Top level
thermography.
The thermal imager testo 890.

testo 890

Thermography for professionals.
Be amazed by the precision and quality of the thermal imager testo 890.

In professional thermography, not every potential measurement task can already be foreseen at the time of purchase. For
this reason, it is essential for professionals to be able to rely on an equally flexible and versatile imager which meets thehighest quality standards. testo 890 is just such an imager. It offers you the precision, image quality and flexibility you needto successfully meet all the thermographic challenges of your working day.

testo 890 in professional building analysis.
When more than a simple thermogram of a window frame
is required, testo 890 is the right measuring instrument.
The imager grants you a detailed view of the object indoors
and outdoors, allowing more precise insights into its exact
status – and, because of its flexibility, it does this evenwhen the building is too high, the roof angle too low or thefront too large.

testo 890 in industrial maintenance.
The objective of preventive maintenance is not to wait
until systems break down, but to ensure that they worksmoothly. In order to accomplish this efficiently, inspections(and their documentation) must on the one hand take as
little time as possible, but on the other hand guaranteethat even the smallest anomalies are identified reliably.
The thermal imager testo 890 enables you to do just this,becoming an indispensable companion on your next
inspection tours.

testo 890 in Research and Development.
In R&D, the thermal imager testo 890 helps you to answer
some fundamental questions: Do all components of a
new development comply with the specifications? Are
heat transfer and cooling working as they should? Are alltemperature limit values adhered to?
And these are only three of many different applications
in which the testo 890 supports you ideally, in productdevelopment as well as in fundamental research.

2

Outstanding features.
Record even better thermographic images.

In addition to the image quality, imager functions which allow you to work even more securely and efficiently are crucial for
top level thermography. This is why our engineers have developed the following technology in testo 890 especially for you.

640 x 480 pixel detector
With 307,200 measurement
points, objects are detected
clearly and precisely in excellent image quality.
SuperResolution technology to 1280 x 960 pixels
With testo SuperResolution
technology, the image quality
is improved by one class, i.e.
the resolution of the thermal images is fourtime higher.

High temperature up to
1200 °C
With the high temperature option, the measuring range canbe extended up to 1200°C .
Thermal sensitivity < 40 mKWith this excellent temperature
resolution, even the smallest
temperature differences are visible.

• Voice recording
Additional information can be
documented by voice recording directlyon site together with the thermal image.
• Intuitive operation
The imager can be operated by joystick
as well as by touchscreen.

SiteRecognition technology
For repeated thermal imagingof similar measuring objects,
the testo SiteRecognition
technology offers immediate
measurement location recognition and automatic assignment and archiving of the thermal
images.

Panorama image assistant
For large measurement objects,
the panorama image assistantallows the analysis anddocumentation of an overall
image stitched together from several individual
images. This means that there is no need tocompare and administer several images.

Autofocus/manual focus
The autofocus creates a sharp
image automatically, so thatthe testo 890 can be used with
one hand. There is also thepossibility of focussing manually.

Exchangeable lenses.
With the standard 42° wideangle lens, large image sections are recorded, and withthe high-quality 15° telephotolens and the high-performance 5° super-telephoto lens, small details can be measured
even at a greater distance.

Display of surface moisture
distribution
For every measurement point,
the value of the relative surface humidity is displayed
according to a traffic light principle, in order
to be able to evaluate the danger of mouldquickly and easily.

Minimum focus distanceWith a minimum focus distance of 10 cm, measurements
at macro level can be takenfrom a close distance, in order
to measure very small objects.

Process analysis package
Thanks to image sequence
capturing in the instrument, and fully radiometricvideo measurement, thermal processes can berecorded, streamed to a
PC, and analyzed as time progressions.

Laser marker
With the laser marker, a laser
spot can be indicated on themeasurement object for orientation purposes. This laser
spot is also displayed parallax-free in theinfrared image.

AF• Digital camera with power LEDs
The integrated digital camera recordsa real image parallel to the thermal
image. The power LEDs assist theillumination of dark areas.

