GENERAL GUIDANCE

7th Edition

Flowmeters and Levelmeters

Contents
FLOW MEASUREMENT AND CONTROL INSTRUMENTS

Metal Tube Variable Area Flowmeter1

Direct Reading Type Flowmeter

2

Sanitary Flowmeter

3

Purgemeter4

Flow Switch / Flow Monitor

5

Flow Switch (for process use)

5

Purge Set

6

Differential Pressure Flowmeter

7

Ultrasonic Flowmeter (for built-in use)

8

Flow Controller (for built-in use)

9

Ultrasonic Flowmeter (for process use)

10

Vortex Flow Sensor

11

Turbine Flowmeter

11

Thermal Flowmeter

12

Thermal Mass Flowmeter

13

Thermal Mass Flow Controller

15

Constant Flow Valve

15

Pitot Tube Flowmeter, Calorie Monitor (for Air conditioning application)

16

Flapper-type Flowmeter

16

Sight Glass

16®

MAGMAX Electromagnetic Flowmeter

17

SWIRLMAX® Vortex Flowmeter

18®

MASSMAX Coriolis Mass Flowmeter

19

Flowmeter for Filling Machines

19

Single-Use Flowmeters

20

Flow Measurement System for Automobile Bench Test

20

Custom-made Product21

Nominal flow rate of variable area flowmeters

22

The following symbols are applicable to flow measurement and control instruments.
Applicable fluids: For liquid measurement

Explosion-proof: For gas measurement: For steam measurement
Contact us for instruments for certified high-pressure gas equipment.

Ex d : Flameproof types available
Ex i : Intrinsically safe types available

LEVEL MEASUREMENT AND CONTROL INSTRUMENTS

Float Type Level Meter

23

Displacer Type Level Meter

25

Microwave Level Meter

25

Micropulse Level Meter

26

Ultrasonic Level Meter

26

Purge Type Level Meter

26

Level Switch

27

Peripheral Instruments for Tank Gauging System

28

MICROCELL Weight Measurement System

29

Oil Leak Detector

29

Level Gauge for LNG Tanks

29

Marine Use Tank Gauging Systems

30

The following symbols are applicable to level measurement and control instruments.
Explosion-proof

Ex d : Flameproof types available

Ex i : Intrinsically safe types available

Contact us for instruments for bonded tanks and high-pressure applications.

RELATED INSTRUMENTS

Level Transmitter

32

Differential Pressure / Pressure Transmitter

32

Receiver32

Metal Tube Variable Area Flowmeter
Metal Tube Variable Area Flowmeter

AM7000 series

Ex d
Ex i

Ex d
FLOW MEASUREMENT AND CONTROL INSTRUMENTS

AM7000/L

Model

AM7000/T

Fluid

Ex d
Ex i

AM7000/R/N/M

AM7000/E/H

Liquid, Gas, Steam

Function·Local indication·Current output·Local totalizer·Pulse output·Alarm output·Local indication·Local indication·Alarm output·R: Reed switch
N: Proximity switch
M: Micro switch·Local indication·E: Current output
H: Current output +HART communication

0.01 to 0.1 m3/h

Measuring range Min.
(water)Max.

20 to 200 m3/h

Measuring range Min.
(air)
Max.

480 to 4800m3/h (nor)Process connection

Flange: 15 mm to 150 mm (1/2" to 6")Standard material

SUS304, SUS316, SUS316L

Available lining material

Rubber, Fluorocarbon resin, PVC, Glass

0.3 to 3 m3/h (nor)Compact Type Metal Tube Variable Area Flowmeter (250 mm unified installation length)

NLZ series

Ex i
Model

Ex d

NLZ1000

NLZ2000

Fluid

Liquid, Gas, Steam

Function·Local indication·Current output·Alarm output·HART communication·FOUNDATION Fieldbus

Measuring range Min.
(water)Max.

0.04 to 0.4 m3/h

Measuring range Min.
(air)
Max.

1.2 to 12 m3/h (nor)Process connection

10 to 100 m3/h
60 to 600 m3/h (nor)Flange: 15 mm to 100 mm (1/2" to 4") for Liquid

15 mm to 80 mm (1/2" to 3") for Gas

Standard material

316L SS, PTFE Lining

Installation length

250 mm

Metal Tube Variable Area Flowmeter for Micro Flow Measurement

AM3000/M-900 series

Ex d
Ex i
Model

AM3000/E

AM3000/H

Fluid
Function·Local indication·Current output·HART communication·Local indication·Current output

0.4 to 2 L/h

Measuring range Min.
(air)
Max.

12 to 60 L/h (nor)

60 to 600 L/h
1700 to 17000 L/h (nor)·Rc 1/4 to 3/4·Flange: 10 mm (3/8") to 25 mm (1")

Standard material

SUS304, SUS316, SUS316L

Option material

Titanium, NW0276

1

M-900

M-950

·Local indication·Local indication·Alarm output(Reed switch)

Liquid, Gas

Measuring range Min.
(water)Max.

Process connection

Ex iMetal Tube Variable Area Flowmeter

MA-900 series

Ex dMA-950

Fluid

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

MA-900Model

MA-920Liquid, Gas

Function·Local indication·Alarm output (Hall IC)·Local indication·Local indication (Digital display)·Current output

Measuring range Min.
(water)Max.

0.1 to 0.5 L/h

0.6 to 3 L/h

60 to 600L/h

60 to 600 L/h

Measuring range Min.
(air)
Max.

3 to 15 L/h (nor)

10 to 100 L/h (nor)

2.2 to 22 m3/h (nor)Process connection

2.2 to 22 m3/h (nor)·Rc 1/4 to 3/4

Standard material·Flange: 10 mm (3/8") to 25 mm (1")
SUS304, SUS316

Direct Reading Type Flowmeter
Glass Tube Variable Area Flowmeter

R series

Ex i
Model

R-101-E

Ex i

R-751-E

R-101

Fluid

Liquid, Gas

LiquidFunction·Local indication·Local indication·Alarm output (reed switch)

R-751-R / R-751
Liquid, Gas·Local indication·Alarm output (reed switch)·Local indication

3

Measuring range Min.
(water)Max.

25 to 250 L/h

0.3 to 3 m /h

0.9 to 9 L/h

7 to 70 L/h

10 to 100 m3/h

8 to 80 m3/h

5.2 to 52 m3/h

5 to 50 m3/h

Measuring range Min.
(air)
Max.

0.45 to 4.5 m3/h (nor)−

15 to 150 L/h (nor)

0.3 to 3 m3/h (nor)

110 to 1100 m3/h (nor)−

100 to 1000 m3/h (nor)

16 to 160 m3/h (nor)Process connection

Flange: 15 mm to 100 mm (1/2" to 4") Flange: 25 mm to 100 mm (1" to 4")

Flange: 10 mm to 100 mm (3/8" to 4") for Liquid
10 mm to 50 mm (3/8" to 2") for Gas

Standard material

SS400, SUS304, PVC, HT-PVC

FC200, SUS304, SUS316, SUS316L, PVC (10 mm to 20 mm)

Resin Tube Variable Area Flowmeter

AC/AC-T series

Ex i
Model

AC (Small size)

Fluid
Function·Local indication·Local indication·Alarm output (Reed switch)

Measuring range Min.
(water)Max.

0.1 to 1 L/min

3 to 30 L/min

10 to 100 L/min

7 to 70 L/min

Process connection
Standard material

AC (Medium size)

AC-T

Liquid·Rc·TS socket 1/2" to 1"·Flange: 15 mm to 25 mm (1/2" to 1")·Local indication·Local indication·Alarm output (Reed switch)·Local indication

0.4 to 4 m3/h

2 to 20 L/min

2 to 20 m3/h

5 to 50 L/min·Rc·TS socket 1-1/2" to 2"·Flange: 40 mm, 50 mm (1-1/2", 2")

Rc 1/2, 3/4

Body: PVC / Tapered tube: Acrylic resin

PVDF/PFA

2

Sanitary Flowmeter
Metal Tube Variable Area Flowmeter for Sanitary Application

AM7000/SR series

Ex d
Ex i
FLOW MEASUREMENT AND CONTROL INSTRUMENTS

Model

AM7000/SR

AM7000/T/SR

Fluid

Function

AM7000/R/N/M/SR

AM7000/E/H/SR·Local indication·Alarm output·R: Reed switch
N: Proximity switch
M: Micro switch·Local indication·E: Current output
H: Current output +HART communication

Liquid·Local indication·Current output·Local totalizer·Pulse output·Alarm output·Local indication

0.01 to 0.1 m3/h

Measuring range Min.
(water)Max.

7 to 70 m3/h

Process connection

1S to 4.5S

Standard material

SUS304, SUS316, SUS316L

Glass Tube Variable Area Flowmeter for Sanitary Application

Electromagnetic Flowmeter for Sanitary Application

R-101-SR series®

6000 series

Model

R-101-SR

R-101-SRE

Fluid

LiquidFunction·Local indication

Measuring range Min.
(water)Max.

5 to 50 L/h

0.025 to 0.25 m3/h

760 to 7600 L/h

2.3 to 23 m3/h

Process connection

1S to 2.5S

1S to 3S

Standard material

SUS304

Optional material

SUS316, SUS316L

3

Model

EGM6300C

Fluid

Conductive liquids

Function·Local indication·Current output·Local totalizer·Pulse output
0 to 0.6 m3/h

Measuring range Min.
(water)Max.

0 to 300 m3/h

Process connection

1S to 4S

Standard material

PFA / Hastelloy® C

Certification

EHEDG / 3A

Purgemeter
P series

P-200

P-400

Fluid

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

P-100

Model

P-510

Liquid, Gas·Local indication·Alarm output

Function·Local indication·Alarm output

Measuring range Min.
(water)Max.
Measuring range Min.
(air)
Max.

5 to 50 mL/min·Local indication·Alarm output
0.1 to 1 L/min

0.2 to 2 L/min

3 to 30 L/min

0.5 to 5 mL/min (nor)

5 to 50 mL/min (nor)80 to 800 mL/min (nor)

2.5 to 25 L/min (nor)

5 to 50 L/min (nor)

6 to 60 L/min (nor)

6 to 60 L/min (nor)

60 to 600 L/min (nor)Rc 1/4Rc 3/8Process connection

Rc 1/8, Rc 1/4

Standard material
Installation length·Local indication

SUS304/SUS316
115 mmSCS14/SUS304

200 mm
·UL-approved reed switch·Optical alarm unit

Alarm

200 mm

200 mm

−·UL-approved reed switch·Optical alarm unit

P series

P-530

Model

P-810

P-820

Liquid, Gas

Function·Local indication·Alarm output·Local indication·Alarm output·Local indication·Alarm output·Local indication·Alarm output

Measuring range Min.
(water)Max.

