MAGMAX®

EGM1300C
Compact Electromagnetic Flowmeter
GENERALMAGMAX® EGM1300C is a combination of EGS1000 primaryhead with PFA liner and high performance converter EGC300.
EGM1300C aims “Low cost - High performance.”
An excitation system extendable up to twice the commercial frequency has been introduced to reduce fluid noise.
And improved self-diagnostic functions include vacancy detection
and detection unit monitoring.
High quality, virgin, clear PFA lining with Hastelloy® C electrodes
covers almost all applications from water lines to chemical process.
8 sizes of 10 to 150mm are available.

FEATURES❏ Punched plate reinforced high quality clear PFA is adopted forhigh anti-corrosive, anti-erosion and anti-penetration capability.
❏ High accuracy of ±0.5% of reading.
❏ High speed data processing for quick response. Suitable forbatch process control and for pulsating flow.
❏ The excitation system extendable up to twice the commercial
frequency allows applications to much fluid noise such as slurry.
❏ Blue dot matrix LCD (with backlight) used for the display. Capable of providing 1- to 3-digit display.
❏ Equipped with a quick setup function to readily respond tochanged flow range, pulse rate, etc.

STANDARD SPECIFICATION
General Specification● Excitation: Square wave

● Nominal size

: 10, 15, 25, 40, 50, 80, 100 and 150mm● Measurement: Flow rate, Flow velocity

function● Measuring range

A touch panel system by an infrared sensor allows you to alter theMin. 0 to 0.3m/s

settings without removing the cover of the conversion section.

Max. 0 to 12m/s❏ 10 kHz high-speed pulse output. Capable of responding to short

Flow rate

batch processes.

Min. 0 to 0.085m3/h❏ Current and pulse output, bi-directional measurement, double(Minimum flow at 10mm size)

range, status output, control input...Full function provided inMax. 0 to 763m3/hcompact design.
❏ HART Communication (standard): Flow velocity(Maximum flow at 150mm size)● Protection class: IP66/67 (IEC 60529)● Housing material

Measuring tube : Stainless steel (SS304)
Primary head : Size 10 to 40 mm ; Cast iron *Size 50 to 150 mm ; Carbon steel ** Anti-corrosive painting● Wetted part material

Liner : PFA
Electrode : Hastelloy® C22
Earth ring : Stainless steel (SS316) [Standard]
Earth ring seal : Supplied for size 10 and 15mm only;
FPM/FKM● Converter housing: Aluminium alloy● Painting: Siloxane coating● Color: Grey (Primary head housing/Converterhousing)
Jade green (Converter cover / Terminalcover)

TG-EM145E-3
RevisedJan 2017 K
1st edition Jun 2006

MAGMAX® Compact Electromagnetic Flowmeter EGM1300C: 2 × G1/2 female thread● Cable entry

2 × 1/2 NPT female thread
2 × M20 with watertight glands(Option : Watertight glands for G1/2)● Status output

Open collector output
Rating : Less than 32V DC, 100mA Max.
Contents of output(Option : Number of wiring connection ; 3)

One of the following selectable:: 100 to 230V AC (85 to 250V AC)

1) No status output (Standard factory setting)

24V DC (9 to 31V)

2) Identification of flow direction● Supply frequency: 48 to 63Hz (AC)3) Over range● Power consumption: AC; approx. 22VA

4) ErrorDC; approx. 12W

5) Flow alarm: –40 to +65°C (Fluid temp. ≤120°C)

6) Identification of range (For double range measurement)–50 to +70°C (For storage)7) Empty detection● Supply voltage● Ambient temp.

● Grounding: Grounding resistance must be less than100Ω.

● Control input

Voltage input● Process connection: Wafer type

● Matching flanges: JIS10K/20K, ANSI class 150, DIN PN16/40

Low : 0 to 2.5V DC High : 19 to 32V DC
Contents* Max. Op. Press is 1.6MPa irrespective of

One of the following selectable:flange rating.

1) No control input (Standard factory setting)2) Signal hold3) Signal lock to 0%

Fluid Specification● Temperature: –25 to +120˚C● Pressure: 0Pa (abs) to 1.6MPa● Conductivity: To be more than 5μS/cm(20μS/cm for water flow measurement)

4) Total counter reset
5) Error reset
6) Range selection (For double range measurement)● Description of input and output terminal

Terminal

Switchover byreprogrammingStandard setup

Indication and Output Specification

A (A, A / A–)

Current output–● Indicator : Blue, dot matrix LCD (With backlight)

B (B / B–)

Status output

Control input

C (C / C–)

Status output–

D (D / D–)

Pulse output

Status output

128 × 64 pixels (59 × 31mm)
Indication function :Changeover (2 screens)
One to three lines are displayed at one screen.
Contents of indication ; Flow rate, velocity, total flow,
Conductivity, and coil temperature● Current output : 4 to 20mA DC (Max. 22mA)

Internal power supply :Less than 1000ohms (Load resistance)
External power supply :32V DC (External voltage)● Pulse output● Low flow cutoff

Current output, Pulse output, Indicator (Separate setting
is possible.)
Setting value : 0.0 to 20.0%
Setting value (Standard) :Current output, Pulse output ; ON 1%, OFF 3% FS
Indicator ; Without low flow cutoff● Damping time constantCurrent output, Pulse output, Indicator (Separate settingis possible.)