• Solar mode
In solar mode, the solar radiation
value can be entered in the imager.
This value is stored with every thermal
image, and is then available in theanalysis software testo IRSoft.
• Digital zoom
Using the digital zoom, you can enlarge
the infrared image details in the display.
This helps optimum focusing and onsite fault analysis.
• Fold-out display and rotatablehandleThe rotatable, fold-out display andthe ergonomic rotatable handle allow
measurements in difficult-to-accessplaces.
• IRSoftWith the high-performance analysis
software testo IRSoft, you cananalyze images quickly and easily,and document them in a report. It
is included in delivery, and can beinstalled licence-free on an unrestricted
number of computers.
• Calibration
The testo 890 is delivered with a works
calibration certificate as standard. ISO
certificates are also optionally available.

3

testo 890

Highest image quality.
Every pixel counts.

The heart of the thermal imager is the detector. For thisThermal image with
160 x 120 pixels

reason, we place great value on the best quality. An
extremely high-quality detector with 640 x 480 pixels is

35 °C

fitted in the testo 890. That means 307,200 measurement
values, which in combination with the special Germanium
optics, ensure the highest level of image quality. With the
testo SuperResolution technology, the testo 890 even takes
extremely high-resolution thermal images in megapixel

Thermal image with
320 x 240 pixels
45 °C

quality (1280 x 960 pixels; 1,228,800 measurement values).
This means that you can also thermograph the smallest or
most distant measuring objects extremely accurately.

Thermal image with
640 x 480 pixels
52 °C

Thermal image with 640 x
480 pixels, in combination
with testo SuperResolution
technology in megapixel
quality.
—> 1280 x 960 Pixel

4

60 °C

The more sensitive the better.

A further important influencing parameter on the quality ofthermal imager is able to detect and visualise. Thermala thermal image is the so-called NETD (Noise Equivalentsensitivity is usually described in °C or mK. With an

Temperature Difference) which describes the thermal

excellent NETD of < 40 mK, the testo 890 can resolve evensensitivity, and thus the temperature differences which

the smallest temperature differences, and present them in

an imager can identify. The better the thermal sensitivity,

the thermal image.

the smaller the minimum temperature difference that theThanks to the outstanding thermal sensitivity of the testo 890, all details are identifiable in this wall.

The combination of high-resolution 640 x 480-pixel detector and Testo SuperResolution technology, andthe excellent NETD of < 40 mK means that you are optimally equipped for every measurement task withthe thermal imager.

5

testo 890

Look behind the facade.
Building thermography: Provide energy consultation, discover building faults.

In professional building analysis, thermography with theWith the thermal imager testo 890, you can quickly andthermal imager testo 890 supports you when carrying out

efficiently discover and impressively visualize thesecomprehensive diagnosis and maintenance work.

thermal anomalies. By additionally creating professional· Analysis of energy loss in the heating or air conditioning ofthermography reports, you emphasize your expertise in thebuildings

field of building energy with the help of the imager.

· Visualization of faulty insulation and thermal bridges· Recording and documentation of energy losses at outer
windows and doors, roller blind casings, radiator recessesor roof constructions

Every detail counts.
When examining a building shell for thermal bridges, as wellas in the evaluation of moisture damage and its effect on
the heat transfer coefficient of a wall, the identification ofvery small temperature differences is of the essence. For
this reason, a thermal imager used for these purposes mustpossess a correspondingly good thermal sensitivity (NETD).
The NETD of the testo 890 is < 40 mK. That means that you
can visualize even the smallest anomalies.
Horizontal and vertical thermal bridges in a building.

Even more versatile thermography.
The 42° wide-angle lens is advantageous for recording
a large image section in the confines of indoor rooms. It
also helps to take thermographic images of a building shellfrom a short distance when recording outdoors. The 15°
telephoto lens is in many cases also indispensable, e.g. forthe examination of abnormalities on roofs or when making
thermal images of multi-storey buildings.

The telephoto lens identifies hotspots even from a great distance.

6

Image of a building with a 42° (left) and a 25° (right) wide-angle lens . As you can see, more detail is visible.

