy
határozott állásba kattan.
Figyelem! Ha a gombot az áramlási
sebességek közé állítja be, az előíráson kívüli
oxigénáramlást eredményez.
4. Most már oxigént kell kapnia. Az oxigénadagolás
elzárásához, fordítsa el az áramlásszabályozó
gombot az óramutató járásával ellentétes irányba,
a kikapcsolt (0) helyzetbe.
Figyelem! Az áramlásszabályzó gombot
mindig fordítsa kikapcsolt (0) helyzetbe, ha
nem használja az eszközt.

Sprint / Stroller / Hi Flow Stroller
5. Használja az alábbi táblázatot útmutatásként a
Sprint/Stroller/Hi Flow Stroller készülék működési
időtartamának meghatározásához:Az áramlásszabályozó szelep beállítása
SprintStroller
Hi Flow
Stroller
Névleges
Névleges
Az
SprintStroller
Hi Flowáramlás�Stroller
szabályozó
szelepállásaKikapcsolva
30:00
60:00
71:00
0,25
18:00
30:00
Nem
alkalmazható
0,5
12:12
23:00
32:00
0,75
9:18
17:24
Nem
alkalmazható
1
7:30
14:06
16:00
1,5
5:30
10:12
Nem
alkalmazható
2
4:18
8:00
8:00
2,5
3:36
6:36
6:20
3
3:06
5:36
5:20
4
2:24
4:24
4:00
5
2:00
3:36
Nem
alkalmazható
6
1:42
3:00
2:40
8
Nem
Nem
2:00
alkalmazható alkalmazható
10
Nem
Nem
1:40
alkalmazható alkalmazható
12
Nem
Nem
1:20
alkalmazható alkalmazható
15
Nem
Nem
1:00
alkalmazható alkalmazható
ModellMegjegyzés: Az idő órákban és percekben van
kifejezve (formátum: 00:00).
Megjegyzés: A „névleges” idők ideális körülményekre
vonatkoznak, úgymint maximális feltöltődöttség,
pontos áramlásértékek, megfelelő veszteségszint,
nem mozgatott egység stb.
Ezek az idők a várható maximális idők.
Megjegyzés: Az Ön egyéni eredményei eltérhetnekettől.

• Gen 4 mérőműszer: Nyomja meg a zöld
kezelőgombot a mérőműszer elején. A LED-ív jelzi afolyadékszintet.

Figyelem! Ha a jelzőrúdon csak az utolsó
szakasz világít, az azt jelzi, hogy a Sprint/
Stroller/HI Flow Stroller kiürült.
Ha az alacsony akkufeszültséget jelző lámpa
a gomb benyomásával kigyullad, értesítseegészségügyi ellátóját.
Megjegyzés: A Sprint/Stroller/HI Flow Stroller készülék
az alacsony akkufeszültséget jelző lámpa kigyúlása
után is szolgáltat oxigént addig, amíg az egységben
rendelkezésre áll folyékony oxigén.
6b. A folyékony oxigén szintjének ellenőrzése az
egységben mechanikus folyadékszintmérővel:• Függessze fel az egységet a szövetből készült
pánttal, közvetlenül a fekete szintjelző rugó
skálakamrája fölé.
• Hagyja, hogy az egység stabilizálódjon, (ne rugózzon
fel-le), majd olvassa le a megjelenő színes skálát,hogy meghatározza a hozzávetőleges folyékony
oxigénszintet.

6a. A folyékony oxigén szintjének ellenőrzése az
egységben elektronikus folyadékszint-mérővel:• Gen 3 mérőműszer: Tartsa benyomva az egység
tetején található nyomógombot (folyadékszintkapcsoló) legalább két másodpercig. A jelzőrúd
kivilágított része jelzi a folyadékszintet.

PN MN236-C4 E | Felhasználói kézikönyv HU - 219

Sprint / Stroller / Hi Flow Stroller
Megjegyzés: Amikor az egység üres, csak a skála vörösre
színezett sávja látható. Ilyen esetben használat előtt fel
kell tölteni az egységet.
7. Adott környezeti körülmények között, folyamatos
használat mellett, és különösen akkor, ha ahordozható részt nem mozgatják, nagyobb
mennyiségű fagy képződhet a porlasztótekercsen
és a burkolaton. Ez úgy távolítható el, hogy
megkopogtatja az egységet és/vagy letörli aburkolatra lerakodott fagyréteget.

