MODEL 3330 G-LOGGERDATA ACQUISITION SYSTEM

•
••••
•••
•
•

•

Compatible with Silicon Designs accelerometers
No additional accelerometer power supply needed
Automatic and manual calibration routineo +/-1G Flip or manually enter from calibration certificate
Adjustable filters and FFT for data analysis in both real time orpost collection
Three input channels support three 1- axis modules or one 3axis module
16 Bit sample rates from 1 to 10,000 samples/second per axis
Recording feature with playback in multiple speeds
Available preconfigured if purchased with a new Silicon Designs
accelerometer module
Automatic setup in less than 5 minutes includes bias, scale
factor, etc.
Last configuration stored in memory and available remotely
Included software features familiar and convenient userinterface built using a LabView platform

SPECIFICATIONSPHYSICAL
Case Size
WeightCase Material

5.5" x 4.25" x 1.5"
275 grams / 9.5 oz.
Die Cast Aluminum, Plastic

OPERATIONAL
Connection
USB Connection
Memory TypeMax SD Card Size

ENVIRONMENTAL
Operating Temperature
Storage Temperature
Humidity

PC REQUIREMENTS
25 Pin Female D-Sub
Operating SystemsMicro USB (B)
Host Connection
SD Card, Micro SD w/ Adaptor Power Supply
32 GB
Max Power Consumption
TCP/IP Remote Operation

0°C to +55°C (max)-40°C to +85°C (max)
0% - 90% Non-condensing

Windows 10, 8, 7, XP
USB2 Type A
Via USB Cable
750 mW
Network Connection Req.

ZERO (DC) TO MEDIUM FREQUENCY APPLICATIONS







ADDITIONAL FEATURES• Real time data monitoring• Collect data in G or volts• Display shows from 100ms to 2 minutes of data
• FFT (Fast Fourier Transform) analysis is anadvanced feature usually found on much more
expensive DAQ systems• Independent scale G/Div settings expand or shrink
each channel’s input for better visibility• collection• PAUSE, RWD, FWD without interrupting data
• Optional offsets provide a staggered display for no overlap• Independent scale G/Div per channel• Oscilloscope (Sweep, Scope, Strip) and Volt Meter modes• Hide or show any or all of the 3 channels• View data from remote locations on network via TCP/IP• Optional offset setting per channel• Export time–stamped data to Excel, MatLab etc.

^W/&/d/KE^^h:ddK,E't/d,KhdEKd/^ŝůŝĐŽŶĞƐŝŐŶƐ͕/ŶĐ͘ͻϭϯϵϬϱEϭϮϴƚŚ^ƚƌĞĞƚ͕<ŝƌŬůĂŶĚtϵϴϬϯϰͻWŚŽŶĞ͗ϰϮϱͲϯϵϭͲϴϯϮϵͻ&Ăǆ͗ϰϮϱͲϯϵϭͲϬϰϰϲǁǁǁ͘ƐŝůŝĐŽŶĚĞƐŝŐŶƐ͘ĐŽŵ΀ƉĂŐĞϭ΁ϭϱͲ:ƵůǇͲϮϬϮϬ

MODEL 3330 G-LOGGERDATA COLLECTION

Data can be collected live or recorded for playback later. Modifying the horizontal scale expands or contracts theperiod of time displayed on the screen from 100ms up to 2 minutes. Each axis is one channel, and these can be
hidden or offset (but will still be recorded) as desired.

SPECTRUM (FFT)
SPECTRUM displays the FFT of the data. This is a more advanced feature of the G-logger 3330. You can analyze
the data to see at which frequencies the maximum vibrations are occurring.

SPECIFICATIONS SUBJECT TO CHANGE WITHOUT NOTICE
Silicon Designs, Inc. • 13905 NE 128th Street, Kirkland WA 98034 • Phone: 425-391-8329 • Fax: 425-391-0446
www.silicondesigns.com[page 2]15-July-2020

MODEL 3330 G-LOGGERPEAK VALUESThe METERS screen provides DC, RMS, and peak values in either Volts or Gs. These values are calculated over thetime interval selected by the horizontal scale selected on the Data tab. The values are updated at that same interval
as well.

CALIBRATE
The default calibration parameters are supplied automatically, or unit-specific calibration parameters can be
manually entered. Manual calibration can be done any time using gravity and performing a simple +/-1G flip.

SPECIFICATIONS SUBJECT TO CHANGE WITHOUT NOTICE
Silicon Designs, Inc. • 13905 NE 128th Street, Kirkland WA 98034 • Phone: 425-391-8329 • Fax: 425-391-0446
www.silicondesigns.com[page 3]15-July-2020

MODEL 3330 G-LOGGERCONNECTOR PIN LAYOUT

TOP ROW PIN NUMBERS
123CH 0CH 0CH 10 Volt
AON0 Volt
Bottom Row Pin Numbers

4CH 1AON5CH 20 Volt

6CH 2AON

14CH 0AOP

17CH 18-32 V18
CH 2AOP19
CH 28-32 V

15
CH 0
8-32 V

16CH 1AOP

78910111213X

X

X

X

X

X

X

202122232425

X

X

X

X

X

X

Included 25 Pin D-Sub
Connector for Accelerometer
Connection

CABLE WIRINGSPECIFICATIONS SUBJECT TO CHANGE WITHOUT NOTICESilicon Designs, Inc. • 13905 NE 128th Street, Kirkland WA 98034 • Phone: 425-391-8329 • Fax: 425-391-0446
www.silicondesigns.com[page 4]15-July-2020