Problem-free thermography of large objects.
Thermographic recording of the entire shell of a building insuch a way that one obtains an overall presentation as well
as being able to identify the relevant details, can present a
problem even for professionals.
Common problems often include the spatial limitations
created by walls, streets or the security zones ofneighbouring objects. These can make the recording of alarge measurement object in only one image impossible.
The panorama image assistant of the testo 890 supports
you in meeting this challenge: It stitches up to three time
three images together into one thermal image, in which

Complete building front in a panorama image.

one can see thermal irregularities in detail over the entire
building surface at a glance.
This function supports you in conducting energy
consultation efficiently, and impressively demonstrating
to your customers at a glance various possibilities for the
optimization of the energy efficiency of a building.

The panorama image assistant guides you through the individual recordings.

7

testo 890

Fast identification of indoor moisture.
In context of indoor building thermography, the visualization of potential damp spots is of great significance, as it ishere that mould growth can occur. Testo's patent-pending"moisture image" shows the relative surface moisture foreach individual measurement value in the thermal image,and presents the various danger zones according to aneasy-to-follow traffic light principle:
Green (< 65 % RH): no mould danger.
Yellow (> 65 to < 80 %RH): possible mould danger.
Red (> 80 %RH): acute mould danger.
Indoor room with acute danger of mould under a pitched roof.

Analyzing thermal images professionally.
For the thermography of buildings, a high-performance
software is also crucial. Because without it, thermograms
cannot be quickly and easily analyzed, evaluated anddocumented in a report.
The licence-free software testo IRSoft was speciallydeveloped for these requirements. In addition tocomprehensive analysis functions and an intuitive operation,
it above all offers numerous options for creating individual
and norm-compliant reports for your own documentation or
your customer. See page 15 for more information.

Professional thermography reports – created with the analysis software
testo IRSoft.

The display of the surface humidity distribution can be optionallycarried out by manual input of
the two parameters temperatureand relative humidity. These
measurement values can be determined with the help of a humidity
measuring instrument. The testo
625 is especially suitable for thispurpose.

8

With the thermal imager testo
890-2, the humidity measurement
can be carried out optionally using
an external wireless humidity
probe. The measurement values
are automatically transmitted tothe imager in real time. Manual
input is not necessary. In addition
to this, the current measurement
values are saved with the image.

So that wear does not become a problem.
Professional maintenance with the thermal imager testo 890.

Ensuring the availability of industrial plants is a great

temperature, not only in mechanics but also in electronics,

responsibility. Because in order to work efficiently andthe thermal imager testo 890 helps you toeconomically, industrial plants are usually operated around· optimize system security

the clock, 7 days a week, 365 days a year – if possible· reduce downtimes

without malfunctions, or even breakdowns. Since problems· lower energy costs

and critical load stati are usually preceded by an increase in· improve product quality· lower maintenance costs.

Visualizing increased electrical resistance.
In order to ensure that an overheating of small components
is not missed in the testing of switching cabinets, it isimportant to use an imager with as high a resolution as
possible. The testo 890 has an infrared resolution of 640 x
480 pixels, and using the testo
SuperResolution technology, iteven accomplishes images in
megapixel quality.

Conspicuous terminal on a contactor (left) in a switching
cabinet.

Precise thermography from a distance.
If, for reasons of safety (e.g. in high-voltage or high-temperature systems) or local peculiarities, it is necessary to takethermographic measurements from a greater distance, thethermal imager must support you here also. In these cases,
you can rely on the telephotoand the super-telephoto lenses
when working with the testo
890, identifying details preciselyfrom a distance.
Telephoto image: Overheated bearing of a transport conveyor belt under the ceiling of a hall.

9

testo 890

Test large plants easily.
When very large objects
need to be recorded thermographically, or when it is
not possible to increase thedistance to the measurement object, you need thepossibility of recording alarge image section. For thisreason, the testo 890 has
a standard 42° wide-angle
Examining a tank at an oil refinery for filling level,
deposits, cooling system and material status.

lens. If this is not sufficient,
the panorama image assistant supports you by creating a large panorama image
from up to three times three

Panorama image from two individual images of the feed-in of arotary kiln.

images. This then allows
overview and detail in one
image.