A kondenzációs betét karbantartása
Miután az egység kiürült és szoba-hőmérsékletűre
melegedett, távolítsa el a nedvességet a kondenzációs
kosárból:
Kemény burkolat esetén• Billentse oldalára az egységet. Fogja meg
a gombot a kondenzációs kosár közepén(a készülék alján), és egyenesen húzza ki.
Ezáltal kiszabadul a kosár.
Megjegyzés: Erős szennyeződés esetén a betét
mosógépben tisztítható (fehérítővel fertőtlenítendő).
• Visszahelyezéskor igazítsa a kosarat az egység
alján található nyílás közepe fölé, nyomja be egy
kattanásig a gombot, hogy a kosár rögzüljön,
majd még egy kattanásig, hogy a helyére
kerüljön.

beszívott nedvességet. Mielőtt visszahelyezné akosárba, engedje teljesen megszáradni a betétet.
Megjegyzés: Erős szennyeződés esetén a betét
mosógépben tisztítható (fehérítővel fertőtlenítendő).
• Visszahelyezéskor egyszerűen helyezze a betétet
az alsó lapra, és húzza vissza az alsó cipzárt.
Megjegyzés: Ha a gombot nem lehet benyomni és a kosár
nem akad be, tartsa meg a kosarat és addig húzza agombot, amíg nyitott pozícióban rögzül. Ezután helyezze
vissza a kosarat az előzőleg bemutatott módon.

A fagyás csökkentése
Adott környezeti körülmények között, folyamatos
használat mellett, és különösen akkor, ha a hordozható
részt nem mozgatják, nagyobb mennyiségű fagyképződhet a porlasztótekercsen és a burkolaton. Ez úgy
távolítható el, hogy megkopogtatja az egységet és/vagy
letörli a burkolatra lerakodott fagyréteget.
FIGYELMEZTETÉS: A TÚLZOTTNEDVESSÉG/JÉG FELHALMOZÓDÁSÁNAKLEHETŐSÉGE MIATT AJÁNLOTT AHI FLOW STROLLER KÉSZÜLÉKET A
FELHASZNÁLÁS UTÁN 1 ÓRÁN KERESZTÜL
ÜRESJÁRATBAN HAGYNI A NEDVESSÉGFELHALMOZÓDÁSÁNAK CSÖKKENTÉSEÉRDEKÉBEN.

Zárva
Csatlakoztatva
NyitvaGomb Kosár

Megjegyzés: Ha a gombot nem lehet benyomni és a kosár
nem akad be, tartsa meg a kosarat és addig húzza agombot, amíg nyitott pozícióban rögzül. Ezután helyezze
vissza a kosarat az előzőleg bemutatott módon.
Puha burkolat esetén• Húzza el az alsó cipzárt és húzza vissza az alsó
burkolatot.
• Vegye ki a betétet a kosárból és csavarja ki a
220 - HU PN MN236-C4 E | Felhasználói kézikönyv

A Hi Flow Stroller készülékkel járó nagyobbáramlási sebesség miatt az egységben páralecsapódásés kismértékű fagyás lép fel, különösen a 10 l/perc vagy ennél magasabb áramlási értékeknél.
Ajánlott az egységet használat után legalább 1 óránát pihentetni, hogy csökkentsük a folyamatos
használat miatt kialakuló jelentős nedvesség- és/vagy
jégfelhalmozódás lehetőségét.

Sprint / Stroller / Hi Flow Stroller
Hibaelhárítás
Az alábbi információk segítséget nyújtanak a Sprint/Stroller/Hi Flow Stroller készülék használata során tapasztalt
egyszerű működési problémák elhárításához és megoldásához.
Probléma
A hordozható eszköz
sziszegő hangot ad.

A hordozható készülékben
az áramlás használat
közben leáll.

A hordozható eszköz nemtöltődik.
A tartály ventilátorszelepe
nem záródik be teljesen atöltési folyamat végén.

A hordozható rész nem
választható le könnyen atartályról a feltöltés után.

