WMP-15A

Medical

Medical Grade Panel PC 15 inch Series
NEWKey Features● Intel Broadwell Fanless Medical Panel PC● 15.6" Wide 16:9 1366x768 LED panel● P-CAP multi-touch with 10 fingers● Medical UL60601-1 3rd edition certification● Anti-bacteria (MRSA) plastic housing● Friendly design of front function hot-keys● Optional PCI-E/PCI expansion slot
● Optional 4KV LAN/COM/USB isolation I/O● Optional built-in battery pack

Food
Processing

Industrial

Box PC

Outline Drawing

75
100300

6.25
4KV

PCI-EIsolationExpansion

CPUBroadwell83

Main System

I/O & Power & Control Keys

Options

CustomizationCPU

FCBGA1168 5th generation Intel® Corei7/i5/i3 processor (15W max.)USB

USB 3.0 x 2 , USB 2.0 x 2 (Default)

802.11 a/b/g/n WLAN module (Mini PCIe)

USB 5,6 or COM 3,4Chipset

SoC

COM

RS232 x 1, RS232/RS422/RS485 x 1(RS-485 auto flow support)

802.11 a/b/g/n WLAN + Bluetooth 4.0 module (Mini PCIe)

Tempered glass

VGAMemoryIntel® integrated HD Graphics 5500 by CPULAN

RJ-45 x 2, Gigabit

5M pixel AF webcam module (USB)

RFID reader (13.56Hz) (USB)Video-out

DP x 2Two 1600 MHz DDR3L SODIMM
socket support dual Channel, non-ECC,up to 16GB

Speaker

2Wx2

LCD PanelSize

15.6“ wide 16:9

Resolution

1366x768

Color numbers

16.7M colors (RGB 6-bit + Hi FRC)

Viewing angle

170 (H) / 160 (V)Contrast

500:1 (Typ)Touch PanelType

Full flat projective capacitive touch panel(10 fingers)

Light transmission
Hardness
Life time

Audio
Power input

Line-out, Microphone-in (3.5 mm phone-jack)Power adaptor

AC input 90 ~ 264V / 47~63 Hz / DC output 12V,
medical grade, 1.2m DC cable

DC-in x 1, DC 7~32V input (9~28V with battery pack)

Power consumption
Control keys

78.2 W ( MAX. loading without adapter )Power, audio volume, brightness, backlight on/off,
Clean me, webcam (disable), programmable button

Mechanical & Environmental
Housing

90% ± 3%

•Front: Full flat design / Touch (standard),
Glass (Customization)•Rear: Plastic + I/O cover w/ antibacterial treatment•Anti-bacteria (MRSA)
IP65 – front side, IPx1 whole system

IP rating

100M times

Cooling
Dimension

430 (H) x 300 (V) x 83 (T) (mm x mm x mm)

Weight (net)

5 kg (w/o power adapter)

Temperature

Operating : 0°C~35°C (chamber)
Storage : -20°C~60°C

Humidity

Operating & storage – 10%~90%,
non-condensing

Regulatory•FCC/CE/VCCI Class B•cULus Mark/UL60601-1•EN60601-1 approved / ISO 14971 / IEC-62133•Energy star 6.1 complianceHDD

2.5” SATA HDD drive bay x 1 ,Expansion slot

•PCI-E (x4) expansion slot x 1(25W max, can't co-exist with isolated module)•Card size: 100mm (W) x 160mm (D) x 14mm (H)•Mini-PCIe x 2 (52 pin card-edge type)

Ordering Information
WMP-15A (i5)15" FANLESS

i5-5200U/4G/500G/P-CAP

WMP-15A (i7)15" FANLESS

i7-5500U/4G/500G/P-CAP

All information is subject to change without prior notice.

Fanless

1Battery pack
PCI-E x 4 riser card

7H pencil, pressure 750g/45°

Storage & Expansion

23430

Power adaptor w/ 2.4m or 3.6m DC cable910

Anti-bacteria desktop stand (white, L type)
Equipotential terminal pin
AG film86

TPM v1.2 module
mSATA module

74KV Isolation Card x 1(Isolation COM x 2, Isolation LAN x 1, Isolation USB x 1)

COM x 2Line-out
Micropone-in
LAN x 2
USB 2.0 x 2

9USB 3.0 x 2
DP x 28 Reset9 DC-in
10 PCI-E/PCI expansion

4KV Isolation I/O

Medical Panel PC Series

24