Industrial kiln with faulty insulation.

testo SuperResolution

Four times as many readings and a

The higher the resolution of yourgeometrical resolution that has been

thermal images, the more anomalies

improved by a factor of 1.6 mean

you can identify. The revolutionary

even greater detail and even greater

SuperResolution Technology will

reliability of measurement for you.

instantly improve the image quality
without testo
SuperResolution

10with testo
SuperResolutionof your thermal imager by one class.

Monitoring plant stati even more efficiently with
Testo SiteRecognition.
For the status-oriented maintenance of plants andmachines, it is important always to be informed of theirstatus, in order to be able to identify changes in time. The
SiteRecognition of testo 890 provides support. You can use
it to create a measurement site archive in the testo IRSoft
analysis software which serves as a database for yourthermal images. For every measurement site stored in thearchive, you can create markers (small symbols similar to
QR codes), and attach them on site. In the next inspection,
you simply record this marker with the SiteRecognition
assistant of the imager, which then automatically storesthe measurement site and the corresponding information

Database structure of the SiteRecognitionarchive in testo IRSoft.
testo SiteRecognition
measurement site marker

together with the thermal imager. When you transfer
these thermal images to the analysis software after themeasurement, they are then fully automatically sorted into
the archive. There is no longer any need for time-consuming
administration and archiving. You can then conveniently,
directly out of the archive, open the images, analyze them(for example to make comparisons or identify negative
trends), or process them in reports.

Electric motor for driving a pump.

Electric motor running at different loads.

11

testo 890

See everything, develop quicker.
Research and Development with testo 890.

In Research and Development, wherever temperatures

Very early on in the development process, checks can because a desired or undesired effect they can be examined

carried out as to whether limit values and the quality stipu-

without contact using the thermal imager testo 890, and if

lations and specifications based on them are adhered to, or

necessary optimized. Particularly in demanding develop-

whether production processes have already reached serial

ment tasks, early analyses (e.g. of temperature distribution)

status. This saves you money and time, and you can avoid

are very useful.

unnecessary correction loops.

Identifying finest structures.
The thermal imager testo 890 has a high-resolution detector
and a standard lens with macro characteristics. This meansthat with a minimum focus distance of only 10 cm, you can
examine even the smallest structures of only 113 µm. If theimager is used in hand-held operation, the testo SuperResolution technology even allows the recognition of 70 µm
structures.

Circuit board with SMD-equipped components.

Injection-moulded plastic part with fine moulding lines.

Smallest identifiable object
with 42° standard lens:12

Cooling thin thread after industrial
yarn extrusion.

Distance

0.3 m

0.25 m0.2 m0.15 m0.1 m

without testo SuperResolution
with testo SuperResolution

0.34 mm0.21 mm0.28 mm0.18 mm0.23 mm0.14 mm0.17 mm0.11 mm

0.113 mm0.07 mm

Optimizing thermal management.
In the context of thermal management, thermal imagers canbe very useful for defining suitable cooling measures. Thiscan reach from the installation of simple heat sinks to complex active refrigeration methods. However, if it is necessary to identify temperature changes as time progressions,such as in the examination of thermal processes, a static
thermal image is not enough.
This is important for example, when the mould releasetemperature and the cooling behaviour of injection-moulded
plastic parts after mould release need to be analyzed. In

Heat sink for an LED module.

electronics too, it is important to examine components orassemblies in operation with different degrees of load. Withthe testo 890, you can detect thermal developments reliably
and precisely.

Heat sink in a TwinPix image.

Temperature analysis of a voltage regulator as a time progression

13

The process analysis package from Testo.
The optional process analysis package consists of the fullyradiometric video measurement and the image sequence
capturing directly in the imager. This allows you to recordand store image sequences directly on site without
connection to a PC, and analyze them in the analysis
software testo IRSoft.
Fully radiometric video measurement at a PC:· Data streaming to a PC for the recording of fast-moving
processes· Remote control from a PC for applications at test benches· Analysis of stored sequences and videos
Image sequence capturing in the instrument:· Storage directly in the instrument without cables· Can also be operated without a PC

Image sequence capturing in the imager.