Megoldás• Sziszegő hang jelentkezhet a helyes üzemi nyomás tartályon belüli fenntartása során. A
sziszegő hang nagy valószínűséggel a tartály feltöltése után vagy a hordozható készülék
helyzetének megváltoztatásakor jelentkezik. A sziszegő hang a feltöltés után körülbelül
10 percig hallatszik. Ezenkívül akkor is előfordulhat, ha az áramlásszabályozó szelep
alacsonyra van állítva.
• Ha a hordozható készüléket nem a megfelelő helyzetbe állították, helyezze vissza állóhelyzetbe, és hagyjon néhány percet az egység stabilizálódására.
• Ellenőrizze, hogy a kanül szorosan rögzítve van-e az oxigén kimenetéhez.
• Ellenőrizze, hogy a kanül nincs-e megtörve.
• Ellenőrizze a tartalom mutatóját/szintjelzőjét, és szükség esetén töltse fel a hordozható
készüléket.
• Bizonyosodjon meg arról, hogy az áramlásszabályzó gomb kikapcsolt ("0") helyzetben van.
• Ellenőrizze, hogy van-e oxigén a tartályban.
• Ellenőrizze, hogy a hordozható eszköz és a tartály töltőcsatlakozója teljesen illeszkedik-e atöltés folyamata során.
• Ha a ventilátorszelep nem záródik be, valamint a sziszegő hang és az oxigén párafelhő
továbbra is jelentkezik, óvatosan vegye le a hordozható részt a tartályon lévő kioldógomb
lenyomásával. A ventilálás a hordozható rész aljából néhány percen belül befejeződik.
Hagyja, hogy az egység felmelegedjen, hogy be tudja zárni a ventilátorszelepet. A
hordozható készüléknek akár 60 percre is szüksége lehet ahhoz, hogy visszaállítsa a
megfelelő nyomást a helyes oxigénáramláshoz. Szükség esetén használjon tartalék
oxigénforrást, például a tartályhoz csatlakoztatott áramlásellenőrző szelepet.
• A hordozható rész és a tartály töltőcsatlakozói befagyhattak.
• NE ERŐLTESSE! Várjon néhány percet, hogy a fagyott részek felmelegedjenek, majd
válassza le a hordozható részt, ha a jég felolvadt. Az egységek egymáshoz fagyásának
megakadályozása érdekében a feltöltés előtt mindig törölje le egy száraz törlőronggyal
a tartályon a töltőcsatlakozó csatlakozódugóját, valamint a hordozható részen atöltőcsatlakozó csatlakozóhüvelyét.

PN MN236-C4 E | Felhasználói kézikönyv HU - 221

Sprint / Stroller / Hi Flow Stroller
Szabványos tisztításVIGYÁZAT: AZ EGYSÉG KIZÁRÓLAG AZUTÁN
TISZTÍTHATÓ, HOGY KIÜRÜLT.
• Enyhe mosogatószeres vízzel tisztítsa.
• A tisztítóoldatot közvetlenül egy szöszmentestörlőrongyra vigye fel. Jóváhagyotttisztítószerek többek között a HydroPure és a
HydroKlean. Ne fújjon tisztítószert közvetlenül
a Sprint/Stroller/Hi Flow Stroller készülékre.
• Törölje tisztára a külső felszínt a szöszmentestörlőronggyal.
Figyelem! Ne használjon magashőmérsékleten és magas nyomásonműködő tisztítóberendezést ezen egységek
tisztításához.
• Ne juttasson be tisztítószert a belső részekre
vagy szelepekre.
• Használat előtt hagyja az egységet alaposanmegszáradni.
Megjegyzés: Megjegyzés az egészségügyi szolgáltató
számára – az újrafeldolgozási eljárásokhoz lásd avonatkozó szervizelési kézikönyvet.

WEEE és RoHS irányelvek

Ez a szimbólum a berendezés tulajdonosátarra figyelmezteti, hogy a készüléket annakélettartama végén az elektromos és elektronikus
készülékek hulladékainak kezeléséről szóló irányelv(WEEE) szerint egy újrafeldolgozó létesítménybe kelljuttatnia.
A termékeink megfelelnek a veszélyes anyagokkorlátozására (RoHS) vonatkozó irányelvnek.
Legfeljebb nyomokban tartalmazhatnak ólmot ésegyéb veszélyes anyagot.

Megjegyzés: A légköri nyomástartomány 700 hPa és
1060 hPa (10 000 láb és –1000 láb, azaz 3000 m és–300 m) közötti lehet.