0.2 to 2 L/min

5 to 50 mL/min

5 to 50 mL/min

0.1 to 1 L/min

1 to 10 L/min

0.2 to 2 L/min

0.2 to 2 L/min

1.5 to 7 L/min

Measuring range Min.
(air)
Max.

10 to 50 L/min (nor)

5 to 50 mL/min (nor)

0.5 to 5 mL/min (nor)

50 to 250 L/min (nor)

6 to 60 L/min (nor)

6 to 60 L/min (nor)Process connection

Rc 3/8

Standard material

SCS14/SUS304

SCS14/SUS316

SCS14/SUS316

SCS14/SUS304

Installation length

150 mm

224 mm115 · 224 mm

76 mm·Optical alarm unit

·UL-approved reed switch·Optical alarm unit

·UL-approved reed switch

AlarmGas, Liquid (equivalent to water)

P-830

Fluid

Rc 1/4, 1/4" SW, 1/4" VCR·UL-approved reed switch

P series

Liquid (equivalent to water)−−
Rc 3/8, 3/8" SW

P series(for micro flow rates)

Model

P-900

Fluid

Gas, Liquid (equivalent to water)

P-850

P-880

Model

Liquid, Gas

Fluid

Gas
·Local indication

Function·Local indication·Local indication

Function

Measuring range Min.
(water)Max.

5 to 50 mL/min

5 to 50 mL/min−

0.25 to 2.5 L/min

0.1 to 1 L/min

Measuring range Min.
(water)Max.

0.15 to 1.5 mL/min (nor)−Measuring range Min.
(air)
Max.

80 to 800 mL/min (nor)

20 to 200 mL/min (nor)

6 to 60 L/min (nor)

2 to 20 L/min (nor)Measuring range Min.
(air)
Max.

Process connection

Rc 1/4, 1/4NPT

Rc 1/8Process connection

Rc 1/4, 1/4" SW, 1/4" VCR

Standard material

SUS304SCS14/SUS304

Standard material

SCS14/SUS316

Installation length

114 · 224 mm

80 mm

Installation length

115·130·145 mm

Alarm

−

−

Alarm

6 to 60 L/min (nor)−4Purgemeter

P series (Resin)

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

P-060

Model

P-620

XP

Fluid

Liquid, Gas

LiquidLiquid, Gas

Function·Local indication·Alarm output·Local indication·Alarm output·Local indication·Alarm output

10 to 100 mL/min

0.1 to 1 L/min

0.02 to 0.1 L/min

1 to 10 L/min

1 to 10 L/min

0.2 to 1 L/min

Measuring range Min.
(water)Max.
Measuring range Min.
(air)
Max.

0.2 to 2 L/min (nor)−

30 to 300 L/min (nor)−Process connection

Rc 1/8 to Rc 3/8

0.1 to 1 L/min (nor)
2 to 20 L/min (nor)Rc 3/8, 3/8" SW, 3/8NPT

Rc 1/4
PolyacetalStandard material

Acryl resin

Acryl resin

Installation length

84 mm

76 mm

80 mm

Alarm

·UL-approved reed switch·UL-approved reed switch·Optical alarm unit

P-700 series(Fluorocarbon resin)

P-771

Model

P-772

Fluid

P-773

P-710

LiquidLiquid, Gas·Local indication·Alarm output·Local indication·Alarm output·Local indication·Alarm output·Local indication·Alarm output

Measuring range Min.
(water)Max.

3 to 15 mL/min

0.06 to 0.6 L/min

0.1 to 1 L/min

3 to 30 mL/min

0.2 to 2 L/min

4.5 to 45 L/min

1 to 10 L/min

Measuring range Min.
(air)
Max.

−

−

−

50 to 500 mL/min (nor)

−

−

−

2 to 20 L/min (nor)Process connection

Rc 1/8, Tube end

Rc, NPT (1/2 · 3/4), Tube end

Rc, NPT (1/4 · 3/8), Tube end

FunctionStandard material
Installation length
Alarm

0.4 to 2 L/min

PFA

Rc 1/8, Fitting
ETFE

80 mm

150 mm

115 mm79 mm·Optical alarm unit

·UL-approved reed switch·Optical alarm unit

·UL-approved reed switch·Optical alarm unit

·Optical alarm unit

Note: Depending on specifications, Model P-772-U (with a valve) falls in “Valves or components thereof” listed in (ii) -7 of row 3 of Appended Table 1 of
the Export Trade Control Order. Consult us for details.

Flow Switch / Flow Monitor

Flow Switch (for process use)

FA series

F-740 series

Ex iFA-3000

Model
Fluid

Ex i

FA4000FA-6000

Liquid (equivalent to water)·Local indication·Alarm output·Local indication·Alarm output·Local indication·Alarm output

Measuring range Min.
(water)Max.

0.3 to 3 L/min

0.1 to 1 L/min

3 to 30 L/min

5 to 50 L/min

13 to 130 L/min

10 to 100 L/min

Process connection

FunctionStandard
material

Model

F-740

Fluid

Liquid (Viscosity: up to 5 mPa · s)

Function·Alarm output

Alarm setting
range(water)Min.

Low alarm 0.1 to 2 m3/h
High alarm 0.13 to 2 m3/h
Low alarm 5 to 70 m3/h

Rc 3/8 to Rc 1

Rc 1/2 to Rc 1-1/2

Rc 1/2 to Rc 1

Tapered tube

Acryl resin

SUS316

Acryl resin

Body

SUS304

SCS14

SCS14

Process connection

Flange 15 mm to 100 mm

0 to 60°C

0 to 100°C

0 to 60°C

Standard material

FC200, SCS13, SCS14

Fluid temperature

5

Ex d
Ex i

Ex i

Max.

High alarm 6.5 to 70 m3/h

Purge Set
CP series
Purege Set

Primary press. control
Secondary press. control

Fluid

CP-11-100, 200, 400
CP-12-100, 200, 400

CP-21-100, 200, 400
CP-22-100, 200, 400

CP-31-500
CP-32-500

Gas

CP-41-500
CP-42-500

Liquid, Gas

Flow control range Min.
(water)Max.

−

5 to 50 mL/min

0.2 to 2 L/min

0.5 to 5 L/min−

0.2 to 2 L/min

0.5 to 5 L/min

1 to 10 L/min

Flow control range Min.
(air)
Max.

10 to 100 mL/min (nor)

0.1 to 1 L/min (nor)

5 to 50 L/min (nor)

15 to 150 L/min (nor)

0.3 to 3 L/min (nor)

5 to 50 L/min (nor)

15 to 150 L/min (nor)

30 to 300 L/min (nor)Process connection

Rc 1/8Rc 1/4Rc 3/8Rc 1/2Standard material
Controllable DP range

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

Model

SUS304
C-11: 0.03 to 0.3 MPa
C-12: 0.05 to 0.3 MPa

0.06 to 0.4 MPa

0.1 to 0.5 MPa

0.1 to 0.6 MPa

C series
Constant flow valve

Model

C-11
C-12

Primary press. control
Secondary press. control

Fluid

C-21
C-22

C-31
C-32

Gas

C-51
C-52

Liquid, Gas

Flow control range Min.
(water)Max.
Flow control range Min.
(air)
Max.

C-41
C-42

−

0.9 to 9 L/h

12 to 120 L/h

30 to 300 L/h

70 to 700 L/h−

12 to 120 L/h

30 to 300 L/h

70 to 700 L/h

120 to 1200 L/h

10 to 100 mL/min (nor)0.015 to 0.15 m3/h (nor)0.36 to 3.6 m3/h (nor)

0.9 to 9 m3/h (nor)

2.1 to 21 m3/h (nor)

0.3 to 3 L/min (nor)0.36 to 3.6 m3/h (nor)

0.9 to 9 m3/h (nor)

2.1 to 21 m3/h (nor)

3.6 to 36 m3/h (nor)Process connection

Rc 1/8Rc 1/4Standard material

SUS304

SCS14·Rc 3/8 ·Flange 15 mm (1/2") ·Rc 1/2 ·Flange 15 mm (1/2")

Controllable DP range

0.03 to 0.3 MPa

0.06 to 0.4 MPa

Flange 20 mm (3/4")

SUS304
0.1 to 0.5 MPa

C series
Constant flow valve

0.1 to 0.6 MPa

CR/CAM series
Purege Set
Combination
withR series

Model

Primary press. control
Secondary press. control

C-61
C-62

Fluid
Flow control range Min.
(water)
Max.

C-71
C-72

C-81
C-82

Liquid, Gas
120 to 1200 L/h

180 to 1800 L/h

300 to 3000 L/h

180 to 1800 L/h

300 to 3000 L/h1000 to 10000 L/h

Flow control range Min.
(air)
Max.

3.6 to 36 m3/h (nor)

5.4 to 54 m3/h (nor)9 to 90 m3/h (nor)

5.4 to 54 m3/h (nor)9 to 90 m3/h (nor)23 to 230 m3/h (nor)Process connection

Flange 20 mm (3/4")

Flange 25 mm (1")

Flange 50 mm (2")

Standard material

SUS304Controllable DP range

0.1 to 0.6 MPa

CombinationwithAM series

6

Differential Pressure Flowmeter
ORIFLOMETER®O/HDT series

Ex i

O-180

Model

O-680

Fluid

O-780

HDT1000

Liquid (Viscosity: up to 3 mPa·s), Gas

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

Function·Local indication·Alarm output(Optical alarm unit)·Local indication·Local indication·Alarm output·Current output·Pulse output·Battery-powered·Local indication·Alarm output(Reed switch)

Standard DP (Liquid)

15 kPaMeasuring range Min.
(water)Max.

0.05 to 0.25 m3/h
300 to 1500 m3/h

340 to 1700 m3/hStandard DP (Gas)

5 kPa

20 kPaMeasuring range Min.
(air)
Max.

0.7 to 3.5 m3/h (nor)0.84 to 4.2 m3/h (nor)0.34 to 3.4m3/h (nor)4600 to 23000 m3/h (nor)9000 to 45000 m3/h (nor)1720 to 17200 m3/h (nor)−0.023 to 0.23 m3/h
106 to 1060 m3/h−

·Rc 1/2 to Rc 4
·Flange / Wafer 15 mm to
300 mm (1/2" to 12")

·Rc 3/8 to Rc 4
·Flange / Wafer 10 mm to 500 mm (3/8" to 20")

Process connection
Standard
material

20 kPaMeasuring pipe

SGP, SUS304, SUS316, PVC, HT-PVC

Indicator

SCS14, PVC, HT-PVC

SUS304

O/HDT series

Ex d
Ex i

O-100

Model

Ex d
Ex i

O3000

Fluid

O7000

Liquid, Gas

Function·Local indication·Alarm output·Local indication·Alarm output·Current output·Pulse output·Local indication·Current output·HART communication

Standard DP (Liquid)

30 kPaMeasuring range Min.
(water)Max.