Fully radiometric video measurement

Thanks to the newly developed image sequence capturing,With the fully radiometric video measurement, you haveyou can record temperature developments at certain freely

on the one hand the possibility of streaming radiometricselectable intervals directly in the thermal imager testo 890,

measurement data by up to 25 Hz* into the the analysis

and save them either as fully radiometric video, (.vmt) or assoftware testo IRSoft in online measurement and ofa sequence of thermal images (.bmt) with the corresponding

recording sequences. This means that all changes are

optional real images. The smallest interval is three seconds,

immediately visible in the thermal image, even in fast-

the longest is one hour and 59 seconds. Up to 1170 thermal

moving processes. The video measurement moreover offers

images can thus be stored in the imager, depending on thethe option of controlling the imager remotely. Individual

format selected. For applications with the image sequence

images can additionally be saved by remote trigger as

capturing, the imager furthermore has a synchronizedthermal images or as JPEG files.

automatic shutter, so an internal adjustment always takes

Stored videos and sequences which you have previously

place at the right time before recording an image. This

recorded in the imager can furthermore be clearly analyzed.

means that an optimum image quality is always achieved

For this purpose, the many useful functions of the analysis

even in very long recordings. The recording of image

software testo IRSoft are available to you:sequences can be started in different ways:- Setting of up to 15 measurement points which can be- Manually, in order to begin recording immediately.
- After a limit violation, in order to check that it is being
adhered to.
- By a countdown, in order to begin after a certain time.

presented as a temperature-time diagram.
- Setting of up to five profile lines, in order to examine thetemperature profile of measurement objects.
- Automatic hot/cold spot recognition, in order to identify
conspicuous temperature values immediately.
* Within the EU and for countries without export restrictions, otherwise 9 Hz

14

Analysis, evaluation, documentation.
With the professional analysis software testo IRSoft.

Thermography at the highest level needs more than justa modern imager system. A high-performance analysis
software is also crucial in order to quickly and easily
analyze and evaluate recorded images, and to present themto your customer professionally. The high-performance
analysis software testo IRSoft was specially developed
for these requirements. It offers comprehensive analysis
functions, intuitive operation and a high level of userfriendliness. The software is included in delivery of all
Testo thermal imagers, and can be installed licence-free onan unrestricted number of computers. And the best part:
regular upgrades can be downloaded online free of charge.

Creating and individually adapting reports.
The report designer integrated into the Testo IRSoft offersa broad selection of different templates from which you canselect your desired report. Whether brief and to the point,
or comprehensive and detailed – you are guided throughthe creation step by step, and can yourself select which
information you wish to use. A report template according to
DIN EN 13187 is available to you specially for the analysis
of building shells for thermal bridges. All reports can be
simply saved as a PDF, RTF, or in Testo's own TIR format.

testo IRSoft – overview of all advantages.
• Licence-free online updates free of charge• Extensive analysis functions• testo SiteRecognition (automatic measurement site
recognition)• TwinPix (real and thermal image overlay)• Fully radiometric video measurement• Panorama image function• Event-based trigger• Emissivity correction• SuperResolution

15

testo 890

Technical data.
testo 890-2

testo 890-1
Infrared image output
Infrared resolution

640 x 480 pixels

Thermal sensitivity (NETD)< 40 mK at +30 °C

Lens version

Standard

Standard

Tele

Super-tele

Field of view/min. focusing distance

42° x 32° /0.1 m

42° x 32° /0.1 m

15° x 11° /0.5 m

6.6° x 5° /2m

1.13 mrad

0.42 mrad0.18 mrad

Geometric resolution (IFOV)

1.13 mrad
1280 x 960 Pixel /0.71 mrad

SuperResolution (pixels/IFOV) - optional

1280 x 960 Pixel / 1280 x 960 Pixel / 1280 x 960 Pixel /0.71 mrad0.26 mrad0.11 mrad