KarbantartásAz Ön szolgáltatója felelős minden olyan
karbantartásért, amelyre ezen eszköz műszaki
kézikönyve alapján szükség lehet. A karbantartásielőírásokért hívja szolgáltatóját.
A kondenzációs betét az egyetlen része a készüléknek,
amely a felhasználó által karbantartható. A készülék
egyetlen más részét sem tarthatja karban a felhasználó.
A várt üzemi élettartam legalább öt év.
A töltőcsatlakozókat tisztítsa meg a rögzített és ahordozható egységeken egyaránt egy tiszta, száraz,szöszmentes törlőronggyal a fagyás és a berendezéslehetséges meghibásodásának elkerülése érdekében.
Megjegyzés: Az esetlegesen szükséges karbantartási
feladatokat képzett szakember vagy szolgáltató végezze.

TartozékokÁrtalmatlanítás
A megfelelő ártalmatlanítás érdekében mindig adja
vissza a Sprint/Stroller/Hi Flow Stroller készüléket
az otthoni ellátásáért felelős szakembernek, atartozékokat is beleértve. Az akkumulátor megfelelő
ártalmatlanítására vonatkozó utasításokkal kapcsolatban
a helyi önkormányzattal is kapcsolatba léphet.

Szállítás és tárolás
Az eszközt álló helyzetben, jól szellőző helyen
kell tárolni. Az eszközt tilos az oldalára fektetni.
Páratartalom: legfeljebb 95%-os (nem lecsapódó).
Hőmérséklet-tartomány: –40 °C és 70 °C (–40 °F és
158 °F) között.
Üzemi hőmérséklet-tartomány: –10°C és 40 °C (14 °F
és 104 °F) között. Páratartalom-tartomány: 30% és
75% között (nem lecsapódó).

222 - HU PN MN236-C4 E | Felhasználói kézikönyv

Hordozókocsi PN 20765565

Sprint / Stroller / Hi Flow Stroller
BiztonságVIGYÁZAT: HORDOZHATÓ RÁDIÓFREKVENCIÁS KOMMUNIKÁCIÓS ESZKÖZT (BELEÉRTVE OLYANPERIFÉRIÁS ESZKÖZÖKET, MINT AZ ANTENNAKÁBELEK ÉS A KÜLSŐ ANTENNÁK) NEM SZABAD
30 CM-NÉL KÖZELEBB HASZNÁLNI A SPRINT/STROLLER/HI FLOW STROLLER KÉSZÜLÉKEK BÁRMELYRÉSZÉHEZ, BELEÉRTVE A GYÁRTÓ ÁLTAL MEGHATÁROZOTT KÁBELEKET. ELLENKEZŐ ESETBEN AZ A
BERENDEZÉS TELJESÍTMÉNYÉNEK ROMLÁSÁT EREDMÉNYEZHETI.
VIGYÁZAT: A BERENDEZÉS GYÁRTÓJA ÁLTAL MEGHATÁROZOTTAKTÓL ÉS BIZTOSÍTOTTAKTÓL ELTÉRŐTARTOZÉKOK, ÁTALAKÍTÓK ÉS KÁBELEK HASZNÁLATA A BERENDEZÉS HELYTELEN MŰKÖDÉSÉT ÉSELEKTROMÁGNESES KIBOCSÁTÁSÁNAK MEGNÖVEKEDÉSÉT EREDMÉNYEZHETI.
VIGYÁZAT: A KÉSZÜLÉKET NE HASZNÁLJA MÁS BERENDEZÉS KÖZVETLEN KÖZELÉBEN VAGY ARRARÁHELYEZVE, MIVEL AZ HELYTELEN MŰKÖDÉST EREDMÉNYEZHET. HA ERRE MÉGIS SZÜKSÉG LENNE, AKKORELLENŐRIZNI KELL, HOGY A KÉSZÜLÉK ILYEN ELRENDEZÉSBEN MEGFELELŐEN MŰKÖDIK-E.
Figyelem! Az elektromos gyógyászati berendezések az elektromágneses összeférhetőség (EMC)
tekintetében speciális óvintézkedéseket igényelnek, és az ebben a kézikönyvben megadott elektromágnesesösszeférhetőségre vonatkozó információknak megfelelően kell azokat telepíteni és üzembe helyezni.
Figyelem! A hordozható eszköz és a mobil rádiófrekvenciás (RF) kommunikációs berendezések
befolyásolhatják az egészségügyi célú elektromos berendezések működését.
1. táblázatÚtmutató és a gyártó nyilatkozata – elektromágneses kibocsátás
A Sprint/Stroller/Hi Flow Stroller készülékek a következőkben részletezett elektromágneses körülmények
között használhatóak. A Sprint/Stroller/Hi Flow Stroller készülékek vevőjének vagy felhasználójának
biztosítania kell, hogy ilyen környezetben használják.
Kibocsátási vizsgálat
RF-kibocsátás