Open collector output
Rating : Less than 32V DC, 20mA (≤10kHz)
Less than 100mA (≤100Hz)
Pulse rate
2 to 36,000,000 pulse/h (0.00056Hz to 10kHz)
Pulse width
One of the following selectable1) Automatic : Pulse width by which duty factor to be 50%

Setting value : 0.0 to 100.0s
Setting value (Standard) :Current output, Indicator ; 3s
Pulse output ; Damping time constant 0● Isolation of input and output

Each circuit of power supply, electrode input, terminal A,terminal B, terminal C, and terminal D are isolated.

at full scale2) Duty factor 1:1 fixed3) Free setting ; 0.05 to 2000m/s

2TOKYO KEISO CO., LTD.

TG-EM145E-3

MAGMAX® Compact Electromagnetic Flowmeter EGM1300C

Standard Functions

Accuracy (*)● Customer’s free measuring unit setting function● Pulse output :Volume (or mass) and time unit in 7 characters can be cre-For velocity ≥1m/s : ±0.5% of readingated.

For velocity <1m/s : ±0.4% of reading + velocity error of±0.001m/s● Automatic zero adjustment function

Zero adjustment is automatically conducted at “ZERO2.0● Bi-directional flow measurement function

A flow-direction distinction signal is outputted in state
output and current.
● Double range measurement function

Possible range setting range ratio 1 : 20 to 1 : 1.25
(Setting range of low range : 5 to 80% of high range)

Error ±% of reading

ADJUST MODE” (Subject to zero flow)

1.5

1.00.5

Range selection ; By automatic or control input signal● Excitation current frequency switching function

Standard mode :1/6 of supply frequency (Standard)High frequency mode :0.002Velocity (m/s)

3Additional error of ±0.01mA be added onto displey andpulse output.

etc.)(*) Basis condition

The following conditions are indicated by error message;

Fluid

Functional diagnosis :Fluid temperature: 10 to 30ºC

Conductivity: 150μS/cm or more

Supply voltage: Rated voltage ±2%

Ambient temperature: 18 to 28ºC

Coil disconnection, CPU, Memory, Software, Output
module, and Output connection
Status diagnosis :12● Current output :1/50 to 2 of supply frequency (For slurry, pulsating flow,● Self diagnosis function

1: Water

Velocity distribution, Linearity, Magnetizing current /Upstream / Downstream pipe length : 10D / 2D (D: Diameter)

frequency, Empty detection, Over range, Counter overWarm-up time: About 10 minutes

flow, and Power fail detection

Measuring time: 100s

● Memory save function for power failOperation parameters and totalization figures are stored formore than 10 years by EEPROM (Non volatile memory).

FLOW RANGE● Testing function

Simulating output function for current and pulse output isintegrated.
Current output test :Arbitrary output (0.0 to 22.0 mA)
Pulse output test :Arbitrary output (1Hz to 10kHz)
Status output test : On / Off● Touch sensor setting function (Infrared radiation)

By four infrared sensors, data setup from exterior is pos-Possible setting range (m3/h)
Nominal size(mm)Min. (Velocity : 0 to 0.3 m/s) Max. (Velocity : 0 to 12 m/s)
100 to 0.0849

0 to 3.39

150 to 0.191

0 to 7.63

25

0 to 0.531

0 to 21.2

40

0 to 1.36

0 to 54.2

50

0 to 2.13

0 to 84.8

80

0 to 5.43

0 to 217

100

0 to 8.49

0 to 339

150

0 to 19.1

0 to 763

sible without removing cover.
● HART communication

Standard

TG-EM145E-3

TOKYO KEISO CO., LTD.

3

MAGMAX® Compact Electromagnetic Flowmeter EGM1300C

DIMENSIONS
Nominal size: 10/15mm
260
137(26)

82

120

140

140

38 38

1076

Groundingterminal(M5)

H1H2(H)

H3ESC

Earth ring *1WL

Flow direction(Standard)

Nominal size: 25/40mm
260137(26)82120140

140

38 38

1076

Groundingterminal(M5)

H1H2(H)

H3ESC

Earth ring *2W

L

Flow direction(Standard)

Nominal size: 50 to 150mm
260137(26)82120140140

38 38

1076

H2Grounding
terminal
(M5)

H1(H)

H3ESC

W

Earth ring *2

L
Flow direction(Standard)

4

TOKYO KEISO CO., LTD.

TG-EM145E-3

MAGMAX® Compact Electromagnetic Flowmeter EGM1300C
Dimensions (mm)Nominal size(mm)L(H)

H1H2H3W

1068315

69

176

246

4961568315

69176

2464962560325

641912616964084340

70200

270

85

750106302

51

181251

102

980

156

342

65

207

277

130

10100

206

369

78

221

291

156

15150

206

433

110

253

323

220

20Mass (kg)*1 Dimension L for nominal size 10 and 15 mm includes earth rings thickness.
The earth rings are fixed onto the primary head.
*2 Dimension L for nominal size 25 to 150 mm includes earth rings thickness.
The earth rings are not fixed onto the primary head.
They are to be installed between the primary head and connection flanges on installation.
Dimension L is shorter by 6mm in case of the primary head only. (Earth ring thickness = 3 mm × 2)

TG-EM145E-3

TOKYO KEISO CO., LTD.