33 Hz*

Image refresh rate

Automatic/manual

Focus

7.5 to 14 µm

Spectral range
Image output visual
Image size / min. focusing distance

3.1 MP / 0.5 m

Image presentation
Image display

4.3" LCD touchscreen with 480 x 272 pixels

Digital zoom

1 to 3 x

Display options

IR image/real image

Video output

USB 2.0

Colour palettes

9 (iron, rainbow, rainbow HC, grey, inverted grey, cold-hot, blue-red, sepia, Testo)

Measurement
Measuring range-30 to +100 °C / 0 to +350 °C (switchable) / 0 to +650 °C (switchable)

High temperature measurement - optional–+350 to +1200 °C(not in connection with the telephoto lens)Accuracy±2 °C, ±2 % of m.v.

Emissivity / reflected temperature setting

0.01 to 1 / manual

Transmission correction (atmosphere)
Measuring functions
Display of surface moisture distribution(via manual input)–

Humidity meas. with wireless humidity probe**(automatic meas. value transfer in real time)

–

(

)

Solar mode
up to 10 measurement points, Hot/Cold Spot Recognition,
up to 5 x area measurement (min/max & average), Isotherm and alarm values

Analysis functions
Standard

16() optional– Not availabletesto 890-1

testo 890-2

Standard lens 42° x 32°

Standard lens 42° x 32°
Telephoto lens 15° x 11°
Super-telephoto lens 6.6° x 5°

Imager equipment
Digital camera with power LEDs
LensesSiteRecognition(meas. site recognition with image management)–

Panorama image assistantLaser*** (laser classification 635 nm,
Class 2 )Laser marker

Speech recording

Bluetooth**** / wired headset–

Up to 3 measuring points

Video measurement (via USB)
Process analysis package: image sequence
capturing in instrument and fully radiometricvideo measurement
Interface–()

LabVIEW, interface description download on the Testo homepage

Image storage
File format individual image.bmt, possible export to .bmp, .jpg, .png, .csv, .xls

Video file format (via USB).wmv, .mpeg-1 / Testo format(fully radiometric video).wmv, .mpeg-1

Storage device

SD card 2 GB (approx. 600-700 images)

Power supply
Battery typeFast-charging, Li-ion battery can be changed on-site

Operating time

4.5 hours

Charging options

In instrument/in charger(optional)

Mains operation

yes

Ambient conditions
Operating temperature range-15 to +50 °C

Storage temperature range-30 to +60 °C
20 % to 80 % non-condensing

Air humidity
Housing protection class (IEC 60529)

IP 54

Vibration (IEC 60068-2-6)

2G

Physical features
1630 gWeightDimensions (L x W x H) in mm

253 x 132 x 111

Tripod mounting

1/4" - 20UNC
ABS

Housing
PC software

Windows 10, Windows Vista, Windows 7 (Service Pack 1), Windows 8,
USB 2.0 interface

System requirements
Standards, tests, warranty

2004/108/EC

EU Directive

2 years

Warranty
Standard() optional– Not available* Within the EU and for countries without export restrictions, otherwise 9 Hz** W ireless humidity probes only in the EU, Norway, Switzerland, USA, Canada, Colombia, Turkey, Brazil, Chile, Mexico, New Zealand, Indonesia*** Excepting USA, Japan and China**** Bluetooth only in the EU, Norway, Switzerland, USA, Canada, Colombia, Turkey, Japan, Russia, Ukraine, India, Australia
17

testo 890

Overview of versions.