Megfelelőség
1. csoport

CISPR 11

Elektromágneses környezet – útmutatás
A Sprint/Stroller/Hi Flow Stroller készülék csak a belső
működéshez használ rádiófrekvenciás energiát.
Ezért a rádiófrekvenciás kibocsátása nagyon alacsony, és
nem valószínű, hogy interferenciát okoz a közelében lévőelektronikus berendezésekben.

RF-kibocsátás CISPR 11

B osztály

Harmonikus kibocsátások
IEC 61000-3-2

Nem alkalmazható

Feszültségingadozások/feszültségrezgések
IEC 61000-3-3

Nem alkalmazható

A Sprint/Stroller/Hi Flow Stroller minden létesítményben
használható, beleértve a lakókörnyezetet, valamint alakossági célokat szolgáló épületeket ellátó alacsony
feszültségű, nyilvános hálózatra közvetlenül csatlakozó
környezeteket is.

PN MN236-C4 E | Felhasználói kézikönyv HU - 223

Sprint / Stroller / Hi Flow Stroller
2. táblázat*: A hordozható és mobil rádiófrekvenciás kommunikációs berendezések és az
Sprint/Stroller/Hi Flow Stroller készülékek közötti ajánlott elkülönítési távolságok
A Sprint/Stroller/Hi Flow Stroller készüléket olyan elektromágneses környezetben való használatra tervezték, amelyben a sugárzott
rádiófrekvenciás zavarok szabályozva vannak. A Sprint/Stroller/Hi Flow Stroller készülék vásárlója vagy felhasználója úgy segíthet
az elektromágneses interferencia megelőzésében, hogy betartja a hordozható és mobil rádiófrekvenciás távközlési eszközök(jeladók) és a Sprint/Stroller/Hi Flow Stroller készülék közötti alábbi ajánlott elkülönítési távolságot, a kommunikációs berendezés
teljes leadott teljesítményének megfelelően.
A jeladó névleges
Elkülönítési távolság az adó frekvenciájától függően
maximális leadottmteljesítménye
80 MHz és 800 MHz
800 MHz és 2,5 GHz
150 kHz és 80 MHz

d = 1,2√P

d = 1,2√P

d = 2,3√PW
0,010,12 m0,12 m0,23 m0,10,38 m0,38 m0,73 m
1
1,2 m
1,2 m
2,3 m
10
3,8 m
3,8 m
7,3 m
100
12 m
12 m
23 m
A fent fel nem sorolt névleges kimeneti teljesítményű adók esetében a méterben (m) kifejezett „d” javasolt izolációs távolságot az
adó frekvenciájára vonatkozó egyenlettel lehet kiszámítani, ahol „P” az adó gyártója által megadott névleges kimeneti teljesítmény
maximális értéke wattban (W).
1. MEGJEGYZÉS: 80 MHz és 800 MHz között a nagyobb frekvenciatartományhoz tartozó távolság érvényes.
2. MEGJEGYZÉS: Ezek az irányelvek nem minden esetben használhatók. Az elektromágneses hullámok terjedését befolyásolja azépítmények, tárgyak és emberek elnyelő és visszaverő hatása.

* Ez a szabványos követelményeket tartalmazó táblázat az olyan berendezésekre vonatkozik, amelyeket
meghatározott mérőszintek és meghatározott frekvenciatartományok között vizsgáltak, és amelyeket aszabályzatoknak megfelelőnek találtak.