0.5 to 2.5 m3/h

0.054 to 0.27 m3/h

0.8 to 4 m3/h

400 to 2000 m3/h

164 to 820 m3/h

600 to 3000 m3/hStandard DP (Gas)

10 kPa

40 kPa60 kPaMeasuring range Min.
(air)
Max.

6.8 to 34 m3/h (nor)

1.1 to 5.5 m3/h (nor)18 to 90 m3/h (nor)

6600 to 33000 m3/h (nor)4800 to 24000 m3/h (nor)Process connection
Standard
material

Measuring pipe
Indicator

60 kPaD-D/2 tap: 100 mm to 500 mm (4" to 20")
Corner tap,Fflange tap: 50 mm to 500 mm (2" to 20")
Vena contracta tap: 200 mm to 500 mm (8" to 20")

14400 to 72000 m3/h (nor)D-D/2 tap: 100 mm to 500 mm (4" to 20")
Corner tap,Fflange tap: 50 mm to 500 mm (2" to 20")
Vena contracta tap: 200 mm to 500 mm (8" to 20")

·Rc 1/2 to 4
·Flange/Wafer: 15 mm to 300 mm
SS400, SUS304, SUS316

SS400, SUS304, SUS316

SS400, SUS304, SUS316

SUS304, SUS316

V-Cone® FlowmeterV series

Ex d
Model

VC

Fluid
Function

VD

Differential pressure port: Rc 1/2 or Rc 1/4·Gas, Saturated steam·Local indication·Current output

Standard material

7·Local indication·Current output·Mass flow

0.4 to 3.47 m3/h−

112 to 1245 m3/h−

7 to 71 m3/h (nor)−

300 to 32859 m3/h (nor)Measuring range Min.
(Saturated steam) Max.
Process connection

VM·Steam, Liquid, Gas

Measuring range Min.
(water)Max.
Measuring range Min.
(air)
Max.

Ex d

6 to 65 kg/h
2600 to 28315 kg/h
Flange 15 mm to 400 mm (1/2" to 16")

Flange 15 mm to 300 mm (1/2" to 12")
SUS304

Differential Pressure Flowmeter®

Wafer-Cone FlowmeterV series Wafer type

VH

Model

VT

Fluid
Function

VNT

VDT

Liquid, Gas, Saturated steam
Differential pressure port:
Rc 1/4 or Rc 1/8

VTW

Liquid, Gas·Local indication·Totalizer·Current output·Pulse output·Local indication·Current output

Liquid·Local indication·Current output·Battery-powered·Local indication

Measuring range Min.
(water)Max.

0.4 to 5.5 1m3/h

0.26 to 1.3 m3/h

9 to 119.73 m3/h

18 to 90 m3/h

Measuring range Min.
(air)
Max.

6 to 77 m3/h (nor)−−

120 to 1587 m3/h (nor)Measuring range Min.
(Saturated steam) Max.

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

Ex d

750 to 7500 m3/h (nor)−8 to 103 kg/h−−

170 to 2421 kg/h−−

Process connection

Wafer: 25 mm to 100 mm (1" to 4")Standard material

SCS14A

Ultrasonic Flowmeter (for built-in use)
Clamp-on type

GST/SFC011GS
seriesModel

UCL/SFC010C/SFC011C
series

GSTPower supply
Fluid
Function
Measuring range Min.
(water)Max.

SFC011GS

Model

UCL

24 V DCPower supply

Water

Fluid

·Flow rate indication·Current output·Pulse output
0 to 0.5 L/min

SFC011C (6 ch)

24 V DC
Water·Flow rate indication·Current output·Pulse output

Function
Measuring range Min.
(water)Max.

0 to 2 L/min

SFC010C (1 ch)

0 to 3 L/min
0 to 80 L/min

Process connection

1/8, 1/4, 4 mm, 5 mm, 6 mm

Process connection

6 mm, 8 mm, 10 mm,
6.35 mm, 9.53 mm,
12.70 mm, 19.05 mm,
25.40 mmStandard material

PPS GF

Standard material

PPS, PP

Integrated Type

UF/UCF/UCM®
seriesModel

UF

UCF006

Power supply
Fluid
Function

UCM-04A/06A

24 V DC
Water·Flow rate indication·Current output·Pulse output

Liquid·Analog output·Pulse output·Flow rate indication·Analog output
0 to 0.2 L/min

Measuring range Min.
(water)Max.

0 to 20 L/min

0 to 3 L/min

0 to 150 L/min

0 to 8 L/min

0 to 8 L/min

Process connection

Tube end: 1/2", 3/4", 1"

Tube end: 3/8"

Tube end: 1/4", 3/8"

Standard material

PFA

8Ultrasonic Flowmeter (for built-in use)
Detector (Sensor)

UCUF® series

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

UCUF-EModel

UCUF-K

UCUF-02M

Fluid

UCUF-04MT

LiquidMeasuring range Min.
(water)Max.

0 to 8 L/min

0 to 50 mL/min
0 to 80 L/min

Process connection

Tube end: 3/8"

Tube end: 3/8" to 1"

0 to 10 mL/min−

0 to 100 mL/min

0 to 2 L/min
Tube end: 1/4"

Standard material

PFA*Max. fluid temperature for UCUF-04MT is 180˚C.

Converter

SFC series

SFC4000

Model

SFC3000

Power supply
Flow rate
Output

SFC-010L

SFC-010T·4 to 20 mA DC·0 to 20 mA DC·4 to 20 mA DC·1 to 5 V DC

24 V DC·4 to 20 mA DC·1 to 5 V DC

4 to 20 mA DC

Pulse
Alarm

Open collector: 0 to 1000 Hz max.
2 points

2 points

Communication
Applicable Detector (Sensor)

RS485 (Protocol: MODBUS)
UCUF-K, UCUF-M, UCUF-E

UCUF-K, UCUF-M, UCUF-E

UCUF-02M

UCUF-04MT

Flow Controller (for built-in use)
Controller

Control Valve

FCA6000 series

ModelPower supply
Function

Setting input
MV (flow rate) output

9

FCV series

FCA6000

FCV-3000Model

24 V DC

Fluid·Flow rate indication·Flow rate output·Alarm output·Pulse output
1 to 5 V DC/
0 to 10 V DC/
4 to 20 mA DC
0 to 10 V DC/
4 to 20 mA DC

Flow control
range (water)

FCV-3000T

FCV-1000S

LiquidMin.

2.5 to 25 mL/min−

0.2 to 2 L/min

Max.

200 to 2000 mL/min

50 to 500 mL/min

1 to 10 L/min

Process connection

Tube end:
ø6.35 × ø4.35

Tube end:
ø4 × ø2.8

Tube end:
ø9.53 × ø6.35
Tube end:
ø12.7 × ø9.53

Standard material

PTFE, PFA

THV

PCTFE, PTFE,PFA

Flow Controller (for built-in use)
All-in-one type

CLFC® series

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

All-in-one Flow Controller with a ultrasonic flowmeter · a control valve and a controller

CLFC300

Model
Power supply

24 V DC

Fluid

LiquidFlow control range Min.
(water)Max.

2.5 to 25 mL/min
200 to 2000 mL/min

Process connection

SUPER300 Type PILLAR FITTING etc.

Standard material

PFA, PTFE

Setting input

4 to 20 mA DC

MV (flow rate) output

4 to 20 mA DC

Ultrasonic Flowmeter (for process use)
Clamp-on TypeUL300
UL6000seriesGeneral purpose · for small pipe sizes

General purpose · for small and medium pipe sizes High-performance · for small to large pipe sizes

UL330

Model

UL350

Fluid

UL6300 V2

Liquid·Flow rate indication·Totalizer·Current output·Pulse output·Status output·Flow rate indication·Totalizer·Current output·Pulse output·Status output·RS485 (optional)·Flow rate indication·Totalizer·Bar graph·Current output (HART)·Pulse output·Status output

Measuring range Min.
(flow velocity)Max.

0 to 0.3 m/s

0 to 0.3 m/s

0 to 0.5m/s

0 to 10 m/s

0 to 10 m/s

0 to 20m/s

Measurable pipe size

25 mm to 400 mm (1" to 16")25 mm to 1000 mm (1" to 40")

15 mm to 4000 mm (1/2" to 160")

Pipe material

Metal, Resin

Function

Metal, Resin, Polyethylene lining

3-Beam In-line TypeSONICMAX ®
UL3400

Ex d

UL3400

Model
Fluid

LiquidFunction·Flow rate indication·Totalizer·Current output·Pulse output·Status output

Measuring range Min.
(flow velocity)Max.

0 to 0.3 m/s

Process connection

Flange: 25 mm to 2000 mm (1" to 80")

Standard material

316L SS

0 to 20 m/s

10Vortex Flow Sensor
VF series

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

VF-2000

Model

VF-2200

VF-2300

Fluid

Liquid (low viscosity)

Type / Function·Current output type
·Pulse output type
·Flow rate indication + Current/Alarm output type−

VF-3000

Measuring range Min.
(water)Max.

0.5 to 4 L/min

10 to 100 L/min

4 to 40 L/min

10 to 150 L/min

25 to 250 L/min

15 to 150 L/min

0.3 to 2.5 L/min

Process connection

R3/8 to 1/2·TS socket DN25 (1")
·Rc 1
·Flange 25 mm (1")·R 1-1/4·Flange 32 mm (1-1/4")

Tube end: 3/8" to 1"

Standard material

PPS resin

PPS resin /PVC

New PFA

Turbine Flowmeter
Axial-flow Flowmeter

Mini-wheel / Mag-wheel / Manifold Mini-wheel Flowmeter

TW series

W series

Model

TW-080/TW-090

Model

Fluid

Cooling water and a variety of fluids

Fluid·Voltage output·Pulse output

Function

W1000

W-500

Cooling water and a variety of fluids·Current output·Voltage output·Pulse output

Function·Local indication·Current output·Pulse output

Measuring range Min.
(water)Max.

0.2 to 2 L/min

Measuring range Min.
(water)Max.

0.3 to 1 L/min

0.7 to 3.5 m3/h

2 to 20 L/min

6 to 60 L/min

50 to 400 m3/h

Process connection

Rc 1/4, 3/8

Process connection

Rc 1/4 to 3/4

Standard material

SCS14

Standard material

P.P./ PVC/ PTFE

·Rc 1/2 to 1
·Flange 15 mm to 200 mm
SUS304
SUS316/ PVC

Mini-wheel / Mag-wheel / Manifold Mini-wheel Flowmeter

W series

Magnetic type

W-2000/2000N

Model
Fluid

W-3000Liquid·Current output·Pulse output

Function·Pulse output

Optical type

MU-1000

W-800

Cooling water

Liquid·Current output·Pulse output·Pulse output

Measuring range Min.

0.5 to 3 L/min

0.6 to 3 L/min

0.04 to 0.2 L/minMax.

