Medical AIO Solution

MATE-2205
22" Fanless Core i7 Medical AIOFeatures● Intel® 8th/9th Generation Core™ i7/ i5 / i3 Processor● Intel® H310 Chipset● Supports Dual Channel DDR4 2666 SODIMM up to 64GB
● 22" Full HD High Contrast MVA LCD● Capacitive Multi-Touch Screen● Two Gigabit Ethernet● High Speed USB 3.0 Ports● RS-232 x 2● PCI-Express [x4] x 1 (optional)● +12V DC Input

Power on / off

LCD on/off
Reading Light

LCD Brightness
Speaker Volume

Specifications
Main Specifications

Touch Screen on/offProcessor

System MemoryChipset
OS Support
Storage Disk DriveUSB2.0 x 2

Security
Wireless Communication

Slim
DVD-RW Drive
Smart CardReader

Speaker
Function Key
Power Requirement

Intel® 9th Generation Core i7-9700TE 8 Cores 3.8GHz
Intel® 9th Generation Core i5-9500TE 6 Cores 3.6GHz
Intel® 8th Generation Core i7-8700T 6 Core 4.0GHz
Intel® 8th Generation Core i5-8500T 6 Cores 3.5GHz
Supports Dual Channel DDR4 2666 SODIMM
up to 64GBIntel® H310
Windows® 10, Linux ®
2.5" SATA Hard Disk Drive/SSD x 1
M.2 2242 SSD x 1
Slim DVD-RW Drive x 1 (optional)
Trusted Platform Module 2.0, RFID Reader (optional)
Smart Card Reader x 1 (optional)
802.11 ac/a/b/g/n(optional),
Bluetooth 4.1 / 5(optional)
5W x 2
Speaker Volume Up/Down, LCD Brightness Up/Down
LCD On/Off, Touch Screen On/Off, Reading Light On/
Off
DC 12V

Display
Size / TypeMax. Resolution
Luminance(cd/m2)(TYP)
Viewing Angle
Contrast Ratio
Back Light Life Time
Touch Screen

I/O

Gigabit LAN x 2
USB 3.0 x 4
COM Port x 1

Mic inDC Input

22" MVA LCD
1920 x 1080
250 nits
178˚(H)/178˚(V)3000:1
30,000 Hours
5-Wire Resistive / Capacitive Multi-Touch

PCI Express[x4](optional)

USB
Serial PortUSB 3.0 x 4, USB 2.0 x 2
RS-232 x 2Ethernet

1.5KV Isolated Gigabit LAN x 2
Display Port 1.2 x1, HDMI 1.4b x 1(Supports resolution up to 3840 x 2160)
Mic-in, Line-out

Video Out
Audio

Mechanical and Environmental
DP x 1COM Port x 1

HDMI x 1

Line out

Reading Light

Operating Temperature
Storage Temperature
Mounting VESA
Degree of Protection
Dimension
Package Size
Gross Weight
Net Weight

Certifications

87

www.onyx-healthcare.com 2020.9.24

0°C ~ 35°C(32°F ~ 95°F)-20°C ~ 60°C(-4°F ~ 140°F)
75/100 mm
IP65 in the front ; IP54 in the back546 x 352 x 66mm
750 x 220 x 490 mm
12 kg (26.45 lb)
8 kg (17.63 lb)
CE: EN 60601-1-2:2015(V4.0),
EN 60601-1:2006/A1:2013(V3.1)
FCC: Part 18 Class B
UL: ANSI/AAMI ES60601-1:2012 (V3.1)
cUL: CAN/CSA-C22.2 No. 60601-1:2014 (V3.1)※ All product specifications are subject to change without notice8822" Fanless Core i7 Medical AIO
Dimension / Unit: mm

6610075351.4

546

Ordering Information» MATE-2205-P1-A1-0010
Medical Panel PC,22” LCD,i7-8700T,DC 12V,Capacitive Touch» MATE-2205-P2-A1-0010
Medical Panel PC,22” LCD,i7-9700TE,DC 12V,Capacitive Touch» MATE-2205-P3-A1-0010
Medical Panel PC,22” LCD,i5-8500T,DC 12V,Capacitive Touch» MATE-2205-P4-A1-0010
Medical Panel PC,22” LCD,i5-9500TE,DC 12V,Capacitive Touch

Optional Accessories» 1255301205
Medical Power Adapter, 120W, 12V» OPM-R03X-A1
PCI Express[x1] Riser Kit» OPM-R04X-A1
PCI Express[x4] Riser Kit» OPM-C12W-A6
WLAN Kit, IEEE 802.11a/b/g/n/ac + Bluetooth 4.1» OPM-C15W-A6WLAN Kit, IEEE 802.11a/b/g/n/ac + Bluetooth 5» OPM-T006-A2Smart Card Reader Kit» OPM-C02C-A1
Two Isolated RS-232 & One Isolated USB 2.0 PCIe Card

» OPM-C02C-A2
Two Isolated RS-232 PCIe Card

» OPM-T013-A1
Slim Slot-in DVD-RW Drive» OPM-C03C-A1
One Isolated Gigabit LAN & One Isolated RS-232 &One Isolated USB 2.0 PCIe Card

» OPM-T017-A2
Hot Swap Rack w/Lock Kit for 2.5” HDD/SSD