3. táblázat: Útmutató és a gyártó nyilatkozata – elektromágneses zavartűrés
A Sprint/Stroller/Hi Flow Stroller készülékek a következőkben részletezett elektromágneses körülmények között használhatóak. A
Sprint/Stroller/Hi Flow Stroller készülékek vevőjének vagy felhasználójának biztosítania kell, hogy ilyen környezetben használják.
Zavartűrési teszt
IEC 60601 tesztszint
Megfelelőségi szint
Elektromágneses környezet – útmutatás± 8 kV érintkezéssel
Elektrosztatikus kisülés ± 8 kV érintkezéssel
A padló legyen fa, beton vagy kerámialap. Ha(ESD)a padló műanyaggal van burkolva, a relatív± 2 kV, ± 4 kV,± 2 kV, ± 4 kV,
IEC 61000-4-2±8 kV, ± 15 kV levegőn át±8 kV, ± 15 kV levegőn át páratartalom legalább 30% legyen.**Gyors elektromos
Tranziens/burst
jelenségIEC 610004-4±2 kV teljesítményű
Tápvezetékek±1 kV bemeneti/kimeneti
vezetékeknél

Nem alkalmazható
egyenárammal működő
készülék esetén
Nem alkalmazható
Nincs adat bemeneti/
kimeneti vezetékek
eseténNem alkalmazható
egyenárammal működő
készülék esetén±1 kV vezeték–vezeték
esetén±2 kV vezeték–föld esetén<5% UT (>95%-os UT sülFeszültségletörések,lyedés UT) 0,5 ciklus erejéig
rövid idejű feszül40% UT (60%-os UT sülNem alkalmazható
tségkimaradás és
lyedés) 5 ciklus erejéig
egyenárammal működő
feszültségváltozások a 70% UT (30%-os UT sülkészülék eseténtápellátó vonalakon
lyedés) 25 ciklus erejéig<5% UT (>95%-os UT sülIEC 61000-4-11
lyedés) 5 mp erejéig
Hálózati frekvenciás(50/60 Hz) mágne3A/m
3A/m
ses tér
50/60 Hz50/60 HzIEC 61000-4-8
Túlfeszültség

IEC 61000-4-5

Nem alkalmazható

Nem alkalmazható

Nem alkalmazható

A váltakozó áram által generált mágnesesmezőnek olyan szintűnek kell lennie, amely
megfelel a tipikus kereskedelmi vagy kórházi
környezetben elvárható értéknek.

Megjegyzés: UT a váltakozó feszültségű táphálózat feszültsége a tesztszint alkalmazása előtt.

** Ez a kijelentés azt jelzi, hogy a szükséges vizsgálatokat ellenőrzött környezetben végezték el, és a Sprint/
Stroller/Hi Flow Stroller készülékek megfelelnek a szabályozásoknak.
224 - HU PN MN236-C4 E | Felhasználói kézikönyv

Sprint / Stroller / Hi Flow Stroller
4. táblázatÚtmutatás és a gyártó nyilatkozata – ME berendezések és ME rendszerek zavartűrése

Útmutatás és a gyártó nyilatkozata – zavartűrés
A Sprint/Stroller/Hi Flow Stroller készülékek a következőkben részletezett elektromágneses körülmények
között használhatóak. A Sprint/Stroller/Hi Flow Stroller készülék vevőjének vagy felhasználójának
biztosítania kell, hogy ilyen környezetben használják.
Zavartűrési vizsgálat

IEC 60601-es mérőszint

Megfelelőségi szint

Elektromágneses környezet – útmutatás

Vezetett
rádiófrekvencia
IEC 61000-4-6

3 Vrms6 Vrms (ISM sávokban)
150 kHz-től 80 MHz-ig

Nem alkalmazható

A hordozható és mobil rádiófrekvenciás
kommunikációs berendezések nem használhatók
a Sprint/Stroller/Hi Flow Stroller készülékekhez,
illetve azok részeihez (beleértve a kábeleket) az adó
frekvenciájára alkalmazható egyenlettel számított,javasolt izolációs távolságon belül.

Akkumulátoros eszköz,
nincs jelbemeneti port
(SIP)/jelkimeneti port
(SOP)

Javasolt izolációs távolság
d = 1,2 √P
Sugárzott
rádiófrekvencia
IEC 61000-4-3

80 MHz-től 2,7 GHz-ig

d = 1,2 √P
d = 2,3 √Pahol P az adó gyártó által megadott legnagyobb
kimeneti teljesítménye Wattban (W), és d a javasolt
izolációs távolság méterben (m).