6 to 60 L/min

2 to 20 L/min

5 to 50 L/min

Rc 3/8 to 3/4

Rc 3/8Tube end
Rc 1/4 to 3/4(water)

Process connection
Standard material

11

SCS14

PFA (PTFE)

Thermal Flowmeter
TH series

TH·Detector

Insertion type

Flange type

Insertion type (Variable length)

Small size

Built-in straightener typeTH-1100

TH-1200

TH-1400

TH-1700

TH-1800

Fluid

Gas

Measuring range Min.
(air)
Max.
Operatingtemperature

0 to 45 m3/h (nor)0 to 45 m3/h (nor)0 to 45 m3/h (nor)0 to 250 L/min (nor)0 to 10 L/min (nor)0 to 680000 m3/h (nor)0 to 7000 m3/h (nor)0 to 680000 m3/h (nor)0 to 14000 L/min (nor)0 to 5800 L/min (nor)Standard typeHigh-temp. type

Process connection /Process pipe size

80˚C Max.
240˚C Max.
Flange: 50 mm to 1500 mm(2" to 60")

180˚C Max.
Flange: 50 mm to 150 mm(2" to 6")Standard material

Flange: 50 mm to 1500 mm(2" to 60")

Flange: 15 mm to 50 mm(1/2" to 2")

Rc 3/8 to 1
Flange: 15 mm to 50 mm (1/2" to 2")

SUS304, SUS316, SUS316L

Built-in purge function

High temperature / Low temperature type

TH·Detector

Model

Insertion type(max. temp. 240°C)

Insertion type(max. temp. 550°C)

TH-1100-SP

TH-3200-SP

Fluid

TH-3200

TH-3300

Gas

Measuring range Min.
(air)
Max.
Operatingtemperatur

Insertion type

TH-3100

0 to 120 m3/h (nor)Standard type

80˚C Max.

High-temp. type

240˚C Max.

Process connection /Process pipe sizeStandard material

0 to 260 m3/h (nor)0 to 80 m3/h (nor)0 to 60 m3/h (nor)0 to 390000 m3/h (nor)550˚C Max.

Flange: 50 mm to 1500 mm(2" to 60")

0 to 380000 m3/h (nor)550˚C Max.
Flange: 80 mm to 1500 mm(3" to 60")-196˚C to 0˚CFlange: 65 mm to 1500 mm(2-1/2" to 60")

SUS304, SUS316, SUS316L

12

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

ModelThermal Flowmeter

TRX·Converter

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

TRX-600

Model
Power supply

Function

OutputTRX-700

TRX-900

100, 110, 115, 200,
220, 240 VAC

96 to 264 VAC·Flow rate
·Totalizer·Temperature·Temp. compensation·Purge control·Current output·Pulse output(Photo MOS open collector)·RS-485·Flow rate
·Totalizer·Alarm contact output·Bar graph indication·Temperature / pressure compensation·Current output·Alarm output·Pulse output (open collector)·RS-485

Cable length

50 m Max.

100 m Max.

Waterproof(for outdoor use)

Housing

Panel mount(for indoor use)

Waterproof(for outdoor use)

SRT seriesCompact & Separate
type

Model

Insertion typeIn-line type (medium size)

In-line type (small size)

Insertion type (Variable length)

SRT1100

SRT1200

SRT1300

SRT1400

Fluid

Gas

Function
Measuring range Min.
(air)
Max.

·Local indication·Analog output· Alarm output·RS-485

0 to 40 m3/h (nor)0 to 2.5 m3/h (nor)0 to 60 m3/h (nor)0 to 890000 m3/h (nor)0 to 10000 m3/h (nor)0 to 300 m3/h (nor)0 to 310000 m3/h (nor)Flange: 50 mm to 1500 mm(2" to 60")

Flange: 40 mm to 150 mm(1-1/2" to 6")·Rc 3/8 to 1
·Flange: 10 mm to 25 mm (3/8" to 1")

Flange: 50 mm to 900 mm

Operating temperature
Process connection /Process pipe size

·Pulse output

0 to 60 m3/h (nor)120˚C Max.

Standard material

SUS316, SUS316L

Thermal Mass Flowmeter
TF series
HM series

Ex d
Model

TF-5000/TF-6000

EP-TF-5300

Fluid

Gas

Function·Voltage output

Measuring range Min.
(air)
Max.

0 to 5 mL/min (nor)Process connection
Standard material

13

H-EP-TF-5300

HM1000HM5000HM9700AGas

0 to 500 L/min (nor)Rc 1/4 to 1

Rc 1/4 to 3/4
SUS316·Voltage output·Local indication·Current output·Alarm output

0 to 5 mL/min (nor)0 to 1 L/min (nor)0 to 20 L/min (nor)0 to 400 L/min (nor)1/4" SW

0 to 200 L/min (nor)1/4" to 1/2" SW
SUS316

Thermal Mass Flowmeter

TF series

Compact typeHigh performance

Compact typeTF-1000

TF-900

TF-4000

TF-600

Fluid

Gas

Function

Indicator / valve built-in

TF-600D/600V

Air, N2, O2

Air, N2, O2

Air, N2·Voltage output·Current output·Voltage output·Local indication·Totalizer·Current output·RS-485·Pulse output·Alarm output·Voltage output·Local indication·Voltage output·Pulse output·Alarm output·RS-485

Measuring range Min.
(air)
Max.

0 to 2 L/min (nor)0 to 10 L/min (nor)0 to 2 L/min (nor)0 to 20 L/min (nor)0 to 5 L/min (nor)0 to 1000 L/min (nor)0 to 100 L/min (nor)0 to 1000 L/min (nor)0 to 100 L/min (nor)0 to 1000 L/min (nor)Process connection

Rc 1/4 to 1

Rc 1/4

Rc 1/4 to 3/4

Rc 1/4

Rc 1/4 to 3/4

Standard material

SUS316SCS14
Polyacetal

SCS14

SCS14, SUS316

TF series

Rotatable indicator type,for large flow rates

Flow rate, Totalizer, Alarm indication+ output

TF-4100

Model

TF-2000T

For medium tolarge flow rate

TF-2261T

TF-1161/TF-1261

Fluid

Air, N2

Gas

Air, N2

Air, N2

Function·Local indication·Current output·Voltage output·Pulse output·Alarm output·RS-485·Local indication·Current output·Pulse output·Alarm output·Local indication·Current output·Pulse output·Alarm output·Current output

Measuring range Min.
(air)
Max.

0 to 4000 L/min (nor)0 to 2 L/min (nor)0 to 16000 L/min (nor)0 to 750 m /h (nor)0 to 1500 m3/h (nor)Process connection

Rc 1 to 2·Rc 1/4 to 2
·Flange 15 mm to 80 mm·Rc 1 to 2
·Flange 25 mm to 80 mm

Standard material

A6061-T6, SCS13

SUS316

SUS304, SUS316

30 to 80 m3/h (nor)Converter for
TF series

Type

TM3000Applicable instrument·TF-900·TF-1000·TF-5000/6000·(H-) EP-TF-5300

Function·Flow rate
indication·Totalizer·Analog output·Pulse output·RS-485

14

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

Small to Large flow
Model

Thermal Mass Flow Controller
TC series

Ex d
FLOW MEASUREMENT AND CONTROL INSTRUMENTS

TC-1000/2000

Model

EP-TC-1000/2000 V

Fluid

TC-3000

Gas

Function·Flow rate control·Voltage output·Flow rate control·4 to 20 mADC
·0 to 5 VDC

0 to 5 mL/min (nor)0 to 2 L/min (nor)Measuring range Min.
(air)
Max.

0 to 500 L/min (nor)0 to 100 L/min (nor)0 to 800 L/min (nor)Process connection

Rc, SW 1/4" to 1"

Rc, SW, VCR 1/4" or 3/8"

Rc 1/4 to 1

Standard material

SUS316

HM series

Converter
for TC series

HM1000

Model
Fluid

HM5000Gas

Function·Flow rate control

Measuring range Min.
(air)

Max.

Process connection

Power supply

85 to 240 V AC

Applicable instrument·TC-1000·TC-2000·TC-3000·EP-TC-1000/2000V

Function·Flow rate indication·Analog output·Flow rate setting·Voltage output

0 to 5 L/min (nor)0 to 5 mL/min (nor)0 to 20 L/min (nor)0 to 400 L/min (nor)1/4" SW

1/4" to 1/2" SW

Standard material

TM-1400

Type

SUS316

Constant Flow ValveCX series

CX-1101

Model
Fluid
Flow range(water/air)LiquidMin.
Max.

Process connection

3 to 10 L/min
25 to 70 L/min
Rc 1/2 to 1-1/2

CX-1500
Gas

CX-1510Water

3

5 to 13 m /h (nor)3

600 to 1000 m /h (nor)

0.7 to 1.1 m /h
3

22 to 60 m /h

·Rc (NPT) 1/2 to 2
·Rc (NPT) 1/2 to 2
·Flange 15 mm to 150 mm (1/2" to 6") ·Flange 15 mm to 100 mm (1/2" to 4")

FPC/RSP/NSPW
series

FPC

Model
Fluid
Flow range(water)

NSPW/NFFW/NFF-S

Water

Water

Water and other liquidsMin.

0.7 to 2 L/min

0.6 to 1.8 L/min

5 to 20 L/min

Max.

4 to 7 L/min

4 to 9 L/min

Rc 1/4 to 1/2

Rc 3/8, Rc 1/2

Process connection

15

RSP

CX-2000
Liquid3

22 to 850 L/min·Rc 3/8 to 1
·Flange 32 mm to 80 mm (1-1/4" to 3")

0.2 to 1.2 m3/h
10 to 70 m3/h
Flange 15 mm to 100 mm(1/2" to 4")

Pitot Tube Flowmeter, Calorie Monitor (for Air conditioning application)
CFW/CDT series

CFW1000

Model

CFW2000

Fluid

CDT1000

CDT2000

CDT3000·Local indication·Detachable·Battery powered·Local indication·Alarm output·Current output·Calorie monitor

Water, Cold water

Measuring range Min.
(water)Max.

·Local indication·Detachable·Local indication·Local indication·Alarm output·Current output·Battery powered

12 to 100 L/min

5 to 50 L/min

0.3 to 3 m3/h

4500 to 35000 L/min

1600 to 16000 L/min

100 to 1000 m3/h

Process pipe size

20 mm to 450 mm (3/4" to 18")

Standard material

SUS316 / C3604

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

Function

Flapper-type Flowmeter
STK series

STK2000

Model
Fluid

STK7400
LiquidFunction
Measuring range Min.
(water)Max.
Process connection
Standard material·Local indication·Alarm output·Local indication
3

0.3 to 1.5 m /h
120 to 600 m3/h
Flange 15 mm to 300 mm(1/2" to 12")SS400, SGPSCS14, SUS304, PVC

Sight Glass
STK series

STK4000

Model
Fluid

LiquidFunction·Local indication

Measuring range Min.
(water)Max.

