Production discontinued, successor model under development.

THERMO ELECTRIC PELTIERLERPELTIE CONTROLLER
Model TA-PB-1
 Features Patententnt *１acquisition
uisiisiDownsizing High Precision Peltier(Uni-Thermo recommended) Controller
Temperature Control  0.1 Cm
External Dimension  W 178D 93(D107.5*)H 44.5 mm(*: Including protrusions such as connectorsors)

High Cost Performance by Simple Basic Function
Drive voltage  5 V ~ 25 V 10 A max.
Temp. Control Range  100 C ~ 150 CSimple Method of HandlingTemperature setting, parameter setting are simple(TFT touch screen, free PC application, homepage, etc.)

PC Communication Function Loaded
Setting, monitoring is possible from the outsidesuch as a PC, PLC via 3 wire type RS232

WiFi Communication Function Loaded
Setting, monitoring is possible by the browsers such as a PC,
tablet, smartphone via WiFi. Or setting, monitoring is possible
from the cloud corresponding to the MQTT format.

Corresponds to Various Sensor Types
Standard Correspondence is Pt 100. When purchasing,
you can select thermistor, Pt 100, thermocouple.
(after purchase, it can be changed for a fee.)

Connect Diagram and Extensibility (Semii Custom,Cu
Full Custom Correspondence)
20 series, 30 series, 40 series, 70 series
Corresponds to various Peltier
Peltier drive output corresponds to 5 V to 25 V, up to 10 A

AUX output (cooling FAN etc.)
Independence and joint ownership are
possible to the driving power supply.

WiFi

32 mountedm
PC, PLC and RS 232
controller
face converters
And various interface
EtheUSB, RS485, Ethernet,
BT, BLE, ZIGBEE,x, OpticO
LoRa, SigFox,
fiber communication etc.

Loaded with a Simple WiFi server
Operable with browsers such as
smartphone (iOS and Android),tablet (including iPad),
PC (Win, Mac, Linux)
MQTT compatible cloud
LOSANT (Paid option)

Drive power supply input 5 V ~ 25 V 10 A
And DC output power supply input 5 V to 25 V 10 A

CAUTIONUTIO : Sensor, Peltier product, power supply,connectable equipment etc. are all sold separately.
*1: Acquired patent: Patent No. 6627951

Temperature sensor input 2 system
Pt 100 (standard)
As purchase option●Thermistor specified by our company 10K @ 25 C●Select one type of K, J, N, R, S, T, E, B thermocouples●Compatible with Pt100, thermistor and thermocouple combinations
Other sensors can be customized

TAISEI Co.,Ltd.
e-peltier_cont_ta-pb-1_p1_03

Thermoelectric Peltier Controller
Production discontinued, successor model under development. Model TA-PB-1
Specifications
Temperature Range
Temperature Setting
Temperature Indication

Indicator, Function

Control Method
PID parameter
Peltier Drive Method
Temp. Sensor
Safety Function
Safety operation function
Drive command
Script command
Log function
Input / Output Connector(Standard loading*)
Use a terminal block madeby Pheonix which can insert
and remove all except DC
jack and D-sub 9. Special
connector etc. unnecessary.
Power Supplies
Input Power Supply
Communication
Working Environment
Working Temp. Range
Working Humidity Range
ange
Outer Dimensions
Weight100 C ~ 150 C
Possible in increments of 0.1 C
Possible in increments of 0.1 CTemperature display, graph display, device control and setting input withth 2.4
2. inch TFT
touch screen.
Connection via WiFi by built-in WiFi server in the main unit. PC (Win,Win Mac, Linux) withbrowser, Tablet (including iPad), Smartphone (iOS and Androidd).
(Custom support)
Operation is possible by cloud corresponding to MQTT format
at message.
mPID control (also supports P control and PI control)
For Kp, Ki and Kd, the numerical value setting of the main
body can be entered in 0.1 steps
n bodfrom 0 to 999. (However, As for the number of the indication
ation figures, indication setting ispossible with up to 3 digits of 0.0-99.9-999 including after the decimal point.)
PWM drive
2 systems loading. Standard Pt 100. As purchase option,optio thermistor and thermocouple canbe selected. Also, after purchase, it can be changed
nged for a fee.
In addition, Each sensor, offset function andadjustment function Loaded.
nd gaingMore equipped, such as runaway measuress due to sensor unconnected state.
Operation management of the main unit settsetting screen with PIN is possible.
Language used for input and output with PC,
P PLC, cloud, etc. The grasp of the control andmain body state is possible.
Supports macro languages, conditional
ition expressions, and theoretical expressions forassembly languages that can operatepera at high speed inside the main unit.
Supports startup from drive commands
and startup scripts.
com
Recording to microSD cardd (16GBor less) that can be set inside, or output in MQTT format
1
via 3 wire type RS232 port and via WiFi.
input : Main unit, for AUX
AU (including for DC OUTLET)
Sensor 1, Sensor
enso 2
output : Peltier outpu
output, AUX output 1, AUX output 2, DC OUTLET 2 systems.
(AUXX and DC output can be set independently of drive voltage, and AUX1 and 2
can be OON / OFF controlled)
Input / output
utput : 3 wire type RS232 (Tx, Rx, GND)*: TTL lev
level serial output (Tx, Rx), USB output, additional interface, Peltier output, sensor
type,, inp
input / output port can be customized for a fee.
Fordriving : 5 ~ 25 V 5 %or drMax. 25 V 10A
Max
RS232 (Easy to connect to converters such as RS485 and USB because it is a 3-wireRsystem), WiFi (with a WEB server), I2C (for custom use, no external terminals are exposedfor standard products).
Inside area
5 C ~ 60 C
85 % max. (No evidence of dew)
W 178  D 93 (D107.5*)  H 44.5 mm (Including protrusions such as connectors)
3955 g (Standard product for Pt100)* The
he ccontent of this catalog is based on the information as of January 2020. Specifications of products are subject to change without notice.

e-peltier_cont_ta-pb-1_p2_03

Manufacturing Division 2
Manufacturing Group 3423-3, Arakawa-Niegawa, Chichibu-shi,
Saitama-ken, 369-1911, JAPAN.
TEL +81-494-54-2211 FAX +81-494-54-2216