,;$Innovative Spectrum & Signal Analyzer,;$LVWKHILHOGSRUWDEOHVROXWLRQIRUYDOLGDWLQJDOODVSHFWVRI***FHOOVLWH
GHSOR\PHQWPDLQWHQDQFHDQGPDQDJHPHQW+DQG+HOG6SHFWUXP$QDO\]HUEDVHGRQ&ORXG6HUYHUIXA is Hand-Held 5G NR enabled Spectrum analyzer based on Cloud Server, which supports autonomous measurement.
This solution offers an opportunity to reduce OPEX and CAPEX across the whole network lifecycle, from initial deploymentto maintenance and upgrades, by saving up front on equipment cost and staffing expense.

8VH,;$WR
Autonomic measurement without
human error,;$6\VWHP&RQILJXUDWLRQQR CodeConnect Wired(USB) / Wireless(WiFi)

Measurement of spurious waves
in the fieldeNB or Repeater Maintenance ServiceRadioStation

IXA
HW

IXA SW(Application)

Linkage
ServerClient PC

Instant measurement and analysis,
DB datafication)HDWXUHV%HQHILWV

Pre-Configured SA

Easy of movement and measurement
through wireless connection withtablet PC

DuplexChannel BW

High efficiency by using autonomous
measurement without human error
LTECloud Client- Economic and low operating cost
- Light Weight & Hand-Held- Spectrum Analysis (5G NR, LTE, WCDMA)- Signal Analysis (5G NR, LTE, WCDMA)- Spurious wave Measurement- Coverage Scan function- Support google map- Rechargeable battery operated- Search eNB using QR Code- Autonomous Measurement Function

Cloud Server- Upgrade Management- The Statistic Function- Set Autonomous Scenario- Autonomous Measurement Records- Predicting spurious point on the serverand sharing this point between members.
- User management- Measured Data Analysis

Measurement
Item

Duplex

WCDMA

Measurement
Item

DuplexChannel BW
Subcarrier spacing



5G NR
Measurement
Item- Channel Power- OBW(Occupied Band-Width)- ACLR(Adjacent Channel Leakage Ratio)- SEM(Spectrum Emission Mask)- SE(Spurious Emission)- Harmonic Distortion
FDD and TDD
1.4/3/5/10/15/20 MHz
Cell Info
PCI, Group ID, Sector ID
RSTP/RSRP, RSRQ,Power Info
OFDM Symbol Power
Frequency Offset,
EVM per Channel,
SignalPower per Channel,
Quality#RB per Channel,
Time Alignment Error,
Index of Antenna Path
FDD
Total Power/Total Active Power
Code
CPICH/PSCH/SSCH Power
DomainMax/Avg Active/In-active PowerRho, EVM (RMS/Peak)Modulation
Peak CDE/Peak Active CDE
AccuracyFrequency Offset/ IQ Origin Offset
TDD
5/10/15/20/25/30/40/50/60/70/80/90/100 MHz
15/30/60 kHz
PCI, Group ID, Sector ID
GSCN
Cell Info
SSB-SCS
RSTP/RSRP, RSRQPower Info
OFDM Symbol Power
Frequency Offset
Time Offset
Signal
EVM per Channel
Quality
Modulation Order#RB per Channel,;$Innovative Spectrum & Signal Analyzer

Spec item

Specification

Freq. Accuracy±0.05ppm + Aging (1ppm/year)

Freq. Range

300kHz ~ 7.5GHz

Power Accuracy

1 MHz to 7.5 GHz : ≤ ±1.3dB (±0.5dB typical) 100kHz RBW, +20°C to +30°C

Ref. Level Range-200 ~ + 100dBm

Freq. Span(MIN~MAX)

0Hz (Zero Span), 10Hz ~ 7.5GHz

RBW/VBW

RBW : 1Hz ~ 10MHz / VBW : 1Hz ~ 10MHz (1-3-10 sequence)

Detect type

Peak & Negative, Peak, sample, Negative, RMS

MarkerNumber of Marker : 6 (Normal/Delta/fix/Off )Max Input level

Average continuous power : +20 dBm

Input Attenuator(MIN~MAX)

0 ~ 60dB

Harmonic distortion

10MHz ~ 700MHz : <-40dBc (Typ.), 700MHz ~ 7.5GHz : <-70dBc (Typ.)Third Order Intercept(TOI)

100MHz ~ 3GHz : +10dBm (Typ.), 3 GHz ~ 7.5 GHz : +12 dBm (Typ.)Preamp Gain(MAX)

15dB

Preamp Reverse Filter

Built-in 34 dB LNA and Uplink Bandpass Filters

10MHz Ref. Clock

Internal, External, GPSTime Gate Trigger Source

Internal, NR SSB, External 1PPS, GPSTrace

Number of Trace : 5 (Clear Write/Average/Min Hold/Max Hold)

Wired/Wireless Communication Speed

40 /25Mbps

Temperature Range (Operating)Battery Discharging : -10 ~ +50˚C, Battery Charging : 0 ~ +40˚C

Temperature Range (Storage)-20˚C ~ +80˚C+81-3-6430-2580+81-3-6430-2589

Fax : +82 31 705 0712