0.16 to 0.6 m3/h

Process connection
Standard material

120 to 600 m3/h
·Rc 1/2 to 3/4·Flange 25 mm to 300 mm(1" to 12")
FCD450, SS400
SCS14, SUS304

16

®

Electromagnetic Flowmeter

Compact Type Electromagnetic Flowmeter

EGM series

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

PFA Lining(Meter size 10 mm to 150 mm)

EGM1050C

Model

EGM1100C

Hard rubber / Polypropylene (PP) Lining(Meter size 25 mm to 1000 mm)

EGM1300C

Fluid

EGM2050C

EGM2100C

EGM2300C

Conductive liquids

Function
Measuring
range·Flow rate indication
Min.
Max.

·Totalizer·Current output (HART)·Pulse output·Status output

0 to 0.09 m3/h

0 to 0.6 m3/h

0 to 700 m3/h

0 to 33000 m3/h

Process connection

Wafer 10 mm to 150 mm (3/8" to 6")

Flange 25 mm to 1000 mm (1" to 40")

Standard material

PFA / Hastelloy® C

Polypropylene (PP) · Hard rubber / Hastelloy® C

EGM series

PFA or other Lining(Meter size 10 mm to 1000 mm)

EGM4050C

Model

EGM4100C

Ceramic type
(Meter size 2.5 mm to 100 mm)

EGM4300C

Fluid

EGM5100C

EGM5300C

Sanitary type
(Meter size
25 mm to 100 mm)

EGM6300C

Conductive liquids

Function·Flow rate indication·Totalizer·Current output (HART)·Pulse output·Status output

Min.

0 to 0.09 m3/h

0 to 0.01 m3/h

0 to 0.6 m3/h

Max.

0 to 33000 m3/h

0 to 300 m3/h

0 to 300 m3/h

Process connection

Flange 10 mm to 1000 mm (1" to 40")

Wafer 10 mm to 100 mm (3/8" to 4")Standard material

PFA · PTFE · ETFE / Hastelloy® C

Zirconia ceramic · Alumina ceramic/Platinum

Measuring
range

Sanitary joint
1S to 4S
PFA / Hastelloy® C/ 316SS

Battery-powered Electromagnetic Watermeter

ETM series

EGM series

Capacitive type
Electromagnetic flow switch Electromagnetic flow switch(Meter size 25 mm to 100 mm) (Meter size 10 to 150 mm) (Meter size 25 to 300 mm)

EGM7300C

Model
Fluid

·Flow rate indication·Totalizer·Current output (HART)·Pulse output·Status output

Function

Measuringrange

EGM2300CS·Flow rate indication·Totalizer·Alarm output (Dry contact)

Min.

0 to 0.6 m3/h

0 to 0.09 m3/h

0 to 0.6 m3/h

Max.

0 to 300 m3/h

0 to 700 m3/h

0 to 3000 m3/h

Wafer 25 mm to 100 mm(1" to 4")
Zirconia ceramic
Alumina ceramic

Wafer 10 mm to 150 mm(3/8" to 6")

Flange 25 mm to 300 mm(1" to 12")PP · Hard rubber /Hastelloy® C

Process connection
Standard material

17

EGM1300CS
Conductive liquids

PFA/ Hastelloy® C

Battery-powered(Meter size 25 mm to 200 mm)
Model

ETM3070

Fluid

Water (50 μS/cm or more)

Function·Flow rate indication·Totalizer·Pulse output

Measuring
range

Min.

0 to 0.9 m3/h

Max.

0 to 1000 m3/h

Process connection
Standard material

Flange 25 mm to 200 mm(1" to 8")
Rilsan (Polyamide resin)/ 304SS®

Electromagnetic Flowmeter

Separate Type Detector

EGS series

Hard rubber / PP Lining(Meter size 25 mm to 1000 mm)

PFA or other Lining(Meter size 10 mm to 1000 mm)

Ceramic type
(Meter size 2.5 mm to 100 mm)

EGS1000

EGS2000

EGS4000

EGS5000

Fluid

Conductive liquidsMin.

0 to 0.09 m3/h

0 to 0.6 m3/h

0 to 0.09 m3/h

0 to 0.01 m3/h

Max.

0 to 700 m3/h

0 to 33000 m3/h

0 to 33000 m3/h

0 to 300 m3/h

Process connection

Wafer 10 mm to 150 mm(3/8" to 6")

Flange 25 mm to 1000 mm(1" to 40")

Flange 10 mm to 1000 mm(3/8" to 40")

Wafer 10 mm to 100 mm(3/8" to 4")Standard material

PFA / Hastelloy® C

Polypropylene (PP) · Hard rubber /Hastelloy® C

PFA · PTFE · ETFE /Hastelloy® C

Zirconia ceramic
Alumina ceramic / Platinum

Measuringrange

Separate Type Converter

EGC series

Standard typeHigh-performance type

EGC050W

Model

EGC100W

EGC300F/W

Power supply

100 to 230 V AC / 24 V DC

Function·Flow rate indication·Totalizer

Applicable instrument

EGS/MGS/IFS series

Electromagnetic flow detectors·Current output (HART)·Pulse output·Status output

SWIRLMAX ® Vortex Flowmeter
VFM series

Ex d
Model

VFM4200

Fluid

Gas, Liquid, Steam

Function·Local indication·Current output

1772 m3/h

Measuring range Min.
(air /0 MPa)
Max.

25250 m3/h (nor)Measuring range Min.
(saturated steam /0.1 MPa) Max.
Process connection
Standard material·Pulse output

0.91 m3/h

Measuring range Min.
(water /20˚C)Max.

12.1 m3/h (nor)11.8 kg/h
22996 kg/h
Flange 15 mm to 300 mm (1/2" to 12")Wafer 15 mm to 100 mm (1/2" to 4")
316L SS

18

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

Model

PFA Lining(Meter size 10 mm to 150 mm)

®

Coriolis Mass Flowmeter

MMM series

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

Ex d

Ex d

Straight twin tube
(Meter size 15 mm to 50 mm)

Straight twin tube
(Meter size 10 mm to 400 mm)

Single tube
(Meter size 1 mm to 4 mm)

Straight single tube
(Meter size 6 mm to 80 mm)

MMM2400R

MMM3400R

MMM7400R

MMM1400R

Model

Ex d

Fluid

LiquidFunction·Flow rate indication·Totalizer·Current output (HART)·Pulse output·Status output·Density measurement (for 15 mm or larger meter sizes)

Measuring
range

Min.

48 kg/h1560 kg/h

0.3 kg/hMax.

125,000 kg/h

4,600,000 kg/h

450 kg/h

Process connection

Flange 15 mm to 80 mm (1/2" to 3") Flange 100 mm to 400 mm (4" to 12")

Standard material

ASTM UNS S31803 / 316L SS

Ex d

12 kg/h
560,000 kg/h

1/4 NPT male

Flange 10 mm to 100 mm (3/8" to 4")

316L SS

Titanium · Hastelloy® C22

MMM series

Ex dTwin tubes, high precision and low and high temperature resistant model(Meter size 8 mm to 200 mm)

MMM6400R

Model
Fluid

Liquid, Gas

Function·Flow rate indication·Totalizer·Current output (HART)·Pulse output·Status output·Density measurement

Measuring
range

Min.

5 kg/hMax.

1,000,000 kg/h

Process connection

Flange 10 mm to 200 mm (3/8" to 8")Standard material

316/316L SS (dual certified)

Flowmeter for Filling Machines
Coriolis Mass Flowmeter

Electromagnetic Flowmeter

MMM series

EGM series

Twin tube(Meter size 10 mm, 15 mm)
Model

MMM4011C

Model

Fluid

LiquidFluid

Function·Pulse output
Flow rate

Recommended
fillingMin. filling
conditions
volume8 mm: 5 to 165 g/sec
10 mm: 10 to 360 g/sec
15 mm: 25 to 1200 g/sec
8 mm: 7.5 g
10 mm: 15 g
15 mm: 37.5 g

Ceramic type
(Meter size 10 mm, 15 mm)

Ceramic type
(Meter size 2.5mm to 40 mm)

EGM5500C

EGM5015C
Conductive liquids

Function
Flow rate
Recommended
Min. filling
filling
volumeconditions
Min. filling time

·Pulse output
10 mm: 60 to 200 mL/sec
15 mm: 150 to 600 mL/sec
10 mm: 100 mL
15 mm: 200 mL

2.5 mm: 3 to 10 mL/sec
40 mm: 1000 to 3000 mL/sec
2.5 mm: 10 mL
40 mm: 1500 mL

1.5 s

1.5 s
Wafer: 10 mm to 40 mm(3/8" to 1-1/2")
Zirconia ceramic
Alumina ceramic / Platinum

1.5 s

Process connection

Process connection

Sanitary joint: 10 mm to 15 mm(3/8" to 1/2")

Wafer: 10 mm to 15 mm(3/8" to 1/2")

Standard material

Zirconia ceramic / Platinum

Standard material

316L SS

Min. filling time19

Single-Use Flowmeters
BIOMAG® Electromagnetic Flowmeter

BIOSONIC® Ultrasonic Flowmeter

BIOMAG
series

BM/BIOMAG4050C

Model

BS

Power supply

24 V DCPower supply

Fluid

LiquidFluid·Current output·Pulse output

Function

Flow range

Min.

0.015 to 3 L/min

Max.

0.5 to 75 L/min

SFC4000-EV

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

Model

BIOSONIC
series24 V DC
Liquid·Flow rate indication·Current output·Pulse output·Alarm output

Function

Flow range

0 to 3 L/min

Min.

0 to 8 L/min

Max.

Process connection

Barb fittings 1/4 to 1

Process connection

Barb fitting 3/8

Standard material

PUS, Hastelloy C22

Standard material

PSU

Flow Measurement System for Automobile Bench Test
CNG Flow Measurement System

TRX series
Converter

TH series Detector

TH-1800-T

Model
Fluid

CNG

Function·Local indication

Measuring
range·Current output

Min.

5 to 365 L/min (nor)Max.

45 to 3795 L/min (nor)Protection category

Equivalent to IP65

Process connection·Rc 3/8 to 1
·Flange: 15 mm to 50 mm (1/2" to 2")

Accuracy±1% R.D (Flow range 5 to 100%)

Standard material

SUS304/ Fluororubber

Option material

SUS316/ Fluororubber

Converter

TRX-700-CNG

Flowmeter for Engine Cooling Water

Model

TRX-700-CNG

Function·Flow indication·Temperature indication or Total flow (continuous)

Power supply

90 to 264 V AC
·4 to 20 mA DC or 1 to 5 V DC
·RS485·Flow alarm

OutputCable length

10 m (100 m Max.)