Featurestesto 890-1

Infrared resolution

testo 890-2

Set testo 890-2

640 x 480 pixels

Thermal sensitivity (NETD)< 40 mK

Image refresh rate

33 Hz*

Measuring range-30 to +650 °C(

SuperResolution

)

(

)

(

)

Exchangeable telephoto lens 15° x 11° *****

–

(

)

Super-tele 6.6° x 5° *****

–

(

)

–

(

)

(

)

(

)

(

)

(

)

(

)

Auto-focus
High temperature measurement up to 1,200 °C
Panorama image assistant–

Site recognition with image management
Laser marker**
Display of surface moisture distribution (via manual input)–

Humidity measurement with wireless humidity probe*** (automatic
measurement value transfer in real time)–

Speech recording using headset****–Process analysis package: Image sequence capturing in instrument andfully radiometric video measurement–

Solar mode
Lens protection glass

(

)

(

)

Additional battery

(

)

(

)

Fast battery charger

(

)

(

)

Included in delivery() optional– Not available* Within the EU and for countries without export restrictions, otherwise 9 Hz** excepting USA, China and Japan*** W
 ireless humidity probes only in the EU, Norway, Switzerland, USA, Canada, Colombia, Turkey, Brazil, Chile, Mexico, New Zealand, Indonesia**** Bluetooth only in the EU, Norway, Switzerland, USA, Canada, Colombia, Turkey, Japan, Russia, Ukraine, India, Australia***** depending on the selected set

18

Ordering data.

testo 890-2 sets with yourselection of lenses
Complete sets in a robust case, including pro software, SD card, USB cable, carrying strap, lenscleaning cloth, mains unit, Li ion rech. battery,lens protection glass, spare rech. battery, fast
charger, headset and lens case.
Order no.
testo 890-2 set with standard and telephoto lens – see above for further set components

0563 0890 V3

testo 890-2 set with standard or telephoto, and super-telephoto lens – see above for further set components

0563 0890 V5

testo 890-2 set with standard, telephoto and super-telephoto lens – see above for further set components

0563 0890 V6

testo 890 thermal imagers

Order no.

Thermal imager testo 890-1 with standard lens in a robust case incl. pro software, SD card, USB cable,
carrying strap, lens-cleaning cloth, mains unit and Li ion rech. battery

0563 0890 V1

Thermal imager testo 890-2 with standard lens in a robust case incl. pro software, SD card, USB cable,
carrying strap, lens-cleaning cloth, mains unit, Li ion rech. battery, headset

0563 0890 V2

Thermal imager testo 890-2 with super-telephoto lens in a robust case incl. pro software, SD card, USB
cable, carrying strap, lens-cleaning cloth, mains unit, Li ion rech. battery, headset

0563 0890 V4

Code1)(First equipment)

Order no.
(Retro-fit)

SuperResolution. Four times more measurement values for even more detailed
analysis of the thermal images.

S10554 7806

Lens protection glass. Special protective glass for optimum protection of the lensfrom dust and scratching.

F10554 0289

Additional battery. Additional Lithium ion rechargeable battery for extending theoperating time.

G10554 8852

Fast battery charger. Desktop charging station for two rechargeable batteries for
optimising the charging time.

H10554 8851

High temperature measurement up to 1200 °C

I1

2)

Humidity measurement with wireless humidity probe 4)

E1

2)

Exchangeable telephoto lens 15° x 11°

D1

2)

Super-telephoto lens 6.6° x 5°

T2

2)

Process analysis package
Image sequence capturing in instrument and fully radiometric video measurement

V10554 8902

Accessories

Emission tape. Adhesive tape e.g. for shiny surfaces (roll, L.: 10 m, W.: 25 mm), ε = 0.95, temperature-proof
up to +250 °C

0554 0051

ISO calibration certificates3)

Calibration points at 0 °C, +25 °C, +50 °C

0520 0489

Calibration points at 0 °C, +100 °C, +200 °C

0520 0490

Freely selectable calibration points in the range -18 to +250 °C

0520 0495

1)

When ordering as first equipment, you receive the accessories directly in the case.
Example: testo 890-1 incl. lens protection glass and SuperResolution: Order no. 0563 0890 V1 F1 S1

2)

Please contact our customer service

3)

Per lens4)

Wireless humidity probes only in the EU, Norway, Switzerland, USA, Canada, Colombia, Turkey, Brazil, Chile, Mexico, New Zealand, Indonesia

19Subject to change without notice.

0981 4024/msp/I/07.2016

testo 890

www.testo.com