10 V/m80 MHz – 2.7 GHz
80% AM 1 kHz-en

A helyhez kötött rádiófrekvenciás adók
elektromágneses helyszínfelmérése sorána
megállapított elektromágneses térerősségének
minden frekvenciatartományban kisebbnek kelllennie a határértéknélb.
A következő szimbólummal ellátott berendezések
közelében interferencia léphet fel:Vizsgálati
frekvencia (MHz)

Sáva)
(MHz)

A készülék javításaa)Modulációb)

385

380-390TETRA 400:450

430–470

GMRS 460, FRS 460

Impulzusmodulációb)
18 Hz
FMC)
±5 kHz eltérés
1 kHz szinusz jel

Maximális
teljesítmény(W)

Távolság(m)

Zavartűrési
vizsgálati szint
(V/m)

1,8

0,3

27

2

0,3

28

710
745
780
810
870
930
1720
1845
1970

704–787

LTE sáv 13, 17

Impulzusmodulációb)
217 Hz

0,2

0,3

9

800–960

GSM 800/900, TETRA 800,
IDEN 820, CDMA 850,
LTE sáv 5

Impulzusmodulációb)
18 Hz

2

0,3

28

1700–
1900

GSM 1800; CDMA 1900;
GSM 1900; DECT; LTE
sáv 1, 3, 4, 25; UMTS

Impulzusmodulációb)
217 Hz

2

0,3

28

2450

2400–
2570

Impulzusmodulációb)
217 Hz

2

0,3

28

5240
5500
5785

Bluetooth,
WLAN 802.11 b/g/n,
RFID 2450, LTE sáv 7

5100–
5800

WLAN 802.11 a/n

Impulzusmodulációb)
217 Hz

0,2

0,3

9

Megjegyzés: Ha szükséges a ZAVARTŰRÉSI VIZSGÁLATI SZINT eléréséhez, az átviteli antenna és a Sprint/Stroller/Hi
Flow Stroller készülék közötti távolság 1 m-re csökkenthető. Az 1 m-es vizsgálati távolságot az IEC 61000-4-3 engedélyezi.
Egyes szolgáltatások esetében csak a felmenő irányú frekvenciák szerepelnek.
A hordozót 50%-os működési ciklusú négyszöghullámú jellel kell modulálni.
Az FM moduláció alternatívájaként lehet 18 Hz-en 50% impulzusmodulációt használni, mert habár nem ad tényleges modulációt, ez lenne alegrosszabb eset.
abc

CAIRE i CAIRE Inc. są zastrzeżonymi znakami towarowymi firmy CAIRE Inc. Aby uzyskać pełną listę znaków towarowych, odwiedź naszą stronę internetową poniżej.Znaki towarowe: www.
careinc.com/corporate/trademarks.
Copyright © 2023 CAIRE Inc. A CAIRE Inc. fenntartja a jogot termékei megszüntetésére vagy termékei árának, anyagainak, berendezéseinek, minőségének, leírásainak, specifikációinak
és/vagy folyamatainak bármikor, előzetes értesítés és kötelezettség és következmény vállalása nélkül való módosítására. Minden itt kifejezetten nem meghatározott jogot adott esetben
fenntartunk.

PN MN236-C4 E | Felhasználói kézikönyv HU - 225

Sprint / Stroller / Hi Flow Stroller
NOTES:

226 - US PN MN236-C4 E | User ManualSprint / Stroller / Hi Flow Stroller
NOTES:PN MN236-C4 E | User Manual US - 227

www.c a ire inc .c om

CAIRE Inc.
2200 Airport Industrial Dr., Ste. 500
Ball Ground, GA 30107 U.S.A.
Medical Product Service GmbH
Borngasse 20
35619 Braunfels, Germany
Email: info@mps-gmbh.eu
Worldwide: www.caireinc.com
Accumed Sagl
Viale Serfontana, 106834 - Morbio Inferiore
Switzerland
Medstar Importacaoe Exportacao Ltda
Rua Valencio Soares Rodrigues, 89, Sala 01 Centro
Vargem Grande Paulista/SP
Brasil06730-00
Tel/Fax: (55)(11)5535-0989
Email: medstar@medstar.com.br

CAIRE and CAIRE Inc. are registered trademarks of CAIRE Inc. Please visit our website
below for a full listing of trademarks. Trademarks: www.caireinc.com/corporate/trademarks/.
Copyright © 2023 CAIRE Inc.
CAIRE Inc. reserves the right to discontinue
its products, or change the prices, materials, equipment, quality, descriptions, specifications
and/or, processes to its products at any time without prior notice and with no further
obligation or consequence. All rights not expressly stated herein are reserved by us, asapplicable.

|MN236-C4-E)~
10 Aug 2023

MN236-C4 E