Flowmeter for Intake AirEF series

GFM series

Detector/Converter Separate typeEF-AUTO

Model
Fluid

Detector/Converter Integrated type

OPTISONIC 7300

Model

Conductive liquids

Fluid

Min.

0 to 1 L/min

Max.

0 to 1400 L/min

Measuringrange

Temperature-20 to 180˚C

Temperature-20 to 100˚C

Pressure

0 to 1 MPa

Pressure

Atmospheric pressure

Process connection

Hose fitting 6 mm to 50 mm (1/4" to 2")

Process connection

Flange: 65 mm to 250 mm (2-1/2" to 10")

Power supply

100 to 230 V AC

Power supply

100 V AC

Output4 to 20 mA DC

Output4 to 20 mA DC

Protection category

Equivalent to IP65

Protection category

IP66

Material

Zirconia ceramic, alumina ceramic

Measuring
range

Air
Min.
Max.

12 to 300 m3/h
88 to 3530 m3/h

20

Flow Measurement System for Automobile Bench Test
Radiator Air Flow Measurement System

RF series

RF-1000 series

RF-2000 series

Propeller sensor

Propeller sensor

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

Model

RS-1038 / RS-1050

RS-1050-IR

Sensor type

Optical sensor with optical cable

Infra-Red sensor with electrical cable

Measuring range

RS-1038: 0.5 to 30 m/s, RS-1050: 0.4 to 30 m/s

0.4 to 30 m/s

Temperature

0 to 120˚C (100˚C in continuous operation)-40 to 120˚C (100˚C in continuous operation)

Accuracy

RS-1038: ± (1.5% of R.D + 0.05 m/s) at 0.5 to 20 m/s
RS-1050: ± (1% of R.D + 0.05 m/s) at 0.4 to 20 m/s± (1% of R.D + 0.05 m/s) at 0.4 to 20 m/s

Sensor diameter

RS-1038: approx. 40 mm, RS-1050: approx. 64 mm

approx. 69 mm

Converter

Converter

Model

RR-5000B

RR5000D

Power supply

12 V DC

12 to 24 V DC

Measuring channel

8 ch

16 ch

Output-5 to 5 V DC

AccuracyFrequency output: ± 1 Hz

Communication

RS485 (RS232 converter + Wind1 required)

0 to 5 V DC
Analog output: ± 0.6% R.D. ± 0.01 V

Frequency output: ± 1 Hz

Analog output: ± 0.6% R.D. ± 0.01 V

RS485, CAN network

Custom-made Product
Metal Tube Variable Area Flowmeter

AM-1000 series

Ex d
Ex i

Ex d

AM-1400

Model

AM-1520

Ex d

AM-1740

Fluid

AM-1690

AM-1310·Local indication·Totalizer·Pulse output·Local indication·Pneumatic output

Liquid, Gas, Steam

Function·Local indication·Local indication·Current output·Local indication·Alarm output

Measuring range Min.
(water)Max.

0.01 to 0.1 m3/h

Measuring range Min.
(air)
Max.

0.35 to 3.5 m3/h (nor)Process connection

Flange: 15 mm to 150 mm (1/2" to 6")Standard material

SUS304, SUS316, SUS316L

Available lining material

Rubber, fluorocarbon resin, PVC, glass

15 to 150 m3/h

450 to 4500 m3/h (nor)M series

Ex d
Ex i

Ex d

M-400

Model

M-520

Fluid
Function·Local indication·Local indication·Current output

Measuring range Min.
(water)Max.

0.01 to 0.1 m3/h

Measuring range Min.

0.3 to 3 m3/h (nor)

(air)

M-740

M-310

·Local indication·Alarm output·Local indication·Pneumatic output

Liquid, Gas, SteamMax.

80 to 800 m3/h

560 to 5590 m3/h (nor)Process connection

Flange: 15 mm to 300 mm (1/2" to 12")

Standard material

SUS304, SUS316, SUS316L

Available lining material

Rubber, fluorocarbon resin, PVC, glass

21

Nominal flow rate of variable area flowmeters
In this catalog, the following fluids are used for the measuring range of each model.
Liquid: Water with a density of 1.0 g/cm3 and a viscosity of 1.0 mPa·s
Gas: Air at 0°C, 0 MPa (1 atm)
If actual operating conditions differ from the above, correct the values with the formulas given below.
For gas measurement

For liquid measurement

Correct the value considering the density, pressure, and temperature of the measuring gas.
1. When the flow rate is indicated in the normal condition

××

273+T0273

×2. When the flow rate is indicated in operating conditions

0.10130.1013+P0

Qair = Q0

Qair : Corrected flow rate
Q 0 : Flow rate of the measuring gas in
actual conditions

×

ρ0
1.293

×

273
273+T0× 0.1013+P
0.10130

Q= Q0

Qair : Corrected flow rate
Q 0 : Flow rate of the measuring gas in
actual conditions(Flow rate in normal conditions: 0°C, 0 MPa)

ρ0 (ρ1-1)( ρ1-ρ0)

Q : Corrected flow rate
Q 0 : Flow rate of themeasuring liquidρ0 : Density of themeasuring liquid (g/cm 3)
ρ1 : Density of the float (g/cm 3)(Flow rate in operating conditions: T0°C, P0 MPa)

ρ 0 : Density of the measuring gas (kg/m 3 (nor))T0 : Fluid temperature (°C)
P0 : Fluid pressure (MPa)×ρ 0 : Density of the measuring gas (kg/m 3 (nor))T0 : Fluid temperature (°C)
P0 : Fluid pressure (MPa)

Table of float density
Float material

Fluorocarbon
resin

Glass

Ruby

PVC

Stainless steel

Titanium(equivalent to
Hastelloy C)

NW0276

Stainless steel
AM7000Density(g/cm 3)

2.2

2.67

4

1.45

7.9

4.5

8.2

7.7

Applicable
instrument

Glass tube flowmeter

Metal tube flowmeter

Note: 1. Some models have weights in the float, which increases the density.
2. Some models will be affected by fluids with a viscosity of 1 mPa·s or larger. See the respective product catalogs.

Properties of gases

Ammonia
Argon
Nitrous oxide
Nitrogen oxide
Carbon monoxide
Carbon dioxide
Sulfurous acid gas
Hydrogen chloride
Chloride
Air
OxygenCyanogenHydrogen bromide
BromineHydrogen
NitrogenFluorine
Hydrogen sulfide
HeliumMolecular Density: kg/m3 (nor) at
formula0°C, 0 MPa
NH 3
Ar
N 2O
NO
CO
CO 2
SO 2
HCℓ
Cℓ2
(AIR)
O2
C2N 2
HBr
Br2
H2
N2
F2
H 2S
He

0.7713
1.783
1.988
1.340
1.250
1.977
2.927
1.639
3.214
1.293
1.429
2.335
3.645
7.139
0.08994
1.251
1.696
1.539
0.1785

Viscosity (mPa·s)
at 0°C
at 20°C
0.00930.01000.02120.02220.01370.01460.01790.01880.01660.01770.01380.01470.01160.01260.01310.01430.01230.01320.01710.01810.01920.02030.0093–0.0170–0.01460.01530.00840.00880.01660.0175––0.01170.01240.01860.0196

Gas

Organic compounds

Inorganic compounds

Gas

Acetylene
AcetoneIsobutane
Isopropyl alcohol
Ethanol
EthaneEthyl ether
Ethylene
Ethyl chloride
Methyl chloride
Methylene chloride
Chloroform
ButanePropanePropyl alcohol
Propylene
HexaneBenzenePentaneMethanol
Methane
Methyl ether
Utility gas

Molecular Density: kg/m3 (nor) at
formula0°C, 0 MPa
C2H 2
C3H6O
C4H10
C3H8O
C2H 6 O
C2H 6
C4H10 O
C2H4
C2H 5CℓCH 3CℓCH 2Cℓ2
CHCℓ 3
C4H10
C3H8
C3H8O
C3H6
C 6 H14
C6H6
C5H12
CH4O
CH4
C2H 6 O
13A

1.171
2.593
2.595
2.683
2.057
1.356
3.309
1.260
2.880
2.308
3.792
5.329
2.703
2.020
2.683
1.879
3.847
3.488
3.221
1.430
0.71682.057
0.8407

Viscosity (mPa·s)
at 0°C
at 20°C
0.00960.01020.0066–0.00690.00740.0070–0.0075–0.00860.00920.0068–0.00940.01010.0094–0.00980.01060.00910.00990.00930.01000.00690.00740.00750.00800.0068–0.00780.00840.0059–0.00680.00740.0062–0.0087–0.01020.01080.00850.0091–0.010522

FLOW MEASUREMENT AND CONTROL INSTRUMENTS

Qair = Q0

ρ0
1.293

When the density of the measuringliquid is not 1.0 g/cm3

Float Type Level Meter
Float & Tape Type Level Meter (Tank Gauge)

FT/FP-1000 series
FT-2000 series

Ex d

FT-1000

Model
Function
Measuringrange

FP-1000·Local indication·Current output·Alarm output

Min.

FT-2000·Pneumatic output·Digital output

3m

Max.

Temperature (at the wetted parts)

30 m

10 m

30 m

-196 to 400˚C

0 to 150˚C-196 to 400˚C

PressureStandard: For low pressure applications

Process connection·Rc 1-1/2

Standard float material

Option: For high pressure applications·Flange 40 mm (1-1/2")

SUS304

Optional float material

SUS316, SUS316L, PVC

SUS316, SUS316L

LEVEL MEASUREMENT AND CONTROL INSTRUMENTS

Analog Transmitter

TR/AT series

Ex d
Model

TR-210, TR-221 to 226

Power supply / Supply air

24 V DC

Function
Output·Current output·Alarm output

4 to 20 mA DC, Alarm contact: 1 to 6 points

Ex d

TR-101 to 106

AT-101W−

Pneumatic pressure: 0.14 MPa·Alarm output·Pneumatic output

Alarm contact: 1 to 6 points

Pneumatic output: 20 to 100 kPa

Digital Transmitter

DM4N series

Ex d
Model

DM4N-1

Power supply

20 to 35 V DC

Function
Communication
Measuring range

23

DM4N-2

DM4N-385 to 264 V AC

Digital output: Level · Temperature · Alarm
One-way communication

Two-way communication
0 to 40 m, 0 to 60 m

One-way communication

Float Type Level Meter
Magnet Float Type Level Transmitter

FP-7100

Ex d
Ex i

FP-7100

Model
Power supply

24 V DC

Function

Level measurement, interface measurement

Output4 to 20 mA DC

Measuring
range

Min.

300 mmMax.