5MAGMAX® Compact Electromagnetic Flowmeter EGM1300C

ELECTRICAL CONNECTION
Mounting position of LCD display(L–) (L+)

direction.

Version A

Version B

Version C

A– A+

B

B–

C

C–

D

D–N

LIndication part of EGM1300C can be changed according to the flowA

The mounting position will be arranged according to the customer’s
request when ordering.
The arrow indicates standard flow direction.
Flow direction can be changed by data setting.

Protection cover is provided for power supply terminals.
Power supply terminal cover

Description

Terminal
L / L+

L+ (+) • L– (–) (AC power supply / DC power supply)

N / L–Grounding
Terminal

Description

D–
C–+–

Status output

C
B–+–

Status output or Control input

B

Current output(4 to 20mA DC / HART:
Internal power supply)A–
A–

Pulse output or Status output

DA+Polarity

Current output(4 to 20mA DC / HART:
Internal power supply)● Terminal type++Current output(4 to 20mA DC / HART:
External power supply)––+: Plug-in type screw terminal● Connection capacity : 0.5 to 2.5mm26TOKYO KEISO CO., LTD.

TG-EM145E-3

MAGMAX® Compact Electromagnetic Flowmeter EGM1300C

MODEL AND SPECIFICATION CODE
● Nominal size : 10 to 150mm

Model : EGM1300C
Primary head
V N 1 7 4Spec. code
Primary head code V N 1 7(Fixed code)4

N 0 1 C 0Process connection(Fixed code)
Type(Fixed code)
Earth ring (Seal for earth ring)(Fixed code)
Calibration(Fixed code)
Special feature

ConverterSpec. code
Converter code
(Fixed code)
TypePower supply(Fixed code)
Cable entry(Fixed code)
Housing(Fixed code)
Output type
(Fixed code)
Special feature

V N 3 0 4 4
V N 3 0Description

Standard

Wafer type / PFA liner / Hastelloy® C electrodes
568BCEFH

Nominal size

0 0 0 0 0 0 2 0 0 0 0 0

always 4Connection flange size
10 to 15A
10mm
1/2"
15A
15mm
1/2"
25A
25mm
1"
40A
40mm
1-1/2"
50A
50mm
2"
80A
80mm
3"
100A
100mm
4"
150A
150mm
6"
N
Wafer type
0
always 01 C
Compact version (EGC300 Converter)
0always 0
Stainless steel (SS316) (FPM/FKM sealing) For size 10 and 15mm
1
Stainless steel (SS316)
For size 25 to 150mm
KOthers
90 0 0 0
always 0000
0Standard calibration
0 2 0 0 0 0 0
always 0200000(Blank) None/ZInvolved *10

2 0 0 1 2 1 0 0 0 0 0

Description

Standard

Type: EGC300 (Cylindrical housing)4
always 4
4
LCD indication
124V DC (18 to 31V)A
100 to 230V AC (85 to 250V)
0always 0
4
1/2 NPT female thread
5G1/2 female thread
6M20 with watertight glands
2 0 0
always 200
1Standard (Aluminium alloy)
2
always 21Standard (Current output + Pulse output + Control input + Status output)
0 0 0 0 0always 00000(Blank) None/ZInvolved *1*1 In case that special feature are involved, put [/z] at the end of spec. code and specify the details.
It is recommended to consult TOKYO KEISO for such availability before ordering.

TG-EM145E-3

TOKYO KEISO CO., LTD.

7

MAGMAX® Compact Electromagnetic Flowmeter EGM1300C

STANDARD ACCESSORIES● Parameter sheet:1● Instruction manual : 1OPTION● G1/2 watertight glands for cable entry : 1 set [Symbol : WG]● Number of wiring connection : 3 [Symbol : 3G]● No converter data (parameter) setting [Symbol : NS]

We will supply with standard data setting in case you haveno request.
Please set the data of flow range, pulse rate and flow
direction etc. that required operating.

ORDERING INSTRUCTIONSSpecify the following when ordering :1. Model and spec. code
Example : Model : EGM1300C
Primary head spec. code :VN1745N01C01000000200000
Converter spec. code : VN3044A0520012100000
2. Flow range (Full scale) (Unnecessary when option is NS.)
3. Option (Specify if necessary.)
Specify the symbol with reference to the option.
4. Fluid name* Specification is subject to change without notice.

Head Office : Shiba Toho Building, 1 – 7 – 24 Shibakoen, Minato-ku, Tokyo 105 – 8558
Tel : +81-3 – 3431 – 1625 (KEY) ; Fax : +81-3 – 3433 – 4922
e-mail : overseas.sales@tokyokeiso.co.jp ; URL : http://www.tokyokeiso.co.jp
8TG-EM145E-3