5000 mm

Temperature(at the wetted parts)·SUS + 0 to 100˚C·PVC + 0 to 60˚C
·PFA + 0 to 100˚C

Pressure

1 MPa max. (0.2 MPa max. for resin)

Process connection

Flange: 80 mm to 200 mm (3" to 8")

Wetted parts material

SUS304, SUS316, SUS316L, PVC, PFA

LT-321

FGY1000

Ex d

Ex d

LT-321

FGY1000

Model

Power supply

24 V DCPower supply

24 V DC

Function

Level measurement, interface measurement

Function

Level measurement

Output4 to 20 mA DC (2-wire system)/HART (Rev.7)

OutputMin.

250 mmMax.

7500 mm

Measuring
range

Model

Measuringrange

Temperature(at the wetted parts)·SUS: -40 to 125˚C·PVC: 0 to 60˚C
·PFA: 0 to 100˚C

Pressure

2 MPa max.

Process connection
Wetted parts material

Flange: 50 mm to 200 mm (2" to 8"),Ferrule: 2.5 S to 6.5 S
Screw: R2, G2
SUS304, SUS316, SUS316L, Titanium,
NW0276, PVC, PFA (tubing)

4 to 20 mA DC (4-wire system)
Min.

250 mmMax.

5000 mm

Temperature(at the wetted parts)·SUS: -40 to 125˚C·PVC: 0 to 60˚C
·PFA: 0 to 100˚C

Pressure

2 MPa max.

Process connection

Flange: 50 mm to 125 mm (2" to 5")

Wetted parts material

SUS304, SUS316, SUS316L, PVC, PFAMAG GAUGE Metal Tube Type Level Meter

FM series

Ex d
Ex i

Ex d
Ex i

Model

FM4000FM-1000

Function·Local indication·Alarm output·4 to 20 mA DC·Alarm output + 4 to 20 mA DC·Local indication·Alarm output·4 to 20 mA DC·Alarm output + 4 to 20 mA DC

Min.

250 mm

250 mmMax.

4900 mm0 to 4380 mm (Depends on chamber material)-196 to 400°C-10 to 120˚C (350˚C Max.)

Measuringrange

Temperature(at the wetted parts)
Pressure

40 MPa (max.)

20 MPa max.

Process connection

Flange: 25 mm (1") (standard)

Flange: 25 mm (1") (standard)Chamber material

SUS304, SUS316, SUS316L·SUS304, SUS316, SUS316L·PVC (HT-PVC)·SUS + PVC lining·SUS + ETFE lining·SUS + PFA lining·SUS + PTFE lining·SUS + Glass lining

24

LEVEL MEASUREMENT AND CONTROL INSTRUMENTS

Magnetostrictive Level TransmitterDisplacer Type Level Meter
Servo-balancing Type Tank Gauge

Torque Tube Type Level Meter

FW9000NN

FST series

Ex d
Ex i

Ex d

FW9000NN

Model

FST4000

Model

Power supply

100 to 240 V AC

Power supply

8 to 40 V DC

Function·Digital output·Current output

Function·Level measurement·Interface measurement·Density measurement

Min.

5m

OutputMax.

60 m

Measuringrange

Measuringrange

4 to 20 mA DC
Min.

300 mmTemperature (at the wetted parts)-200 to 300˚C

Pressure·Atmospheric (Low pressure version)·3 MPa max. (High pressure version)

Temperature (at the connect. part)-196 to 450˚C
ASME Class 150 to 2500

3000 mm

Flange: 80 mm to 150 mm (3" to 6")

Pressure

Low press. version

AC2A, SCS13, SCS14

Process connection

Flange: 40 mm to 100 mm (1-1/2" to 4")

High press. version

SCS13, SCS14

Displacer material

SUS304, SUS316, SUS316L, NW0276, MONEL, PVC

Process connection
Pressurizedparts material

Max.

LEVEL MEASUREMENT AND CONTROL INSTRUMENTS

Spring-balancing Type Level Meter

FS series

Ex d
Model
Function

FS-110FS-115

FS-313

FS-512

FS-517·Local indication·Local indication·Alarm output·Local indication·Pneumatic output
20 to 100 kPa·Local indication·Current output·Local indication·Current output·Alarm output

Level measurement · Interface measurement · Density measurement
Power supply / Supply air
Measuring
range

−

−

0.14 MPaMin.

300 mmMax.

3000 mm12 to 30 V DC

Temperature (at the wetted parts)-10 to 150˚C (-40 to 350˚C)

Pressure

9.8 MPa max.

Process connection

Flange: 80 mm to 125 mm (3" to 5")

Displacer material

SUS304, SUS316, SUS316L, NW0276, PVC

Microwave Level Meter
TLR series

Ex d
Ex i

TLR7400

Model
Measuring object
Frequency band

TLR3500

0.2 m

Max.

100 m

Wetted part material
Process connection

80 GHz2-wire, 4 to 20 mA DC / 24V DC

Min.

Pressure

25

TLR7500

24 GHzLiquid temperature

Ex d
Ex i

Liquid, Paste, Slurry

Output/Power supply
Measuring
range

Ex d
Ex i

0.1 m
50 m

25 m

−50 to 200°C∙ Horn antenna: 0 kPa (abs) to 4 MPa∙ Drop antenna: 0 kPa (abs) to 4 MPa
SS316L: Process connection, metal horn antenna
PTFE, PEEK: Drop antenna
G1-1/2
Flange: 40 mm to 200 mm (1-1/2" to 8")−50 to 150°C∙ Lens antenna: 0 kPa (abs) to 4 MPa∙ Lens antenna: 0 kPa (abs) to 2.5 MPa

SS316L: Process connection
PEEK: Lens antenna
G3/4 to G3
Flange: 50 mm to 200 mm (2" to 8")

Sanitary joint: 1.5S to 2S

Micropulse Level Meter
TGR/TGF series

Model

TGF7200

TGF6200

TGF1100

Measuring object

Liquid, slurry

GranulesLiquid, Slurry, Granules

Output4 to 20 mA DC

Power supply

24 V DCMin.

−·Single rod probes: 8 mm in dia. and 10 mm india. (PTFE coating)·Twin rod probe: 8 mm in dia., 2 pieces·Single cable probe: 4 mm in dia.
·Twin cable probe: 4 mm in dia. 2 pieces·Coaxial cylinders: 22 mm in dia., 42 mm in dia.

·Single cable probe: 8 mm in dia.

·Single cable: 10 m (20 m)·Coaxial: 4 m

Temperature (process)-50°C to 250°C-50°C to 200°C-50 to 100˚C

Pressure

0 kPa (abs) to 10 MPa

0 kPa (abs) to 4 MPa

0 kPa (abs) to 1.6 MPa

Wetted part material

SS316L, Hastelloy C

SS316L (excluding probe)

SS316L (excluding probe)

Probe material

SS316, SS316L, Hastelloy C, FEP, PTFE

SS316, SUS316L, PTFE

SS316, SS316L

Process connection

G 3/4 to 1-1/2, 3/4 to 1-1/2 NPT
Flange: 40 mm to 200 mm (•" to •")G 1-1/2, 1-1/2 NPT
Flange: 40 mm to 200 mm (•" to •")G 3/4 to 1, 3/4 to 1 NPT

Measuring
range

Max.

LEVEL MEASUREMENT AND CONTROL INSTRUMENTS

Ultrasonic Level MeterUW series

UW3200UW5000Measuring object

Liquid, Slurry

Liquid, Slurry, Powder, Granule, Gravel

Power supply

12 to 24 V DC

100 to 240 V AC

Output4 to 20 mA DC

4 to 20 mA DC

Measuring range

0.3 to 10 m

0 to 5 m (min.)0 to 60 m (max.)

Model

Purge Type Level Meter
CP series Purge

PGT seriesModel

CP-22-100-B

Model

Supply fluid

Air, N2

Measuring object

LiquidPGT

Supply pressure

0.3 to 0.99 MPa
0 to 1.2 L/min (std)(20˚C, 1 atm)

Max. measuring length

Standard flow range·Stainless st.: 16000 mm·PVC: 4000 mm

Process connection

Mounting type

·Panel mount type·U-bolt mount type·Box type·Dual type

·Fixed flange:
10 mm to 25 mm JIS10K·Sliding flange:
15 mm to 40 mm JIS10K

Purge connection

Rc 1/4

Panel material

SPCC, SUS304

Material

SUS304, SUS316
PVC

26

Level Switch
Float Type / Displacer Type Level Switch

FP-4000 series

Ex d
Model

FP-4000

Alarm

1 to 5 points

Ex dFP-4100

FP-4200
1 to 6 points

Max. length of guide pipe

·4.9 m for stainless steel model (3.9 m for Ex-proof model)

Contact switch·3.9 m for resin model

Reed switch

Temperature (at the wetted parts)

0 to 60˚C (-20 to 150˚C)

0 to 60˚C (-5 to 80˚C)

Pressure

0.66 MPa0.13 MPa

Process connection

Flange 50 mm (2")

Float material

SUS304, SUS316, SUS316L, PVC,HT-PVC, PP, PFA

Flange 80 mm (3")
SUS304, SUS316 SUS316L, PVC,HT-PVC, PP, PFALEVEL MEASUREMENT AND CONTROL INSTRUMENTSFB series

FR series
FS series

Ex d

Ex d
Spring-balancing
Displacer type

Float type
Model

SUS304, SUS316, SUS316L, PVC,HT-PVC, PP, PTFE, PFA

FR-6000

FS-100Alarm

1 point

1, 2, 3 or 4 points

Contact switch

Micro switch

Micro switch

Temperature (at the wettedparts, for stainless steel model)-25 to +400˚C-60 to +400˚C

Pressure

10 K, 20 K, 30 K Class

4.9 MPa max.

Process connection

Flange 100 mm (4")

Flange 80 mm to 150 mm(3" to 8")

Float material

SUS304, SUS316,
SUS316L

SUS316, SUS316L, TP340

Ex d

Model

Ex d

Float type

Float type

FB-5000

FB-7000

Alarm

1 point

1 point

Contact switch

Reed switch

Micro switch

Temperature (at the wetted parts)-5 to +90˚C-170 to +400˚C

Pressure

2 MPa max.

4 MPa max.

Process connection

R 1-1/2
Flange 40 mm (1-1/2")

Flange 80 mm (3")

Float material

SUS316

SUS316L

Capacitance Type Level Switch

Relay Driver

CA-1000 series

RD series

Ex i
Model

CA-1000

CA-1000S

Alarm

1 point

Temperature (at the wetted parts)-10 to 80˚C (-10 to 300˚C )Pressure
Process connection
Probe material

27

RD-1000

Model
Function

Amplifier with low-capacity
contacts

1 MPa max.

Combination instrument

Combination with level switches

Flange: 25mm (1")

Power supply

100 / 200 V AC

SUS304, SUS316, SUS316L

Output1 contact

Peripheral Instruments for Tank Gauging System
Receiver for Digital Tank Gauging System

CATAMS® series
NMR series

NMR5000

IFX30000

Function

Tank data monitoring software

CATAMS

Tank Data ReceiverInterface unit

Main display mode·Calibration with 12 different screens·Real-time tank data
·Data after loading/unloading·Data of all tanks·Liquid level in the bar graph·Data history·Error codes and details·Alarm record·Others∙Main menu
∙One tank∙Tank lists∙Tank status∙Tank graphs∙History∙Slot status

Model

Power supply

Depending on PC specifications

Host interface·LAN·RS-232C·Others−

Communication−∙LAN∙RS-232C∙Others·Max. 256 transmitters (2 way)
·Max. 128 transmitters (1 way)·Max. 160 transmitters(communication with third-party
products)−·For constant monitoring of a single
tank·For small tank yards with a few
tanks·Interface for host systems

24 V DC

85 to 250 V AC

·LAN
·RS-232C·RS-485·USB Type-C·Max. 32 transmitters (2 way)
·Max. 16 transmitters (1 way)·Max. 20 transmitters(communication with third-party
products)·RS-232C·RS-485 Others
LEVEL MEASUREMENT AND CONTROL INSTRUMENTS

Number of transmitters

100 to 240 V AC

DIR-530
Tank data receiver·Max. 16 transmitters (2 way)
·Max. 4 transmitters (1 way)

Tank-side Indicator

DIR series

Ex dDIR-110NN

Model
Function

Tank-side indication

Indication·Liquid level·Temperature·Other statuses

Power supply

100 to 240 V AC

OutputRelay contact (6 points)

Applicable tank gauge

FW9000NN

Temperature Sensor for Tanks

AT series

Ex d

Ex d

Model

ATM

ATS

Function

Averaging temperature sensor

Multi-spot type averaging
temperature sensor

Measuring range

0 to 100˚C

Probe length

30 m max.

Number of elements

16 max.

Process connection

Flange: 40 mm to 100 mm (1-1/2" to 4")

28

MICROCELL Weight Measurement System

Oil Leak Detector

SVS series

CSR series

Ex i
Model

SVS2000

Model

Measuring method

MICROCELL

Measuring method

CSR-3005/AS-40
Capacitance typeMeasuring object·Direct measurement of product mass in vessels·Measuring object: Silo, hopper, etc.

Measuring object

Floated oil

Power supply

100 V ACPower supply

100 / 200 V AC

Output4 to 20 mA DC

OutputContact output

Measuring range

3 to 5 mm oil layer

Level Gauge for LNG Tanks
LEVEL MEASUREMENT AND CONTROL INSTRUMENTS

Servo-balancing Type Tank Gauge

FW9000NN series

FW-2200 series

Ex d

Ex d

FW9000NN

Model

All-wire type

Power supply

100 to 240 V AC

Model

Output·Digital output·Current output

Power supply

Measuring
range

Min.
Max.

5m
60 m

Temperature (at the wetted parts)-200 to 300˚C

Pressure·Atmospheric (Low pressure version)·3 MPa max. (High pressure version)

Process connection
Pressurizedparts material

29

For lowpressures
For highpressures

Flange: 80 mm to 150 mm (3" to 6")
AC2A, SCS13, SCS14
SCS13, SCS14

FW-2200
100 V AC (standard)

OutputMeasuring
range

Digital output
Min.

30 m

Max.

60 m

Temperature (at the wetted parts)-200 to 80˚C

Pressure

2 MPa max.

Process connection
Pressurizedparts material

For lowpressures
For highpressures

Flange: 40 mm (1-1/2")
AC4C, SCS13, SCS14
SCS13, SCS14

Marine Use Tank Gauging Systems
Magnetic Float Type Level Gauge

SPT series

Ex i

Ex i

SPT3500N

Model

SPT7200N

Measuring method·Level: Magnetic float (Hall IC)·Temperature: Pt 1000 RTD (max. 3 points)·Pressure: Ceramic sensor·Level: Magnetic float (reed switch)·Temperature: Pt 100 RTD (max. 2 points)

Measuring object

Level, temperature, and pressure of liquid cargo

Level and temperature of liquid cargo

Indication

2-Line LCD with backlight(Separate indicator is available)

Output2-wire system for power supply and serial BCD signal

Measuring range

Accuracy

Optional material
Explosion proof

Resistance output·Level: 35 m max.
·Temperature: –25 to 115°C·Pressure: 80 to 200 kPa·Level: ±20 mm (standard) or ±10 mm (precision type)·Temperature: ±1.5°C·Pressure: ±0.5% F.S.
·Guide pipe: SUS304·Flange: SUS304·Float: SUS316L·Guide pipe: SUS316 / SUS316L·Flange: SUS316 / SUS316L·Float: TP340 (LPG specification)·Level: 30 m max.
·Temperature: –25 to 125°C·Level: ±1% F.S.
·Temperature: ±2°C
·Guide pipe: SUS304·Flange: SUS304·Float: SUS316L·Guide pipe: SUS316, SUS316L·Flange: SUS316, SUS316L
Ex ia IIC T6 ... T4 Ga
Ex ia IIIC T80°C, T150°C, T200°C Da

EX ia IIC T6…T4

Radar Level Gauge

LEVEL MEASUREMENT AND CONTROL INSTRUMENTS

Standard material−

High-level Alarming Device

TA series

FP/MIA series

Ex i

TA840

Model
MethodRadar

Measuring range

0.5 to 42 m

Resolution

0.1 mm

Accuracy± 10 mm

Application

For cargo tankCommunication

RS485

Standard material

SUS316L

Explosion proof

Intrinsically safe (ATEX)

FP-7091

Model

Ex i

MIA-LIDEC

MethodFloat type level switch

Acoustic wave type level switch

SensorReed switch

Piezo-electric element

Contact signal

2 points detection (high level and overflow)

Accuracy± 10 mm± 10 mm

Application·Cargo·Slop and fuel tanks·Bilge alarm etc.

Cargo tank etc.

Power supply−

Standard material·Float: SUS316·Guide pipe: SUS304, SUS316
SUS316L

Construction·Non-explosion-proof (IP65)·Intrinsically safe (Ex ia IICT6)

24 V DC (2-wire)·Sensor: SUS316L·Support: SUS304, SUS316,
SUS316L·‌Non-explosion-proof(Equivalent IP66)·Intrinsically safe (Ex ia IICT6)30

Marine Use Tank Gauging Systems
Instrument for Control Rooms

Super DIR series
ReceiverModel
LCD display

Super DIR-M3200

Super DIR-M8000

CALTIS II Windows

10.4" touch panel·10.4" touch panel·19" touch panel

Delivery: ·Pre-installed PC(or CD-ROM for software only)

Power supply
Max. number of input sensors

Connecting instrument (input)

Connecting instrument (output)

24VDC

Your computer specification

32 units (for SPT3500N level gauge or
80 units
4 to 20 mA sensors)* 32 units of SPT3500N + 48 units of* Up to 16 units if SPT3500N and
4 to 20 mA sensors
4 to 20 mA output sensors are mixed.
·SP (T) 3500N·SP (T) 7200N·TA840·Others: 4 to 20 mA level, temperature, and pressure sensors, etc.
·DIR-700-DB II·Serial communication RS232C/RS485·Ethernet (LAN) communication

LEVEL MEASUREMENT AND CONTROL INSTRUMENTS

TAN series
DIR series

Model

TAN-M1600-RP

DIR-700-DB II

Function

Annunciator unit

Power supply

24 V DC

24 V DC

Output / Indication

Output·Contact output for buzzers·CPU trouble self-diagnosis contact
output·Contact output for external alarm
lamps

Indication·Bar graph of level·Cargo data digital (Indication
selectable: level, temperature andpressure)·Alarm status·ErrorNumber of input
Input signal

31

16 Max.
·Non-voltage contact·Open collector contact

Multi-monitor for cargo tanks−
RS232C(from Super DIR-M3200 / M8000
series)·Super DIR-M3200/M8000 series·Serial communication RS232C/RS485·Ethernet (LAN) communication

Level Transmitter
FCX-AIII series

Ex d
Ex i

FKE

Model
Measuring object

LiquidFunction·Local indication·Current output

Measuring
range

Min.

0.32 kPaMax.

500 kPa

Process connection

Flange: 40 to 100 mm (1-1/2" to 4")Standard material

SUS316

Differential Pressure / Pressure Transmitter
FCX-A III series

DT series

Ex d
Ex i

Ex d
Ex i

FKC

FKG

Measuring object

Liquid, Gas, Steam

Liquid, Gas, Steam

Liquid, Gas, Steam

Measuring object

Liquid, Gas

Function·Local indication·Current output·Local indication·Current output·Local indication·Current output

Function·Local indication·Current output·Pulse output

Min.

0.1 kPa

1.3 kPa

0.32 kPa

Max.

3000 kPa

50000 kPa

500 kPa

Process connection

Rc 1/4

Rc 1/4

Flange: 15 to 100 mm(1/2" to 4")

Process connection

Rc 1/4Standard material

SUS316

SUS316

SUS316

Standard material

SUS316

Model

FKD

DT

Model

Measuring
range

Min.

1 kPa

Max.

50 kPaRELATED INSTRUMENTS

Measuring
range

Ex d
Ex i

Receiver
IR/RR series

ModelIR4600IR-6000·85 to 264 V AC
·12 to 24 V DC
·Flow rate
·Totalizer·Open collector pulse·4 to 20 mA DC·1 to 5 V DC
0 to 5 V DC (optional)·Voltage pulse·Square root extraction
function·85 to 264 V AC
·24 V DC
·Flow rate
·Totalizer

OutputPower supply for the sensor

Power supply
Function / Indication

Input

IR1700

RR930N

RR940N·24 V DC

10 to 27 V DC

10 to 27 V DC

·Flow rate
·Totalizer·Flow rate
·Totalizer·Flow rate
·Totalizer·Open collector pulse·4 to 20 mA DC·1 to 5 V DC
0 to 5 V DC (optional)·Voltage pulse·Open collector pulse·Voltage pulse·Open collector pulse·Open collector pulse·4 to 20 mA DC·Pulse·Alarm·4 to 20 mA DC·Pulse·Alarm·4 to 20 mA DC·1c contact relay output(optional)·Alarm output (flow rate/
totalizer)

Re-output of pulse

Selectable from4 to 20 mA DC or
Voltage pulse

24 V DC

24 V DC

12 V DC

12 V DC

12 V DC32

Overseas base

EUROPEMIDDLE
EAST

ASIA

NORTH
AMERICAAFRICA* Specification is subject to change without notice.

Head Office : Shiba Toho Building, 1 – 7 – 24 Shibakoen, Minato-ku, Tokyo 105 – 8558
Tel : +81-3 – 3431 – 1625 (KEY) ; Fax : +81-3 – 3433 – 4922
URL : https://www.tokyokeiso.co.jp

SG-S2032-E01

Jan 2023 K